Ceccuzzi, Silvio; Jandieri, Vakhtang; Baccarelli, Paolo; Ponti, Cristina; Schettini, Giuseppe
2016-04-01
Comparison of the beam-shaping effect of a field radiated by a line source, when an ideal infinite structure constituted by two photonic crystals and an actual finite one are considered, has been carried out by means of two different methods. The lattice sums technique combined with the generalized reflection matrix method is used to rigorously investigate the radiation from the infinite photonic crystals, whereas radiation from crystals composed of a finite number of rods along the layers is analyzed using the cylindrical-wave approach. A directive radiation is observed with the line source embedded in the structure. With an increased separation distance between the crystals, a significant edge diffraction appears that provides the main radiation mechanism in the finite layout. Suitable absorbers are implemented to reduce the above-mentioned diffraction and the reflections at the boundaries, thus obtaining good agreement between radiation patterns of a localized line source coupled to finite and infinite photonic crystals, when the number of periods of the finite structure is properly chosen.
Kuzay, Tuncer M.; Shu, Deming
1995-01-01
A photon beam position monitor for use in the front end of a beamline of a high heat flux and high energy photon source such as a synchrotron radiation storage ring detects and measures the position and, when a pair of such monitors are used in tandem, the slope of a photon beam emanating from an insertion device such as a wiggler or an undulator inserted in the straight sections of the ring. The photon beam position monitor includes a plurality of spaced blades for precisely locating the photon beam, with each blade comprised of chemical vapor deposition (CVD) diamond with an outer metal coating of a photon sensitive metal such as tungsten, molybdenum, etc., which combination emits electrons when a high energy photon beam is incident upon the blade. Two such monitors are contemplated for use in the front end of the beamline, with the two monitors having vertically and horizontally offset detector blades to avoid blade "shadowing". Provision is made for aligning the detector blades with the photon beam and limiting detector blade temperature during operation.
Photon propagators at finite temperature
International Nuclear Information System (INIS)
Yee, J.H.
1982-07-01
We have used the real time formalism to compute the one-loop finite temperature corrections to the photon self energies in spinor and scalar QED. We show that, for a real photon, only the transverse components develop the temperature-dependent masses, while, for an external static electromagnetic field applied to the finite temperature system, only the static electric field is screened by thermal fluctuations. After showing how to compute systematically the imaginary parts of the finite temperature Green functions, we have attempted to give a microscopic interpretation of the imaginary parts of the self energies. (author)
Photon Collider Physics with Real Photon Beams
International Nuclear Information System (INIS)
Gronberg, J; Asztalos, S
2005-01-01
Photon-photon interactions have been an important probe into fundamental particle physics. Until recently, the only way to produce photon-photon collisions was parasitically in the collision of charged particles. Recent advances in short-pulse laser technology have made it possible to consider producing high intensity, tightly focused beams of real photons through Compton scattering. A linear e + e - collider could thus be transformed into a photon-photon collider with the addition of high power lasers. In this paper they show that it is possible to make a competitive photon-photon collider experiment using the currently mothballed Stanford Linear Collider. This would produce photon-photon collisions in the GeV energy range which would allow the discovery and study of exotic heavy mesons with spin states of zero and two
Photonic Crystal Polarizing and Non-Polarizing Beam Splitters
International Nuclear Information System (INIS)
Chun-Ying, Guan; Jin-Hui, Shi; Li-Boo, Yuan
2008-01-01
A polarizing beam splitter (PBS) and a non-polarizing beam splitter (NPBS) based on a photonic crystal (PC) directional coupler are demonstrated. The photonic crystal directional coupler consists of a hexagonal lattice of dielectric pillars in air and has a complete photonic band gap. The photonic band structure and the band gap map are calculated using the plane wave expansion (PWE) method. The splitting properties of the splitter are investigated numerically using the finite difference time domain (FDTD) method
Virtual photon spectra for finite nuclei
International Nuclear Information System (INIS)
Wolynec, E.; Martins, M.N.
1988-01-01
The experimental results of an isochromat of the virtual photon spectrum, obtained by measuring the number of ground-state protons emitted by the 16.28 MeV isobaric analogue state in 90 Zr as a function of electron incident energy in the range 17-105 MeV, are compared with the values predicted by a calculation of the E1 DWBA virtual photon spectra for finite nuclei. It is found that the calculations are in excellent agreement with the experimental results. The DWBA virtual photon spectra for finite nuclei for E2 and M1 multipoles are also assessed. (author) [pt
A beam profile monitor for a tagged photon beam
International Nuclear Information System (INIS)
Arends, J.; Breuer, M.; Dahmen, H.D.; Detemple, P.; Noeldeke, G.; Schneider, W.; Zucht, B.
1990-10-01
A beam profile monitor for electron and photon beams is described, which operates at the low intensities encountered in a tagged bremsstrahlung beam environment, typically 10 10 electrons/s and 10 7 photons/s. The method is based on a wire scanner and utilizes the presence of a tagging spectrometer. The accuracy of the measurements can be tuned in a wide range (at the expense of measuring time) to meet the requirements set by the actual beam size. Examples of measured electron and photon beam profiles at the tagged photon beam of the PHOENICS experiment at the electron stretcher ring ELSA are given. (orig.)
A beam profile monitor for a tagged photon beam
Energy Technology Data Exchange (ETDEWEB)
Arends, J.; Breuer, M.; Dahmen, H.D.; Detemple, P.; Noeldeke, G.; Schneider, W.; Zucht, B.
1990-10-01
A beam profile monitor for electron and photon beams is described, which operates at the low intensities encountered in a tagged bremsstrahlung beam environment, typically 10{sup 10} electrons/s and 10{sup 7} photons/s. The method is based on a wire scanner and utilizes the presence of a tagging spectrometer. The accuracy of the measurements can be tuned in a wide range (at the expense of measuring time) to meet the requirements set by the actual beam size. Examples of measured electron and photon beam profiles at the tagged photon beam of the PHOENICS experiment at the electron stretcher ring ELSA are given. (orig.).
Beam profile monitors for a tagged photon beam facility
International Nuclear Information System (INIS)
Arends, J.; Breuer, M.; Dahmen, H.D.; Detemple, P.; Schneider, W.; Urban, D.; Zucht, B.
1991-01-01
A beam profile monitor for electron and photon beams is described, which operates at the low intensities encountered in a tagged bremsstrahlung beam environment, typically 10 10 electrons/s and 10 7 photons/s. The method is based on a wire scanner and utilizes the presence of a tagging spectrometer. The accuracy of the measurements can be tuned in a wide range to meet the requirements set by the actual beam parameters. Examples of measured electron and photon beam profiles at the tagged photon beam of the PHOENICS experiment at the electron stretcher ring ELSA in Bonn are given. (orig.)
Finite element modelling of composite castellated beam
Directory of Open Access Journals (Sweden)
Frans Richard
2017-01-01
Full Text Available Nowadays, castellated beam becomes popular in building structural as beam members. This is due to several advantages of castellated beam such as increased depth without any additional mass, passing the underfloor service ducts without changing of story elevation. However, the presence of holes can develop various local effects such as local buckling, lateral torsional buckling caused by compression force at the flange section of the steel beam. Many studies have investigated the failure mechanism of castellated beam and one technique which can prevent the beam fall into local failure is the use of reinforced concrete slab as lateral support on castellated beam, so called composite castellated beam. Besides of preventing the local failure of castellated beam, the concrete slab can increase the plasticity moment of the composite castellated beam section which can deliver into increasing the ultimate load of the beam. The aim of this numerical studies of composite castellated beam on certain loading condition (monotonic quasi-static loading. ABAQUS was used for finite element modelling purpose and compared with the experimental test for checking the reliability of the model. The result shows that the ultimate load of the composite castellated beam reached 6.24 times than the ultimate load of the solid I beam and 1.2 times compared the composite beam.
Photon-Electron Interaction and Condense Beams
International Nuclear Information System (INIS)
Chattopadhyay, S.
1998-01-01
We discuss beams of charged particles and radiation from multiple perspectives. These include fundamental acceleration and radiation mechanisms, underlying electron-photon interaction, various classical and quantum phase-space concepts and fluctuational interpretations
Beam shaping of light sources using circular photonic crystal funnel
Kumar, Mrityunjay; Kumar, Mithun; Dinesh Kumar, V.
2012-10-01
A novel two-dimensional circular photonic crystal (CPC) structure with a sectorial opening for shaping the beam of light sources was designed and investigated. When combined with light sources, the structure acts like an antenna emitting a directional beam which could be advantageously used in several nanophotonic applications. Using the two-dimensional finite-difference time-domain (2D FDTD) method, we examined the effects of geometrical parameters of the structure on the directional and transmission properties of emitted radiation. Further, we examined the transmitting and receiving properties of a pair of identical structures as a function of distance between them.
Photon interactions in a cesium beam
International Nuclear Information System (INIS)
Nygaard, K.J.; Jones, J.D.; Hebner, R.E. Jr
1974-01-01
Photoionization of excited cesium atoms in the 6 2 P3/2 - state has been studied in a triple crossed-beam experiment. A thermal beam of cesium atoms was intersected by one photon beam of wavelength 8521A that served to excite the atoms and another photon beam with wavelengths below 5060A that served to ionize the excited atoms. The resulting ions were detected with a channel electron multiplier. All background effects were discriminated against by chopping the beam of exciting radiation and by analyzing the net count rate with digital synchronous techniques. The relative cross section for photoionization fo Cs(6 2 P3/2) has been measured from threshold (5060A) to 2500A. The results fall off faster than the theoretical calculations of Weisheit and Norcross
Nuclear photon science with inverse compton photon beam
International Nuclear Information System (INIS)
Fujiwara, Mamoru
2007-01-01
Recent developments of the synchrotron radiation facilities and intense lasers are now guiding us to a new research frontier with probes of a high energy GeV photon beam and an intense and short pulse MeV γ-ray beam. New directions of the science developments with photo-nuclear reactions are discussed. The inverse Compton γ-ray has two good advantages for searching for a microscopic quantum world; they are 1) good emittance and 2) high linear and circular polarizations. With these advantages, photon beams in the energy range from MeV to GeV are used for studying hadron structure, nuclear structure, astrophysics, materials science, as well as for applying medical science. (author)
Photoelectric effect photon beam position monitors
International Nuclear Information System (INIS)
Anon.
1991-01-01
Stability of the electron orbit is of critical importance at the NSLS. Many experimenters utilizing the NSLS photon beams can take full advantage of the small transverse dimensions of the source only if the electron orbit variation is kept below 10 to 20% of the transverse electron bunch size. Clearly the first step in a program to stabilize the orbit is to develop position monitors with the required sensitivity, reliability and dynamic range. Of great importance are monitors detecting the photon beams themselves, and also monitors measuring the position of the electron beam. In this section the authors discuss photon beam position monitors utilizing the photoelectric effects, and in the following section the use of capacitively coupled pick-up electrodes to detect electron beam position will be described. In what follows they shall proceed to consider two generic types of monitor geometries (1) Gap monitors, which are designed with the idea that the fringes of the synchrotron radiation will be measured, and the hot or fundamental beam will pass through the monitor unimpeded. (2) Area monitors, which are comprised of two triangular elements nested together similar to the electrodes of a split ion chamber or the diodes described by Siddons and Kraner or Mitsuhashi et al
Photonuclear physics with low intensity photon beams
International Nuclear Information System (INIS)
Mecking, B.A.
1985-01-01
Experiments in photonuclear physics are discussed that require a low intensity photon beam and large acceptance detectors. This combination is especially suitable for the investigation of photoprocesses on nucleons and light nuclei. A specific experimental setup for the electron stretcher ring ELSA is presented. (orig.)
From Electron Beams to Photon Beams
International Nuclear Information System (INIS)
Ranieri, Alberto
2015-01-01
n this article I try to report at the best the events and the emotions I experienced, together with my colleagues, when I was a young researcher working at the Frascati Center of CNEN. In the middle of 70’s the high energy physics activities carried out in Frascati were transferred from CNEN to INFN (Istituto Nazionale Fisica Nucleare) and the personnel had the chance to chose to continue to work at the CNEN (obviously in a different research field) or to continue to work in high energy physics, but at the INFN. I decided to remain at the CNEN and, consequently, I had to change my research activity. I moved from the high energy accelerators research field to the lasers research field in which, at that time at the CNEN, a new interesting project on “uranium laser isotope separation” was just starting. This article is focused on the theoretical and experimental development activity, carried out in the years 70’s-80’s at the CNEN Frascati Center, on a quite particular kind of laser to be utilized in that project. In this laser the active medium is not made of atoms or molecules but is a beam of free electrons running along a spatially periodic magnetic structure: this laser is the “Free Electron Laser” [it
Finite element modeling of micromachined MEMS photon devices
Evans, Boyd M., III; Schonberger, D. W.; Datskos, Panos G.
1999-09-01
The technology of microelectronics that has evolved over the past half century is one of great power and sophistication and can now be extended to many applications (MEMS and MOEMS) other than electronics. An interesting application of MEMS quantum devices is the detection of electromagnetic radiation. The operation principle of MEMS quantum devices is based on the photoinduced stress in semiconductors, and the photon detection results from the measurement of the photoinduced bending. These devices can be described as micromechanical photon detectors. In this work, we have developed a technique for simulating electronic stresses using finite element analysis. We have used our technique to model the response of micromechanical photon devices to external stimuli and compared these results with experimental data. Material properties, geometry, and bimaterial design play an important role in the performance of micromechanical photon detectors. We have modeled these effects using finite element analysis and included the effects of bimaterial thickness coating, effective length of the device, width, and thickness.
Finite Element Modeling of Micromachined MEMS Photon Devices
International Nuclear Information System (INIS)
Datskos, P.G.; Evans, B.M.; Schonberger, D.
1999-01-01
The technology of microelectronics that has evolved over the past half century is one of great power and sophistication and can now be extended to many applications (MEMS and MOEMS) other than electronics. An interesting application of MEMS quantum devices is the detection of electromagnetic radiation. The operation principle of MEMS quantum devices is based on the photoinduced stress in semiconductors, and the photon detection results from the measurement of the photoinduced bending. These devices can be described as micromechanical photon detectors. In this work, we have developed a technique for simulating electronic stresses using finite element analysis. We have used our technique to model the response of micromechanical photon devices to external stimuli and compared these results with experimental data. Material properties, geometry, and bimaterial design play an important role in the performance of micromechanical photon detectors. We have modeled these effects using finite element analysis and included the effects of bimaterial thickness coating, effective length of the device, width, and thickness
High energy high intensity coherent photon beam for the SSC
International Nuclear Information System (INIS)
Tannenbaum, M.J.
1984-01-01
What is proposed for the 20 TeV protons hitting a fixed target is to make a tertiary electron beam similar to that which is the basis of the tagged photon beam at Fermilab. Briefly, a zero degree neutral beam is formed by sweeping out the primary proton beam and any secondary charged particles. Then the photons, from the decay of π 0 in the neutral beam, are converted to e + e - pairs in a lead converter and a high quality electron beam is formed. This beam is brought to the target area where it is converted to a photon beam by Bremsstrahlung in a radiator
On beam quality and flatness of radiotherapy megavoltage photon beams
International Nuclear Information System (INIS)
Hossain, Murshed; Rhoades, Jeffrey
2016-01-01
Ratio of percentage depth dose (PDD) at two depths, PDD at a depth of 10 cm (PDD 10 ), and beam flatness are monitored regularly for radiotherapy beams for quality assurance. The purpose of this study is to understand the effects of changes in one of these parameters on the other. Is it possible to monitor only the beam flatness and not PDD? The investigation has two components. Naturally occurring i.e., unintended changes in PDD ratio and in-plane flatness for 6 and 10 MV photon beams for one particular Siemens Artiste Linac are monitored for a period of about 4 years. Secondly, deliberate changes in the beam parameters are induced by changing the bending magnet current (BMI). Relationships between various beam parameters for unintended changes as well as deliberate changes are characterized. Long term unintentional changes of PDD ratio are found to have no systematic trend. The flatness in the in plane direction for 6 and 10 MV beams show slow increase of 0.43 and 0.75 % respectively in about 4 years while the changes in the PDD ratio show no such trend. Over 10 % changes in BMI are required to induce changes in the beam quality indices at 2 % level. PDD ratio for the 10 MV beam is found to be less sensitive, while the depth of maximum dose, d max , is more sensitive to the changes in BMI compared to the 6 MV beam. Tolerances are more stringent for PDD 10 than PDD ratio for the 10 MV beam. PDD ratio, PDD 10 , and flatness must be monitored independently. Furthermore, off axis ratio alone cannot be used to monitor flatness. The effect of beam quality change in the absolute dose is clinically insignificant.
Finite-amplitude, pulsed, ultrasonic beams
Coulouvrat, François; Frøysa, Kjell-Eivind
An analytical, approximate solution of the inviscid KZK equation for a nonlinear pulsed sound beam radiated by an acoustic source with a Gaussian velocity distribution, is obtained by means of the renormalization method. This method involves two steps. First, the transient, weakly nonlinear field is computed. However, because of cumulative nonlinear effects, that expansion is non-uniform and breaks down at some distance away from the source. So, in order to extend its validity, it is re-written in a new frame of co-ordinates, better suited to following the nonlinear distorsion of the wave profile. Basically, the nonlinear coordinate transform introduces additional terms in the expansion, which are chosen so as to counterbalance the non-uniform ones. Special care is devoted to the treatment of shock waves. Finally, comparisons with the results of a finite-difference scheme turn out favorable, and show the efficiency of the method for a rather large range of parameters.
Data analysis of photon beam position at PLS-II
Energy Technology Data Exchange (ETDEWEB)
Ko, J.; Shin, S., E-mail: tlssh@postech.ac.kr; Huang, Jung-Yun; Kim, D.; Kim, C.; Kim, Ilyou; Lee, T.-Y.; Park, C.-D.; Kim, K. R. [Pohang Accelerator Laboratory, Pohang, Kyungbuk 790-834 (Korea, Republic of); Cho, Moohyun [Department of Physics, POSTECH, Pohang, Kyungbuk 790-834 (Korea, Republic of)
2016-07-27
In the third generation light source, photon beam position stability is critical issue on user experiment. Generally photon beam position monitors have been developed for the detection of the real photon beam position and the position is controlled by feedback system in order to keep the reference photon beam position. In the PLS-II, photon beam position stability for front end of particular beam line, in which photon beam position monitor is installed, has been obtained less than rms 1μm for user service period. Nevertheless, detail analysis for photon beam position data in order to demonstrate the performance of photon beam position monitor is necessary, since it can be suffers from various unknown noises. (for instance, a back ground contamination due to upstream or downstream dipole radiation, undulator gap dependence, etc.) In this paper, we will describe the start to end study for photon beam position stability and the Singular Value Decomposition (SVD) analysis to demonstrate the reliability on photon beam position data.
Magnetic fields with photon beams: Use of circular current loops
International Nuclear Information System (INIS)
Jette, David
2001-01-01
Strong transverse magnetic fields can produce very large dose enhancements and reductions in localized regions of a patient under irradiation by a photon beam. Through EGS4 Monte Carlo simulations, we have examined the effects of applying a magnetic field produced by a pair of circular current loops to a photon beam penetrating a water phantom of finite thickness. We have indeed found very substantial localized dose enhancements, albeit with no corresponding dose reduction just distal to the region of dose enhancement. (However, dose reduction does occur near the distal end of the phantom.) We have also observed two phenomena to be concerned with, for this configuration: significant broadening of the penumbra close to the current loop, and narrowness of the enhanced dose region in a plane parallel to the planes of the loops. We have also examined the use of a single current loop to produce the magnetic field, and have found great asymmetry in the dose distribution; this asymmetry appears to make it impossible to treat with a single circular magnet a tumor of large dimension extending below the application surface
Measuring Beam Quality of Hollow Core Photonic Crystal Fibers
DEFF Research Database (Denmark)
Shephard, J.D.; Roberts, John; Jones, J.D.C.
2006-01-01
In this paper, the authors measure the quality of the delivered beam from hollow core photonic crystal fibers (HC-PCFs). The$M^2$parameter is determined, and the near- to far-field transition is examined. The influence on these properties due to the presence of a core surround mode is evaluated.......17 for the same output beam. This highlights the need for careful consideration when measuring and describing the beam quality delivered by these novel photonic fibers....
Finite element analysis of rotating beams physics based interpolation
Ganguli, Ranjan
2017-01-01
This book addresses the solution of rotating beam free-vibration problems using the finite element method. It provides an introduction to the governing equation of a rotating beam, before outlining the solution procedures using Rayleigh-Ritz, Galerkin and finite element methods. The possibility of improving the convergence of finite element methods through a judicious selection of interpolation functions, which are closer to the problem physics, is also addressed. The book offers a valuable guide for students and researchers working on rotating beam problems – important engineering structures used in helicopter rotors, wind turbines, gas turbines, steam turbines and propellers – and their applications. It can also be used as a textbook for specialized graduate and professional courses on advanced applications of finite element analysis.
200-MeV bremsstrahlung tagged photon beams at Sendai
International Nuclear Information System (INIS)
Hirose, K.; Chiba, M.; Inoue, M.; Kanda, H.; Kimura, R.; Kino, K.; Kobayashi, Y.; Konno, O.; Maeda, K.; Miyase, H.; Miyamoto, A.; Ohtsuki, T.; Saito, A.; Suda, T.; Takahashi, K.; Tamae, T.; Terasaki, Y.; Terasawa, T.; Tsubota, H.; Tsuruta, T.; Utoyama, M.; Yuuki, H.; Yamaguchi, Y.; Yamazaki, H.
2006-01-01
A new beam line for photonuclear reaction experiments using tagged photons has been constructed to take advantage of the completion of the 1.2-GeV STretcher Booster (STB) ring at the Laboratory of Nuclear Science (LNS), Tohoku University. A photon tagging system was installed at the end of the new beam line. It provides bremsstrahlung tagged photon beams in an energy range from 0.2E 0 to 0.8E 0 MeV at the incident electron energy E 0 with an energy resolution of ΔE/E∼10 -2 . The tagged photon intensity I= 6 photons/s is available for typical photonuclear reaction experiments. We introduce the basic parameters of the tagged photons by showing the commissioning data
Production of high energy photon beam at TAC
International Nuclear Information System (INIS)
Akkurt, I.; Tekin, H. O.; Demir, N.; Cakirli, R. B.; Akkus, B.; Kupa, I.
2010-01-01
When an electron pass through an electric field, the electron loose its part of energy and photon is generated. This process is known as Bremsstrahlung (means 'radiation breaking' in German) and this photon can be used in a variety of different application. The TAC will be first Turkish Accelerator Center (TAC) where a IR-FEL and Beamstrahlung photon beam facilities will be established in first stage. The electrons will be accelerated up to 40 MeV by two LINAC and these beam will be used to generate Bremsstrahlung photon. In this study, the main parameters for Bremsstrahlung photon beam facility will be established at TAC will be detailed and fields to be used Bremsstrahlung beam will also be presented.
Characteristics of photon beams from Philips SL25 linear accelerators
International Nuclear Information System (INIS)
Palta, J.R.; Ayyangar, K.; Daftari, I.; Suntharalingam, N.
1990-01-01
The Philips SL25 accelerator is a multimodality machine offering asymmetric collimator jaws and a new type of beam bending and transport system. It produces photon beams, nominally at 6 and 25 MV, and a scattered electron beam with nine selectable energies between 4 and 22 MeV. Dosimetric characteristics for the 6- and 25-MV photon beams are presented with respect to field flatness, surface and depth dose characteristics, isodose distribution, field size factors for both open and wedged fields, and narrow beam transmission data in different materials
Diagnostic phosphors for photon beams at the ALS and APS
International Nuclear Information System (INIS)
Shu, Deming; Warwick, T.; Johnson, E.D.
1991-01-01
Some tests have been made of phosphors for photon beam diagnosis. Plasma sprayed coatings were checked for damage and decay during a period of several ampere-hours of irradiation from the NSLS Xray ring. Surface temperatures were measured and extrapolated to give an indication of the temperatures expected in more powerful photon beams. Based on our findings we have chosen plasma-sprayed coatings of Europium-doped Yttrium Oxide as the best candidate for use on masks and flags in ''white'' photon beams of the new facilities. 4 refs., 2 figs
The Monte Carlo simulation of the Ladon photon beam facility
International Nuclear Information System (INIS)
Strangio, C.
1976-01-01
The backward compton scattering of laser light against high energy electrons has been simulated with a Monte Carlo method. The main features of the produced photon beam are reported as well as a careful description of the numerical calculation
Study of the exotic Θ+ with polarized photon beams
International Nuclear Information System (INIS)
Zhao Qiang
2004-01-01
We carry out an analysis of the pentaquark Θ + photoproduction with polarized photon beams. Kinematical and dynamical aspects are examined for the purpose of determining Θ + 's spin and parity. It shows that the polarized photon beam asymmetry in association with certain dynamical properties of the production mechanism would provide further information on its quantum numbers. Facilities at SPring-8, JLab, ELSA, and ESRF will have access to them
Optimum steering of photon beam lines in SPEAR
International Nuclear Information System (INIS)
Corbett, W.J.; Fong, B.; Lee, M.; Ziemann, V.
1993-04-01
A common operational requirement for many synchrotron light sources is to maintain steered photon beamlines with minimum corrector strength values. To solve this problem for SPEAR, we employed the Singular Value Decomposition (SVD) matrix-inversion technique to minimize corrector strengths while constraining the photon beamlines to remain on target. The result was a reduction in corrector strengths, yielding increased overhead for the photon-beam position feedback systems
Optical microscope using an interferometric source of two-color, two-beam entangled photons
Dress, William B.; Kisner, Roger A.; Richards, Roger K.
2004-07-13
Systems and methods are described for an optical microscope using an interferometric source of multi-color, multi-beam entangled photons. A method includes: downconverting a beam of coherent energy to provide a beam of multi-color entangled photons; converging two spatially resolved portions of the beam of multi-color entangled photons into a converged multi-color entangled photon beam; transforming at least a portion of the converged multi-color entangled photon beam by interaction with a sample to generate an entangled photon specimen beam; and combining the entangled photon specimen beam with an entangled photon reference beam within a single beamsplitter. An apparatus includes: a multi-refringent device providing a beam of multi-color entangled photons; a condenser device optically coupled to the multi-refringent device, the condenser device converging two spatially resolved portions of the beam of multi-color entangled photons into a converged multi-color entangled photon beam; a beam probe director and specimen assembly optically coupled to the condenser device; and a beam splitter optically coupled to the beam probe director and specimen assembly, the beam splitter combining an entangled photon specimen beam from the beam probe director and specimen assembly with an entangled photon reference beam.
A photon beam position monitor for SSRL beamline 9
International Nuclear Information System (INIS)
Cerino, J.A.; Rabedeau, T.; Bowen, W.
1995-10-01
We present here the concept of a simple one dimensional photon beam position monitor for use with high power synchrotron radiation beams. It has micron resolution, reasonable linearity in an inexpensive design. Most important, is its insensitivity to diffusely scattered low energy radiation from components upstream of the monitor
Creating aperiodic photonic structures by synthesized Mathieu-Gauss beams
Vasiljević, Jadranka M.; Zannotti, Alessandro; Timotijević, Dejan V.; Denz, Cornelia; Savić, Dragana M. Jović
2017-08-01
We demonstrate a kind of aperiodic photonic structure realized using the interference of multiple Mathieu-Gauss beams. Depending on the beam configurations, their mutual distances, angles of rotation, or phase relations we are able to observe different classes of such aperiodic optically induced refractive index structures. Our experimental approach is based on the optical induction in a single parallel writing process.
Design of a novel multi channel photonic crystal fiber polarization beam splitter
Zhao, Yunyan; Li, Shuguang; Wang, Xinyu; Wang, Guangyao; Shi, Min; Wu, Junjun
2017-10-01
A kind of multi channel dual-core photonic crystal fiber polarization beam splitter is designed. We analyze the effects of the lattice parameters and the thickness of gold layer on the beam splitting by the finite element method. Numerical results show that the thickness of metal layer and the size of the air holes near the fiber cores are closely linked with the nature of the polarization beam splitter. We also obtain that extinction ratio can reach -73.87 dB at 1 . 55 μm wavelength and at 1 . 41 μm, 1 . 65 μm extinction ratio can reach 30.8978 dB and 31.1741 dB, respectively. The comparison of the effect on the characteristic of the photonic crystal fiber with coating no gold is also taken into account.
Laboratory Astrophysics Using High Energy Density Photon and Electron Beams
Bingham, Robert
2005-01-01
The development of intense laser and particle beams has opened up new opportunities to study high energy density astrophysical processes in the Laboratory. With even higher laser intensities possible in the near future vacuum polarization processes such as photon - photon scattering with or without large magnetic fields may also be experimentally observed. In this talk I will review the status of laboratory experiments using intense beans to investigate extreme astrophysical phenomena such as supernovae explosions, gamma x-ray bursts, ultra-high energy cosmic accelerators etc. Just as intense photon or electron beams can excite relativistic electron plasma waves or wakefields used in plasma acceleration, intense neutrino beams from type II supernovae can also excite wakefields or plasma waves. Other instabilities driven by intense beams relevant to perhaps x-ray bursts is the Weibel instability. Simulation results of extreme processes will also be presented.
Fast IMRT with narrow high energy scanned photon beams
Energy Technology Data Exchange (ETDEWEB)
Andreassen, Bjoern; Straaring t, Sara Janek; Holmberg, Rickard; Naefstadius, Peder; Brahme, Anders [Department of Medical Radiation Physics, Karolinska Institutet and Stockholm University, P.O. Box 260, SE-171 76 Stockholm (Sweden); Department of Hospital Physics, Karolinska University Hospital, SE-171 76 Stockholm (Sweden); Department of Medical Radiation Physics, Karolinska Institutet and Stockholm University, P.O. Box 260, SE-171 76 Stockholm, Sweden and Department of Hospital Physics, Karolinska University Hospital, SE-171 76 Stockholm (Sweden)
2011-08-15
Purpose: Since the first publications on intensity modulated radiation therapy (IMRT) in the early 1980s almost all efforts have been focused on fairly time consuming dynamic or segmental multileaf collimation. With narrow fast scanned photon beams, the flexibility and accuracy in beam shaping increases, not least in combination with fast penumbra trimming multileaf collimators. Previously, experiments have been performed with full range targets, generating a broad bremsstrahlung beam, in combination with multileaf collimators or material compensators. In the present publication, the first measurements with fast narrow high energy (50 MV) scanned photon beams are presented indicating an interesting performance increase even though some of the hardware used were suboptimal. Methods: Inverse therapy planning was used to calculate optimal scanning patterns to generate dose distributions with interesting properties for fast IMRT. To fully utilize the dose distributional advantages with scanned beams, it is necessary to use narrow high energy beams from a thin bremsstrahlung target and a powerful purging magnet capable of deflecting the transmitted electron beam away from the generated photons onto a dedicated electron collector. During the present measurements the scanning system, purging magnet, and electron collimator in the treatment head of the MM50 racetrack accelerator was used with 3-6 mm thick bremsstrahlung targets of beryllium. The dose distributions were measured with diodes in water and with EDR2 film in PMMA. Monte Carlo simulations with geant4 were used to study the influence of the electrons transmitted through the target on the photon pencil beam kernel. Results: The full width at half-maximum (FWHM) of the scanned photon beam was 34 mm measured at isocenter, below 9.5 cm of water, 1 m from the 3 mm Be bremsstrahlung target. To generate a homogeneous dose distribution in a 10 x 10 cm{sup 2} field, the authors used a spot matrix of 100 equal intensity
Photon beam convolution using polyenergetic energy deposition kernels
International Nuclear Information System (INIS)
Hoban, P.W.; Murray, D.C.; Round, W.H.
1994-01-01
In photon beam convolution calculations where polyenergetic energy deposition kernels (EDKs) are used, the primary photon energy spectrum should be correctly accounted for in Monte Carlo generation of EDKs. This requires the probability of interaction, determined by the linear attenuation coefficient, μ, to be taken into account when primary photon interactions are forced to occur at the EDK origin. The use of primary and scattered EDKs generated with a fixed photon spectrum can give rise to an error in the dose calculation due to neglecting the effects of beam hardening with depth. The proportion of primary photon energy that is transferred to secondary electrons increases with depth of interaction, due to the increase in the ratio μ ab /μ as the beam hardens. Convolution depth-dose curves calculated using polyenergetic EDKs generated for the primary photon spectra which exist at depths of 0, 20 and 40 cm in water, show a fall-off which is too steep when compared with EGS4 Monte Carlo results. A beam hardening correction factor applied to primary and scattered 0 cm EDKs, based on the ratio of kerma to terma at each depth, gives primary, scattered and total dose in good agreement with Monte Carlo results. (Author)
International Nuclear Information System (INIS)
Cho, Yeong-Kwon; Kim, Ki-Hong
2014-01-01
The propagation of optical vortex beams through disordered nonlinear photonic lattices is numerically studied. The vortex beams are generated by using a superposition of several Gaussian laser beams arranged in a radially-symmetric manner. The paraxial nonlinear Schroedinger equation describing the longitudinal propagation of the beam array through nonlinear triangular photonic lattices with two-dimensional disorder is solved numerically by using the split-step Fourier method. We find that due to the spatial disorder, the vortex beam is destabilized after propagating a finite distance and new vortex-antivortex pairs are nucleated at the positions of perfect destructive interference. We also find that in the presence of a self-focusing nonlinearity, the vortex-antivortex pair nucleation is suppressed and the vortex beam becomes more stable, while a self-defocusing nonlinearity enhances the vortex-antivortex pair nucleation.
A novel optical beam splitter based on photonic crystal with hybrid lattices
International Nuclear Information System (INIS)
Zhu Qing-Yi; Fu Yong-Qi; Zhang Zhi-Min; Hu De-Qing
2012-01-01
A novel optical beam splitter constructed on the basis of photonic crystal (PC) with hybrid lattices is proposed in this paper. The band gap of square-lattice PC is so designed that the incident light is divided into several branch beams. Triangular-lattice graded-index PCs are combined for focusing each branch. Computational calculations are carried out on the basis of finite-different time-domain algorithm to prove the feasibility of our design. The waveguide is unnecessary in the design. Thus the device has functions of both splitting and focusing beams. Size of the divided beam at site of full-width at half-maximum is of the order of λ/2. The designed splitter has the advantages that it has a small volume and can be integrated by conventional semiconductor manufacturing process. (electromagnetism, optics, acoustics, heat transfer, classical mechanics, and fluid dynamics)
Dosimetric Uncertainties in Verification of Intensity Modulated Photon Beams
International Nuclear Information System (INIS)
Jurkovic, S.
2010-01-01
The doctoral thesis presents method for the calculation of the compensators' shape to modulate linear accelerators' beams. Characteristic of the method is more strict calculation of the scattered radiation in beams with an inhomogeneous cross-section than it was before. Method could be applied in various clinical situations. It's dosimetric verification was made in phantoms, measuring dose distributions using ionization chambers as well as radiographic film. Therefore, ionization chambers were used for the evaluation of modulator shape and film was used for the evaluation of two-dimensional dose distributions. It is well known that dosimetry of the intensity modulated photon beams is rather complicated regarding inhomogeneity of the dose distribution. The main reason for that is the beam modulator which changes spectral distribution of the beam. Possibility of use different types of detectors for the measurements of dose distributions in modulated photon beams and their accuracy were examined. Small volume ionization chambers, different diodes and amorphus silicon detector and radigraphic film were used. Measured dose distributions were compared between each other as well as with distributions simulated using Monte Carlo particle transport algorithm. In this way the most accurate method for the verification of modulate photon beams is suggested. (author)
Photon beam commissioning of an Elekta Synergy linear accelerator
Al Mashud, Md Abdullah; Tariquzzaman, M.; Jahangir Alam, M.; Zakaria, GA
2017-12-01
The aim of this study is to present the results of commissioning of Elekta Synergy linear accelerator (linac). The acceptance test and commissioning were performed for three photon beams energies 4 MV, 6 MV and 15 MV and for the multileaf collimator (MLC). The percent depth doses (PDDs), in-plane and cross-plane beam profiles, head scatter factors (Sc), relative photon output factors (Scp), universal wedge transmission factor and MLC transmission factors were measured. The size of gantry, collimator, and couch isocenter were also measured.
Dynamics of optical beams with finite beam width
International Nuclear Information System (INIS)
Deng Ximing
1993-01-01
A postulation of the pseudo-polarization energy was introduced to the electromagnetic field in the free space. The angular momentum, velocity of the energy flow, static mass density, diffracted divergence, generalization of the principle of Fermat etc. of the electromagnetic field can be described satisfactorily by using this postulation. In the authors research on the transmission of optical beams for more than ten years, the movement of the electromagnetic field has been divided to an orbital motion and an intrinsic motion, and these motions have been described by only a single cartesian coordinate and its first-order partial differential. In this paper, on the basis of past results, the author uses the energy density of the field to replace the single cartesian coordinate component to make the description more precise and complete. On the other hand, as a basic postulation, a pseudo-polarization energy density is introduced to make the description and analysis of the field movement more abstract, deeper, and clearer. 3 refs
Yu, Tianbao; Huang, Jiehui; Liu, Nianhua; Yang, Jianyi; Liao, Qinghua; Jiang, Xiaoqing
2010-04-10
We propose and simulate a new kind of compact polarizing beam splitter (PBS) based on a photonic crystal ring resonator (PCRR) with complete photonic bandgaps. The two polarized states are separated far enough by resonant and nonresonant coupling between the waveguide modes and the microring modes. Some defect holes are utilized to control the beam propagation. The simulated results obtained by the finite-difference time-domain method show that high transmission (over 95%) is obtained and the polarization separation is realized with a length as short as 3.1 microm. The design of the proposed PBS can be flexible, thanks to the advantages of PCRRs.
To test photon statistics by atomic beam deflection
International Nuclear Information System (INIS)
Wang Yuzhu; Chen Yudan; Huang Weigang; Liu Liang
1985-02-01
There exists a simple relation between the photon statistics in resonance fluorescence and the statistics of the momentum transferred to an atom by a plane travelling wave [Cook, R.J., Opt. Commun., 35, 347(1980)]. Using an atomic beam deflection by light pressure, we have observed sub-Poissonian statistics in resonance fluorescence of two-level atoms. (author)
Imaging high energy photons with PILATUS II at the tagged photon beam at MAX-lab
Energy Technology Data Exchange (ETDEWEB)
Lee, V. [School of Physics, University of Melbourne, Parkville 3010 (Australia)], E-mail: leev@physics.unimelb.edu.au; Peake, D.J.; Sobott, B. [School of Physics, University of Melbourne, Parkville 3010 (Australia); Schroeder, B. [MAX-lab, Lund University, Lund (Sweden); Broennimann, Ch. [DECTRIS Ltd., Baden (Switzerland); Henrich, B. [Paul Scherrer Institut, 5232 Villigen PSI (Switzerland); Hansen, K. [MAX-lab, Lund University, Lund (Sweden); O' Keefe, G.J. [Centre for PET, Austin Hospital, Heidelberg, Victoria 3084 (Australia); School of Physics, University of Melbourne, Parkville 3010 (Australia); Taylor, G.N. [School of Physics, University of Melbourne, Parkville 3010 (Australia); Boland, M.J. [Australian Synchrotron, Clayton, Victoria 3168 (Australia); School of Physics, University of Melbourne, Parkville 3010 (Australia); Thompson, M.N.; Rassool, R.P. [School of Physics, University of Melbourne, Parkville 3010 (Australia)
2009-05-21
In photonuclear experiments precise location of the photon beam relative to the experimental sample is critical. Previously used techniques such as using photographic film to identify the position, intensity and centroid of the beam is time-consuming and a faster method is required. PILATUS is a single-photon-counting pixel detector developed at the Paul Scherrer Institute (PSI), Switzerland. It is a silicon-based, two-dimensional detector with a large dynamic range and zero readout noise. Designed as an X-ray detector, its optimal quantum efficiency is between 3 and 30 keV. This paper reports measurements carried out at the MAX-lab tagged photon facility in Lund, Sweden. The beam endpoint energy of approximately 200 MeV is far above the designed optimal energy detection range of PILATUS, and provides a critical test of the use of PILATUS under high energy conditions. The detector was placed in the photon beam and images were taken both downstream of other experiments, and in close range of a 19 mm collimator. The successful measurements demonstrate the versatility and robustness of the detector and provide an effective way of quickly and accurately monitoring beam position and profile in real time.
The bremsstrahlung tagged photon beam in Hall B at JLab
Sober, D I; Longhi, A; Matthews, S K; O'Brien, J T; Berman, B L; Briscoe, W J; Cole, P L; Connelly, J P; Dodge, W R; Murphy, L Y; Philips, S A; Dugger, M K; Lawrence, D; Ritchie, B G; Smith, E S; Lambert, J M; Anciant, E; Audit, G; Auger, T; Marchand, C; Klusman, M; Napolitano, J; Khandaker, M A; Salgado, C W; Sarty, A J
2000-01-01
We describe the design and commissioning of the photon tagging beamline installed in experimental Hall B at the Thomas Jefferson National Accelerator Facility (JLab). This system can tag photon energies over a range from 20% to 95% of the incident electron energy, and is capable of operation with beam energies up to 6.1 GeV. A single dipole magnet is combined with a hodoscope containing two planar arrays of plastic scintillators to detect energy-degraded electrons from a thin bremsstrahlung radiator. The first layer of 384 partially overlapping small scintillators provides photon energy resolution, while the second layer of 61 larger scintillators provides the timing resolution necessary to form a coincidence with the corresponding nuclear interaction triggered by the tagged photon. The definitions of overlap channels in the first counter plane and of geometric correlation between the two planes are determined using digitized time information from the individual counters. Auxiliary beamline devices are briefl...
Temporal nonlinear beam dynamics in infiltrated photonic crystal fibers
DEFF Research Database (Denmark)
Bennet, Francis; Rosberg, Christian Romer; Neshev, Dragomir N.
Liquid-infiltrated photonic crystal fibers (PCFs) offer a new way of studying light propagation in periodic and discrete systems. A wide range of available fiber structures combined with the ease of infiltration opens up a range of novel experimental opportunities for optical detection and bio...... the evolution of the fiber output beam in the few micro or milliseconds after the beam is turned on. The characterization of the temporal behavior of the thermal nonlinear response provides important information about the nonlocality associated with heat diffusion inside the fiber, thus enabling studies of long...... and technological potential of liquid-infiltrated PCFs it is important to understand the temporal dynamics of nonlinear beam propagation in such structures. In this work we consider thermally induced spatial nonlinear effects in infiltrated photonic crystal fibers. We experimentally study the temporal dynamics...
Ultracompact multiway beam splitters using multiple coupled photonic crystal waveguides
International Nuclear Information System (INIS)
Yu Tianbao; Zhou Haifeng; Yang Jianyi; Jiang Xiaoqing; Wang Minghua; Gong Zhao
2008-01-01
Ultracompact 1 x N (N > 2) beam splitters based on coupling of multiple photonic crystal waveguides (PCWs) are numerically demonstrated. The operation of the devices is on the basis of the self-imaging phenomenon. Variation of the effective index of modified rods induces the transverse redistribution of the N-fold images with the same coupling length, and uniform or free splitting can be achieved. The devices with three and four output channels are discussed in details as examples. Results show that this kind of beam splitters are very short. At the operating wavelength of 1.55 μm, the splitting length of the devices is only 35 μm even if the output channel number reaches 20. It provides a new method and a compact model to export freely the beam to N channels in PCW devices and can find practical applications in future photonic integrated circuits
Ultracompact multiway beam splitters using multiple coupled photonic crystal waveguides
Energy Technology Data Exchange (ETDEWEB)
Yu Tianbao; Zhou Haifeng; Yang Jianyi; Jiang Xiaoqing; Wang Minghua [Department of Information Science and Electronic Engineering, Zhejiang University, 310027 Hangzhou (China); Gong Zhao [Zhejiang University City College, 310027 Hangzhou (China)
2008-05-07
Ultracompact 1 x N (N > 2) beam splitters based on coupling of multiple photonic crystal waveguides (PCWs) are numerically demonstrated. The operation of the devices is on the basis of the self-imaging phenomenon. Variation of the effective index of modified rods induces the transverse redistribution of the N-fold images with the same coupling length, and uniform or free splitting can be achieved. The devices with three and four output channels are discussed in details as examples. Results show that this kind of beam splitters are very short. At the operating wavelength of 1.55 {mu}m, the splitting length of the devices is only 35 {mu}m even if the output channel number reaches 20. It provides a new method and a compact model to export freely the beam to N channels in PCW devices and can find practical applications in future photonic integrated circuits.
Comparison of photon beam qualities for treatment of deep seated tumours
International Nuclear Information System (INIS)
Ssengabi, J.
1977-06-01
Physical parameters that influence the quality of photon beams have been examined. The interaction of photon beams of different qualities from cobolt-60 gamma rays to 42 MV X-rays, with a patient-target region system has been investigated with a view to compare the photon beam qualities under specified irradiation conditions. The concept of integral dose and its use in photon beam intercomparison has been investigated. The results of the study have shown the inadequacy of a single beam parameter, such as the central axis depth dose data, in the intercomparison of photon beam qualities for the treatment of deep seated tumours. (author)
Finite element analysis of FRP-strengthened RC beams
Directory of Open Access Journals (Sweden)
Teeraphot Supaviriyakit
2004-05-01
Full Text Available This paper presents a non-linear finite element analysis of reinforced concrete beam strengthened with externally bonded FRP plates. The finite element modeling of FRP-strengthened beams is demonstrated. Concrete and reinforcing bars are modeled together as 8-node isoparametric 2D RC element. The FRP plate is modeled as 8-node isoparametric 2D elastic element. The glue is modeled as perfect compatibility by directly connecting the nodes of FRP with those of concrete since there is no failure at the glue layer. The key to the analysis is the correct material models of concrete, steel and FRP. Cracks and steel bars are modeled as smeared over the entire element. Stress-strain properties of cracked concrete consist of tensile stress model normal to crack, compressive stress model parallel to crack and shear stress model tangential to crack. Stressstrain property of reinforcement is assumed to be elastic-hardening to account for the bond between concrete and steel bars. FRP is modeled as elastic-brittle material. From the analysis, it is found that FEM can predict the load-displacement relation, ultimate load and failure mode of the beam correctly. It can also capture the cracking process for both shear-flexural peeling and end peeling modes similar to the experiment.
Fundamental limits on beam stability at the Advanced Photon Source
International Nuclear Information System (INIS)
Decker, G. A.
1998-01-01
Orbit correction is now routinely performed at the few-micron level in the Advanced Photon Source (APS) storage ring. Three diagnostics are presently in use to measure and control both AC and DC orbit motions: broad-band turn-by-turn rf beam position monitors (BPMs), narrow-band switched heterodyne receivers, and photoemission-style x-ray beam position monitors. Each type of diagnostic has its own set of systematic error effects that place limits on the ultimate pointing stability of x-ray beams supplied to users at the APS. Limiting sources of beam motion at present are magnet power supply noise, girder vibration, and thermal timescale vacuum chamber and girder motion. This paper will investigate the present limitations on orbit correction, and will delve into the upgrades necessary to achieve true sub-micron beam stability
Postal dosimetry audit test for small photon beams
International Nuclear Information System (INIS)
Espinosa, María del Mar; Núñez, Luis; Muñiz, José Luis; Lagares, Juan Ignacio; Embid, Miguel; Gómez-Ros, José María
2012-01-01
Background and purpose: Small radiation beams ( 3 ) of TLD-100 inserted at 5 and 10 cm of depth in a cylindrical PMMA phantom designed for this purpose. This experimental system is mailed to the audited centres to be irradiated with beams of 1 and 3 cm of side or diameter. The prescribeddose is 1.5 Gy at 10 cm. The properties of this system were studied experimentally and by Monte Carlo (MC) simulation, before the external test. Results: Deviations between the prescribed and measured absorbed doses are below 5% for 69% (1 × 1 cm 2 beam) and 64% (3 × 3 cm 2 beam) of the audited centres. When deviations are above 5%, their causes have been investigated and led to corrections. Conclusion: The developed postal audit is suitable to verify the absorbed doses in small photon beams with an accuracy of 2.9% (1s).
Beam position monitor data acquisition for the Advanced Photon Source
International Nuclear Information System (INIS)
Lenkszus, F.R.; Kahana, E.; Votaw, A.J.; Decker, G.A.; Chung, Y.; Ciarlette, D.J.; Laird, R.J.
