Diastolic blood pressure; Systolic blood pressure; Blood pressure reading; Measuring blood pressure ... or your health care provider will wrap the blood pressure cuff snugly around your upper arm. The ...
Nolte, R. W.
The Blood Pressure Measuring System is described. It measures blood pressure by the noninvasive Korotkoff sound technique on a continual basis as physical stress is imposed during experiment M092, Lower Body Negative Pressure, and experiment M171, Metabolic Activity.
Rabbia, F; Del Colle, S; Testa, E; Naso, D; Veglio, F
Blood pressure measurement is the cornerstone for the diagnosis, the treatment and the research on arterial hypertension, and all of the decisions about one of these single aspects may be dramatically influenced by the accuracy of the measurement. Over the past 20 years or so, the accuracy of the conventional Riva-Rocci/Korotkoff technique of blood pressure measurement has been questioned and efforts have been made to improve the technique with automated devices. In the same period, recognition of the phenomenon of white coat hypertension, whereby some individuals with an apparent increase in blood pressure have normal, or reduced, blood pressures when measurement is repeated away from the medical environment, has focused attention on methods of measurement that provide profiles of blood pressure behavior rather than relying on isolated measurements under circumstances that may in themselves influence the level of blood pressure recorded. These methodologies have included repeated measurements of blood pressure using the traditional technique, self-measurement of blood pressure in the home or work place, and ambulatory blood pressure measurement using innovative automated devices. The purpose of this review to serve as a source of practical information about the commonly used methods for blood pressure measurement: the traditional Riva-Rocci method and the automated methods. PMID:17016412
Conclusion: The palpatory method would be very useful where frequent blood pressure measurement are being done manually like in wards, in busy OPD, patient on treadmill and also whenever stethoscope is not available. The blood pressure can be measured in noisy environment too.
The Leg Volume Measuring System is used to measure leg calf girth changes that occur during exposure to lower body negative pressure as a result of pooling of blood and other fluids in the lower extremities.
... the walls of the arteries is called blood pressure. Blood pressure is measured both as the heart contracts, which ... as it relaxes, which is called diastole. Normal blood pressure is considered to be a systolic blood pressure ...
Frese, Ethel M; Fick, Ann; Sadowsky, H. Steven
Vital sign measurement and assessment are important components of the review of systems in a physical therapy examination for individuals with and without documented cardiopulmonary disease. The measurement of blood pressure gives the therapist information regarding the patient's baseline cardiovascular status, response to exercise/activity, and guides exercise prescription. Accurate measurement of blood pressure is critical for making appropriate clinical decisions especially if physical the...
Campbell, N. R.; Chockalingam, A; Fodor, J. G.; McKay, D. W.
The diagnosis of mild hypertension and the treatment of hypertension require accurate measurement of blood pressure. Blood pressure readings are altered by various factors that influence the patient, the techniques used and the accuracy of the sphygmomanometer. The variability of readings can be reduced if informed patients prepare in advance by emptying their bladder and bowel, by avoiding over-the-counter vasoactive drugs the day of measurement and by avoiding exposure to cold, caffeine con...
Pulse pressure in the aorta was evaluated by the measurement of pulse wave velocity (PWV) and blood flow velocity (BFV). PWV reflects the elasticity of the vessel and was determined by a time-of-flight method. BFV was measured by analyzing the change of magnetization decay due to flow in multiecho experiments. If one neglects pulse wave reflections at vascular branch points and flow resistance due to blood viscosity, pulse pressure is proportional to PWV and BFV. Noninvasive MR imaging measurements were obtained in 12 patients, all of whom underwent correlative arterial catheterization. Values varied between 35 and 100 mm Hg. The results demonstrated a high correlation between the two methods
Drawz, Paul E; ABDALLA, MOHAMED; Rahman, Mahboob
Blood pressure has traditionally been measured in the clinic setting using the auscultory method and a mercury sphygmomanometer. Technological advances have led to improvements in measuring clinic blood pressure and allowed for measuring blood pressures outside the clinic. This review outlines various methods for evaluating blood pressure and the clinical utility of each type of measurement. Home blood pressures and 24 hour ambulatory blood pressures have improved our ability to evaluate risk...
Barbé, K.; Van Moer, W.; Lauwers, L.
In this paper, the oscillometric waveform measured by automatic non-invasive blood pressure meters (NIBP) is analyzed by transforming the data from the time domain to the frequency domain. The signal's spectrum of the oscillometric waveform is in current literature badly understood or explored. The only known link between the oscillometric waveform and the blood pressure is the maximum of the oscillometry's envelope equalling the mean arterial pressure (MAP). This link is established under the assumption that the oscillometry is an AM-signal. Unfortunately, computing the MAP is difficult in practice due to the non-sinusoidal nature of the actual measured signals. In this paper, we construct the best AM-signal approximation of the oscillometry and explore its use to compute the MAP.
Hayashi, Satoru; Hayase, Toshiyuki; Shirai, Atsushi; Maruyama, Masaru
In this paper, a simulation model based on the partially pressurized collapsible tube model for reproducing noninvasive blood pressure measurement is presented. The model consists of a collapsible tube, which models the pressurized part of the artery, rigid pipes connected to the collapsible tube, which model proximal and distal region far from the pressurized part, and the Windkessel model, which represents the capacitance and the resistance of the distal part of the circulation. The blood flow is simplified to a one-dimensional system. Collapse and expansion of the tube is represented by the change in the cross-sectional area of the tube considering the force balance acting on the tube membrane in the direction normal to the tube axis. They are solved using the Runge-Kutta method. This simple model can easily reproduce the oscillation of inner fluid and corresponding tube collapse typical for the Korotkoff sounds generated by the cuff pressure. The numerical result is compared with the experiment and shows good agreement. PMID:16995754
Rushton, Melanie; Smith, Joyce
Rationale and key points This article aims to help nurses to measure blood pressure (BP) manually using an aneroid sphygmomanometer. ▶ BP measurement is an essential clinical skill, and nurses must be competent in performing this procedure and taking accurate readings. ▶ Nurses should be aware of manual BP measurement techniques and understand the patient and environmental factors that may result in inaccurate readings that could compromise patient care. ▶ Nurses should regularly undertake manual BP measurement to ensure they remain competent to perform the procedure. Reflective activity Clinical skills articles can help update your practice and ensure it remains evidence based. Apply this article to your practice. Reflect on and write a short account of: 1. How reading this article will change your practice. 2. Further learning needs to extend your professional development. Subscribers can upload their reflective accounts at: rcni.com/portfolio . PMID:26786460
... right away. continue How Do Doctors Measure Blood Pressure? Blood pressure readings are fast and painless. Blood pressure ... same age, height, and gender have lower blood pressure. Blood pressure between 90% and 95% of the normal ...
Rosholm, Jens-Ulrik; Pedersen, Sidsel Arnspang; Matzen, Lars;
Auscultatory measurement using a sphygmomanometer has been the predominant method for clinical estimation of blood pressure, but it is now rapidly being replaced by oscillometric measurement.......Auscultatory measurement using a sphygmomanometer has been the predominant method for clinical estimation of blood pressure, but it is now rapidly being replaced by oscillometric measurement....
Pedersen, Sidsel Arnspang
Abstract Background. Auscultatory measurement using a sphygmomanometer has been the predominant method for clinical estimation of blood pressure, but it is now rapidly being replaced by oscillometric measurement. Objective. To compare blood pressure by auscultatory and oscillometric measurements in...... patients 80 years. Method. 100 patients had blood pressure measured by auscultation with a sphygmomanometer and by an electronic device using the oscillometric method. For each patient the mean of two blood pressures with each method measured within 15 min were compared. Results. The mean age of...... participants was 85.8 years; 55.8% were women. The correlation coeffi cient for systolic blood pressure was 0.88 and for diastolic 0.79. Differences between auscultatory and oscillometric values were less than 10 mmHg in 70.6% of systolic blood pressures and in 83.2% for diastolic. Arrhythmia and hypertension...
Gu, Liangling; Yang, Yongming; Yu, Chengbo; Guo, Qiaohui; Zhu, Gang
Blood pressure reflects a person's health.It is proposed here that the method of detecting blood pressure may be the key to improving the precision of blood pressure measurements. The oscillometric blood pressure measurement technique is widely used in automatic blood pressure measurement instruments correctly. A method of blood pressure measurement by oscillometric method is first presented. In the oscillometric method, the basic principle of the "feature point" method and the "amplitude characteristic ratios" method is also explained and discussed here. A new method of blood pressure measurement, namely the coefficient difference comparative method, is proposed here,which is based on the feature point method and amplitude characteristic ratios method. The method is proved both effective and reliable through the analysis of many cases and clinical tests. Utilizing Visual C++, software for this new and novel method was developed and passed criterion simulation apparatus test. When applied in hospital situation, its error was +/-5%. It is concluded that the oscillometric blood pressure measurement method can provide better means of blood pressure measurements reference for doctors.
<正>The principle for blood pressure measurement using pulse transit time is introduced in this paper.And the math model of synergetics theory is studied in detail.The synergetics theory is applied in the analysis of blood pressure measurement data.The simulation results show that the application of synergetics theory is helpful to judge the normal blood pressure,and the accuracy is up to 80%.
Accurate systolic and diastolic pressure estimation, using automated blood pressure measurement, is difficult to achieve when the transduced signals are contaminated with noise or interference, such as movement artifact. This study presents an algorithm for automated signal quality assessment in blood pressure measurement by determining the feasibility of accurately detecting systolic and diastolic pressures when corrupted with various levels of movement artifact. The performance of the proposed algorithm is compared to a manually annotated reference scoring (RS). Based on visual representations and audible playback of Korotkoff sounds, the creation of the RS involved two experts identifying sections of the recorded sounds and annotating sections of noise contamination. The experts determined the systolic and diastolic pressure in 100 recorded Korotkoff sound recordings, using a simultaneous electrocardiograph as a reference signal. The recorded Korotkoff sounds were acquired from 25 healthy subjects (16 men and 9 women) with a total of four measurements per subject. Two of these measurements contained purposely induced noise artifact caused by subject movement. Morphological changes in the cuff pressure signal and the width of the Korotkoff pulse were extracted features which were believed to be correlated with the noise presence in the recorded Korotkoff sounds. Verification of reliable Korotkoff pulses was also performed using extracted features from the oscillometric waveform as recorded from the inflatable cuff. The time between an identified noise section and a verified Korotkoff pulse was the key feature used to determine the validity of possible systolic and diastolic pressures in noise contaminated Korotkoff sounds. The performance of the algorithm was assessed based on the ability to: verify if a signal was contaminated with any noise; the accuracy, sensitivity and specificity of this noise classification, and the systolic and diastolic pressure
Wagner, Stefan; Kamper, Christina H.; Toftegaard, Thomas Skjødeberg;
Background: Pregnant diabetic patients are often required to self- measure their blood pressure in the waiting room before consulta- tion. Currently used blood pressure devices do not guarantee valid measurements when used unsupervised. This could lead to misdi- agnosis and treatment error. The aim...... of this study was to investigate current use of blood pressure self-measurement in the waiting room in order to identify challenges that could influence the resulting data quality. Also, we wanted to investigate the potential for addressing these challenges with e-health and telemedicine technology...... obtain a reliable blood pressure reading. Results: We found that the patients did not adhere to given instructions when performing blood pressure self-measurement in the waiting room. None of the 81 patients adhered to all six inves- tigated recommendations, while around a quarter adhered to five out of...
Saul, F; Aristidou, Y; Klaus, D; Wiemeyer, A; Lösse, B
Indirectly measured blood pressure at the wrist or upper arm was compared with directly measured values in the aortic arch during routinely performed diagnostic cardiac catheterization in 100 patients (31-80 years, mean 59.3 years, 60% males). The noninvasive measurements were carried out by oscillometric devices, NAiS Blood Pressure Watch for measurements at the wrist, and Hestia OZ80 at the upper arm. Systolic blood pressure measured at the wrist was 4.3 +/- 14.1 mm Hg, and the diastolic value 6.0 +/- 8.9 mm Hg higher than when measured at the aortic arch; the difference was significant in both cases. Correlation coefficients were 0.85 for systolic and 0.71 for diastolic blood pressure. In 16% of the patients the systolic blood pressure at the wrist differed more than +/- 20 mm Hg. The diastolic blood pressure at the wrist measured more than +/- 20 mm Hg higher than in the aorta in 5% of the patients. At the upper arm mean systolic values were not different to the aorta. The diastolic pressure was 9.3 +/- 9.8 mm Hg higher in the aorta than at the upper arm. To verify the accuracy of values measured with the NAiS Blood Pressure Watch compared with the standard technique at the upper arm, sequential measurements were made at wrist and ipsilateral upper arm in the same group of 100 patients. The systolic blood pressure at the left wrist was 3.4 +/- 13.3 mm Hg higher and the diastolic pressure 3.8 +/- 9.5 mm Hg lower than at the upper arm. Only 53% of systolic values lay within a range of +/- 10 mm Hg. The correspondence between wrist and upper arm values was better for diastolic blood pressure, the values differing by less than +/- 10 mm Hg in two-thirds of patients. Self-measurement of arterial blood pressure with an oscillometric device at the wrist can be recommended only in individual cases with a difference of simultaneously measured values at the upper arm of less than +/- 10 mm Hg for systolic and diastolic blood pressures. The standard method for indirectly
Laugesen, Esben; Knudsen, Søren T; Hansen, Klavs W;
Aortic systolic blood pressure (BP) represents the hemodynamic cardiac and cerebral burden more directly than office systolic BP. Whether invasively measured aortic systolic BP confers additional prognostic value beyond office BP remains debated. In this study, office systolic BP and invasively...
Full Text Available Abstract Background Automatic measurement of arterial blood pressure is important, but the available commercial automatic blood pressure meters, mostly based on oscillometry, are of low accuracy. Methods In this study, we present a cuff-based technique for automatic measurement of systolic blood pressure, based on photoplethysmographic signals measured simultaneously in fingers of both hands. After inflating the pressure cuff to a level above systolic blood pressure in a relatively slow rate, it is slowly deflated. The cuff pressure for which the photoplethysmographic signal reappeared during the deflation of the pressure-cuff was taken as the systolic blood pressure. The algorithm for the detection of the photoplethysmographic signal involves: (1 determination of the time-segments in which the photoplethysmographic signal distal to the cuff is expected to appear, utilizing the photoplethysmographic signal in the free hand, and (2 discrimination between random fluctuations and photoplethysmographic pattern. The detected pulses in the time-segments were identified as photoplethysmographic pulses if they met two criteria, based on the pulse waveform and on the correlation between the signal in each segment and the signal in the two neighboring segments. Results Comparison of the photoplethysmographic-based automatic technique to sphygmomanometry, the reference standard, shows that the standard deviation of their differences was 3.7 mmHg. For subjects with systolic blood pressure above 130 mmHg the standard deviation was even lower, 2.9 mmHg. These values are much lower than the 8 mmHg value imposed by AAMI standard for automatic blood pressure meters. Conclusion The photoplethysmographic-based technique for automatic measurement of systolic blood pressure, and the algorithm which was presented in this study, seems to be accurate.
Marcelle Morgana Vieira de Assis
Full Text Available Blood pressure checking and its theoretical knowledge are crucial to obtain reliable data in clinical examination. Although it is considered a simple technique to be learned and applied, it is not fully dealt with at physiotherapy university courses. Therefore, a wide range of mistakes and misinterpretations are likely to threaten treatment quality. This work aims to evaluate the theory background of physiotherapists for blood pressure measurement. From June to October 2002, 55 physiotherapists answered a 20- question multiple-choice questionnaire on general knowledge related to concepts, anatomy and physiology, suitable equipment, indirect measurement, blood pressure values, mistakes and misinterpretation factors related to blood pressure measurement. The results disclose faulty theory concepts in the sample studied, indicating the need of deeper approach to this complex theme during course and ongoing updating of professionals.
The problem: To develop an instrument that will provide an external (indirect) measurement of arterial blood pressure in the form of an easily interpreted graphic trace that can be correlated with standard clinical blood-pressure measurements. From sphygmograms produced by conventional sphygmographs, it is very difficult to differentiate the systolic and diastolic blood-pressure pulses and to correlate these indices with the standard clinical values. It is nearly impossible to determine these indices when the subject is under physical or emotional stress. The solution: An electronic blood-pressure system, basically similar to conventional ausculatory sphygmomanometers, employing a standard occluding cuff, a gas-pressure source, and a gas-pressure regulator and valve. An electrical output transducer senses cuff pressure, and a microphone positioned on the brachial artery under the occluding cuff monitors the Korotkoff sounds from this artery. The output signals present the conventional systolic and diastolic indices in a clear, graphical display. The complete system also includes an electronic timer and cycle-control circuit.
HuangMengcai; GuZhong; HangWenjing; ZhongQuan; TangFuyong
Describe some new fully automatic instruments for the measurements of the blood capillary pressure (Pcap) and arterial elastic properties in human fingers using a photoelectric plethysmographic technique, With these instruments, the value of Pcap was in good agreement with those reported by other investigators, the arterial elastic properties in human fingers have been successfully measured. The measurements of Pcap and arterial elasticity are now required in clinics because they provide useful and important information for evaluating vascular haemodynamics.
Lassen, N A; Krähenbühl, B; Hirai, M
A miniaturized blood pressure cuff made of plastic material and applicable to fingers and toes is described. The cuff was compared to rubber cuffs and to bladder-free cuffs. It was found to be more reliable than the former type and much easier to use than the latter type. It is recommended for us...... in conjunction with a mercury-in-Silastic strain gauge for routine measurement of digital blood pressure and blood flow in patients with arterial disease....
Vaur; Dubroca; Dutrey-Dupagne; Genès; Chatellier; Bouvier-d'Yvoire; Elkik; Ménard
OBJECTIVE: To compare the effects on office blood pressure and home blood pressure of placebo and active drug administration. DESIGN: After a 2-week wash-out period, patients with mild-to-moderate hypertension entered a 2-week single-blind placebo period and then a 4-week double-blind period. Patients were randomly assigned to be administered either 2 mg trandolapril once daily or its placebo in a 2:1 proportion. Office blood pressure was measured by a physician at the end of each period, using a mercury sphygmomanometer (mean of three consecutive measurements). Home blood pressure was measured during the last week of each period according to standard procedure carefully taught to each patient by the physician. Compliance was checked by using electronic pill boxes. RESULTS: Data for 34 of the 44 patients who entered the study were eligible for analysis. Baseline systolic blood pressure/diastolic blood pressure were significantly (P = 0.0001/P = 0.0001) higher for office blood pressure (161/101 mmHg) than they were for home blood pressure (145/93 mmHg). There was no statistically significant difference between the placebo and active-treatment groups at baseline. During the single-blind period, blood pressures measured at the office and at home did not change significantly. Office blood pressure decreased by 2.7 +/- 10 mmHg for systolic blood pressure and by 0.5 +/- 4 mmHg for diastolic blood pressure whereas home blood pressure increased by 0.8 +/- 6 mmHg for systolic blood pressure and by 0.7 +/- 4 mmHg for diastolic blood pressure. During the double-blind period, office blood pressure fell significantly with trandolapril treatment (systolic by 10.2 +/- 12 mmHg, diastolic by 8.3 +/- 6 mmHg; P = 0.0005/0.0001, versus single-blind placebo period) but this decrease was not significantly different (P = 0.45/0.92) from the fall in members of the placebo group (systolic by 6.9 +/- 9 mmHg, diastolic by 8.0 +/-6 mmHg; P = 0.04/0.002, versus single-blind placebo period
Nielsen, S L; Lassen, N A
A double-inlet plastic cuff was designed for local cooling and systolic blood pressure measurement on the middle phalanx of the fingers. With a tourniquet on the proximal phalanx of one finger, cooling for 5 min made the digital artery temperature equal the skin temperature. The difference between...... the systolic pressure in a control finger and in the cooled finger give the reopening pressure in the digital arteries. At 30, 25, 20, 15, and 10 degrees C, respectively the percent decrease of the finger pressure was 0.2 (0.2), 1.5 (2.5), 8.5 (3.7), 11.4 (3.4), and 15.3 (3.1) in normal young women...
Hogan, Jennifer L
OBJECTIVE: The accurate measurement of blood pressure requires the use of a large cuff in subjects with a high mid-arm circumference (MAC). This prospective study examined the need for a large cuff during pregnancy and its correlation with maternal obesity. METHODS: Maternal body mass index (BMI), fat mass, and MAC were measured. RESULTS: Of 179 women studied, 15.6% were obese. With a BMI of level 1 obesity, 44% needed a large cuff and with a BMI of level 2 obesity 100% needed a large cuff. CONCLUSION: All women booking for antenatal care should have their MAC measured to avoid the overdiagnosis of pregnancy hypertension.
Blood pressure monitoring is an essential component of neonatal intensive care. We compared invasive and noninvasive (Dinamap, Marquette, and Dash) recordings in newborns and also noninvasive values obtained from upper and lower limbs. Infants\\' blood pressure was recorded every 6 hours for 72 hours using three noninvasive devices and compared with invasive readings taken simultaneously. Twenty-five babies were enrolled in the study, with birth weights of 560 to 4500 g and gestation 24 + 1 to 40 + 5 weeks. Three hundred thirty-two recordings were obtained. Comparison between invasive and noninvasive readings revealed that all three noninvasive monitors overread mean blood pressure. There was no significant difference between the cuff recordings obtained from the upper or lower limbs. All three noninvasive devices overestimated mean blood pressure values compared with invasive monitoring. Clinicians may be falsely reassured by noninvasive monitoring. Mean blood pressure values obtained from the upper and lower limb are similar.