1993-01-01
This paper describes the Beam Position Monitor (BPM) data acquisition scheme for the Advanced Photon Source (APS) storage ring. The storage ring contains 360 beam position monitors distributed around its 1104-meter circumference. The beam position monitor data acquisition system is capable of making turn-by-turn measurements of all BPMs simultaneously. It is VXI-based with each VXI crate containing the electronics for 9 BPMS. The VXI Local Bus is used to provide sustained data transfer rates of up to 13 mega-transfers per second to a scanner module. The system provides single-bunch tracking, bunch-to-bunch measurements, fast digital-averaged positions, beam position history buffering, and synchronized multi-turn measurements. Data is accessible to the control system VME crates via an MXI bus. Dedicated high-speed ports are provided to supply position data to beam orbit feedback systems
International Nuclear Information System (INIS)
Aniel, Thierry.
1982-06-01
The performance of photon beams obtained by two different tagging processes (tagging of ''hard'' annihilation photons with ''soft'' associated photons, tagging of bremstrahlung photons with associated electrons) on the same facility was studied. The two processes are described and experimental results on the characteristics of the resulting beams given. The respective advantages of both methods are compared with one another and with those of a quasi-monochromatic beam obtained by the in-flight annihilation of a positron beam. A development based on the second process is then studied together with its applications to photonuclear physics [fr
Acoustics of finite asymmetric exotic beams: Examples of Airy and fractional Bessel beams
Mitri, F. G.
2017-12-01
The purpose of this investigation is to examine the properties of finite asymmetric exotic scalar (acoustic) beams with unusual properties using the angular spectrum decomposition in plane waves. Such beams possess intrinsic uncommon characteristics that make them attractive from the standpoint of particle manipulation, handling and rotation, and possibly other applications in particle clearing and separation. Assuming a specific apodization function at the acoustic source, the angular spectrum function is calculated and used to synthesize the radiated pressure field (i.e., excluding evanescent waves that decay away from the source) in the forward direction of wave motion (i.e., away from the source). Moreover, a generalized hybrid method combining the angular spectrum approach with the multipole expansion formalism in spherical coordinates is developed, which is applicable to any finite beam of arbitrary wavefront. The improved approach allows adequate computation of the resonance scattering, radiation force, and spin torque components on an object of arbitrary shape, located on or off the axis of the incident beam in space. Considering the illustrative example of a viscous fluid sphere submerged in a non-viscous liquid and illuminated by finite asymmetric beams such as the Airy and the Bessel vortex beam with fractional order, numerical computations for the scattering, radiation force, and torque components are performed with an emphasis on the distance from the source, the arbitrary location of the particle ,and the asymmetric nature of the incident field. Moreover, beamforming calculations are presented with supplementary animations for the pressure field distribution in space, with an emphasis on the intrinsic properties of the selected beams. The numerical predictions illustrate the scattering, radiation force, and spin torque properties depending on the beam parameters and the distance separating the sphere from the source. This study provides a generalized
Simultaneous optimization of photons and electrons for mixed beam radiotherapy.
Mueller, S; Fix, M K; Joosten, A; Henzen, D; Frei, D; Volken, W; Kueng, R; Aebersold, D M; Stampanoni, M F M; Manser, P
2017-06-26
The aim of this work is to develop and investigate an inverse treatment planning process (TPP) for mixed beam radiotherapy (MBRT) capable of performing simultaneous optimization of photon and electron apertures. A simulated annealing based direct aperture optimization (DAO) is implemented to perform simultaneous optimization of photon and electron apertures, both shaped with the photon multileaf collimator (pMLC). Validated beam models are used as input for Monte Carlo dose calculations. Consideration of photon pMLC transmission during DAO and a weight re-optimization of the apertures after deliverable dose calculation are utilized to efficiently reduce the differences between optimized and deliverable dose distributions. The TPP for MBRT is evaluated for an academic situation with a superficial and an enlarged PTV in the depth, a left chest wall case including the internal mammary chain and a squamous cell carcinoma case. Deliverable dose distributions of MBRT plans are compared to those of modulated electron radiotherapy (MERT), photon IMRT and if available to those of clinical VMAT plans. The generated MBRT plans dosimetrically outperform the MERT, photon IMRT and VMAT plans for all investigated situations. For the clinical cases of the left chest wall and the squamous cell carcinoma, the MBRT plans cover the PTV similarly or more homogeneously than the VMAT plans, while OARs are spared considerably better with average reductions of the mean dose to parallel OARs and D 2% to serial OARs by 54% and 26%, respectively. Moreover, the low dose bath expressed as V 10% to normal tissue is substantially reduced by up to 45% compared to the VMAT plans. A TPP for MBRT including simultaneous optimization is successfully implemented and the dosimetric superiority of MBRT plans over MERT, photon IMRT and VMAT plans is demonstrated for academic and clinical situations including superficial targets with and without deep-seated part.
Integral window/photon beam position monitor and beam flux detectors for x-ray beams
Shu, Deming; Kuzay, Tuncer M.
1995-01-01
A monitor/detector assembly in a synchrotron for either monitoring the position of a photon beam or detecting beam flux may additionally function as a vacuum barrier between the front end and downstream segment of the beamline in the synchrotron. A base flange of the monitor/detector assembly is formed of oxygen free copper with a central opening covered by a window foil that is fused thereon. The window foil is made of man-made materials, such as chemical vapor deposition diamond or cubic boron nitrate and in certain configurations includes a central opening through which the beams are transmitted. Sensors of low atomic number materials, such as aluminum or beryllium, are laid on the window foil. The configuration of the sensors on the window foil may be varied depending on the function to be performed. A contact plate of insulating material, such as aluminum oxide, is secured to the base flange and is thereby clamped against the sensor on the window foil. The sensor is coupled to external electronic signal processing devices via a gold or silver lead printed onto the contact plate and a copper post screw or alternatively via a copper screw and a copper spring that can be inserted through the contact plate and coupled to the sensors. In an alternate embodiment of the monitor/detector assembly, the sensors are sandwiched between the window foil of chemical vapor deposition diamond or cubic boron nitrate and a front foil made of similar material.
Finite geometry effect on the interaction of a hot beam with a plasma
International Nuclear Information System (INIS)
Shoucri, M.M.; Gagne, R.R.J.
1977-01-01
The effect of finite geometry on the interaction of a hot low-density beam with a uniform plasma filling a circular waveguide is studied. An expression is derived for the growth rate of the instabilities developing at the harmonic of the beam gyrofrequency, taking the finite beam gyroradius into account. The calculations are done in the quasistatic approximation. (author)
Dosimetric evaluation of fattening filter free photon beams
International Nuclear Information System (INIS)
Lechner, W.
2014-01-01
Very recently, conventional treatment machines able to deliver flattening filter free photon have been introduced into clinical practice. An Elekta Pricise linear accelerator LINAC which was able to deliver 6MV and 10 MV flattened (FF) and flattening filter free (FFF) photon beams was used throughout this work. The investigated modalities were 9-Field intensity modulated radiotherapy (Intensity modulated radiotherapy (IMRT)) and 360° - single arc volumetric modulated arc therapy (Volumetric modulated arc therapy (VMAT)) with flattened and unflattened photon beams. For treatment plan evaluation the concept of Pareto optimal fronts was employed. Additionally, the efficiency of these modalities was assessed. The evaluation of the treatment plan quality showed no significant difference between FF- and FFF-beams. With respect to treatment plan efficiency, a significant decrease of delivery time of IMRT treatment plans without flattening filter compared to those with filter was found. On average, the delivery time decreased by 18% and 4% for prostate and head-and-neck cases, respectively. In contrast to that, the delivery time of VMAT treatment plans without flatting filter was significantly increased by 22% and 16% for prostate and head- and-neck cases, respectively. The applicability of 14 different detectors to small field dosimetry in FF- as well as in FFF-beams was investigated by measuring output factors of 6MV and 10 MV FF- and FFF- beams. An additional MLC was attached to the treatment machine in order to generate field sizes between 0.6x0.6 cm 2 and 10x10 cm 2 . Alanine pellets were used as reference detectors for the calculation of correction factors for small field dosimetry. Compared to alanine, the solid sate detectors and the liquid filled ionization chamber generally overestimated the output factors of small fields, whereas the air filled ionzation chambers underestimated the output factors. For the shielded diodes the correction factors ranged between 8
Photon beam properties at the European XFEL (December 2010 revision)
Energy Technology Data Exchange (ETDEWEB)
Schneidmiller, E.A.; Yurkov, M.V.
2011-09-15
A new set of baseline parameters of the electron beam and undulator for the European XFEL project has been defined recently. Changes refer to the electron beam emittance, charge, operation at different electron energies, and change of undulator period. According to the present concept, it is planned to vary charge from 20 pC to 1 nC allowing control of the FWHM radiation pulse duration. Operation at different electron energies of 17.5 GeV, 14 GeV, and 10.5 GeV will allow extension of the wavelength range to longer wavelengths. Electron bunches with different charges possess different properties. These features have an impact on photon beam properties which should be taken into account at the design stage optical beamlines and instruments and planning user experiments. In this note we present an overview of the radiation properties generated by SASE FEL radiators driven by electron beam with revised baseline parameters. (orig.)
The physics of small megavoltage photon beam dosimetry.
Andreo, Pedro
2018-02-01
The increased interest during recent years in the use of small megavoltage photon beams in advanced radiotherapy techniques has led to the development of dosimetry recommendations by different national and international organizations. Their requirement of data suitable for the different clinical options available, regarding treatment units and dosimetry equipment, has generated a considerable amount of research by the scientific community during the last decade. The multiple publications in the field have led not only to the availability of new invaluable data, but have also contributed substantially to an improved understanding of the physics of their dosimetry. This work provides an overview of the most important aspects that govern the physics of small megavoltage photon beam dosimetry. Copyright © 2017 Elsevier B.V. All rights reserved.
New limits on hidden photons from past electron beam dumps
International Nuclear Information System (INIS)
Andreas, Sarah; Niebuhr, Carsten; Ringwald, Andreas
2012-09-01
Hidden sectors with light extra U(1) gauge bosons, so called hidden photons, have recently attracted some attention because they are a common feature of physics beyond the Standard Model like string theory and SUSY and additionally are phenomenologically of great interest regarding recent astrophysical observations. The hidden photon is already constrained by various laboratory experiments and presently searched for in running as well as upcoming experiments. We summarize the current status of limits on hidden photons from past electron beam dump experiments including two new limits from such experiments at KEK and Orsay that have so far not been considered. All our limits take into account the experimental acceptances obtained from Monte Carlo simulations.
New limits on hidden photons from past electron beam dumps
Energy Technology Data Exchange (ETDEWEB)
Andreas, Sarah; Niebuhr, Carsten; Ringwald, Andreas
2012-09-15
Hidden sectors with light extra U(1) gauge bosons, so called hidden photons, have recently attracted some attention because they are a common feature of physics beyond the Standard Model like string theory and SUSY and additionally are phenomenologically of great interest regarding recent astrophysical observations. The hidden photon is already constrained by various laboratory experiments and presently searched for in running as well as upcoming experiments. We summarize the current status of limits on hidden photons from past electron beam dump experiments including two new limits from such experiments at KEK and Orsay that have so far not been considered. All our limits take into account the experimental acceptances obtained from Monte Carlo simulations.
Characteristics of mobile MOSFET dosimetry system for megavoltage photon beams.
Kumar, A Sathish; Sharma, S D; Ravindran, B Paul
2014-07-01
The characteristics of a mobile metal oxide semiconductor field effect transistor (mobile MOSFET) detector for standard bias were investigated for megavoltage photon beams. This study was performed with a brass alloy build-up cap for three energies namely Co-60, 6 and 15 MV photon beams. The MOSFETs were calibrated and the performance characteristics were analyzed with respect to dose rate dependence, energy dependence, field size dependence, linearity, build-up factor, and angular dependence for all the three energies. A linear dose-response curve was noted for Co-60, 6 MV, and 15 MV photons. The calibration factors were found to be 1.03, 1, and 0.79 cGy/mV for Co-60, 6 MV, and 15 MV photon energies, respectively. The calibration graph has been obtained to the dose up to 600 cGy, and the dose-response curve was found to be linear. The MOSFETs were found to be energy independent both for measurements performed at depth as well as on the surface with build-up. However, field size dependence was also analyzed for variable field sizes and found to be field size independent. Angular dependence was analyzed by keeping the MOSFET dosimeter in parallel and perpendicular orientation to the angle of incidence of the radiation with and without build-up on the surface of the phantom. The maximum variation for the three energies was found to be within ± 2% for the gantry angles 90° and 270°, the deviations without the build-up for the same gantry angles were found to be 6%, 25%, and 60%, respectively. The MOSFET response was found to be independent of dose rate for all three energies. The dosimetric characteristics of the MOSFET detector make it a suitable in vivo dosimeter for megavoltage photon beams.
Resonant production of dark photons in positron beam dump experiments
Nardi, Enrico; Carvajal, Cristian D. R.; Ghoshal, Anish; Meloni, Davide; Raggi, Mauro
2018-05-01
Positrons beam dump experiments have unique features to search for very narrow resonances coupled superweakly to e+e- pairs. Due to the continued loss of energy from soft photon bremsstrahlung, in the first few radiation lengths of the dump a positron beam can continuously scan for resonant production of new resonances via e+ annihilation off an atomic e- in the target. In the case of a dark photon A' kinetically mixed with the photon, this production mode is of first order in the electromagnetic coupling α , and thus parametrically enhanced with respect to the O (α2)e+e-→γ A' production mode and to the O (α3)A' bremsstrahlung in e- -nucleon scattering so far considered. If the lifetime is sufficiently long to allow the A' to exit the dump, A'→e+e- decays could be easily detected and distinguished from backgrounds. We explore the foreseeable sensitivity of the Frascati PADME experiment in searching with this technique for the 17 MeV dark photon invoked to explain the Be 8 anomaly in nuclear transitions.
A high energy photon beam derived from neutral strange particle decay
International Nuclear Information System (INIS)
Reibel, K.; Ruchti, R.
1982-01-01
Conventional methods for generating photon beams include: tagged beams in which the photons are derived from electron bremsstrahlung in a radiator target; and broad band beams in which the photons are derived from π/sup 0/ decay - the hadronic component (n, K/sub s//sup 0/) accompanying such a beam is usually suppressed by passage of the beam through a low Z (D/sub 2/) filter. Although one can generate high energy photons by these techniques, the major drawback to these beams is that the photon energy spectrum obtained is peaked at very low E/sub γ/. (Recall that the bremsstrahlung spectrum falls as 1/k). With very high energy proton beams (20 TeV/c), one can image other alternatives for photon beam design. The authors consider one such option here
Entanglement of two-qubit photon beam by magnetic field
Energy Technology Data Exchange (ETDEWEB)
Levin, A.D.; Castro, R.A. [University of Sao Paulo, Institute of Physics, CP 66318, Sao Paulo (Brazil); Gitman, D.M. [University of Sao Paulo, Institute of Physics, CP 66318, Sao Paulo (Brazil); P.N. Lebedev Physical Institute, Moscow (Russian Federation); Tomsk State University, Tomsk (Russian Federation)
2014-09-15
We study the possibility of affecting the entanglement in a two-qubit system consisting of two photons with different fixed frequencies but with two arbitrary linear polarizations, moving in the same direction, with the help of an applied external magnetic field. The interaction between the magnetic field and the photons in our model is achieved through intermediate electrons that interact both with the photons and the magnetic field. The possibility of an exact theoretical analysis of this scheme is based on well-known exact solutions that describe the interaction of an electron subjected to an external magnetic field (or a medium of electrons not interacting with each other) with a quantized field of two photons. We adapt these exact solutions to the case under consideration. Using explicit wave functions for the resulting electromagnetic field, we calculate the entanglement measures (the information and the Schmidt ones) of the photon beam as functions of the applied magnetic field and the parameters of the electron medium. (orig.)
International Nuclear Information System (INIS)
Centini, M.; Sciscione, L.; Sibilia, C.; Bertolotti, M.; Perina, J. Jr.; Scalora, M.; Bloemer, M.J.
2005-01-01
A description of spontaneous parametric down-conversion in finite-length one-dimensional nonlinear photonic crystals is developed using semiclassical and quantum approaches. It is shown that if a suitable averaging is added to the semiclassical model, its results are in very good agreement with the quantum approach. We propose two structures made with GaN/AlN that generate both degenerate and nondegenerate entangled photon pairs. Both structures are designed so as to achieve a high efficiency of the nonlinear process
A numerical study of the characteristics of the LEALE photon beam
International Nuclear Information System (INIS)
Lucherini, V.; De Sanctis, E.; Di Giacomo, P.
1978-01-01
At the LEALE laboratory a monochromatic photon beam with energy in the range 80/300 MeV is available. Photons are produced by positron annihilation on a liquid hydrogen target. The characteristics of the beam are calculated for various conditions (positron energy, photon collimator, target thickness), taking into account the effects contributing to the beam spreading (energy loss and multiple scattering of protons in the annihilation target, energy distribution and angular divergence of the positron beam). (author)
Application of spherical diodes for megavoltage photon beams dosimetry
International Nuclear Information System (INIS)
Barbés, Benigno; Azcona, Juan D.; Burguete, Javier; Martí-Climent, Josep M.
2014-01-01
Purpose: External beam radiation therapy (EBRT) usually uses heterogeneous dose distributions in a given volume. Designing detectors for quality control of these treatments is still a developing subject. The size of the detectors should be small to enhance spatial resolution and ensure low perturbation of the beam. A high uniformity in angular response is also a very important feature in a detector, because it has to measure radiation coming from all the directions of the space. It is also convenient that detectors are inexpensive and robust, especially to performin vivo measurements. The purpose of this work is to introduce a new detector for measuring megavoltage photon beams and to assess its performance to measure relative dose in EBRT. Methods: The detector studied in this work was designed as a spherical photodiode (1.8 mm in diameter). The change in response of the spherical diodes is measured regarding the angle of incidence, cumulated irradiation, and instantaneous dose rate (or dose per pulse). Additionally, total scatter factors for large and small fields (between 1 × 1 cm 2 and 20 × 20 cm 2 ) are evaluated and compared with the results obtained from some commercially available ionization chambers and planar diodes. Additionally, the over-response to low energy scattered photons in large fields is investigated using a shielding layer. Results: The spherical diode studied in this work produces a high signal (150 nC/Gy for photons of nominal energy of 15 MV and 160 for 6 MV, after 12 kGy) and its angular dependence is lower than that of planar diodes: less than 5% between maximum and minimum in all directions, and 2% around one of the axis. It also has a moderated variation with accumulated dose (about 1.5%/kGy for 15 MV photons and 0.7%/kGy for 6 MV, after 12 kGy) and a low variation with dose per pulse (±0.4%), and its behavior is similar to commercial diodes in total scatter factor measurements. Conclusions: The measurements of relative dose using
Application of spherical diodes for megavoltage photon beams dosimetry.
Barbés, Benigno; Azcona, Juan D; Burguete, Javier; Martí-Climent, Josep M
2014-01-01
External beam radiation therapy (EBRT) usually uses heterogeneous dose distributions in a given volume. Designing detectors for quality control of these treatments is still a developing subject. The size of the detectors should be small to enhance spatial resolution and ensure low perturbation of the beam. A high uniformity in angular response is also a very important feature in a detector, because it has to measure radiation coming from all the directions of the space. It is also convenient that detectors are inexpensive and robust, especially to perform in vivo measurements. The purpose of this work is to introduce a new detector for measuring megavoltage photon beams and to assess its performance to measure relative dose in EBRT. The detector studied in this work was designed as a spherical photodiode (1.8 mm in diameter). The change in response of the spherical diodes is measured regarding the angle of incidence, cumulated irradiation, and instantaneous dose rate (or dose per pulse). Additionally, total scatter factors for large and small fields (between 1 × 1 cm(2) and 20 × 20 cm(2)) are evaluated and compared with the results obtained from some commercially available ionization chambers and planar diodes. Additionally, the over-response to low energy scattered photons in large fields is investigated using a shielding layer. The spherical diode studied in this work produces a high signal (150 nC/Gy for photons of nominal energy of 15 MV and 160 for 6 MV, after 12 kGy) and its angular dependence is lower than that of planar diodes: less than 5% between maximum and minimum in all directions, and 2% around one of the axis. It also has a moderated variation with accumulated dose (about 1.5%/kGy for 15 MV photons and 0.7%/kGy for 6 MV, after 12 kGy) and a low variation with dose per pulse (± 0.4%), and its behavior is similar to commercial diodes in total scatter factor measurements. The measurements of relative dose using the spherical diode described in this
Consistency check of photon beam physical data after recommissioning process
International Nuclear Information System (INIS)
Kadman, B; Chawapun, N; Ua-apisitwong, S; Asakit, T; Chumpu, N; Rueansri, J
2016-01-01
In radiotherapy, medical linear accelerator (Linac) is the key system used for radiation treatments delivery. Although, recommissioning was recommended after major modification of the machine by AAPM TG53, but it might not be practical in radiotherapy center with heavy workloads. The main purpose of this study was to compare photon beam physical data between initial commissioning and recommissioning of 6 MV Elekta Precise linac. The parameters for comparing were the percentage depth dose (PDD) and beam profiles. The clinical commissioning test cases followed IAEA-TECDOC-1583 were planned on REF 91230 IMRT Dose Verification Phantom by Philips’ Pinnacle treatment planning system. The Delta 4PT was used for dose distribution verification with 90% passing criteria of the gamma index (3%/3mm). Our results revealed that the PDDs and beam profiles agreed within a tolerance limit recommended by TRS430. Most of the point doses and dose distribution verification passed the acceptance criteria. This study showed the consistency of photon beam physical data after recommissioning process. There was a good agreement between initial commissioning and recommissioning within a tolerance limit, demonstrated that the full recommissioning process might not be required. However, in the complex treatment planning geometry, the initial data should be applied with great caution. (paper)
Intermediate Megavoltage Photon Beams for Improved Lung Cancer Treatments.
Directory of Open Access Journals (Sweden)
Ying Zhang
Full Text Available The goal of this study is to evaluate the effects of intermediate megavoltage (3-MV photon beams on SBRT lung cancer treatments. To start with, a 3-MV virtual beam was commissioned on a commercial treatment planning system based on Monte Carlo simulations. Three optimized plans (6-MV, 3-MV and dual energy of 3- and 6-MV were generated for 31 lung cancer patients with identical beam configuration and optimization constraints for each patient. Dosimetric metrics were evaluated and compared among the three plans. Overall, planned dose conformity was comparable among three plans for all 31 patients. For 21 thin patients with average short effective path length (< 10 cm, the 3-MV plans showed better target coverage and homogeneity with dose spillage index R50% = 4.68±0.83 and homogeneity index = 1.26±0.06, as compared to 4.95±1.01 and 1.31±0.08 in the 6-MV plans (p < 0.001. Correspondingly, the average/maximum reductions of lung volumes receiving 20 Gy (V20Gy, 5 Gy (V5Gy, and mean lung dose (MLD were 7%/20%, 9%/30% and 5%/10%, respectively in the 3-MV plans (p < 0.05. The doses to 5% volumes of the cord, esophagus, trachea and heart were reduced by 9.0%, 10.6%, 11.4% and 7.4%, respectively (p < 0.05. For 10 thick patients, dual energy plans can bring dosimetric benefits with comparable target coverage, integral dose and reduced dose to the critical structures, as compared to the 6-MV plans. In conclusion, our study indicated that 3-MV photon beams have potential dosimetric benefits in treating lung tumors in terms of improved tumor coverage and reduced doses to the adjacent critical structures, in comparison to 6-MV photon beams. Intermediate megavoltage photon beams (< 6-MV may be considered and added into current treatment approaches to reduce the adjacent normal tissue doses while maintaining sufficient tumor dose coverage in lung cancer radiotherapy.
Magnetic monopole and finite photon mass: are they compatible?
International Nuclear Information System (INIS)
Ignatiev, A.Y.; Joshi, G.C.
1994-01-01
The role played by the gauge invariance for the existence of Dirac monopole is analysed. It is shown that the derivation of the Dirac quantization condition based on the angular momentum algebra cannot be generalized to the case of massive electrodynamics. The work revealed a new and important relation between the two fundamental facts: the masslessness (massiveness?) of the photons and the non-existence (existence?) of the magnetic monopole. Possible implications of this result are briefly discussed. 12 refs
Fission product yield measurements using monoenergetic photon beams
Krishichayan; Bhike, M.; Tonchev, A. P.; Tornow, W.
2017-09-01
Measurements of fission products yields (FPYs) are an important source of information on the fission process. During the past couple of years, a TUNL-LANL-LLNL collaboration has provided data on the FPYs from quasi monoenergetic neutron-induced fission on 235U, 238U, and 239Pu and has revealed an unexpected energy dependence of both asymmetric fission fragments at energies below 4 MeV. This peculiar FPY energy dependence was more pronounced in neutron-induced fission of 239Pu. In an effort to understand and compare the effect of the incoming probe on the FPY distribution, we have carried out monoenergetic photon-induced fission experiments on the same 235U, 238U, and 239Pu targets. Monoenergetic photon beams of Eγ = 13.0 MeV were provided by the HIγS facility, the world's most intense γ-ray source. In order to determine the total number of fission events, a dual-fission chamber was used during the irradiation. These irradiated samples were counted at the TUNL's low-background γ-ray counting facility using high efficient HPGe detectors over a period of 10 weeks. Here we report on our first ever photofission product yield measurements obtained with monoenegetic photon beams. These results are compared with neutron-induced FPY data.
Fission product yield measurements using monoenergetic photon beams
Directory of Open Access Journals (Sweden)
Krishichayan
2017-01-01
Full Text Available Measurements of fission products yields (FPYs are an important source of information on the fission process. During the past couple of years, a TUNL-LANL-LLNL collaboration has provided data on the FPYs from quasi monoenergetic neutron-induced fission on 235U, 238U, and 239Pu and has revealed an unexpected energy dependence of both asymmetric fission fragments at energies below 4 MeV. This peculiar FPY energy dependence was more pronounced in neutron-induced fission of 239Pu. In an effort to understand and compare the effect of the incoming probe on the FPY distribution, we have carried out monoenergetic photon-induced fission experiments on the same 235U, 238U, and 239Pu targets. Monoenergetic photon beams of Eγ = 13.0 MeV were provided by the HIγS facility, the world's most intense γ-ray source. In order to determine the total number of fission events, a dual-fission chamber was used during the irradiation. These irradiated samples were counted at the TUNL's low-background γ-ray counting facility using high efficient HPGe detectors over a period of 10 weeks. Here we report on our first ever photofission product yield measurements obtained with monoenegetic photon beams. These results are compared with neutron-induced FPY data.
Finite element analysis of composite concrete-timber beams
Directory of Open Access Journals (Sweden)
N. C. S. FORTI
Full Text Available AbstractIn the search for sustainable construction, timber construction is gaining in popularity around the world. Sustainably harvested wood stores carbon dioxide, while reforestation absorbs yet more CO2. One technique involves the combination of a concrete slab and a timber beam, where the two materials are assembled by the use of flexible connectors. Composite structures provide reduced costs, environmental benefits, a better acoustic performance, when compared to timber structures, and maintain structural safety. Composite structures combine materials with different mechanical properties. Their mechanical performance depends on the efficiency of the connection, which is designed to transmit shear longitudinal forces between the two materials and to prevent vertical detachment. This study contributes with the implementation of a finite element formulation for stress and displacement determination of composite concrete-timber beams. The deduced stiffness matrix and load vector are presented along to numerical examples. Numerical examples are compared to the analytical equations available in Eurocode 5 and to experimental data found in the literature.
Laser-driven particle and photon beams and some applications
International Nuclear Information System (INIS)
Ledingham, K W D; Galster, W
2010-01-01
Outstanding progress has been made in high-power laser technology in the last 10 years with laser powers reaching petawatt (PW) values. At present, there are 15 PW lasers built or being built around the world and plans are afoot for new, even higher power, lasers reaching values of exawatt (EW) or even zetawatt (ZW) powers. Petawatt lasers generate electric fields of 10 12 V m -1 with a large fraction of the total pulse energy being converted to relativistic electrons with energies reaching in excess of 1 GeV. In turn these electrons result in the generation of beams of protons, heavy ions, neutrons and high-energy photons. These laser-driven particle beams have encouraged many to think of carrying out experiments normally associated with conventional nuclear accelerators and reactors. To this end a number of introductory articles have been written under a trial name 'Laser Nuclear Physics' (Ledingham and Norreys 1999 Contemp. Phys. 40 367, Ledingham et al 2002 Europhys. News. 33 120, Ledingham et al 2003 Science 300 1107, Takabe et al 2001 J. Plasma Fusion Res. 77 1094). However, even greater strides have been made in the last 3 or 4 years in laser technology and it is timely to reassess the potential of laser-driven particle and photon beams. It must be acknowledged right from the outset that to date laser-driven particle beams have yet to compete favourably with conventional nuclear accelerator-generated beams in any way and so this is not a paper comparing laser and conventional accelerators. However, occasionally throughout the paper as a reality check, it will be mentioned what conventional nuclear accelerators can do.
Laser-driven particle and photon beams and some applications
Energy Technology Data Exchange (ETDEWEB)
Ledingham, K W D; Galster, W, E-mail: K.Ledingham@phys.strath.ac.u [SUPA, Department of Physics, University of Strathclyde, Glasgow G4 0NG (United Kingdom)
2010-04-15
Outstanding progress has been made in high-power laser technology in the last 10 years with laser powers reaching petawatt (PW) values. At present, there are 15 PW lasers built or being built around the world and plans are afoot for new, even higher power, lasers reaching values of exawatt (EW) or even zetawatt (ZW) powers. Petawatt lasers generate electric fields of 10{sup 12} V m{sup -1} with a large fraction of the total pulse energy being converted to relativistic electrons with energies reaching in excess of 1 GeV. In turn these electrons result in the generation of beams of protons, heavy ions, neutrons and high-energy photons. These laser-driven particle beams have encouraged many to think of carrying out experiments normally associated with conventional nuclear accelerators and reactors. To this end a number of introductory articles have been written under a trial name 'Laser Nuclear Physics' (Ledingham and Norreys 1999 Contemp. Phys. 40 367, Ledingham et al 2002 Europhys. News. 33 120, Ledingham et al 2003 Science 300 1107, Takabe et al 2001 J. Plasma Fusion Res. 77 1094). However, even greater strides have been made in the last 3 or 4 years in laser technology and it is timely to reassess the potential of laser-driven particle and photon beams. It must be acknowledged right from the outset that to date laser-driven particle beams have yet to compete favourably with conventional nuclear accelerator-generated beams in any way and so this is not a paper comparing laser and conventional accelerators. However, occasionally throughout the paper as a reality check, it will be mentioned what conventional nuclear accelerators can do.
Performance of EPI diodes as dosimeters for photon beam radiotherapy
Energy Technology Data Exchange (ETDEWEB)
Santos, Thais C. dos; Bizetto, Cesar A., E-mail: ccbueno@ipen.br [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil); Neves-Junior, Wellington F.P.; Haddad, Cecilia M.K. [Hospital Sirio Libanes (HSL), Sao Paulo, SP (Brazil); Goncalves, Josemary A.C.; Bueno, Carmen C. [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil); Pontificia Universidade Catolica de Sao Paulo (PUC-SP), SP (Brazil)
2011-07-01
In this work we present the preliminary results about the performance of an epitaxial (EPI) diode as on-line dosimeter for photon beam radiotherapy. The diode used was processed at University of Hamburg on n-type 75 {mu}m thick epitaxial silicon layer grown on a highly doped n-type 300 {mu}m thick Czochralski (Cz) silicon substrate. The measurements were performed with a diode which not received any type of pre-dose. In order to use this device as a dosimeter, it was enclosed in a black polymethylmethacrylate (PMMA) probe. The diode was connected to an electrometer Keithley 6517B in the photovoltaic mode. During all measurements, the diode was held between PMMA plates, placed at 10.0 cm depth and centered in a radiation field of 10 x 10 cm{sup 2}, with the source-to-surface distance (SSD) kept at 100 cm. The short-term repeatability was measured with photon beams of 6 and 18 MV energy by registering five consecutive current signals for the same radiation dose. The current signals induced showed good instantaneous repeatability of the diode, characterized by a smallest coefficient of variation (CV) of 0.21%. Furthermore, the dose-response curves of the diode were quite linear with the highest charge sensitivity achieved of 5.0 {mu}C/Gy. It worth noting that still remains to be investigated the pre-dose influence on epitaxial silicon diode response in radiotherapy photon beam dosimetry, the long term stability and the radiation hardness of these diodes for absorbed doses higher than that investigated in this work. All these studies are under way. (author)
Cardiac single-photon emission-computed tomography using combined cone-beam/fan-beam collimation
International Nuclear Information System (INIS)
Gullberg, Grant T.; Zeng, Gengsheng L.
2004-01-01
The objective of this work is to increase system sensitivity in cardiac single-photon emission-computed tomography (SPECT) studies without increasing patient imaging time. For imaging the heart, convergent collimation offers the potential of increased sensitivity over that of parallel-hole collimation. However, if a cone-beam collimated gamma camera is rotated in a planar orbit, the projection data obtained are not complete. Two cone-beam collimators and one fan-beam collimator are used with a three-detector SPECT system. The combined cone-beam/fan-beam collimation provides a complete set of data for image reconstruction. The imaging geometry is evaluated using data acquired from phantom and patient studies. For the Jaszazck cardiac torso phantom experiment, the combined cone-beam/fan-beam collimation provided 1.7 times greater sensitivity than standard parallel-hole collimation (low-energy high-resolution collimators). Also, phantom and patient comparison studies showed improved image quality. The combined cone-beam/fan-beam imaging geometry with appropriate weighting of the two data sets provides improved system sensitivity while measuring sufficient data for artifact free cardiac images
Packaging consideration of two-dimensional polymer-based photonic crystals for laser beam steering
Dou, Xinyuan; Chen, Xiaonan; Chen, Maggie Yihong; Wang, Alan Xiaolong; Jiang, Wei; Chen, Ray T.
2009-02-01
In this paper, we report the theoretical study of polymer-based photonic crystals for laser beam steering which is based on the superprism effect as well as the experiment fabrication of the two dimensional photonic crystals for the laser beam steering. Superprism effect, the principle for beam steering, was separately studied in details through EFC (Equifrequency Contour) analysis. Polymer based photonic crystals were fabricated through double exposure holographic interference method using SU8-2007. The experiment results were also reported.
Using of germanium detectors in nuclear experiments with photon beams
International Nuclear Information System (INIS)
Kapitonov, I.M.; Tutin, I.A.
1995-01-01
Full text: The study of atomic nuclei with real photons is very important source of the information about nuclear structure. In such experiments the basic electromagnetic interaction between the photon and the target nuclei is well known. Experiments with photon beams become especially valuable when outcoming particles are also photons. In these cases completely model-independent information on nuclear structure can be extracted. The use of semiconductor Ge-spectrometers with excellent resolution and large sensitive volumes for recording outcoming photons gives us such an additional important advantage as possibility to observe individual closely spaced levels of the final nuclei. In the report an experience of using Ge-detectors in two types of nuclear experiments is described. Both of them - nuclear resonance fluorescence (NRF) and nuclear photodisintegration - are carried out in beams of bremsstrahlung gamma radiation. The central element of the setup recording gamma quanta in these experiments is germanium detector. NRF is unique method for studying low-lying excited nuclear states. The spins of the states can be determined easily from the measured angular distributions of scattered photons. Model independent parity assignments in NRF can be achieved by measuring polarization observables. There are two experimental possibilities: the use of linearly polarized photons (off-axis bremsstrahlung) in the entrance channel and the measurement of the linear polarization of the scattered photons using Compton polarimeters. For both methods several germanium detectors (3-5) must be used simultaneously. Nowadays Compton polarimeter can also be done from single large Ge-crystal by segmenting the outer electrode. Advantages and drawbacks of the methods and background conditions are discussed and requirements to Ge-crystals are formulated. The importance of using a new generation of electron accelerators with continuous wave (cw) beams for NRF-measurements is stressed. The
Reduction of metal artifacts: beam hardening and photon starvation effects
Yadava, Girijesh K.; Pal, Debashish; Hsieh, Jiang
2014-03-01
The presence of metal-artifacts in CT imaging can obscure relevant anatomy and interfere with disease diagnosis. The cause and occurrence of metal-artifacts are primarily due to beam hardening, scatter, partial volume and photon starvation; however, the contribution to the artifacts from each of them depends on the type of hardware. A comparison of CT images obtained with different metallic hardware in various applications, along with acquisition and reconstruction parameters, helps understand methods for reducing or overcoming such artifacts. In this work, a metal beam hardening correction (BHC) and a projection-completion based metal artifact reduction (MAR) algorithms were developed, and applied on phantom and clinical CT scans with various metallic implants. Stainless-steel and Titanium were used to model and correct for metal beam hardening effect. In the MAR algorithm, the corrupted projection samples are replaced by the combination of original projections and in-painted data obtained by forward projecting a prior image. The data included spine fixation screws, hip-implants, dental-filling, and body extremity fixations, covering range of clinically used metal implants. Comparison of BHC and MAR on different metallic implants was used to characterize dominant source of the artifacts, and conceivable methods to overcome those. Results of the study indicate that beam hardening could be a dominant source of artifact in many spine and extremity fixations, whereas dental and hip implants could be dominant source of photon starvation. The BHC algorithm could significantly improve image quality in CT scans with metallic screws, whereas MAR algorithm could alleviate artifacts in hip-implants and dentalfillings.
SU-E-T-577: Commissioning of a Deterministic Algorithm for External Photon Beams
International Nuclear Information System (INIS)
Zhu, T; Finlay, J; Mesina, C; Liu, H
2014-01-01
Purpose: We report commissioning results for a deterministic algorithm for external photon beam treatment planning. A deterministic algorithm solves the radiation transport equations directly using a finite difference method, thus improve the accuracy of dose calculation, particularly under heterogeneous conditions with results similar to that of Monte Carlo (MC) simulation. Methods: Commissioning data for photon energies 6 – 15 MV includes the percentage depth dose (PDD) measured at SSD = 90 cm and output ratio in water (Spc), both normalized to 10 cm depth, for field sizes between 2 and 40 cm and depths between 0 and 40 cm. Off-axis ratio (OAR) for the same set of field sizes was used at 5 depths (dmax, 5, 10, 20, 30 cm). The final model was compared with the commissioning data as well as additional benchmark data. The benchmark data includes dose per MU determined for 17 points for SSD between 80 and 110 cm, depth between 5 and 20 cm, and lateral offset of up to 16.5 cm. Relative comparisons were made in a heterogeneous phantom made of cork and solid water. Results: Compared to the commissioning beam data, the agreement are generally better than 2% with large errors (up to 13%) observed in the buildup regions of the FDD and penumbra regions of the OAR profiles. The overall mean standard deviation is 0.04% when all data are taken into account. Compared to the benchmark data, the agreements are generally better than 2%. Relative comparison in heterogeneous phantom is in general better than 4%. Conclusion: A commercial deterministic algorithm was commissioned for megavoltage photon beams. In a homogeneous medium, the agreement between the algorithm and measurement at the benchmark points is generally better than 2%. The dose accuracy for a deterministic algorithm is better than a convolution algorithm in heterogeneous medium
Output calibration in solid water for high energy photon beams
International Nuclear Information System (INIS)
Reft, C.S.
1989-01-01
The AAPM Protocol recommends the use of water, polystyrene or acrylic media for measuring the output of high energy photon beams. It provides the appropriate restricted mass stopping powers and mass energy absorption coefficients for converting the dose to these media to dose to water. A water-equivalent solid has been developed for dosimetric applications. [C. Constantinou, F. Attix, and B. Paliwal, Med. Phys. 9, 436 (1982)]. Calculated values for the restricted mass stopping powers and mass energy absorption coefficients have been published for this material. [A. Ho and B. Paliwal, Med. Phys. 13, 403 (1986)]. The accuracy of these calculations was investigated by making output measurements, following the Protocol, with a Farmer type chamber in four materials for Co-60, 4, 6, 10, 18, and 24 MV photon beams. The results show that the scaled dose to water for the different media agree to better than 1%, and the analysis supports the methodology of the Protocol for obtaining the dose to water from the different media
Simulation of the electron acoustic instability for a finite-size electron beam system
International Nuclear Information System (INIS)
Lin, C.S.; Winske, D.
1987-01-01
Satellite observations at midlatitudes (≅20,000 km) near the earth's dayside polar cusp boundary layer indicate that the upward electron beams have a narrow latitudinal width up to 0.1 0 . In the cusp boundary layer where the electron population consists of a finite-size electron beam in a background of uniform cold and hot electrons, the electron acoustic mode is unstable inside the electron beam but damped outside the electron beam. Simulations of the electron acoustic instability for a finite-size beam system are carried out with a particle-in-cell code to investigate the heating phenomena associated with the instability and the width of the heating region. The simulations show that the finite-size electron beam radiates electrostatic electron acoustic waves. The decay length of the electron acoustic waves outside the beam in the simulation agrees with the spatial decay length derived from the linear dispersion equation
Integration of a photonic crystal polarization beam splitter and waveguide bend.
Zheng, Wanhua; Xing, Mingxin; Ren, Gang; Johnson, Steven G; Zhou, Wenjun; Chen, Wei; Chen, Lianghui
2009-05-11
In this work, we present the design of an integrated photonic-crystal polarization beam splitter (PC-PBS) and a low-loss photonic-crystal 60 degrees waveguide bend. Firstly, the modal properties of the PC-PBS and the mechanism of the low-loss waveguide bend are investigated by the two-dimensional finite-difference time-domain (FDTD) method, and then the integration of the two devices is studied. It shows that, although the individual devices perform well separately, the performance of the integrated circuit is poor due to the multi-mode property of the PC-PBS. By introducing deformed airhole structures, a single-mode PC-PBS is proposed, which significantly enhance the performance of the circuit with the extinction ratios remaining above 20 dB for both transverse-electric (TE) and transverse-magnetic (TM) polarizations. Both the specific result and the general idea of integration design are promising in the photonic crystal integrated circuits in the future.
Beam position feedback system for the advanced photon source
International Nuclear Information System (INIS)
Chung, Y.
1994-01-01
The Advanced Photon Source (APS) will implement both global and local beam position feedback systems to stabilize the particle and x-ray beams for the storage ring. The systems consist of 20 VME crates distributed around the ring, each running multiple digital signal processors (DSP) running at 4 kHz sampling rate with a proportional, integral, and derivative (PID) control algorithm. The particle and x-ray beam position data is shared by the distributed processors through networked reflective memory. A theory of closed orbit correction using the technique of singular value decomposition (SVD) of the response matrix and simulation of its application to the APS storage ring will be discussed. This technique combines the global and local feedback systems and resolves the conflict among multiple local feedback systems due to local bump closure error. Maximum correction efficiency is achieved by feeding back to the global orbit data to the local feedback systems. The effect of the eddy current induced in the relatively thick (1/2 in.) vacuum chamber by the ac corrector magnet field for local feedback systems is compensated by digital filters. Results of experiments conducted on the x-ray ring of the National Synchrotron Light Source and the SPEAR at Stanford Synchrotron Radiation Laboratory will also be presented
Beam position feedback system for the advanced photon source
International Nuclear Information System (INIS)
Chung, Y.
1994-01-01
The Advanced Photon Source (APS) will implement both global and local beam position feedback systems to stabilize the particle and X-ray beams for the storage ring. The systems consist of 20 VME crates distributed around the ring, each running multiple digital signal processors (DSP) running at 4 kHz sampling rate with a proportional, integral, and derivative (PID) control algorithm. The particle and X-ray beam position data is shared by the distributed processors through networked reflective memory. A theory of closed orbit correction using the technique of singular value decomposition (SVD) of the response matrix and simulation of its application to the APS storage ring will be discussed. This technique combines the global and local feedback systems and resolves the conflict among multiple local feedback systems due to local bump closure error. Maximum correction efficiency is achieved by feeding back the global orbit data to the local feedback systems. The effect of the vacuum chamber eddy current induced by the AC corrector magnet field for local feedback systems is compensated by digital filters. Results of experiments conducted on the X-ray ring of the National Synchrotron Light Source and the SPEAR at Stanford Synchrotron Radiation Laboratory will be presented. copyright 1994 American Institute of Physics
Beam position feedback system for the Advanced Photon Source
International Nuclear Information System (INIS)
Chung, Y.
1993-01-01
The Advanced Photon Source (APS) will implement both global and local beam position feedback systems to stabilize the particle and X-ray beams for the storage ring. The systems consist of 20 VME crates distributed around the ring, each running multiple digital signal processors (DSP) running at 4kHz sampling rate with a proportional, integral, and derivative (PID) control algorithm. The particle and X-ray beam position data is shared by the distributed processors through networked reflective memory. A theory of closed orbit correction using the technique of singular value decomposition (SVD) of the response matrix and simulation of its application to the APS storage ring will be discussed. This technique combines the global and local feedback systems and resolves the conflict among multiple local feedback systems due to local bump closure error. Maximum correction efficiency is achieved by feeding back the global orbit data to the local feedback systems. The effect of the vacuum chamber eddy current induced by the AC corrector magnet field for local feedback systems is compensated by digital filters. Results of experiments conducted on the X-ray ring of the National Synchrotron Light Source and the SPEAR at Stanford Synchrotron Radiation Laboratory will be presented
Beam position feedback system for the Advanced Photon Source
International Nuclear Information System (INIS)
Chung, Y.
1993-01-01
The Advanced Photon Source (APS) will implement both global and local beam position feedback systems to stabilize the particle and X-ray beams for the storage ring. The systems consist of 20 VME crates distributed around the ring, each running multiple digital signal processors (DSP) running at 4kHz sampling rate with a proportional, integral, and derivative (PID) control algorithm. The particle and X-ray beam position data is shared by the distributed processors through networked reflective memory. A theory of closed orbit correction using the technique of singular value decomposition (SVD) of the response matrix and simulation of its application to the APS storage ring will be discussed. This technique combines the global and local feedback systems and resolves the conflict among multiple local feedback systems due to local bump closure error. Maximum correction efficiency is achieved by feeding back the global orbit data to the local feedback systems. The effect of the eddy current induced in the relatively thick (1/2 inch) vacuum chamber by the AC corrector magnet field for local feedback systems is compensated by digital filters. Results of experiments conducted on the X-ray ring of the National Synchrotron Light Source and the SPEAR at Stanford Synchrotron Radiation Laboratory will also be presented
Mini-beam collimator applications at the Advanced Photon Source
Energy Technology Data Exchange (ETDEWEB)
Xu Shenglan, E-mail: sxu@anl.gov [GM/CA CAT, Biosciences Division, Argonne National Laboratory, Argonne, IL 60439 (United States); Keefe, Lisa J.; Mulichak, Anne [IMCA CAT, Argonne National Laboratory, Argonne, IL 60439 (United States); Yan Lifen; Alp, Ercan E.; Zhao Jiyong [X-ray Sciences Division, Argonne National Laboratory, Argonne, IL 60439 (United States); Fischetti, Robert F. [GM/CA CAT, Biosciences Division, Argonne National Laboratory, Argonne, IL 60439 (United States)
2011-09-01
In 2007, the General Medicine and Cancer Institutes Collaborative Access Team (GM/CA CAT, Sector 23, Advanced Photon Source) began providing mini-beam collimators to its users. These collimators contained individual, 5- or 10-{mu}m pinholes and were rapidly exchangeable, thereby allowing users to tailor the beam size to their experimental needs. The use of these collimators provided a reduction in background noise, and thus improved the signal-to-noise ratio . Recent improvements in the collimator design include construction of the device from a monolithic piece of molybdenum with multiple pinholes mounted inside . This allows users to select from various size options from within the beamline control software without the realignment that was previously necessary. In addition, a new, 20-{mu}m pinhole has been added to create a 'quad-collimator', resulting in greater flexibility for the users. The mini-beam collimator is now available at multiple crystallographic beamlines and also is a part of the first Moessbauer Microscopic system at sector 3-ID.
Mini-beam collimator applications at the Advanced Photon Source
International Nuclear Information System (INIS)
Xu Shenglan; Keefe, Lisa J.; Mulichak, Anne; Yan Lifen; Alp, Ercan E.; Zhao Jiyong; Fischetti, Robert F.
2011-01-01
In 2007, the General Medicine and Cancer Institutes Collaborative Access Team (GM/CA CAT, Sector 23, Advanced Photon Source) began providing mini-beam collimators to its users. These collimators contained individual, 5- or 10-μm pinholes and were rapidly exchangeable, thereby allowing users to tailor the beam size to their experimental needs. The use of these collimators provided a reduction in background noise, and thus improved the signal-to-noise ratio . Recent improvements in the collimator design include construction of the device from a monolithic piece of molybdenum with multiple pinholes mounted inside . This allows users to select from various size options from within the beamline control software without the realignment that was previously necessary. In addition, a new, 20-μm pinhole has been added to create a 'quad-collimator', resulting in greater flexibility for the users. The mini-beam collimator is now available at multiple crystallographic beamlines and also is a part of the first Moessbauer Microscopic system at sector 3-ID.
Volume dose ratios relevant for alanine dosimetry in small, 6 MV photon beams
DEFF Research Database (Denmark)
Cronholm, Rickard O.; Andersen, Claus Erik; Behrens, Claus F.
2012-01-01
therapy). To this end, we here present the results of a Monte Carlo simulation study with DOSRZnrc that investigated the influence of field and detector size for small 6 MV photon beams. The study focusses on doses averaged over the volume of the detector rather than point doses.The ratio of volume...... averaged doses to water (D¯W) and alanine (D¯det) was found to be approximately 1.025 for most situations studied, and a constant ratio is likely to be representative for many applications in radiation therapy. However, D¯W/D¯det was found to be as low as 0.9908 ± 0.0037 in situations where one might...... expect significant deviations from charged particle equilibrium (i.e. at shallow depths and when the field size was smaller than the range of the secondary electrons). These effects therefore need consideration when finite-size alanine dosimeters are used under such conditions....
International Nuclear Information System (INIS)
Ren Gang; Zheng Wanhua; Wang Ke; Du Xiaoyu; Xing Mingxin; Chen Lianghui
2008-01-01
In this paper a compact polarization beam splitter based on a deformed photonic crystal directional coupler is designed and simulated. The transverse-electric (TE) guided mode and transverse-magnetic (TM) guided mode are split due to different guiding mechanisms. The effect of the shape deformation of the air holes on the coupler is studied. It discovered that the coupling strength of the coupled waveguides is strongly enhanced by introducing elliptical airholes, which reduce the device length to less than 18.5μm. A finite-difference time-domain simulation is performed to evaluate the performance of the device, and the extinction ratios for both TE and TM polarized light are higher than 20 dB. (classical areas of phenomenology)
Linear theory of a cold relativistic beam in a strongly magnetized finite-geometry plasma
International Nuclear Information System (INIS)
Gagne, R.R.J.; Shoucri, M.M.
1976-01-01
The linear theory of a finite-geometry cold relativistic beam propagating in a cold homogeneous finite-geometry plasma, is investigated in the case of a strongly magnetized plasma. The beam is assumed to propagate parallel to the external magnetic field. It is shown that the instability which takes place at the Cherenkov resonance ωapprox. =k/subz/v/subb/ is of the convective type. The effect of the finite geometry on the instability growth rate is studied and is shown to decrease the growth rate, with respect to the infinite geometry, by a factor depending on the ratio of the beam-to-plasma radius
Assessment of Structural Behavior of Non-corroded and Corroded RCC Beams Using Finite Element Method
Directory of Open Access Journals (Sweden)
Anand Parande
2008-09-01
Full Text Available A three dimensional finite element model is developed to examine the structural behaviour of corroded reinforced concrete beam and non corroded reinforced concrete beam. Non linear finite element analysis is performed using the ANSYS program. SOLID 65, LINK 8 element represent concrete and discrete reinforcing steel bars, based on each component actual characteristics, non linear material properties are defined for both elements. The effect of corrosion in reinforced concrete is studied by finite element analysis; an approach is developed to model the corrosion product expansion causing concrete cover cracking for this, beam has been modeled using ANSYS and using this data the beam has been casted with M20 concrete after 28 days the beam will be tested for flexural strength. The comparison between ANSYS prediction and field data are made in terms of deflection, stress, strain, bond strength and crack pattern of concrete beam.
Scattering of thermal photons by a 46 GeV positron beam at LEP
International Nuclear Information System (INIS)
Bini, C.; De Zorzi, G.; Diambrini-Palazzi, G.; Di Cosimo, G.; Di Domenico, A.; Gauzzi, P.; Zanello, D.
1991-01-01
The scattering of thermal photons present in the vacuum pipe of LEP against the high energy positron beam has been detected. The spectrum of the back-scattered photons is presented for a positron beam energy of 46.1 GeV. Measurements have been performed in the interaction region 1 with the LEP-5 experiment calorimeter. (orig.)
Ultracompact photonic crystal polarization beam splitter based on multimode interference.
Lu, Ming-Feng; Liao, Shan-Mei; Huang, Yang-Tung
2010-02-01
We propose a theoretical design for a compact photonic crystal (PC) polarization beam splitter (PBS) based on the multimode interference (MMI) effect. The size of a conventional MMI device designed by the self-imaging principle is not compact enough; therefore, we design a compact PC PBS based on the difference of the interference effect between TE and TM modes. Within the MMI coupler, the dependence of interference of modes on propagation distance is weak for a TE wave and strong for a TM wave; as a result, the length of the MMI section can be only seven lattice constants. Simulation results show that the insertion losses are 0.32 and 0.89 dB, and the extinction ratios are 14.4 and 17.5 dB for Port 1 (TE mode) and Port 2 (TM mode), respectively.
Rf-synchronized imaging for particle and photon beam characterizations
Energy Technology Data Exchange (ETDEWEB)
Lumpkin, A.H.
1993-07-01
The usefulness of imaging electro-optics for rf-driven accelerators can be enhanced by synchronizing the instruments to the system fundamental frequency or an appropriate subharmonic. This step allows one to obtain micropulse bunch length and phase during a series of linac bunches or storage ring passes. Several examples now exist of the use of synchroscan and dual-sweep streak cameras and/or image dissector tubes to access micropulse scale phenomena (10 to 30 ps) during linac and storage ring operations in the US, Japan, and Europe. As space permits, selections will be presented from the list of phase stability phenomena on photoelectric injectors, micropulse length during a macropulse, micropulse elongation effects, transverse Wakefield effects within a micropulse, and submicropulse phenomena on a stored beam. Potential applications to the subsystems of the Advanced Photon Source (APS) will be briefly addressed.
Parallel processing of dose calculation for external photon beam therapy
International Nuclear Information System (INIS)
Kunieda, Etsuo; Ando, Yutaka; Tsukamoto, Nobuhiro; Ito, Hisao; Kubo, Atsushi
1994-01-01
We implemented external photon beam dose calculation programs into a parallel processor system consisting of Transputers, 32-bit processors especially suitable for multi-processor configuration. Two network conformations, binary-tree and pipeline, were evaluated for rectangular and irregular field dose calculation algorithms. Although computation speed increased in proportion to the number of CPU, substantial overhead caused by inter-processor communication occurred when a smaller computation load was delivered to each processor. On the other hand, for irregular field calculation, which requires more computation capability for each calculation point, the communication overhead was still less even when more than 50 processors were involved. Real-time responses could be expected for more complex algorithms by increasing the number of processors. (author)
THE PAIR BEAM PRODUCTION SPECTRUM FROM PHOTON-PHOTON ANNIHILATION IN COSMIC VOIDS
Energy Technology Data Exchange (ETDEWEB)
Schlickeiser, R.; Ibscher, D. [Institut fuer Theoretische Physik, Lehrstuhl IV: Weltraum- und Astrophysik, Ruhr-Universitaet Bochum, D-44780 Bochum (Germany); Elyiv, A. [Institut d' Astrophysique et de Geophysique, Universite de Liege, B-4000 Liege (Belgium); Miniati, F., E-mail: rsch@tp4.rub.de, E-mail: ibscher@tp4.rub.de, E-mail: elyiv@astro.ulg.ac.be, E-mail: fm@phys.ethz.ch [Physics Department, Wolfgang-Pauli-Strasse 27, ETH-Zuerich, CH-8093 Zuerich (Switzerland)
2012-10-20
Highly beamed relativistic e {sup {+-}}-pair energy distributions result in double photon collisions of the beamed gamma rays from TeV blazars at cosmological distances with the isotropically distributed extragalactic background light (EBL) in the intergalactic medium. The typical energies k {sub 0} {approx_equal} 10{sup -7} in units of m{sub e}c {sup 2} of the EBL are more than 10 orders of magnitude smaller than the observed gamma-ray energies k {sub 1} {>=} 10{sup 7}. Using the limit k {sub 0} << k {sub 1}, we demonstrate that the angular distribution of the generated pairs in the lab frame is highly beamed in the direction of the initial gamma-ray photons. For the astrophysically important case of power-law distributions of the emitted gamma-ray beam up to the maximum energy M interacting with Wien-type N(k {sub 0}){proportional_to}k{sup q} {sub 0}exp (- k {sub 0}/{Theta}) soft photon distributions with total number density N {sub 0}, we calculate analytical approximations for the electron production spectrum. For distant objects with luminosity distances d{sub L} >> r {sub 0} = ({sigma} {sub T} N {sub 0}){sup -1} = 0.49N {sup -1} {sub 0} Mpc (with Thomson cross section {sigma} {sub T}), the implied large values of the optical depth {tau}{sub 0} = d{sub L} /r {sub 0} indicate that the electron production spectra differ at energies inside and outside the interval [({Theta}ln {tau}{sub 0}){sup -1}, {tau}{sub 0}/{Theta}], given the maximum gamma-ray energy M >> {Theta}{sup -1}. In the case M >> {Theta}{sup -1}, the production spectrum is strongly peaked near E {approx_equal} {Theta}{sup -1}, being exponentially reduced at small energies and decreasing with the steep power law {proportional_to}E {sup -1-p} up to the maximum energy E = M - (1/2).
SU-F-T-478: Effect of Deconvolution in Analysis of Mega Voltage Photon Beam Profiles
Energy Technology Data Exchange (ETDEWEB)
Muthukumaran, M [Apollo Speciality Hospitals, Chennai, Tamil Nadu (India); Manigandan, D [Fortis Cancer Institute, Mohali, Punjab (India); Murali, V; Chitra, S; Ganapathy, K [Apollo Speciality Hospital, Chennai, Tamil Nadu (India); Vikraman, S [JAYPEE HOSPITAL- RADIATION ONCOLOGY, Noida, UTTAR PRADESH (India)
2016-06-15
Purpose: To study and compare the penumbra of 6 MV and 15 MV photon beam profiles after deconvoluting different volume ionization chambers. Methods: 0.125cc Semi-Flex chamber, Markus Chamber and PTW Farmer chamber were used to measure the in-plane and cross-plane profiles at 5 cm depth for 6 MV and 15 MV photons. The profiles were measured for various field sizes starting from 2×2 cm till 30×30 cm. PTW TBA scan software was used for the measurements and the “deconvolution” functionality in the software was used to remove the volume averaging effect due to finite volume of the chamber along lateral and longitudinal directions for all the ionization chambers. The predicted true profile was compared and the change in penumbra before and after deconvolution was studied. Results: After deconvoluting the penumbra decreased by 1 mm for field sizes ranging from 2 × 2 cm till 20 x20 cm. This is observed for along both lateral and longitudinal directions. However for field sizes from 20 × 20 till 30 ×30 cm the difference in penumbra was around 1.2 till 1.8 mm. This was observed for both 6 MV and 15 MV photon beams. The penumbra was always lesser in the deconvoluted profiles for all the ionization chambers involved in the study. The variation in difference in penumbral values were in the order of 0.1 till 0.3 mm between the deconvoluted profile along lateral and longitudinal directions for all the chambers under study. Deconvolution of the profiles along longitudinal direction for Farmer chamber was not good and is not comparable with other deconvoluted profiles. Conclusion: The results of the deconvoluted profiles for 0.125cc and Markus chamber was comparable and the deconvolution functionality can be used to overcome the volume averaging effect.
Degirmenci, Elif; Landais, Pascal
2013-10-20
Photonic band gap and transmission characteristics of 2D metallic photonic crystals at THz frequencies have been investigated using finite element method (FEM). Photonic crystals composed of metallic rods in air, in square and triangular lattice arrangements, are considered for transverse electric and transverse magnetic polarizations. The modes and band gap characteristics of metallic photonic crystal structure are investigated by solving the eigenvalue problem over a unit cell of the lattice using periodic boundary conditions. A photonic band gap diagram of dielectric photonic crystal in square lattice array is also considered and compared with well-known plane wave expansion results verifying our FEM approach. The photonic band gap designs for both dielectric and metallic photonic crystals are consistent with previous studies obtained by different methods. Perfect match is obtained between photonic band gap diagrams and transmission spectra of corresponding lattice structure.
Dou, Xinyuan; Chen, Xiaonan; Chen, Maggie Yihong; Wang, Alan Xiaolong; Jiang, Wei; Chen, Ray T
2010-03-01
In this paper, we report the theoretical study of polymer-based photonic crystals for laser beam steering which is based on the superprism effect as well as the experiment fabrication of the two dimensional photonic crystals for the laser beam steering. Superprism effect, the principle for beam steering, was separately studied in details through EFC (Equifrequency Contour) analysis. Polymer based photonic crystals were fabricated through double exposure holographic interference method using SU8-2007. The experiment results showed a beam steering angle of 10 degree for 30 nm wavelength variation.
International Nuclear Information System (INIS)
Xu Xingsheng; Chen Hongda; Xiong Zhigang; Jin Aizi; Gu Changzhi; Cheng Bingying; Zhang Daozhong
2007-01-01
In this paper, we introduced the fabrication of photonic crystals on several kinds of semiconductor materials by using focused-ion beam machine, it shows that the method of focused-ion beam can fabricate two-dimensional photonic crystal and photonic crystal device efficiently, and the quality of the fabricated photonic crystal is high. Using the focused-ion beam method, we fabricate photonic crystal wavelength division multiplexer, and its characteristics are analyzed
Two-Photon-Absorption Scheme for Optical Beam Tracking
Ortiz, Gerardo G.; Farr, William H.
2011-01-01
A new optical beam tracking approach for free-space optical communication links using two-photon absorption (TPA) in a high-bandgap detector material was demonstrated. This tracking scheme is part of the canonical architecture described in the preceding article. TPA is used to track a long-wavelength transmit laser while direct absorption on the same sensor simultaneously tracks a shorter-wavelength beacon. The TPA responsivity was measured for silicon using a PIN photodiode at a laser beacon wavelength of 1,550 nm. As expected, the responsivity shows a linear dependence with incident power level. The responsivity slope is 4.5 x 10(exp -7) A/W2. Also, optical beam spots from the 1,550-nm laser beacon were characterized on commercial charge coupled device (CCD) and complementary metal-oxide semiconductor (CMOS) imagers with as little as 13.7 microWatts of optical power (see figure). This new tracker technology offers an innovative solution to reduce system complexity, improve transmit/receive isolation, improve optical efficiency, improve signal-to-noise ratio (SNR), and reduce cost for free-space optical communications transceivers.
CHARACTERIZATION OF GEOLOGICAL MATERIALS USING ION AND PHOTON BEAMS
International Nuclear Information System (INIS)
TOROK, SZ.B.; JONES, K.W.; TUNIZ, C.
1998-01-01
Geological specimens are often complex materials that require different analytical methods for their characterization. The parameters of interest may include the chemical composition of major, minor and trace elements. The chemical compounds incorporated in the minerals, the crystal structure and isotopic composition need to be considered. Specimens may be highly heterogeneous thus necessitating analytical methods capable of measurements on small sample volumes with high spatial resolution and sensitivity. Much essential information on geological materials can be obtained by using ion or photon beams. In this chapter we describe the principal analytical techniques based on particle accelerators, showing some applications that are hardly possible with conventional methods. In particular, the following techniques will be discussed: (1) Synchrotron radiation (SR) induced X-ray emission (SRIXE) and particle-induced X-ray emission (PEE) and other ion beam techniques for trace element analysis. (2) Accelerator mass spectrometry (AMS) for ultra sensitive analysis of stable nuclides and long-lived radionuclides. In most of the cases also the possibilities of elemental and isotopic analysis with high resolution will be discussed
Shimozato, Tomohiro; Aoyama, Yuichi; Matsunaga, Takuma; Tabushi, Katsuyoshi
2017-01-01
This work investigated the dosimetric properties of a 10-MV photon beam emitted from a medical linear accelerator (linac) with no flattening filter (FF). The aim of this study is to analyze the radiation fluence and energy emitted from the flattening filter free (FFF) linac using Monte Carlo (MC) simulations. The FFF linac was created by removing the FF from a linac in clinical use. Measurements of the depth dose (DD) and the off-axis profile were performed using a three-dimensional water phantom with an ionization chamber. A MC simulation for a 10-MV photon beam from this FFF linac was performed using the BEAMnrc code. The off-axis profiles for the FFF linac exhibited a chevron-like distribution, and the dose outside the irradiation field was found to be lower for the FFF linac than for a linac with an FF (FF linac). The DD curves for the FFF linac included many contaminant electrons in the build-up region. Therefore, for clinical use, a metal filter is additionally required to reduce the effects of the electron contamination. The mean energy of the FFF linac was found to be lower than that of the FF linac owing to the absence of beam hardening caused by the FF.
The transmission of finite amplitude sound beam in multi-layered biological media
Liu, Xiaozhou; Li, Junlun; Yin, Chang; Gong, Xiufen; Zhang, Dong; Xue, Honghui
2007-02-01
Based on the Khokhlov Zabolotskaya Kuznetsov (KZK) equation, a model in the frequency domain is given to describe the transmission of finite amplitude sound beam in multi-layered biological media. Favorable agreement between the theoretical analyses and the measured results shows this approach could effectively describe the transmission of finite amplitude sound wave in multi-layered biological media.
The transmission of finite amplitude sound beam in multi-layered biological media
Energy Technology Data Exchange (ETDEWEB)
Liu, Xiaozhou [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China)]. E-mail: xzliu@nju.edu.cn; Li, Junlun [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China); Yin, Chang [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China); Gong, Xiufen [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China); Zhang, Dong [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China); Xue, Honghui [Key Lab of Modern Acoustics, Institute of Acoustics, Nanjing University, Nanjing 210093 (China)
2007-02-19
Based on the Khokhlov-Zabolotskaya-Kuznetsov (KZK) equation, a model in the frequency domain is given to describe the transmission of finite amplitude sound beam in multi-layered biological media. Favorable agreement between the theoretical analyses and the measured results shows this approach could effectively describe the transmission of finite amplitude sound wave in multi-layered biological media.
The transmission of finite amplitude sound beam in multi-layered biological media
International Nuclear Information System (INIS)
Liu, Xiaozhou; Li, Junlun; Yin, Chang; Gong, Xiufen; Zhang, Dong; Xue, Honghui
2007-01-01
Based on the Khokhlov-Zabolotskaya-Kuznetsov (KZK) equation, a model in the frequency domain is given to describe the transmission of finite amplitude sound beam in multi-layered biological media. Favorable agreement between the theoretical analyses and the measured results shows this approach could effectively describe the transmission of finite amplitude sound wave in multi-layered biological media
On natural frequencies of non-uniform beams modulated by finite periodic cells
International Nuclear Information System (INIS)
Xu, Yanlong; Zhou, Xiaoling; Wang, Wei; Wang, Longqi; Peng, Fujun; Li, Bin
2016-01-01
It is well known that an infinite periodic beam can support flexural wave band gaps. However, in real applications, the number of the periodic cells is always limited. If a uniform beam is replaced by a non-uniform beam with finite periodicity, the vibration changes are vital by mysterious. This paper employs the transfer matrix method (TMM) to study the natural frequencies of the non-uniform beams with modulation by finite periodic cells. The effects of the amounts, cross section ratios, and arrangement forms of the periodic cells on the natural frequencies are explored. The relationship between the natural frequencies of the non-uniform beams with finite periodicity and the band gap boundaries of the corresponding infinite periodic beam is also investigated. Numerical results and conclusions obtained here are favorable for designing beams with good vibration control ability. - Highlights: • The transfer matrix method to study the natural frequencies of the finite periodic non-uniform beams is derived. • The transfer matrix method to study the band gaps of the infinite periodic non-uniform beams is derived. • The effects of the periodic cells on the natural frequencies are explored. • The relationships of the natural frequencies and band gap boundaries are investigated.
On natural frequencies of non-uniform beams modulated by finite periodic cells
Energy Technology Data Exchange (ETDEWEB)
Xu, Yanlong, E-mail: xuyanlong@nwpu.edu.cn [School of Aeronautics, Northwestern Polytechnical University, Xi' an 710072, Shaanxi (China); Zhou, Xiaoling [Shanghai Institute of Aerospace System Engineering, Shanghai 201109 (China); Wang, Wei [School of Aeronautics, Northwestern Polytechnical University, Xi' an 710072, Shaanxi (China); Wang, Longqi [School of Civil & Environmental Engineering, Nanyang Technological University, 50 Nanyang Avenue, Singapore 639798 (Singapore); Peng, Fujun [Shanghai Institute of Aerospace System Engineering, Shanghai 201109 (China); Li, Bin [School of Aeronautics, Northwestern Polytechnical University, Xi' an 710072, Shaanxi (China)
2016-09-23
It is well known that an infinite periodic beam can support flexural wave band gaps. However, in real applications, the number of the periodic cells is always limited. If a uniform beam is replaced by a non-uniform beam with finite periodicity, the vibration changes are vital by mysterious. This paper employs the transfer matrix method (TMM) to study the natural frequencies of the non-uniform beams with modulation by finite periodic cells. The effects of the amounts, cross section ratios, and arrangement forms of the periodic cells on the natural frequencies are explored. The relationship between the natural frequencies of the non-uniform beams with finite periodicity and the band gap boundaries of the corresponding infinite periodic beam is also investigated. Numerical results and conclusions obtained here are favorable for designing beams with good vibration control ability. - Highlights: • The transfer matrix method to study the natural frequencies of the finite periodic non-uniform beams is derived. • The transfer matrix method to study the band gaps of the infinite periodic non-uniform beams is derived. • The effects of the periodic cells on the natural frequencies are explored. • The relationships of the natural frequencies and band gap boundaries are investigated.
Present status of beam position stabilization at photon factory storage ring
International Nuclear Information System (INIS)
Nakamura, Norio
1990-01-01
Stabilization of photon beam position became a major issue in the operation of the storage rings dedicated as synchrotron radiation source. At the Photon Factory storage ring (PF ring), the orbit movement appeared remarkably when the low-emittance operation started. This orbit movement became a serious problem to synchrotron radiation users because the photon beam to drift with a large amplitude. The horizontal and vertical orbit feedback systems were constructed and developed in order to suppress the orbit movement globally. As a result, the horizontal and vertical orbit movements were reduced by a factor of five and ten, respectively. In addition, another type of feedback system using a local bump was constructed. In the test operation, this system could remove the fast photon beam motion as well as the slow photon beam drift for a beamline. (author)
A dark hollow beam from a selectively liquid-filled photonic crystal fibre
International Nuclear Information System (INIS)
Mei-Yan, Zhang; Shu-Guang, Li; Yan-Yan, Yao; Bo, Fu; Lei, Zhang
2010-01-01
This paper reports that, based on the electromagnetic scattering theory of the multipole method, a high-quality hollow beam is produced through a selectively liquid-filled photonic crystal fibre. Instead of a doughnut shape, a typical hollow beam is produced by other methods; the mode-field images of the hollow-beam photonic crystal fibre satisfy sixth-order rotation symmetry, according to the symmetry of the photonic crystal fibre (PCF) structure. A dark spot size of the liquid-filled photonic crystal fibre-generated hollow beam can be tuned by inserting liquid into the cladding region and varying the photonic crystal fibre structure parameters. The liquid-filled PCF makes a convenient and flexible tool for the guiding and trapping of atoms and the creation of all-fibre optical tweezers. (condensed matter: electronic structure, electrical, magnetic, and optical properties)
International Nuclear Information System (INIS)
Lumpkin, A.; Wang, X.; Sellyey, W.; Patterson, D.; Kahana, E.
1994-01-01
During the first half of 1994, a number of the diagnostic systems for measurement of the charged-particle beam parameters throughout the subsystems of the Advanced Photon Source (APS) have been installed and tested. The particle beams eventually will involve 450-MeV to 7-GeV positrons and with different pulse formats. The first test and commissionin results for beam profiles, beam position monitors, loss rate monitors, current monitors, and synchrotron radiation photon monitors hve been obtained using 200- to 350-MeV electron beams injected into the subsystems. Data presented are principally from the transport lines and the positron accumulator ring
International Nuclear Information System (INIS)
Jangda, A.Q.; Hussein, S.
2012-01-01
In external beam radiation therapy (EBRT), the quality assurance (QA) of the radiation beam is crucial to the accurate delivery of the prescribed dose to the patient. One of the dosimetric parameters that require monitoring is the beam output, specified as the dose rate on the central axis under reference conditions. The aim of this project was to validate dose rate calibration of megavoltage photon beams using the International Atomic Energy Agency (IAEA)/World Health Organisation (WHO) postal audit dosimetry service. Three photon beams were audited: a 6 MV beam from the low-energy linac and 6 and 18 MV beams from a dual high-energy linac. The agreement between our stated doses and the IAEA results was within 1% for the two 6 MV beams and within 2% for the 18 MV beam. The IAEA/WHO postal audit dosimetry service provides an independent verification of dose rate calibration protocol by an international facility. (author)
Jangda, Abdul Qadir; Hussein, Sherali
2012-05-01
In external beam radiation therapy (EBRT), the quality assurance (QA) of the radiation beam is crucial to the accurate delivery of the prescribed dose to the patient. One of the dosimetric parameters that require monitoring is the beam output, specified as the dose rate on the central axis under reference conditions. The aim of this project was to validate dose rate calibration of megavoltage photon beams using the International Atomic Energy Agency (IAEA)/World Health Organisation (WHO) postal audit dosimetry service. Three photon beams were audited: a 6 MV beam from the low-energy linac and 6 and 18 MV beams from a dual high-energy linac. The agreement between our stated doses and the IAEA results was within 1% for the two 6 MV beams and within 2% for the 18 MV beam. The IAEA/WHO postal audit dosimetry service provides an independent verification of dose rate calibration protocol by an international facility.
Charge and finite size corrections for virtual photon spectra in second order Born approximation
International Nuclear Information System (INIS)
Durgapal, P.
1982-01-01
The purpose of this work is to investigate the effects of finite nuclear size and charge on the spectrum of virtual photons emitted when a relativistic electron is scattered in the field of an atomic nucleus. The method consisted in expanding the scattering cross section in terms of integrals over the nuclear inelastic form factor with a kernel which was evaluated in second order Born approximation and was derived from the elastic-electron scattering form factor. The kernel could be evaluated analytically provided the elastic form factor contained only poles. For this reason the author used a Yukawa form factor. Before calculating the second order term the author studied the first order term containing finite size effects in the inelastic form factor. The author observed that the virtual photon spectrum is insensitive to the details of the inelastic distribution over a large range of energies and depends only on the transition radius. This gave the author the freedom of choosing an inelastic distribution for which the form factor has only poles and the author chose a modified form of the exponential distribution, which enabled the author to evaluate the matrix element analytically. The remaining integral over the physical momentum transfer was performed numerically. The author evaluated the virtual photon spectra for E1 and M1 transitions for a variety of electron energies using several nuclei and compared the results with the distorted wave calculations. Except for low energy and high Z, the second order results compared well with the distorted wave calculations
Excitation of lower hybrid waves by electron beams in finite geometry plasmas
International Nuclear Information System (INIS)
Shoucri, M.m.; Gagne, R.R.J.
1978-01-01
The quasi-static lower hybrid eigenmodes of a plasma column in a cylindrical waveguide are determined, and their linear excitation by a small density electron beam is discussed for the cases of a hot electron beam as well as for a cold electron beam. It is shown that under certain conditions, finite geometry effects introduce important quantitative and qualitative differences with respect to the results obtained in an infinite geometry. (author)
Matching beams on photon/electron linear accelerators
International Nuclear Information System (INIS)
Oliver, L.; Vial, P.; Hunt, P.
2004-01-01
Full text: There are a number of obvious reasons to match megavoltage X-ray and electron beams for clinical purposes. If two dual-purpose X-ray/electron linear accelerators are of the same design and manufacturer, then this might be possible. The issue is however whether the beams can be matched sufficiently close to be considered the same for patient treatments and planning data for dose calculation purposes. If successfully achieved, there are significant advantages in reduced commissioning time, less work in planning and flexibility in the treatment of patients between the two treatment machines. We have investigated matching a new Varian Clinac 21EX with our 1993 Varian Clinac 2100 C/D. A Varian Clinac 1800 was the first linear accelerator installed at RNSH in 1987. When the Clinac 2100 C/D was installed in 1993, we attempted to match all the X-ray and electron beams with the original Clinac 1800 physical data. The X-ray beam characteristics were satisfactory but the electron beams were not sufficiently compatible for planning or patient treatment purposes. A different designed scattering foil and electron applicator were the cause of the different electron beam physical characteristics between the two models. In replacing the Clinac 1800 with the Clinac 21EX, we have used the original 1993 data of the Clinac 2100 C/D as the gold standard to aim for. Initial measurements during acceptance tests showed that all beams satisfied the manufacturer's specification. The energy was then matched to the existing clinical physics data by adjusting the bending magnet power supply and re-tuning the accelerator. This involved matching % depth dose and the corresponding ratio of 10 and 20 cm % depth dose ratio for 6MV and 18 MV X-ray beams. For 6, 9, 12, 16 and 20 MeV electron beams the normal physical parameters of depth of maximum (R max ), the practical range (R p ), the depth of 50% (R 50 ), the slope (G), the average energy at the surface (E 0 ) and the % photon
Liquid gallium cooling of silicon crystals in high intensity photon beams
International Nuclear Information System (INIS)
Smither, R.K.; Forster, G.A.; Bilderback, D.H.; Bedzyk, M.; Finkelstein, K.; Henderson, C.; White, J.; Berman, L.E.; Stefan, P.; Oversluizen, T.
1989-01-01
The high-brilliance, insertion-device-based photon beams of the next generation of synchrotron sources (Argonne's APS and Grenoble's ESRF) will deliver large thermal loads (1--10 kW) to the first optical elements. Considering the problems that present synchrotron users are experiencing with beams from recently installed insertion devices, new and improved methods of cooling these first optical elements, particularly when they are diffraction crystals, are clearly needed. A series of finite element calculations were performed to test the efficiency of new cooling geometries and various cooling fluids. The best results were obtained with liquid Ga metal flowing in channels just below the surface of the crystal. Ga was selected because of its good thermal conductivity and thermal capacity, low melting point, high boiling point, low kinetic viscosity, and very low vapor pressure. Its very low vapor pressure, even at elevated temperatures, makes it especially attractive in UHV conditions. A series of experiments were conducted at CHESS in February of 1988 that compared liquid gallium-cooled silicon diffraction crystals with water-cooled crystals. A six-pole wiggler beam was used to perform these tests on three different Si crystals, two with new cooling geometries and the one presently in use. A special high-pressure electromagnetic induction pump, recently developed at Argonne, was used to circulate the liquid gallium through the silicon crystals. In all experiments, the specially cooled crystal was used as the first crystal in a two crystal monochromator. An infrared camera was used to monitor the thermal profiles and correlated them with rocking curve measurements. A second set of cooling experiments were conducted in June of 1988 that used the intense, highly collimated beam from the newly installed ANL/CHESS undulator
Liquid gallium cooling of silicon crystals in high intensity photon beam
International Nuclear Information System (INIS)
Smither, R.K.; Forster, G.A.; Bilderback, D.H.
1988-11-01
The high-brilliance, insertion-device-based, photon beams of the next generation of synchrotron sources will deliver large thermal loads (1 kW to 10 kW) to the first optical elements. Considering the problems that present synchrotron users are experiencing with beams from recently installed insertion devices, new and improved methods of cooling these first optical elements, particularly when they are diffraction crystals, are clearly needed. A series of finite element calculations were performed to test the efficiency of new cooling geometries and new cooling fluids. The best results were obtained with liquid Ga metal flowing in channels just below the surface of the crystal. Ga was selected because of its good thermal conductivity and thermal capacity, low melting point, high boiling point, low kinetic viscosity, and very low vapor pressure. Its very low vapor pressure, even at elevated temperatures, makes it especially attractive in uhv conditions. A series of experiments were conducted at CHESS in February of 1988 that compared liquid gallium cooled silicon diffraction crystals with water cooled crystals. 2 refs., 16 figs., 1 tab
Casimir effect at finite temperature for pure-photon sector of the minimal Standard Model Extension
Energy Technology Data Exchange (ETDEWEB)
Santos, A.F., E-mail: alesandroferreira@fisica.ufmt.br [Instituto de Física, Universidade Federal de Mato Grosso, 78060-900, Cuiabá, Mato Grosso (Brazil); Department of Physics and Astronomy, University of Victoria, 3800 Finnerty Road Victoria, BC (Canada); Khanna, Faqir C., E-mail: khannaf@uvic.ca [Department of Physics and Astronomy, University of Victoria, 3800 Finnerty Road Victoria, BC (Canada)
2016-12-15
Dynamics between particles is governed by Lorentz and CPT symmetry. There is a violation of Parity (P) and CP symmetry at low levels. The unified theory, that includes particle physics and quantum gravity, may be expected to be covariant with Lorentz and CPT symmetry. At high enough energies, will the unified theory display violation of any symmetry? The Standard Model Extension (SME), with Lorentz and CPT violating terms, has been suggested to include particle dynamics. The minimal SME in the pure photon sector is considered in order to calculate the Casimir effect at finite temperature.
Reflection and Transmission of a Focused Finite Amplitude Sound Beam Incident on a Curved Interface
Makin, Inder Raj Singh
Reflection and transmission of a finite amplitude focused sound beam at a weakly curved interface separating two fluid-like media are investigated. The KZK parabolic wave equation, which accounts for thermoviscous absorption, diffraction, and nonlinearity, is used to describe the high intensity focused beam. The first part of the work deals with the quasilinear analysis of a weakly nonlinear beam after its reflection and transmission from a curved interface. A Green's function approach is used to define the field integrals describing the primary and the nonlinearly generated second harmonic beam. Closed-form solutions are obtained for the primary and second harmonic beams when a Gaussian amplitude distribution at the source is assumed. The second part of the research uses a numerical frequency domain solution of the KZK equation for a fully nonlinear analysis of the reflected and transmitted fields. Both piston and Gaussian sources are considered. Harmonic components generated in the medium due to propagation of the focused beam are evaluated, and formation of shocks in the reflected and transmitted beams is investigated. A finite amplitude focused beam is observed to be modified due to reflection and transmission from a curved interface in a manner distinct from that in the case of a small signal beam. Propagation curves, beam patterns, phase plots and time waveforms for various parameters defining the source and media pairs are presented, highlighting the effect of the interface curvature on the reflected and transmitted beams. Relevance of the current work to biomedical applications of ultrasound is discussed.
Photon polarization tensor in the light front field theory at zero and finite temperatures
International Nuclear Information System (INIS)
Silva, Charles da Rocha; Perez, Silvana; Strauss, Stefan
2012-01-01
Full text: In recent years, light front quantized field theories have been successfully generalized to finite temperature. The light front frame was introduced by Dirac , and the quantization of field theories on the null-plane has found applications in many branches of physics. In order to obtain the thermal contribution, we consider the hard thermal loop approximation. This technique was developed by Braaten and Pisarski for the thermal quantum field theory at equal times and is particularly useful to extract the leading thermal contributions to the amplitudes in perturbative quantum field theories. In this work, we consider the light front quantum electrodynamics in (3+1) dimensions and evaluate the photon polarization tensor at one loop for both zero and finite temperatures. In the first case, we apply the dimensional regularization method to extract the finite contribution and find the transverse structure for the amplitude in terms of the light front coordinates. The result agrees with one-loop covariant calculation. For the thermal corrections, we generalize the hard thermal loop approximation to the light front and calculate the dominant temperature contribution to the polarization tensor, consistent with the Ward identity. In both zero as well as finite temperature calculations, we use the oblique light front coordinates. (author)
Modelling Convergence of Finite Element Analysis of Cantilever Beam
African Journals Online (AJOL)
Convergence studies are carried out by investigating the convergence of numerical results as the number of elements is increased. If convergence is not obtained, the engineer using the finite element method has absolutely no indication whether the results are indicative of a meaningful approximation to the correct solution ...
Integrated single- and two-photon light sheet microscopy using accelerating beams
DEFF Research Database (Denmark)
Piksarv, Peeter; Marti, Dominik; Le, Tuan
2017-01-01
We demonstrate the first light sheet microscope using propagation invariant, accelerating Airy beams that operates both in single- and two-photon modes. The use of the Airy beam permits us to develop an ultra compact, high resolution light sheet system without beam scanning. In two-photon mode......, an increase in the field of view over the use of a standard Gaussian beam by a factor of six is demonstrated. This implementation for light sheet microscopy opens up new possibilities across a wide range of biomedical applications, especially for the study of neuronal processes....
The APS x-ray undulator photon beam position monitor and tests at CHESS and NSLS
International Nuclear Information System (INIS)
Shu, D.; Rodricks, B.; Barraza, J.; Sanchez, T.; Kuzay, T.M.
1992-01-01
The advent of third generation synchrotron radiation sources, like the Advanced Photon Source (APS), will provide significant increases in brilliance over existing synchrotron sources. The APS x-ray undulators will increase the brilliance in the 3-40 KeV range by several orders of magnitude. Thus, the design of the photon beam position monitor is a challenging engineering task. The beam position monitors must withstand the high thermal load, be able to achieve sub-micron spatial resolution while maintaining their stability, and be compatible with both undulators and wigglers. A preliminary APS prototype photon beam position monitor consisting of a CVD-diamond-based, tungsten-coated blade was tested on the APS/CHESS undulator at the Cornell High Energy Synchrotron Radiation Source (CHESS) and on the NSLS X-13 undulator beamline. Results from these tests, as well as the design of this prototype APS photon beam position monitor, will be discussed in this paper
The APS X-ray undulator photon beam position monitor and tests at CHESS and NSLS
International Nuclear Information System (INIS)
Shu, D.; Rodricks, B.; Barraza, J.; Sanchez, T.; Kuzay, T.M.
1992-01-01
The advent of thirs generation synchrotron sources, like the Advanced Photon Source (APS), will provide significant increases in brilliance over existing synchrotron sources. The APS X-ray undulators will increase the brilliance in the 3-40 keV range by several orders of magnitude. Thus, the design of the photon beam position monitor is a challenging engineering task. The beam position monitors must withstand the high thermal load, be able to achieve submicron spatial resolution while maintaining their stability, and be compatible with both undulators and wigglers. A preliminary APS prototype photon beam position monitor consisting of a CVD-diamond-based, tungsten-coated blade was tested on the APS/CHESS undulator at the Cornell High Energy Synchrotron Radiation Source (CHESS) and on the NSLS X-13 undulator beamline. Results from these tests, as well as the design of this prototype APS photon beam position monitor, will be discussed in this paper. (orig.)
Theoretical and experimental study of an energy-reinforced braking radiation photon beam
International Nuclear Information System (INIS)
Bertin, Pierre-Yves
1966-01-01
This research thesis reports the theoretical study of a photon beam raised towards high energies, its experimental implementation, the definition of a gamma spectrometry method which aimed at checking various hypotheses used in the beam theoretical study. After a presentation of the theory of phenomena of electron braking radiation, of materialisation of photons into positon-negaton pair, and of issues related to multiple Coulomb diffusion, the author reports the study of the different solutions which allow a photon beam to be obtained. A braking radiation of mono-kinetic electron has been used. This braking radiation is reinforced by absorption of low energy protons in a column of lithium hydride. The author describes how the beam is built up, and the experimental approach. He describes how raw data are processed to get rid of the influence of the multiple Coulomb diffusion and of the braking radiation. Experimental results are compared with those obtained by convolution of photon spectra and differential cross section
International Nuclear Information System (INIS)
Ganguli, G.; Palmadesso, P.
1984-01-01
Finite geometry effects on the stability properties of a charged beam propagating through an intense relativistic annular electron beam have been studied. The stability of the system under transverse oscillation has been examined in detail in a parameter domain pertinent to the collective particle accelerator, currently under development at the Naval Research Laboratory. Both the normal mode and the convective aspects of this instability have been investigated. Despite a substantial temporal growth rate as predicted by the normal mode approach, this instability does not prevent successful acceleration of a portion of the axial beam. Thus the transverse oscillation is not fatal to the collective particle accelerator operation
International Nuclear Information System (INIS)
Irazola, L.; Terrón, J.A.; Bedogni, R; Pola, A.; Lorenzoli, M.; Sánchez-Nieto, B.; Gómez, F.; Sánchez-Doblado, F.
2016-01-01
The increasing interest of the medical community to radioinduced second malignancies due to photoneutrons in patients undergoing high-energy radiotherapy, has stimulated in recent years the study of peripheral doses, including the development of some dedicated active detectors. Although these devices are designed to respond to neutrons only, their parasitic photon response is usually not identically zero and anisotropic. The impact of these facts on measurement accuracy can be important, especially in points close to the photon field-edge. A simple method to estimate the photon contribution to detector readings is to cover it with a thermal neutron absorber with reduced secondary photon emission, such as a borated rubber. This technique was applied to the TNRD (Thermal Neutron Rate Detector), recently validated for thermal neutron measurements in high-energy photon radiotherapy. The positive results, together with the accessibility of the method, encourage its application to other detectors and different clinical scenarios. - Highlights: • Neutron-to-photon discrimination of a thermal neutron detector used in radiotherapy. • Photon and anisotropic response study with distance and beam incidence of thermal neutron detector. • Borated rubber for estimating photon contribution in any thermal neutron detector.
Absolute calibration of photon-number-resolving detectors with an analog output using twin beams
Czech Academy of Sciences Publication Activity Database
Peřina Jr., J.; Haderka, Ondřej; Allevi, A.; Bondani, M.
2014-01-01
Roč. 104, č. 4 (2014), "041113-1"-"041113-4" ISSN 0003-6951 R&D Projects: GA ČR GAP205/12/0382 Institutional support: RVO:68378271 Keywords : photon- number resolving detector * twin beams * photon fields Subject RIV: BH - Optics, Masers, Lasers Impact factor: 3.302, year: 2014
Corpuscular Model of Two-Beam Interference and Double-Slit Experiments with Single Photons
Jin, Fengping; Yuan, Shengjun; De Raedt, Hans; Michielsen, Kristel; Miyashita, Seiji
We introduce an event-based corpuscular simulation model that reproduces the wave mechanical results of single-photon double-slit and two-beam interference experiments and (of a one-to-one copy of an experimental realization) of a single-photon interference experiment with a Fresnel biprism. The
Energy Technology Data Exchange (ETDEWEB)
Lonski, Peta; Kron, Tomas [Peter MacCallum Cancer Centre, Melbourne (Australia); RMIT Univ., Melbourne (Australia); Ramachandran, Prabhakar; Franich, Rick [Peter MacCallum Cancer Centre, Melbourne (Australia)
2017-07-01
This study examines the difference in surface dose between flat and flattening filter free (FFF) photon beams in the context of breast radiotherapy. The surface dose was measured for 6 MV, 6 MV FFF, 10 MV, 10 MV FFF and 18 MV photon beams using a thin window ionisation chamber for various field sizes. Profiles were acquired to ascertain the change in surface dose off-axis. Out-of-field measurements were included in a clinically representative half beam block tangential breast field. In the field centres of FFF beams the surface dose was found to be increased for small fields and decreased for large fields compared to flat beams. For FFF beams, surface dose was found to decrease off-axis and resulted in lower surface dose out-of-field compared to flat beams.
Finite Element Analysis of Walking Beam of a New Compound Adjustment Balance Pumping Unit
Wu, Jufei; Wang, Qian; Han, Yunfei
2017-12-01
In this paper, taking the designer of the new compound balance pumping unit beam as our research target, the three-dimensional model is established by Solid Works, the load and the constraint are determined. ANSYS Workbench is used to analyze the tail and the whole of the beam, the stress and deformation are obtained to meet the strength requirements. The finite element simulation and theoretical calculation of the moment of the center axis beam are carried out. The finite element simulation results are compared with the calculated results of the theoretical mechanics model to verify the correctness of the theoretical calculation. Finally, the finite element analysis is consistent with the theoretical calculation results. The theoretical calculation results are preferable, and the bending moment value provides the theoretical reference for the follow-up optimization and research design.
Wavelet-based spectral finite element dynamic analysis for an axially moving Timoshenko beam
Mokhtari, Ali; Mirdamadi, Hamid Reza; Ghayour, Mostafa
2017-08-01
In this article, wavelet-based spectral finite element (WSFE) model is formulated for time domain and wave domain dynamic analysis of an axially moving Timoshenko beam subjected to axial pretension. The formulation is similar to conventional FFT-based spectral finite element (SFE) model except that Daubechies wavelet basis functions are used for temporal discretization of the governing partial differential equations into a set of ordinary differential equations. The localized nature of Daubechies wavelet basis functions helps to rule out problems of SFE model due to periodicity assumption, especially during inverse Fourier transformation and back to time domain. The high accuracy of WSFE model is then evaluated by comparing its results with those of conventional finite element and SFE results. The effects of moving beam speed and axial tensile force on vibration and wave characteristics, and static and dynamic stabilities of moving beam are investigated.
The ESTRO-EQUAL quality assurance network for photon and electron radiotherapy beams in Germany
International Nuclear Information System (INIS)
Ferreira, I.H.; Dutreix, A.; Richter, J.; Bridier, A.; Chavaudra, J.; Svensson, H.
2001-01-01
Background: In 1998 an ESTRO Quality Assurance Network for radiotherapy (EQUAL) has been set up for 25 European countries for photon and electron beams in reference and non-reference conditions. Material and Methods: Measurements are done using LiF powder (DTL937-Philitech, France) that is processed with the PCL3 automatic reader (Fimel-PTW). The participating centers irradiate the TLDs with an absorbed dose of 2 Gy according to the clinical routine. Results: Until September 2000 EQUAL has checked 135 photon beams (including the beams rechecked) from 51 radiotherapy centers in Germany out of 86 accepted centers. The results show that 2% of the beam outputs in reference conditions and 3% of the percentage depth doses are outside the tolerance level (deviation > ± 5%). 6% of the beam output variations and of the wedge transmission factors show deviations > ± 5%. The global analysis of results shows deviations > ± 5% in at least one parameter for 18 beams out of the 135 beams checked. Five rechecked beams present one ''real dosimetric'' problem in one or more parameters, corresponding to 4% of the 114 beams for which the deviations cannot be attributed to set-up errors. - The EQUAL network has checked 89 electron beams in Germany. The results show that all beam outputs checked are within the tolerance level. The standard deviation for the beam output in reference conditions is 2.0% and 2.2% for the beam output for the others field sizes. The percentage of deviations > 3% and ≤ 5% for the reference beam output is higher for electron beams than for photon beam checks. Therefore the electron beam calibration and the TPS algorithms should be improved to increase the accuracy of the patient dosimetry for radiotherapy. (orig.) [de
Heck, Martijn J. R.
2017-01-01
Technologies for efficient generation and fast scanning of narrow free-space laser beams find major applications in three-dimensional (3D) imaging and mapping, like Lidar for remote sensing and navigation, and secure free-space optical communications. The ultimate goal for such a system is to reduce its size, weight, and power consumption, so that it can be mounted on, e.g. drones and autonomous cars. Moreover, beam scanning should ideally be done at video frame rates, something that is beyond the capabilities of current opto-mechanical systems. Photonic integrated circuit (PIC) technology holds the promise of achieving low-cost, compact, robust and energy-efficient complex optical systems. PICs integrate, for example, lasers, modulators, detectors, and filters on a single piece of semiconductor, typically silicon or indium phosphide, much like electronic integrated circuits. This technology is maturing fast, driven by high-bandwidth communications applications, and mature fabrication facilities. State-of-the-art commercial PICs integrate hundreds of elements, and the integration of thousands of elements has been shown in the laboratory. Over the last few years, there has been a considerable research effort to integrate beam steering systems on a PIC, and various beam steering demonstrators based on optical phased arrays have been realized. Arrays of up to thousands of coherent emitters, including their phase and amplitude control, have been integrated, and various applications have been explored. In this review paper, I will present an overview of the state of the art of this technology and its opportunities, illustrated by recent breakthroughs.
Directory of Open Access Journals (Sweden)
Heck Martijn J.R.
2016-06-01
Full Text Available Technologies for efficient generation and fast scanning of narrow free-space laser beams find major applications in three-dimensional (3D imaging and mapping, like Lidar for remote sensing and navigation, and secure free-space optical communications. The ultimate goal for such a system is to reduce its size, weight, and power consumption, so that it can be mounted on, e.g. drones and autonomous cars. Moreover, beam scanning should ideally be done at video frame rates, something that is beyond the capabilities of current opto-mechanical systems. Photonic integrated circuit (PIC technology holds the promise of achieving low-cost, compact, robust and energy-efficient complex optical systems. PICs integrate, for example, lasers, modulators, detectors, and filters on a single piece of semiconductor, typically silicon or indium phosphide, much like electronic integrated circuits. This technology is maturing fast, driven by high-bandwidth communications applications, and mature fabrication facilities. State-of-the-art commercial PICs integrate hundreds of elements, and the integration of thousands of elements has been shown in the laboratory. Over the last few years, there has been a considerable research effort to integrate beam steering systems on a PIC, and various beam steering demonstrators based on optical phased arrays have been realized. Arrays of up to thousands of coherent emitters, including their phase and amplitude control, have been integrated, and various applications have been explored. In this review paper, I will present an overview of the state of the art of this technology and its opportunities, illustrated by recent breakthroughs.