... that is healthy for you. Checking Your Blood Pressure Your blood pressure can be measured at many places, including: ... Alternative Names Controlling hypertension Images Taking your blood pressure at home Blood pressure check Low sodium diet References American Diabetes ...
Sandager, Puk; Lindahl, Camilla; Schlütter, Jacob Mørup;
The importance of accurate measurement of blood pressure in the screening and management of hypertension during pregnancy is well established. Blood pressure levels can be measured manually by healthcare staff or by using a blood pressure self-measurement device, either at home or in the clinic. In...... blood pressure self-measurement process. Preliminary results indicate that such active and context-aware guidance leads to more reliable measurements by inhibiting non-adherent patient behavior...
Chin, K Y; Panerai, R B
Arterial volume clamping uses external compression of an artery to provide continuous non-invasive measurement of arterial blood pressure. It has been assumed that mean arterial pressure (MAP) corresponds to the point where unloading leads to the maximum oscillation of the arterial wall as reflected by photoplethysmogram (PPG), an assumption that has been challenged. Five subjects were recruited for the study (three males, mean age (SD) = 32 (15) years). The PPG waveform was analysed to identify the relationship between the external compressing pressure, PPG pulse amplitude and MAP. Two separate tests were carried out at compression step intervals of 10 mmHg and 2 mmHg, respectively. No significant differences were found between the two tests. The bias between the compressing pressure and the MAP was -4.7 ± 5.63 mmHg (p < 0.001) showing a normal distribution. Further research is needed to identify optimal algorithms for estimation of MAP using PPG associated with arterial compression. PMID:25429784
Karamanou, Marianna; Papaioannou, Theodore G; Tsoucalas, Gregory; Tousoulis, Dimitris; Stefanadis, Christodoulos; Androutsos, George
The profound observations of William Harvey (1578-1657), in blood circulation and the progress of physical science laid the foundation for the development of the Iatrophysical School that contributed to the evolution of clinical sphygmomanometry. The pioneer work of Reverend Stephen Hales (1677-1761) demonstrated the dynamics of the vascular system. One century later the French physician Jean-Léonard-Marie Poiseuille (1797-1867) invented a U-tube mercury manometer and in 1860 the physiologist Etienne- Jules Marey (1830-1904) devised the first portable sphygmograph for recording the pulse wave. The non-invasive techniques of blood pressure measurement were completed by Scipione Riva-Rocci (1896-1937) sphygmomanometer and the description of "Korotkov sounds" by the Russian surgeon Nikolai- Sergeyevich Korotkov (1874-1920). PMID:25341864
... blood pressure with the development of a practical method to measure it. Physicians began to note associations between hypertension and risk of heart failure, stroke, and kidney failure. Although scientists had yet to prove that lowering blood pressure ...
Maria Pauline; Avadhany, Sandhya T.; K.N. Maruthy
Background: Non invasive, simple and economical instrument to measure blood pressure in r365-ats is important in cardiovascular research. Methods: Systolic blood pressure measuring instrument was fabricated using a tail cuff, photoplethysmograph, pressure transducer and PC with Biopac Software for recording. Tail cuff was used to occlude the tail artery, photoplethysmograph picked the blood flow pulses in the rat tail and the pressure transducer measured the cuff pressure and converted it int...
Amankwah, Ernest; Campbell, Norman R C; Maxwell, Colleen; Onysko, Jay; Quan, Hude
Regular blood pressure (BP) measurements are required to identify people with hypertension and to optimally manage their cardiovascular risk. Analyses of data from the 2000-2001 Canadian Community Health Survey showed that most adult Canadians have had a BP assessment in the previous 2 years and few have never had one. Large numbers of persons without BP recordings were observed, however, among persons who were younger, were male, who did not have either a regular doctor or physician contact in the previous year, who were recent immigrants or visible minorities (nonwhite and non-Aboriginal), and who spoke neither French nor English. Common reasons reported for not having a BP assessment included believing it was not necessary and simply not getting around to it. Education programs targeting those at risk as well as more convenient BP screening may improve awareness and testing. PMID:18046100
Leblanc, Marie-Ève; Cloutier, Lyne; Poirier, Paul
For blood pressure assessment, it has been reported that forearm blood pressure measurement appears to be as valid as an upper-arm blood pressure measurement method in individuals with severe obesity when correlated to the intra-arterial method, considered as the gold standard. Sensitivity, specificity, and predictive values for the presence of systemic hypertension in 25 severely obese patients from 352 blood pressure measurements were calculated. The sensitivity (0.98) and the positive predictive value (0.93) for hypertension on forearm blood pressure measurements are excellent, indicating that the forearm approach is a promising alternative to systemic hypertension diagnosis in severe obesity. PMID:25757220
M. F. VALIM; Barros, S.
Blood pressure is the force of blood against artery walls. It is measured in millimeters of mercury (mm Hg) and recorded as two numbers: systolic pressure (as the heart contracts) over diastolic pressure (as the heart relaxes between beats). High blood pressure (hypertension) is defined as chronically elevated high blood pressure, with systolic blood pressure (SBP) of 140 mm Hg or greater, and diastolic blood pressure (DBP) of 90 mm Hg or greater. High blood pressure ...
McGrath, Barry P
Home blood pressure monitoring is the self-measurement of blood pressure by patients. In the diagnosis and management of high blood pressure it is complementary to 24-hour ambulatory blood pressure monitoring and clinic blood pressure measurements. Home monitoring can also help to identify white-coat and masked hypertension.
Kok, R.H.; Beltman, F.W.; Terpstra, W.F.; Smit, A.J.; May, J.F.; de Graeff, P.A.; Meyboom-de Jong, B.
OBJECTIVE: To evaluate the reproducibility and relationship with left ventricular mass index of home blood pressure in comparison with ambulatory and office blood pressures. METHODS: We measured home, ambulatory and office blood pressures of 84 previously untreated hypertensive patients, aged 60-74
There are few reports on the relationship of blood pressure with cognitive function in maintenance dialysis patients. The Cognition and Dialysis Study is an ongoing investigation of cognitive function and its risk factors in six Boston area hemodialysis units. In this analysis, we evaluated the rela...
Hoffmann-Petersen, N; Lauritzen, T; Bech, J N;
of the measurements and subsequent communication by telephone or E-mail. In the control group, patients received usual care. Primary outcome was reduction in daytime ambulatory blood pressure measurements (ABPM) from baseline to 3 months' follow-up. RESULTS: In both groups, daytime ABPM decreased significantly......: Participants (n = 356) were recruited from a prevalence study among citizens aged 55-64 years in the municipality of Holstebro, Denmark. The study was a randomised, controlled, unblinded 3 months trial. In the intervention group, antihypertensive treatment was based on TBPM with transmission....... The decrease in daytime ABPM in the intervention group was systolic/diastolic, -8 ± 12/-4 ± 7 mmHg. This did not differ significantly from the control group's -8 ± 13/-4 ± 8 mmHg. An equal number of participants obtained normal daytime ABPM, in the intervention group 17% (31/175) versus control 21% (37...
Background: The use of sphygmomanometers may lead to problems in investigations on health of young people. The purpose of this paper is to present the validation of the blood pressure (BP) collected during the survey 'Youth Heart Health' in Hainaut by using second sample of young people in Hainaut for which BP was measured by a manual taking of tension and by an electronic device. Methods: Validation was done with a control sample of 343 young with five successive BP measures: twice with the mercury sphygmomanometer and three with DXL. We compared the manual and the electronic measures in order to study the correlation between the two methods. The control sample was used in order to compare the BP measurements with the results of the survey on the health of young people in Hainaut. Results: The differences between manual systolic BP and Dinamap measures are significant (differences in averages 3.6 mmHg; d.s. 7.8; 95% CI: 2.8-4.4 mmHg; p < 0.001) and the regression coefficient is -0.015. Diastolic BP is significantly higher with the manual method than with Dinamap (differences in means: 8.2 mmHg; d.s. 7.0; 95% CI: 7.4-8.9 mmHg; p < 0.001) and the coefficient of regression is 0.096. We observe a difference in BP between the general survey and the control group (10.3 mmHg for systolic manual control and of 10.9 mmHg for systolic electronic control; 3.3 mmHg for diastolic manual control and of 11.5 mmHg for diastolic electronic control). Conclusions: The values of BP of the 'Youth Heart Health' are significantly higher. These observations indicate the difficulties in the choice of the tool for measurements of BP in epidemiological investigations in the teenagers. In order to decrease skews of observations in the measurement of BP, it is recommended to use a valid electronic instrument. Dinamap XL is an instrument of choice in such studies of young people. Environmental and organisational factors may also explain the observed difference, which means that survey
Verberk, Willem J; Kroon, Abraham A.; Jongen-Vancraybex, Heidi A; Peter W de Leeuw
Purpose To review the literature on home blood pressure measurement (HBPM), to examine its validity and applicability for clinical practice and to provide recommendations regarding HBPM assessment. Findings HBPM can eliminate the white coat effect and offers the possibility to obtain multiple measurements under standardized conditions, which increases knowledge of overall blood pressure value. Although it is not entirely capable of replacing ambulatory blood pressure measurement (ABPM), HBPM ...
... your doctor prescribes it, medicine. What Is Blood Pressure? Blood pressure is the force of blood flow inside ... Will I Know if I Have High Blood Pressure? High blood pressure is a silent problem — you won't ...
Niiranen, T. J.; Asayama, K.; Thijs, L; Johansson, J K; Ohkubo, T.; Kikuya, M; Boggia, J.; Hozawa, A.; Sandoya, E.; Stergiou, G. S.; Tsuji, I; Jula, A. M.; Imai, Y.; Staessen, J. A.
The lack of outcome-driven operational thresholds limits the clinical application of home blood pressure (BP) measurement. Our objective was to determine an outcome-driven reference frame for home BP measurement. We measured home and clinic BP in 6470 participants (mean age, 59.3 years; 56.9% women; 22.4% on antihypertensive treatment) recruited in Ohasama, Japan (n=2520); Montevideo, Uruguay (n=399); Tsurugaya, Japan (n=811); Didima, Greece (n=665); and nationwide in Finland (n=2075). In mul...
Chow, P K; Ng, T H; Heng, D; Mack, P O
The pig is a commonly used large animal model in experimental studies. Few non-invasive techniques exist however for the measurement of blood pressure in the porcine model. This study evaluates the novel use of the easily available neonatal blood pressure cuff for measuring blood pressure in the pig. Six Yorkshire pigs were used for the study. Blood pressure measurements obtained by the application of neonatal blood pressure cuff (Hewlett Packard) around the base of the tail were compared with results obtained from intra-arterial measurements in the normotensive range as well as in experimentally created hypertensive (intravenous dopamine) and hypotensive (hypovolaemic shock) ranges. Results of the two techniques are closely correlated (Pearson's coefficient = 0.95, 0.97, 0.90). Systematic bias was however detected at the extremes of hypertensive and hypotensive blood pressure. Analysis of the limits of agreement (method of Bland and Altman) showed that neonatal blood pressure cuff measurements fall within--2 to 2.5 mmHg of the readings obtained from the invasive technique (95% confidence interval). The neonatal blood pressure cuff technique is a good substitute for the standard invasive intra-arterial measurement of blood pressure in the pig model. PMID:10374020
Gholamhosseini, Hamid; Meintjes, Andries; Baig, Mirza; Linden, Maria
The increasing availability of low cost and easy to use personalized medical monitoring devices has opened the door for new and innovative methods of health monitoring to emerge. Cuff-less and continuous methods of measuring blood pressure are particularly attractive as blood pressure is one of the most important measurements of long term cardiovascular health. Current methods of noninvasive blood pressure measurement are based on inflation and deflation of a cuff with some effects on arteries where blood pressure is being measured. This inflation can also cause patient discomfort and alter the measurement results. In this work, a mobile application was developed to collate the PhotoPlethysmoGramm (PPG) waveform provided by a pulse oximeter and the electrocardiogram (ECG) for calculating the pulse transit time. This information is then indirectly related to the user's systolic blood pressure. The developed application successfully connects to the PPG and ECG monitoring devices using Bluetooth wireless connection and stores the data onto an online server. The pulse transit time is estimated in real time and the user's systolic blood pressure can be estimated after the system has been calibrated. The synchronization between the two devices was found to pose a challenge to this method of continuous blood pressure monitoring. However, the implemented continuous blood pressure monitoring system effectively serves as a proof of concept. This combined with the massive benefits that an accurate and robust continuous blood pressure monitoring system would provide indicates that it is certainly worthwhile to further develop this system. PMID:27225558
Full Text Available Introduction. Home blood pressure monitoring has several advantages over blood pressure monitoring at a physician's office, and has become a useful instrument in the management of hypertension. Objective. To explore the rate and characteristics of patients who measure their blood pressure at home. Methods. A sample of 2,752 patients with diagnosis of essential arterial hypertension was selected from 12596 consecutive office visitors. Data of 2,639 patients was appropriate for analysis. The data concerning home blood pressure measurement and patients' characteristics were obtained from the patients' case histories. Results 1,835 (69.5% out of 2,639 patients measured their blood pressure at home. 1,284 (70.0% of home blood pressure patients had their own blood pressure measurement device. There were some important differences between these two groups: home blood pressure patients were more frequently male, of younger age, better educated, from urban area, mostly non-smokers, more likely to have diabetes mellitus and ischemic heart disease and had higher number of co-morbidities and were on other drugs beside antihypertensive medication. Using the logistic regression analysis we found that the most powerful predictors of home blood pressure monitoring had higher education level than primary school OR=1.80 (95% CI 1.37-2.37, non-smoking OR=2.16 (95% CI 1.40-3.33 and having a physician in urban area OR=1.32 (95% CI 1.02-1.71. Conclusion. Home blood pressure monitoring is popular in Slovenia. Patients who measured blood pressure at home were different from patients who did not. Younger age, higher education, non-smoking, having a physician in urban area and longer duration of hypertension were predictors of home blood pressure monitoring.
... What is the AHA recommendation for healthy blood pressure? This blood pressure chart reflects categories defined by the American ... unusually low blood pressure readings. How is high blood pressure diagnosed? Your healthcare providers will want to get ...
... Connected Home » High Blood Pressure Heath and Aging High Blood Pressure What Is Blood Pressure? Do ... high blood pressure increases as you get older. Gender. Before age 55, men have a greater chance ...
Current noninvasive blood pressure (BP) measurement methods, such as the oscillometric method, estimate the systolic and diastolic blood pressure (SBP and DBP) at two random instants in time and do not take into account the natural variability in BP. The standard for automated BP devices sets a maximum allowable system error of ±5 mmHg, even though natural BP variability often exceeds these limits. This paper proposes a new approach using simultaneous recordings of the oscillometric and continuous arterial pulse waveforms to augment the conventional noninvasive measurement by providing (1) the mean SBP and DBP over the measurement interval and the associated confidence intervals of the mean, (2) the standard deviation of SBP and DBP over the measurement interval, which indicates the degree of fluctuation in BP and (3) an indicator as to whether or not the oscillometric reading is an outlier. Recordings with healthy subjects demonstrate the potential utility of this approach to characterize BP, to detect outlier measurements, and that it does not suffer from bias relative to the conventional oscillometric method. (paper)
Komine Hidehiko; Asai Yoshiyuki; Yokoi Takashi; Yoshizawa Mutsuko
Abstract Background Arterial stiffness is a major contributor to cardiovascular diseases. Because current methods of measuring arterial stiffness are technically demanding, the purpose of this study was to develop a simple method of evaluating arterial stiffness using oscillometric blood pressure measurement. Methods Blood pressure was conventionally measured in the left upper arm of 173 individuals using an inflatable cuff. Using the time series of occlusive cuff pressure and the amplitudes ...
Nexøe, Jørgen; Damsbo, Bent; Lund, Jens Otto;
BACKGROUND: Low ankle brachial index (ABI) is a sensitive measure of 'burden' of atherosclerosis, indicating cardiovascular risk of the asymptomatic patient. Conventionally, ABI values......BACKGROUND: Low ankle brachial index (ABI) is a sensitive measure of 'burden' of atherosclerosis, indicating cardiovascular risk of the asymptomatic patient. Conventionally, ABI values...
... cuff, or an automatic device may measure the pressure. Blood pressure is given as 2 numbers. The first ... described above. High Blood Pressure NIHSeniorHealth: High Blood Pressure High Blood Pressure and Kidney Disease Heart and Vascular Diseases ...
Hirai, M; Nielsen, S L; Lassen, N A
The aim of the present paper was to study the methodological problems involved in measuring systolic blood pressure in all five fingers by the strain gauge technique. In 24 normal subjects, blood pressure at the proximal phalanx of finger I and both at the proximal and the intermediate phalanx of...
... pressure and should be taken seriously. Over time, consistently high blood pressure weakens and damages ... of landmark NIH blood pressure study confirm that lower blood pressure target can reduce ...
Methodology and technology for peripheral and central blood pressure and blood pressure variability measurement: current status and future directions - Position statement of the European Society of Hypertension Working Group on blood pressure monitoring and cardiovascular variability.
Stergiou, George S; Parati, Gianfranco; Vlachopoulos, Charalambos; Achimastos, Apostolos; Andreadis, Emanouel; Asmar, Roland; Avolio, Alberto; Benetos, Athanase; Bilo, Grzegorz; Boubouchairopoulou, Nadia; Boutouyrie, Pierre; Castiglioni, Paolo; de la Sierra, Alejandro; Dolan, Eamon; Head, Geoffrey; Imai, Yutaka; Kario, Kazuomi; Kollias, Anastasios; Kotsis, Vasilis; Manios, Efstathios; McManus, Richard; Mengden, Thomas; Mihailidou, Anastasia; Myers, Martin; Niiranen, Teemu; Ochoa, Juan Eugenio; Ohkubo, Takayoshi; Omboni, Stefano; Padfield, Paul; Palatini, Paolo; Papaioannou, Theodore; Protogerou, Athanasios; Redon, Josep; Verdecchia, Paolo; Wang, Jiguang; Zanchetti, Alberto; Mancia, Giuseppe; O'Brien, Eoin
Office blood pressure measurement has been the basis for hypertension evaluation for almost a century. However, the evaluation of blood pressure out of the office using ambulatory or self-home monitoring is now strongly recommended for the accurate diagnosis in many, if not all, cases with suspected hypertension. Moreover, there is evidence that the variability of blood pressure might offer prognostic information that is independent of the average blood pressure level. Recently, advancement in technology has provided noninvasive evaluation of central (aortic) blood pressure, which might have attributes that are additive to the conventional brachial blood pressure measurement. This position statement, developed by international experts, deals with key research and practical issues in regard to peripheral blood pressure measurement (office, home, and ambulatory), blood pressure variability, and central blood pressure measurement. The objective is to present current achievements, identify gaps in knowledge and issues concerning clinical application, and present relevant research questions and directions to investigators and manufacturers for future research and development (primary goal). PMID:27214089
Pan, Fan; Zheng, Dingchang; He, Peiyu; Murray, Alan
Abstract This study aimed to investigate the effect of stethoscope position and contact pressure on auscultatory blood pressure (BP) measurement. Thirty healthy subjects were studied. Two identical stethoscopes (one under the cuff, the other outside the cuff) were used to simultaneously and digitally record 2 channels of Korotkoff sounds during linear cuff pressure deflation. For each subject, 3 measurements with different contact pressures (0, 50, and 100 mm Hg) on the stethoscope outside th...
Vachon, Catherine; Belanger, Marie C; Burns, Patrick M
Two non-invasive blood pressure (NIBP) devices (oscillometry and Doppler) were compared to invasive blood pressure using a Bland-Altman analysis, in anesthetized and conscious dogs. When considering the systolic arterial pressure only during general anesthesia, both NIBP devices slightly underestimated the systolic arterial blood pressure however the precision and the limits of agreement for the Doppler were of a greater magnitude. This indicates a worse clinical performance by the Doppler. The performance of both NIBP devices deteriorated as measured in conscious animals. In general, for the oscillometric device, determination of invasive diastolic and mean arterial pressures was better than the invasive systolic arterial pressure. Overall, the oscillometric device satisfied more of the criteria set by the American College of Veterinary Internal Medicine consensus statement. Based upon these results, the oscillometric device is more reliable than the Doppler in the determination of blood pressure in healthy medium to large breed dogs. PMID:24924217
... Blood Pressure Tools & Resources Stroke More Medications and Blood Pressure Updated:Jul 6,2016 When your blood pressure ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
Full Text Available Background: Non invasive, simple and economical instrument to measure blood pressure in r365-ats is important in cardiovascular research. Methods: Systolic blood pressure measuring instrument was fabricated using a tail cuff, photoplethysmograph, pressure transducer and PC with Biopac Software for recording. Tail cuff was used to occlude the tail artery, photoplethysmograph picked the blood flow pulses in the rat tail and the pressure transducer measured the cuff pressure and converted it into analog voltage. PPG signals were converted into voltage and amplified by the recording system with two channel amplifiers and in addition it also amplified analog voltage converted by the pressure transducer. Results: Calibration of the instrument entailed a simple Bland Altman’s plot of pressure recorded as voltage changes against pressure changes in the mercury sphygmomanometer which is considered as gold standard. A regression value of r=0.9 and p=0.00 was obtained. A pilot study was done on ten female rats. Three blood pressure readings were taken on two occasions. Between - animal variation of BP was 123±7 (mean SD, CV=5.9% and within - animal variation of BP was 120 ±7, CV=5.5%. Conclusion: The tail cuff and PPG based technique to measure systolic blood pressure in rats is simple, economic, accurate and reliable.
Widdop, R E; Li, X C
1. The non-invasive measurement of tail cuff systolic blood pressure in conscious rats is routinely used in long-term cardiovascular studies. There are a number of commercially available tail cuff systems, however, these apparatus are generally expensive and are dedicated for single-task operations. In the present study, a simple method for measuring systolic blood pressure, which requires only minor modifications to the existing hardware found in most cardiovascular laboratories, is described. 2. Systolic blood pressure measurements were made in the conventional manner by determining the systolic blood pressure which coincided with the restoration of the caudal artery pulse. This was achieved by using an inexpensive piezo-electric pulse transducer to detect the pulse, and this was coupled to a standard data-acquisition system (MacLab, ADInstruments) normally set up to record blood pressure. This method was compared with another established tail cuff method, as well as with direct intra-arterial recordings. 3. It was found that the results obtained using both tail cuff systems were in good agreement when systolic blood pressure was measured in Wistar-Kyoto rats and spontaneously hypertensive rats. In addition, systolic blood pressure was measured over 4 weeks in 2K1C rats and sham-operated rats, with both tail cuff methods producing similar results, which were not significantly different from direct intra-arterial recordings in the same animals. 4. Thus, in the present study, with only minor modifications, the same equipment was used for both direct and indirect determinations of systolic blood pressure. This situation differs from other conventional tail cuff systems since these items are designed for a single purpose. Therefore, the current method using piezo-electric sensor/MacLab-technology should be viewed as a relatively simple, flexible and cheap alternative method to measure tail cuff systolic blood pressure in conscious rats. PMID:9337632
de Morais, Polyana Resende Silva; Sousa, Ana Luiza Lima; Jardim, Thiago de Souza Veiga; Nascente, Flávia Miquetichuc Nogueira; Mendonça, Karla Lorena; Povoa, Thaís Inácio Rolim; Carneiro, Carolina de Souza; Ferreira, Vanessa Roriz; de Souza, Weimar Kunz Sebba Barroso; Jardim, Paulo César Brandão Veiga
Background Blood pressure is directly related to body mass index, and individuals with increased waist circumference have higher risk of developing hypertension, insulin resistance, and other metabolic changes, since adolescence. Objective to evaluate the correlation of blood pressure with insulin resistance, waist circumference and body mass index in adolescents. Methods Cross-section study on a representative sample of adolescent students. One group of adolescents with altered blood pressure detected by casual blood pressure and/or home blood pressure monitoring (blood pressure > 90th percentile) and one group of normotensive adolescents were studied. Body mass index, waist circumference were measured, and fasting glucose and plasma insulin levels were determined, using the HOMA-IR index to identify insulin resistance. Results A total of 162 adolescents (35 with normal blood pressure and 127 with altered blood pressure) were studied; 61% (n = 99) of them were boys and the mean age was 14.9 ± 1.62 years. Thirty-eight (23.5%) adolescents had altered HOMA-IR. The group with altered blood pressure had higher values of waist circumference, body mass index and HOMA-IR (p<0.05). Waist circumference was higher among boys in both groups (p<0.05) and girls with altered blood pressure had higher HOMA-IR than boys (p<0.05). There was a significant moderate correlation between body mass index and HOMA-IR in the group with altered blood pressure (ρ = 0.394; p < 0.001), and such correlation was stronger than in the normotensive group. There was also a significant moderate correlation between waist circumference and HOMA-IR in both groups (ρ = 0.345; p < 0.05). Logistic regression showed that HOMA-IR was as predictor of altered blood pressure (odds ratio - OR = 2.0; p = 0.001). Conclusion There was a significant association of insulin resistance with blood pressure and the impact of insulin resistance on blood pressure since childhood. The correlation and association between
... you are at risk for: Bleeding from the aorta, the large blood vessel that supplies blood to ... tests Blood pressure check Blood pressure References American Diabetes Association. Standards of medical care in diabetes-2015 ...
... normal blood pressure 140/90 or higher is high blood pressure Between 120 and 139 for the top number, ... prehypertension. Prehypertension means you may end up with high blood pressure, unless you take steps to prevent it. High ...
National High Blood Pressure Education Program Working Group on High Blood Pressure in Children and Adolescents. The fourth report on the diagnosis, evaluation, and treatment of high blood pressure in children and adolescents. Pediatrics . ...
... Heart Disease Cholesterol Salt Million HeartsÂ® WISEWOMAN High Blood Pressure Facts Recommend on Facebook Tweet Share Compartir ... facts about high blood pressure [PDF-255K] . High Blood Pressure in the United States About 70 million ...
Nielsen, P E; Bell, G; Lassen, N A
The systolic blood pressure on the finger, toe, and ankle has been measured by a strain gauge technique in 10 normal subjects aged 17-31 years and 14 normal subjects aged 43-57 years. The standard deviation in repeated measurements lies between 2 and 6 mm Hg. The finger pressure in the younger...
Full Text Available Abstract Background Arterial stiffness is a major contributor to cardiovascular diseases. Because current methods of measuring arterial stiffness are technically demanding, the purpose of this study was to develop a simple method of evaluating arterial stiffness using oscillometric blood pressure measurement. Methods Blood pressure was conventionally measured in the left upper arm of 173 individuals using an inflatable cuff. Using the time series of occlusive cuff pressure and the amplitudes of pulse oscillations, we calculated local slopes of the curve between the decreasing cuff pressure and corresponding arterial volume. Whole pressure-volume curve was derived from numerical integration of the local slopes. The curve was fitted using an equation and we identified a numerical coefficient of the equation as an index of arterial stiffness (Arterial Pressure-volume Index, API. We also measured brachial-ankle (baPWV PWV and carotid-femoral (cfPWV PWV using a vascular testing device and compared the values with API. Furthermore, we assessed carotid arterial compliance using ultrasound images to compare with API. Results The slope of the calculated pressure-volume curve was steeper for compliant (low baPWV or cfPWV than stiff (high baPWV or cfPWV arteries. API was related to baPWV (r = -0.53, P r = -0.49, P r = 0.32, P Conclusions These results suggest that our method can simply and simultaneously evaluate arterial stiffness and blood pressure based on oscillometric measurements of blood pressure.
The most common method of clinical measurement of arterial blood pressure is by means of the cuff sphygmomanometer. This instrument has provided fundamental quantitative information on arterial pressure in individual subjects and in populations and facilitated estimation of cardiovascular risk related to levels of blood pressure obtained from the brachial cuff. Although the measurement is taken in a peripheral limb, the values are generally assumed to reflect the pressure throughout the arterial tree in large conduit arteries. Since the arterial pressure pulse becomes modified as it travels away from the heart towards the periphery, this is generally true for mean and diastolic pressure, but not for systolic pressure, and so pulse pressure. The relationship between central and peripheral pulse pressure depends on propagation characteristics of arteries. Hence, while the sphygmomanometer gives values of two single points on the pressure wave (systolic and diastolic pressure), there is additional information that can be obtained from the time-varying pulse waveform that enables an improved quantification of the systolic load on the heart and other central organs. This topical review will assess techniques of pressure measurement that relate to the use of the cuff sphygmomanometer and to the non-invasive registration and analysis of the peripheral and central arterial pressure waveform. Improved assessment of cardiovascular function in relation to treatment and management of high blood pressure will result from future developments in the indirect measurement of arterial blood pressure that involve the conventional cuff sphygmomanometer with the addition of information derived from the peripheral arterial pulse. (topical review)
Avolio, Alberto P; Butlin, Mark; Walsh, Andrew
The most common method of clinical measurement of arterial blood pressure is by means of the cuff sphygmomanometer. This instrument has provided fundamental quantitative information on arterial pressure in individual subjects and in populations and facilitated estimation of cardiovascular risk related to levels of blood pressure obtained from the brachial cuff. Although the measurement is taken in a peripheral limb, the values are generally assumed to reflect the pressure throughout the arterial tree in large conduit arteries. Since the arterial pressure pulse becomes modified as it travels away from the heart towards the periphery, this is generally true for mean and diastolic pressure, but not for systolic pressure, and so pulse pressure. The relationship between central and peripheral pulse pressure depends on propagation characteristics of arteries. Hence, while the sphygmomanometer gives values of two single points on the pressure wave (systolic and diastolic pressure), there is additional information that can be obtained from the time-varying pulse waveform that enables an improved quantification of the systolic load on the heart and other central organs. This topical review will assess techniques of pressure measurement that relate to the use of the cuff sphygmomanometer and to the non-invasive registration and analysis of the peripheral and central arterial pressure waveform. Improved assessment of cardiovascular function in relation to treatment and management of high blood pressure will result from future developments in the indirect measurement of arterial blood pressure that involve the conventional cuff sphygmomanometer with the addition of information derived from the peripheral arterial pulse. PMID:19940350
Braun, Fabian; Proença, Martin; Rapin, Michael; Lemay, Mathieu; Adler, Andy; Grychtol, Bartłomiej; Solà, Josep; Thiran, Jean-Philippe
Electrical impedance tomography (EIT) allows the measurement of intra-thoracic impedance changes related to cardiovascular activity. As a safe and low-cost imaging modality, EIT is an appealing candidate for non-invasive and continuous haemodynamic monitoring. EIT has recently been shown to allow the assessment of aortic blood pressure via the estimation of the aortic pulse arrival time (PAT). However, finding the aortic signal within EIT image sequences is a challenging task: the signal has a small amplitude and is difficult to locate due to the small size of the aorta and the inherent low spatial resolution of EIT. In order to most reliably detect the aortic signal, our objective was to understand the effect of EIT measurement settings (electrode belt placement, reconstruction algorithm). This paper investigates the influence of three transversal belt placements and two commonly-used difference reconstruction algorithms (Gauss-Newton and GREIT) on the measurement of aortic signals in view of aortic blood pressure estimation via EIT. A magnetic resonance imaging based three-dimensional finite element model of the haemodynamic bio-impedance properties of the human thorax was created. Two simulation experiments were performed with the aim to (1) evaluate the timing error in aortic PAT estimation and (2) quantify the strength of the aortic signal in each pixel of the EIT image sequences. Both experiments reveal better performance for images reconstructed with Gauss-Newton (with a noise figure of 0.5 or above) and a belt placement at the height of the heart or higher. According to the noise-free scenarios simulated, the uncertainty in the analysis of the aortic EIT signal is expected to induce blood pressure errors of at least ± 1.4 mmHg. PMID:26007662
Rey, Markus; Weber, Edgar W; Hess, Patrick D
Here we evaluated the ability of a new, dual blood-pressure telemetry transmitter to simultaneously measure pulmonary and systemic blood pressure and the electrocardiogram in rats. The transmitter was implanted in normotensive and monocrotaline-induced pulmonary hypertensive Wistar rats, with sensing catheters placed in the pulmonary artery (channel 1) and descending aorta (channel 2). Biopotential electrodes were positioned to record an apex-based lead II electrocardiogram. Pulmonary and sys...
Rego Filho, E A; Mello, S F; Silva, C R; Vituri, D W; Bazoni, E; Gordan, L N
OBJECTIVE: The aim of this study is to analyze the substitution of the standard auscultatory method by the oscillometric blood pressure monitor, independently of the validity of the intraarterial blood pressure measurement. The accuracy of the automatic oscillometric monitor was compared to the auscultatory mercury manometer blood pressure measurement in apparently healthy school age children. METHODS: A device able to perform 3 simultaneous readings are used: one reading by the monitor and the others by two "blind" observers. We studied 72 school age children with the following characteristics: mean age 9.5 (6.1-16.1) and 39 males (54.2%). RESULTS: The difference for the systolic and diastolic blood pressure obtained by the monitor was in average + 6.2 mmHg and + 10.0 mmHg, respectively, when compared to the observer's readings. There was neither a good correlation nor a good agreement between the two observers and the monitor in the blood pressure determination. CONCLUSIONS: We concluded that the substitution of the standard auscultatory method for the non-invasive oscillometric method to measure blood pressure in school age children can not be generally recommended. PMID:14685547
Chen, Diliang; Chen, Fei; Murray, Alan; Zheng, Dingchang
Background Accurate blood pressure (BP) measurement depends on the reliability of oscillometric cuff pressure pulses (OscP) and Korotkoff sounds (KorS) for automated oscillometric and manual techniques. It has been widely accepted that respiration is one of the main factors affecting BP measurement. However, little is known about how respiration affects the signals from which BP measurement is obtained. The aim was to quantify the modulation effect of respiration on oscillometric pulses and K...
Thien, T.; Keltjens, E.B.; Lenders, J.W.M.; Deinum, J.
OBJECTIVE: To establish whether the results of blood pressure (BP) measurements are affected by wearing clothing underneath the BP cuff during measurement. METHODS: Normotensive and hypertensive patients (n=133; 65 men) of an outpatient clinic participated in this study. BP was measured according to
Full Text Available Blood pressure (BP measurement by the use of sphygmomanometer and auscultation of brachial artery by stethoscope is the most widespread way of measurement in daily clinical practice. The aim of the present study was to explore nurses' knowledge regarding correct way of measurement of blood pressure by the use of a sphygmomanometer. Method and material: The sample –studied consisted of individuals from all levels of nursing personnel. Data were collected by the use the completion of a specially designed questionnaire which apart from demographic variables, it included items concerning knowledge about the correct way of blood pressure measurement. Results: From the 360 participants, 88% (316 were women, while the mean age of the sample-studied was 36,1 years old. 51,4% of the sample-studied was of tertiary education. In terms of the knowledge of participants, 79,6% knew the correct preparation, 98,6% knew that arm position for blood pressure measurement should be on a pillow or a side table and the 94,8% knew the back should be supported on a chair back. 60,9% of the sample-studied knew that at first time, blood pressure (BP should be measured in both arms, while 45,9% didn't know that BP should be measured in arm with higher measurement. 78,2% of the participants knew that the correct measurement includes the use of sphygmomanometer and auscultation of brachial artery by stethoscope. 67,9% knew that body and arm position should be in ankle of 45o degrees and that the cuff should be placed around the upper arm at the same vertical height as the heart. 59,8% knew that the cuff should be placed 2-3 cm above the point of auscultation of brachial artery. 45,1% considered that the arm cuff should be inflated manually until when the height of the column of mercury approaches 200 mmHg. 52,3% knew the correct rhythm (2-3 mmHg/sec that the pressure in the cuff should be released. 88,4% knew the sounds corresponding to the measurement of Systolic blood
Paaske, William; Tønnesen, K H
The clinical course was followed and the ankle and toe blood pressures were measured with the strain gauge technique on 5 occasions during 2 years in 43 patients with pain at rest and/or ischaemic ulceration due to severe ischaemia of the legs on the basis of occlusive arterial disease. Although...... arteriosclerosis of the legs in non-diabetic patients is generally considered a benign disease from the standpoint of limb survival, the critical level of TPI (systolic toe blood pressure/systolic arm blood pressure) was found to be 0.07 as a TPI below this value was associated with an overall 82% risk of...... amputation. With TPI above 0.07, the chance of successful conservative therapy was about 40%. Diabetics with severe ischaemia must be regarded as a high risk group in respect of amputation (64%) and lethality (64%). A variance analysis was made on the pressure data: In patients with low pressure peripheral...
Willem J Verberk
Full Text Available Willem J Verberk, Abraham A Kroon, Heidi A Jongen-Vancraybex, Peter W de LeeuwUniversity Hospital Maastricht, Department of Internal Medicine and Cardiovascular Research Institute Maastricht (CARIM, Maastricht University, The NetherlandsPurpose: To review the literature on home blood pressure measurement (HBPM, to examine its validity and applicability for clinical practice and to provide recommendations regarding HBPM assessment.Findings: HBPM can eliminate the white coat effect and offers the possibility to obtain multiple measurements under standardized conditions, which increases knowledge of overall blood pressure value. Although it is not entirely capable of replacing ambulatory blood pressure measurement (ABPM, HBPM correlates better with target organ damage and cardiovascular mortality than office blood pressure measurement (OBPM, it enables prediction of sustained hypertension in patients with borderline hypertension, and proves to be an appropriate tool for assessing drug efficacy. Additional advantages of HBPM are that it may increase drug compliance and patient’s awareness of hypertension. Overall, OBPM yield higher blood pressure values than HBPM. Differences between OBPM and HBPM tend to increase with age and are generally higher in patients without antihypertensive treatment than in patients with antihypertensive treatment.Recommendations: Measurements should be performed according to accepted guidelines and recordings should be performed with a memory equipped automatic validated device. From the data reviewed here, we recommend that HBPM be assessed monthly by taking two measurements in the morning within 1 hour after awakening and two in the evening for three consecutive days, the data from the first day should be dismissed. A subject should be labeled hypertensive if his/her HBPM value is equal to or greater than 137 mmHg systolic and/or 84 mmHg diastolic.Keywords: blood pressure, hypertension, self-measurement, home
Wagner, Stefan; Kamper, C. H.; Rasmussen, Niels H;
Background: Patients often fail to adhere to clinical recommendations when using current blood pressure self-measurement (BPSM) methods and equipment. As existing BPSM equipment is not able to detect non-adherent behavior, this could result in misdiagnosis and treatment error. To overcome this...... patients scheduled for self-measuring their blood pressure (BP) in the waiting room at an obstetrics department's outpatient clinic to perform an additional BPSM using ValidAid. We then compared the automatically measured and classified values from ValidAid with our manual observations. Results: We found...
Seymour, John; Pennington, Gary
Heart disease, the leading cause of death in the US, may be spotted early on by looking at photoplethysmogram (PPG) data. This experiment explores a new method of continuously monitoring the blood pressure pulse with PPG data. In contrast to the traditional sphygmomanometer (cuff) method, which yields only the systolic and diastolic pressure during measurement, this method tracks the blood pressure pulse wave in a non-invasive continuous manner. This procedure allows for fast, inexpensive, and detailed analysis of the patient's blood pressure implementable on a large scale. We also explore the second derivative of the PPG data. In combination with the above method, the patient's heart risk can be effectively detected. We acknowledge Fisher Endowment Grant support from the Jess and Mildred Fisher College of Science and Mathematics, Towson University.
Pan, Cheng-Rui; Staessen, Jan A.; Li, Yan; Wang, Ji-Guang
The ankle-brachial blood pressure index (ABI) predicts cardiovasular disease. To our knowledge, no study has compared manual ABI measurements with an automated electronic oscillometric method in a population sample. We enrolled 946 residents (50.8% women; mean age, 43.5 years) from 8 villages in JingNing County, Zhejiang Province, P.R. China. We computed ABI as the ratio of ankle-to-arm systolic blood pressures from consecutive auscultatory or Doppler measurements at the posterior tibial and ...
Babbs Charles F
Full Text Available Abstract Background The oscillometric method of measuring blood pressure with an automated cuff yields valid estimates of mean pressure but questionable estimates of systolic and diastolic pressures. Existing algorithms are sensitive to differences in pulse pressure and artery stiffness. Some are closely guarded trade secrets. Accurate extraction of systolic and diastolic pressures from the envelope of cuff pressure oscillations remains an open problem in biomedical engineering. Methods A new analysis of relevant anatomy, physiology and physics reveals the mechanisms underlying the production of cuff pressure oscillations as well as a way to extract systolic and diastolic pressures from the envelope of oscillations in any individual subject. Stiffness characteristics of the compressed artery segment can be extracted from the envelope shape to create an individualized mathematical model. The model is tested with a matrix of possible systolic and diastolic pressure values, and the minimum least squares difference between observed and predicted envelope functions indicates the best fit choices of systolic and diastolic pressure within the test matrix. Results The model reproduces realistic cuff pressure oscillations. The regression procedure extracts systolic and diastolic pressures accurately in the face of varying pulse pressure and arterial stiffness. The root mean squared error in extracted systolic and diastolic pressures over a range of challenging test scenarios is 0.3 mmHg. Conclusions A new algorithm based on physics and physiology allows accurate extraction of systolic and diastolic pressures from cuff pressure oscillations in a way that can be validated, criticized, and updated in the public domain.
Alam, Mohammad Afaque; Parks, Cory; Mancarella, Salvatore
During the development of new vasoactive agents, arterial blood pressure monitoring is crucial for evaluating the efficacy of the new proposed drugs. Indeed, research focusing on the discovery of new potential therapeutic targets using genetically altered mice requires a reliable, long-term assessment of the systemic arterial pressure variation. Currently, the gold standard for obtaining long-term measurements of blood pressure in ambulatory mice uses implantable radio-transmitters, which require artery cannulation. This technique eliminates the need for tethering, restraining, or anesthetizing the animals which introduce stress and artifacts during data sampling. However, arterial blood pressure monitoring in mice via catheterization can be rather challenging due to the small size of the arteries. Here we present a step-by-step guide to illustrate the crucial key passages for a successful subcutaneous implantation of radio-transmitters and carotid artery cannulation in mice. We also include examples of long-term blood pressure activity taken from freely moving mice after a period of post-surgery recovery. Following this procedure will allow reliable direct blood pressure recordings from multiple animals simultaneously. PMID:27286041
Sannino, Giovanna; Melillo, Paolo; Stranges, Saverio; De Pietro, Giuseppe; Pecchia, Leandro
Orthostatic Hypotension is defined as a reduction of systolic and diastolic blood pressure within 3 minutes of standing, and may cause dizziness and loss of balance. Orthostatic Hypotension has been considered an important risk factor for falls since 1960. This paper presents a model to predict the systolic blood pressure drop due to orthostatic hypotension, relying on heart rate variability measurements extracted from 5 minute ECGs recorded before standing. This model was developed and validated with the leave-one-out cross-validation technique involving 10 healthy subjects, and finally tested with an additional 5 healthy subjects, whose data were not used during the training and cross-validation process. The results show that the model predicts correctly the systolic blood pressure drop in 80 % of all experiments, with an error rate below the measurement error of a sphygmomanometer digital device. PMID:26345451
McCarthy, B. M.; O'Flynn, B.; Mathewson, A.