International Nuclear Information System (INIS)
Palm, A.; Mattsson, Olof
2002-01-01
The implementation of protocols based on absorbed dose to water standards requires beam quality conversion factors, k Q . Calculated values of k Q are available for ionization chambers used for reference dosimetry. Ideally, k Q should be experimentally determined at the same beam qualities as that of the user. In this work we measure k Q factors in clinical photon beams and compare them with calculated and measured values. Beam quality conversion factors are determined for clinical photon beams of nominal energies 4 MV, 6 MV, 15 MV, and 25 MV, for commonly used cylindrical ionization chambers. Twelve chambers of eight different types are used. For three of them, no experimental data have previously been available. The experimental procedure is based on measurements with ionization chambers and Fricke dosimetry in the reference beam ( 60 Co γ radiation) and in clinical linear accelerator beams. The k Q values determined in this work generally agree within 0.5% with previously reported experimental values both when %dd(10) x and TPR 20,10 are used for beam quality specification. The agreement with calculated data is generally within 0.5%, except for the 15 MV beam. For this beam the measured values are usually between 0.5% and 1% lower than the data taken from the TG-51 protocol or the TRS-398 code of practice. For three NE2571 chambers and three NE2581 chambers, the maximum observed deviation of individual k Q values is 0.2% and 0.4%, respectively
FINITE ELEMENT MODELING OF CAMBER OF PRESTRESSED CONCRETE BEAMS
Directory of Open Access Journals (Sweden)
Peter P. Gaigerov
2017-12-01
Full Text Available For large-span reinforced concrete beam structures developed by the method of determining the camber due to the prestressing of a steel rope on the concrete. Performed numerical experiments to study the impact of various schemes layout prestressed reinforcement without bonding with concrete on the distribution of the relief efforts along the path of the reinforcement.
Finite element formulation for dynamics of planar flexible multi-beam system
International Nuclear Information System (INIS)
Liu Zhuyong; Hong Jiazhen; Liu Jinyang
2009-01-01
In some previous geometric nonlinear finite element formulations, due to the use of axial displacement, the contribution of all the elements lying between the reference node of zero axial displacement and the element to the foreshortening effect should be taken into account. In this paper, a finite element formulation is proposed based on geometric nonlinear elastic theory and finite element technique. The coupling deformation terms of an arbitrary point only relate to the nodal coordinates of the element at which the point is located. Based on Hamilton principle, dynamic equations of elastic beams undergoing large overall motions are derived. To investigate the effect of coupling deformation terms on system dynamic characters and reduce the dynamic equations, a complete dynamic model and three reduced models of hub-beam are prospected. When the Cartesian deformation coordinates are adopted, the results indicate that the terms related to the coupling deformation in the inertia forces of dynamic equations have small effect on system dynamic behavior and may be neglected, whereas the terms related to coupling deformation in the elastic forces are important for system dynamic behavior and should be considered in dynamic equation. Numerical examples of the rotating beam and flexible beam system are carried out to demonstrate the accuracy and validity of this dynamic model. Furthermore, it is shown that a small number of finite elements are needed to obtain a stable solution using the present coupling finite element formulation
FINITE ELEMENT ANALYSIS OF DEEP BEAM UNDER DIRECT AND INDIRECT LOAD
Directory of Open Access Journals (Sweden)
Haleem K. Hussain
2018-05-01
Full Text Available This research study the effect of exist of opening in web of deep beam loaded directly and indirectly and the behavior of reinforced concrete deep beams without with and without web reinforcement, the opening size and shear span ratio (a/d was constant. Nonlinear analysis using the finite element method with ANSYS software release 12.0 program was used to predict the ultimate load capacity and crack propagation for reinforced concrete deep beams with openings. The adopted beam models depend on experimental test program of reinforced concrete deep beam with and without openings and the finite element analysis result showed a good agreement with small amount of deference in ultimate beam capacity with (ANSYS analysis and it was completely efficient to simulate the behavior of reinforced concrete deep beams. The mid-span deflection at ultimate applied load and inclined cracked were highly compatible with experimental results. The model with opening in the shear span shows a reduction in the load-carrying capacity of beam and adding the vertical stirrup has improve the capacity of ultimate beam load.
Energy Technology Data Exchange (ETDEWEB)
Kosunen, A
1999-08-01
In radiation therapy using electron and photon beams the dosimetry chain consists of several sequential phases starting by the realisation of the dose quantity in the Primary Standard Dosimetry Laboratory and ending to the calculation of the dose to a patient. A similar procedure can be described for the dosimetry of epithermal neutron beams in boron neutron capture therapy (BNCT). To achieve the required accuracy of the dose delivered to a patient the quality of all steps in the dosimetry procedure has to be considered. This work is focused on two items in the dosimetry chains: the determination of the dose in the reference conditions and the evaluation of the accuracy of dose calculation methods. The issues investigated and discussed in detail are: a)the calibration methods of plane parallel ionisation chambers used in electron beam dosimetry, (b) the specification of the critical dosimetric parameter i.e. the ratio of stopping powers for water to air, (S I ?){sup water} {sub air}, in photon beams, (c) the feasibility of the twin ionization chamber technique for dosimetry in epithermal neutron beams applied to BNCT and (d) the determination accuracy of the calculated dose distributions in phantoms in electron, photon, and epithermal neutron beams. The results demonstrate that up to a 3% improvement in the consistency of dose determinations in electron beams is achieved by the calibration of plane parallel ionisation chambers in high energy electron beams instead of calibrations in {sup 60}Co gamma beams. In photon beam dosimetry (S I ?){sup water} {sub air} can be determined with an accuracy of 0.2% using the percentage dose at the 10 cm depth, %dd(10), as a beam specifier. The use of %odd(10) requires the elimination of the electron contamination in the photon beam. By a twin ionisation chamber technique the gamma dose can be determined with uncertainty of 6% (1 standard deviation) and the total neutron dose with an uncertainty of 15 to 20% (1 standard deviation
International Nuclear Information System (INIS)
Kosunen, A.
1999-08-01
In radiation therapy using electron and photon beams the dosimetry chain consists of several sequential phases starting by the realisation of the dose quantity in the Primary Standard Dosimetry Laboratory and ending to the calculation of the dose to a patient. A similar procedure can be described for the dosimetry of epithermal neutron beams in boron neutron capture therapy (BNCT). To achieve the required accuracy of the dose delivered to a patient the quality of all steps in the dosimetry procedure has to be considered. This work is focused on two items in the dosimetry chains: the determination of the dose in the reference conditions and the evaluation of the accuracy of dose calculation methods. The issues investigated and discussed in detail are: a)the calibration methods of plane parallel ionisation chambers used in electron beam dosimetry, (b) the specification of the critical dosimetric parameter i.e. the ratio of stopping powers for water to air, (S I ?) water air , in photon beams, (c) the feasibility of the twin ionization chamber technique for dosimetry in epithermal neutron beams applied to BNCT and (d) the determination accuracy of the calculated dose distributions in phantoms in electron, photon, and epithermal neutron beams. The results demonstrate that up to a 3% improvement in the consistency of dose determinations in electron beams is achieved by the calibration of plane parallel ionisation chambers in high energy electron beams instead of calibrations in 60 Co gamma beams. In photon beam dosimetry (S I ?) water air can be determined with an accuracy of 0.2% using the percentage dose at the 10 cm depth, %dd(10), as a beam specifier. The use of %odd(10) requires the elimination of the electron contamination in the photon beam. By a twin ionisation chamber technique the gamma dose can be determined with uncertainty of 6% (1 standard deviation) and the total neutron dose with an uncertainty of 15 to 20% (1 standard deviation). To improve the accuracy
An EXAFS spectrometer on beam line 10B at the Photon Factory
International Nuclear Information System (INIS)
Oyanagi, Hiroyuki; Matsushita, Tadashi; Ito, Masahisa; Kuroda, Haruo.
1984-03-01
An EXAFS spectrometer installed on the beam line 10B at the Photon Factory is designed to cover the photon energy between 4 and 30 keV. Utilizing either a channel-cut or two flat silicon crystals as a monochromator, a beam intensity between 10 8 and 10 9 photons/sec is obtained at 9 keV with a resolution of 1 eV. The performance of the spectrometer, such as a signal-to-noise ratio or an energy resolution is demonstrated with examples of K edge absorption spectra of bromine, germanium, gallium arsenide, and zinc selenide. (author)
Photon beam asymmetry Σ for η and η′ photoproduction from the proton
P. Collins; B.G. Ritchie; M. Dugger; A.V. Anisovich; M. Döring; E. Klempt; V.A. Nikonov; D. Rönchen; D. Sadasivan; A. Sarantsev; K.P. Adhikari; Z. Akbar; M.J. Amaryan; S. Anefalos Pereira; H. Avakian
2017-01-01
Measurements of the linearly-polarized photon beam asymmetry $\\Sigma$ for photoproduction from the proton of $\\eta$ and $\\eta^\\prime$ mesons are reported. A linearly-polarized tagged photon beam produced by coherent bremsstrahlung was incident on a cryogenic hydrogen target within the CEBAF Large Acceptance Spectrometer. Results are presented for the $\\gamma p \\to \\eta p$ reaction for incident photon energies from 1.070 to 1.876 GeV, and from 1.516 to 1.836 GeV for the $\\gamma p \\to \\eta^\\pri...
Directory of Open Access Journals (Sweden)
M.A. Najafgholipour
Full Text Available Abstract Reinforced concrete (RC beam-column connections especially those without transverse reinforcement in joint region can exhibit brittle behavior when intensive damage is concentrated in the joint region during an earthquake event. Brittle behavior in the joint region can compromise the ductile design philosophy and the expected overall performance of structure when subjected to seismic loading. Considering the importance of joint shear failure influences on strength, ductility and stability of RC moment resisting frames, a finite element modeling which focuses on joint shear behavior is presented in this article. Nonlinear finite element analysis (FEA of RC beam-column connections is performed in order to investigate the joint shear failure mode in terms of joint shear capacity, deformations and cracking pattern. A 3D finite element model capable of appropriately modeling the concrete stress-strain behavior, tensile cracking and compressive damage of concrete and indirect modeling of steel-concrete bond is used. In order to define nonlinear behavior of concrete material, the concrete damage plasticity is applied to the numerical model as a distributed plasticity over the whole geometry. Finite element model is then verified against experimental results of two non-ductile beam-column connections (one exterior and one interior which are vulnerable to joint shear failure. The comparison between experimental and numerical results indicates that the FE model is able to simulate the performance of the beam-column connections and is able to capture the joint shear failure in RC beam-column connections.
Compact high-efficiency vortex beam emitter based on a silicon photonics micro-ring
DEFF Research Database (Denmark)
Li, Shimao; Ding, Yunhong; Guan, Xiaowei
2018-01-01
Photonic integrated devices that emit vortex beam carrying orbital angular momentum are becoming key components for multiple applications. Here we propose and demonstrate a high-efficiency vortex beam emitter based on a silicon micro-ring resonator integrated with a metal mirror. Such a compact...
SU-F-T-428: An Optimization-Based Commissioning Tool for Finite Size Pencil Beam Dose Calculations
Energy Technology Data Exchange (ETDEWEB)
Li, Y; Tian, Z; Song, T; Jia, X; Gu, X; Jiang, S [UT Southwestern Medical Center, Dallas, TX (United States)
2016-06-15
Purpose: Finite size pencil beam (FSPB) algorithms are commonly used to pre-calculate the beamlet dose distribution for IMRT treatment planning. FSPB commissioning, which usually requires fine tuning of the FSPB kernel parameters, is crucial to the dose calculation accuracy and hence the plan quality. Yet due to the large number of beamlets, FSPB commissioning could be very tedious. This abstract reports an optimization-based FSPB commissioning tool we have developed in MatLab to facilitate the commissioning. Methods: A FSPB dose kernel generally contains two types of parameters: the profile parameters determining the dose kernel shape, and a 2D scaling factors accounting for the longitudinal and off-axis corrections. The former were fitted using the penumbra of a reference broad beam’s dose profile with Levenberg-Marquardt algorithm. Since the dose distribution of a broad beam is simply a linear superposition of the dose kernel of each beamlet calculated with the fitted profile parameters and scaled using the scaling factors, these factors could be determined by solving an optimization problem which minimizes the discrepancies between the calculated dose of broad beams and the reference dose. Results: We have commissioned a FSPB algorithm for three linac photon beams (6MV, 15MV and 6MVFFF). Dose of four field sizes (6*6cm2, 10*10cm2, 15*15cm2 and 20*20cm2) were calculated and compared with the reference dose exported from Eclipse TPS system. For depth dose curves, the differences are less than 1% of maximum dose after maximum dose depth for most cases. For lateral dose profiles, the differences are less than 2% of central dose at inner-beam regions. The differences of the output factors are within 1% for all the three beams. Conclusion: We have developed an optimization-based commissioning tool for FSPB algorithms to facilitate the commissioning, providing sufficient accuracy of beamlet dose calculation for IMRT optimization.
Thomson scattering of polarized photons in an intense laser beam
Energy Technology Data Exchange (ETDEWEB)
Byung Yunn
2006-02-21
We present a theoretical analysis of the Thomson scattering of linearly and circularly polarized photons from a pulsed laser by electrons. The analytical expression for the photon distribution functions presented in this paper should be useful to designers of Thomson scattering experiments.
Neutral-beam deposition in large, finite-beta noncircular tokamak plasmas
International Nuclear Information System (INIS)
Wieland, R.M.; Houlberg, W.A.
1982-02-01
A parametric pencil beam model is introduced for describing the attenuation of an energetic neutral beam moving through a tokamak plasma. The nonnegligible effects of a finite beam cross section and noncircular shifted plasma cross sections are accounted for in a simple way by using a smoothing algorithm dependent linearly on beam radius and by including information on the plasma flux surface geometry explicitly. The model is benchmarked against more complete and more time-consuming two-dimensional Monte Carlo calculations for the case of a large D-shaped tokamak plasma with minor radius a = 120 cm and elongation b/a = 1.6. Deposition profiles are compared for deuterium beam energies of 120 to 150 keV, central plasma densities of 8 x 10 13 - 2 x 10 14 cm -3 , and beam orientation ranging from perpendicular to tangential to the inside wall
Three-dimensional simulation of diamagnetic cavity formation by a finite-sized plasma beam
International Nuclear Information System (INIS)
Thomas, V.A.
1989-01-01
The problem of collisionless coupling between a plasma beam and a background plasma is examined using a three-dimensional hybrid code. The beam is assumed to be moving parallel to an ambient magnetic field at a speed greater than the local Alfven speed. In addition, the beam has a finite spatial extent in the directions perpendicular to the magnetic field and is uniform and infinite in the direction parallel to the ambient magnetic field. Such a system is susceptible to coupling of the beam ions with the background ions via an electromagnetic ion beam instability. This instability isotropizes the beam and energizes the background plasma. A large-amplitude Alfven wave traveling radially away from the interaction region is associated with the energized background plasma. The process described here is one which may be responsible for the formation of diamagnetic cavities observed in the solar wind. copyright American Geophysical Union 1989
Modeling silicon diode energy response factors for use in therapeutic photon beams.
Eklund, Karin; Ahnesjö, Anders
2009-10-21
Silicon diodes have good spatial resolution, which makes them advantageous over ionization chambers for dosimetry in fields with high dose gradients. However, silicon diodes overrespond to low-energy photons, that are more abundant in scatter which increase with large fields and larger depths. We present a cavity-theory-based model for a general response function for silicon detectors at arbitrary positions within photon fields. The model uses photon and electron spectra calculated from fluence pencil kernels. The incident photons are treated according to their energy through a bipartition of the primary beam photon spectrum into low- and high-energy components. Primary electrons from the high-energy component are treated according to Spencer-Attix cavity theory. Low-energy primary photons together with all scattered photons are treated according to large cavity theory supplemented with an energy-dependent factor K(E) to compensate for energy variations in the electron equilibrium. The depth variation of the response for an unshielded silicon detector has been calculated for 5 x 5 cm(2), 10 x 10 cm(2) and 20 x 20 cm(2) fields in 6 and 15 MV beams and compared with measurements showing that our model calculates response factors with deviations less than 0.6%. An alternative method is also proposed, where we show that one can use a correlation with the scatter factor to determine the detector response of silicon diodes with an error of less than 3% in 6 MV and 15 MV photon beams.
Modeling silicon diode energy response factors for use in therapeutic photon beams
International Nuclear Information System (INIS)
Eklund, Karin; Ahnesjoe, Anders
2009-01-01
Silicon diodes have good spatial resolution, which makes them advantageous over ionization chambers for dosimetry in fields with high dose gradients. However, silicon diodes overrespond to low-energy photons, that are more abundant in scatter which increase with large fields and larger depths. We present a cavity-theory-based model for a general response function for silicon detectors at arbitrary positions within photon fields. The model uses photon and electron spectra calculated from fluence pencil kernels. The incident photons are treated according to their energy through a bipartition of the primary beam photon spectrum into low- and high-energy components. Primary electrons from the high-energy component are treated according to Spencer-Attix cavity theory. Low-energy primary photons together with all scattered photons are treated according to large cavity theory supplemented with an energy-dependent factor K(E) to compensate for energy variations in the electron equilibrium. The depth variation of the response for an unshielded silicon detector has been calculated for 5 x 5 cm 2 , 10 x 10 cm 2 and 20 x 20 cm 2 fields in 6 and 15 MV beams and compared with measurements showing that our model calculates response factors with deviations less than 0.6%. An alternative method is also proposed, where we show that one can use a correlation with the scatter factor to determine the detector response of silicon diodes with an error of less than 3% in 6 MV and 15 MV photon beams.
Absorbed-dose beam quality conversion factors for cylindrical chambers in high energy photon beams.
Seuntjens, J P; Ross, C K; Shortt, K R; Rogers, D W
2000-12-01
Recent working groups of the AAPM [Almond et al., Med. Phys. 26, 1847 (1999)] and the IAEA (Andreo et al., Draft V.7 of "An International Code of Practice for Dosimetry based on Standards of Absorbed Dose to Water," IAEA, 2000) have described guidelines to base reference dosimetry of high energy photon beams on absorbed dose to water standards. In these protocols use is made of the absorbed-dose beam quality conversion factor, kQ which scales an absorbed-dose calibration factor at the reference quality 60Co to a quality Q, and which is calculated based on state-of-the-art ion chamber theory and data. In this paper we present the measurement and analysis of beam quality conversion factors kQ for cylindrical chambers in high-energy photon beams. At least three chambers of six different types were calibrated against the Canadian primary standard for absorbed dose based on a sealed water calorimeter at 60Co [TPR10(20)=0.572, %dd(10)x=58.4], 10 MV [TPR10(20)=0.682, %dd(10)x=69.6), 20 MV (TPR10(20)=0.758, %dd(10)x= 80.5] and 30 MV [TPR10(20) = 0.794, %dd(10)x= 88.4]. The uncertainty on the calorimetric determination of kQ for a single chamber is typically 0.36% and the overall 1sigma uncertainty on a set of chambers of the same type is typically 0.45%. The maximum deviation between a measured kQ and the TG-51 protocol value is 0.8%. The overall rms deviation between measurement and the TG-51 values, based on 20 chambers at the three energies, is 0.41%. When the effect of a 1 mm PMMA waterproofing sleeve is taken into account in the calculations, the maximum deviation is 1.1% and the overall rms deviation between measurement and calculation 0.48%. When the beam is specified using TPR10(20), and measurements are compared with kQ values calculated using the version of TG-21 with corrected formalism and data, differences are up to 1.6% when no sleeve corrections are taken into account. For the NE2571 and the NE2611A chamber types, for which the most literature data are
Study of dose distribution in high energy photon beam used in radiotherapy
International Nuclear Information System (INIS)
Rafaravavy, R.; Raoelina Andriambololona; Bridier, A.
2007-01-01
The dose distribution in a medium traversed by a photon beam depends on beam energy, field size and medium nature. Percent depth dose (PDD), Dose Profile (DP) and Opening Collimator Factor (OCF) curves will be established to study this distribution. So, the PDD curves are composed by tree parts: the build-up region, the maximal dose and the quasi-equilibrium region. The maximum dose depth and the dose in depth increase with increasing photon beam energy but the dose surface decreases. The PDD increases with increasing field size.
Photonic guiding structures in lithium niobate crystals produced by energetic ion beams
Chen, Feng
2009-10-01
A range of ion beam techniques have been used to fabricate a variety of photonic guiding structures in the well-known lithium niobate (LiNbO3 or LN) crystals that are of great importance in integrated photonics/optics. This paper reviews the up-to-date research progress of ion-beam-processed LiNbO3 photonic structures and reports on their fabrication, characterization, and applications. Ion beams are being used with this material in a wide range of techniques, as exemplified by the following examples. Ion beam milling/etching can remove the selected surface regions of LiNbO3 crystals via the sputtering effects. Ion implantation and swift ion irradiation can form optical waveguide structures by modifying the surface refractive indices of the LiNbO3 wafers. Crystal ion slicing has been used to obtain bulk-quality LiNbO3 single-crystalline thin films or membranes by exfoliating the implanted layer from the original substrate. Focused ion beams can either generate small structures of micron or submicron dimensions, to realize photonic bandgap crystals in LiNbO3, or directly write surface waveguides or other guiding devices in the crystal. Ion beam-enhanced etching has been extensively applied for micro- or nanostructuring of LiNbO3 surfaces. Methods developed to fabricate a range of photonic guiding structures in LiNbO3 are introduced. Modifications of LiNbO3 through the use of various energetic ion beams, including changes in refractive index and properties related to the photonic guiding structures as well as to the materials (i.e., electro-optic, nonlinear optic, luminescent, and photorefractive features), are overviewed in detail. The application of these LiNbO3 photonic guiding structures in both micro- and nanophotonics are briefly summarized.
The Effect of Aquaplast on Surface Dose of Photon Beam
International Nuclear Information System (INIS)
Oh, Do Hoon; Bae, Hoon Sik
1995-01-01
Purpose : To evaluate the effect on surface dose due to Aquaplast used for immobilizing the patients with head and neck cancers in photon beam radiotherapy. Materials and Methods : To assess surface and buildup region dose for 6MV X-ray from linear accelerator(Siemens Mevatron 6740), we measured percent ionization value with the Markus chamber model 30-329 manufactured by PTW Frieburg and Capintec electrometer, model WK92. For measurement of surface ionization value, the chamber was embedded in 25 X 25 X 3 cm 3 acrylic phantom and set on 25 X 25 X 5 cm 3 , polystyrene phantom to allow adequate scattering. The measurements of percent depth ionization were made by placing the polystyrene layers of appropriate thickness over the chamber. The measurements were taken at 10 cm SSD for 5 X 5 cm 2 , 10 X 10 cm 2 , and 15 X 15 cm 2 field sizes, respectively. Placing the layer of Aquaplast over the chamber, the same procedures were repeated. We evaluated two types o Aquaplast: 1.6mm layer of original Aquaplast(manufactured by WFR Aquaplast Corp.) and transformed Aquaplast similar to moulded one for immobilizing the patients practically. We also measured surface ionization values with blocking tray in presence or absence of transformed Aquaplast. In calculating percent depth dose, we used the formula suggested by Gerbi and khan to correct over response of the Markus chamber. Results : The surface doses for open fields of 5 X 5 cm 2 , 10 X 10 cm 2 , 15 X 15 cm 2 were 7.9%, 13.6%, and 18.7% respectively. He original Aquaplast increased the surface doses upto 38.4%, 43.6% and 47.4% respectively. There were little differences in percent depth dose values beyond the depth of Dmax. Increasing field size, the blocking tray caused increase of the surface dose by 0.2%, 1.7%, 3.0% without Aquaplast, 0.2%, 1.9%, 3.7% with transformed Aquaplast, respectively. Conclusion : The original and transformed Aquaplast increased the surface dose moderately. The percent depth doses beyond Dmax
Energy Technology Data Exchange (ETDEWEB)
Steel, Jared; Stewart, Allan; Satory, Philip [Auckland Regional Blood and Cancer Service, Auckland City Hospital, 2 Park Road, Grafton, Auckland 1023 (New Zealand)
2009-09-15
Purpose: Matching the penumbra of a 6 MeV electron beam to the penumbra of a 6 MV photon beam is a dose optimization challenge, especially when the electron beam is applied from an extended source-to-surface distance (SSD), as in the case of some head and neck treatments. Traditionally low melting point alloy blocks have been used to define the photon beam shielding over the spinal cord region. However, these are inherently time consuming to construct and employ in the clinical situation. Multileaf collimators (MLCs) provide a fast and reproducible shielding option but generate geometrically nonconformal approximations to the desired beam edge definition. The effects of substituting Cerrobend for the MLC shielding mode in the context of beam matching with extended-SSD electron beams are the subject of this investigation. Methods: Relative dose beam data from a Varian EX 2100 linear accelerator were acquired in a water tank under the 6 MeV electron beam at both standard and extended-SSD and under the 6 MV photon beam defined by Cerrobend and a number of MLC stepping regimes. The effect of increasing the electron beam SSD on the beam penumbra was assessed. MLC stepping was also assessed in terms of the effects on both the mean photon beam penumbra and the intraleaf dose-profile nonuniformity relative to the MLC midleaf. Computational techniques were used to combine the beam data so as to simulate composite relative dosimetry in the water tank, allowing fine control of beam abutment gap variation. Idealized volumetric dosimetry was generated based on the percentage depth-dose data for the beam modes and the abutment geometries involved. Comparison was made between each composite dosimetry dataset and the relevant ideal dosimetry dataset by way of subtraction. Results: Weighted dose-difference volume histograms (DDVHs) were produced, and these, in turn, summed to provide an overall dosimetry score for each abutment and shielding type/angle combination. Increasing the
International Nuclear Information System (INIS)
Steel, Jared; Stewart, Allan; Satory, Philip
2009-01-01
Purpose: Matching the penumbra of a 6 MeV electron beam to the penumbra of a 6 MV photon beam is a dose optimization challenge, especially when the electron beam is applied from an extended source-to-surface distance (SSD), as in the case of some head and neck treatments. Traditionally low melting point alloy blocks have been used to define the photon beam shielding over the spinal cord region. However, these are inherently time consuming to construct and employ in the clinical situation. Multileaf collimators (MLCs) provide a fast and reproducible shielding option but generate geometrically nonconformal approximations to the desired beam edge definition. The effects of substituting Cerrobend for the MLC shielding mode in the context of beam matching with extended-SSD electron beams are the subject of this investigation. Methods: Relative dose beam data from a Varian EX 2100 linear accelerator were acquired in a water tank under the 6 MeV electron beam at both standard and extended-SSD and under the 6 MV photon beam defined by Cerrobend and a number of MLC stepping regimes. The effect of increasing the electron beam SSD on the beam penumbra was assessed. MLC stepping was also assessed in terms of the effects on both the mean photon beam penumbra and the intraleaf dose-profile nonuniformity relative to the MLC midleaf. Computational techniques were used to combine the beam data so as to simulate composite relative dosimetry in the water tank, allowing fine control of beam abutment gap variation. Idealized volumetric dosimetry was generated based on the percentage depth-dose data for the beam modes and the abutment geometries involved. Comparison was made between each composite dosimetry dataset and the relevant ideal dosimetry dataset by way of subtraction. Results: Weighted dose-difference volume histograms (DDVHs) were produced, and these, in turn, summed to provide an overall dosimetry score for each abutment and shielding type/angle combination. Increasing the
Steel, Jared; Stewart, Allan; Satory, Philip
2009-09-01
Matching the penumbra of a 6 MeV electron beam to the penumbra of a 6 MV photon beam is a dose optimization challenge, especially when the electron beam is applied from an extended source-to-surface distance (SSD), as in the case of some head and neck treatments. Traditionally low melting point alloy blocks have been used to define the photon beam shielding over the spinal cord region. However, these are inherently time consuming to construct and employ in the clinical situation. Multileaf collimators (MLCs) provide a fast and reproducible shielding option but generate geometrically nonconformal approximations to the desired beam edge definition. The effects of substituting Cerrobend for the MLC shielding mode in the context of beam matching with extended-SSD electron beams are the subject of this investigation. Relative dose beam data from a Varian EX 2100 linear accelerator were acquired in a water tank under the 6 MeV electron beam at both standard and extended-SSD and under the 6 MV photon beam defined by Cerrobend and a number of MLC stepping regimes. The effect of increasing the electron beam SSD on the beam penumbra was assessed. MLC stepping was also assessed in terms of the effects on both the mean photon beam penumbra and the intraleaf dose-profile nonuniformity relative to the MLC midleaf. Computational techniques were used to combine the beam data so as to simulate composite relative dosimetry in the water tank, allowing fine control of beam abutment gap variation. Idealized volumetric dosimetry was generated based on the percentage depth-dose data for the beam modes and the abutment geometries involved. Comparison was made between each composite dosimetry dataset and the relevant ideal dosimetry dataset by way of subtraction. Weighted dose-difference volume histograms (DDVHs) were produced, and these, in turn, summed to provide an overall dosimetry score for each abutment and shielding type/angle combination. Increasing the electron beam SSD increased
Derivation of electron and photon energy spectra from electron beam central axis depth dose curves
Energy Technology Data Exchange (ETDEWEB)
Deng Jun [Department of Radiation Oncology, Stanford University School of Medicine, Stanford, CA 94305 (United States)]. E-mail: jun@reyes.stanford.edu; Jiang, Steve B.; Pawlicki, Todd; Li Jinsheng; Ma, C.M. [Department of Radiation Oncology, Stanford University School of Medicine, Stanford, CA 94305 (United States)
2001-05-01
A method for deriving the electron and photon energy spectra from electron beam central axis percentage depth dose (PDD) curves has been investigated. The PDD curves of 6, 12 and 20 MeV electron beams obtained from the Monte Carlo full phase space simulations of the Varian linear accelerator treatment head have been used to test the method. We have employed a 'random creep' algorithm to determine the energy spectra of electrons and photons in a clinical electron beam. The fitted electron and photon energy spectra have been compared with the corresponding spectra obtained from the Monte Carlo full phase space simulations. Our fitted energy spectra are in good agreement with the Monte Carlo simulated spectra in terms of peak location, peak width, amplitude and smoothness of the spectrum. In addition, the derived depth dose curves of head-generated photons agree well in both shape and amplitude with those calculated using the full phase space data. The central axis depth dose curves and dose profiles at various depths have been compared using an automated electron beam commissioning procedure. The comparison has demonstrated that our method is capable of deriving the energy spectra for the Varian accelerator electron beams investigated. We have implemented this method in the electron beam commissioning procedure for Monte Carlo electron beam dose calculations. (author)
DEFF Research Database (Denmark)
Kim, Taeseong; Hansen, Anders Melchior; Branner, Kim
2013-01-01
In this paper a new anisotropic beam finite element for composite wind turbine blades is developed and implemented into the aeroelastic nonlinear multibody code, HAWC2, intended to be used to investigate if use of anisotropic material layups in wind turbine blades can be tailored for improved...
Ion-acoustic wave propagation in plasmas with ion beams having a finite cross section--
International Nuclear Information System (INIS)
Huld, T.A.; Pe'cseli, H.L.; Rasmussen, J.J.
1990-01-01
The propagation of a low-density-modulated ion beam with finite cross section in a homogeneous plasma is considered. Analytical expressions describing a Cerenkov-like radiation pattern are obtained. An experimental setup is described that is suitable for investigating these phenomena. The results are in qualitive agreement with the analytical expressions
The shock formation distance in a bounded sound beam of finite amplitude.
Tao, Chao; Ma, Jian; Zhu, Zhemin; Du, Gonghuan; Ping, Zihong
2003-07-01
This paper investigates the shock formation distance in a bounded sound beam of finite amplitude by solving the Khokhlov-Zabolotskaya-Kuznetsov (KZK) equation using frequency-domain numerical method. Simulation results reveal that, besides the nonlinearity and absorption, the diffraction is another important factor that affects the shock formation of a bounded sound beam. More detailed discussions of the shock formation in a bounded sound beam, such as the waveform of sound pressure and the spatial distribution of shock formation, are also presented and compared for different parameters.
Dang, Zhiya; Breese, Mark Bh; Recio-Sánchez, Gonzalo; Azimi, Sara; Song, Jiao; Liang, Haidong; Banas, Agnieszka; Torres-Costa, Vicente; Martín-Palma, Raúl José
2012-07-23
A method for fabrication of three-dimensional (3D) silicon nanostructures based on selective formation of porous silicon using ion beam irradiation of bulk p-type silicon followed by electrochemical etching is shown. It opens a route towards the fabrication of two-dimensional (2D) and 3D silicon-based photonic crystals with high flexibility and industrial compatibility. In this work, we present the fabrication of 2D photonic lattice and photonic slab structures and propose a process for the fabrication of 3D woodpile photonic crystals based on this approach. Simulated results of photonic band structures for the fabricated 2D photonic crystals show the presence of TE or TM gap in mid-infrared range.
Single photon emission computed tomography by using fan beam collimator
International Nuclear Information System (INIS)
Akiyama, Yoshihisa
1992-01-01
A multislice fan beam collimator which has parallel collimation along the cephalic-caudul axis of a patient and converging collimation within planes that are perpendicular to that axis was designed for a SPECT system with a rotating scintillation camera, and it was constructed by the lead casting method which was developed in recent years. A reconstruction algorithm for fan beam SPECT was formed originally by combining the reconstruction algorithm of the parallel beam SPECT with that of the fan beam X-ray CT. The algorithm for fan beam SPECT was confirmed by means of computer simulation and a head phantom filled with diluted radionuclide. Not only 99m Tc but also 123 I was used as a radionuclide. A SPECT image with the fan beam collimator was compared with that of a parallel hole, low energy, high resolution collimator which was routinely used for clinical and research SPECT studies. Both system resolution and sensitivity of the fan beam collimator were ∼20% better than those of the parallel hole collimator. Comparing SPECT images obtained from fan beam collimator with those of parallel hole collimator, the SPECT images using fan beam collimator had far better resolution. A fan beam collimator is a useful implement for the SPECT study. (author)
Pimpinella, Maria; Caporali, Claudio; Guerra, Antonio Stefano; Silvi, Luca; De Coste, Vanessa; Petrucci, Assunta; Delaunay, Frank; Dufreneix, Stéphane; Gouriou, Jean; Ostrowsky, Aimé; Rapp, Benjamin; Bordy, Jean-Marc; Daures, Josiane; Le Roy, Maïwenn; Sommier, Line; Vermesse, Didier
2018-01-01
To investigate the feasibility of using the ratio of dose-area product at 20 cm and 10 cm water depths (DAPR 20,10 ) as a beam quality specifier for radiotherapy photon beams with field diameter below 2 cm. Dose-area product was determined as the integral of absorbed dose to water (D w ) over a surface larger than the beam size. 6 MV and 10 MV photon beams with field diameters from 0.75 cm to 2 cm were considered. Monte Carlo (MC) simulations were performed to calculate energy-dependent dosimetric parameters and to study the DAPR 20,10 properties. Aspects relevant to DAPR 20,10 measurement were explored using large-area plane-parallel ionization chambers with different diameters. DAPR 20,10 was nearly independent of field size in line with the small differences among the corresponding mean beam energies. Both MC and experimental results showed a dependence of DAPR 20,10 on the measurement setup and the surface over which D w is integrated. For a given setup, DAPR 20,10 values obtained using ionization chambers with different air-cavity diameters agreed with one another within 0.4%, after the application of MC correction factors accounting for effects due to the chamber size. DAPR 20,10 differences among the small field sizes were within 1% and sensitivity to the beam energy resulted similar to that of established beam quality specifiers based on the point measurement of D w . For a specific measurement setup and integration area, DAPR 20,10 proved suitable to specify the beam quality of small photon beams for the selection of energy-dependent dosimetric parameters. Copyright © 2017 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.
Modelling of electron contamination in clinical photon beams for Monte Carlo dose calculation
International Nuclear Information System (INIS)
Yang, J; Li, J S; Qin, L; Xiong, W; Ma, C-M
2004-01-01
The purpose of this work is to model electron contamination in clinical photon beams and to commission the source model using measured data for Monte Carlo treatment planning. In this work, a planar source is used to represent the contaminant electrons at a plane above the upper jaws. The source size depends on the dimensions of the field size at the isocentre. The energy spectra of the contaminant electrons are predetermined using Monte Carlo simulations for photon beams from different clinical accelerators. A 'random creep' method is employed to derive the weight of the electron contamination source by matching Monte Carlo calculated monoenergetic photon and electron percent depth-dose (PDD) curves with measured PDD curves. We have integrated this electron contamination source into a previously developed multiple source model and validated the model for photon beams from Siemens PRIMUS accelerators. The EGS4 based Monte Carlo user code BEAM and MCSIM were used for linac head simulation and dose calculation. The Monte Carlo calculated dose distributions were compared with measured data. Our results showed good agreement (less than 2% or 2 mm) for 6, 10 and 18 MV photon beams
Dynamic analysis of an axially moving beam subject to inner pressure using finite element method
Energy Technology Data Exchange (ETDEWEB)
Hua, Hongliang; Qiu, Ming; Liao, Zhenqiang [Nanjing University of Science and Technology, Nanjing (China)
2017-06-15
A dynamic model of an axially moving flexible beam subject to an inner pressure is present. The coupling principle between a flexible beam and inner pressure is analyzed first, and the potential energy of the inner pressure due to the beam bending is derived using the principle of virtual work. A 1D hollow beam element contain inner pressure is established. The finite element method and Lagrange’s equation are used to derive the motion equations of the axially moving system. The dynamic responses are analyzed by Newmark-β time integration method. Based on the computed dynamic responses, the effects of inner pressure on beam dynamics are discussed. Some interesting phenomenon is observed.
Conversion of optical wave polarizations in 1D finite anisotropic photonic crystal
International Nuclear Information System (INIS)
Ouchani, N.; Nougaoui, N.; Daoudi, A.; Bria, D.
2006-07-01
We show that by using one dimensional anisotropic photonic structures, it is possible to realize optical wave polarization conversion by transmission or by reflection. Thus a single incident S(P) polarized plane wave can produce a single reflected P(S) polarized wave and a single transmitted P(S) polarized wave. This polarization conversion property can be fulfilled with a simple finite superlattice constituted by anisotropic dielectric materials. We discuss the appropriate choices of the material and geometrical properties to realize such structures. The transmission and reflection coefficients are discussed in relation with the dispersion curves of the finite structure embedded between two isotropic substrates. Both transmission and reflection coefficients are calculated in the framework of Green's function method. The amplitude and the polarization characteristics of reflected and transmitted waves are determined as function of frequency ω , and wave vector k parallel ( parallel to the interface) and the orientations of the principal axes of the layers constituting the SL. Moreover, this structure exhibits a coupling between S and P waves that does not exist in SL composed only of isotropic materials. Specific applications of these results are given for a superlattice consisting of alternating biaxial anisotropic layers NaNO 2 /SbSi sandwiched between two identical semi-infinite isotropic media. (author)
Assessment of radiotherapy photon beams: A practical and low cost methodology
International Nuclear Information System (INIS)
Reis, C.Q.M.; Nicolucci, P.
2017-01-01
Dosimetric properties of radiation beams used in radiotherapy are directly related to the energy spectrum produced by the treatment unit. Therefore, the development of methodologies to evaluate in a simple and accurate way the spectra of clinical beams can help establishing the quality control of the treatment. The purpose of this study is to present a practical and low cost methodology for determining primary spectra of radiotherapy photon beams from transmission measurements in attenuators of aluminum and using the method of the inverse Laplace transform. Monte Carlo simulation with PENELOPE code was used in order to evaluate and validate the reconstructed spectra by the calculation of dosimetric parameters that characterize the beam. Percentage depth dose values simulated with a 6 MV reconstructed spectrum shows maximum difference of 4.4% when compared to values measured at the corresponding clinical beam. For a 10 MV beam that difference was around 4.2%. Results obtained in this study confirm the adequacy of the proposed methodology for assessing primary photon beams produced by clinical accelerators. - Highlights: • Primary spectra of radiotherapy photon beams are determined from transmission measurements. • Monte Carlo calculations are used to evaluate the method of the inverse Laplace transform. • The proposed methodology is practical and of low cost for clinical purposes. • Results are in fair agreement with literature and clinical data.
Local density of optical states in the band gap of a finite one-dimensional photonic crystal
Yeganegi Dastgerdi, Elahe; Lagendijk, Aart; Mosk, Allard; Vos, Willem L.
2014-01-01
We study the local density of states (LDOS) in a finite photonic crystal, in particular in the frequency range of the band gap. We propose an original point of view on the band gap, which we consider to be the result of vacuum fluctuations in free space that tunnel in the forbidden range in the
Asena, A; Kairn, T; Crowe, S B; Trapp, J V
2015-05-01
This study investigates the effects of temporary tissue expanders (TTEs) on the dose distributions in breast cancer radiotherapy treatments under a variety of conditions. Using EBT2 radiochromic film, both electron and photon beam dose distribution measurements were made for different phantoms, and beam geometries. This was done to establish a more comprehensive understanding of the implant's perturbation effects under a wider variety of conditions. The magnetic disk present in a tissue expander causes a dose reduction of approximately 20% in a photon tangent treatment and 56% in electron boost fields immediately downstream of the implant. The effects of the silicon elastomer are also much more apparent in an electron beam than a photon beam. Evidently, each component of the TTE attenuates the radiation beam to different degrees. This study has demonstrated that the accuracy of photon and electron treatments of post-mastectomy patients is influenced by the presence of a tissue expander for various beam orientations. The impact of TTEs on dose distributions establishes the importance of an accurately modelled high-density implant in the treatment planning system for post-mastectomy patients. Copyright © 2015 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.
Surface dose measurements and comparison of unflattened and flattened photon beams
Directory of Open Access Journals (Sweden)
Ashokkumar Sigamani
2016-01-01
Full Text Available The purpose of this study was to evaluate the central axis dose in the build-up region and the surface dose of a 6 MV and 10 MV flattened photon beam (FB and flattening filter free (FFF therapeutic photon beam for different square field sizes (FSs for a Varian Truebeam linear accelerator using parallel-plate ionization chamber and Gafchromic film. Knowledge of dosimetric characteristics in the build-up region and surface dose of the FFF is essential for clinical care. The dose measurements were also obtained empirically using two different commonly used dosimeters: a p-type photon semiconductor dosimeter and a cylindrical ionization chamber. Surface dose increased linearly with FS for both FB and FFF photon beams. The surface dose values of FFF were higher than the FB FSs. The measured surface dose clearly increases with increasing FS. The FFF beams have a modestly higher surface dose in the build-up region than the FB. The dependence of source to skin distance (SSD is less significant in FFF beams when compared to the flattened beams at extended SSDs.
Prediction of the behavior of reinforced concrete deep beams with web openings using the finite ele
Directory of Open Access Journals (Sweden)
Ashraf Ragab Mohamed
2014-06-01
Full Text Available The exact analysis of reinforced concrete deep beams is a complex problem and the presence of web openings aggravates the situation. However, no code provision exists for the analysis of deep beams with web opening. The code implemented strut and tie models are debatable and no unique solution using these models is available. In this study, the finite element method is utilized to study the behavior of reinforced concrete deep beams with and without web openings. Furthermore, the effect of the reinforcement distribution on the beam overall capacity has been studied and compared to the Egyptian code guidelines. The damaged plasticity model has been used for the analysis. Models of simply supported deep beams under 3 and 4-point bending and continuous deep beams with and without web openings have been analyzed. Model verification has shown good agreement to literature experimental work. Results of the parametric analysis have shown that web openings crossing the expected compression struts should be avoided, and the depth of the opening should not exceed 20% of the beam overall depth. The reinforcement distribution should be in the range of 0.1–0.2 beam depth for simply supported deep beams.
The comparison of Co-60 and 4MV photons matching dosimetry during half-beam technique
International Nuclear Information System (INIS)
Cakir, Aydin; Bilge, Hatice; Dadasbilge, Alpar; Kuecuecuek, Halil; Okutan, Murat; Merdan Fayda, Emre
2005-01-01
In this phantom study, we tried to compare matching dosimetry differences between half-blocking of Co-60 and asymmetric collimation of the 4MV photons during craniospinal irradiation. The dose distributions are compared and discussed. Firstly, some gaps with different sizes are left between cranial and spinal field borders. Secondly, the fields are overlapped in the same sizes. We irradiate the films located in water-equivalent solid phantoms with Co-60 and 4MV photon beams. This study indicates that the field placement errors in +/- 1mm are acceptable for both Co-60 and 4MV photon energies during craniospinal irradiation with half-beam block technique. Within these limits the dose variations are specified in +/- 5%. However, the setup errors that are more than 1mm are unacceptable for both asymmetric collimation of 4MV photon and half-blocking of Co-60
Directory of Open Access Journals (Sweden)
E. Carrera
2011-01-01
Full Text Available This paper presents hierarchical finite elements on the basis of the Carrera Unified Formulation for free vibrations analysis of beam with arbitrary section geometries. The displacement components are expanded in terms of the section coordinates, (x, y, using a set of 1-D generalized displacement variables. N-order Taylor type expansions are employed. N is a free parameter of the formulation, it is supposed to be as high as 4. Linear (2 nodes, quadratic (3 nodes and cubic (4 nodes approximations along the beam axis, (z, are introduced to develop finite element matrices. These are obtained in terms of a few fundamental nuclei whose form is independent of both N and the number of element nodes. Natural frequencies and vibration modes are computed. Convergence and assessment with available results is first made considering different type of beam elements and expansion orders. Additional analyses consider different beam sections (square, annular and airfoil shaped as well as boundary conditions (simply supported and cantilever beams. It has mainly been concluded that the proposed model is capable of detecting 3-D effects on the vibration modes as well as predicting shell-type vibration modes in case of thin walled beam sections.
Gamma beams generation with high intensity lasers for two photon Breit-Wheeler pair production
D'Humieres, Emmanuel; Ribeyre, Xavier; Jansen, Oliver; Esnault, Leo; Jequier, Sophie; Dubois, Jean-Luc; Hulin, Sebastien; Tikhonchuk, Vladimir; Arefiev, Alex; Toncian, Toma; Sentoku, Yasuhiko
2017-10-01
Linear Breit-Wheeler pair creation is the lowest threshold process in photon-photon interaction, controlling the energy release in Gamma Ray Bursts and Active Galactic Nuclei, but it has never been directly observed in the laboratory. Using numerical simulations, we demonstrate the possibility to produce collimated gamma beams with high energy conversion efficiency using high intensity lasers and innovative targets. When two of these beams collide at particular angles, our analytical calculations demonstrate a beaming effect easing the detection of the pairs in the laboratory. This effect has been confirmed in photon collision simulations using a recently developed innovative algorithm. An alternative scheme using Bremsstrahlung radiation produced by next generation high repetition rate laser systems is also being explored and the results of first optimization campaigns in this regime will be presented.
A closed-loop photon beam control study for the Advanced Light Source
International Nuclear Information System (INIS)
Portmann, G.; Bengtsson, J.
1993-05-01
The third generation Advanced Light Source (ALS) will produce extremely bright photon beams using undulators and wigglers. In order to position the photon beams accurate to the micron level, a closed-loop feedback system is being developed. Using photon position monitors and dipole corrector magnets, a closed-loop system can automatically compensate for modeling uncertainties and exogenous disturbances. The following paper will present a dynamics model for the perturbations of the closed orbit of the electron beam in the ALS storage ring including the vacuum chamber magnetic field penetration effects. Using this reference model, two closed-loop feedback algorithms will be compared -- a classical PI controller and a two degree-of-freedom approach. The two degree-of-freedom method provides superior disturbance rejection while maintaining the desired performance goals. Both methods will address the need to gain schedule the controller due to the time varying dynamics introduced by changing field strengths when scanning the insertion devices
Benchmarking of Touschek Beam Lifetime Calculations for the Advanced Photon Source
Energy Technology Data Exchange (ETDEWEB)
Xiao, A.; Yang, B.
2017-06-25
Particle loss from Touschek scattering is one of the most significant issues faced by present and future synchrotron light source storage rings. For example, the predicted, Touschek-dominated beam lifetime for the Advanced Photon Source (APS) Upgrade lattice in 48-bunch, 200-mA timing mode is only ~ 2 h. In order to understand the reliability of the predicted lifetime, a series of measurements with various beam parameters was performed on the present APS storage ring. This paper first describes the entire process of beam lifetime measurement, then compares measured lifetime with the calculated one by applying the measured beam parameters. The results show very good agreement.
Chen, Chin-Hui; Klamkin, Jonathan; Nicholes, Steven C; Johansson, Leif A; Bowers, John E; Coldren, Larry A
2009-09-01
We present an extensive study of an ultracompact grating-based beam splitter suitable for photonic integrated circuits (PICs) that have stringent density requirements. The 10 microm long beam splitter exhibits equal splitting, low insertion loss, and also provides a high extinction ratio in an integrated coherent balanced receiver. We further present the design strategies for avoiding mode distortion in the beam splitter and discuss optimization of the widths of the detectors to improve insertion loss and extinction ratio of the coherent receiver circuit. In our study, we show that the grating-based beam splitter is a competitive technology having low fabrication complexity for ultracompact PICs.
Attenuation of photon beams from radionuclides. Pt. 6
International Nuclear Information System (INIS)
Dorner, R.; Vogt, H.G.
1982-04-01
Point kernel shielding calculations have been performed for water for the photons of radioactive nuclides, comprising gamma rays, characteristic X-rays, annihilation photons as well as internal and external bremsstrahlung. The calculations refer to an isotropic point source and an infinite homogeneous medium. The results are presented for 137 radionuclides by tables of dose rate constants and sets of figures of the reciprocal attenuation factor S -1 . The shielding curves cover the range of 10 -8 -1 <1 and water thicknesses up to 340 cm. (orig./HP)
International Nuclear Information System (INIS)
Liu Lizhong; Prasad, Satish C.; Bassano, Daniel A.
2003-01-01
Two commercially available water-equivalent solid phantom materials were evaluated for output calibration in both photon (6-15 MV) and electron (6-20 MeV) beams. The solid water 457 and virtual water materials have the same chemical composition but differ in manufacturing process and density. A Farmer-type ionization chamber was used for measuring the output of the photon beams at 5- and 10-cm depth and electron beams at maximum buildup depth in the solid phantoms and in natural water. The water-equivalency correction factor for the solid materials is defined as the ratio of the chamber reading in natural water to that in the solid at the same linear depth. For photon beams, the correction factor was found to be independent of depth and was 0.987 and 0.993 for 6- and 15-MV beams, respectively, for solid water. For virtual water, the corresponding correction factors were 0.993 and 0.998 for 6- and 15-MV beams, respectively. For electron beams, the correction factors ranged from 1.013 to 1.007 for energies of 6 to 20 MeV for both solid materials. This indicated that the water-equivalency of these materials is within ± 1.3%, making them suitable substitutes for natural water in both photon and electron beam output measurements over a wide energy range. These correction factors are slightly larger than the manufacturers' advertised values (± 1.0% for solid water and ± 0.5% for virtual water). We suggest that these corrections are large enough in most cases and should be applied in the calculation of beam outputs
Monte Carlo simulation of Varian Linac for 6 MV photon beam with BEAMnrc code
Mohammed, Maged; El Bardouni, T.; Chakir, E.; Boukhal, H.; Saeed, M.; Ahmed, Abdul-Aziz
2018-03-01
The purpose of this study is to investigate the effects of the initial electron beam parameters on the absorbed dose distribution calculated with EGSnrc Monte Carlo code, for 6 MV photon beam. A proposed methodology for benchmarking the BEAMnrc model of Varian Linac has been used. Also, a new photon cross section data based on ENDF/B-VII release 8 evaluation has been employed. The parameters tested include mean energy, radial intensity distribution and angular spread of the initial electron beam. Mean energy and angular spread were tested for a square irradiation field 10 × 10 cm2, whereas beam width of the electron beam was studied for 10 × 10 cm2 at different depths and 30 × 30 cm2 at depth of 10 cm. The results obtained are compared with measurement data to select the optimal electron beam parameters. The differences between MC calculation and measurements data are analyzed using gamma index criteria which fixed within 1% -1 mm accuracy. The obtained results indicated that the depth-dose and dose-profile curves were considerably influenced by the mean energy of the electron beam. The depth-dose curves were unaffected by the beam width of the electron beam, for both irradiation fields. On the contrary, lateral dose-profile curves were affected by the beam width of initial electron beam. Both dose-profile and depth-dose curves were unaffected to the angular spread of the electron beam. A deep depth of 10 × 10 cm2 is very accurate to tune the beam width. Mean energy and beam width must be tuned precisely, to get the MC does distribution with acceptable accuracy.
Determining clinical photon beam spectra from measured depth dose with the Cimmino algorithm
International Nuclear Information System (INIS)
Bloch, P.; Altschuler, M.D.; Bjaerngard, B.E.; Kassaee, A.; McDonough, J.
2000-01-01
A method to determine the spectrum of a clinical photon beam from measured depth-dose data is described. At shallow depths, where the range of Compton-generated electrons increases rapidly with photon energy, the depth dose provides the information to discriminate the spectral contributions. To minimize the influence of contaminating electrons, small (6x6cm2 ) fields were used. The measured depth dose is represented as a linear combination of basis functions, namely the depth doses of monoenergetic photon beams derived by Monte Carlo simulations. The weights of the basis functions were obtained with the Cimmino feasibility algorithm, which examines in each iteration the discrepancy between predicted and measured depth dose. For 6 and 15 MV photon beams of a clinical accelerator, the depth dose obtained from the derived spectral weights was within about 1% of the measured depth dose at all depths. Because the problem is ill conditioned, solutions for the spectrum can fluctuate with energy. Physically realistic smooth spectra for these photon beams appeared when a small margin (about ±1%) was attributed to the measured depth dose. The maximum energy of both derived spectra agreed with the measured energy of the electrons striking the target to within 1 MeV. The use of a feasibility method on minimally relaxed constraints provides realistic spectra quickly and interactively. (author)
Monte Carlo simulation of spectrum changes in a photon beam due to a brass compensator
Energy Technology Data Exchange (ETDEWEB)
Custidiano, E.R., E-mail: ernesto7661@gmail.com [Department of Physics, FaCENA, UNNE, Av., Libertad 5470, C.P.3400, Corrientes (Argentina); Valenzuela, M.R., E-mail: meraqval@gmail.com [Department of Physics, FaCENA, UNNE, Av., Libertad 5470, C.P.3400, Corrientes (Argentina); Dumont, J.L., E-mail: Joseluis.Dumont@elekta.com [Elekta CMS Software, St.Louis, MO (United States); McDonnell, J., E-mail: josemc@express.com.ar [Cumbres Institute, Riobamba 1745, C.P.2000, Rosario, Santa Fe (Argentina); Rene, L, E-mail: luismrene@gmail.com [Radiotherapy Center, Crespo 953, C.P.2000, Rosario, Santa Fe (Argentina); Rodriguez Aguirre, J.M., E-mail: juakcho@gmail.com [Department of Physics, FaCENA, UNNE, Av., Libertad 5470, C.P.3400, Corrientes (Argentina)
2011-06-15
Monte Carlo simulations were used to study the changes in the incident spectrum when a poly-energetic photon beam passes through a static brass compensator. The simulated photon beam spectrum was evaluated by comparing it against the incident spectra. We also discriminated the changes in the transmitted spectrum produced by each of the microscopic processes. (i.e. Rayleigh scattering, photoelectric effect, Compton scattering, and pair production). The results show that the relevant process in the energy range considered is the Compton Effect, as expected for composite materials of intermediate atomic number and energy range considered.
Monte Carlo simulation of spectrum changes in a photon beam due to a brass compensator
International Nuclear Information System (INIS)
Custidiano, E.R.; Valenzuela, M.R.; Dumont, J.L.; McDonnell, J.; Rene, L; Rodriguez Aguirre, J.M.
2011-01-01
Monte Carlo simulations were used to study the changes in the incident spectrum when a poly-energetic photon beam passes through a static brass compensator. The simulated photon beam spectrum was evaluated by comparing it against the incident spectra. We also discriminated the changes in the transmitted spectrum produced by each of the microscopic processes. (i.e. Rayleigh scattering, photoelectric effect, Compton scattering, and pair production). The results show that the relevant process in the energy range considered is the Compton Effect, as expected for composite materials of intermediate atomic number and energy range considered.
International Nuclear Information System (INIS)
Garcia-Garduno, Olivia A.; Larraga-Gutierrez, Jose M.; Rodriguez-Villafuerte, Mercedes; Martinez-Davalos, Arnulfo; Celis, Miguel A.
2010-01-01
This work reports the use of both GafChromic EBT film immersed in a water phantom and Monte Carlo (MC) simulations for small photon beam stereotactic radiosurgery dosimetry. Circularly collimated photon beams with diameters in the 4-20 mm range of a dedicated 6 MV linear accelerator (Novalis (registered) , BrainLAB, Germany) were used to perform off-axis ratios, tissue maximum ratios and total scatter factors measurements, and MC simulations. GafChromic EBT film data show an excellent agreement with MC results (<2.7%) for all measured quantities.
Prototype photon position monitors for undulator beams at the advanced light source
International Nuclear Information System (INIS)
Warwick, T.; Shu, D.; Rodricks, B.; Johnson, E.D.
1992-01-01
Design criteria are described, and test results are presented, for prototype ALS undulator beam position monitors. The design is based on monitors presently in use at NSLS, with modifications to account for the widely varying and large K values of the undulators to be installed at the ALS. In particular, we have modified the design to simplify the thermal engineering and we have explored techniques to suppress the response of the monitors to soft photons, so that the beam position can be determined by measuring the higher energy photons which are better collimated
Application of focused ion beam technology for photonic nanostructures
Ay, F.; Gadgil, V.J.; Geskus, D.; Aravazhi, S.; Worhoff, Kerstin; Pollnau, Markus
Al2O3 and KY(WO4)2 are promising materials for photonic applications with excellent optical properties and of interest for obtaining on chip resonator structures. However, there is no method available to fabricate these structures except FIB technology. We will discuss strategies to optimize the
Hidden photons in beam dump experiments and in connection with dark matter
Energy Technology Data Exchange (ETDEWEB)
Andreas, Sarah
2012-12-15
Hidden sectors with light extra U(1) gauge bosons, so-called hidden photons, recently received much interest as natural feature of beyond standard model scenarios like string theory and SUSY and because of their possible connection to dark matter. This paper presents limits on hidden photons from past electron beam dump experiments including two new limits from experiments at KEK and Orsay. Additionally, various hidden sector models containing both a hidden photon and a dark matter candidate are discussed with respect to their viability and potential signatures in direct detection.
Propagation of photons induced by a proton beam in a quartz bar
International Nuclear Information System (INIS)
Plaszczynski, S.; Torassa, E.; Wormser, G.; Aleksan, R.; Amadon, A.; Besson, P.; Bourgeois, P.; Monchenault, G.H.; Jarry, P.; Lemaire, M.C.; Lemeur, J.; London, G.; Robert, J.P.
1996-01-01
For the BABAR experiment, an amplitude and timing analysis of photons produced by a proton beam in a quartz bar has been performed. The photoelectron yield is obtained for different angles and positions. It is found that the linear speed of propagation of the Cherenkov photons through the bar is well described y internal reflections. The timing analysis reveals also the existence of an unexpected parasite light component, not described by the simulation. However, all the results can be reproduced by adding an isotropic scintillation of about 4 photons/cm in the quartz bar on the trajectory of the primary proton, in 50 % of the cases. (author)
International Nuclear Information System (INIS)
Kandic, A.; Jevremovic, T.; Boreli, F.
1989-01-01
Monte Carlo simulation (without secondary radiation) of the standard photon interactions (Compton scattering, photoelectric absorption and pair protection) for the complex slab's geometry is used in numerical code ACCA. A typical ACCA run will yield: (a) transmission of primary photon radiation differential in energy, (b) the spectrum of energy deposited in the target as a function of position and (c) the cumulative percent energy deposition as a function of position. A cumulative percent energy deposition of photon monoenergetic beam incident on simplest and complexity tissue slab and Fe slab are presented in this paper. (author). 5 refs.; 2 figs
Hidden photons in beam dump experiments and in connection with dark matter
International Nuclear Information System (INIS)
Andreas, Sarah
2012-12-01
Hidden sectors with light extra U(1) gauge bosons, so-called hidden photons, recently received much interest as natural feature of beyond standard model scenarios like string theory and SUSY and because of their possible connection to dark matter. This paper presents limits on hidden photons from past electron beam dump experiments including two new limits from experiments at KEK and Orsay. Additionally, various hidden sector models containing both a hidden photon and a dark matter candidate are discussed with respect to their viability and potential signatures in direct detection.
Status of positron beams for dark photons experiments
Directory of Open Access Journals (Sweden)
Valente Paolo
2017-01-01
Full Text Available High energy positron beams are an important tool for fixed-target experiments searching for new particles produced in the annihilation on atomic electrons of a target. The status of existing or planned infrastructures is reviewed.
Sources of electron contamination for the Clinac-35 25-MV photon beam
International Nuclear Information System (INIS)
Petti, P.L.; Goodman, M.S.; Sisterson, J.M.; Biggs, P.J.; Gabriel, T.A.; Mohan, R.
1983-01-01
A detailed Monte Carlo approach has been employed to investigate the sources of electron contamination for the 25-MV photon beam generated by Varian's Clinac-35. Three sources of contamination were examined: (a) the flattening filter and beam monitor chamber, (b) the fixed primary collimators downstream from the monitor chamber and the adjustable photon jaws, and (c) the air volume separating the treatment head from the observation point. Five source-to-surface distances (SSDs) were considered for a single field size, 28 cm in diameter at 80 cm SSD. It was found that for small SSDs (80-100 cm), the dominant sources of electron contamination were the flattening filter and the beam monitor chamber which accounted for 70% of the unwanted electrons. Thirteen percent of the remaining electrons originated in the downstream primary collimators and the photon jaws, and 17% were produced in air. At larger SSDs, the fraction of unwanted electrons originating in air increased. At 400 cm SSD, 61% of the contaminating electrons present in the beam were produced in air, 34% originated in the flattening filter and beam monitor chamber, and 5% were due to interactions in the fixed collimators downstream from the monitor chamber and the adjustable photon jaws. These calculated results are substantiated by recent experiments
Finite element beam flexural properties of cement composites of fiber reinforced PVA
Yang, Chengzhi; Pei, Changchun
2018-05-01
In this paper, the initial cracking state and the mid span bending moment and deflection of ECC beam under different PVA fiber and fly ash mixing rate are studied by finite element simulation analysis. The results show that the bending moment of the ECC beam increases with the increase of the PVA fiber content, and the deflection decreases. When the ratio of PVA fiber is 1.5%, the middle bending moment is the largest and the deflection is the least. With the increase of fly ash content, the mid span bending moment of ECC beam increases first and then decreases. When the fly ash ratio is 60%, the middle bending moment is the largest and the deflection is the least. Through the study, the formula for calculating the flexural capacity of the cross section suitable for ECC beams is derived.
Directory of Open Access Journals (Sweden)
Husain M. Husain
2013-05-01
Full Text Available In this work a program is developed to carry out the nonlinear analysis (material nonlinearity of prestressed concrete beams using tendons of carbon fiber reinforced polymer (CFRP instead of steel. The properties of this material include high strength, light weight, and insusceptibility to corrosion and magnetism. This material is still under investigation, therefore it needs continuous work to make it beneficial in concrete design. Four beams which are tested experimentally by Yan et al. are examined by the developed computer program to reach a certain analytical approach of the design and analysis of such beams because there is no available restrictions or recommendations covering this material in the codes. The program uses the finite element analysis by dividing the beams into isoparametric 20-noded brick elements. The results obtained are good in comparison with experimental results.
The development of a curved beam element model applied to finite elements method
International Nuclear Information System (INIS)
Bento Filho, A.
1980-01-01
A procedure for the evaluation of the stiffness matrix for a thick curved beam element is developed, by means of the minimum potential energy principle, applied to finite elements. The displacement field is prescribed through polynomial expansions, and the interpolation model is determined by comparison of results obtained by the use of a sample of different expansions. As a limiting case of the curved beam, three cases of straight beams, with different dimensional ratios are analised, employing the approach proposed. Finally, an interpolation model is proposed and applied to a curved beam with great curvature. Desplacements and internal stresses are determined and the results are compared with those found in the literature. (Author) [pt
Type I parametric down conversion of highly focused Gaussian beams in finite length crystals
International Nuclear Information System (INIS)
Jeronimo-Moreno, Yasser; Jáuregui, R
2014-01-01
This paper presents a study of the correlations in wave vector space of photon pairs generated by type I spontaneous parametric down conversion using a Gaussian pump beam. The analysis covers both moderate focused and highly focused regimes, paying special attention to the angular spectrum and the conditional angular spectrum. Simple analytic expressions are derived that allow a detailed study of the dependence of these spectra on the waist of the source and the length of the nonlinear crystal. These expressions are in good agreement with numerical expectations and reported experimental results. They are used to make a systematic search of optimization parameters that improve the feasibility of using highly focused Gaussian beams to generate idler and signal photons with predetermined mean values and spread of their transverse wave vectors. (papers)
Polarised Photon Beams for the BGO-OD Experiment at ELSA
Zimmermann, T.; Bella, A.; Alef, S.; Bayadilov, D.; Beck, R.; Becker, M.; Bielefeldt, P.; Boese, S.; Braghieri, A.; Brinkmann, K.; Cole, P.; Curciarello, F.; De Leo, V.; Di Salvo, R.; Dutz, H.; Elsner, D.; Fantini, A.; Freyermuth, O.; Friedrich, S.; Frommberger, F.; Ganenko, V.; Gervino, G.; Ghio, F.; Giardina, G.; Goertz, S.; Gridnev, A.; Gutz, E.; Hammann, D.; Hannappel, J.; Hartmann, P.; Hillert, W.; Ignatov, A.; Jahn, R.; Joosten, R.; Jude, T. C.; Klein, F.; Koop, K.; Krusche, B.; Lapik, A.; Levi Sandri, P.; Lopatin, I. V.; Mandaglio, G.; Messi, F.; Messi, R.; Metag, V.; Moricciani, D.; Mushkarenkov, A.; Nanova, M.; Nedorezov, V.; Novinskiy, D.; Pedroni, P.; Reitz, B.; Romaniuk, M.; Rostomyan, T.; Rudnev, N.; Schaerf, C.; Scheluchin, G.; Schmieden, H.; Stugelev, A.; Sumachev, V.; Tarakanov, V.; Vegna, V.; Walther, D.; Watts, D.; Zaunick, H.
The new BGO-OD experiment at the electron accelerator ELSA, of the University of Bonn, is designed to study the reaction dynamics of nucleon excitations in meson photoproduction. It consists of a central BGO calorimeter with a magnetic spectrometer in forward direction. The physics programme includes the measurement of polarisation observables using linearly and circularly polarised photon beams. Linear polarisation is obtained by coherent bremsstrahlung off a diamond crystal, and circular polarisation is obtained via bremsstrahlung from longitudinally polarised electrons. The degree of linear polarisation is determined from the bremsstrahlung spectrum itself. To determine the polarisation of the circularly polarised photon beam, the polarisation of the electron beam is measured by a Møller polarimeter. As a preliminary consistency check, the (linear) polarisation observable, Σ, was compared to world data for π0 and η photoproduction. To determine the degree of circular polarisation, a Møller polarimeter was setup and first measurements of the electron beam polarisation performed.
Cascaded two-photon spectroscopy of Yb atoms with a transportable effusive atomic beam apparatus
Energy Technology Data Exchange (ETDEWEB)
Song, Minsoo; Yoon, Tai Hyun [Department of Physics, Korea University, Anam-dong, Seongbuk-gu, Seoul 136-713 (Korea, Republic of)
2013-02-15
We present a transportable effusive atomic beam apparatus for cascaded two-photon spectroscopy of the dipole-forbidden transition (6s{sup 2} {sup 1}S{sub 0}{r_reversible} 6s7s {sup 1}S{sub 0}) of Yb atoms. An ohmic-heating effusive oven is designed to have a reservoir volume of 1.6 cm{sup 3} and a high degree of atomic beam collimation angle of 30 mrad. The new atomic beam apparatus allows us to detect the spontaneously cascaded two-photons from the 6s7s{sup 1}S{sub 0} state via the intercombination 6s6p{sup 3}P{sub 1} state with a high signal-to-noise ratio even at the temperature of 340 Degree-Sign C. This is made possible in our apparatus because of the enhanced atomic beam flux and superior detection solid angle.
Cascaded two-photon spectroscopy of Yb atoms with a transportable effusive atomic beam apparatus
International Nuclear Information System (INIS)
Song, Minsoo; Yoon, Tai Hyun
2013-01-01
We present a transportable effusive atomic beam apparatus for cascaded two-photon spectroscopy of the dipole-forbidden transition (6s 2 1 S 0 ↔ 6s7s 1 S 0 ) of Yb atoms. An ohmic-heating effusive oven is designed to have a reservoir volume of 1.6 cm 3 and a high degree of atomic beam collimation angle of 30 mrad. The new atomic beam apparatus allows us to detect the spontaneously cascaded two-photons from the 6s7s 1 S 0 state via the intercombination 6s6p 3 P 1 state with a high signal-to-noise ratio even at the temperature of 340 °C. This is made possible in our apparatus because of the enhanced atomic beam flux and superior detection solid angle.
Commissioning of a medical accelerator photon beam Monte Carlo simulation using wide-field profiles
International Nuclear Information System (INIS)
Pena, J; Franco, L; Gomez, F; Iglesias, A; Lobato, R; Mosquera, J; Pazos, A; Pardo, J; Pombar, M; RodrIguez, A; Sendon, J
2004-01-01
A method for commissioning an EGSnrc Monte Carlo simulation of medical linac photon beams through wide-field lateral profiles at moderate depth in a water phantom is presented. Although depth-dose profiles are commonly used for nominal energy determination, our study shows that they are quite insensitive to energy changes below 0.3 MeV (0.6 MeV) for a 6 MV (15 MV) photon beam. Also, the depth-dose profile dependence on beam radius adds an additional uncertainty in their use for tuning nominal energy. Simulated 40 cm x 40 cm lateral profiles at 5 cm depth in a water phantom show greater sensitivity to both nominal energy and radius. Beam parameters could be determined by comparing only these curves with measured data
Commissioning of a medical accelerator photon beam Monte Carlo simulation using wide-field profiles
Energy Technology Data Exchange (ETDEWEB)
Pena, J [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Franco, L [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Gomez, F [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Iglesias, A [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Lobato, R [Hospital ClInico Universitario de Santiago, Santiago de Compostela (Spain); Mosquera, J [Hospital ClInico Universitario de Santiago, Santiago de Compostela (Spain); Pazos, A [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Pardo, J [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Pombar, M [Hospital ClInico Universitario de Santiago, Santiago de Compostela (Spain); RodrIguez, A [Departamento de Fisica de PartIculas, Facultade de Fisica, 15782 Santiago de Compostela (Spain); Sendon, J [Hospital ClInico Universitario de Santiago, Santiago de Compostela (Spain)
2004-11-07
A method for commissioning an EGSnrc Monte Carlo simulation of medical linac photon beams through wide-field lateral profiles at moderate depth in a water phantom is presented. Although depth-dose profiles are commonly used for nominal energy determination, our study shows that they are quite insensitive to energy changes below 0.3 MeV (0.6 MeV) for a 6 MV (15 MV) photon beam. Also, the depth-dose profile dependence on beam radius adds an additional uncertainty in their use for tuning nominal energy. Simulated 40 cm x 40 cm lateral profiles at 5 cm depth in a water phantom show greater sensitivity to both nominal energy and radius. Beam parameters could be determined by comparing only these curves with measured data.
International Nuclear Information System (INIS)
Fogliata, A.; Garcia, R.; Knöös, T.; Nicolini, G.; Clivio, A.; Vanetti, E.; Khamphan, C.; Cozzi, L.
2012-01-01
Purpose: Flattening filter free (FFF) beams generated by medical linear accelerators have recently started to be used in radiotherapy clinical practice. Such beams present fundamental differences with respect to the standard filter flattened (FF) beams, making the generally used dosimetric parameters and definitions not always viable. The present study will propose possible definitions and suggestions for some dosimetric parameters for use in quality assurance of FFF beams generated by medical linacs in radiotherapy. Methods: The main characteristics of the photon beams have been analyzed using specific data generated by a Varian TrueBeam linac having both FFF and FF beams of 6 and 10 MV energy, respectively. Results: Definitions for dose profile parameters are suggested starting from the renormalization of the FFF with respect to the corresponding FF beam. From this point the flatness concept has been translated into one of “unflatness” and other definitions have been proposed, maintaining a strict parallelism between FFF and FF parameter concepts. Conclusions: Ideas for quality controls used in establishing a quality assurance program when introducing FFF beams into the clinical environment are given here, keeping them similar to those used for standard FF beams. By following the suggestions in this report, the authors foresee that the introduction of FFF beams into a clinical radiotherapy environment will be as safe and well controlled as standard beam modalities using the existing guidelines.
Six-beam homodyne laser Doppler vibrometry based on silicon photonics technology.
Li, Yanlu; Zhu, Jinghao; Duperron, Matthieu; O'Brien, Peter; Schüler, Ralf; Aasmul, Soren; de Melis, Mirko; Kersemans, Mathias; Baets, Roel
2018-02-05
This paper describes an integrated six-beam homodyne laser Doppler vibrometry (LDV) system based on a silicon-on-insulator (SOI) full platform technology, with on-chip photo-diodes and phase modulators. Electronics and optics are also implemented around the integrated photonic circuit (PIC) to enable a simultaneous six-beam measurement. Measurement of a propagating guided elastic wave in an aluminum plate (speed ≈ 909 m/s @ 61.5 kHz) is demonstrated.
A finite element beam propagation method for simulation of liquid crystal devices.
Vanbrabant, Pieter J M; Beeckman, Jeroen; Neyts, Kristiaan; James, Richard; Fernandez, F Anibal
2009-06-22
An efficient full-vectorial finite element beam propagation method is presented that uses higher order vector elements to calculate the wide angle propagation of an optical field through inhomogeneous, anisotropic optical materials such as liquid crystals. The full dielectric permittivity tensor is considered in solving Maxwell's equations. The wide applicability of the method is illustrated with different examples: the propagation of a laser beam in a uniaxial medium, the tunability of a directional coupler based on liquid crystals and the near-field diffraction of a plane wave in a structure containing micrometer scale variations in the transverse refractive index, similar to the pixels of a spatial light modulator.
Monte Carlo simulation of MOSFET detectors for high-energy photon beams using the PENELOPE code
Panettieri, Vanessa; Amor Duch, Maria; Jornet, Núria; Ginjaume, Mercè; Carrasco, Pablo; Badal, Andreu; Ortega, Xavier; Ribas, Montserrat
2007-01-01
The aim of this work was the Monte Carlo (MC) simulation of the response of commercially available dosimeters based on metal oxide semiconductor field effect transistors (MOSFETs) for radiotherapeutic photon beams using the PENELOPE code. The studied Thomson&Nielsen TN-502-RD MOSFETs have a very small sensitive area of 0.04 mm2 and a thickness of 0.5 µm which is placed on a flat kapton base and covered by a rounded layer of black epoxy resin. The influence of different metallic and Plastic water™ build-up caps, together with the orientation of the detector have been investigated for the specific application of MOSFET detectors for entrance in vivo dosimetry. Additionally, the energy dependence of MOSFET detectors for different high-energy photon beams (with energy >1.25 MeV) has been calculated. Calculations were carried out for simulated 6 MV and 18 MV x-ray beams generated by a Varian Clinac 1800 linear accelerator, a Co-60 photon beam from a Theratron 780 unit, and monoenergetic photon beams ranging from 2 MeV to 10 MeV. The results of the validation of the simulated photon beams show that the average difference between MC results and reference data is negligible, within 0.3%. MC simulated results of the effect of the build-up caps on the MOSFET response are in good agreement with experimental measurements, within the uncertainties. In particular, for the 18 MV photon beam the response of the detectors under a tungsten cap is 48% higher than for a 2 cm Plastic water™ cap and approximately 26% higher when a brass cap is used. This effect is demonstrated to be caused by positron production in the build-up caps of higher atomic number. This work also shows that the MOSFET detectors produce a higher signal when their rounded side is facing the beam (up to 6%) and that there is a significant variation (up to 50%) in the response of the MOSFET for photon energies in the studied energy range. All the results have shown that the PENELOPE code system can
Monte Carlo simulation of MOSFET detectors for high-energy photon beams using the PENELOPE code.
Panettieri, Vanessa; Duch, Maria Amor; Jornet, Núria; Ginjaume, Mercè; Carrasco, Pablo; Badal, Andreu; Ortega, Xavier; Ribas, Montserrat
2007-01-07
The aim of this work was the Monte Carlo (MC) simulation of the response of commercially available dosimeters based on metal oxide semiconductor field effect transistors (MOSFETs) for radiotherapeutic photon beams using the PENELOPE code. The studied Thomson&Nielsen TN-502-RD MOSFETs have a very small sensitive area of 0.04 mm(2) and a thickness of 0.5 microm which is placed on a flat kapton base and covered by a rounded layer of black epoxy resin. The influence of different metallic and Plastic water build-up caps, together with the orientation of the detector have been investigated for the specific application of MOSFET detectors for entrance in vivo dosimetry. Additionally, the energy dependence of MOSFET detectors for different high-energy photon beams (with energy >1.25 MeV) has been calculated. Calculations were carried out for simulated 6 MV and 18 MV x-ray beams generated by a Varian Clinac 1800 linear accelerator, a Co-60 photon beam from a Theratron 780 unit, and monoenergetic photon beams ranging from 2 MeV to 10 MeV. The results of the validation of the simulated photon beams show that the average difference between MC results and reference data is negligible, within 0.3%. MC simulated results of the effect of the build-up caps on the MOSFET response are in good agreement with experimental measurements, within the uncertainties. In particular, for the 18 MV photon beam the response of the detectors under a tungsten cap is 48% higher than for a 2 cm Plastic water cap and approximately 26% higher when a brass cap is used. This effect is demonstrated to be caused by positron production in the build-up caps of higher atomic number. This work also shows that the MOSFET detectors produce a higher signal when their rounded side is facing the beam (up to 6%) and that there is a significant variation (up to 50%) in the response of the MOSFET for photon energies in the studied energy range. All the results have shown that the PENELOPE code system can successfully
International Nuclear Information System (INIS)
Johnson, M; McNabb, D
2006-01-01
An important applied technology is a tunable mono-energetic photon source [1]. These sources are made of relativistic electron accelerators coupled to low-energy lasers, which produce high-energy, mono-energetic-rays. One challenge associated with systems such as this is a continuum of bremsstrahlung background created when an electron beam passes through an aperture of some sort and the electron bunch or its halo impinges on the aperture pictured in figure 1. For instance, in the current T-REX [1] design for the interaction point between the laser- and electron-beam, the electron-beam passes through the center of a mirror used to reflect the laser. There is a potential with this design that bremsstrahlung radiation may be produced at the edges of the mirror openings and contaminate the mono-energetic photon beam. Certain applications [2] may be sensitive to this contamination. To reduce the bremsstrahlung contaminate a collimator (thickness ∼24in. (calculated from XCOM database [3]) to attenuate by a factor of 10 -3 the 112MeV photons expected in the T-REX demonstration [1]) is situated between the aperture and target. To maximize the brightness of the photon-beam, the collimator opening must be no less than the size of the photon-beam spot size expected to be about 1mm. This fixes the collimator opening. a priori the aperture size must be greater than the collimator opening and is a function distance between the aperture and collimator. In this paper we focus on two approaches to estimate the aperture size, given a collimator and a target whose sizes and distances from the aperture are given. In the next section we will discuss these approaches
Czech Academy of Sciences Publication Activity Database
Centini, M.; Peřina ml., Jan; Sciscione, L.; Sibilia, C.; Scalora, M.; Bloemer, M.J.; Bertolotti, M.
2005-01-01
Roč. 72, 03 (2005), 033806/1-033806/11 ISSN 1050-2947 R&D Projects: GA MŠk(CZ) OC P11.003 Institutional research plan: CEZ:AV0Z10100522 Keywords : photon pair * photonic crystals * spontaneous parametric down-conversion Subject RIV: BH - Optics, Masers, Lasers Impact factor: 2.997, year: 2005
Finite element modelling of FRC beams containing PVA and Basalt fibres: A comparative study
Ayub, Tehmina; Khan, Sadaqat Ullah
2017-09-01
The endeavour of current study is to compare the flexural behaviour and three dimensional (3D) finite element analysis (FEA) and the results of FEM are compared with the experimental results of 07 HPFRC beams. Out of seven (07), 01 beam of plain concrete without fibres was cast as a control beam. Three (03) beams containing 1, 2 and 3% volume of PVA fibres were prepared by using HPFRC mixes while, the remaining other three (03) beams were prepared using HPFRC mixes containing 1, 2 and 3% volume of Basalt fibres. In order to ensure flexural failure, three-point bending load was applied at the mid span of all beams. The maximum flexural load and corresponding deflection and strains at the mid span attained prior to the failure were obtained as flexural test results. The FEM results were obtained by simulating all beams in ATENA 3D program and verified through flexural test results. Both of the results of FEM and Experiment showed good agreement with each other.
Directory of Open Access Journals (Sweden)
Litesh N. Sulbhewar
Full Text Available Abstract The conventional Timoshenko piezoelectric beam finite elements based on First-order Shear Deformation Theory (FSDT do not maintain the accuracy and convergence consistently over the applicable range of material and geometric properties. In these elements, the inaccuracy arises due to the induced potential effects in the transverse direction and inefficiency arises due to the use of independently assumed linear polynomial interpolation of the field variables in the longitudinal direction. In this work, a novel FSDT-based piezoelectric beam finite element is proposed which is devoid of these deficiencies. A variational formulation with consistent through-thickness potential is developed. The governing equilibrium equations are used to derive the coupled field relations. These relations are used to develop a polynomial interpolation scheme which properly accommodates the bending-extension, bending-shear and induced potential couplings to produce accurate results in an efficient manner. It is noteworthy that this consistently accurate and efficient beam finite element uses the same nodal variables as of conventional FSDT formulations available in the literature. Comparison of numerical results proves the consistent accuracy and efficiency of the proposed formulation irrespective of geometric and material configurations, unlike the conventional formulations.
Giant nonlinear interaction between two optical beams via a quantum dot embedded in a photonic wire
Nguyen, H. A.; Grange, T.; Reznychenko, B.; Yeo, I.; de Assis, P.-L.; Tumanov, D.; Fratini, F.; Malik, N. S.; Dupuy, E.; Gregersen, N.; Auffèves, A.; Gérard, J.-M.; Claudon, J.; Poizat, J.-Ph.
2018-05-01
Optical nonlinearities usually appear for large intensities, but discrete transitions allow for giant nonlinearities operating at the single-photon level. This has been demonstrated in the last decade for a single optical mode with cold atomic gases, or single two-level systems coupled to light via a tailored photonic environment. Here, we demonstrate a two-mode giant nonlinearity with a single semiconductor quantum dot (QD) embedded in a photonic wire antenna. We exploit two detuned optical transitions associated with the exciton-biexciton QD level scheme. Owing to the broadband waveguide antenna, the two transitions are efficiently interfaced with two free-space laser beams. The reflection of one laser beam is then controlled by the other beam, with a threshold power as low as 10 photons per exciton lifetime (1.6 nW ). Such a two-color nonlinearity opens appealing perspectives for the realization of ultralow-power logical gates and optical quantum gates, and could also be implemented in an integrated photonic circuit based on planar waveguides.
International Nuclear Information System (INIS)
Woo, M.K.; Cunningham, J.R.
1990-01-01
In the convolution/superposition method of photon beam dose calculations, inhomogeneities are usually handled by using some form of scaling involving the relative electron densities of the inhomogeneities. In this paper the accuracy of density scaling as applied to primary electrons generated in photon interactions is examined. Monte Carlo calculations are compared with density scaling calculations for air and cork slab inhomogeneities. For individual primary photon kernels as well as for photon interactions restricted to a thin layer, the results can differ significantly, by up to 50%, between the two calculations. However, for realistic photon beams where interactions occur throughout the whole irradiated volume, the discrepancies are much less severe. The discrepancies for the kernel calculation are attributed to the scattering characteristics of the electrons and the consequent oversimplified modeling used in the density scaling method. A technique called the kernel integration technique is developed to analyze the general effects of air and cork inhomogeneities. It is shown that the discrepancies become significant only under rather extreme conditions, such as immediately beyond the surface after a large air gap. In electron beams all the primary electrons originate from the surface of the phantom and the errors caused by simple density scaling can be much more significant. Various aspects relating to the accuracy of density scaling for air and cork slab inhomogeneities are discussed
Experimental Investigation of Propagation and Reflection Phenomena in Finite Amplitude Sound Beams.
Averkiou, Michalakis Andrea
Measurements of finite amplitude sound beams are compared with theoretical predictions based on the KZK equation. Attention is devoted to harmonic generation and shock formation related to a variety of propagation and reflection phenomena. Both focused and unfocused piston sources were used in the experiments. The nominal source parameters are piston radii of 6-25 mm, frequencies of 1-5 MHz, and focal lengths of 10-20 cm. The research may be divided into two parts: propagation and reflection of continuous-wave focused sound beams, and propagation of pulsed sound beams. In the first part, measurements of propagation curves and beam patterns of focused pistons in water, both in the free field and following reflection from curved targets, are presented. The measurements are compared with predictions from a computer model that solves the KZK equation in the frequency domain. A novel method for using focused beams to measure target curvature is developed. In the second part, measurements of pulsed sound beams from plane pistons in both water and glycerin are presented. Very short pulses (less than 2 cycles), tone bursts (5-30 cycles), and frequency modulated (FM) pulses (10-30 cycles) were measured. Acoustic saturation of pulse propagation in water is investigated. Self-demodulation of tone bursts and FM pulses was measured in glycerin, both in the near and far fields, on and off axis. All pulse measurements are compared with numerical results from a computer code that solves the KZK equation in the time domain. A quasilinear analytical solution for the entire axial field of a self-demodulating pulse is derived in the limit of strong absorption. Taken as a whole, the measurements provide a broad data base for sound beams of finite amplitude. Overall, outstanding agreement is obtained between theory and experiment.
Photonic density of states in the vicinity of a single-wall finite-length carbon nanotube
International Nuclear Information System (INIS)
Nemilentsau, A; Ya Slepyan, G; Maksimenko, S A
2009-01-01
Photonic density of states in the vicinity of a single-wall finite-length carbon nanotube (CNT) is investigated theoretically in this paper. The analysis is based on the fluctuation-dissipative theorem in the Callen-Welton form. The Dyson equation for the Green dyadic of the electromagnetic field in the presence of CNT is formulated and a method for its numerical solution is elaborated. We show that the photonic density of states spectrum has a nontrivial resonant structure in the terahertz range in the vicinity of the metallic single-wall CNT. The origin of these resonances is the surface plasmon resonances on the CNT's edges.
Residual Strength Analysisof Asymmetrically Damaged Ship Hull GirderUsing Beam Finite Element Method
Directory of Open Access Journals (Sweden)
Muhammad Zubair Muis Alie
2016-04-01
Full Text Available The objective of the present study is to analyze the residual strength of asymmetrically damaged ship hull girder under longitudinal bending. Beam Finite Element Method isused for the assessment of the residual strength of two single hull bulk carriers (Ship B1 and Ship B4 and a three-cargo-hold model of a single-side Panamax Bulk Carrierin hogging and sagging conditions. The Smith’s method is adopted and implemented into Beam Finite Element Method. An efficient solution procedure is applied; i.e. by assuming the cross section remains plane, the vertical bending moment is applied to the cross section and three-cargo-hold model. As a fundamental case, the damage is simply created by removing the elements from the cross section, neglecting any welding residual stress and initial imperfection. Also no crack extension is considered. The result obtained by Beam Finite Element Method so-called Beam-HULLST is compared to the progressive collapse analysis obtained by HULLST for the validation of the present work. Then, for the three-hold-model, the Beam-HULLST is used to investigate the effect of the rotation of the netral axisboth intact and damage condition taking the one and five frame spaces into account.
Electromagnetic-field amplification in finite one-dimensional photonic crystals
International Nuclear Information System (INIS)
Gorelik, V. S.; Kapaev, V. V.
2016-01-01
The electromagnetic-field distribution in a finite one-dimensional photonic crystal is studied using the numerical solution of Maxwell’s equations by the transfer-matrix method. The dependence of the transmission coefficient T on the period d (or the wavelength λ) has the characteristic form with M–1 (M is the number of periods in the structure) maxima with T = 1 in the allowed band of an infinite crystal and zero values in the forbidden band. The field-modulus distribution E(x) in the structure for parameters that correspond to the transmission maxima closest to the boundaries of forbidden bands has maxima at the center of the structure; the value at the maximum considerably exceeds the incident-field strength. For the number of periods M ~ 50, more than an order of magnitude increase in the field amplification is observed. The numerical results are interpreted with an analytic theory constructed by representing the solution in the form of a linear combination of counterpropagating Floquet modes in a periodic structure.
International Nuclear Information System (INIS)
Elata, A.; Hassan, A. M. E.; Ali, E.; Calzolari, P.; Bettega, D.
2009-02-01
The main objective of this study was to perform a radiobiological characterization of different energy photon beams (6 MV and 15 MV) from linear accelerator used in radiotherapy, and comparison of different treatment modalities, with special regard to late effects of radiation. Using two end points, cell survival and micronucleus induction, in the biological system (Chines hamster V79 cell line). Chromosomes number was counted and found to be 22 chromosomes per cell. Cells were kept in confluent growth for two days and then exposed to two photon beams and immediately after irradiation were counted and re seeded in different numbered for each dose. For evaluation of surviving fraction samples were incubated at 37o C for 6 days, five samples were counted for each dose. At the same time three samples were seeded for the micronuclei frequency and incubated at 37o C after 24 hours cytochalasin-B was added to block cells in cytokinesis. The survival curve showed similar curves for the two beams and decreased with dose. The micronuclei frequency was positively correlated with dose and the energy of the photon. This indicates the presence of low dose of photoneutrons produced by using high energy photon beams. (Author)
Focused-ion-beam nano-structuring of photonic cavities in dielectric materials
Ay, F.; Pollnau, Markus
Focused ion beam (FIB) milling is an emerging technology that enables fast, reliable and well-controlled nanometer-size feature definition. In this work we will discuss applications of the tool in the area of photonics. The FIB technique can be adapted and optimized almost for any material system
Characteristic of EBT-XD and EBT3 radiochromic film dosimetry for photon and proton beams
Khachonkham, Suphalak; Dreindl, Ralf; Heilemann, Gerd; Lechner, Wolfgang; Fuchs, Hermann; Palmans, Hugo; Georg, Dietmar; Kuess, Peter
2018-03-01
Recently, a new type of radiochromic film, the EBT-XD film, has been introduced for high dose radiotherapy. The EBT-XD film contains the same structure as the EBT3 film but has a slightly different composition and a thinner active layer. This study benchmarks the EBT-XD against EBT3 film for 6 MV and 10 MV photon beams, as well as for 97.4 MeV and 148.2 MeV proton beams and 15-100 kV x-rays. Dosimetric and film reading characteristics, such as post irradiation darkening, film orientation effect, lateral response artifact (LRA), film sensitivity, energy and beam quality dependency were investigated. Furthermore, quenching effects in the Bragg peak were investigated for a single proton beam energy for both film types, in addition measurements were performed in a spread-out Bragg peak. EBT-XD films showed the same characteristic on film darkening as EBT3. The effects between portrait and landscape orientation were reduced by 3.1% (in pixel value) for EBT-XD compared to EBT3 at a dose of 2000 cGy. The LRA is reduced for EBT-XD films for all investigated dose ranges. The sensitivity of EBT-XD films is superior to EBT3 for doses higher than 500 cGy. In addition, EBT-XD showed a similar dosimetric response for photon and proton irradiation with low energy and beam quality dependency. A quenching effect of 10% was found for both film types. The slight decrease in the thickness of the active layer and different composition configuration of EBT-XD resulted in a reduced film orientation effect and LRA, as well as a sensitivity increase in high-dose regions for both photon and proton beams. Overall, the EBT-XD film improved regarding film reading characteristics and showed advantages in the high-dose region for photon and proton beams.
Characteristic of EBT-XD and EBT3 radiochromic film dosimetry for photon and proton beams.
Khachonkham, Suphalak; Dreindl, Ralf; Heilemann, Gerd; Lechner, Wolfgang; Fuchs, Hermann; Palmans, Hugo; Georg, Dietmar; Kuess, Peter
2018-03-15
Recently, a new type of radiochromic film, the EBT-XD film, has been introduced for high dose radiotherapy. The EBT-XD film contains the same structure as the EBT3 film but has a slightly different composition and a thinner active layer. This study benchmarks the EBT-XD against EBT3 film for 6 MV and 10 MV photon beams, as well as for 97.4 MeV and 148.2 MeV proton beams and 15-100 kV x-rays. Dosimetric and film reading characteristics, such as post irradiation darkening, film orientation effect, lateral response artifact (LRA), film sensitivity, energy and beam quality dependency were investigated. Furthermore, quenching effects in the Bragg peak were investigated for a single proton beam energy for both film types, in addition measurements were performed in a spread-out Bragg peak. EBT-XD films showed the same characteristic on film darkening as EBT3. The effects between portrait and landscape orientation were reduced by 3.1% (in pixel value) for EBT-XD compared to EBT3 at a dose of 2000 cGy. The LRA is reduced for EBT-XD films for all investigated dose ranges. The sensitivity of EBT-XD films is superior to EBT3 for doses higher than 500 cGy. In addition, EBT-XD showed a similar dosimetric response for photon and proton irradiation with low energy and beam quality dependency. A quenching effect of 10% was found for both film types. The slight decrease in the thickness of the active layer and different composition configuration of EBT-XD resulted in a reduced film orientation effect and LRA, as well as a sensitivity increase in high-dose regions for both photon and proton beams. Overall, the EBT-XD film improved regarding film reading characteristics and showed advantages in the high-dose region for photon and proton beams.
Measurement of electron beam polarization from unstrained GaAs via two-photon photoemission
Energy Technology Data Exchange (ETDEWEB)
McCarter, J.L., E-mail: jlm2ar@virginia.edu [Department of Physics, University of Virginia, Charlottesville, VA 22901 (United States); Afanasev, A. [Department of Physics, The George Washington University, Washington, DC 20052 (United States); Gay, T.J. [Jorgensen Hall, University of Nebraska, Lincoln, NE 68588 (United States); Hansknecht, J. [Thomas Jefferson National Accelerator Facility, 12050 Jefferson Avenue, Suite 500, Newport News, VA 23606 (United States); Kechiantz, A. [Department of Physics, The George Washington University, Washington, DC 20052 (United States); Poelker, M. [Thomas Jefferson National Accelerator Facility, 12050 Jefferson Avenue, Suite 500, Newport News, VA 23606 (United States)
2014-02-21
Two-photon absorption of 1560 nm light was used to generate polarized electron beams from unstrained GaAs photocathodes of varying thickness: 625 μm, 0.32 μm, and 0.18 μm. For each photocathode, the degree of spin polarization of the photoemitted beam was less than 50%, contradicting earlier predictions based on simple quantum mechanical selection rules for spherically-symmetric systems but consistent with the more sophisticated model of Bhat et al. (Phys. Rev. B 71 (2005) 035209). Polarization via two-photon absorption was the highest from the thinnest photocathode sample and comparable to that obtained via one-photon absorption (using 778 nm light), with values 40.3±1.0% and 42.6±1.0%, respectively.
International Nuclear Information System (INIS)
Palta, J.R.; Ayyangar, K.M.; Suntharalingam, N.
1988-01-01
Dosimetric measurements have been made of a 6 MV photon beam from a linear accelerator equipped with asymmetric jaws. The field size factors for asymmetrically set fields are compared to those for symmetrically set fields. The change of beam quality has been measured as a function of off-axis position of the asymmetric fields to assess its effect on depth dose. Additional measurements include beam penumbra and shape of isodose curves for open and wedge fields as the field opening is moved asymmetrically from the central ray
Two-photon polymerization of cylinder microstructures by femtosecond Bessel beams
Energy Technology Data Exchange (ETDEWEB)
Yang, Liang [Department of Precision Machinery and Precision Instrumentation, University of Science and Technology of China, Hefei 230026 (China); Laser Zentrum Hannover e.V., 30419 Hannover (Germany); El-Tamer, Ayman; Hinze, Ulf; Chichkov, Boris N [Laser Zentrum Hannover e.V., 30419 Hannover (Germany); Li, Jiawen, E-mail: jwl@ustc.edu.cn; Hu, Yanlei; Huang, Wenhao; Chu, Jiaru [Department of Precision Machinery and Precision Instrumentation, University of Science and Technology of China, Hefei 230026 (China)
2014-07-28
In this work, we present an approach to modulate femtosecond laser beams into Bessel beams with a spatial light modulator (SLM) for two-photon polymerization applications. Bessel beams with different parameters are generated and annular optical fields are produced at the focal plane of the objective. Uniform cylinder microstructures are fabricated by a single illumination during a few seconds without stage translation. By modulating the holograms encoded on the SLM, the diameters of the fabricated annular structures can be flexibly controlled in a wide range with no need of changing the optical elements and realignment of the optical path.