The objective of this paper is to examine the Pulse Transit Method (PTT) as a non-invasive means to track Blood Pressure over a short period of time. PTT was measured as the time it takes for an ECG R-wave to propagate to the finger, where it is detected by a photoplethysmograph sensor. The PTT method is ideal for continuous 24-hour Blood Pressure Measurement (BPM) since it is both cuff-less and non-invasive and therefore comfortable and unobtrusive for the patient. Other techniques, such as the oscillometric method, have shown to be accurate and reliable but require a cuff for operation, making them unsuitable for long term monitoring. Although a relatively new technique, the PTT method has shown to be able to accurately track blood pressure changes over short periods of time, after which re-calibration is necessary. The purpose of this study is to determine the accuracy of the method.
The objective of this paper is to examine the Pulse Transit Method (PTT) as a non-invasive means to track Blood Pressure over a short period of time. PTT was measured as the time it takes for an ECG R-wave to propagate to the finger, where it is detected by a photoplethysmograph sensor. The PTT method is ideal for continuous 24-hour Blood Pressure Measurement (BPM) since it is both cuff-less and non-invasive and therefore comfortable and unobtrusive for the patient. Other techniques, such as the oscillometric method, have shown to be accurate and reliable but require a cuff for operation, making them unsuitable for long term monitoring. Although a relatively new technique, the PTT method has shown to be able to accurately track blood pressure changes over short periods of time, after which re-calibration is necessary. The purpose of this study is to determine the accuracy of the method.
McCarthy, B M; O' Flynn, B; Mathewson, A, E-mail: firstname.lastname@example.org [Tyndall National Institute, UCC, Lee Maltings, Prospect Row, Cork (Ireland)
The objective of this paper is to examine the Pulse Transit Method (PTT) as a non-invasive means to track Blood Pressure over a short period of time. PTT was measured as the time it takes for an ECG R-wave to propagate to the finger, where it is detected by a photoplethysmograph sensor. The PTT method is ideal for continuous 24-hour Blood Pressure Measurement (BPM) since it is both cuff-less and non-invasive and therefore comfortable and unobtrusive for the patient. Other techniques, such as the oscillometric method, have shown to be accurate and reliable but require a cuff for operation, making them unsuitable for long term monitoring. Although a relatively new technique, the PTT method has shown to be able to accurately track blood pressure changes over short periods of time, after which re-calibration is necessary. The purpose of this study is to determine the accuracy of the method.
Patzak, Andreas; Mendoza, Yuri; Gesche, Heiko; Konermann, Martin
Continuous blood pressure (BP) measurement allows the investigation of transient changes in BP and thus may give insights into mechanisms of BP control. We validated a continuous, non-invasive BP measurement based on the pulse transit time (PTT), i.e., BP(PTT), by comparing it with the intra-arterial BP (BP(i.a.)) measurement. Twelve subjects (five females and seven males) were included. BP(i.a.) was obtained from the radial artery using a system from ReCor Medical. Systolic and diastolic BP were calculated using the PTT (BP(PTT), SOMNOscreen). (PTT) was determined from the electrocardiogram and the peripheral pulse wave. The BP was modulated by application of increasing doses of dobutamine (5, 10, 20 μg/kg body mass). Systolic BP(PTT) and systolic BP(i.a.) correlated significantly (R = 0.94). The limits of agreement in the Bland-Altman plot were ± 19 mmHg; the mean values differed by 1 mmHg. The correlation coefficient for the diastolic BP measurements was R = 0.42. The limits of agreement in the Bland-Altman plot were ± 18 mmHg, with a mean difference of 5 mmHg in favour of the BP(PTT). The study demonstrates a significant correlation between the measurement methods for systolic BP. The results encourage the application of PTT-based BP measurement for the evaluation of BP dynamics and pathological BP changes. PMID:25857601
In this paper the authors try to classify circulatory measurements as to certain characteristics such as ''activeness'', ''invasivenesss'', etc. Next they turn to defining some fundamental measurements in the blood circulation and focus attention on the continuous measurement of systemic arterial pressure and flow to finally emphasize the inaccuracy of clinical cardiac output methods. The conclusion is that clinical circulatory measurement methods have accuracies at least an order of magnitude worse than common physical methods used in electrical or mechanical engineering. (Auth.)
chart were uncontrolled according to 24 hour ABPM criteria. This suggests that the white coat effect makes blood pressure measurements in the doctor's offices, at least as currently done, not sufficiently accurate for determining treatment endpoint.
Forearm skin blood flow was measured during external pressure loading in normal human subjects using 133Xe washout from intracutaneous injection sites. Pressures ranging between 5 and 150 mmHg were applied through a 3-cm-diameter disc placed over the site of flow determination. The pressure was maintained constant by a servo-controlled loading mechanism. Flow decreased with pressures from 5 to 10 and 30 to 150 mmHg, but remained constant with pressures from 10 to 30 mmHg. Reactive hyperemia occurred following removal of pressures of 90 mmHg or greater, but did not occur following removal of lower pressures. The pressure-flow curve for parasacral skin of paraplegic subjects closely paralleled the pressure-flow curve of normal skin at pressures tested: 5 to 15 mmHg. These data are interpreted to demonstrate autoregulation of skin blood flow. Autoregulation in parasacral skin of paraplegic subjects suggests a peripheral mechanism. The occurrence of hyperemia at pressures which exceed the ability of skin to autoregulate suggests that both autoregulation and post occlusion hyperemia may have the same mechanism
High blood pressure, also called hypertension, usually has no symptoms. But it can cause serious problems such as stroke, heart ... kidney failure. If you cannot control your high blood pressure through lifestyle changes such as losing weight and ...
Tønnesen, K H; Noer, Ivan; Paaske, William; Sager, P
Systolic blood pressures at toe and ankle were measured in 459 consecutive patients with occlusive arterial disease. Fifty-eight per cent had intermittent claudication with arterial disease of all degrees of severity. Seventeen per cent complained of rest pain having toe systolic pressures below 30...... occlusive arterial disease which was located distally on the legs. A classification in three groups is suggested: (1) ischemia only during exercise; (2) ischemia at rest with or without ulcerations: and (3) diabetics with chronic ulcerations....
Harja, Juha; Myllylä, Teemu S.; Sorvoja, Hannu S. S.; Myllylä, Risto A.; Elseoud, Ahmed A.; Nikkinen, Juha; Kiviniemi, Vesa; Tervonen, Osmo
This report focuses on designing and implementing a non-invasive blood pressure measuring device capable of being used during magnetic resonance imaging. This device is based on measuring pulse wave velocity in arterial blood and using the obtained result to estimate diastolic blood pressure. Pulse transit times are measured by two fibre optical accelerometers placed over chest and carotid artery. The fabricated accelerometer contains two static fibres and a cantilever beam. The free end of the beam is angled at 90 degrees to act as a reflecting surface. Optical fibres are used for both illuminating the surface and receiving the reflected light. Acceleration applied to the sensor causes deflection of the beam, whereupon the amount of reflected light changes. The sensor output voltage is proportional to the intensity of the reflected light. Tests conducted on the electronics and sensors inside an MRI room during scanning proved that the device is MR conditional. No artifacts or distortions were detected.
Lass, J; Meigas, K; Karai, D; Kattai, R; Kaik, J; Rossmann, M
This paper gives an overview of a research, which is focused on the development of the convenient device for continuous non-invasive monitoring of arterial blood pressure. The blood pressure estimation method is based on a presumption that there is a singular relationship between the pulse wave propagation time in arterial system and blood pressure. The parameter used in this study is pulse wave transit time (PWTT). The measurement of PWTT involves the registration of two time markers, one of which is based on ECG R peak detection and another on the detection of pulse wave in peripheral arteries. The reliability of beat to beat systolic blood pressure calculation during physical exercise was the main focus for the current paper. Sixty-one subjects (healthy and hypertensive) were studied with the bicycle exercise test. As a result of current study it is shown that with the correct personal calibration it is possible to estimate the beat to beat systolic arterial blood pressure during the exercise with comparable accuracy to conventional noninvasive methods. PMID:17272172
Mehlsen, Jesper; Wiinberg, Niels; Bruce, Christopher
Blood pressure at the ankle level is a reliable indicator of peripheral arterial disease (PAD) and the ankle brachial index (ABI) is a useful non-invasive screening tool for the early detection of atherosclerosis. In the first part of the study, systolic blood pressures obtained by oscillometry and...... PAD was sufficiently high in subjects over the age of 60 years to warrant screening. The ankle brachial index based on measurements with an oscillometric device was shown reliable in the exclusion of PAD, thereby fulfilling an important criterion for the use in screening....
Suganthi V, Navin Rajaratnam, Suzanne Maria D’cruz
Background: The prevalence of obesity is increasing in Indian youth and obesity is associated with complications like systemic hypertension. Often, due to the non-availability of appropriate sized cuffs, standard cuff bladders are used to measure blood pressure in the forearms of obese young adults. Aim: To compare the upper arm arterial blood pressure measured using an appropriate cuff with the forearm arterial blood pressure measured using a standard cuff and conventional sphygmomanometry i...
Rithalia, S V; Edwards, D
Non-invasive oscillometric blood pressure and pulse measured by an Omron HEM-703CP monitor were compared with arterial values obtained from direct measurements of the radial artery. An excellent correlation and agreement was found between the two methods (systolic r = 0.99; diastolic r = 0.97; pulse r = 0.99), although there was some variability among individual subjects. The range of difference between them was 0 to 10 mmHg for systolic and -6 to +5 mmHg for diastolic pressures. When tested on the bench using the Metron QA-1280 non-invasive blood pressure analyser the HEM-703CP monitor rarely exhibited errors exceeding 2-3 mmHg over a measurement range of 50-200 mmHg. PMID:7776359
A new arterial distensibility measurement technique was assessed in 100 healthy normotensive subjects. Arterial transmural pressures on the whole right arm were reduced with a 50 cm long cuff inflated to 10, 20, 30 and 40 mmHg. The electrocardiogram, and finger and ear photoplethysmograms were recorded simultaneously. Arm pulse propagation time, pulse wave velocity (PWV) and arterial volume distensibility were determined. With a 40 mmHg reduction in transmural pressure, arm pulse propagation time increased from 61 to 83 ms, PWV decreased from 12 to 8 m s−1 and arterial distensibility increased from 0.102% to 0.232% per mmHg (all P < 0.0001). At all cuff pressures, arterial distensibility was significantly related to resting mean arterial pressure (MAP), diastolic blood pressure (DBP) and age, and for systolic blood pressure at 30 and 40 mmHg (all P < 0.05). At 40 mmHg cuff pressure, arterial distensibility fell by 54% for a MAP increase from 75 to 105 mmHg, 57% for a DBP increase from 60 to 90 mmHg and 47% for an age increase from 20 to 70 years. These changes were more than double than those without cuff pressure. Our technique showed that systemic volume distensibility of the peripheral arm artery reduced with age, with a greater effect at higher external and lower transmural pressures
Amar M. Taksande; Aishwarya Jadhav; Jyoti Nair
Background: When the upper arm (UA) is inaccessible or a standard-sized blood pressure (BP) cuff is unavailable, some healthcare workers use the forearm (FA) to measure BP with a mercury sphygmomanometer. Objective: The objective was to determine the accuracy of BP measurement in the arm and FA. Design: Prospective, randomized study. Setting: Department of Pediatrics, JNMC, Sawangi (Meghe) Participants: A total of 72 children aged 5-15 years. Measurements: Mercury and Automatic (OMRON Tokyo, ...
M. F. VALIM
Full Text Available
Blood pressure is the force of blood against artery walls. It is measured in millimeters of mercury (mm Hg and recorded as two numbers: systolic pressure (as the heart contracts over diastolic pressure (as the heart relaxes between beats. High blood pressure (hypertension is defined as chronically elevated high blood pressure, with systolic blood pressure (SBP of 140 mm Hg or greater, and diastolic blood pressure (DBP of 90 mm Hg or greater. High blood pressure (HBP, smoking, abnormal blood lipid levels, obesity and diabetes are risk factors for coronary heart disease, the leading cause of death in the US. Lifestyle modifications such as engaging in regular physical activity, quitting smoking and eating a healthy diet (limiting intake of saturated fat and sodium and increasing consumption of fiber, fruits and vegetables are advocated for the prevention, treatment, and control of HBP. As multiple factors influence blood pressure, the effects of each factor are typically modest, particularly in normotensive subjects, yet the combined effects can be substantial. Nutrition plays an important role in influencing blood pressure. Orange juice should be included as part of any low sodium diet and/or any blood pressure reducing eating plan, as it is sodium free, fat-free and can help meet recommended levels of potassium intake that may contribute to lower BP.
Drake, M J P; Hill, J S
Upper-arm non-invasive blood pressure measurement during caesarean section can be uncomfortable and unreliable because of movement artefact in the conscious parturient. We aimed to determine whether ankle blood pressure measurement could be used instead in this patient group by comparing concurrent arm and ankle blood pressure measured throughout elective caesarean section under regional anaesthesia in 64 term parturients. Bland-Altman analysis of mean difference (95% limits of agreement [range]) between the ankle and arm was 11.2 (-20.3 to +42.7 [-67 to +102]) mmHg for systolic arterial pressure, -0.5 (-21.0 to +19.9 [-44 to +91]) mmHg for mean arterial pressure and -3.8 (-25.3 to +17.8 [-41 to +94]) mmHg for diastolic arterial pressure. Although ankle blood pressure measurement is well tolerated and allows greater mobility of the arms than measurement from the arm, the degree of discrepancy between the two sites is unacceptable to allow routine use of ankle blood pressure measurement, especially for systolic arterial pressure. However, ankle blood pressure measurement may be a useful alternative in situations where arm blood pressure measurement is difficult or impossible. PMID:23480469
Jong-Gu Choi; Keun-Ho Kim; Woo-Beom Lee; Dong-Hyun Nam; Il-Ho Son; You-Sik Hong; Sang-Suk Lee
To measure precise blood pressure (BP) and pulse rate without using a cuff, we have developed an arterial pulsimeter consisting of a small, portable apparatus incorporating a Hall device. Regression analysis of the pulse wave measured during testing of the arterial pulsimeter was conducted using two equations of the BP algorithm. The estimated values of BP obtained by the cuffless arterial pulsimeter over 5 s were compared with values obtained using electronic or liquid mercury BP meters. The...
Full Text Available Introduction. The analysis of frequency of kidney diseases, causes of terminal renal failure in patients subjected to repeated dialysis in major dialysis centers in the region of Kosovo and Metohija as well as mortality rate in these patients showed a stable number of patients with kidney diseases, evidencing that the populations of Vitina and Gnjilane were the most commonly affected. These towns are designated as regions with the highest incidence of Balkan endemic nephropathy. The present analysis was aimed at: 1. establishing the incidence rate of hypertension and renal function impairment by using clinical and laboratory analyses in individuals from Vitina and neighboring villages, and 2. investigating the correlation between creatinine clearance, age and arterial blood pressure in the studied group. Material and methods. The analysis included 510 individuals (excluding diabetics over the one-month period during 2004 (201 males with their age ranging from 18 to 90 years. Blood pressure measurements were performed in all the individuals, while renal function was determined by glomerular filtration rate (GFR, as calculated according to Cockcroft-Gault formula. Results. Creatinine clearance was lower than 100 ml/min in 237 (46.5% individuals, out of whom 62 (12% had GFR below 60 ml/min. Among the remaining 273 (53.5% individuals with GFR above 100 ml/min, 68 individuals had GFR above 140 ml/min. The distribution of patients depending on their systolic blood pressure values revealed that systolic blood pressure was lower in 44 subjects and higher in 302 subjects (59%. The correlation test revealed significant association between the studied parameters. The positive correlation between patients' age and kidney function means the higher the age of subjects the lower creatinine clearance value (r=-0.622, p<0.001. The negative correlation of the systolic blood pressure value and kidney function was found, meaning that higher the systolic blood
Morel, D; Suter, P
A new automatic apparatus for the measurement of arterial pressure by a non-invasive technique was compared with direct intra-arterial measurement in 20 adult patients in a surgical intensive care unit. The apparatus works on the basis of the principle of oscillometry. Blood pressure is determined with a microprocessor by analysis of the amplitude of the oscillations produced by a cuff which is inflated then deflated automatically. Thus mean arterial pressure corresponds to the maximum amplitude. Systolic and diastolic pressures are deduced by extrapolation to zero of the amplitudes on either side of the maximum reading. Mean arterial pressure (AP) proved to be very reliable within the limits studied: 8.0 - 14.7 kPa (60 - 110 mmHg) with a difference in mean direct AP and indirect AP of 0,09 +/- 0.9 kPa SD (0.71 +/- 7 mmHg) and a coefficient of linear correlation between the two methods of r = 0.82. This non-invasive technique determined systolic arterial pressure (sAP) in a less reliable fashion than AP when compared with the invasive technique, with a tendency to flatten the extreme values. The correlation coefficient here was 0.68. Finally, diastolic arterial pressure (dAP) showed a better degree of agreement through with a difference in mean indirect AP and mean direct AP of 1.0 +/- 0.8 kPa (7.6 +/- 6.0 mmHg). These results indicate a good degree of agreement for measurements of mean arterial pressure, clinically the most important, between the two methods used. Measurements of diastolic pressure and above all of diastolic pressure seemed to be less in agreement. This difference could be due to an error in determination of the automatic apparatus tested or to the peripheral site (radial artery) of the intra-arterial catheter used, itself falsifying the humeral arterial pressure. PMID:6113805
İlhan, İlhan; Yıldız, İbrahim; Kayrak, Mehmet
Today, smart mobile devices (telephones and tablets) are very commonly used due to their powerful hardware and useful features. According to an eMarketer report, in 2014 there were 1.76 billion smartphone users (excluding users of tablets) in the world; it is predicted that this number will rise by 15.9% to 2.04 billion in 2015. It is thought that these devices can be used successfully in biomedical applications. A wireless blood pressure measuring device used together with a smart mobile device was developed in this study. By means of an interface developed for smart mobile devices with Android and iOS operating systems, a smart mobile device was used both as an indicator and as a control device. The cuff communicating with this device through Bluetooth was designed to measure blood pressure via the arm. A digital filter was used on the cuff instead of the traditional analog signal processing and filtering circuit. The newly developed blood pressure measuring device was tested on 18 patients and 20 healthy individuals of different ages under a physician's supervision. When the test results were compared with the measurements made using a sphygmomanometer, it was shown that an average 93.52% accuracy in sick individuals and 94.53% accuracy in healthy individuals could be achieved with the new device. PMID:26626086
... also known as blood vessels and capillaries. The pressure --- blood pressure --- is the result of two forces. The ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) Introduction What ...
... much alcohol. Signs and Symptoms of High Blood Pressure High blood pressure usually has no warning signs or symptoms , ... they are at high risk for high blood pressure . Blood Pressure Levels Normal systolic: less than 120 mmHg ...
... Radiation-Emitting Products Vaccines, Blood & Biologics Animal & Veterinary Cosmetics Tobacco Products For Consumers Home For Consumers Consumer Information by Audience For Women High Blood Pressure (Hypertension) Share Tweet Linkedin Pin ...
Reidpath, Daniel D.; Ling, Mei Lee; Yasin, Shajahan; Rajagobal, Kanason; Allotey, Pascale
Introduction: Population monitoring and screening of blood pressure is an important part of any population health strategy. Qualified health workers are expensive and often unavailable for screening. Non-health workers with electronic blood pressure monitors are increasingly used in community-based research. This approach is unvalidated. In a poor, urban community we compared blood pressure measurements taken by non-health workers using electronic devices against qualified health workers usin...
Parker, Emily D; Sinaiko, Alan R; Daley, Matt F.; Kharbanda, Elyse O; Trower, Nicole K.; Tavel, Heather M.; Sherwood, Nancy E.; Magid, David J.; Margolis, Karen L.; O’Connor, Patrick J.
Introduction Elevated blood pressure in childhood may predict increased cardiovascular risk in young adulthood. The Task Force on the Diagnosis, Evaluation and Treatment of High Blood pressure in Children and Adolescents recommends that blood pressure be measured in children aged 3 years or older at all health care visits. Guidelines from both Bright Futures and the Expert Panel of Integrated Guidelines for Cardiovascular Health and Risk Reduction in Children and Adolescents recommend annual ...
Maltz, Jonathan S.; Tison, Geoffrey H; Alley, Hugh F.; Budinger, Thomas F; Owens, Christopher D.; Olgin, Jeffrey
The integrity of endothelial function in major arteries (EFMA) is a powerful independent predictor of heart attack and stroke. Existing ultrasound-based non-invasive assessment methods are technically challenging and suitable only for laboratory settings. EFMA, like blood pressure (BP), is both acutely and chronically affected by factors such as lifestyle and medication. Consequently, lab-based measurements cannot fully gauge the effects of medical interventions on EFMA. EFMA and BP have, arg...
Zimei, Su; Wei, Xu; Hui, Yu; Fei, Du; Jicun, Wang; Kexin, Xu
In this study the pulse wave characteristics were used as a new approach to measure the human blood pressure. Based the principle of pulse wave and theory of the elastic vascular, the authors analyzed the characteristic of the pulse waveforms and revealed the characteristics points which could be used to represent the blood pressure. In this investigation the relevant mathematical feature was used to identify the relationship between the blood pressure and pulse wave parameters in a more accurate way. It also provided an experimental basis to carry out continuing non-invasive blood pressure monitoring using the pulse wave method.