Finite element modelling of concrete beams reinforced with hybrid fiber reinforced bars
Smring, Santa binti; Salleh, Norhafizah; Hamid, NoorAzlina Abdul; Majid, Masni A.
2017-11-01
Concrete is a heterogeneous composite material made up of cement, sand, coarse aggregate and water mixed in a desired proportion to obtain the required strength. Plain concrete does not with stand tension as compared to compression. In order to compensate this drawback steel reinforcement are provided in concrete. Now a day, for improving the properties of concrete and also to take up tension combination of steel and glass fibre-reinforced polymer (GFRP) bars promises favourable strength, serviceability, and durability. To verify its promise and support design concrete structures with hybrid type of reinforcement, this study have investigated the load-deflection behaviour of concrete beams reinforced with hybrid GFRP and steel bars by using ATENA software. Fourteen beams, including six control beams reinforced with only steel or only GFRP bars, were analysed. The ratio and the ordinate of GFRP to steel were the main parameters investigated. The behaviour of these beams was investigated via the load-deflection characteristics, cracking behaviour and mode of failure. Hybrid GFRP-Steel reinforced concrete beam showed the improvement in both ultimate capacity and deflection concomitant to the steel reinforced concrete beam. On the other hand, finite element (FE) modelling which is ATENA were validated with previous experiment and promising the good result to be used for further analyses and development in the field of present study.
The tagged photon beam polarization of the jet target experiment
International Nuclear Information System (INIS)
Bianchi, N.; Muccifora, V.
1989-01-01
The applicability of the residual electron selection method to the tagging method of the jet target laboratory has been studied. With this end in view the behaviour of the polarized bremsstrahlung cross section in the range considered has been analysed, while the polarization increase by means of the RES has been evaluated. The vertical conditions of the focusing of the tagging spectrometer as a function of energy have been determined. Finally the gamma beam density and the tagging efficiency have been calculated
Dosimetric aspects of the therapeutic photon beams from a dual-energy linear accelerator
International Nuclear Information System (INIS)
Al-Ghazi, M.S.A.L.; Arjune, B.; Fiedler, J.A.; Sharma, P.D.
1988-01-01
Parameters of the photon beams (6 and 20 MV) from a dual-energy linear accelerator (Mevatron-KD, Siemens Medical Laboratories, CA) are presented. The depth dose characteristics of the photon beams are d/sub max/ of 1.8 and 3.8 cm and percentage depth dose of 68% and 80% at 10-cm depth and 100-cm source--surface distance for a field size of 10 x 10 cm 2 for 6 and 20 MV, respectively. The 6 and 20 MV beams were found to correspond to nominal accelerating potentials of 4.7 and 17 MV, respectively. The stability of output is within +- 1% and flatness and symmetry are within +- 3%. These figures compare favorably with the manufacturer's specifications
On the absorbed dose determination method in high energy photon beams
International Nuclear Information System (INIS)
Scarlat, F.; Scarisoreanu, A.; Oane, M.; Mitru, E.; Avadanei, C.
2008-01-01
The absorbed dose determination method in water, based on standards of air kerma or exposure in high energy photon beams generated by electron with energies in the range of 1 MeV to 50 MeV is presented herein. The method is based on IAEA-398, AAPM TG-51, DIN 6800-2, IAEA-381, IAEA-277 and NACP-80 recommendations. The dosimetry equipment is composed of UNIDOS T 10005 electrometer and different ionization chambers calibrated in air kerma method in a Co 60 beam. Starting from the general formalism showed in IAEA-381, the determination of absorbed dose in water, under reference conditions in high energy photon beams, is given. This method was adopted for the secondary standard dosimetry laboratory (SSDL) in NILPRP-Bucharest
A Liquid Scintillator System for Dosimetry of Photon and Proton Beams
International Nuclear Information System (INIS)
Beddar S
2010-01-01
We have developed a 3D system based on liquid scintillator (LS) for the dosimetry of photon and proton therapy. We have validated the LS detector system for fast and accurate quality assurance of IMRT and proton therapy fields. Further improvements are required to optimize the quantitative analysis of the light output provided by the system in photon beams. We have also demonstrated its usefulness for protons as it can determine the position and the range of proton beams. This system has also been shown to be capable of fast, sub-millimeter spatial localization of proton spots delivered in a 3D volume and could be used for quality assurance of IMPT. Further developments are on-going to measure beam intensities in 3D.
Two-photon equivalent weighting of spatial excimer laser beam profiles
Eva, Eric; Bauer, Harry H.; Metzger, K.; Pfeiffer, A.
2001-04-01
Damage in optical materials for semiconductor lithography applications caused by exposure to 248 or 193 nm light is usually two-photon driven, hence it is a nonlinear function of incident intensity. Materials should be tested with flat- topped temporal and spatial laser beam profiles to facilitate interpretation of data, but in reality this is hard to achieve. Sandstrom provided a formula that approximates any given temporal pulse shape with a two- photon equivalent rectangular pulse (Second Symposium on 193 nm Lithography, Colorado Springs 1997). Known as the integral-square pulse duration, this definition has been embraced as an industry standard. Originally faced with the problem of comparing results obtained with pseudo-Gaussian spatial profiles to literature data, we found that a general solution for arbitrarily inhomogeneous spatial beam profiles exists which results in a definition much similar to Sandstrom's. In addition, we proved the validity of our approach in experiments with intentionally altered beam profiles.
International Nuclear Information System (INIS)
Chen Xi-Yao; Lin Gui-Min; Li Jun-Jun; Xu Xiao-Fu; Jiang Jun-Zhen; Qiang Ze-Xuan; Qiu Yi-Shen; Li Hui
2012-01-01
A polarization beam splitter based on a self-collimation Michelson interferometer (SMI) in a hole-type silicon photonic crystal is proposed and numerically demonstrated. Utilizing the polarization dependence of the transmission spectra of the SMI and polarization peak matching method, the SMI can work as a polarization beam splitter (PBS) by selecting an appropriate path length difference in the structure. Based on its novel polarization beam splitting mechanics, the polarization extinction ratios (PERs) for TM and TE modes are as high as 18.4 dB and 24.3 dB, respectively. Since its dimensions are only several operating wavelengths, the PBS may have practical applications in photonic integrated circuits. (fundamental areas of phenomenology(including applications))
International Nuclear Information System (INIS)
Hawileh, R.A.
2015-01-01
Highlights: • Modeling of concrete beams reinforced steel and FRP bars. • Developed finite element models achieved good results. • The models are validated via comparison with experimental results. • Parametric studies are performed. - Abstract: Corrosion of steel bars has an adverse effect on the life-span of reinforced concrete (RC) members and is usually associated with crack development in RC beams. Fiber reinforced polymer (FRP) bars have been recently used to reinforce concrete members in flexure due to their high tensile strength and superior corrosion resistance properties. However, FRP materials are brittle in nature, thus RC beams reinforced with such materials would exhibit a less ductile behavior when compared to similar members reinforced with conventional steel reinforcement. Recently, researchers investigated the performance of concrete beams reinforced with a hybrid combination of steel and Aramid Fiber Reinforced Polymer (AFRP) reinforcement to maintain a reasonable level of ductility in such members. The function of the AFRP bars is to increase the load-carrying capacity, while the function of the steel bars is to ensure ductility of the flexural member upon yielding in tension. This paper presents a three-dimensional (3D) finite element (FE) model that predicted the load versus mid-span deflection response of tested RC beams conducted by other researchers with a hybrid combination of steel and AFRP bars. The developed FE models account for the constituent material nonlinearities and bond–slip behavior between the reinforcing bars and adjacent concrete surfaces. It was concluded that the developed models can accurately capture the behavior and predicts the load-carrying capacity of such RC members. In addition, a parametric study is conducted using the validated models to investigate the effect of AFRP bar size, FRP material type, bond–slip action, and concrete compressive strength on the performance of concrete beams when reinforced
The response of a radiophotoluminescent glass dosimeter in megavoltage photon and electron beams.
Araki, Fujio; Ohno, Takeshi
2014-12-01
This study investigated the response of a radiophotoluminescent glass dosimeter (RGD) in megavoltage photon and electron beams. The RGD response was compared with ion chamber measurements for 4-18 MV photons and 6-20 MeV electrons in plastic water phantoms. The response was also calculated via Monte Carlo (MC) simulations with EGSnrc/egs_chamber and Cavity user-codes, respectively. In addition, the response of the RGD cavity was analyzed as a function of field sizes and depths according to Burlin's general cavity theory. The perturbation correction factor, PQ, in the RGD cavity was also estimated from MC simulations for photon and electron beams. The calculated and measured RGD energy response at reference conditions with a 10 × 10 cm(2) field and 10 cm depth in photons was lower by up to 2.5% with increasing energy. The variation in RGD response in the field size range of 5 × 5 cm(2) to 20 × 20 cm(2) was 3.9% and 0.7%, at 10 cm depth for 4 and 18 MV, respectively. The depth dependence of the RGD response was constant within 1% for energies above 6 MV but it increased by 2.6% and 1.6% for a large (20 × 20 cm(2)) field at 4 and 6 MV, respectively. The dose contributions from photon interactions (1 - d) in the RGD cavity, according to Burlin's cavity theory, decreased with increasing energy and decreasing field size. The variation in (1 - d) between field sizes became larger with increasing depth for the lower energies of 4 and 6 MV. PQ for the RGD cavity was almost constant between 0.96 and 0.97 at 10 MV energies and above. Meanwhile, PQ depends strongly on field size and depth for 4 and 6 MV photons. In electron beams, the RGD response at a reference depth, dref, varied by less than 1% over the electron energy range but was on average 4% lower than the response for 6 MV photons. The RGD response for photon beams depends on both (1 - d) and perturbation effects in the RGD cavity. Therefore, it is difficult to predict the energy dependence of RGD response by
A Polarized High-Energy Photon Beam for Production of Exotic Mesons
Energy Technology Data Exchange (ETDEWEB)
Senderovich, Igor [Univ. of Connecticut, Storrs, CT (United States)
2012-01-01
This work describes design, prototyping and testing of various components of the Jefferson Lab Hall D photon beamline. These include coherent bremsstrahlung radiators to be used in this facility for generating the photon beam, a fine resolution hodoscope for the facility's tagging spectrometer, and a photon beam position sensor for stabilizing the beam on a collimator. The principal instrumentation project was the hodoscope: its design, implementation and beam testing will be thoroughly described. Studies of the coherent bremsstrahlung radiators involved X-ray characterization of diamond crystals to identify the appropriate line of manufactured radiators and the proper techniques for thinning them to the desired specification of the beamline. The photon beam position sensor project involved completion of a designed detector and its beam test. The results of these shorter studies will also be presented. The second part of this work discusses a Monte Carlo study of a possible photo-production and decay channel in the GlueX experiment that will be housed in the Hall D facility. Specifically, the γ p → Xp → b_{1 π} → ω π^{+1} π^{-1} channel was studied including its Amplitude Analysis. This exercise attempted to generate a possible physics signal, complete with internal angular momentum states, and be able to reconstruct the signal in the detector and find the proper set of JPC quantum numbers through an amplitude fit. Derivation of the proper set of amplitudes in the helicity basis is described, followed by a discussion of the implementation, generation of the data sets, reconstruction techniques, the amplitude fit and results of this study.
Equivalent (uniform) square field sizes of flattening filter free photon beams
Lechner, Wolfgang; Kuess, Peter; Georg, Dietmar; Palmans, Hugo
2017-10-01
Various types of treatment units, such as CyberKnife, TomoTherapy and C-arm linear accelerators (LINACs) are operated using flattening filter free (FFF) photon beams. Their reference dosimetry, however, is currently based on codes of practice that provide data which were primarily developed and tested for high-energy photon beams with flattening filter (WFF). The aim of this work was to introduce equivalent uniform square field sizes of FFF beams to serve as a basis of a unified reference dosimetry procedure applicable to all aforementioned FFF machines. For this purpose, in-house determined experimental data together with published data of the ratio of doses at depths of 20 cm and 10 cm in water (D 20,10) were used to characterize the depth dose distribution of 6 and 10 MV WFF and FFF beams. These data were analyzed for field sizes ranging from 2 × 2 cm2 to 40 × 40 cm2. A scatter function that takes the lateral profiles of the individual beams into account was fitted to the experimental data. The lateral profiles of the WFF beams were assumed to be uniform, while those of the FFF beams were approximated using fourth or sixth order polynomials. The scatter functions of the FFF beams were recalculated using a uniform lateral profile (the same as the physical profile of the WFF beams), and are henceforth denoted as virtual uniform FFF beams (VUFFF). The field sizes of the VUFFF beams having the same scatter contribution as the corresponding FFF beams at a given field size were defined as the equivalent uniform square field (EQUSF) size. Data from four different LINACs with 18 different beams in total, as well as a CyberKnife beam, were analyzed. The average values of EQUSFs over all investigated LINACs of the conventional 10 × 10 cm2 reference fields of 6 MV and 10 MV FFF beams for C-arm LINACs and machine-specific reference fields for CyberKnife and TomoTherapy were 9.5 cm, 9 cm, 5.0 cm and 6.5 cm respectively. The
Self-collimating photonic crystal polarization beam splitter.
Zabelin, V; Dunbar, L A; Le Thomas, N; Houdré, R; Kotlyar, M V; O'Faolain, L; Krauss, T F
2007-03-01
We present theoretical and experimental results of a polarization splitter device that consists of a photonic crystal (PhC) slab, which exhibits a large reflection coefficient for TE and a high transmission coefficient for TM polarization. The slab is embedded in a PhC tile operating in the self-collimation mode. Embedding the polarization-discriminating slab in a PhC with identical lattice symmetry suppresses the in-plane diffraction losses at the PhC-non-PhC interface. The optimization of the PhC-non-PhC interface is thereby decoupled from the optimization of the polarizing function. Transmissions as high as 35% for TM- and 30% for TE-polarized light are reported.
Analytical description of photon beam phase spaces in inverse Compton scattering sources
Directory of Open Access Journals (Sweden)
C. Curatolo
2017-08-01
Full Text Available We revisit the description of inverse Compton scattering sources and the photon beams generated therein, emphasizing the behavior of their phase space density distributions and how they depend upon those of the two colliding beams of electrons and photons. The main objective is to provide practical formulas for bandwidth, spectral density, brilliance, which are valid in general for any value of the recoil factor, i.e. both in the Thomson regime of negligible electron recoil, and in the deep Compton recoil dominated region, which is of interest for gamma-gamma colliders and Compton sources for the production of multi-GeV photon beams. We adopt a description based on the center of mass reference system of the electron-photon collision, in order to underline the role of the electron recoil and how it controls the relativistic Doppler/boost effect in various regimes. Using the center of mass reference frame greatly simplifies the treatment, allowing us to derive simple formulas expressed in terms of rms momenta of the two colliding beams (emittance, energy spread, etc. and the collimation angle in the laboratory system. Comparisons with Monte Carlo simulations of inverse Compton scattering in various scenarios are presented, showing very good agreement with the analytical formulas: in particular we find that the bandwidth dependence on the electron beam emittance, of paramount importance in Thomson regime, as it limits the amount of focusing imparted to the electron beam, becomes much less sensitive in deep Compton regime, allowing a stronger focusing of the electron beam to enhance luminosity without loss of mono-chromaticity. A similar effect occurs concerning the bandwidth dependence on the frequency spread of the incident photons: in deep recoil regime the bandwidth comes out to be much less dependent on the frequency spread. The set of formulas here derived are very helpful in designing inverse Compton sources in diverse regimes, giving a
Araki, Fujio
2017-07-01
The dosimetric properties of the recently developed SW557 phantom have been investigated by comparison with those of the existing SW457 phantom in megavoltage photon beams. The electron fluence ratio φ pl w , and chamber ionization ratio k pl , of water to SW457 and water to SW557 for 4-15MV photons were calculated as a function of depth using Monte Carlo simulations, and compared with measured values. Values of φ pl w for SW457 were in the range of 1.004-1.014 for 4MV, and 1.014-1.018 for 15MV photons. The φ pl w for SW557 ranged from 1.005 to 1.008 for 4MV and from 1.010 to 1.015 for 15MV photons and the variation of φ pl w with depth for each beam energy was within ±0.5%. Values of k pl were obtained with a PTW 30013 Farmer-type ionization chamber. The k pl for SW457 ranged from 0.997 to 1.011 for 4-15MV photons. Values of k pl for SW557 were almost unity for 4 and 6MV photons, while in the case of 10 and 15MV photons they were less than 1.006, excepting the build-up region. The measured and calculated k pl values of water to SW557 were in the range of 0.997-1.002 and 1.000-1.006, respectively, for 4-15MV photons, at a depth of 10cm with a source-to-axis distance of 100cm. The measured and calculated k pl values were in agreement within their uncertainty ranges. As a water-equivalent phantom, SW557 can be used with a dosimetric difference within±0.6%, for 4-15MV photons, and is more water-equivalent than SW457 in megavoltage photon beams. Copyright © 2017 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.
International Nuclear Information System (INIS)
Jaszczak, R.J.; Jianying Li; Huili Wang; Coleman, R.E.
1992-01-01
Single photon emission computed tomography (SPECT) using cone beam (CB) collimation exhibits increased sensitivity compared with acquisition geometries using parallel (P) hole collimation. However, CB collimation has a smaller field-of-view which may result in truncated projections and image artifacts. A primary objective of this work is to investigate maximum likelihood-expectation maximization (ML-EM) methods to reconstruct simultaneously acquired parallel and cone beam (P and CB) SPECT data. Simultaneous P and CB acquisition can be performed with commercially available triple camera systems by using two cone-beam collimators and a single parallel-hole collimator. The loss in overall sensitivity (relative to the use of three CB collimators) is about 15 to 20%. The authors have developed three methods to combine P and CB data using modified ML-EM algorithms. (author)
International Nuclear Information System (INIS)
Koivunoro, H.; Uusi-Simola, J.; Savolainen, S.; Kotiluoto, P.; Auterinen, I.; Kosunen, A.
2006-01-01
At FiR 1 BNCT facility in Finland, neutron-insensitive Mg(Ar) ionization chambers are used for photon dose measurements in an epithermal neutron beam. Previously, photon sensitivity factors for the chamber for the measurements in a water phantom in FiR 1 beam have been determined experimentally from measurements in 60 Co gamma and in a 6 MV clinical accelerator photon beams. However, the response of the ionization chamber in a water phantom depends on energy spectrum and angle of the photons and the secondary electrons created inside the phantom and may differ depending on type of the irradiation source (accelerator vs. an epithermal neutron beam). Also, the experimental sensitivity factor does not take into account the possible perturbations in the photon production in phantom caused by the ionization chamber materials. Therefore, it is necessary to determine the photon quality correction factors (k Qγ ) for the Mg(Ar) chamber at the FiR 1 beam through computer simulations. In this study, the k Qγ factors have been determined for Mg(Ar) chamber from Monte Carlo calculations of absorbed photon dose at two depths in a water phantom using MCNP code. The k qγ factors obtained with this method are compared to the sensitivity factors determined with measurements in an accelerator photon beam and to the k Qγ factors published previously. (author)
Beam-bending in spatially variant photonic crystals at telecommunications wavelengths
Digaum, Jennefir L.; Sharma, Rashi; Batista, Daniel; Pazos, Javier J.; Rumpf, Raymond C.; Kuebler, Stephen M.
2016-03-01
This work reports the fabrication of micron-scale spatially variant photonic crystals (SVPCs) and their use for steering light beams through turns with bending radius Rbend on the order of ten times the optical wavelength λ0. Devices based on conventional photonic crystals, metamaterials, plasmonics and transformation optics have all been explored for controlling light beams and steering them through tight turns. These devices offer promise for photonic interconnects, but they are based on exotic materials, including metals, that make them impractically lossy or difficult to fabricate. Waveguides can also be used to steer light using total internal reflection; however, Rbend of a waveguide must be hundreds of times λ0 to guide light efficiently, which limits their use in optical circuits. SVPCs are spatially variant 3D lattices which can be created in transparent, low-refractive-index media and used to control the propagation of light through the self-collimation effect. SVPCs were fabricated by multi-photon lithography using the commercially available photo-polymer IP-DIP. The SVPCs were structurally and optically characterized and found to be capable of bending light having λ0 = 1.55 μm through a 90-degree turn with Rbend = 10 μm. Curved waveguides with Rbend = 15 μm and 35 μm were also fabricated using IP-DIP and optically characterized. The SVPCs were able to steer the light beams through tighter turns than either waveguide and with higher efficiency.
Rapid volumetric imaging with Bessel-Beam three-photon microscopy
Chen, Bingying; Huang, Xiaoshuai; Gou, Dongzhou; Zeng, Jianzhi; Chen, Guoqing; Pang, Meijun; Hu, Yanhui; Zhao, Zhe; Zhang, Yunfeng; Zhou, Zhuan; Wu, Haitao; Cheng, Heping; Zhang, Zhigang; Xu, Chris; Li, Yulong; Chen, Liangyi; Wang, Aimin
2018-01-01
Owing to its tissue-penetration ability, multi-photon fluorescence microscopy allows for the high-resolution, non-invasive imaging of deep tissue in vivo; the recently developed three-photon microscopy (3PM) has extended the depth of high-resolution, non-invasive functional imaging of mouse brains to beyond 1.0 mm. However, the low repetition rate of femtosecond lasers that are normally used in 3PM limits the temporal resolution of point-scanning three-photon microscopy. To increase the volumetric imaging speed of 3PM, we propose a combination of an axially elongated needle-like Bessel-beam with three-photon excitation (3PE) to image biological samples with an extended depth of focus. We demonstrate the higher signal-to-background ratio (SBR) of the Bessel-beam 3PM compared to the two-photon version both theoretically and experimentally. Finally, we perform simultaneous calcium imaging of brain regions at different axial locations in live fruit flies and rapid volumetric imaging of neuronal structures in live mouse brains. These results highlight the unique advantage of conducting rapid volumetric imaging with a high SBR in the deep brain in vivo using scanning Bessel-3PM.
Determination of the quality index (Q) for photon beams at arbitrary field sizes.
Sauer, Otto A
2009-09-01
A commonly used beam quality index (Q) for high-energy photon beams is the tissue phantom ratio (TPR20,10) for a square field of 10 x 10 cm2 and SDD of 100 cm. On some specialized radiotherapy treatment equipment such a reference collimator setting is not achievable. Likewise a flat beam profile, not explicitly required in dosimetry protocols, but certainly influences the measurement of Q, is not always produced. In this work, a method was developed in order to determine Q at any field size, especially for small and nonflattened beams. An analytical relationship was derived between TPR20,10 for arbitrary field sizes and Q [the TPR20,10 (10 x 10 cm2)] as quality index. The proposed model equation was fitted to the measured and published data in order to achieve three general fit parameters. The procedure was then tested with published data from TomoTherapy and CyperKnife treatment devices. For standard flattened photon fields, the uncertainty in Q measured at any field size using the parameters derived from this study is better than 1%. In flattening-filter free beams, the proposed procedure results in a reliable Q for any field size setting. A method is introduced and successfully tested in order to measure the beam quality under nonstandard conditions. It can be used, e.g., to get energy dependent correction factors as tabulated in dosimetry codes of practice even if standard conditions are not adjustable.
SU-E-T-238: Deriving Electron Contamination Spectra From Pure and Clinical Photon Beams
International Nuclear Information System (INIS)
Smit, C; Plessis, F du
2015-01-01
Purpose: To extract the electron contamination energy spectra for an Elekta Precise Linac, based on pure photon and measured clinical beam percentage depth dose data. And to include this as an additional source in isource 4 in DOSXYZnrc. Methods: A pure photon beam was simulated for the Linac using isource 4 in the DOSXYZnrc Monte Carlo (MC) code. Percentage depth dose (PDD) data were extracted afterwards for a range of field sizes (FS). These simulated dose data were compared to actual measured dose PDD data, with the data normalized at 10 cm depth. The resulting PDD data resembled the electron contamination depth dose. Since the dose fall-off is a strictly decreasing function, a method was adopted to derive the contamination electron spectrum. Afterwards this spectrum was used in a DOSXYZnrc MC simulation run to verify that the original electron depth dose could be replicated. Results: Various square aperture FS’s for 6, 8 and 15 megavolt (MV) photon beams were modeled, simulated and compared to their respective actual measured PDD data. As FS increased, simulated pure photon depth-dose profiles shifted deeper, thus requiring electron contamination to increase the surface dose. The percentage of electron weight increased with increase in FS. For a FS of 15×15 cm 2 , the percentage electron weight is 0.1%, 0.2% and 0.4% for 6, 8 and 15 MV beams respectively. Conclusion: From the PDD results obtained, an additional electron contamination source was added to the photon source model so that simulation and measured PDD data could match within 2 % / 2 mm gamma-index criteria. The improved source model could assure more accurate simulations of surface doses. This research project was funded by the South African Medical Research Council (MRC) with funds from National Treasury under its Economic Competitiveness and Support package
SU-E-T-238: Deriving Electron Contamination Spectra From Pure and Clinical Photon Beams
Energy Technology Data Exchange (ETDEWEB)
Smit, C; Plessis, F du [University of the Free State, Bloemfontein, Free State (South Africa)
2015-06-15
Purpose: To extract the electron contamination energy spectra for an Elekta Precise Linac, based on pure photon and measured clinical beam percentage depth dose data. And to include this as an additional source in isource 4 in DOSXYZnrc. Methods: A pure photon beam was simulated for the Linac using isource 4 in the DOSXYZnrc Monte Carlo (MC) code. Percentage depth dose (PDD) data were extracted afterwards for a range of field sizes (FS). These simulated dose data were compared to actual measured dose PDD data, with the data normalized at 10 cm depth. The resulting PDD data resembled the electron contamination depth dose. Since the dose fall-off is a strictly decreasing function, a method was adopted to derive the contamination electron spectrum. Afterwards this spectrum was used in a DOSXYZnrc MC simulation run to verify that the original electron depth dose could be replicated. Results: Various square aperture FS’s for 6, 8 and 15 megavolt (MV) photon beams were modeled, simulated and compared to their respective actual measured PDD data. As FS increased, simulated pure photon depth-dose profiles shifted deeper, thus requiring electron contamination to increase the surface dose. The percentage of electron weight increased with increase in FS. For a FS of 15×15 cm{sup 2}, the percentage electron weight is 0.1%, 0.2% and 0.4% for 6, 8 and 15 MV beams respectively. Conclusion: From the PDD results obtained, an additional electron contamination source was added to the photon source model so that simulation and measured PDD data could match within 2 % / 2 mm gamma-index criteria. The improved source model could assure more accurate simulations of surface doses. This research project was funded by the South African Medical Research Council (MRC) with funds from National Treasury under its Economic Competitiveness and Support package.
Dose calculation methods in photon beam therapy using energy deposition kernels
International Nuclear Information System (INIS)
Ahnesjoe, A.
1991-01-01
The problem of calculating accurate dose distributions in treatment planning of megavoltage photon radiation therapy has been studied. New dose calculation algorithms using energy deposition kernels have been developed. The kernels describe the transfer of energy by secondary particles from a primary photon interaction site to its surroundings. Monte Carlo simulations of particle transport have been used for derivation of kernels for primary photon energies form 0.1 MeV to 50 MeV. The trade off between accuracy and calculational speed has been addressed by the development of two algorithms; one point oriented with low computional overhead for interactive use and one for fast and accurate calculation of dose distributions in a 3-dimensional lattice. The latter algorithm models secondary particle transport in heterogeneous tissue by scaling energy deposition kernels with the electron density of the tissue. The accuracy of the methods has been tested using full Monte Carlo simulations for different geometries, and found to be superior to conventional algorithms based on scaling of broad beam dose distributions. Methods have also been developed for characterization of clinical photon beams in entities appropriate for kernel based calculation models. By approximating the spectrum as laterally invariant, an effective spectrum and dose distribution for contaminating charge particles are derived form depth dose distributions measured in water, using analytical constraints. The spectrum is used to calculate kernels by superposition of monoenergetic kernels. The lateral energy fluence distribution is determined by deconvolving measured lateral dose distributions by a corresponding pencil beam kernel. Dose distributions for contaminating photons are described using two different methods, one for estimation of the dose outside of the collimated beam, and the other for calibration of output factors derived from kernel based dose calculations. (au)
B-factory via conversion of 1-TeV electron beams into 1-TeV photon beams
International Nuclear Information System (INIS)
Mtingwa, S.K.
1991-01-01
This paper reports on the study of CP violation and rare decays of beauty particles which are pressing problems in high-energy physics. It is known that one should analyze beauty decays of at least the order of 10 8 or 19 9 . Thus, numerous proposals for beauty factories are being discussed now, although some of these projects are likely to supply much smaller numbers of beauty events. At the same time, at present several projects, such as CLIC (Cern Linear Collider), expect to build linear e + e - colliders with beam energies up to 1 TeV. The aim of this work is to show that the possibility exists of using the unique features of the discussed teraelectron volt electron linacs to obtain a facility for the production of beauty via photoproduction of nuclei. Unique features of high-energy photoproduction are as follows. The rather large fraction (∼2 x 10 -4 ) of events with beauty at E γ ∼ 1 TeV. Beauty particles are produced with about equally large momenta ∼0.05 E γ and at rather large transverse momenta p t ∼ m b . The following scheme can be envisioned. The 1-TeV electron beam is Compton scattered off a low-energy (∼ 1-eV) laser pulse. The laser photons are thus converted into a highly collimated beam of energy E γ ∼ E e , directed along the electron's original line of motion. Such schemes to produce high-energy photon beams have been discussed. These 1-TeV photons are subsequently scattered onto a nuclear target to produce b bar b pairs
Finite Element Analysis of the Pseudo-elastic Behavior of Shape Memory Alloy Truss and Beam
Directory of Open Access Journals (Sweden)
Kamal M. Bajoria
2010-07-01
Full Text Available The pseudo-elastic behavior of Shape memory alloy (SMA truss and cantilever beam are investigated. Brinson’s one-dimensional material model, which uses the twinned and detwinned martensite fractions separately as internal variables, is applied in the algorithm to establish the SMA stress-strain characteristics. This material model also incorporates different young’s modulus for austenitic and martensite phase to represent the true SMA characteristics. In this model, a cosine function was used to express the evolution of the stress induced martensite fractions during the forward and reverse martensite phase transformation. A finite element formulation for the SMA truss member considering the geometric nonlinearity is proposed and the results are compared with the corresponding linear analysis. As a step forward, a finite element formulation for an SMA cantilever beam with an applied end moment is proposed. The load displacement characteristic for both the loading and unloading phases are considered to check the full pseudo-elastic hysteretic loop. In the numerical investigation, the stress-strain variation along the beam depth is also examined during the loading and unloading process to investigate the forward and reverse martensite phase transformation phenomena. Newton-Raphson’s iterative method is applied to get convergence to the equilibrium for each loading steps. During a complete loading-unloading process, the temperature is kept constant as the model is essentially an isothermal model. Numerical simulation is performed considering two different temperatures to demonstrate the effect of temperature on the hysteretic loop.
Small-Field Dosimetry in A 6 MV Photon Beam Using Alanine and Liquid Ionisation Chamber
DEFF Research Database (Denmark)
Zimmermann, S.; Riis, H. L.; Hjelm-Hansen, M.
2012-01-01
Purpose/Objective: Dosimetry of small field sizes in MV photon beams is an increasingly important subject, and a generally accepted guideline for clinical measurements is still lacking. The present comparative study was carried out to further investigate the use of alanine and the PTW microLion...... of each field and depth. This dose maximum was measured for each field using a Scanditronix Wellhöfer photon field diode. The same measurements were carried out using a liquid ionchamber, PTW microLion, irradiated by 500 MU. The output of the accelerator was controlled by a PTW semiflex ion chamber...
Correlated two-photon interference in a dual-beam Michelson interferometer
International Nuclear Information System (INIS)
Kwiat, P.G.; Vareka, W.A.; Hong, C.K.; Nathel, H.; Chiao, R.Y.
1990-01-01
We report on an interference effect arising from a two-photon entangled state produced in a potassium dihydrogen phosphate (KDP) crystal pumped by an ultraviolet argon-ion laser. Two conjugate beams of signal and idler photons were injected in a parallel configuration into a single Michelson interferometer, and detected separately by two photomultipliers, while the difference in its arm lengths was slowly scanned. The coincidence rate exhibited fringes with a visibility of nearly 50%, and a period given by half the ultraviolet (not the signal or idler) wavelength, while the singles rate exhibited no fringes
A high-speed tunable beam splitter for feed-forward photonic quantum information processing.
Ma, Xiao-Song; Zotter, Stefan; Tetik, Nuray; Qarry, Angie; Jennewein, Thomas; Zeilinger, Anton
2011-11-07
We realize quantum gates for path qubits with a high-speed, polarization-independent and tunable beam splitter. Two electro-optical modulators act in a Mach-Zehnder interferometer as high-speed phase shifters and rapidly tune its splitting ratio. We test its performance with heralded single photons, observing a polarization-independent interference contrast above 95%. The switching time is about 5.6 ns, and a maximal repetition rate is 2.5 MHz. We demonstrate tunable feed-forward operations of a single-qubit gate of path-encoded qubits and a two-qubit gate via measurement-induced interaction between two photons.
Propagation of photons induced by a proton beam in a quartz bar
International Nuclear Information System (INIS)
Plaszczynski, S.; Torassa, E.; Wormser, G.; Aleksan, R.; Amadon, A.; Besson, P.; Bourgeois, P.; Monchenault, G.H.
1995-01-01
For the BABAR experiment, an amplitude and timing analysis produced by a proton beam in a quartz bar has been performed. The photoelectron yield is obtained for different angles and positions. It is found that the linear speed of propagation of the Cherenkov photons through the bar is well described by internal reflections. The timing analysis reveals also the existence of an unexpected parasite light component, not described by the simulation. However, all the results can be reproduced by adding an isotopic scintillation of about 4 photons/cm in the quartz bar on the trajectory of the primary proton, in 50 % of the cases. (authors)
Hamilton, Mark F.
1989-08-01
Four projects are discussed in this annual summary report, all of which involve basic research in nonlinear acoustics: Scattering of Sound by Sound, a theoretical study of two nonconlinear Gaussian beams which interact to produce sum and difference frequency sound; Parametric Receiving Arrays, a theoretical study of parametric reception in a reverberant environment; Nonlinear Effects in Asymmetric Sound Beams, a numerical study of two dimensional finite amplitude sound fields; and Pulsed Finite Amplitude Sound Beams, a numerical time domain solution of the KZK equation.
Peripheral photon and neutron doses from prostate cancer external beam irradiation.
Bezak, Eva; Takam, Rundgham; Marcu, Loredana G
2015-12-01
Peripheral photon and neutron doses from external beam radiotherapy (EBRT) are associated with increased risk of carcinogenesis in the out-of-field organs; thus, dose estimations of secondary radiation are imperative. Peripheral photon and neutron doses from EBRT of prostate carcinoma were measured in Rando phantom. (6)LiF:Mg,Cu,P and (7)LiF:Mg,Cu,P glass-rod thermoluminescence dosemeters (TLDs) were inserted in slices of a Rando phantom followed by exposure to 80 Gy with 18-MV photon four-field 3D-CRT technique. The TLDs were calibrated using 6- and 18-MV X-ray beam. Neutron dose equivalents measured with CR-39 etch-track detectors were used to derive readout-to-neutron dose conversion factor for (6)LiF:Mg,Cu,P TLDs. Average neutron dose equivalents per 1 Gy of isocentre dose were 3.8±0.9 mSv Gy(-1) for thyroid and 7.0±5.4 mSv Gy(-1) for colon. For photons, the average dose equivalents per 1 Gy of isocentre dose were 0.2±0.1 mSv Gy(-1) for thyroid and 8.1±9.7 mSv Gy(-1) for colon. Paired (6)LiF:Mg,Cu,P and (7)LiF:Mg,Cu,P TLDs can be used to measure photon and neutron doses simultaneously. Organs in close proximity to target received larger doses from photons than those from neutrons whereas distally located organs received higher neutron versus photon dose. © The Author 2015. Published by Oxford University Press. All rights reserved. For Permissions, please email: journals.permissions@oup.com.
Photon beam polarization and non-dipolar angular distributions
International Nuclear Information System (INIS)
Peshkin, M.
1996-01-01
Angular distributions of ejecta from unoriented atoms and molecules depend upon the polarization state of the incident x-rays as well as upon the dynamics of the physical systems being studied. I recommend a simple geometrical way of looking at the polarization and its effects upon angular distributions. The polarization is represented as a vector in a parameter space that faithfully represents the polarization of the beam. The simple dependence of the angular dependence of the angular distributions on the polarization vector enables easy extraction of the dynamical information contained in those angular distributions. No new physical results emerge from this geometrical approach, but known consequences of the symmetries appear in an easily visualized form that I find pleasing and that has proved to be useful for planning experiments and for analyzing data
A beam position feedback system for beam lines at the photon factory
International Nuclear Information System (INIS)
Katsura, T.; Kamiya, Y.; Haga, K.; Mitsuhashi, T.
1987-01-01
The beam position of the synchrotron radiation produced from the Storage Ring was stabilized by a twofold position feedback system. A digital feedback system was developed to suppress the diurnal beam movement (one cycle of sin-like drifting motion per day) which became a serious problem in low-emittance operation. The feedback was applied to the closed-orbit-distortion (COD) correction system in order to cancel the position variation at all the beam lines proportionately to the variation monitored at one beam line. An analog feedback system is also used to suppress frequency components faster than the slow diurnal movement
Static Analysis of Steel Fiber Concrete Beam With Heterosis Finite Elements
Directory of Open Access Journals (Sweden)
James H. Haido
2014-08-01
Full Text Available Steel fiber is considered as the most commonly used constructional fibers in concrete structures. The formulation of new nonlinearities to predict the static performance of steel fiber concrete composite structures is considered essential. Present study is devoted to investigate the efficiency of utilizing heterosis finite elements analysis in static analysis of steel fibrous beams. New and simple material nonlinearities are proposed and used in the formulation of these elements. A computer program coded in FORTRAN was developed to perform current finite element static analysis with considering four cases of elements stiffness matrix determination. The results are compared with the experimental data available in literature in terms of central deflections, strains, and failure form, good agreement was found. Suitable outcomes have been observed in present static analysis with using of tangential stiffness matrix and stiffness matrix in second iteration of the load increment.
Which accelerator photon beams are 'clinic-like' for reference dosimetry purposes?
International Nuclear Information System (INIS)
Kalach, N.I.; Rogers, D.W.O.
2003-01-01
Previous work has demonstrated that, for photon beam dosimetry, TPR 10 20 is not an ideal beam quality specifier for all bremsstrahlung beams, especially for lightly filtered beams in some standards laboratories. This paper addresses the following questions: Is TPR 10 20 an adequate beam quality specifier for all modern clinical therapy accelerators? When can nonclinical beams in standards laboratories be used to calibrate ion chambers or measure k Q factors as a function of TPR 10 20 ? Based on detailed Monte Carlo simulations of Varian, Siemens, Elekta, and GE (Saturn) accelerators one can conclude that TPR 10 20 is an adequate beam quality specifier for all these machines in the sense that for a given value of TPR 10 20 , the value of stopping-power ratios is the same. Agreement between calculations for specific accelerators and measured beam quality specifiers is shown to be good, but agreement with published data for a variety of clinical accelerators is not as good. Possible reasons for the discrepancy are discussed
Challenges of fabricating plasmonic and photonic structures with Neon ion beam milling
DEFF Research Database (Denmark)
Leißner, Till; Fiutowski, Jacek; Bozhevolnyi, Sergey I.
-established electron beam lithography and focussed ion beam milling (FIB) using Gallium ions. These techniques, however, are to some extend limited in their resolution, and in addition Gallium and Carbon are implanted and deposited into the plasmonic structures during FIB process, potentially changing plasmonic...... properties. We are currently studying the capabilities of focussed Helium and Neon ion beam milling for the fabricating of plasmonic and photonic devices. We found that Neon ion beam milling enables us to prepare plasmonic structures, such as trenches (see Fig. 1) and V-grooves without doping and alloying...... effects specific to Galium FIB. Neon FIB milling is superior to Helium FIB milling in terms of the processing speed and smaller levels of implanted ions. From our perspective it is the most promising technique for the fabrication of individual plasmonic devices with a few nanometers precision. The main...
Surface dose investigation of the flattening filter-free photon beams.
Wang, Yuenan; Khan, Mohammad K; Ting, Joseph Y; Easterling, Stephen B
2012-06-01
Flattening filter-free (FFF) x-rays can provide more efficient use of photons and a significant increase of dose rate compared with conventional flattened x-rays, features that are especially beneficial for stereotactic radiosurgery (SRS) and stereotactic body radiotherapy (SBRT). The available data on the entrance doses of the FFF photon beams remain limited. The purpose of this study was to investigate the entrance dose of FFF photons in the buildup region and to compare it with that of conventional flattened photons. A Varian TrueBeam linear accelerator has been in full clinical operation with 6-MV and 10-MV FFF and flattened x-ray photons. Entrance dose at the surface was measured using a parallel plate ionization chamber in a solid water phantom with buildup depth = 0~15 mm for 6X and 0~25 mm for 10X. Different field size (FS) patterns were created in the Eclipse Treatment Planning System by multileaf collimator (MLC) rather than jaws (FS = 2 × 2, 3 × 3, 4 × 4, 6 × 6, and 10 × 10 cm(2) by MLC and jaw size = 2.2 × 2.2, 3.2 × 3.2, 4.2 × 4.2, 6 × 6, and 10 × 10 cm(2)). The smallest FS was about four times larger than the ion chamber dimension. All buildup dose measurements were normalized to FS = 10 × 10 cm(2) at the depth of dose maximum (dmax). Good repeatability was demonstrated and surface dose increased linearly with FS for both flattened and FFF photons. The entrance dose of the FFF photons was modestly larger than that of the corresponding flattened photons for both 6X and 10X for different FS ranging from 2 × 2 cm(2) to 10 × 10 cm(2). The FFF photons have a higher entrance dose than that of the corresponding flattened photons for FS smaller than 10 × 10 cm(2). However, the difference is not substantial and may be clinically insignificant. Published by Elsevier Inc.
DEFF Research Database (Denmark)
Ormarsson, Sigurdur; Dahlblom, Ola
2013-01-01
Wood is a hygro-mechanical, non-isotropic and inhomogeneous material concerning both modulus of elasticity (MOE) and shrinkage properties. In stress calculations associated with ordinary timber design, these matters are often not dealt with properly. The main reason for this is that stress...... and the longitudinal shrinkage coefficient vary considerably from pith to bark. The question is how much these variations affect the stress distribution in wooden structures exposed to variable moisture climate. The paper presents a finite element implementation of a beam element with the aim of studying how wooden...
International Nuclear Information System (INIS)
Griessbach, Irmgard; Lapp, Markus; Bohsung, Joerg; Gademann, Guenther; Harder, Dietrich
2005-01-01
Shielded p-silicon diodes, frequently applied in general photon-beam dosimetry, show certain imperfections when applied in the small photon fields occurring in stereotactic or intensity modulated radiotherapy (IMRT), in electron beams and in the buildup region of photon beam dose distributions. Using as a study object the shielded p-silicon diode PTW 60008, well known for its reliable performance in general photon dosimetry, we have identified these imperfections as effects of electron scattering at the metallic parts of the shielding. In order to overcome these difficulties a new, unshielded diode PTW 60012 has been designed and manufactured by PTW Freiburg. By comparison with reference detectors, such as thimble and plane-parallel ionization chambers and a diamond detector, we could show the absence of these imperfections. An excellent performance of the new unshielded diode for the special dosimetric tasks in small photon fields, electron beams and build-up regions of photon beams has been observed. The new diode also has an improved angular response. However, due to its over-response to low-energy scattered photons, its recommended range of use does not include output factor measurements in large photon fields, although this effect can be compensated by a thin auxiliary lead shield
Pignol, Jean-Philippe; Keller, Brian M.
2009-05-01
Our team has previously published that submegavoltage photons could significantly improve the radiological penumbra for small size radiation fields. The present work uses Monte Carlo simulation to evaluate the contributions of secondary electrons and photon scatter to the penumbra region for various field sizes (5, 10, 20, and 40 mm in diameters) and for various monoenergetic photon beams (200, 400, 600, 800, 1000, and 2000 keV, and a standard 6 MV beam), minimizing geometrical and transmission penumbra. For field sizes less than 2 cm in diameter, photon scatter is negligible such that the secondary electrons are the main contributor to the radiological penumbra. Reducing the photon beam energy to the submegavoltage range reduces the range of secondary electrons and eventually improves the beam boundary sharpness. Provided that the geometrical penumbra and patient immobilization system are optimized, submegavoltage photon beams with effective photon energies in the 300 to 600 keV range, present significant advantages for multiple beam stereotactic irradiations of tumors less than 2 cm in diameter.
An efficient coupled polynomial interpolation scheme for shear mode sandwich beam finite element
Directory of Open Access Journals (Sweden)
Litesh N. Sulbhewar
Full Text Available An efficient piezoelectric sandwich beam finite element is presented here. It employs the coupled polynomial field interpolation scheme for field variables which incorporates electromechanical coupling at interpolation level itself; unlike conventional sandwich beam theory (SBT based formulations available in the literature. A variational formulation is used to derive the governing equations, which are used to establish the relationships between field variables. These relations lead to the coupled polynomial field descriptions of variables, unlike conventional SBT formulations which use assumed independent polynomials. The relative axial displacement is expressed only by coupled terms containing contributions from other mechanical and electrical variables, thus eliminating use of the transverse displacement derivative as a degree of freedom. A set of coupled shape function based on these polynomials has shown the improvement in the convergence characteristics of the SBT based formulation. This improvement in the performance is achieved with one nodal degree of freedom lesser than the conventional SBT formulations.
Finite element analysis of composite beam-to-column connection with cold-formed steel section
Firdaus, Muhammad; Saggaff, Anis; Tahir, Mahmood Md
2017-11-01
Cold-formed steel (CFS) sections are well known due to its lightweight and high structural performance which is very popular for building construction. Conventionally, they are used as purlins and side rails in the building envelopes of the industrial buildings. Recent research development on cold-formed steel has shown that the usage is expanded to the use in composite construction. This paper presents the modelling of the proposed composite connection of beam-to-column connection where cold-formed steel of lipped steel section is positioned back-to-back to perform as beam. Reinforcement bars is used to perform the composite action anchoring to the column and part of it is embedded into a slab. The results of the finite element and numerical analysis has showed good agreement. The results show that the proposed composite connection contributes to significant increase to the moment capacity.
Lee, Yang-Sub
A time-domain numerical algorithm for solving the KZK (Khokhlov-Zabolotskaya-Kuznetsov) nonlinear parabolic wave equation is developed for pulsed, axisymmetric, finite amplitude sound beams in thermoviscous fluids. The KZK equation accounts for the combined effects of diffraction, absorption, and nonlinearity at the same order of approximation. The accuracy of the algorithm is established via comparison with analytical solutions for several limiting cases, and with numerical results obtained from a widely used algorithm for solving the KZK equation in the frequency domain. The time domain algorithm is used to investigate waveform distortion and shock formation in directive sound beams radiated by pulsed circular piston sources. New results include predictions for the entire process of self-demodulation, and for the effect of frequency modulation on pulse envelope distortion. Numerical results are compared with measurements, and focused sources are investigated briefly.
Accelerator-based atomic physics experiments with photon and ion beams
International Nuclear Information System (INIS)
Johnson, B.M.; Jones, K.W.; Meron, M.