Wagner, Stefan Rahr; Rasmussen, Niels Holm; Ahrendt, Peter;
Blood pressure self-measurement (BPSM) requires the patient to follow a range of recommendations. Patients must remain silent during measurements, be seated correctly with back support and legs uncrossed, and must have rested at least 5 minutes prior to taking the measurement. Current blood...... process, as well as a sensor seat for measuring patient posture and activity before and during the BPSM process....
Joensen, Jette Bang; Juul, Søren; Abrahamsen, Jimmi;
This study assesses measurement variation in the measurement of ankle systolic blood pressure (ABP) when measured with Doppler ultrasound and with the strain gauge method. Ninety-seven patients were included. ABP was measured with Doppler ultrasound and with the strain gauge method. The methods...... were compared graphically by scatterplots and analyzed by paired t test, analysis of variance, and Pitman's paired variance ratio test. ABP was measured by strain gauge in all extremities, whereas no Doppler signal was obtainable in 7 limbs (4%). There was no systematic difference in measurements...... between the means of the two measurements. However, a substantial difference of more than 25 mm Hg was found in 15% of limbs and more than 20 mm Hg in 20%. In the majority of patients, measurements of ABP by Doppler ultrasound and the strain gauge method give similar results, but for a minority the...
... Heart Disease Cholesterol Salt Million HeartsÂ® WISEWOMAN Preventing High Blood Pressure: Healthy Living Habits Recommend on Facebook Tweet Share ... meal and snack options can help you avoid high blood pressure and its complications. Be sure to eat plenty ...
... whether imbalances in this system cause high blood pressure. Blood Vessel Structure and Function Changes in the structure ... can affect blood pressure. Genetic Causes of High Blood Pressure Much of the understanding of the body systems ...
Stolarz-Skrzypek, Katarzyna; Thijs, Lutgarde; Wizner, Barbara;
Thomas G. Pickering spent most of his scientific career in carrying out research on clinical hypertension and blood pressure (BP) measurement. In our review of Pickering's seminal work, we first focused on white-coat hypertension and masked hypertension, two terms that he had introduced. Next, we...... involving white-coat hypertension and masked hypertension, diurnal BP variability, and self-measured BP. Recent studies validated Pickering's observations in terms of cardiovascular outcome and bridged the path from concept to application in clinical practice....
Bülow, J; Jelnes, Rolf
In order to evaluate if elevation of the ischaemic limb above heart level is an alternative to the conventionally applied method with external counterpressure for estimation of skin perfusion pressure, femoral and popliteal artery pressures were measured directly in eight patients with occlusion of...... the superficial femoral artery. The measurements were done in the horizontal position and during elevation of the calf above heart level. During elevation relative blood flow, measured by arterio-venous oxygen saturation differences, decreased compared with the horizontal position. In contrast the...... lower extremity cannot be determined non-invasively by elevation of the extremity, probably due to collapse of segments of the vascular bed increasing the vascular resistance considerably....
Rath, W; Schrader, J; Guhlke, U; Buhr-Schinner, H; Haupt, A; Kramer, A; Kuhn, W
Noninvasive 24-hour ambulatory blood pressure monitoring was performed in 17 normotensive and 19 preeclamptic pregnant women. The normotensive women showed a significant nightly decline in their systolic and diastolic blood pressure. In contrast, the preeclamptic women demonstrated either an attenuated circadian rhythm or no circadian rhythm at all. This result was even more pronounced in patients with severe hypertension, some of whom had a nocturnal increase in blood pressure in spite of being treated with antihypertensive drugs in an evening dose. The lack of nocturnal blood pressure decrease was also found 24 hours post partum. In summary, these results suggest that preeclamptic women are endangered by hypertensive emergencies mostly during the night. Therefore blood pressure controls should be extended into the night, and antihypertensive drugs should also be given in a sufficient evening dose. PMID:2214601
Suganthi V, Navin Rajaratnam, Suzanne Maria D’cruz
Full Text Available Background: The prevalence of obesity is increasing in Indian youth and obesity is associated with complications like systemic hypertension. Often, due to the non-availability of appropriate sized cuffs, standard cuff bladders are used to measure blood pressure in the forearms of obese young adults. Aim: To compare the upper arm arterial blood pressure measured using an appropriate cuff with the forearm arterial blood pressure measured using a standard cuff and conventional sphygmomanometry in obese otherwise healthy first year medical students. Materials and Methods: Blood pressure was measured in 27 obese otherwise healthy first year medical students after five minutes of rest using a mercury sphygmomanometer with the subjects seated and the arm and forearm at heart level, using an appropriate sized cuff for the upper arm according to American Heart Association standards and a standard cuff for the fore arm. Results: A statistically significant difference in both systolic [t-test (paired = -6.921; df = 26; sig = .000 (2- tailed] and diastolic blood pressure [t-test (paired = -8.508; df = 26; sig = .000 (2- tailed] was found, with the blood pressure readings being higher in the forearm. The correlations between upper arm and forearm systolic and diastolic blood pressure were 0.785 (p = .000 and 0.870 (p = .000. Conclusion: Both systolic and diastolic blood pressure measurements were significantly higher in the forearm. Further studies with larger sample size should be conducted to confirm that forearm blood pressure measurements using standard cuff bladders cannot be considered equal to upper arm measurements made using an appropriate sized cuff in all young obese individuals
Full Text Available Microalbuminuria is associated with hypertension and is a strong risk factor for subsequent chronic disease, both renal and coronary heart disease (CHD, Presently there are several methods available for measurement of microalbuminuria. The aim of this study was to evaluate if the three different methods gave similar information or if one of the assays were superior to the others. Blood pressure, inflammatory markers and cardiovascular mortality and morbidity were correlated with urine albumin analysed with a point-of-care testing (POCT instrument, nephelometric determination of albumin and albumin/creatinine ratio in elderly males. The study population consisted of 103 diabetic and 603 nondiabetic males (age 77 years in a cross-sectional study. We analyzed urine albumin with a HemoCue® Urine Albumin POCT instrument and a ProSpec® nephelometer and albumin/creatinine ratio. There were strong correlations between both systolic and diastolic blood pressure and all three urine albumin methods (p < 0.0001. There were also significant correlations between the different urine albumin measurements and serum amyloid A component, high-sensitivity C-reactive protein and interleukin-6. The three different urine albumin methods studied provided similar information in relation to cardiovascular disease. There was a strong correlation between systolic and diastolic blood pressure and microalbuminuria in both the whole study population and in nondiabetic males emphasizing the role of hypertension in glomerular damage. The good correlation between the studied urine albumin measurements show that all three methods can be used for monitoring urine albumin excretion.
Low frequency arterial wall movements for indirect blood pressure measurement in man. Validation of a method for non-invasive assessment of blood pressure under the influence of isoprenaline and angiotensin.
Dietz, U; Belz, G G
In order to measure blood pressure noninvasively, the second derivative of the low frequency wall movements of the brachial artery were registered with a piezo-electric pressure probe during deflation of a Riva-Rocci cuff along with the actual cuff pressure. Two characteristic phenomena of this signal have been suggested to reflect systolic and diastolic blood pressure. Appearance of a positive spike phenomenon (S) was suggested to indicate systolic blood pressure and disappearance of a negative preanacrotic notch (D) to indicate diastolic blood pressure. To prove the validity of these suggestions, these phenomena were assessed in 10 young healthy males during isoprenaline and angiotensin induced changes of blood pressure. Intraarterial (A. radialis) and auscultatory (A. brachialis) blood pressures were recorded simultaneously. Determination of systolic blood pressure with the S phenomenon agreed well with invasive and auscultatory results. Invasive diastolic values agreed well with the cuff pressure at the last signal before disappearance of the preanacrotic notch (D1). Data from auscultation agreed less well with the D1 phenomenon. With increasing doses of isoprenaline, the diastolic measurements (D1) tended to be lower than the invasive ones. However, this discrepancy was far discreeter than that seen with ordinary auscultatory blood pressure measurement. We therefore conclude that registrations of low frequency arterial wall movements yield distinct characteristic spike phenomena useful for measurement of blood pressure in good agreement with the invasive method. In addition, the method provides clearly documented records and should be useful in situations which rely on a valid indirect method. PMID:1898428
... Pressure High Blood Pressure Tools & Resources Stroke More Blood Pressure vs. Heart Rate Updated:Aug 30,2016 Blood ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) Introduction What ...
Full Text Available To measure precise blood pressure (BP and pulse rate without using a cuff, we have developed an arterial pulsimeter consisting of a small, portable apparatus incorporating a Hall device. Regression analysis of the pulse wave measured during testing of the arterial pulsimeter was conducted using two equations of the BP algorithm. The estimated values of BP obtained by the cuffless arterial pulsimeter over 5 s were compared with values obtained using electronic or liquid mercury BP meters. The standard deviation between the estimated values and the measured values for systolic and diastolic BP were 8.3 and 4.9, respectively, which are close to the range of values of the BP International Standard. Detailed analysis of the pulse wave measured by the cuffless radial artery pulsimeter by detecting changes in the magnetic field can be used to develop a new diagnostic algorithm for BP, which can be applied to new medical apparatus such as the radial artery pulsimeter.
Daniel D. Reidpath
Full Text Available Introduction: Population monitoring and screening of blood pressure is an important part of any population health strategy. Qualified health workers are expensive and often unavailable for screening. Non-health workers with electronic blood pressure monitors are increasingly used in community-based research. This approach is unvalidated. In a poor, urban community we compared blood pressure measurements taken by non-health workers using electronic devices against qualified health workers using mercury sphygmomanometers. Method: Fifty-six adult volunteers participated in the research. Data were collected by five qualified health workers, and six non-health workers. Participants were randomly allocated to have their blood pressure measured on four consecutive occasions by alternating a qualified health worker with a non-health worker. Descriptive statistics and graphs, and mixed effects linear models to account for the repeated measurement were used in the analysis. Results: Blood pressure readings by non-health workers were more reliable than those taken by qualified health workers. There was no significant difference between the readings taken by qualified health workers and those taken by non-health workers for systolic blood pressure. Non-health workers were, on average, 5–7 mmHg lower in their measures of blood pressure than the qualified health workers (95%HPD: −2.9 to −10.0 for diastolic blood pressure. Conclusion: The results provide empirical evidence that supports the practice of non-health workers using electronic devices for BP measurement in community-based research and screening. Non-health workers recorded blood pressures that differed from qualified health workers by no more than 10 mmHg. The approach is promising, but more research is needed to establish the generalisability of the results.
Deng, Zhipeng; Liang, Fuyou
An inflatable cuff wrapped around the upper arm is widely used in noninvasive blood pressure measurement. However, the mechanical interaction between cuff and arm tissues, a factor that potentially affects the accuracy of noninvasive blood pressure measurement, remains rarely addressed. In the present study, finite element (FE) models were constructed to quantify intra-arm stresses generated by cuff compression, aiming to provide some theoretical evidence for identifying factors of importance for blood pressure measurement or explaining clinical observations. Obtained results showed that the simulated tissue stresses were highly sensitive to the distribution of cuff pressure on the arm surface and the contact condition between muscle and bone. In contrast, the magnitude of cuff pressure and small variations in elastic properties of arm soft tissues had little influence on the efficiency of pressure transmission in arm tissues. In particular, it was found that a thickened subcutaneous fat layer in obese subjects significantly reduced the effective pressure transmitted to the brachial artery, which may explain why blood pressure overestimation occurs more frequently in obese subjects in noninvasive blood pressure measurement.
Du Toit, Hanette
Background: Nurses are responsible for the monitoring and assessment of blood pressure in the clinical setting. Increasing evidence has demonstrated that inaccurate measurement technique often leads to the misclassification of large numbers of individuals as hypertensive. The impact of untreated or poorly treated hypertension, due to misclassification of patients, is a major contributor to the overall burden of adult diseases in any population. Accurate measurement of blood pressure relies on...
Amar M Taksande
Full Text Available Background: When the upper arm (UA is inaccessible or a standard-sized blood pressure (BP cuff is unavailable, some healthcare workers use the forearm (FA to measure BP with a mercury sphygmomanometer. Objective: The objective was to determine the accuracy of BP measurement in the arm and FA. Design: Prospective, randomized study. Setting: Department of Pediatrics, JNMC, Sawangi (Meghe Participants: A total of 72 children aged 5-15 years. Measurements: Mercury and Automatic (OMRON Tokyo, 108-0075 Japan BP measurements were recorded from the arm and FA at 2 min intervals. Results: In our study, 72 children of both sexes were enrolled. The mean age of the children was 10.13 ± 2.82 years, and 48% were females. Pearson′s correlation coefficient between FA and UA systolic BP (SBP measured by mercury was 0.782, and for diastolic BP (DBP it was 0.824. Similarly, Pearson′s correlation coefficient between FA and UA SBP measured with an automated device (OMRON was 0.843, and for DBP it was 0.846. The average readings for the SBP and DBP were higher in the FA than in the UA by approximately 3 mmHg. There was a statistically significant difference in both SBP and DBP. Conclusions: The FA is an acceptable method of BP monitoring when the UA cannot be accessed. The pressure from FA is probably higher than it would be from UA.
... from the NHLBI on Twitter. Prevention of High Blood Pressure Healthy lifestyle habits, proper use of medicines, and ... high blood pressure or its complications. Preventing High Blood Pressure Onset Healthy lifestyle habits can help prevent high ...
... dangerous as elevations of both systolic and diastolic pressure. Blood pressure is elevated for two main reasons: too ... and Angiotensin II receptor blockers (ARBs), reduce blood pressure by relaxing blood vessels Beta blockers, which also cause the heart ...
Full Text Available ... of High Blood Pressure For most patients, health care providers diagnose high blood pressure when blood pressure ... painless and can be done in a health care provider’s office or clinic. To prepare for the ...
Full Text Available ... providers diagnose high blood pressure when blood pressure readings are consistently 140/90 mmHg or above. Confirming ... minutes before the test. To track blood pressure readings over a period of time, the health care ...
Yap, Yee Guan; Duong, Trinh; Bland, J Martin; Malik, Marek; Torp-Pedersen, Christian Tobias; Køber, Lars; Connolly, Stuart J; Gallagher, Mark M; Camm, A John
, CAMIAT, SWORD, TRACE and DIAMOND-MI studies with left ventricular ejection fraction less than 40% or asymptomatic ventricular arrhythmia surviving more than 45 days after MI were pooled. Systolic and diastolic blood pressures and pulse pressures were measured soon after MI (median 6 days, range 0-53 days......BACKGROUND: The prognostic value of blood pressure measured during hospitalization after acute myocardial infarction (MI) has not been investigated, particularly with regard to arrhythmic death. METHODS: A total of 3311 placebo patients (2612 men, median age 64 years; range 23-92) from the EMIAT...... pressure measured during hospitalization after acute MI significantly increased the risk of all-cause mortality [hazard ratio (HR) for 10% increase in systolic blood pressure 0.80, 95% confidence interval (CI) 0.71-0.90; P < 0.001] and arrhythmic mortality (HR 0.73, 95% CI 0.61-0.86; P = 0.001). Reduced...
Bertoglio, Cristobal; Núñez, Rodolfo; Galarce, Felipe; Nordsletten, David; Osses, Axel
The pressure gradient across stenotic blood vessels is an important clinical index for diagnosis of the pathological severity of the cardiovascular disease. While the clinical gold standard for its measurement is invasive catheterization, Phase-Contrast MR-imaging has emerged as a promising tool for enabling a non-invasive quantification of the relative pressures, by linking the (highly spatially resolved) velocity measurements with the relative pressure via the Navier-Stokes equations. In th...
Holstein, P; Lassen, N A
(SPPH) as measured with a photocell. Thirty-two patients (35 feet with ulcerations) had diabetes mellitus. The treatment was conservative. In 42 feet the ulcers healed after an average period of 5.8 months; in 24 feet major amputation became necessary after an average of 4.3 months. The frequency of...... cases (9%) healed. Of the 11 cases with SDBP of 20 to 29 mmHg seven cases (64%) healed and of the 33 cases with SDBP of30 mmHg or above all cases (100%) healed. There was no significant difference between the 35 diabetic feet and the 31 non-diabetic feet as regards the healing rates, although infection...... and peripheral neuropathy were frequent in the diabetic group. The data show that the systolic digital blood pressure is a particularly valuable prognostic parameter....
... Blood Pressure Tools & Resources Stroke More Types of Blood Pressure Medications Updated:Aug 26,2016 Many medications known ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
... High Blood Pressure Tools & Resources Stroke More High Blood Pressure and Women Updated:Aug 13,2014 Many people ... was last reviewed on 08/04/14. High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
Full Text Available Appropriate control of blood pressure (BP is essential for prevention of future cardiovascular events. However, BP control among treated hypertensive patients has been insufficient. Recently, the usefulness of self-measured BP at home (home BP measurement for the management of hypertension has been reported in many studies. We evaluated BP control both at home and in the office among treated hypertensive patients in primary care settings in Japan (the J-HOME study. We found poor control of home and office BPs and clarified some factors affecting control. We also examined factors associated with the magnitude of the white-coat effect, the morning–evening BP difference, and home heart rate in this J-HOME study.
Full Text Available ... Clinical Trials What Are Clinical Trials? Children & Clinical Studies NHLBI ... providers diagnose high blood pressure when blood pressure readings are consistently 140/90 mmHg or above. Confirming High Blood ...
... Clinical Trials What Are Clinical Trials? Children & Clinical Studies NHLBI ... providers diagnose high blood pressure when blood pressure readings are consistently 140/90 mmHg or above. Confirming High Blood ...
... Blood Pressure Page Content What is high blood pressure? Blood pressure is the force of blood against the ... do I know if I have high blood pressure? High blood pressure is often called "the silent killer" because ...
Maltz, Jonathan S; Tison, Geoffrey H; Alley, Hugh F; Budinger, Thomas F; Owens, Christopher D; Olgin, Jeffrey
The integrity of endothelial function in major arteries (EFMA) is a powerful independent predictor of heart attack and stroke. Existing ultrasound-based non-invasive assessment methods are technically challenging and suitable only for laboratory settings. EFMA, like blood pressure (BP), is both acutely and chronically affected by factors such as lifestyle and medication. Consequently, laboratory-based measurements cannot fully gauge the effects of medical interventions on EFMA. EFMA and BP have, arguably, comparable (but complementary) value in the assessment of cardiovascular health. Widespread deployment of EFMA assessment is thus a desirable clinical goal. To this end, we propose a device based on modifying the measurement protocol of a standard electronic sphygmomanometer. The protocol involves inflating the cuff to sub-diastolic levels to enable recording of the pulse waveform before and after vasodilatory stimulus. The mechanical unloading of the arterial wall provided by the cuff amplifies the distension that occurs with each pulse, which is measured as a pressure variation in the cuff. We show that the height of the rising edge of each pulse is proportional to the change in lumen area between diastole and systole. This allows the effect of vasodilatory stimuli on the artery to be measured with high sensitivity. We compare the proposed cuff flow-mediated dilation (cFMD) method to ultrasound flow-mediated dilation (uFMD). We find significant correlation (r = 0.55, p = 0.003, N = 27) between cFMD- and uFMD-based metrics obtained when the release of a 5 min cuff occlusion is employed to induce endothelial stimulus via reactive hyperemia. cFMD is approximately proportional to the square of uFMD, representing a typical increase in sensitivity to vasodilation of 300-600%. This study illustrates the potential for an individual to conveniently measure his/her EFMA by using a low-cost reprogrammed home sphygmomanometer. PMID:26393958
Dennis; Bloomfield; Alex; Park
The advent of ambulatory blood pressure monitoring permitted examination of blood pressures during sleep and recognition of the associated circadian fall in pressure during this period. The fall in pressure,called the "dip",is defined as the difference between daytime mean systolic pressure and nighttime mean systolic pressure expressed as a percentage of the day value. Ten percent to 20% is considered normal. Dips less than 10%,referred to as blunted or absent,have been considered as predicting an adverse cardiovascular event. This view and the broader concept that white coat hypertension itself is a forerunner of essential hypertension is disputable. This editorial questions whether mean arterial pressures over many hours accurately represent the systolic load,whether nighttime dipping varies from measure to measure or is a fixed phenomenon,whether the abrupt morning pressure rise is a risk factor or whether none of these issues are as important as the actual night time systolic blood pressure itself. The paper discusses the difference between medicated and nonmedicated white coat hypertensives in regard to the cardiovascular risk and suggests that further work is necessary to consider whether the quality and duration of sleep are important factors.
2010319 Effects of combined application of Xuezhikang capsule with hypotensive drugs on arterial compliance and smoothness of the dynamic blood pressure. ZHU Zongtao(朱宗涛),et al. Dept Cardiol, Centr People’s Hosp, Tengzhou 277500.Chin J Integr Tradit & West Med 2010;30
Midttun, Mette; Azad, B.B.S.; Broholm, R;
Distal blood pressure and local skin perfusion pressure were compared to measurement of blood flow rate (BFR) measured by the heat-washout method in orthopaedic patients with and without diabetes, all with a foot ulcer in one foot, compared to healthy controls. The correlation was good between heat......-washout and distal blood pressure in patients with diabetes with and without an ulcer (P = 0·024 and 0·059, respectively). The correlation was weak in patients without diabetes with and without an ulcer, most probably due to power problems (P = 0·118 and 0·116, respectively). The correlation in the healthy...... oxygen from the surrounding tissue, and therefore, measurements are easier made in these subjects. BFR in the first toe increased significantly in all patients when the foot was moved from heart level to 50 cm below heart level (P = between 0·03 and 0·05) as previously seen in patients with claudication...
Nguyen, Xuan P.; Kronemayer, Ralf; Herrmann, Peter; Mejía, Atila; Daw, Zamira; Nguyen, Xuan D.; Kränzlin, Bettina; Gretz, Norbert
In the present article, we describe the validation of a new non-invasive method for measuring blood pressure (BP) which also enables to determine the three BP values: systolic, diastolic and mean value. Our method is based on the pulse transit time (PTT) measurement along an artery directly at the BP cuff. The accuracy of this method was evaluated by comparison with the direct simultaneous measurement of blood pressure from 40 anesthetized female mice. Close correlation ...
FOKKEMA, DS; SMIT, K; VANDERGUGTEN, J; KOOLHAAS, JM
Behavioral and physiological responses of 18 chronically cannulated male TMD-S3 rats were assessed during various social interactions with conspecifics, both with and without the possibility for physical contact (social vs. psychosocial stimulation). Response magnitudes (behavior, blood pressure, pl
Blood pressure measurement in the diagnosis and management of hypertension, including the technique required for ambulatory blood pressure monitoring and home blood pressure monitoring, will be reviewed in this article. Home and ambulatory measurements are widely used, both to confirm the diagnosis and to improve adherence to therapy. The major advantage of out-of-office blood pressure monitoring is that it provides a large number of blood pressure measurements away from the medical environment, which represents a more reliable assessment of actual blood pressure than office blood pressure. The advantage of ambulatory blood pressure monitoring is its unique ability to measure nocturnal blood pressure. Although not fully validated in large-scale clinical trials, ambulatory blood pressure monitoring appears to correlate best with prognosis. Ambulatory blood pressure monitoring and home blood pressure monitoring provide somewhat different information on the subject's blood pressure status, and the two methods should thus be regarded as complementary, rather than competitive or alternative. PMID:25019016
Celler, Branko G; Basilakis, Jim; Goozee, Kathryn; Ambikairajah, Eliathamby
Accurate non-invasive measurement of blood pressure in unsupervised environments continues to be a challenge, particularly in the presence of movement artefact, electrical noise and most importantly cardiac arrhythmia which are common in those aged over 65 suffering from a range of chronic conditions. Large intra personal variability in signal morphometry and amplitudes further complicates the development of reliable signal processing algorithms for NIBP measurement. In this paper we demonstrate the effect of this variability and propose that the traditional methods of human blood pressure determination by sphygmomanometry should no longer be considered a gold standard for the calibration of NIBP devices. PMID:26737650
... Pressure Tools & Resources Stroke More Stroke and High Blood Pressure Updated:Jan 6,2015 Stroke is a leading ... heart disease and stroke. Start exploring today ! High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
... Pressure Tools & Resources Stroke More Potassium and High Blood Pressure Updated:Mar 1,2016 A diet that includes ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
... Pressure Tools & Resources Stroke More Myths About High Blood Pressure Updated:Aug 12,2014 You CAN manage your ... was last reviewed on 08/04/2014. High Blood Pressure • Home • About High Blood Pressure (HBP) Introduction What ...
Full Text Available ... pressure. Using the results of your blood pressure test, your health care provider will diagnose prehypertension or high blood pressure ... same age, gender, and height . Once your health care provider ... he or she can order additional tests to determine if your blood pressure is due ...
Benmira, A; Perez-Martin, A; Schuster, I; Aichoun, I; Coudray, S; Bereksi-Reguig, F; Dauzat, M
The auscultatory technique remains the point of reference for the validation of non-invasive blood pressure measurement devices, although the exact origin of the Korotkoff sounds is still debated and comparison with intra-arterial measurement shows limits and pitfalls. Automatic oscillometric devices are now widely used by nurses, physicians, and patients. However, many available devices have not been duly validated. Moreover, they calculate systolic and diastolic blood pressures using undisclosed algorithms. Therefore, these devices are not interchangeable, and their reliability may be questionable in some clinical situations. Nevertheless, oscillometry is increasingly used, beside NIBP, for the assessment of central blood pressure and systemic arterial wall stiffness. Awareness of its limits and causes of error is all the more necessary. PMID:26641026
Mehlsen, Jesper; Wiinberg, Niels
The present study aimed at examining the interarm difference in blood pressure and its use as an indicator of peripheral arterial disease (PAD). Data were included from consecutive patients referred from their general practitioner to our vascular laboratory for possible PAD aged 50 years or older...... without known cardiac disease, renal disease, or diabetes mellitus. 824 patients (453 women) with mean age of 72 years (range: 50-101) were included. 491 patients had a diagnosis of hypertension and peripheral arterial disease (PAD) was present in 386 patients. Systolic blood pressure was 143 ± 24 mm......Hg and 142 ± 24 mmHg on the right and left arm, respectively (P = 0.015). The interarm difference was greater in patients with hypertension (P = 0.002) and PAD (P < 0.0005). 443 patients were measured on two separate occasions and the interarm difference for systolic blood pressure was reproducible for...
Contreras, Danitza J; Vogel, Eugenio E; Saravia, Gonzalo; Stockins, Benjamin
We provide ambulatory blood pressure (BP) exams with tools based on information theory to quantify fluctuations thus increasing the capture of dynamic test components. Data from 515 ambulatory 24-hour BP exams were considered. Average age was 54 years, 54% were women, and 53% were under BP treatment. The average systolic pressure (SP) was 127 ± 8 mm Hg. A data compressor (wlzip) designed to recognize meaningful information is invoked to measure mutability which is a form of dynamical variability. For patients with the same average SP, different mutability values are obtained which reflects the differences in dynamical variability. In unadjusted linear regression models, mutability had low association with the mean systolic BP (R(2) = 0.056; P < .000001) but larger association with the SP deviation (R(2) = 0.761; P < .001). Wlzip allows detecting levels of variability in SP that could be hazardous. This new indicator can be easily added to the 24-hour BP monitors improving information toward diagnosis. PMID:26965751
Mietsch, M; Einspanier, A
The common marmoset (Callithrix jacchus, C. j.) is an established primate model in biomedical research and for human-related diseases. Monitoring of cardiovascular parameters including blood pressure (BP) is important for the health surveillance of these experimental animals and the quantification of diseases or pharmaceutical substances influencing BP. Measurement guidelines for C. j. do not exist yet; therefore, the present study was carried out to establish a practicable protocol based on recommendations of the American College of Veterinary Internal Medicine (ACVIM). Furthermore, BP data of 49 marmosets (13.8-202.4 months of age) were obtained via high-definition oscillometry to further knowledge of physiological parameters and gender-related differences in this primate. The thighs proved to be the most suitable measurement localization, since systolic values were less variable (left 4.03 ± 2.90%, right 5.96 ± 2.77%) compared with the tail (12.7 ± 6.96%). BP values were similar in the morning and in the afternoon (P > 0.05). Data were highly reproducible within and between several sessions on three consecutive days (P > 0.05) as well as over the course of 20 months (P > 0.05). Furthermore, the measurement time for females was significantly shorter than for males (5:14 ± 1:59 min versus 6:50 ± 1:58 min, P = 0.007). Measurement recommendations for the common marmoset were successfully established. Standardized values enabled a reliable comparison of BP parameters, e.g. for cardiovascular, toxicological or metabolic research. PMID:25552521
Chia, F; Ang, A T; Wong, T W; Tan, K W; Fung, K P; Lee, J; Khin, K
Four hundred thirty-one paired sets of readings of systolic and diastolic blood pressure and 438 paired sets of readings of mean arterial BP from 49 ill newborns, including 21 very low birth weight infants, were analyzed for the extent and pattern of agreement and the linear relationship between the Dinamap oscillometric monitor and the direct intraarterial blood pressure readings. Agreement between the two methods was measured by the intraclass correlation, whereas the linear relationship was assessed by the product-moment correlation. The intraclass correlations for systolic, diastolic and mean blood pressures were 0.696, 0.766, and 0.781, respectively. The product-moment correlations for systolic, diastolic and mean blood pressures were 0.706, 0.768, and 0.786, respectively. BP measurements by the Dinamap monitor showed reasonably close agreement to those obtained by the intraarterial mean arterial pressure ranges above 40 mmHg. For mean arterial pressure of 40 mmHg and lower, BP readings by the Dinamap monitor tended to be higher than those obtained by the intraarterial method. These findings appeared to be consistent regardless of the birth weight of the newborn. PMID:2340688
Krähenbühl, B; Nielsen, S L; Lassen, N A
Finger systolic blood pressure (FSP) was measured indirectly in normal subjects and patients with primary Raynaud phenomenon by applying a thin-walled plastic cuff around the finger and a strain gauge more distally to detect volume changes. Inducing a high vascular tone in one or more fingers by...... direct cooling or intra-arterial noradrenaline infusion caused a marked drop in FSP in the exposed fingers, but not in the non-exposed fingers of the same hand. The fact that the non-exposed fingers retained the normal (arm systolic) pressure level is taken to indicate that palmar arch blood pressure...... also remained normal. In the high vascular tone state, a large transmural pressure difference must apparently be established before the digital arteries are forced open. The lowered opening pressure constitutes a manifestation of the closure phenomenon of the digital arteries described in patients with...
İsmail Özanli; Sebahat Tülpar; Yunus Yılmaz; Fatih Yıldız
Objective: The prevalence of hypertension in childhood and adolescence is gradually increasing. We aimed to investigate the blood pressure (BP) values of children aged 7-18 years. Methods: This study was conducted in a total of 3375 (1777 females, 1598 males) children from 27 schools. Blood pressures of children were measured using sphygmomanometer appropriate to arm circumference. Results: A positive relationship was found between systolic blood pressure (SBP) and diastolic blood pr...
Midanik, L T; Resnick, B; Hurley, L B; Smith, E J; Mccarthy, M.
A clinical trial of 204 untreated patients with mild hypertension was conducted to assess the effect of home blood pressure monitoring on blood pressure level, pharmacologic treatment, reduction of risk factors, and use of health services. After 1 year, no statistically significant differences were found between the treatment and control groups. The findings indicate that, while home blood pressure monitoring may be useful, it has no measurable short-term impact on these aspects of blood pres...
Sheppard, James P; Stevens, Richard; Gill, Paramjit; Martin, Una; Godwin, Marshall; Hanley, Janet; Heneghan, Carl; Hobbs, F D Richard; Mant, Jonathan; McKinstry, Brian; Myers, Martin; Nunan, David; Ward, Alison; Williams, Bryan; McManus, Richard J
Patients often have lower (white coat effect) or higher (masked effect) ambulatory/home blood pressure readings compared with clinic measurements, resulting in misdiagnosis of hypertension. The present study assessed whether blood pressure and patient characteristics from a single clinic visit can accurately predict the difference between ambulatory/home and clinic blood pressure readings (the home-clinic difference). A linear regression model predicting the home-clinic blood pressure difference was derived in 2 data sets measuring automated clinic and ambulatory/home blood pressure (n=991) using candidate predictors identified from a literature review. The model was validated in 4 further data sets (n=1172) using area under the receiver operator characteristic curve analysis. A masked effect was associated with male sex, a positive clinic blood pressure change (difference between consecutive measurements during a single visit), and a diagnosis of hypertension. Increasing age, clinic blood pressure level, and pulse pressure were associated with a white coat effect. The model showed good calibration across data sets (Pearson correlation, 0.48-0.80) and performed well-predicting ambulatory hypertension (area under the receiver operator characteristic curve, 0.75; 95% confidence interval, 0.72-0.79 [systolic]; 0.87; 0.85-0.89 [diastolic]). Used as a triaging tool for ambulatory monitoring, the model improved classification of a patient's blood pressure status compared with other guideline recommended approaches (93% [92% to 95%] classified correctly; United States, 73% [70% to 75%]; Canada, 74% [71% to 77%]; United Kingdom, 78% [76% to 81%]). This study demonstrates that patient characteristics from a single clinic visit can accurately predict a patient's ambulatory blood pressure. Usage of this prediction tool for triaging of ambulatory monitoring could result in more accurate diagnosis of hypertension and hence more appropriate treatment. PMID:27001299
INTRODUCTION: Orthostatic hypotension (OH) is associated with troublesome symptoms and increased mortality. It is treatable and deserving of accurate diagnosis. This can be time consuming. The current reference standard for its diagnosis is head-up tilt (HUT) testing with continuous beat-to-beat plethysmography. Our objective was to assess the accuracy of sit-stand testing with semi-automatic sphygmomanometry for the diagnosis of OH. DESIGN: Retrospective test of diagnostic accuracy. METHODS: This was a retrospective study performed using a database maintained by a busy syncope unit. HUT testing was performed using an automated tilt table with Finometer monitoring. A 3 min 70 degrees HUT was performed following 5 min supine. Sitting blood pressure (BP) was measured following 3 min rest. Standing BP was measured within 30 s of assuming the upright posture. The results of sit-stand testing were compared with HUT testing as a reference standard. Both tests happened within 5 min of each other and patients underwent no intervention between tests. RESULTS: From a total of 1452 consecutive HUTs, we identified 730 with pre-test measures of sitting and standing BP. The mean age of this group was 70.57 years (SD = 15.1), 62% were female. The sensitivity of sit-stand testing was calculated as 15.5%, specificity as 89.9%, positive predictive value as 61.7%, negative predictive value as 50.2% and the likelihood ratio as 1.6. The area under the Receiver Operator Curve was 0.564. CONCLUSION: We have demonstrated that sit-stand testing for OH has very low diagnostic accuracy. We recommend that the more time-consuming reference standard method of diagnosis be used if the condition is suspected.
Schoenfeld, A; Ziv, I; Tzeel, A; Ovadia, J
In order to make the earliest possible prediction of the type of woman likely to develop pregnancy-induced hypertension (PIH), one hundred and ninety-six primigravidas underwent a roll-over test (ROT) during wk 28-32 of their pregnancy. Blood pressure (BP) readings were taken with a standard 12 cm cuff as well as with cuffs adapted to various arm circumferences. We found that the prediction rate of ROT readings with a standard 12 cm cuff was relatively low (38.5%) as compared with Gant's study (94%) (Amer. J. Obstet. Gynec., 120 (1974) 1). When a suitably sized cuff was used, the prediction rate dropped (to 14.7%). Data analysis at term for the whole population of this study shows that, by measuring with a standard 12 cm cuff, 10.2% of the women were found to have PIH, whereas measuring with a suitable cuff showed PIH in only 2.55% of the cases (1:4 ratio). We suggest that the low prediction rates in this and other studies demonstrate that the ROT test is not sufficiently reliable as a tool for predicting which women are liable to develop PIH, but there is definitely enough in it to predict which group will not develop PIH (in this study 89-93%). It has been recommended that ROT be considered only as a test of possible reliability. It should be done according to proper criteria for BP measuring, and a repeat ROT should be considered after several days before starting any kind of treatment. PMID:3979651
Full Text Available ... above. Confirming High Blood Pressure A blood pressure test is easy and painless and can be done ... provider’s office or clinic. To prepare for the test: Don’t drink coffee or smoke cigarettes for ...
... this page: //medlineplus.gov/ency/article/007483.htm High blood pressure and diet To use the sharing features on ... diet is a proven way to help control high blood pressure . These changes can also help you lose weight ...
Full Text Available ... track blood pressure readings over a period of time, the health care provider may ask you to ... the office on different days and at different times to take your blood pressure. The health care ...
Full Text Available ... visit the health care provider, he or she should tell you what your blood pressure numbers are; if he or she does not, you should ask for your readings. Blood Pressure Severity and ...
... of the baby. Controlling your blood pressure during pregnancy and getting regular prenatal care are important for ... your baby. Treatments for high blood pressure in pregnancy may include close monitoring of the baby, lifestyle ...
... Research Matters NIH Research Matters September 26, 2011 Genes that Influence Blood Pressure In one of the ... 16 previously unknown variations. Six were found in genes already suspected of regulating blood pressure. The remaining ...
De Mey, C; Schroeter, V; Butzer, R; Roll, S; Belz, G G
1. The agreement of blood pressure measurements by stethoscope auscultation (SBPa, DBPa-IV and DBPa-V), oscillometry (Dinamap; SBPo, and DBPo) and digital photoplethysmography (Finapres; SBPf, and DBPf) with the graphical analysis of the analogue microphone signals of vascular wall motion sound (SBPg and DBPg) was evaluated in eight healthy subjects in the presence of responses to the intravenous infusion of 1 microgram min-1 isoprenaline. 2. In general, there was good agreement between the SBP/DBP-measurements based on auscultatory Korotkoff-I- and IV-criteria and the reference method; the average method difference in estimating the isoprenaline responses for SBPa-SBPg was: -1.1, 95% CI: -5.4 to 3.1 mm Hg with a within-subject between-method repeatability coefficient (REP) of 11.6 mm Hg and for DBPa-IV-DBPg: 3.5, 95% CI: -0.5 to 6.5 mm Hg, REP: 11.5 mm Hg. The ausculatation of Korotkoff-V substantially overestimated the isoprenaline induced reduction of DBP: method difference DBPa-V-DBPg: -11.3, 95% CI: -17.8 to -4.7 mm Hg, REP: 31.8 mm Hg. 3. Oscillometry yielded good approximations for the SBP response to isoprenaline (average method difference SBPo-SBPg: -2.9, 95% CI: -9.0 to 3.3 mm Hg, REP: 17.6 mm Hg) but was poorly sensitive with regard to the DBP responses: method difference DBPo-DBPg: 6.5, 95% CI: -1.3 to 14.3 mm Hg, REP: 25.7 mm Hg. 4. Whilst the finger pulse pressure agreed well with regard to DBP (method difference for the DBP responses to isoprenaline: DBPf-DBPg: 1.8, 95% CI: -5.1 to 8.6 mm Hg, REP: 18.5 mm Hg) it was rather unsatisfactory with regard to SBP (method difference SBPf-SBPg: -14.1, 95% CI: -28.2 to -0.1 mm Hg, REP: 49.9 mm Hg).(ABSTRACT TRUNCATED AT 250 WORDS) PMID:8554929
Pogue, Velvie; Rahman, Mahboob; Lipkowitz, Michael; Toto, Robert; Miller, Edgar; Faulkner, Marquetta; Rostand, Stephen; Hiremath, Leena; Sika, Mohammed; Kendrick, Cynthia; Hu, Bo; Greene, Tom; Appel, Lawrence; Phillips, Robert A
Ambulatory blood pressure (ABP) monitoring provides unique information about day-night patterns of blood pressure (BP). The objectives of this article were to describe ABP patterns in African Americans with hypertensive kidney disease, to examine the joint distribution of clinic BP and ABP, and to determine associations of hypertensive target organ damage with clinic BP and ABP. This study is a cross-sectional analysis of baseline data from the African American Study of Kidney Disease Cohort Study. Masked hypertension was defined by elevated daytime (>or= 135/85 mm Hg) or elevated nighttime (>or= 120/70 mm Hg) ABP in those with controlled clinic BP (disease, in large part because of increased nighttime BP. Whether lowering nighttime BP improves clinical outcomes is unknown but should be tested given the substantial burden of BP-related morbidity in this population. PMID:19047584
Non-invasive monitoring of blood pressure has become increasingly important in research. High-Definition Oscillometry (HDO) delivers not only accurate, reproducible and thus reliable blood pressure but also visualises the pulse waves on screen. This allows for on-screen feedback in real time on data validity but even more on additional parameters like systemic vascular resistance (SVR), stroke volume (SV), stroke volume variances (SVV), rhythm and dysrhythmia. Since complex information on drug effects are delivered within a short period of time, almost stress-free and visible in real time, it makes HDO a valuable technology in safety pharmacology and toxicology within a variety of fields like but not limited to cardiovascular, renal or metabolic research. PMID:26091643
Staessen, J.A.; Thijs, L.; Ohkubo, T.; Kikuya, M.; Richart, T.; Boggia, J.; Adiyaman, A.; Dechering, D.G.; Kuznetsova, T.; Thien, Th.; Leeuw, P de; Imai, Y.; O'brien, E.; Parati, G.
OBJECTIVE: The goal of this review study is to summarize 30 years of research on cut-off limits for the self-measured blood pressure. METHODS: We reviewed two meta-analyses, several prospective outcome studies in populations and hypertensive patients, studies in pregnant women, three clinical trials
Jurák, Pavel; Kára, T.; Halámek, Josef; Kuba, R.; Chládek, Jan; Vondra, Vlastimil
Liptovský Mikuláš : Vojenská Akadémia FZV, 2002 - (Puttera, J.), s. 96 - 99 ISBN 80-8040-180-2. [KTERP. Tatranské Zruby (SK), 24.04.2002-26.04.2002] R&D Projects: GA ČR GA102/00/1262 Institutional research plan: CEZ:AV0Z2065902 Keywords : Vagus Nerve Stimulation * heart rate * blood pressure Subject RIV: FA - Cardiovascular Diseases incl. Cardiotharic Surgery
Temporal variability of parameters which describe dynamic cerebral autoregulation (CA), usually quantified by the short-term relationship between arterial blood pressure (BP) and cerebral blood flow velocity (CBFV), could result from continuous adjustments in physiological regulatory mechanisms or could be the result of artefacts in methods of measurement, such as the use of non-invasive measurements of BP in the finger. In 27 subjects (61 ± 11 years old) undergoing coronary artery angioplasty, BP was continuously recorded at rest with the Finapres device and in the ascending aorta (Millar catheter, BPAO), together with bilateral transcranial Doppler ultrasound in the middle cerebral artery, surface ECG and transcutaneous CO2. Dynamic CA was expressed by the autoregulation index (ARI), ranging from 0 (absence of CA) to 9 (best CA). Time-varying, continuous estimates of ARI (ARI(t)) were obtained with an autoregressive moving-average (ARMA) model applied to a 60 s sliding data window. No significant differences were observed in the accuracy and precision of ARI(t) between estimates derived from the Finapres and BPAO. Highly significant correlations were obtained between ARI(t) estimates from the right and left middle cerebral artery (MCA) (Finapres r = 0.60 ± 0.20; BPAO r = 0.56 ± 0.22) and also between the ARI(t) estimates from the Finapres and BPAO (right MCA r = 0.70 ± 0.22; left MCA r = 0.74 ± 0.22). Surrogate data showed that ARI(t) was highly sensitive to the presence of noise in the CBFV signal, with both the bias and dispersion of estimates increasing for lower values of ARI(t). This effect could explain the sudden drops of ARI(t) to zero as reported previously. Simulated sudden changes in ARI(t) can be detected by the Finapres, but the bias and variability of estimates also increase for lower values of ARI. In summary, the Finapres does not distort time-varying estimates of dynamic CA obtained with a sliding window combined with an ARMA model, but
Blood pressure measuring represents a routine investigation in general medicine. Nokolai Korotkoff was only 31 years old when he made a short presentation to the Scientific Meeting of the Military Hospital of the Academy on 5 November 1905 concerning an easy non-invasive method of blood pressure (BP) measurement, entitled ‘Concerning the problems of the methods of blood pressure measurement’. If the pressure in the cuff is relieved, blood starts coming through the compressed arterial segment...