1984-01-01
Accelerator-based atomic physics experiments at Brookhaven presently use heavy-ion beams from the Dual MP Tandem Van de Graaff Accelerator Facility for atomic physics experiments of several types. Work is presently in progress to develop experiments which will use the intense photon beams which will be available in the near future from the ultraviolet (uv) and x-ray rings of the National Synchrotron Light Source (NSLS). Plans are described for experiments at the NSLS and an exciting development in instrumentation for heavy-ion experiments is summarized
A polarized {sup 3}He target for the photon beam at MAMI
Energy Technology Data Exchange (ETDEWEB)
Krimmer, J., E-mail: j.krimmer@ipnl.in2p3.fr [Institut fuer Physik, Johannes Gutenberg-Universitaet, Staudinger Weg 7, 55099 Mainz (Germany); Institut de Physique Nucleaire de Lyon, 4 rue Enrico Fermi, 69622 Villeurbanne cedex (France); Aguar Bartolome, P.; Ahrens, J. [Institut fuer Kernphysik, Johannes Gutenberg-Universitaet, Johann-Joachim-Becher-Weg 45, 55099 Mainz (Germany); Altieri, S. [INFN Sezione di Pavia, Pavia (Italy); Dipartimento di Fisica Nucleare e Teorica, Universita di Pavia, Pavia (Italy); Arends, H.J. [Institut fuer Kernphysik, Johannes Gutenberg-Universitaet, Johann-Joachim-Becher-Weg 45, 55099 Mainz (Germany); Heil, W.; Karpuk, S.; Otten, E.W. [Institut fuer Physik, Johannes Gutenberg-Universitaet, Staudinger Weg 7, 55099 Mainz (Germany); Pedroni, P. [INFN Sezione di Pavia, Pavia (Italy); Salhi, Z. [Institut fuer Physik, Johannes Gutenberg-Universitaet, Staudinger Weg 7, 55099 Mainz (Germany); Thomas, A. [Institut fuer Kernphysik, Johannes Gutenberg-Universitaet, Johann-Joachim-Becher-Weg 45, 55099 Mainz (Germany)
2011-08-21
A polarized {sup 3}He target has been installed for the first time inside the 4{pi} Crystal Ball detector at the tagged photon beam of the MAinz MIcrotron (MAMI). It has been demonstrated that the system works reliably and that the polarization losses during handling of the polarized gas are under control. Initial polarization values up to 70% and total relaxation times up to 20 h could be obtained during a first test beam time devoted to the measurement of the double polarized photoabsorption cross-section in the {Delta}(1232) baryon resonance region.
Mosalaei, Homeira; Karnas, Scott; Shah, Sheel; Van Doodewaard, Sharon; Foster, Tim; Chen, Jeff
2012-01-01
Electrons are ideal for treating shallow tumors and sparing adjacent normal tissue. Conventionally, electron beams are collimated by cut-outs that are time-consuming to make and difficult to adapt to tumor shape throughout the course of treatment. We propose that electron cut-outs can be replaced using photon multileaf collimator (MLC). Two major problems of this approach are that the scattering of electrons causes penumbra widening because of a large air gap, and available commercial treatment planning systems (TPSs) do not support MLC-collimated electron beams. In this study, these difficulties were overcome by (1) modeling electron beams collimated by photon MLC for a commercial TPS, and (2) developing a technique to reduce electron beam penumbra by adding low-energy intensity-modulated radiation therapy (IMRT) photons (4 MV). We used blocks to simulate MLC shielding in the TPS. Inverse planning was used to optimize boost photon beams. This technique was applied to a parotid and a central nervous system (CNS) clinical case. Combined photon and electron plans were compared with conventional plans and verified using ion chamber, film, and a 2D diode array. Our studies showed that the beam penumbra for mixed beams with 90 cm source to surface distance (SSD) is comparable with electron applicators and cut-outs at 100 cm SSD. Our mixed-beam technique yielded more uniform dose to the planning target volume and lower doses to various organs at risk for both parotid and CNS clinical cases. The plans were verified with measurements, with more than 95% points passing the gamma criteria of 5% in dose difference and 5 mm for distance to agreement. In conclusion, the study has demonstrated the feasibility and potential advantage of using photon MLC to collimate electron beams with boost photon IMRT fields. Copyright Â© 2012 American Association of Medical Dosimetrists. Published by Elsevier Inc. All rights reserved.
Automated Sample Exchange Robots for the Structural Biology Beam Lines at the Photon Factory
International Nuclear Information System (INIS)
Hiraki, Masahiko; Watanabe, Shokei; Yamada, Yusuke; Matsugaki, Naohiro; Igarashi, Noriyuki; Gaponov, Yurii; Wakatsuki, Soichi
2007-01-01
We are now developing automated sample exchange robots for high-throughput protein crystallographic experiments for onsite use at synchrotron beam lines. It is part of the fully automated robotics systems being developed at the Photon Factory, for the purposes of protein crystallization, monitoring crystal growth, harvesting and freezing crystals, mounting the crystals inside a hutch and for data collection. We have already installed the sample exchange robots based on the SSRL automated mounting system at our insertion device beam lines BL-5A and AR-NW12A at the Photon Factory. In order to reduce the time required for sample exchange further, a prototype of a double-tonged system was developed. As a result of preliminary experiments with double-tonged robots, the sample exchange time was successfully reduced from 70 seconds to 10 seconds with the exception of the time required for pre-cooling and warming up the tongs
A finite Hankel algorithm for intense optical beam propagation in saturable medium
International Nuclear Information System (INIS)
Bardin, C.; Babuel-Peyrissac, J.P.; Marinier, J.P.; Mattar, F.P.
1985-01-01
Many physical problems, especially light-propagation, that involve the Laplacian operator, are naturally connected with Fourier or Hankel transforms (in case of axial symmetry), which both remove the Laplacian term in the transformed space. Sometimes the analytical calculation can be handled at its end, giving a series or an integral representation of the solution. Otherwise, an analytical pre-treatment of the original equation may be done, leading to numerical computation techniques as opposed to self-adaptive stretching and rezoning techniques, which do not use Fourier or Hankel transforms. The authors present here some basic mathematical properties of infinite and finite Hankel transform, their connection with physics and their adaptation to numerical calculation. The finite Hankel transform is well-suited to numerical computation, because it deals with a finite interval, and the precision of the calculation can be easily controlled by the number of zeros of J 0 (x) to be taken. Moreover, they use a special quadrature formula which is well connected to integral conservation laws. The inconvenience of having to sum a series is reduced by the use of vectorized computers, and in the future will be still more reduced with parallel processors. A finite-Hankel code has been performed on CRAY-XMP in order to solve the propagation of a CW optical beam in a saturable absorber. For large diffractions or when a very small radial grid is required for the description of the optical field, this FHT algorithm has been found to perform better than a direct finite-difference code
Finite element analysis of beam-to-column joints in steel frames under cyclic loading
Directory of Open Access Journals (Sweden)
Elsayed Mashaly
2011-03-01
Full Text Available The aim of this paper is to present a simple and accurate three-dimensional (3D finite element model (FE capable of predicting the actual behavior of beam-to-column joints in steel frames subjected to lateral loads. The software package ANSYS is used to model the joint. The bolted extended-end-plate connection was chosen as an important type of beam–column joints. The extended-end-plate connection is chosen for its complexity in the analysis and behavior due to the number of connection components and their inheritable non-linear behavior. Two experimental tests in the literature were chosen to verify the finite element model. The results of both the experimental and the proposed finite element were compared. One of these tests was monotonically loaded, whereas the second was cyclically loaded. The finite element model is improved to enhance the defects of the finite element model used. These defects are; the long time need for the analysis and the inability of the contact element type to follow the behavior of moment–rotation curve under cyclic loading. As a contact element, the surface-to-surface element is used instead of node-to-node element to enhance the model. The FE results show good correlation with the experimental one. An attempt to improve a new technique for modeling bolts is conducted. The results show that this technique is supposed to avoid the defects above, give much less elements number and less solution time than the other modeling techniques.
A double expansion method for the frequency response of finite-length beams with periodic parameters
Ying, Z. G.; Ni, Y. Q.
2017-03-01
A double expansion method for the frequency response of finite-length beams with periodic distribution parameters is proposed. The vibration response of the beam with spatial periodic parameters under harmonic excitations is studied. The frequency response of the periodic beam is the function of parametric period and then can be expressed by the series with the product of periodic and non-periodic functions. The procedure of the double expansion method includes the following two main steps: first, the frequency response function and periodic parameters are expanded by using identical periodic functions based on the extension of the Floquet-Bloch theorem, and the period-parametric differential equation for the frequency response is converted into a series of linear differential equations with constant coefficients; second, the solutions to the linear differential equations are expanded by using modal functions which satisfy the boundary conditions, and the linear differential equations are converted into algebraic equations according to the Galerkin method. The expansion coefficients are obtained by solving the algebraic equations and then the frequency response function is finally determined. The proposed double expansion method can uncouple the effects of the periodic expansion and modal expansion so that the expansion terms are determined respectively. The modal number considered in the second expansion can be reduced remarkably in comparison with the direct expansion method. The proposed double expansion method can be extended and applied to the other structures with periodic distribution parameters for dynamics analysis. Numerical results on the frequency response of the finite-length periodic beam with various parametric wave numbers and wave amplitude ratios are given to illustrate the effective application of the proposed method and the new frequency response characteristics, including the parameter-excited modal resonance, doubling-peak frequency response
A Monte Carlo multiple source model applied to radiosurgery narrow photon beams
International Nuclear Information System (INIS)
Chaves, A.; Lopes, M.C.; Alves, C.C.; Oliveira, C.; Peralta, L.; Rodrigues, P.; Trindade, A.
2004-01-01
Monte Carlo (MC) methods are nowadays often used in the field of radiotherapy. Through successive steps, radiation fields are simulated, producing source Phase Space Data (PSD) that enable a dose calculation with good accuracy. Narrow photon beams used in radiosurgery can also be simulated by MC codes. However, the poor efficiency in simulating these narrow photon beams produces PSD whose quality prevents calculating dose with the required accuracy. To overcome this difficulty, a multiple source model was developed that enhances the quality of the reconstructed PSD, reducing also the time and storage capacities. This multiple source model was based on the full MC simulation, performed with the MC code MCNP4C, of the Siemens Mevatron KD2 (6 MV mode) linear accelerator head and additional collimators. The full simulation allowed the characterization of the particles coming from the accelerator head and from the additional collimators that shape the narrow photon beams used in radiosurgery treatments. Eight relevant photon virtual sources were identified from the full characterization analysis. Spatial and energy distributions were stored in histograms for the virtual sources representing the accelerator head components and the additional collimators. The photon directions were calculated for virtual sources representing the accelerator head components whereas, for the virtual sources representing the additional collimators, they were recorded into histograms. All these histograms were included in the MC code, DPM code and using a sampling procedure that reconstructed the PSDs, dose distributions were calculated in a water phantom divided in 20000 voxels of 1x1x5 mm 3 . The model accurately calculates dose distributions in the water phantom for all the additional collimators; for depth dose curves, associated errors at 2σ were lower than 2.5% until a depth of 202.5 mm for all the additional collimators and for profiles at various depths, deviations between measured
International Nuclear Information System (INIS)
Payne, M.G.; Garrett, W.R.
1983-01-01
Multiphoton ionization in the region near a three-photon resonance is treated for focused, plane-polarized Gaussian beams with diffraction-limited beam divergence. In this situation, a third-harmonic field is generated within the laser beam. At, and very near, three-photon resonance the driving rate for the upper-state probability amplitude due to one-photon absorption of third-harmonic light becomes nearly equal to the corresponding three-photon rate due to the laser field, but these effects are 180 0 out of phase. As a consequence of this cancellation between two pumping terms, the three-photon resonance line essentially disappears at moderate concentrations and the observed ionization has a line shape that is close to the phase-matching curve for third-harmonic generation. The ionization signal, near but not on the resonance, is due almost entirely to absorption of third-harmonic photons plus other laser photons; three-photon resonantly enhanced multiphoton ionization by the laser is much weaker. This is particularly true on the blue side of the three-photon resonance at detunings where phase matching occurs. The problem is treated quite generally with predictions of the full line shape for n-photon ionization and third-harmonic light generation near three-photon resonance, including the rather strong influences of positively dispersive buffer gases. We also show that the cancellation between the one-photon and the three-photon process is partially spoiled in the presence of a counterpropagating beam at the same frequency
Relative dosimetry of photon beam of 6 MV with a liquid ionization chamber
International Nuclear Information System (INIS)
Benitez Villegas, E. M.; Casado Villalon, F. J.; Martin-Cueto, J. A.; Caudepon Moreno, F.; Garcia Pareja, S.; Galan Montenegro, P.
2011-01-01
The increasing use of reduced size fields in the special techniques of treatment generates regions with high dose gradients. It therefore requires the use of detectors that present high spatial resolution. The aim of this study is to compare the dosimetric measurements obtained with a liquid ionization chamber PTW MicroLion recently acquired with other commonly used detectors for a photon beam of 6 MV linear electron accelerator Varian 600DBX.
International Nuclear Information System (INIS)
Hawileh, Rami A.; El-Maaddawy, Tamer A.; Naser, Mohannad Z.
2012-01-01
Highlights: ► A 3D nonlinear FE model is developed of RC deep beams with web openings. ► We used cohesion elements to simulate bond. ► The developed FE model is suitable for analysis of such complex structures. -- Abstract: This paper aims to develop 3D nonlinear finite element (FE) models for reinforced concrete (RC) deep beams containing web openings and strengthened in shear with carbon fiber reinforced polymer (CFRP) composite sheets. The web openings interrupted the natural load path either fully or partially. The FE models adopted realistic materials constitutive laws that account for the nonlinear behavior of materials. In the FE models, solid elements for concrete, multi-layer shell elements for CFRP and link elements for steel reinforcement were used to simulate the physical models. Special interface elements were implemented in the FE models to simulate the interfacial bond behavior between the concrete and CFRP composites. A comparison between the FE results and experimental data published in the literature demonstrated the validity of the computational models in capturing the structural response for both unstrengthened and CFRP-strengthened deep beams with openings. The developed FE models can serve as a numerical platform for performance prediction of RC deep beams with openings strengthened in shear with CFRP composites.
Experiences of calibration in photon beams for the personal dose equivalent
International Nuclear Information System (INIS)
Lindborg, L.; Gullberg, O.
1994-01-01
The calibration quantity shall, according to ICRU, be the personal dose equivalent, H p (d), in a phantom having the composition of the ICRU tissue and the same shape and size as the recommended PMMA calibration phantom, 30 x 30 x 15 cm 3 . There exist differences in backscattering between PMMA and tissue that for certain photon energies could be of importance. This could either be treated as a systematic uncertainty or be incorporated in the definition. However, monoenergetic beams seldom appear in reality and the difference in backscatter is not thought to be important. The calibration quantity for photons was chosen as the absorbed dose to ICRU tissue (times a quality factor 1) at 10 mm depth in a tissue-equivalent slab phantom. In Sweden 13 different services run personal dosimetry. Their initial hesitation about the change of quantity disappeared after testing their photon energy responses. It was found that most TLD systems were measuring the new quantity better than the old one and that the film systems needed only minor corrections. Most TLD systems now report 5% larger dose equivalents for the same irradiation in a photon beam from a 137 Cs source. (author)
Photon beam asymmetry Σ for η and η′ photoproduction from the proton
Directory of Open Access Journals (Sweden)
P. Collins
2017-08-01
Full Text Available Measurements of the linearly-polarized photon beam asymmetry Σ for photoproduction from the proton of η and η′ mesons are reported. A linearly-polarized tagged photon beam produced by coherent bremsstrahlung was incident on a cryogenic hydrogen target within the CEBAF Large Acceptance Spectrometer. Results are presented for the γp→ηp reaction for incident photon energies from 1.070 to 1.876 GeV, and from 1.516 to 1.836 GeV for the γp→η′p reaction. For γp→ηp, the data reported here considerably extend the range of measurements to higher energies, and are consistent with the few previously published measurements for this observable near threshold. For γp→η′p, the results obtained are consistent with the few previously published measurements for this observable near threshold, but also greatly expand the incident photon energy coverage for that reaction. Initial analysis of the data reported here with the Bonn–Gatchina model strengthens the evidence for four nucleon resonances – the N(18951/2−, N(19003/2+, N(21001/2+ and N(21203/2− resonances – which presently lack the “four-star” status in the current Particle Data Group compilation, providing examples of how these new measurements help refine models of the photoproduction process.
Photon beam asymmetry Σ for η and η‧ photoproduction from the proton
Collins, P.; Ritchie, B. G.; Dugger, M.; Anisovich, A. V.; Döring, M.; Klempt, E.; Nikonov, V. A.; Rönchen, D.; Sadasivan, D.; Sarantsev, A.; Adhikari, K. P.; Akbar, Z.; Amaryan, M. J.; Anefalos Pereira, S.; Avakian, H.; Ball, J.; Balossino, I.; Bashkanov, M.; Battaglieri, M.; Bedlinskiy, I.; Biselli, A. S.; Briscoe, W. J.; Brooks, W. K.; Burkert, V. D.; Cao, Frank Thanh; Carman, D. S.; Celentano, A.; Chandavar, S.; Charles, G.; Chetry, T.; Ciullo, G.; Clark, L.; Colaneri, L.; Cole, P. L.; Compton, N.; Contalbrigo, M.; Cortes, O.; Crede, V.; D'Angelo, A.; Dashyan, N.; De Vita, R.; De Sanctis, E.; Deur, A.; Djalali, C.; Dupre, R.; Egiyan, H.; El Alaoui, A.; El Fassi, L.; Elouadrhiri, L.; Eugenio, P.; Fanchini, E.; Fedotov, G.; Filippi, A.; Fleming, J. A.; Ghandilyan, Y.; Gilfoyle, G. P.; Giovanetti, K. L.; Girod, F. X.; Glazier, D. I.; Gleason, C.; Golovatch, E.; Gothe, R. W.; Griffioen, K. A.; Guo, L.; Hafidi, K.; Hakobyan, H.; Hanretty, C.; Harrison, N.; Heddle, D.; Hicks, K.; Holtrop, M.; Hughes, S. M.; Ilieva, Y.; Ireland, D. G.; Ishkhanov, B. S.; Isupov, E. L.; Jenkins, D.; Jo, H. S.; Joosten, S.; Keller, D.; Khachatryan, G.; Khachatryan, M.; Khandaker, M.; Kim, A.; Kim, W.; Klein, A.; Klein, F. J.; Kubarovsky, V.; Lanza, L.; Lenisa, P.; Livingston, K.; MacGregor, I. J. D.; Markov, N.; McKinnon, B.; Meyer, C. A.; Mirazita, M.; Mokeev, V.; Montgomery, R. A.; Movsisyan, A.; Munoz Camacho, C.; Murdoch, G.; Nadel-Turonski, P.; Niccolai, S.; Niculescu, G.; Niculescu, I.; Osipenko, M.; Ostrovidov, A. I.; Paolone, M.; Paremuzyan, R.; Park, K.; Pasyuk, E.; Phelps, W.; Pisano, S.; Pogorelko, O.; Price, J. W.; Prok, Y.; Protopopescu, D.; Raue, B. A.; Ripani, M.; Rizzo, A.; Rosner, G.; Roy, P.; Sabatié, F.; Salgado, C.; Schumacher, R. A.; Sharabian, Y. G.; Skorodumina, Iu.; Smith, G. D.; Sokhan, D.; Sparveris, N.; Stepanyan, S.; Strakovsky, I. I.; Strauch, S.; Taiuti, M.; Tian, Ye; Torayev, B.; Ungaro, M.; Voskanyan, H.; Voutier, E.; Walford, N. K.; Wei, X.; Zachariou, N.; Zhang, J.
2017-08-01
Measurements of the linearly-polarized photon beam asymmetry Σ for photoproduction from the proton of η and η‧ mesons are reported. A linearly-polarized tagged photon beam produced by coherent bremsstrahlung was incident on a cryogenic hydrogen target within the CEBAF Large Acceptance Spectrometer. Results are presented for the γp → ηp reaction for incident photon energies from 1.070 to 1.876 GeV, and from 1.516 to 1.836 GeV for the γp →η‧ p reaction. For γp → ηp, the data reported here considerably extend the range of measurements to higher energies, and are consistent with the few previously published measurements for this observable near threshold. For γp →η‧ p, the results obtained are consistent with the few previously published measurements for this observable near threshold, but also greatly expand the incident photon energy coverage for that reaction. Initial analysis of the data reported here with the Bonn-Gatchina model strengthens the evidence for four nucleon resonances - the N (1895) 1 /2-, N (1900) 3 /2+, N (2100) 1 /2+ and N (2120) 3 /2- resonances - which presently lack the ;four-star; status in the current Particle Data Group compilation, providing examples of how these new measurements help refine models of the photoproduction process.
International Nuclear Information System (INIS)
Moon, Jinsoo; Won Jang, Kyoung; Jae Yoo, Wook; Han, Ki-Tek; Park, Jang-Yeon; Lee, Bongsoo
2012-01-01
In this study, we fabricated a one-dimensional scintillating fiber-optic dosimeter, which consists of 9 scintillating fiber-optic dosimeters, septa, and PMMA blocks for measuring surface and percentage depth doses of a therapeutic photon beam. Each dosimeter embedded in the 1-D scintillating fiber-optic dosimeter is composed of square type organic scintillators and plastic optical fibers. Also black PVC films are used as septa to minimize cross-talk between the scintillating fiber-optic dosimeters. To construct a dosimeter system, a 1-D scintillating fiber-optic dosimeter and a CMOS image sensor were combined with 20 m-length plastic optical fibers. Using the dosimeter system, we measured surface and percentage depth doses of 6 and 15 MV photon beams and compared the results with those of EBT films and an ionization chamber. - Highlights: ► Fabrication of a one-dimensional scintillating fiber-optic dosimeter. ► The one-dimensional scintillating fiber-optic dosimeter has 9 scintillating fiber-optic dosimeters. ► Measurements of surface and percentage depth doses of a therapeutic photon beam. ► The results were compared with those of EBT films and an ionization chamber.
Pattern generation using axicon lens beam shaping in two-photon polymerisation
International Nuclear Information System (INIS)
Bhuian, B.; Winfield, R.J.; O'Brien, S.; Crean, G.M.
2007-01-01
The fabrication of three-dimensional microstructures by two-photon polymerisation has been widely reported as a viable route to the development of photonic crystals, rotors, bridges and other complex artefacts requiring nanoscale resolution. Conventionally, single point serial writing is used to write the structures but recently multipoint beam delivery using beam division optics has been reported as a method of introducing parallel processing. In this paper we present an alternative and novel approach using an axicon lens to give profiled beam delivery. This enables complete three-dimensional annular structure fabrication without the use of scanning stages. In addition, the concept of axicon delivery is developed further to investigate three-dimensional structure as a function of axicon geometry. A Ti:sapphire laser, with wavelength 795 nm, 80 MHz repetition rate, 100 fs pulse duration and an average power of 700 mW, was used to initiate two-photon polymerisation. The axicon was used, in combination with a 100x microscope objective, to form representative three-dimensional structures based on the annular cell with varying diameter. The structures were written in a Zr-loaded resin prepared on a glass substrate using dip coating deposition of a Zr/PMMA hybrid prepared by the sol-gel method. Annuli with diameters up to 50 μm were characterised in terms of topography and surface roughness using SEM and Zygo interferometer. The writing technique was also extended to fabrication of stacked structures
Energy Technology Data Exchange (ETDEWEB)
Caporali, C; Guerra, A S; Laitano, R F; Pimpinella, M [ENEA-Casaccia, Inst. Nazionale di Meterologia delle Radiazioni Ionizzanti, Rome (Italy). Dipt. Ambiente
1996-08-01
Ion chambers calibrated in terms of absorbed dose to water need an additional factor conventionally designed by k{sub Q} in order to determine the absorbed dose. The quantity k{sub Q} depends on beam quality and chamber characteristics. Rogers and Andreo provided calculations of the k{sub Q} factors for most commercially available ionization chambers for clinical dosimetry. Experimental determinations of the k{sub Q} factors for a number of cylindrical ion chambers have been made and are compared with the calculated values so far published. Measurements were made at 6 MV and 15 MV clinical photon beams at a point in water phantom where the ion chambers and a Fricke dosimeter were alternatively irradiated. The uncertainty on the experimental k{sub Q} factors resulted about {+-} 0.6%. The theoretical and experimental k{sub Q} values are in fairly good agreement. (author). 12 refs, 3 tabs.
A model based on the Fermi-Dirac distribution to determine the dose profile of a photon beam
International Nuclear Information System (INIS)
Guerrero, R.; Silva, P.; Gutt, F.; Diaz, J.
1998-01-01
The objective of this work is to propose a new model based in the Fermi-Dirac distribution, in which it is considered that the photon beams possess a particles behavior when they interact with the matter. (Author)
A finite element modeling of a multifunctional hybrid composite beam with viscoelastic materials
Wang, Ya; Inman, Daniel J.
2013-04-01
The multifunctional hybrid composite structure studied here consists of a ceramic outer layer capable of withstanding high temperatures, a functionally graded ceramic layer combining shape memory alloy (SMA) properties of NiTi together with Ti2AlC (called Graded Ceramic/Metal Composite, or GCMeC), and a high temperature sensor patch, followed by a polymer matrix composite laced with vascular cooling channels all held together with various epoxies. Due to the recoverable nature of SMA and adhesive properties of Ti2AlC, the damping behavior of the GCMeC is largely viscoelastic. This paper presents a finite element formulation for this multifunctional hybrid structure with embedded viscoelastic material. In order to implement the viscoelastic model into the finite element formulation, a second order three parameter Golla-Hughes-McTavish (GHM) method is used to describe the viscoelastic behavior. Considering the parameter identification, a strategy to estimate the fractional order of the time derivative and the relaxation time is outlined. The curve-fitting aspects of both GHM and ADF show good agreement with experimental data obtained from dynamic mechanics analysis. The performance of the finite element of the layered multifunctional beam is verified through experimental model analysis.
Sum rules for multi-photon spectroscopy of ions in finite symmetry
International Nuclear Information System (INIS)
Kibler, M.; Daoud, M.
1993-05-01
Models describing one- and two-photon transitions for ions in crystalline environments are unified and extended to the case of parity-allowed and parity-forbidden p-photon transitions. The number of independent parameters for characterizing the polarization dependence is shown to depend on an ensemble of properties and rules which combine symmetry considerations and physical models. (author) 26 refs
Aglieri-Rinella, G; Van Lysebetten, A; Piedigrossi, D; Wyllie, K; Bellunato, T F; Calvi, M; Matteuzzi, C; Musy, M; Perego, D L; Somerville, L P; Newby, C; Easo, S; Wotton, S
2006-01-01
A test-beam study was performed at CERN with a Ring Imaging Cherenkov (RICH) prototype using three pixel Hybrid Photon Detectors. Results on the photon yield and Cherenkov angle resolution are presented here, for the Aerogel radiator and also for reference runs taken with Nitrogen radiator.
Directory of Open Access Journals (Sweden)
N. Dadashzadeh
2013-09-01
Full Text Available Ultra-short pulse is a promising technology for achieving ultra-high data rate transmission which is required to follow the increased demand of data transport over an optical communication system. Therefore, the propagation of such type of pulses and the effects that it may suffer during its transmission through an optical waveguide has received a great deal of attention in the recent years. We provide an overview of recent theoretical developments in a numerical modeling of Maxwell's equations to analyze the propagation of short laser pulses in photonic structures. The process of short light pulse propagation through 2D periodic and quasi-periodic photonic structures is simulated based on Finite-Difference Time-Domain calculations of Maxwell’s equations.
International Nuclear Information System (INIS)
Eastham, P. R.; Littlewood, P. B.
2006-01-01
We consider polariton condensation in a generalized Dicke model, describing a single-mode cavity containing quantum dots, and extend our previous mean-field theory to allow for finite-size fluctuations. Within the fluctuation-dominated regime the correlation functions differ from their (trivial) mean-field values. We argue that the low-energy physics of the model, which determines the photon statistics in this fluctuation-dominated crossover regime, is that of the (quantum) anharmonic oscillator. The photon statistics at the crossover are different in the high-temperature and low-temperature limits. When the temperature is high enough for quantum effects to be neglected we recover behavior similar to that of a conventional laser. At low enough temperatures, however, we find qualitatively different behavior due to quantum effects
Correlation of photon beam motion with vacuum chamber cooling on the NSLS x-ray ring
International Nuclear Information System (INIS)
Johnson, E.D.; Fauchet, A.M.; Zhang, Xiaohao.
1991-01-01
The NSLS X-ray ring exhibits a direct correlation between photon beam motion, and distortion of the ring vacuum chamber induced by fluctuations in the cooling system. We have made long term measurements of photon beam vertical position, accelerator vacuum chamber motion, process water temperatures, and angular motions of the magnets around one superperiod of the NSLS x-ray ring. Short term transients in water temperature cause deflection of the ring vacuum chamber which have in turn been shown to induce very small angular rotations of the magnets, on the order of 10 micro-radians. A larger and more difficult to correct effect is the drift in beam position over the course of a fill. This problem has been shown to be related to the thermal gradients that develop across the vacuum chamber which, as a consequence of the configuration of the chamber cooling, depend upon stored current. Orbit simulations based upon the measured rotations are in agreement with the observed beam motions, and reveal that certain patterns of correlated motions of the magnets can produce much larger errors than random motion or concerted motion of all the magnets. During the course of these measurements global orbit feedback was installed, and found to significantly reduce the orbit errors which could not be corrected at their source
International Nuclear Information System (INIS)
Mazal, A.; Gaboriauid, G.; Zefkili, S.; Rosenwald, J.C.; Boutaudon, S.; Pontvert, D.
1999-01-01
The irradiation of small targets in the brain in a singe fraction (radiosurgery) or with a fractionated approach (stereotactic radiosurgery) with small beams of photons requires specific conditions to measure and to model the dosimetric data needed for treatment planning. In this work we present the method and materials adopted in our institution since 1988 to perform the dosimetry of high energy (6-23) circular photon beams with diameters ranging from 10 to 40 mm at the isocenter of linear accelerators, and its evolution as new dosimetric material became commercially available. in circular ionization chambers of small dimensions. We want to answer the following questions: Which are the minimal basic data needed to model small circular beams of high energy photons? Can we extrapolate or convert data from conventional data of larger beams? Which are the detectors well adapted for these kind of measurements and for which range of beam sizes?
Dosimetric properties characterization of silicon diodes used in photon beam radiotherapy
International Nuclear Information System (INIS)
Bizetto, Cesar Augusto
2013-01-01
In the current work it was studied the performance of epitaxial (EPI) and float zone (FZ) silicon diodes as on-line dosimeters for megavoltage (EPI diode) and orthovoltage (EPI and FZ diode) photon beam radiotherapy. In order to be used as dosimeters the diodes were enclosed in black polymethylmethacrylate (PMMA) probes. The devices were then connected, on photovoltaic mode, to an electrometer Keithley® 6517B to allow measurements of the photocurrent. The irradiations were performed with 6 and 18 MV photon beams (Siemens Primus® linear accelerator), 6 and 15 MV (Novalis TX®) and 10, 25, 30 and 50 kV of a Pantak / Seifert X ray radiation device. During the measurements with the Siemens Primus the diodes were held between PMMA plates placed at 10.0 cm depth. When using Novalis TX® the devices were held between solid water plates placed at 50 cm depth. In both cases the diodes were centered in a radiation field of 10 x 10 cm 2 , with the source-to-surface distance (SSD) kept at 100 cm. In measurements with orthovoltage photon beams the diodes were placed 50.0 cm from the tube in a radiation field of 8 cm diameter. The dose-rate dependency was studied for 6 and 15 MV (varying the dose-rate from 100 to 600 monitor units per minute) and for the 50 kV beam by varying the current tube from 2 to 20 mA. All devices showed linear response with dose rate and, within uncertainties the charge collected is independent of dose rate. The current signals induced showed good instantaneous repeatability of the diodes, characterized by coefficients of variation of current (CV) smaller than 1.14% (megavoltage beams) and 0.15% for orthovoltage beams and coefficients of variation of charge (CV) smaller than 1.84% (megavoltage beams) and 1.67% (orthovoltage beams). The dose response curves were quite linear with linear correlation coefficients better than 0.9999 for all diodes. (author)
Directory of Open Access Journals (Sweden)
Najeeb ur Rahman
Full Text Available A one-dimensional finite element model for buckling analysis of hybrid piezoelectric beams under electromechanical load is presented in this work. The coupled zigzag theory is used for making the model. The inplane displacement is approximated as a combination of a global third order variation across the thickness with an additional layer wise linear variation. The longitudinal electric field is also taken into account. The deflection field is approximated to account for the transverse normal strain induced by electric fields. Two nodded elements with four mechanical and a variable number of electric degrees of freedom at each node are considered. To meet the convergence requirements for weak integral formulation, cubic Hermite interpolation function is used for deflection and electric potential at the sub-layers and linear interpolation function is used for axial displacement and shear rotation. The expressions for the variationally consistent stiffness matrix and load vector are derived and evaluated in closed form using exact integration. The present 1D-FE formulation of zigzag theory is validated by comparing the results with the analytical solution for simply-supported beam and 2D-FE results obtained using ABAQUS. The finite element model is free of shear locking. The critical buckling parameters are obtained for clamped-free and clamped-clamped hybrid beams. The obtained results are compared with the 2D-FE results to establish the accuracy of the zigzag theory for above boundary conditions. The effect of lamination angle on critical buckling load is also studied.
Finite Element Modelling for Static and Free Vibration Response of Functionally Graded Beam
Directory of Open Access Journals (Sweden)
Ateeb Ahmad Khan
Full Text Available Abstract A 1D Finite Element model for static response and free vibration analysis of functionally graded material (FGM beam is presented in this work. The FE model is based on efficient zig-zag theory (ZIGT with two noded beam element having four degrees of freedom at each node. Linear interpolation is used for the axial displacement and cubic hermite interpolation is used for the deflection. Out of a large variety of FGM systems available, Al/SiC and Ni/Al2O3 metal/ceramic FGM system has been chosen. Modified rule of mixture (MROM is used to calculate the young's modulus and rule of mixture (ROM is used to calculate density and poisson's ratio of FGM beam at any point. The MATLAB code based on 1D FE zigzag theory for FGM elastic beams is developed. A 2D FE model for the same elastic FGM beam has been developed using ABAQUS software. An 8-node biquadratic plane stress quadrilateral type element is used for modeling in ABAQUS. Three different end conditions namely simply-supported, cantilever and clamped- clamped are considered. The deflection, normal stress and shear stress has been reported for various models used. Eigen Value problem using subspace iteration method is solved to obtain un-damped natural frequencies and the corresponding mode shapes. The results predicted by the 1D FE model have been compared with the 2D FE results and the results present in open literature. This proves the correctness of the model. Finally, mode shapes have also been plotted for various FGM systems.
Symmetry adaptation and two-photon spectroscopy of ions in molecular or solid-state finite symmetry
International Nuclear Information System (INIS)
Kibler, M.; Daoud, M.
1991-01-01
Finite symmetry adaptation techniques are applied to the determination of the intensity strength of two-photon transitions for ions with one partly-filled shell nl in crystalline environments of symmetry G. The case of intra-configurational (nl N →nl N ) transitions as well as the case of inter-configurational (nl N →nl N-1 n'l' with (-) l+( l')=-1) transitions is treated. In both cases, the Wigner-Racah algebra of the chain O(3) contains G allows to extract the polarization dependence from the intensity. The reported results are valid for any strength of the crystalline field. (author) 19 refs
Dosimetric properties of radiophotoluminescent glass detector in low-energy photon beams.
Kadoya, Noriyuki; Shimomura, Kouhei; Kitou, Satoshi; Shiota, Yasuo; Fujita, Yukio; Dobashi, Suguru; Takeda, Ken; Jingu, Keiichi; Matsushita, Haruo; Namito, Yoshihito; Ban, Syuichi; Koyama, Syuji; Tabushi, Katsuyoshi
2012-10-01
A radiophotoluminescent glass rod dosimeter (RGD) has recently become commercially available. It is being increasingly used for dosimetry in radiotherapy to measure the absorbed dose including scattered low-energy photons on the body surface of a patient and for postal dosimetry audit. In this article, the dosimetric properties of the RGD, including energy dependence of the dose response, reproducibly, variation in data obtained by the RGD for each energy, and angular dependence in low-energy photons, are discussed. An RGD (GD-301, Asahi Techno Glass Corporation, Shizuoka, Japan) was irradiated with monochromatic low-energy photon beams generated by synchrotron radiation at Photon Factory, High Energy Accelerator Research Organization (KEK). The size of GD-301 was 1.5 mm in diameter and 8.5 mm in length and the active dose readout volume being 1 mm diameter and 0.6 mm depth located 0.7 mm from the end of the detector. The energy dependence of the dose response and reproducibility and variation were investigated for RGDs irradiated with a plastic holder and those irradiated without the plastic holder. Response of the RGD was obtained by not only conventional single field irradiation but also bilateral irradiation. Angular dependence of the RGD was measured in the range of 0°-90° for 13, 17, 40, and 80 keV photon beams by conventional single field irradiation. The dose responses had a peak at around 40 keV. For the energy range of less than 25 keV, all dose response curves steeply decreased in comparison with the ratio of mass energy absorption coefficient of the RGD to that of air. As for the reproducibility and variation in data obtained by the RGD, the coefficient of variance increased with decrease in photon energy. Furthermore, the variation for bilateral irradiation was less than that for single field irradiation. Regarding angular dependence of the RGD, for energies of 13 and 17 keV, the response decreased with increase in the irradiation angle, and the
Cerenkov light spectrum in an optical fiber exposed to a photon or electron radiation therapy beam
International Nuclear Information System (INIS)
Lambert, Jamil; Yin Yongbai; McKenzie, David R.; Law, Sue; Suchowerska, Natalka
2009-01-01
A Cerenkov signal is generated when energetic charged particles enter the core of an optical fiber. The Cerenkov intensity can be large enough to interfere with signals transmitted through the fiber. We determine the spectrum of the Cerenkov background signal generated in a poly(methyl methacrylate) optical fiber exposed to photon and electron therapeutic beams from a linear accelerator. This spectral measurement is relevant to discrimination of the signal from the background, as in scintillation dosimetry using optical fiber readouts. We find that the spectrum is approximated by the theoretical curve after correction for the wavelength dependent attenuation of the fiber. The spectrum does not depend significantly on the angle between the radiation beam and the axis of the fiber optic but is dependent on the depth in water at which the fiber is exposed to the beam.
Photon beam audits for radiation therapy clinics: A pilot mailed dosemeter study in Turkey
International Nuclear Information System (INIS)
Yegingil, Z.; DeWerd, L. A.; Davis, S. D.; Hammer, C.; Kunugi, K.
2012-01-01
A thermoluminescent dosemeter (TLD) mailed dose audit programme was performed at five radiotherapy clinics in Turkey. The intercomparison was organised by the Univ. of Wisconsin Radiation Calibration Laboratory (UWRCL), which was responsible for the technical aspects of the study including reference irradiations, distribution, collection and evaluation. The purpose of these audits was to perform an independent dosimetry check of the radiation beams using TLDs sent by mail. Acrylic holders, each with five TLD chips inside and instructions for their irradiation to specified absorbed dose to water of 2 Gy, were mailed to all participating clinics. TLD irradiations were performed with a 6 MV linear accelerator and 60 Co photon beams. The deviations from the TL readings of UWRCL were calculated. Discrepancies inside the limits of ±5 % between the participant-stated dose, and the TLD-measured dose were considered acceptable. One out of 10 beams checked was outside this limit, with a difference of 5.8 %. (authors)
Clinical application of intensity and energy modulated radiotherapy with photon and electron beams
International Nuclear Information System (INIS)
Xiangkui Mu
2005-01-01
In modern, advanced radiotherapy (e.g. intensity modulated photon radiotherapy, IMXT) the delivery time for each fraction becomes prolonged to 10-20 minutes compared with the conventional, commonly 2-5 minutes. The biological effect of this prolongation is not fully known. The large number of beam directions in IMXT commonly leads to a large integral dose in the patient. Electrons would reduce the integral dose but are not suitable for treating deep-seated tumour, due to their limited penetration in tissues. By combining electron and photon beams, the dose distributions may be improved compared with either used alone. One obstacle for using electron beams in clinical routine is that there is no available treatment planning systems that optimise electron beam treatments in a similar way as for IMXT. Protons have an even more pronounced dose fall-off, larger penetration depth and less penumbra widening than electrons and are therefore more suitable for advanced radiotherapy. However, proton facilities optimised for advanced radiotherapy are not commonly available. In some instances electron beams may be an acceptable surrogate. The first part of this study is an experimental in vitro study where the situation in a tumour during fractionated radiotherapy is simulated. The effect of the prolonged fraction time is compared with the predictions by radiobiological models. The second part is a treatment planning study to analyse the mixing of electron and photon beams for at complex target volume in comparison with IMXT. In the next step a research version of an electron beam optimiser was used for the improvement of treatment plans. The aim was to develop a method for translating crude energy and intensity matrices for optimised electrons into a deliverable treatment plan without destroying the dose distribution. In the final part, different methods of treating the spinal canal in medulloblastoma were explored in a treatment planning study that was evaluated with
Energy Technology Data Exchange (ETDEWEB)
Yaddanapudi, S; Cai, B; Sun, B; Noel, C; Goddu, S; Mutic, S [Washington University School of Medicine, Saint Louis, MO (United States)
2015-06-15
Purpose: Electronic portal imaging devices (EPIDs) have proven to be useful for measuring several parameters of interest in linear accelerator (linac) quality assurance (QA). The purpose of this project was to evaluate the feasibility of using EPIDs for determining linac photon beam energies. Methods: Two non-clinical Varian TrueBeam linacs (Varian Medical Systems, Palo Alto, CA) with 6MV and 10MV photon beams were used to perform the measurements. The linacs were equipped with an amorphous silicon based EPIDs (aSi1000) that were used for the measurements. We compared the use of flatness versus percent depth dose (PDD) for predicting changes in linac photon beam energy. PDD was measured in 1D water tank (Sun Nuclear Corporation, Melbourne FL) and the profiles were measured using 2D ion-chamber array (IC-Profiler, Sun Nuclear) and the EPID. Energy changes were accomplished by varying the bending magnet current (BMC). The evaluated energies conformed with the AAPM TG142 tolerance of ±1% change in PDD. Results: BMC changes correlating with a ±1% change in PDD corresponded with a change in flatness of ∼1% to 2% from baseline values on the EPID. IC Profiler flatness values had the same correlation. We observed a similar trend for the 10MV beam energy changes. Our measurements indicated a strong correlation between changes in linac photon beam energy and changes in flatness. For all machines and energies, beam energy changes produced change in the uniformity (AAPM TG-142), varying from ∼1% to 2.5%. Conclusions: EPID image analysis of beam profiles can be used to determine linac photon beam energy changes. Flatness-based metrics or uniformity as defined by AAPM TG-142 were found to be more sensitive to linac photon beam energy changes than PDD. Research funding provided by Varian Medical Systems. Dr. Sasa Mutic receives compensation for providing patient safety training services from Varian Medical Systems, the sponsor of this study.
Zhang, Qing; Li, Maozhong; Liao, Tingdi; Cui, Xudong
2018-03-01
Under the trend of miniaturization and reduction of system complexity, conventional bulky photonic elements are expected to be replaced by new compact and ultrathin dielectric metasurface elements. In this letter, we propose an αTiO2 dielectric metasurface (DM) platform that could be exploited to design high efficiency wave-front control devices at visible wavelength. Combining with fundamental principles and full wave simulations (Lumerical FDTD 3D solver ®), we successfully realize four DM devices, such as anomalous beam deflectors, polarization insensitive metalens, wave plates and polarization beam splitters. All these devices can achieve high transmission efficiencies (larger than 80%). Among them, the anomalous refraction beam deflectors can bend light propagation to any desired directions; the polarization insensitive metalens maintains diffraction limited focus (focal spot as small as 0.67 λ); the quarter-wave and half-wave plates have broadband working wavelengths from 550 to 1000 nm; and the polarization beam splitter can split an arbitrarily polarized incident beam into two orthogonally polarized beams, the TM components is deflected to the right side, and the TE components is deflected to the left side. These devices may find applications in the areas of imaging, polarization control, spectroscopy, and on-chip optoelectronic systems etc., and our studies may richen the design of all-dielectric optical elements at visible wavelength.
Ultracompact 1×4 TM-polarized beam splitter based on photonic crystal surface mode.
Jiang, Bin; Zhang, Yejin; Wang, Yufei; Liu, Anjin; Zheng, Wanhua
2012-05-01
We provide an improved surface-mode photonic crystal (PhC) T-junction waveguide, combine it with an improved PhC bandgap T-junction waveguide, and then provide an ultracompact 1×4 TM-polarized beam splitter. The energy is split equally into the four output waveguides. The maximal transmission ratio of each output waveguide branch equals 24.7%, and the corresponding total transmission ratio of the ultracompact 1×4 beam splitter equals 98.8%. The normalized frequency of maximal transmission ratio is 0.397(2πc/a), and the bandwidth of the ultracompact 1×4 TM-polarized beam splitter is 0.0106(2πc/a). To the best of our knowledge, this is the first time such a high-efficiency 1×4 beam splitter exploiting the nonradiative surface mode as a guided mode has been proposed. Although we only employed a 1×4 beam splitter, our design can easily be extended to other 1×n beam splitters.
Cone beam tomography of the heart using single-photon emission-computed tomography
International Nuclear Information System (INIS)
Gullberg, G.T.; Christian, P.E.; Zeng, G.L.; Datz, F.L.; Morgan, H.T.
1991-01-01
The authors evaluated cone beam single-photon emission-computed tomography (SPECT) of the heart. A new cone beam reconstruction algorithm was used to reconstruct data collected from short scan acquisitions (of slightly more than 180 degrees) of a detector anteriorally traversing a noncircular orbit. The less than 360 degrees acquisition was used to minimize the attenuation artifacts that result from reconstructing posterior projections of 201T1 emissions from the heart. The algorithm includes a new method for reconstructing truncated projections of background tissue activity that eliminates reconstruction ring artifacts. Phantom and patient results are presented which compare a high-resolution cone beam collimator (50-cm focal length; 6.0-mm full width at half maximum [FWHM] at 10 cm) to a low-energy general purpose (LEGP) parallel hole collimator (8.2-mm FWHM at 10 cm) which is 1.33 times more sensitive. The cone beam tomographic results are free of reconstruction artifacts and show improved spatial and contrast resolution over that obtained with the LEGP parallel hole collimator. The limited angular sampling restrictions and truncation problems associated with cone beam tomography do not deter from obtaining diagnostic information. However, even though these preliminary results are encouraging, a thorough clinical study is still needed to investigate the specificity and sensitivity of cone beam tomography
Energy Technology Data Exchange (ETDEWEB)
Garcia-Pareja, S.; Galan, P.; Manzano, F.; Brualla, L.; Lallena, A. M. [Servicio de Radiofisica Hospitalaria, Hospital Regional Universitario ' ' Carlos Haya' ' , Avda. Carlos Haya s/n, E-29010 Malaga (Spain); Unidad de Radiofisica Hospitalaria, Hospital Xanit Internacional, Avda. de los Argonautas s/n, E-29630 Benalmadena (Malaga) (Spain); NCTeam, Strahlenklinik, Universitaetsklinikum Essen, Hufelandstr. 55, D-45122 Essen (Germany); Departamento de Fisica Atomica, Molecular y Nuclear, Universidad de Granada, E-18071 Granada (Spain)
2010-07-15
Purpose: In this work, the authors describe an approach which has been developed to drive the application of different variance-reduction techniques to the Monte Carlo simulation of photon and electron transport in clinical accelerators. Methods: The new approach considers the following techniques: Russian roulette, splitting, a modified version of the directional bremsstrahlung splitting, and the azimuthal particle redistribution. Their application is controlled by an ant colony algorithm based on an importance map. Results: The procedure has been applied to radiosurgery beams. Specifically, the authors have calculated depth-dose profiles, off-axis ratios, and output factors, quantities usually considered in the commissioning of these beams. The agreement between Monte Carlo results and the corresponding measurements is within {approx}3%/0.3 mm for the central axis percentage depth dose and the dose profiles. The importance map generated in the calculation can be used to discuss simulation details in the different parts of the geometry in a simple way. The simulation CPU times are comparable to those needed within other approaches common in this field. Conclusions: The new approach is competitive with those previously used in this kind of problems (PSF generation or source models) and has some practical advantages that make it to be a good tool to simulate the radiation transport in problems where the quantities of interest are difficult to obtain because of low statistics.
International Nuclear Information System (INIS)
Garcia-Pareja, S.; Galan, P.; Manzano, F.; Brualla, L.; Lallena, A. M.