Schoot, Tessa S; Weenk, Mariska; van de Belt, Tom H; Engelen, Lucien JLPG; van Goor, Harry
Background Cuffless blood pressure (BP) monitoring devices, based on pulse transit time, are being developed as an easy-to-use, more convenient, fast, and relatively cheap alternative to conventional BP measuring devices based on cuff occlusion. Thereby they may provide a great alternative to BP self-measurement. Objective The objective of our study was to evaluate the performance of the first release of the Checkme Health Monitor (Viatom Technology), a cuffless BP monitor, in a real-life setting. Furthermore, we wanted to investigate whether the posture of the volunteer and the position of the device relative to the heart level would influence its outcomes. Methods Study volunteers fell into 3 BP ranges: high (>160 mmHg), normal (130–160 mmHg), and low (International Protocol (ESH-IP) for the validation of BP measurement devices. After calibrating the Checkme device, we measured systolic BP with Checkme and a validated, oscillometric reference BP monitor (RM). Measurements were performed in randomized order both in supine and in sitting position, and with Checkme at and above heart level. Results We recruited 52 volunteers, of whom we excluded 15 (12 due to calibration failure with Checkme, 3 due to a variety of reasons). The remaining 37 volunteers were divided into low (n=14), medium (n=13), and high (n=10) BP ranges. There were 18 men and 19 women, with a mean age of 54.1 (SD 14.5) years, and mean recruitment systolic BP of 141.7 (SD 24.7) mmHg. BP results obtained by RM and Checkme correlated well. In the supine position, the difference between the RM and Checkme was >5 mmHg in 17 of 37 volunteers (46%), of whom 9 of 37 (24%) had a difference >10 mmHg and 5 of 37 (14%) had a difference >15 mmHg. Conclusions BP obtained with Checkme correlated well with RM BP, particularly in the position (supine) in which the device was calibrated. These preliminary results are promising for conducting further research on cuffless BP measurement in the clinical and
Full Text Available ... Diagnosis of High Blood Pressure For most patients, health care providers diagnose high blood pressure when blood pressure ... and painless and can be done in a health care provider’s office or clinic. To prepare for the ...
Full Text Available ... page from the NHLBI on Twitter. Diagnosis of High Blood Pressure For most patients, health care providers diagnose high ... are consistently 140/90 mmHg or above. Confirming High Blood Pressure A blood pressure test is easy and painless ...
... from the NHLBI on Twitter. Living With High Blood Pressure If you have high blood pressure, the best thing to do is to talk ... care provider and take steps to control your blood pressure by making healthy lifestyle changes and taking medications, ...
Rischell, Robert E.
Additional uses found for equipment described in "Implanted Blood-Pressure-Measuring Device" (GSC-13042). Implanted with device electronic circuitry that measures, interprets, and transmits data via inductive link through patient's skin to external receiver. Receiver includes audible alarm generator activated when patient's blood pressure exceeds predetermined threshold. Also included in receiver a blood-pressure display, recorder, or both, for use by patient or physician.
... to Remember Clinical Trials What is high blood pressure? Blood pressure is the force of blood pushing against ... filtering units called nephrons. [ Top ] How does high blood pressure affect the kidneys? High blood pressure can damage ...
... lead an active, normal life. What Is Blood Pressure? Blood pressure is the force that blood puts on ... and medications. continue Long-Term Effects of High Blood Pressure When someone has high blood pressure, the heart ...
Heishima, Yasuhiro; Hori, Yasutomo; Chikazawa, Seishiro; Kanai, Kazutaka; Hoshi, Fumio; Itoh, Naoyuki
We investigated the basic characteristics of indirect arterial blood pressure (ABP) measurement using a device that combines oscillometry and photoplethysmography in cats. Dobutamine was infused intravenously in four anesthetized cats. Direct ABP was measured by a catheter. Indirect ABP was measured from the left forelimb. Dobutamine significantly elevated both systolic arterial pressure (SAP) and mean arterial pressure (MAP) in a dose-dependent manner. The indirect SAP, MAP and diastolic arterial pressure (DAP) values were closely correlated with the direct ABP values (r=0.88, 0.89 and 0.83, respectively). The mean bias for SAP, MAP and DAP was 3.4, 0.2 and -2.4 mmHg, respectively. The indirect ABP measured by this device may be used to reliably monitor ABP changes in anesthetized cats. PMID:27003226
Omboni, Stefano; Verberk, Willem J
Objective Timely detection of atrial fibrillation (AF) may effectively prevent cardiovascular consequences. However, traditional diagnostic tools are either poorly reliable (pulse palpation) or not readily accessible (ECG) in general practice. We tested whether an automatic oscillometric blood pressure (BP) monitor embedded with an algorithm for AF detection might be effective for opportunistic screening of asymptomatic AF in the community. Setting A community-based screening campaign in an unselected population to verify the feasibility of AF screening with a Microlife WatchBP Office BP monitor with a patented AFIB algorithm. When possible AF was detected (≥2 of 3 BP measurements reporting AF), a doctor immediately performed a single-lead ECG in order to confirm or exclude the presence of the arrhythmia. The main demographic and clinical data were also collected. Participants 220 consecutive participants from an unselected sample of individuals in a small Italian community. Primary and secondary outcome measures Number of patients detected with AF and diagnosed risk factors for AF. Results In 12 of 220 participants, the device detected possible AF during the BP measurement: in 4 of them (1.8%), the arrhythmia was confirmed by the ECG. Patients with AF were more likely to be older (77.0±1.2 vs 57.2±15.2 years, p=0.010), obese (50.0 vs 14.4%, p=0.048) and to suffer from a cardiovascular disease (50.0 vs 10.6%, p=0.014) than patients without AF. Participants with a positive BP AF reading and non-AF arrhythmias (n=8) did not differ in their general characteristics from participants with a negative BP AF reading and were younger than patients with AF (mean age 56.4±14.8, p=0.027; 5 of 8 participants aged <65 years). Conclusions Opportunistic screening of AF by BP measurement is feasible to diagnose this arrhythmia in unaware participants, particularly in those older than 65 years, who are the target patient group recommended by current AF screening
Gingival blood flow under total-combs by functional pressure evaluated with Laser-Doppler Flowmetry, a non-invasive method of blood flow measurement. Microcirculation of gum's capillary system can be measured non-invasive by Laser-Doppler-Flowmetry (LDF). Circulation, defined by the number of floating erythrocytes per unit of time, is measured by a fibro-optical Laser-Doppler-Flowmetry. The task was to examine, if there is any change of gum's circulation during strain and relief. Circulation on defined measurepoints, divided on the four quadrants, was determined among maximal strain and subsequent relief, on one probationer (complete denture bearer). Before every measure session systemic pressure was taken. LDF-value was taken on top of jaw-comb, in doing so, to get reproducible result and a satisfying fixation of the probe, there was made an artificial limb of the upper and lower comb. In the upper comb a dynamometer-box, which determined minimal and maximal comb pressure, was integrated. The received results of the LDF-measurement, expressed as perfusion units (PU) were lower under applied pressure than by pressure points more distant. Hyperemia, resulting during relief, seemed the more intense, the less perfusion was before. This new, non-invasive kind of circulation measurement seems to be quite predestined to be used for gingival diagnostic under artificial limb in the future. (author)
Kotovskaya, Yulia V; Kobalava, Zhanna D; Orlov, Artemy V
Background The objective of this study was to validate the novel integration of oscillometric (Vasotens®) technology into a BPLab® ambulatory blood pressure (BP) monitoring system to measure central BP, the aortic augmentation index, and pulse wave velocity (PWV) compared with the recommended and widely accepted tonometric method. Methods The ARTERY Society guidelines for comparison of PWV measurement techniques were used as the basis for recruitment of 99 individuals (mean age 44±19 years, 5...
Kusk, Karina; Nielsen, D B; Thylstrup, Troels;
Blood pressure self-measurement requires the patient to follow a range of recommendations in order to be considered valid for diagnostic use. We investigated how a lightweight and context-aware system designed for sensing and correcting incorrect usage of home blood pressure self......-measurement devices was perceived by a group of four patients. We found that using a context-aware system for improving adherence is feasible for overcoming the challenges related to moving from the supervised clinical setting to the unsupervised home setting. We also found that patients are willing to accept more...... advanced monitoring systems and increased surveillance, as long as it is only used during self-measurement periods and is proportional to the severity of the disease...
Bachmann, K; Wortmann, A; Engels, G
Indirect arterial blood pressure measurement has not changed substantially since its introduction by Riva-Rocci in 1986, Korotkoff in 1905 and Recklinghausen in 1906. Random measurements in the clinic or practice reflect only incompletely the dynamic nature of the blood pressure. Blood pressure recordings by patients themselves have provided more information through better temporal resolution, however, exact characterization of the pressure response throughout the entire day and, in particular, during physical exertion are not enabled; the latter are especially important with regard to diagnosis and treatment of hypertension. In 1966, therefore, radiotelemetric transmission of direct, continuously-measured arterial blood pressure was developed which enabled beat-to-beat registration of blood pressure, outside the laboratory, during normal daily life and sport activities. The initial results showed a marked variability of the blood pressure during the course of the day (Figure 1). Excessive blood pressure increases were observed during exposure to cold, static and dynamic exercise and to a lesser degree during automobile driving and exposure to heat (Figure 3). Recording of the pressure curves via transmission by radiotelemetry shows a high degree of accuracy and temporal resolution, spatial and situational freedom but is invasive and costly in terms of personnel. The same holds true for direct continuous blood pressure registration and storage on a portable tape recorder. Portable, automatic blood pressure measuring units for ambulatory monitoring employ indirect auscultatory or oscillometric recording with a cuff. As compared with the radiotelemetric direct continuous blood pressure measuring method, the indirect method has subordinate temporal resolution, that is, the measurements are only intermittent.(ABSTRACT TRUNCATED AT 250 WORDS) PMID:2676813
... the NHLBI on Twitter. Risk Factors for High Blood Pressure Anyone can develop high blood pressure; however, age, ... can increase your risk for developing high blood pressure. Age Blood pressure tends to rise with age. About 65 ...
Shiga, Yuka; Nojiri, Fumiyo; Yoshizawa, Atsuto; Shimbo, Takuro; Kawachi, Shoji
INTRODUCTION Thalidomide was available for use over-the-counter between 1958 and 1962, and more than 300 thalidomide-impaired people have been confirmed in Japan. Currently, thalidomide-impaired people are nearing the age of 50 years and sometimes require medical treatment or surgery. However, a sphygmomanometer cannot be used to measure the blood pressure in some thalidomide-impaired people because of upper-limb shortening or hypoplastic defects. We encountered a patient with thalidomide-rel...
Kotovskaya YV; Kobalava ZD; Orlov AV
Yulia V Kotovskaya,1 Zhanna D Kobalava,1 Artemy V Orlov21Propedeutics Department, Peoples’ Friendship University of Russia, 2Competitive System Analysis Department (No 65), National Research Nuclear University MEPhI, Moscow, RussiaBackground: The objective of this study was to validate the novel integration of oscillometric (Vasotens®) technology into a BPLab® ambulatory blood pressure (BP) monitoring system to measure central BP, the aortic augmentation index, and pulse wave ve...
Blood pressure measurement in the diagnosis and management of hypertension, including the technique required for ambulatory blood pressure monitoring and home blood pressure monitoring, will be reviewed in this article. Home and ambulatory measurements are widely used, both to confirm the diagnosis and to improve adherence to therapy. The major advantage of out-of-office blood pressure monitoring is that it provides a large number of blood pressure measurements away from the medical environme...
Gebuhr, Peter Henrik; Soelberg, M; Henriksen, Jens Henrik Sahl
Twenty-nine patients due to have a total hip replacement had their systemic systolic and segmental blood pressures measured prior to operation and 1 and 6 weeks postoperatively. No patients had signs of ischemia. The segmental blood pressure was measured at the ankle and at the toes. A significant...... drop was found in all pressures 1 week postoperatively. The decrease followed the systemic pressure and was restored to normal after 6 weeks. In a group of six patients with preoperatively decreased ankle pressure, a significant transient further decrease in the ankle-toe gradient pressure was found...
Full Text Available Blood pressure measuring represents a routine investigation in general medicine. Nokolai Korotkoff was only 31 years old when he made a short presentation to the Scientific Meeting of the Military Hospital of the Academy on 5 November 1905 concerning an easy non-invasive method of blood pressure (BP measurement, entitled ‘Concerning the problems of the methods of blood pressure measurement’. If the pressure in the cuff is relieved, blood starts coming through the compressed arterial segment during systole and causes auscultatory sound and, the first sound, which, in Korotkoff’s opinion, is a measurement of systolic blood pressure. In the same BP measurement, when normal blood flow is fully restored, the auscultatory sounds disappear and, the last sound, which, in Korotkoff's opinion is a measurement of diastolic blood pressure. Listening to Korotkoff sounds (K-sounds to determine systolic and diastolic blood pressure (BP has been the standard for noninvasive BP measurement in medical practices for nearly 100 years and it is the essential tool used for evaluation and assessment of patients with hypertension and risks of cardiovascular diseases (CVD by physicians and nurses despite limited understanding of the nature of K-sounds. This article focuses particularly on the cardiovascular biomechanics of the first and last auscultatory sound and suggests two new terminologies; Highest systolic blood pressure and Baseline systolic blood pressure to represent the systolic pressure and diastolic pressure, respectively. Experimenting blood pressures on the basis of these two new suggested terminologies may reveal various additional undiscovered aspects of normal BP and abnormal BP. KEY WORDS: Highest systolic blood pressure, Baseline systolic blood pressure, Korotkoff sounds
Joffe, Rachel; Duff, Jonathan; Garcia Guerra, Gonzalo; Pugh, Jodie; Ari R. Joffe
Background The accuracy of arterial lines (AL) using the flush test or stopcock test has not been described in children, nor has the difference between invasive arterial blood pressure (IABP) versus non-invasive cuff (NIBP) blood pressure. Methods After ethics approval and consent, we performed the flush test and stopcock test on AL (to determine over damping, under damping, and optimal damping), and determined the difference (NIBP–IABP) in systolic, diastolic, and mean blood pressure (ΔSBP, ...
刘新平; 王爱芹; 袁芳; 李智艳
目的 探讨无创血压袖带缠绕方向和位置对血压测量的影响.方法 将96例危重患者进行自身对照研究,将传感器置于肱动脉上和上臂外侧,每次间隔5分钟,进行正向和反向测量血压,记录测量数值.结果 不同组间收缩压、舒张压和平均压总体比较差异无统计学意义(P＞ 0.05),进一步行各组间多重比较差异无统计学意义(P＞0.05).结论 心电监护仪无创血压测量时,袖带的缠绕方向和位置不影响血压值的准确性.%Objective To explore the effects of the winding direction and position of noninvasive blood pressure cuff on blood pressure measurement.Methods A self-control study was conducted in 96 critically ill patients.We placed the sensor ofa sphygmomanometer on brachial artery and the lateral side of the upper arm, then measured blood pressure every 5 minutes in forward and reverse directions,and then documented the readings.Results The systolic pressure,diastolic pressure,and mean blood pressure did not differ significantly among different ( P＞ 0.05 ); multiple comparisons did not show significant differences ( P ＞ 0.05 ).Conclusions The winding direction and location of noninvasive blood pressure cuff have no effects on the accuracy of blood measurement by using a ECG monitor.
Full Text Available ... does not, you should ask for your readings. Blood Pressure Severity and Type Your health care provider usually takes 2–3 ... any other location. Health care providers diagnose this type of high blood pressure by reviewing readings in the office and ...
People with diabetes have a higher risk for heart attacks and strokes. Your doctor or nurse should check your blood pressure ... People with diabetes have a higher risk for heart attacks and strokes. Your doctor or nurse should check your blood pressure ...
The aim of this work is to design and create a product for automated measurement of blood pressure by auscultatory method by using a computer. Thesis contains a description of the blood pressure, also there are explained ways of its monitoring, basic rules for its measurement and the methods by which it is possible to measure blood pressure. In the practical part is described measuring system serving to measuring blood pressure by auscultatory method using a computer. There is also shown a de...
Débora C. Alavarce
Full Text Available A medida da pressão arterial faz parte do exame físico e deve ser realizada em toda consulta médica. Para avaliar se a pressão arterial é medida nas consultas médicas, foram consultados 500 prontuários de pacientes de primeira consulta ambulatorial, sendo 335 (67% provenientes de clínicas médicas e cirúrgicas e 165 (33% das clínicas ginecológica e obstétrica. A pressão arterial foi anotada em 39% das consultas realizadas (135±32/85±19 mm Hg, sendo que em 11% destas anotações a pressão diastólica estava acima de 90 mm Hg. Em relação ao diagnóstico prévio de hipertensão arterial verificou-se que: a em 62% dos prontuários não havia esta informação; b 20% referiram ter hipertensão, e em 79% destes a pressão arterial foi anotada; e c 18% referiram não ter hipertensão e 46% destes tiveram registro da pressão arterial. Portanto, a maior parte dos pacientes não teve sua pressão arterial anotada.The blood pressure measure is part of the physical exam and it should be accomplished in every medical consultation. To verify the blood pressure measured in the accomplished medical consultations record of first they were consulted it consults 500 patient's ambulatorial being 335 (67% coming of medical and surgical clinics and 165 (33% of the gynecological and obstetric clinics. The blood pressure was written down in 39% of the accomplished consultations (135+32 mm Hg 85+19 mm Hg, and 11% of this annotations the pressure diastólica was above 90 mm Hg. In relation to the previous diagnosis of hypertension it was verified that: a in 62% of the records there was not this information; b 20% referred to have hypertension, and in 79% of these the arterial pressure was scored; and c 18% referred not to have hypertension and 46% of these had registration of the arterial pressure. Therefore, most of the patients didn't have its logged arterial pressure.
Weiss, S T; Munoz, A.; Stein, A.; Sparrow, D.; Speizer, F E
We examined the relationship of blood lead level to systolic and diastolic blood pressure in a longitudinal study of 89 Boston, MA, policemen. At the second examination blood lead level and blood pressure were measured in triplicate. Blood pressure measurements were taken in a similar fashion in years 3, 4, and 5. Multivariate analysis using a first-order autoregressive model revealed that after adjusting for previous systolic blood pressure, body mass index, age, and cigarette smoking, an el...
... your blood to pass through more easily. Alpha-beta blockers not only reduce nerve impulses, but also make ... with less force. They combine the effects of beta blockers and alpha blockers. Angiotensin-converting enzyme inhibitors (also ...
... failure . Your heart may not be able to circulate enough blood to meet your body's needs. Endocrine ... related to heart disease and stroke. Start exploring today ! Related Tools HBP Risk Calculator HBP Trackers Videos ...
The paper presents a method for the Central Aortic Pressure (CAP) waveform estimation from the measured radial Electrical Bio-Impedance (EBI). The method proposed here is a non-invasive and health-safe approach to estimate the cardiovascular system parameters, such as the Augmentation Index (AI). Reconstruction of the CAP curve from the EBI data is provided by spectral domain transfer functions (TF), found on the bases of data analysis. Clinical experiments were carried out on 30 patients in the Center of Cardiology of East-Tallinn Central Hospital during coronary angiography on patients in age of 43 to 80 years. The quality and reliability of the method was tested by comparing the evaluated augmentation indices obtained from the invasively measured CAP data and from the reconstructed curve. The correlation coefficient r = 0.89 was calculated in the range of AICAP values from 5 to 28. Comparing to the traditional tonometry based method, the developed one is more convenient to use and it allows long-term monitoring of the AI, what is not possible with tonometry probes.
Gebuhr, Peter Henrik; Soelberg, M; Henriksen, Jens Henrik
Twenty-nine patients due to have a total hip replacement had their systemic systolic and segmental blood pressures measured prior to operation and 1 and 6 weeks postoperatively. No patients had signs of ischemia. The segmental blood pressure was measured at the ankle and at the toes. A significant...... drop was found in all pressures 1 week postoperatively. The decrease followed the systemic pressure and was restored to normal after 6 weeks. In a group of six patients with preoperatively decreased ankle pressure, a significant transient further decrease in the ankle-toe gradient pressure was found on...... the operated side. None of the patients had symptoms from the lowered pressure. We conclude that in patients without signs of ischemia, the postoperative segmental pressure decrease is reversible and therefore not dangerous....
Full Text Available ... Budget, Planning, & Legislative Advisory Committees Contact Us FAQs Home » Health Information for the Public » Health Topics » High ... also may ask you to check readings at home or at other locations that have blood pressure ...
Full Text Available ... Explore High Blood Pressure What Is... Other Names Causes Who Is at Risk Signs & Symptoms Diagnosis Treatments ... stress, which can occur during the medical appointment, causes white coat hypertension. Rate This Content: NEXT >> Updated: ...
Full Text Available ... are consistently higher than 120/80 mmHg. Your child’s blood pressure numbers are outside average numbers for children of the same age, gender, and height . Once your health care provider ...
Full Text Available ... Explore High Blood Pressure What Is... Other Names Causes Who Is at Risk Signs & Symptoms Diagnosis Treatments Prevention Living With Clinical Trials Links Related Topics Atherosclerosis DASH Eating Plan Overweight and Obesity Smoking and Your Heart ...
Petersen, M E; Williams, T R; Sutton, R
Simultaneous intra-radial and non-invasive (Finapres, Ohmeda) blood pressures were compared during prolonged head-up tilt, in eight patients (mean age 49 years) with malignant vasovagal syncope. Twelve tilts were performed, of which eight resulted in vasovagal syncope. The mean bias (difference between Finapres and intra-arterial pressures) for systolic pressure was +0.7 mmHg (standard deviation 11.3 mmHg) and for diastolic pressure was +5.4 mmHg (standard deviation 7 mmHg). The within-tilt precision (standard deviation of the bias) of the non-invasive measurements varied between 2.9-12.4 mmHg (median 4.5 mmHg) for systolic comparisons, and 1.6-8.4 mmHg (median 4.4 mmHg) for diastolic comparisons. In all but one tilt highly significant positive increases in both systolic (median 7.1 mmHg) and diastolic bias (median 8.1 mmHg) occurred on tilt with respect to resting pre-tilt levels. Independent of the absolute level of agreement, the non-invasive measurements followed changes in intra-arterial pressure closely, with 89% of beat-to-beat changes in systolic pressure, and 95% of beat-to-beat changes in diastolic pressure followed to within +/- 2 mmHg. This study suggests that the Finapres is well suited for use during diagnostic tilt testing, demonstrating an acceptable within-tilt precision and closely following pressure changes during vasovagal syncope. PMID:8881861
... Talk to your doctor about any over-the-counter medications you are taking or are thinking about taking. Does Hypertension or Preeclampsia During Pregnancy Cause Long-Term Heart and Blood Vessel Problems? The effects of high blood pressure during pregnancy vary depending on the disorder and ...
Full Text Available ... Entire Site Health Topics News & Resources Intramural Research Public Health Topics Education & Awareness Resources Contact The Health Information ... Contact Us FAQs Home » Health Information for the Public » Health Topics » High Blood Pressure » Diagnosis of High Blood ...
... page: //medlineplus.gov/ency/article/000999.htm High blood pressure and eye disease To use the sharing features ... are sent to the brain. Causes High blood pressure can damage blood vessels in the retina. The higher the blood ...
... from the NHLBI on Twitter. How Is High Blood Pressure Treated? Based on your diagnosis, health care providers ... the medicine suspected of causing your high blood pressure. If high blood pressure persists or is first diagnosed as primary ...
... Resources Stroke More Avoid the Consequences of High Blood Pressure Infographic Updated:Jun 19,2014 View a downloadable version of this infographic High Blood Pressure • Home • About High Blood Pressure (HBP) • Why HBP ...
... Research Training & Career Development Grant programs for students, postdocs, and faculty Research at NIDDK Labs, faculty, and ... you have high blood pressure. How can I control or prevent high blood pressure? High blood pressure ...
Lee, Yong Jin
This device provides non-invasive beat-to-beat blood pressure measurements and can be worn over the upper arm for prolonged durations. Phase and waveform analyses are performed on filtered proximal and distal photoplethysmographic (PPG) waveforms obtained from the brachial artery. The phase analysis is used primarily for the computation of the mean arterial pressure, while the waveform analysis is used primarily to obtain the pulse pressure. Real-time compliance estimate is used to refine both the mean arterial and pulse pressures to provide the beat-to-beat blood pressure measurement. This wearable physiological monitor can be used to continuously observe the beat-to-beat blood pressure (B3P). It can be used to monitor the effect of prolonged exposures to reduced gravitational environments and the effectiveness of various countermeasures. A number of researchers have used pulse wave velocity (PWV) of blood in the arteries to infer the beat-to-beat blood pressure. There has been documentation of relative success, but a device that is able to provide the required accuracy and repeatability has not yet been developed. It has been demonstrated that an accurate and repeatable blood pressure measurement can be obtained by measuring the phase change (e.g., phase velocity), amplitude change, and distortion of the PPG waveforms along the brachial artery. The approach is based on comparing the full PPG waveform between two points along the artery rather than measuring the time-of-flight. Minimizing the measurement separation and confining the measurement area to a single, well-defined artery allows the waveform to retain the general shape between the two measurement points. This allows signal processing of waveforms to determine the phase and amplitude changes.
Full Text Available Better understanding is needed for antihypertensive medication initiation and lifestyle modification among younger populations with elevated blood pressure. This study aimed to assess health behavior change after receiving a report of elevated blood pressure among African Americans and Caucasians younger than 50 years old. We used the Coronary Artery Risk Development in Young Adults (CARDIA repository dataset. By examination year twenty, 424 out of 2,478 Caucasian and 2,637 African American participants had received feedback from the CARDIA study due to elevated blood pressure readings. Blood pressure was measured by trained CARDIA researchers at the participant's home and was repeatedly recorded at seven examinations over twenty years. A feedback/referral letter was sent to participants with an elevated blood pressure reading. On average, participants first had an elevated blood pressure reading at the age of 34. After receiving the feedback letter, 44% of the previously undiagnosed participants received a formal diagnosis. In addition, 23% initiated the use of antihypertensive medication if they had not received medication treatment before. Among the participants with at-risk lifestyle behaviors, 40% reduced alcohol consumption, 14% increased exercise level, 11% stopped smoking, and 8% reached normal weight. While none of the studied patient factors were associated with lifestyle modification, age had a positive impact on antihypertensive medication initiation (p<0.05. We found no evidence of differences in health behavior change between African American and Caucasian participants after receiving the feedback letter. This research is one of the first to study what followed after receiving a feedback letter about elevated blood pressure outside of healthcare settings. Although additional referral care and behavior interventions are needed to facilitate medication initiation and lifestyle modification, our observations suggest that providing
Pu, Jia; Chewning, Betty A; Johnson, Heather M; Vanness, David J; Young, Henry N; Kreling, David H
Better understanding is needed for antihypertensive medication initiation and lifestyle modification among younger populations with elevated blood pressure. This study aimed to assess health behavior change after receiving a report of elevated blood pressure among African Americans and Caucasians younger than 50 years old. We used the Coronary Artery Risk Development in Young Adults (CARDIA) repository dataset. By examination year twenty, 424 out of 2,478 Caucasian and 2,637 African American participants had received feedback from the CARDIA study due to elevated blood pressure readings. Blood pressure was measured by trained CARDIA researchers at the participant's home and was repeatedly recorded at seven examinations over twenty years. A feedback/referral letter was sent to participants with an elevated blood pressure reading. On average, participants first had an elevated blood pressure reading at the age of 34. After receiving the feedback letter, 44% of the previously undiagnosed participants received a formal diagnosis. In addition, 23% initiated the use of antihypertensive medication if they had not received medication treatment before. Among the participants with at-risk lifestyle behaviors, 40% reduced alcohol consumption, 14% increased exercise level, 11% stopped smoking, and 8% reached normal weight. While none of the studied patient factors were associated with lifestyle modification, age had a positive impact on antihypertensive medication initiation (pbehavior change between African American and Caucasian participants after receiving the feedback letter. This research is one of the first to study what followed after receiving a feedback letter about elevated blood pressure outside of healthcare settings. Although additional referral care and behavior interventions are needed to facilitate medication initiation and lifestyle modification, our observations suggest that providing blood pressure feedback may have promise as part of a multi-method approach
Jennifer M Ryan
Full Text Available Children with cerebral palsy (CP have poor cardiorespiratory fitness in comparison to their peers with typical development, which may be due to low levels of physical activity. Poor cardiorespiratory fitness may contribute to increased cardiometabolic risk.The aim of this study was to determine the association between sedentary behaviour, physical activity and cardiorespiratory fitness in children with CP. An objective was to determine the association between cardiorespiratory fitness, anthropometric measures and blood pressure in children with CP.This study included 55 ambulatory children with CP [mean (SD age 11.3 (0.2 yr, range 6-17 yr; Gross Motor Function Classification System (GMFCS levels I and II]. Anthropometric measures (BMI, waist circumference and waist-height ratio and blood pressure were taken. Cardiorespiratory fitness was measured using a 10 m shuttle run test. Children were classified as low, middle and high fitness according to level achieved on the test using reference curves. Physical activity was measured by accelerometry over 7 days. In addition to total activity, time in sedentary behaviour and light, moderate, vigorous, and sustained moderate-to-vigorous activity (≥10 min bouts were calculated.Multiple regression analyses revealed that vigorous activity (β = 0.339, p<0.01, sustained moderate-to-vigorous activity (β = 0.250, p<0.05 and total activity (β = 0.238, p<0.05 were associated with level achieved on the shuttle run test after adjustment for age, sex and GMFCS level. Children with high fitness spent more time in vigorous activity than children with middle fitness (p<0.05. Shuttle run test level was negatively associated with BMI (r2 = -0.451, p<0.01, waist circumference (r2 = -0.560, p<0.001, waist-height ratio (r2 = -0.560, p<0.001 and systolic blood pressure (r2 = -0.306, p<0.05 after adjustment for age, sex and GMFCS level.Participation in physical activity, particularly at a vigorous intensity, is
... Consumers Consumer Information by Audience For Women High Blood Pressure--Medicines to Help You Share Tweet Linkedin Pin ... is your pressure when your heart relaxes ( diastolic pressure ). High Blood Pressure Medicines Use this guide to help you ...
... nlm.nih.gov/medlineplus/news/fullstory_157555.html Smartphone Blood-Pressure App Often Wrong, Study Finds Measurements ... HealthDay News) -- A popular app that uses your smartphone to check your blood pressure is inaccurate, missing ...
Dimier-David, L.; Billon, N; Costagliola, D.; Jaillon, P; Funck-Brentano, C
1. Spectral analyses of blood pressure and heart rate oscillations are increasingly used to assess the influences of diseases and drugs on the autonomic nervous system. Such influences can only be interpreted in view of the spontaneous variability of these oscillations. We therefore studied the reproducibility of power spectral analyses of blood pressure and heart rate fluctuations measured by a non-invasive finger plethysmographic method in 24 healthy volunteers. 2. Intra-observer reproducib...
We have developed a non-invasive blood pressure measurement system using pulse transit time (PTT) from the heart to the common carotid artery, measured by using an electrocardiogram (ECG) R-wave and carotid arterial Doppler signals at the anterior neck. In this study, we examined the validity of our system by comparing PTT derived systolic blood pressure (Dopp_SBP) with invasive radial systolic arterial pressure (Inv_SBP) with calibration every 15 min in the ICU setting.Methods: 17 patients u...
Full Text Available Yulia V Kotovskaya,1 Zhanna D Kobalava,1 Artemy V Orlov21Propedeutics Department, Peoples’ Friendship University of Russia, 2Competitive System Analysis Department (No 65, National Research Nuclear University MEPhI, Moscow, RussiaBackground: The objective of this study was to validate the novel integration of oscillometric (Vasotens® technology into a BPLab® ambulatory blood pressure (BP monitoring system to measure central BP, the aortic augmentation index, and pulse wave velocity (PWV compared with the recommended and widely accepted tonometric method.Methods: The ARTERY Society guidelines for comparison of PWV measurement techniques were used as the basis for recruitment of 99 individuals (mean age 44±19 years, 52 males. The standard for comparison was the conventional “classic” SphygmoCor device.Results: Accordance of the two methods was satisfactory (r=0.98, mean difference of 2.9±3.5 mmHg for central systolic BP; r=0.98, mean difference of −1.1±2.3 mmHg for central diastolic BP; r=0.83, mean difference of −2.6%±13% for aortic augmentation index; r=0.85, mean difference of 0.69±1.4 for PWV.Conclusion: The performance of Vasotens algorithms using an oscillometric ambulatory BP monitoring system is feasible for accurate diagnosis, risk assessment, and evaluation of the effects of antihypertensive drugs.Keywords: validation study, ambulatory, 24-hour, monitoring, arterial stiffness, pulse wave velocity, augmentation index, central blood pressure
Arveschoug, Anne Kirstine; Vammen, Birthe; Yoshinaka, Emmy;
. Material and methods. Group I comprised 31 healthy persons aged between 61 and 87 years and group II 14 healthy middle-aged hospital staff members aged between 45 and 58 years. Strict rules of inclusion were followed. Results. For group I, significantly greater gradients (DBPtoe - systolic arm blood...
... patientinstructions/000770.htm DASH diet to lower high blood pressure To use the sharing features on this page, ... Hypertension. The DASH diet can help lower high blood pressure and cholesterol and other fats in your blood. ...
Veye, Florent; Mestre, Sandrine; Perez-Martin, Antonia; Triboulet, Jean
International audience Ultrasound examination is the first line procedure for the diagnosis and follow-up of cardiovascular diseases. Instrumenting an ultrasound probe with a force sensor may improve the non-invasive measurement of arterial biomechanical parameters (diameter, pulsatility, intima-media thickness and flow-dependent dilation) by measuring and controlling the force exerted by the sonographer. We present here the results obtained with this approach coupled with image processing...
In the U.S., nearly one third of the adult population have high blood pressure, the leading risk factor for heart disease and stroke - two of the nation's leading causes of death. Created: 10/2/2012 by National Center for Chronic Disease Prevention and Health Promotion (NCCDPHP). Date Released: 10/17/2012.
Full Text Available ... Explore High Blood Pressure What Is... Other Names Causes Who Is at Risk Signs & Symptoms Diagnosis Treatments Prevention Living With Clinical Trials Links Related Topics Atherosclerosis DASH Eating Plan Overweight and Obesity Smoking and Your Heart Stroke Send a link ...
Full Text Available ... Technology Transfer Clinical Trials What Are Clinical Trials? Children & Clinical Studies NHLBI Trials Clinical Trial Websites News & ... are consistently higher than 120/80 mmHg. Your child’s blood pressure numbers are outside average numbers for ...
Full Text Available ... Health Topics Education & Awareness Resources Contact The Health Information Center Health Professionals Systematic Evidence Reviews & Clinical Practice ... Legislative Advisory Committees Contact Us FAQs Home » Health Information for the Public » Health Topics » High Blood Pressure » ...
Full Text Available ... Related Topics Atherosclerosis DASH Eating Plan Overweight and Obesity Smoking and Your Heart Stroke Send a link ... are consistently higher than 120/80 mmHg. Your child’s blood pressure numbers are outside average numbers for ...
Stiffening of the small artery may be the earliest sign of arteriosclerosis. However, there is no adequate method for directly assessing small arterial stiffness. In this study, the finger arterial elasticity index (FEI) was defined as the parameter n which denotes the curvilinearity of an exponential model of pressure (P)–volume (Va) relationship (Va = a − b exp (−nP)). For the original estimation, the FEI was calculated from a compliance index from the finger photoplethysmogram whilst occluding the finger. A simple estimation of the FEI was devised by utilizing normalized pulse volume instead of the compliance index. Both estimations yielded close agreement with the exponential model in healthy young participants (study 1: n = 19). Since the FEI was dependent on finger mean blood pressure, normalized finger arterial stiffness index (FSI) was defined as standardized residual from their relationship: mean and standard deviation (SD) of the FSI were 50 ± 10 (study 2: n = 174). The mean coefficient of variation of the FSI for four measurements was 5.72% (study 3: n = 6). The mean and SD of the FSI in seven arteriosclerotic patients were 100.0 ± 13.5. In conclusion, the FEI and FSI by simple estimation are valid and useful for arteriosclerosis research
目的：探讨无创血压监护仪在心房纤颤病人血压测量中的准确性。方法对60例心房纤颤病人分别采用无创血压监护仪和汞柱台式血压计进行血压监测，对比其测量结果。结果无创血压监护仪和汞柱台式血压计血压测量结果比较，收缩压无显著差异（P＞0.05），而舒张压有显著差异（P＜0.05）。结论无创血压监护仪在心房纤颤病人血压测量中的准确性较差，值得临床注意。%Objective To investigate the accuracy of non-invasive blood pressure monitor in blood pressure measurement of patients with atrial fibrillation. Methods The blood pressure of 60 patients with atrial fibrillation was measured by using non-invasive blood pressure monitor and mercury sphygmomanometer, respectively. Then the measuring results of blood pressure of non-invasive blood pressure monitor were compared with those of mercury sphygmomanometer. Results There was no signiifcant difference between the measuring results of blood pressure of non-invasive blood pressure and those of mercury sphygmomanometer on systolic pressure (P>0.05) while there were signiifcant differences between the measuring results of blood pressure of non-invasive blood pressure and those of mercury sphygmomanometer on diastolic pressure (P<0.05). Conclusion It is inaccurate to measure the blood pressure of patients with atrial ifbrillation by using non-invasive blood pressure monitor, which should be paid attention to in clinic.
Measurement of cerebral blood flow (CBF) and computed tomography (CT) cisternography were performed in 37 patients with a tentative diagnosis of normal pressure hydrocephalus (NPH) to predict their surgical outcome. The mean CBF of the whole brain was measured quantitatively by single photon emission computed tomography with technetium-99m-hexamethylpropylene amine oxime before surgery. The results of CT cisternography were classified into four patients: type I, no ventricular stasis at 24 hours; type II, no ventricular stasis with delayed clearance of cerebral blush; type III, persistent ventricular stasis with prominent cerebral blush; type IV, persistent ventricular stasis with diminished cerebral blush and/or asymmetrical filling of the sylvian fissures. The mean CBF was significantly lower than that of age-matched controls (p<0.005). Patients with a favorable outcome had a significantly higher mean CBF than patients with an unfavorable outcome (p<0.005). Patients with the type I pattern did not respond to shunting. Some patients with type II and III patterns responded to shunting but improvement was unsatisfactory. Patients with type IV pattern responded well to shunting, and those with a mean CBF of 35 ml/100 g/min or over achieved a favorable outcome. The combination of CBF measurement and CT cisternography can improve the prediction of surgical outcome in patients with suspected NPH. (author)
Hinderliter Alan L
Full Text Available Abstract Background Ambulatory blood pressure monitoring (ABPM is increasingly used to measure blood pressure (BP in research studies. We examined ease of use, comfort, degree of disturbance, reported adverse effects, factors associated with poor tolerability, and association of poor tolerability with data acquisition of 24-hour ABPM using the Oscar 2 monitor in the research setting. Methods Sixty adults participating in a research study of people with a history of borderline clinic BP reported on their experience with ABPM on two occasions one week apart. Poor tolerability was operationalized as an overall score at or above the 75th percentile using responses to questions adapted from a previously developed questionnaire. In addition to descriptive statistics (means for responses to Likert-scaled "0 to 10" questions and proportions for Yes/No questions, we examined reproducibility of poor tolerability as well as associations with poor tolerability and whether poor tolerability was associated with removal of the monitor or inadequate number of BP measurements. Results The mean ambulatory BP of participants by an initial ABPM session was 148/87 mm Hg. After wearing the monitor the first time, the degree to which the monitor was felt to be cumbersome ranged from a mean of 3.0 to 3.8, depending on whether at work, home, driving, or other times. The most bother was interference with normal sleeping pattern (mean 4.2. Wearers found the monitor straightforward to use (mean 7.5. Nearly 67% reported that the monitor woke them after falling asleep, and 8.6% removed it at some point during the night. Reported adverse effects included pain (32%, skin irritation (37%, and bruising (7%. Those categorized as having poor tolerability (kappa = 0.5 between sessions, p = 0.0003 were more likely to report being in fair/poor health (75% vs 22%, p = 0.01 and have elevated 24-hour BP average (systolic: 28% vs 17%, p = 0.56; diastolic: 30% vs 17%, p = 0.37. They were
Mitchell, Patrick; Gregson, Barbara A; Piper, Ian; Citerio, Giuseppe; Mendelow, A David; Chambers, Iain R
Objective To determine the statistical characteristics of blood pressure (BP) readings from a large number of head‐injured patients. Methods The BrainIT group has collected high time‐resolution physiological and clinical data from head‐injured patients who require intracranial pressure (ICP) monitoring. The statistical features of this dataset of BP measurements with time resolution of 1 min from 200 patients is examined. The distributions of BP measurements and their relationship with simultaneous ICP measurements are described. Results The distributions of mean, systolic and diastolic readings are close to normal with modest skewing towards higher values. There is a trend towards an increase in blood pressure with advancing age, but this is not significant. Simultaneous blood pressure and ICP values suggest a triphasic relationship with a BP rising at 0.28 mm Hg/mm Hg of ICP, for ICP up to 32 mm Hg, and 0.9 mm Hg/mm Hg of ICP for ICP from 33 to 55 mm Hg, and falling sharply with rising ICP for ICP >55 mm Hg. Conclusions Patients with head injury appear to have a near normal distribution of blood pressure readings that are skewed towards higher values. The relationship between BP and ICP may be triphasic. PMID:17138594
... pressure from normal systolic blood pressure (120 mm Hg) in middle age to high (160 mm Hg). Normal blood pressure in middle age but a steep rise to very high (200 mm Hg). Moderately high blood pressure (140 mm Hg) in ...