2010-01-01
Purpose: In this work, the authors describe an approach which has been developed to drive the application of different variance-reduction techniques to the Monte Carlo simulation of photon and electron transport in clinical accelerators. Methods: The new approach considers the following techniques: Russian roulette, splitting, a modified version of the directional bremsstrahlung splitting, and the azimuthal particle redistribution. Their application is controlled by an ant colony algorithm based on an importance map. Results: The procedure has been applied to radiosurgery beams. Specifically, the authors have calculated depth-dose profiles, off-axis ratios, and output factors, quantities usually considered in the commissioning of these beams. The agreement between Monte Carlo results and the corresponding measurements is within ∼3%/0.3 mm for the central axis percentage depth dose and the dose profiles. The importance map generated in the calculation can be used to discuss simulation details in the different parts of the geometry in a simple way. The simulation CPU times are comparable to those needed within other approaches common in this field. Conclusions: The new approach is competitive with those previously used in this kind of problems (PSF generation or source models) and has some practical advantages that make it to be a good tool to simulate the radiation transport in problems where the quantities of interest are difficult to obtain because of low statistics.
García-Pareja, S; Galán, P; Manzano, F; Brualla, L; Lallena, A M
2010-07-01
In this work, the authors describe an approach which has been developed to drive the application of different variance-reduction techniques to the Monte Carlo simulation of photon and electron transport in clinical accelerators. The new approach considers the following techniques: Russian roulette, splitting, a modified version of the directional bremsstrahlung splitting, and the azimuthal particle redistribution. Their application is controlled by an ant colony algorithm based on an importance map. The procedure has been applied to radiosurgery beams. Specifically, the authors have calculated depth-dose profiles, off-axis ratios, and output factors, quantities usually considered in the commissioning of these beams. The agreement between Monte Carlo results and the corresponding measurements is within approximately 3%/0.3 mm for the central axis percentage depth dose and the dose profiles. The importance map generated in the calculation can be used to discuss simulation details in the different parts of the geometry in a simple way. The simulation CPU times are comparable to those needed within other approaches common in this field. The new approach is competitive with those previously used in this kind of problems (PSF generation or source models) and has some practical advantages that make it to be a good tool to simulate the radiation transport in problems where the quantities of interest are difficult to obtain because of low statistics.
Bending, Buckling and Vibration of a Functionally Graded Porous Beam Using Finite Elements
Directory of Open Access Journals (Sweden)
Noha Fouda
2017-10-01
Full Text Available This study presents the effect of porosity on mechanical behaviors of a power distribution functionally graded beam. The Euler-Bernoulli beam is assumed to describe the kinematic relations and constitutive equations. Because of technical problems, particle size shapes and micro-voids are created during the fabrication which should be taken into consideration. Two porosity models are proposed. The first one describes properties in the explicit form as linear functions of the porosity parameter. The second is a modified model which presents porosity and Young’s modulus in an implicit form where the density is assumed as a function of the porosity parameter and Young’s modulus as a ratio of mass with porosity to the mass without porosity. The modified proposed model is more applicable than the first model. The finite element model is developed to solve the problem by using the MATLAB software. Numerical results are presented to show the effects of porosity on mechanical behaviors of functionally graded beams.
Finite Element Model for Nonlinear Analysis of Reinforced Concrete Beams and Plane Frames
Directory of Open Access Journals (Sweden)
R.S.B. STRAMANDINOLI
Full Text Available Abstract In this work, a two-dimensional finite element (FE model for physical and geometric nonlinear analysis of reinforced concrete beams and plane frames, developed by the authors, is presented. The FE model is based on the Euler-Bernoulli Beam Theory, in which shear deformations are neglected. The bar elements have three nodes with a total of seven degrees of freedom. Three Gauss-points are utilized for the element integration, with the element section discretized into layers at each Gauss point (Fiber Model. It is assumed that concrete and reinforcing bars are perfectly bonded, and each section layer is assumed to be under a uniaxial stress-state. Nonlinear constitutive laws are utilized for both concrete and reinforcing steel layers, and a refined tension-stiffening model, developed by the authors, is included. The Total Lagrangean Formulation is adopted for geometric nonlinear consideration and several methods can be utilized to achieve equilibrium convergence of the nonlinear equations. The developed model is implemented into a computer program named ANEST/CA, which is validated by comparison with some tests on RC beams and plane frames, showing an excellent correlation between numerical and experimental results.
Two-photon physics and online beam monitoring using the DELPHI detector at LEP
International Nuclear Information System (INIS)
Bjarne, J.
1994-01-01
This thesis is based on work done during 1989-1993 using the DELPHI detector at LEP, which is summarized in five articles. It consists of three main parts. The first part describes the Very Small Angle Tagger (VSAT), which is a sub-detector of the DELPHI detector at LEP. It consists of four silicon-tungsten electromagnetic calorimeter modules having a silicon strip planes for position determination. The modules are placed adjacent to the beam pipe, at ±7.7 m from the interaction point and after superconducting quadrupole magnets, allowing the detection of electrons in a polar angle range of 4 to 13 mrad. The second part is devoted to two-photon physics at DELPHI, with strong emphasis on a VSAT single-tagged event analysis. Here is shown, for the first time, evidence of hard scattering processes in single-tagged two-photon collisions. A QCD Resolved Photon Contribution (QCD-RPC) model is introduced. Data is then seen to be well described by a full VDM+(QCD-RPC) model. Different parton density functions are compared with data. The third part first describes the system for online monitoring of LEP beam background and luminosity at the DELPHI interaction point. Details are given of contributing sub-detector signals and program structure. Then follows a description of the VSAT online monitoring program (VSAT-MONITOR). A good agreement is found between the VSAT-MONITOR estimates of luminosity and beam spot and those of other detectors. Finally, results are presented of VSAT measurements of a LEP beam separation scan. 75 refs, figs
In-beam PET at high-energy photon beams: a feasibility study
Müller, H.; Enghardt, W.
2006-04-01
For radiation therapy with carbon ion beams, either for the stable isotope 12C or for the radioactive one 11C, it has been demonstrated that the β+-activity distribution created or deposited, respectively, within the irradiated volume can be visualized by means of positron emission tomography (PET). The PET images provide valuable information for quality assurance and precision improvement of ion therapy. Dedicated PET scanners have been integrated into treatment sites at the Heavy Ion Medical Accelerator at Chiba (HIMAC), Japan, and the Gesellschaft für Schwerionenforschung (GSI), Germany, to make PET imaging feasible during therapeutic irradiation (in-beam PET). A similar technique may be worthwhile for radiotherapy with high-energy bremsstrahlung. In addition to monitoring the dose delivery process which in-beam PET has been primarily developed for, it may be expected that radiation response of tissue can be detected by means of in-beam PET. We investigate the applicability of PET for treatment control in the case of using bremsstrahlung spectra produced by 15-50 MeV electrons. Target volume activation due to (γ, n) reactions at energies above 20 MeV yields moderate β+-activity levels, which can be employed for imaging. The radiation from positrons produced by pair production is not presently usable because the detectors are overloaded due to the low duty factor of medical electron linear accelerators. However, the degradation of images caused by positron motion between creation and annihilation seems to be tolerable.
Directory of Open Access Journals (Sweden)
Litesh N. Sulbhewar
Full Text Available The convergence characteristic of the conventional two-noded Euler-Bernoulli piezoelectric beam finite element depends on the configuration of the beam cross-section. The element shows slower convergence for the asymmetric material distribution in the beam cross-section due to 'material-locking' caused by extension-bending coupling. Hence, the use of conventional Euler-Bernoulli beam finite element to analyze piezoelectric beams which are generally made of the host layer with asymmetrically surface bonded piezoelectric layers/patches, leads to increased computational effort to yield converged results. Here, an efficient coupled polynomial interpolation scheme is proposed to improve the convergence of the Euler-Bernoulli piezoelectric beam finite elements, by eliminating ill-effects of material-locking. The equilibrium equations, derived using a variational formulation, are used to establish relationships between field variables. These relations are used to find a coupled quadratic polynomial for axial displacement, having contributions from an assumed cubic polynomial for transverse displacement and assumed linear polynomials for layerwise electric potentials. A set of coupled shape functions derived using these polynomials efficiently handles extension-bending and electromechanical couplings at the field interpolation level itself in a variationally consistent manner, without increasing the number of nodal degrees of freedom. The comparison of results obtained from numerical simulation of test problems shows that the convergence characteristic of the proposed element is insensitive to the material configuration of the beam cross-section.
Optical signal response pf the alanine gel solution for photons and electrons clinical beams
International Nuclear Information System (INIS)
Silva, Cleber Feijo; Campos, Leticia Lucente
2009-01-01
Alanine gel dosimeter is a new gel material developed at IPEN that presents significant improvement on previous alanine systems developed by Costa (1994). The measure technique is based on the transformation of ferrous ions (Fe 2+ ) in ferric ions (Fe 3+ ) after irradiation. The DL-Alanine (C 3 H 7 NO 2 ) is an aminoacid tissue equivalent that improves the production of ferric ions in the solution. This work aims to study the comparison of optical signal response of the alanine gel solution for photons and electrons clinical beams. It was observed that the calibration factor can be considered independent of quality of the radiation for photons and electrons clinical beams. Therefore, it can be used the same calibration factor for evaluating the absorbed dose in photons and electrons fields in the energy of 6 MeV. Alanine Gel Dosimeter presents good performance and can be useful as alternative dosimeter in the radiotherapy area using MRI technique for 3D dose distribution evaluation. (author)
Abat, E.; et al., [Unknown; Ferrari, P.; Gorfine, G.; Liebig, W.
2011-01-01
The reconstruction of photons in the ATLAS detector is studied with data taken during the 2004 Combined Test Beam, where a full slice of the ATLAS detector was exposed to beams of particles of known energy at the CERN SPS. The results presented show significant differences in the longitudinal
2013-01-01
This study presents the beam data measurement results from the commissioning of three TrueBeam linear accelerators. An additional evaluation of the measured beam data within the TrueBeam linear accelerators contrasted with two other linear accelerators from the same manufacturer (i.e., Clinac and Trilogy) was performed to identify and evaluate any differences in the beam characteristics between the machines and to evaluate the possibility of beam matching for standard photon energies. We performed a comparison of commissioned photon beam data for two standard photon energies (6 MV and 15 MV) and one flattening filter‐free (“FFF”) photon energy (10 FFF) between three different TrueBeam linear accelerators. An analysis of the beam data was then performed to evaluate the reproducibility of the results and the possibility of “beam matching” between the TrueBeam linear accelerators. Additionally, the data from the TrueBeam linear accelerator was compared with comparable data obtained from one Clinac and one Trilogy linear accelerator models produced by the same manufacturer to evaluate the possibility of “beam matching” between the TrueBeam linear accelerators and the previous models. The energies evaluated between the linear accelerator models are the 6 MV for low energy and the 15 MV for high energy. PDD and output factor data showed less than 1% variation and profile data showed variations within 1% or 2 mm between the three TrueBeam linear accelerators. PDD and profile data between the TrueBeam, the Clinac, and Trilogy linear accelerators were almost identical (less than 1% variation). Small variations were observed in the shape of the profile for 15 MV at shallow depths (linear accelerators; the TrueBeam data resulted in a slightly greater penumbra width. The diagonal scans demonstrated significant differences in the profile shapes at a distance greater than 20 cm from the central axis, and this was more notable for the 15 MV energy. Output factor
Resolution and drift measurements on the Advanced Photon Source beam position monitors
International Nuclear Information System (INIS)
Chung, Y.; Kahana, E.
1994-01-01
The resolution and long-term drift of the Advanced Photon Source (APS) beam position monitor (BPM) electronics were measured using the charged particle beams in the ESRF storage ring with various beam current and configurations (single bunch, 8 and 16 equally spaced bunches, and 1/3-fill). The energy of the stored electrons was 6 GeV. The integrated BPM electronics system as used for this work is capable of measuring the beam position on a turn-by-turn basis, which can be accumulated for N turns (N = 2 n , n = 1, 2, ... , 11). Estimation of the BPM resolution apart from the low-frequency beam motion was made by measuring the standard deviation in the measured beam position with different Ns. The analysis of the results indicates a BPM resolution of 18/√ N [μm] for the APS storage ring, which is equivalent to 0.07 μm/√Hz. For the miniature insertion device BPM with 2.8 times higher sensitivity, the resolution will be 0.02 μm/√Hz. The long-term drift of the BPM electronics independent of the actual beam motion was measured at 2 μm/hr, which settled after approximately 1.5 hours. This drift can be attributed mainly to the temperature effect. Comparison of the results with the laboratory measurements shows good agreement. Implication of the BPM resolution limit on the proposed global and local beam position feedback systems for the APS storage ring will also be discussed
Resolution and drift measurements on the advanced photon source beam position monitor
International Nuclear Information System (INIS)
Chung, Y.; Kahana, E.
1995-01-01
The resolution and long-term drift of the Advanced Photon Source (APS) beam position monitor (BPM) electronics were measured using the charged particle beams in the ESRF storage ring with various beam current and configurations (single bunch, 8 and 16 equally spaced bunches, and 1/3-fill). The energy of the stored electrons was 6 GeV. The integrated BPM electronics system as used for this work is capable of measuring the beam position on a turn-by-turn basis, which can be accumulated for N turns (N=2 n , n=1,2,...,11) . Estimation of the BPM resolution apart from the low-frequency beam motion was made by measuring the standard deviation in the measured beam position with different Ns. The analysis of the results indicates a BPM resolution of 18/√N [μm] for the APS storage ring, which is equivalent to 0.07 μm/√Hz. For the miniature insertion device BPM with 2.8 times higher sensitivity, the resolution will be 0.02 μm/√Hz. The long-term drift of the BPM electronics independent of the actual beam motion was measured at 2 μm/hr, which settled after approximately 1.5 hours. This drift can be attributed mainly to the temperature effect. Comparison of the results with the laboratory measurements shows good agreement. Implication of the BPM resolution limit on the proposed global and local beam position feedback systems for the APS storage ring will also be discussed. copyright 1995 American Institute of Physics
Garcia de Gorordo, Alvaro; Hallock, Gary A.; Kandadai, Nirmala
2008-11-01
The Heavy Ion Beam Probe (HIBP) diagnostic has successfully measured the electric potential in a number of major plasma devices in the fusion community. In contrast to a Langmuir probe, the HIBP measures the exact electric potential rather than the floating potential. It is also has the advantage of being a very nonperturbing diagnostic. We propose a new photon-assisted beam probe technique that would extend the HIBP type of diagnostics into the low temperature plasma regime. We expect this method to probe plasmas colder than 10 eV. The novelty of the proposed diagnostic is a VUV laser that ionizes the probing particle. Excimer lasers produce the pulsed VUV radiation needed. The lasers on the market don't have a short enough wavelength too ionize any ion directly and so we calculate the population density of excited states in a NLTE plasma. These new photo-ionization techniques can take an instantaneous one-dimensional potential measurement of a plasma and are ideal for nonmagnitized plasmas where continuous time resolution is not required. Also the status of the Neutral Beam Probe installation on the Helimak experiment will be presented.
Detection system for forward emitted XUV photons from relativistic ion beams at the ESR
Energy Technology Data Exchange (ETDEWEB)
Egelkamp, C.; Hannen, V.; Ortjohann, H.W.; Vollbrecht, J.; Weinheimer, C.; Winzen, D. [Institut fuer Kernphysik, Uni Muenster (Germany); Kuehl, T. [Institut fuer Kernchemie, Uni Mainz (Germany); GSI, Darmstadt (Germany); Helmholtz Institut Jena (Germany); Noertershaeuser, W. [Institut fuer Kernchemie, Uni Mainz (Germany); GSI, Darmstadt (Germany); Sanchez, R.; Winters, D. [GSI, Darmstadt (Germany); Stoehlker, T. [GSI, Darmstadt (Germany); Helmholtz Institut Jena (Germany); Uni Jena (Germany)
2016-07-01
Highly charged heavy ions stored at relativistic velocities provide a unique possibility to test atomic structure calculations. A possibility to investigate electron-electron correlations is the study of the {sup 3}P{sub 0} → {sup 3}P{sub 1} fine structure transition in Be-like Krypton ({sup 84}Kr{sup 32+}) in laser spectroscopy experiments. For this purpose Be-like krypton ions are stored in the experimental storage ring (ESR) at GSI at a velocity of β = 0.69. Through an anticollinear arrangement of the excitation laser and the ions the wavelength in the rest frame of the ions can be matched. After the excitation to the {sup 3}P{sub 1} level the ions immediately decay to the ground state, emitting λ ∼ 17 nm photons. Due to the Lorentz boost, the photons are emitted mainly in the forward direction and experience a Doppler shift to wavelengths < 10 nm. To collect these photons a moveable cathode plate with a central slit is brought into the beam line. The XUV photons mostly produce low energy secondary electrons on the plate which are electromagnetically guided onto a MCP detector. The design and working principle, as well as simulations and test measurements of the detector are presented.
An MCNP-based model of a medical linear accelerator x-ray photon beam.
Ajaj, F A; Ghassal, N M
2003-09-01
The major components in the x-ray photon beam path of the treatment head of the VARIAN Clinac 2300 EX medical linear accelerator were modeled and simulated using the Monte Carlo N-Particle radiation transport computer code (MCNP). Simulated components include x-ray target, primary conical collimator, x-ray beam flattening filter and secondary collimators. X-ray photon energy spectra and angular distributions were calculated using the model. The x-ray beam emerging from the secondary collimators were scored by considering the total x-ray spectra from the target as the source of x-rays at the target position. The depth dose distribution and dose profiles at different depths and field sizes have been calculated at a nominal operating potential of 6 MV and found to be within acceptable limits. It is concluded that accurate specification of the component dimensions, composition and nominal accelerating potential gives a good assessment of the x-ray energy spectra.
An alternative approach to compensators design for photon beams used in radiotherapy
International Nuclear Information System (INIS)
Jurkovic, S.; Zauhar, G.; Bistrovic, M.; Faj, D.; Kaliman, Z.; Smilovic Radojcic, D.
2007-01-01
The use of compensators in order to achieve desired dose distribution has a long history and is a well-established technique in radiation therapy planning. There are several different calculation methods for determining a compensator's thickness. An alternative method that is based on the Cunningham's modification of Clarkson's method to calculate scattered radiation in beams with an inhomogeneous cross-section is proposed. It is well known that the total dose distribution of radiotherapy photon beam consists of the contributions of the primary beam, attenuated by the tissue layer, and the scattered radiation generated by the primary radiation in single and multiple photon scatter events. The scattered component can be represented as a function of the primary radiation. The central point of our method is the numerical estimation of the primary distribution required to achieve the desired total distribution. Now using the calculated primary distribution, the shape of the modulator could be determined. In this way the contribution of the scattered component is validated in a more accurate way than using effective attenuation coefficients, which is a common practice. The method is verified in various clinical situations and compared with the standard method. The accuracy, although dependent on geometry, was improved by at least 2%. With more complex geometries there is an even higher gain in accuracy with our method when compared to the standard method
Liquid scintillator for 2D dosimetry for high-energy photon beams
International Nuclear Information System (INIS)
Poenisch, Falk; Archambault, Louis; Briere, Tina Marie; Sahoo, Narayan; Mohan, Radhe; Beddar, Sam; Gillin, Michael T.
2009-01-01
Complex radiation therapy techniques require dosimetric verification of treatment planning and delivery. The authors investigated a liquid scintillator (LS) system for application for real-time high-energy photon beam dosimetry. The system was comprised of a transparent acrylic tank filled with liquid scintillating material, an opaque outer tank, and a CCD camera. A series of images was acquired when the tank with liquid scintillator was irradiated with a 6 MV photon beam, and the light data measured with the CCD camera were filtered to correct for scattering of the optical light inside the liquid scintillator. Depth-dose and lateral profiles as well as two-dimensional (2D) dose distributions were found to agree with results from the treatment planning system. Further, the corrected light output was found to be linear with dose, dose rate independent, and is robust for single or multiple acquisitions. The short time needed for image acquisition and processing could make this system ideal for fast verification of the beam characteristics of the treatment machine. This new detector system shows a potential usefulness of the LS for 2D QA.
Liquid scintillator for 2D dosimetry for high-energy photon beams
Energy Technology Data Exchange (ETDEWEB)
Poenisch, Falk; Archambault, Louis; Briere, Tina Marie; Sahoo, Narayan; Mohan, Radhe; Beddar, Sam; Gillin, Michael T. [Department of Radiation Physics, University of Texas M. D. Anderson Cancer Center, 1515 Holcombe Boulevard., Unit 94, Houston, Texas 77030 (United States)
2009-05-15
Complex radiation therapy techniques require dosimetric verification of treatment planning and delivery. The authors investigated a liquid scintillator (LS) system for application for real-time high-energy photon beam dosimetry. The system was comprised of a transparent acrylic tank filled with liquid scintillating material, an opaque outer tank, and a CCD camera. A series of images was acquired when the tank with liquid scintillator was irradiated with a 6 MV photon beam, and the light data measured with the CCD camera were filtered to correct for scattering of the optical light inside the liquid scintillator. Depth-dose and lateral profiles as well as two-dimensional (2D) dose distributions were found to agree with results from the treatment planning system. Further, the corrected light output was found to be linear with dose, dose rate independent, and is robust for single or multiple acquisitions. The short time needed for image acquisition and processing could make this system ideal for fast verification of the beam characteristics of the treatment machine. This new detector system shows a potential usefulness of the LS for 2D QA.
Initial time-resolved particle beam profile measurements at the Advanced Photon Source
International Nuclear Information System (INIS)
Yang, B.X.; Lumpkin, A.H.
1995-01-01
The commissioning of the 7-GeV Advanced Photon Source (APS) storage ring began in early 1995. Characterization of the stored particle beam properties involved time-resolved transverse and longitudinal profile measurements using optical synchrotron radiation (OSR) monitors. Early results include the observation of the beam on a single turn, measurements of the transverse beam sizes after damping using a 100 μs integration time (σ x ∼ 150 ± 25 μm, σ γ ∼ 65 ± 25 μm, depending on vertical coupling), and measurement of the bunch length (σ τ ∼ 25 to 55 ps, depending on the charge per bunch). The results are consistent with specifications and predictions based on the 8.2 nm-rad natural emittance, the calculated lattice parameters, and vertical coupling less than 10%. The novel, single-element focusing mirror for the photon transport line and the dual-sweep streak camera techniques which allow turn-by-turn measurements will also be presented. The latter measurements are believed to be the first of their kind on a storage ring in the USA
Probing WWγ and WWγγ couplings with high energy photon beams
International Nuclear Information System (INIS)
Choi, S.Y.; Schrempp, F.
1991-12-01
We examine the potential of a future 500 GeV linear e + e - collider for probing anomalous WW γ and WW γγ couplings in the so-called γ(γ)model, corresponding to colliding γe and γγ beams from Compton backscattering of laser light. We consider in detail the 'minimal' set (k γ , λ γ ) of CP conserving anomalous couplings and present first results for the CP violating 'partner' couplings (anti K γ , anti l γ ) as well. The reactions under consideration are γe → Wν, γγ → W + W - and, as a reference, also e + e - → W + W - . We discuss the impact of both circular polarization of laser photons and polarized e(anti e) beams. Photon 'beams' due to classical Bremsstrahlung are also studied for comparison. We analyze in detail, how changes of the assumed machine parameters, cuts and systematic errors affect the sensitivity to the anomalous couplings. (orig.)
Andrews, David L
2015-01-01
Discusses the basic physical principles underlying Biomedical Photonics, spectroscopy and microscopy This volume discusses biomedical photonics, spectroscopy and microscopy, the basic physical principles underlying the technology and its applications. The topics discussed in this volume are: Biophotonics; Fluorescence and Phosphorescence; Medical Photonics; Microscopy; Nonlinear Optics; Ophthalmic Technology; Optical Tomography; Optofluidics; Photodynamic Therapy; Image Processing; Imaging Systems; Sensors; Single Molecule Detection; Futurology in Photonics. Comprehensive and accessible cov
Andrews, David L
2015-01-01
Discusses the basic physical principles underlying the technology instrumentation of photonics This volume discusses photonics technology and instrumentation. The topics discussed in this volume are: Communication Networks; Data Buffers; Defense and Security Applications; Detectors; Fiber Optics and Amplifiers; Green Photonics; Instrumentation and Metrology; Interferometers; Light-Harvesting Materials; Logic Devices; Optical Communications; Remote Sensing; Solar Energy; Solid-State Lighting; Wavelength Conversion Comprehensive and accessible coverage of the whole of modern photonics Emphas
Andrews, David L
2015-01-01
Discusses the basic physical principles underlying thescience and technology of nanophotonics, its materials andstructures This volume presents nanophotonic structures and Materials.Nanophotonics is photonic science and technology that utilizeslight/matter interactions on the nanoscale where researchers arediscovering new phenomena and developing techniques that go wellbeyond what is possible with conventional photonics andelectronics.The topics discussed in this volume are: CavityPhotonics; Cold Atoms and Bose-Einstein Condensates; Displays;E-paper; Graphene; Integrated Photonics; Liquid Cry
Andrews, David L
2015-01-01
This book covers modern photonics accessibly and discusses the basic physical principles underlying all the applications and technology of photonicsThis volume covers the basic physical principles underlying the technology and all applications of photonics from statistical optics to quantum optics. The topics discussed in this volume are: Photons in perspective; Coherence and Statistical Optics; Complex Light and Singular Optics; Electrodynamics of Dielectric Media; Fast and slow Light; Holography; Multiphoton Processes; Optical Angular Momentum; Optical Forces, Trapping and Manipulation; Pol
Qualification of a Monte Carlo model of photon beams of a Lilac Elekta Precise
International Nuclear Information System (INIS)
Linares R, H. M.; Laguardia, R. A.; Lara M, E.
2014-08-01
For the simulation of the accelerator head the parameters determination that characterize the electrons primary beam that affect in the target is a step that involves a fundamental role in the precision of the Monte Carlo calculations. Applying the proposed methodology by Pena et al. [2007], in this work was carried out the qualification of the photon beams (6 MV and 15 MV) of an accelerator Elekta Precise, using the Monte Carlo code EGSnrc. The influence exerted by the characteristics of the electrons primary beam on the distribution of absorbed dose for the two energy of this equipment was studied. Using different mid energy combinations and FWHM of the electrons primary beam was calculated the dose deposited in a segmented water mannequin with its surface to 100 cm of the source. Starting from the deposited dose in the mannequin the dose curves in depth and dose profiles to different depths were built. These curves were compared with measured values in a similar experimental arrangement to the carried out simulation, applying acceptability criteria based on confidence intervals [Venselaar et al. 2001]. The dose profiles for small fields were like it was expected, to be strongly influenced by the radial distribution (FWHM). The energy/FWHM combinations that better reproduce the experimental curves of each photon beam were determined. One time determined the best combination (5.75 MeV/2 mm and 11.25 MeV/2 mm, respectively) was used for the generation of the phase spaces and the field factors calculation. A good correspondence was obtained between the simulations and the measurements for a wide range of field sizes, as well as for different types of detectors, being all the results inside of the tolerance margins. (author)
Optimization of parameters for fitting linear accelerator photon beams using a modified CBEAM model
International Nuclear Information System (INIS)
Ayyangar, K.; Daftari, I.; Palta, J.; Suntharalingam, N.
1989-01-01
Measured beam profiles and central-axis depth-dose data for 6- and 25-MV photon beams are used to generate a dose matrix which represents the full beam. A corresponding dose matrix is also calculated using the modified CBEAM model. The calculational model uses the usual set of three parameters to define the intensity at beam edges and the parameter that accounts for collimator transmission. An additional set of three parameters is used for the primary profile factor, expressed as a function of distance from the central axis. An optimization program has been adapted to automatically adjust these parameters to minimize the χ 2 between the measured and calculated data. The average values of the parameters for small (6x6 cm 2 ), medium (10x10 cm 2 ), and large (20x20 cm 2 ) field sizes are found to represent the beam adequately for all field sizes. The calculated and the measured doses at any point agree to within 2% for any field size in the range 4x4 to 40x40 cm 2
Proton beam writing of long, arbitrary structures for micro/nano photonics and fluidics applications
International Nuclear Information System (INIS)
Udalagama, Chammika; Teo, E.J.; Chan, S.F.; Kumar, V.S.; Bettiol, A.A.; Watt, F.
2011-01-01
The last decade has seen proton beam writing maturing into a versatile lithographic technique able to produce sub-100 nm, high aspect ratio structures with smooth side walls. However, many applications in the fields of photonics and fluidics require the fabrication of structures with high spatial resolution that extends over several centimetres. This cannot be achieved by purely magnetic or electrostatic beam scanning due to the large off-axis beam aberrations in high demagnification systems. As a result, this has limited us to producing long straight structures using a combination of beam and stage scanning. In this work we have: (1) developed an algorithm to include any arbitrary pattern into the writing process by using a more versatile combination of beam and stage scanning while (2) incorporating the use of the ubiquitous AutoCAD DXF (drawing exchange format) into the design process. We demonstrate the capability of this approach in fabricating structures such as Y-splitters, Mach-Zehnder modulators and microfluidic channels that are over several centimetres in length, in polymer. We also present optimisation of such parameters as scanning speed and scanning loops to improve on the surface roughness of the structures. This work opens up new possibilities of using CAD software in PBW for microphotonics and fluidics device fabrication.
Proton beam writing of long, arbitrary structures for micro/nano photonics and fluidics applications
Udalagama, Chammika; Teo, E. J.; Chan, S. F.; Kumar, V. S.; Bettiol, A. A.; Watt, F.
2011-10-01
The last decade has seen proton beam writing maturing into a versatile lithographic technique able to produce sub-100 nm, high aspect ratio structures with smooth side walls. However, many applications in the fields of photonics and fluidics require the fabrication of structures with high spatial resolution that extends over several centimetres. This cannot be achieved by purely magnetic or electrostatic beam scanning due to the large off-axis beam aberrations in high demagnification systems. As a result, this has limited us to producing long straight structures using a combination of beam and stage scanning. In this work we have: (1) developed an algorithm to include any arbitrary pattern into the writing process by using a more versatile combination of beam and stage scanning while (2) incorporating the use of the ubiquitous AutoCAD DXF (drawing exchange format) into the design process. We demonstrate the capability of this approach in fabricating structures such as Y-splitters, Mach-Zehnder modulators and microfluidic channels that are over several centimetres in length, in polymer. We also present optimisation of such parameters as scanning speed and scanning loops to improve on the surface roughness of the structures. This work opens up new possibilities of using CAD software in PBW for microphotonics and fluidics device fabrication.
Proton beam writing of long, arbitrary structures for micro/nano photonics and fluidics applications
Energy Technology Data Exchange (ETDEWEB)
Udalagama, Chammika, E-mail: chammika@nus.edu.sg [Centre for Ion Beam Applications (CIBA), Department of Physics, National University of Singapore (NUS), 2 Science Drive 3, Singapore 117542 (Singapore); Teo, E.J. [Centre for Ion Beam Applications (CIBA), Department of Physics, National University of Singapore (NUS), 2 Science Drive 3, Singapore 117542 (Singapore); Chan, S.F. [Centre for Ion Beam Applications (CIBA), Department of Physics, National University of Singapore (NUS), 2 Science Drive 3, Singapore 117542 (Singapore); NUS Nanoscience and Nanotechnology Initiative, 2 Science Drive 3, 117542 (Singapore); Department of Chemistry, NUS, 3 Science Drive 3, 117543 (Singapore); Kumar, V.S.; Bettiol, A.A.; Watt, F. [Centre for Ion Beam Applications (CIBA), Department of Physics, National University of Singapore (NUS), 2 Science Drive 3, Singapore 117542 (Singapore)
2011-10-15
The last decade has seen proton beam writing maturing into a versatile lithographic technique able to produce sub-100 nm, high aspect ratio structures with smooth side walls. However, many applications in the fields of photonics and fluidics require the fabrication of structures with high spatial resolution that extends over several centimetres. This cannot be achieved by purely magnetic or electrostatic beam scanning due to the large off-axis beam aberrations in high demagnification systems. As a result, this has limited us to producing long straight structures using a combination of beam and stage scanning. In this work we have: (1) developed an algorithm to include any arbitrary pattern into the writing process by using a more versatile combination of beam and stage scanning while (2) incorporating the use of the ubiquitous AutoCAD DXF (drawing exchange format) into the design process. We demonstrate the capability of this approach in fabricating structures such as Y-splitters, Mach-Zehnder modulators and microfluidic channels that are over several centimetres in length, in polymer. We also present optimisation of such parameters as scanning speed and scanning loops to improve on the surface roughness of the structures. This work opens up new possibilities of using CAD software in PBW for microphotonics and fluidics device fabrication.
Photon beam dose distributions for patients with implanted temporary tissue expanders
Asena, A.; Kairn, T.; Crowe, S. B.; Trapp, J. V.
2015-01-01
This study examines the effects of temporary tissue expanders (TTEs) on the dose distributions of photon beams in breast cancer radiotherapy treatments. EBT2 radiochromic film and ion chamber measurements were taken to quantify the attenuation and backscatter effects of the inhomogeneity. Results illustrate that the internal magnetic port present in a tissue expander causes a dose reduction of approximately 25% in photon tangent fields immediately downstream of the implant. It was also shown that the silicone elastomer shell of the tissue expander reduced the dose to the target volume by as much as 8%. This work demonstrates the importance for an accurately modelled high-density implant in the treatment planning system for post-mastectomy breast cancer patients.
Ralicon anodes for image photon counting fabricated by electron beam lithography
International Nuclear Information System (INIS)
Burton, W.M.
1982-01-01
The Anger wedge and strip anode event location system developed for microchannel plate image photon detectors at the Space Sciences Laboratory of the University of California, Berkeley, has been extended in the present work by the use of electron beam lithography (EBL). This method of fabrication can be used to produce optical patterns for the subsequent manufacture of anodes by conventional photo-etching methods and has also enabled anodes to be produced directly by EBL microfabrication techniques. Computer-aided design methods have been used to develop several types of RALICON (Readout Anodes of Lithographic Construction) for use in photon counting microchannel plate imaging detectors. These anodes are suitable for linear, two dimensional or radial position measurements and they incorporate novel design features made possible by the EBL fabrication technique which significantly extend their application relative to published wedge-strip anode designs. (author)
Why diamond dimensions and electrode geometry are crucial for small photon beam dosimetry
International Nuclear Information System (INIS)
Marsolat, F.; Tromson, D.; Tranchant, N.; Pomorski, M.; Bergonzo, P.; Bassinet, C.; Huet, C.; Derreumaux, S.; Chea, M.; Cristina, K.; Boisserie, G.; Buchheit, I.; Marchesi, V.; Gaudaire-Josset, S.; Lisbona, A.; Lazaro, D.; Hugon, R.
2015-01-01
Recent use of very small photon beams (down to 4 mm) in stereotactic radiotherapy requires new detectors to accurately determine the delivered dose. Diamond detectors have been presented in the literature as an attractive candidate for this application, due to their small detection volume and the diamond atomic number (Z = 6) which is close to water effective atomic number (Zeff ∼ 7.42). However, diamond exhibits a density 3.51 times greater than that of water and recent studies using Monte Carlo simulations have demonstrated the drawback of a high-density detector on small beam output factors. The current study focuses on geometrical parameters of diamond detector, namely, the diamond dimensions and the electrode geometry, in order to solve the dosimetric issues still observed in small photon beams with diamond detectors. To give better insights to these open questions, we have used both computational method and experimental analysis. This study highlighted that reducing diamond dimensions is crucial for small beam output factor measurements and to limit the influence of its high density. Furthermore, electrodes covering the whole diamond surface were essential for a dose rate independence of the diamond detector. The optimal dosimeter derived from this work presented small diamond dimensions of approximately 1 × 1 × 0.15 mm 3 , with diamond-like-carbon electrodes covering the whole diamond surface. A dose rate independence of this diamond detector (better than 0.5% over a wide range of dose rates available on a stereotactic dedicated facility) was obtained due to the electrode geometry. Concerning the output factor measurements, a good agreement (better than 1.1%) was observed between this carbon material detector and two types of passive dosimeters (LiF microcubes and EBT2 radiochromic films) for all beam sizes except the smallest field of 0.6 × 0.6 cm 2 with a deviation of 2.6%. This new study showed the high performance of this
Why diamond dimensions and electrode geometry are crucial for small photon beam dosimetry
Energy Technology Data Exchange (ETDEWEB)
Marsolat, F.; Tromson, D.; Tranchant, N.; Pomorski, M.; Bergonzo, P. [CEA, LIST, Diamond Sensors Laboratory, 91191 Gif-sur-Yvette (France); Bassinet, C.; Huet, C. [IRSN, PRP-HOM/SDE/LDRI, 31 Av. de la Division Leclerc, 92260 Fontenay-aux-Roses (France); Derreumaux, S. [IRSN, PRP-HOM/SER/UEM, 31 Av. de la Division Leclerc, 92260 Fontenay-aux-Roses (France); Chea, M.; Cristina, K.; Boisserie, G. [Pitié Salpêtrière Hospital, 47-83 Blvd de l' Hôpital, 75013 Paris (France); Buchheit, I.; Marchesi, V. [Institut de Cancérologie de Lorraine, 6 Av. de Bourgogne, 54500 Vandoeuvre-lès-Nancy (France); Gaudaire-Josset, S.; Lisbona, A. [Institut de Cancérologie de l' Ouest, Blvd Prof. Jacques Monod, 44805 Saint-Herblain (France); Lazaro, D.; Hugon, R. [CEA, LIST, LM2S, 91191 Gif-sur-Yvette (France)
2015-12-21
Recent use of very small photon beams (down to 4 mm) in stereotactic radiotherapy requires new detectors to accurately determine the delivered dose. Diamond detectors have been presented in the literature as an attractive candidate for this application, due to their small detection volume and the diamond atomic number (Z = 6) which is close to water effective atomic number (Zeff ∼ 7.42). However, diamond exhibits a density 3.51 times greater than that of water and recent studies using Monte Carlo simulations have demonstrated the drawback of a high-density detector on small beam output factors. The current study focuses on geometrical parameters of diamond detector, namely, the diamond dimensions and the electrode geometry, in order to solve the dosimetric issues still observed in small photon beams with diamond detectors. To give better insights to these open questions, we have used both computational method and experimental analysis. This study highlighted that reducing diamond dimensions is crucial for small beam output factor measurements and to limit the influence of its high density. Furthermore, electrodes covering the whole diamond surface were essential for a dose rate independence of the diamond detector. The optimal dosimeter derived from this work presented small diamond dimensions of approximately 1 × 1 × 0.15 mm{sup 3}, with diamond-like-carbon electrodes covering the whole diamond surface. A dose rate independence of this diamond detector (better than 0.5% over a wide range of dose rates available on a stereotactic dedicated facility) was obtained due to the electrode geometry. Concerning the output factor measurements, a good agreement (better than 1.1%) was observed between this carbon material detector and two types of passive dosimeters (LiF microcubes and EBT2 radiochromic films) for all beam sizes except the smallest field of 0.6 × 0.6 cm{sup 2} with a deviation of 2.6%. This new study showed the high performance
Finite element formulation of viscoelastic sandwich beams using fractional derivative operators
Galucio, A. C.; Deü, J.-F.; Ohayon, R.
This paper presents a finite element formulation for transient dynamic analysis of sandwich beams with embedded viscoelastic material using fractional derivative constitutive equations. The sandwich configuration is composed of a viscoelastic core (based on Timoshenko theory) sandwiched between elastic faces (based on Euler-Bernoulli assumptions). The viscoelastic model used to describe the behavior of the core is a four-parameter fractional derivative model. Concerning the parameter identification, a strategy to estimate the fractional order of the time derivative and the relaxation time is outlined. Curve-fitting aspects are focused, showing a good agreement with experimental data. In order to implement the viscoelastic model into the finite element formulation, the Grünwald definition of the fractional operator is employed. To solve the equation of motion, a direct time integration method based on the implicit Newmark scheme is used. One of the particularities of the proposed algorithm lies in the storage of displacement history only, reducing considerably the numerical efforts related to the non-locality of fractional operators. After validations, numerical applications are presented in order to analyze truncation effects (fading memory phenomena) and solution convergence aspects.
Zhang, Yunkai; Hsi, Wen C; Chu, James C H; Bernard, Damian B; Abrams, Ross A
2005-01-01
The effect of gantry rotation on beam profiles of photon and electron beams is an important issue in quality assurance for radiotherapy. To address variations in the profiles of photon and electron beams at different gantry angles, a Dynamic Phantom scanner composed of a 20 x 12 x 6 cm3 scanning Lucite block was designed as a cross-beam-profile scanner. To our knowledge, differences between scanned profiles acquired at different gantry angles with a small size Lucite block and those acquired a full-size (60 x 60 x 50 cm3) water phantom have not been previously investigated. We therefore performed a feasibility study for a first prototype Dynamic Phantom scanner without a gantry attachment mount. Radiation beams from a Varian LINAC 21EX and 2100C were used. Photon beams (6 MV and 18 MV) were shaped by either collimator jaws or a Varian 120 Multileaf (MLC) collimator, and electron beams (6 MeV, 12 MeV, and 20 MeV) were shaped by a treatment cone. To investigate the effect on profiles by using a Lucite block, a quantitative comparison of scanned profiles with the Dynamic Phantom and a full-size water phantom was first performed at a 0 degrees gantry angle for both photon and electron beams. For photon beam profiles defined by jaws at 1.0 cm and 5.0 cm depths of Lucite (i.e., at 1.1 cm and 5.7 cm depth of water), a good agreement (less than 1% variation) inside the field edge was observed between profiles scanned with the Dynamic Phantom and with a water phantom. The use of Lucite in the Dynamic Phantom resulted in reduced penumbra width (about 0.5 mm out of 5 mm to 8mm) and reduced (1% to 2%) scatter dose beyond the field edges for both 6 MV and 18 MV beams, compared with the water phantom scanner. For profiles of the MLC-shaped 6 MV photon beam, a similar agreement was observed. For profiles of electron beams scanned at 2.9 cm depth of Lucite (i.e., at 3.3 cm depth of water), larger disagreements in profiles (3% to 4%) and penumbra width (3 mm to 4 mm out of 12 mm
International Nuclear Information System (INIS)
Santos, Walter Menezes
2003-08-01
Radiosurgery and stereotactic radiotherapy are irradiation techniques that use small diameter photon beams for treating intracranial lesions such as pituitary adenomas, acoustic tumors and arterio-venous malformations which are inaccessible for surgery. These treatment techniques are characterized by the use of very small radiation beams which deliver a precisely measured dose to the target volume, while sparing the surrounding healthy tissue. Treatment can be performed by using multiple 60 Co gamma-ray sources (in the so-called 'Gamma Knife'), charged particles or X-ray beams produced by linear accelerators. The prescribed dose can be given in a single session or in multiple fractions, as in conventional radiotherapy. The success of the treatment depends, among other factors, of the accurate determination of the parameters that characterize the radiation beam produced by the equipment, as well as, of a well designed quality assurance program. In this study, the dosimetric parameters of a set of collimating cones of a Radionics TM treatment system applied to two 6 MV- photon beams (Clinac 600C - Varian TM , and Mevatron MD2 - Siemens TM ) were evaluated by using a water filled PMMA simulator. Measurements were carried out for photon beam diameters ranging from 12.5 to 40.0 mm for the Clinac-600C and from 5.0 to 50.0 mm for the Mevatron MD2. The parameters were evaluated by using a parallel plate ionization chamber (Markus), Kodak X-Omat V dosimetric films, thermoluminescent dosemeters (Harschaw, TLD-100) and photodiodes. The maximum tissue-ratio, the off-axis profile and the output factors were determined and the results were compared to those reported elsewhere. A study of the dosimetric characteristics of some commercially available phototransistors was also carried out. The results showed that these electronic components can be successfully used for measuring the dosimetric parameters of small diameter photon beans used in radiosurgery. Measurements were also
Skin damage probabilities using fixation materials in high-energy photon beams
International Nuclear Information System (INIS)
Carl, J.; Vestergaard, A.
2000-01-01
Patient fixation, such as thermoplastic masks, carbon-fibre support plates and polystyrene bead vacuum cradles, is used to reproduce patient positioning in radiotherapy. Consequently low-density materials may be introduced in high-energy photon beams. The aim of the this study was to measure the increase in skin dose when low-density materials are present and calculate the radiobiological consequences in terms of probabilities of early and late skin damage. An experimental thin-windowed plane-parallel ion chamber was used. Skin doses were measured using various overlaying low-density fixation materials. A fixed geometry of a 10 x 10 cm field, a SSD = 100 cm and photon energies of 4, 6 and 10 MV on Varian Clinac 2100C accelerators were used for all measurements. Radiobiological consequences of introducing these materials into the high-energy photon beams were evaluated in terms of early and late damage of the skin based on the measured surface doses and the LQ-model. The experimental ion chamber save results consistent with other studies. A relationship between skin dose and material thickness in mg/cm 2 was established and used to calculate skin doses in scenarios assuming radiotherapy treatment with opposed fields. Conventional radiotherapy may apply mid-point doses up to 60-66 Gy in daily 2-Gy fractions opposed fields. Using thermoplastic fixation and high-energy photons as low as 4 MV do increase the dose to the skin considerably. However, using thermoplastic materials with thickness less than 100 mg/cm 2 skin doses are comparable with those produced by variation in source to skin distance, field size or blocking trays within clinical treatment set-ups. The use of polystyrene cradles and carbon-fibre materials with thickness less than 100 mg/cm 2 should be avoided at 4 MV at doses above 54-60 Gy. (author)
Directory of Open Access Journals (Sweden)
W.R. Azzam
2015-08-01
Full Text Available This paper reports the application of using a skirted foundation system to study the behavior of foundations with structural skirts adjacent to a sand slope and subjected to earthquake loading. The effect of the adopted skirts to safeguard foundation and slope from collapse is studied. The skirts effect on controlling horizontal soil movement and decreasing pore water pressure beneath foundations and beside the slopes during earthquake is investigated. This technique is investigated numerically using finite element analysis. A four story reinforced concrete building that rests on a raft foundation is idealized as a two-dimensional model with and without skirts. A two dimensional plain strain program PLAXIS, (dynamic version is adopted. A series of models for the problem under investigation were run under different skirt depths and lactation from the slope crest. The effect of subgrade relative density and skirts thickness is also discussed. Nodal displacement and element strains were analyzed for the foundation with and without skirts and at different studied parameters. The research results showed a great effectiveness in increasing the overall stability of the slope and foundation. The confined soil footing system by such skirts reduced the foundation acceleration therefore it can be tended to damping element and relieved the transmitted disturbance to the adjacent slope. This technique can be considered as a good method to control the slope deformation and decrease the slope acceleration during earthquakes.
Gaussian beam-to-slab waveguide coupler by graded index photonic crystal lens
International Nuclear Information System (INIS)
Bahari, B; Abrishamian, M S
2013-01-01
In this numerical study, a Gaussian beam-to-slab waveguide coupler for both modes of TM and TE has been studied. For this purpose, a concrete structure is suggested, in which the graded index photonic crystal lens and the slab waveguide are in the same structure composed of Si material, and can be fabricated with a single-step lithography process. For maximum power coupling, half-holes have been used as an input matching layer. Power coupling of 80% over a 450 nm bandwidth for the TM mode, and 60% over a 180 nm bandwidth for the TE mode is achieved. (paper)
Finite-element 3D simulation tools for high-current relativistic electron beams
Humphries, Stanley; Ekdahl, Carl
2002-08-01
The DARHT second-axis injector is a challenge for computer simulations. Electrons are subject to strong beam-generated forces. The fields are fully three-dimensional and accurate calculations at surfaces are critical. We describe methods applied in OmniTrak, a 3D finite-element code suite that can address DARHT and the full range of charged-particle devices. The system handles mesh generation, electrostatics, magnetostatics and self-consistent particle orbits. The MetaMesh program generates meshes of conformal hexahedrons to fit any user geometry. The code has the unique ability to create structured conformal meshes with cubic logic. Organized meshes offer advantages in speed and memory utilization in the orbit and field solutions. OmniTrak is a versatile charged-particle code that handles 3D electric and magnetic field solutions on independent meshes. The program can update both 3D field solutions from the calculated beam space-charge and current-density. We shall describe numerical methods for orbit tracking on a hexahedron mesh. Topics include: 1) identification of elements along the particle trajectory, 2) fast searches and adaptive field calculations, 3) interpolation methods to terminate orbits on material surfaces, 4) automatic particle generation on multiple emission surfaces to model space-charge-limited emission and field emission, 5) flexible Child law algorithms, 6) implementation of the dual potential model for 3D magnetostatics, and 7) assignment of charge and current from model particle orbits for self-consistent fields.
Accounting for straight parts effects on elbow's flexibilities in a beam type finite element program
International Nuclear Information System (INIS)
Millard, A.
1983-01-01
An extension of Von Karman's theory is applied to the calculations of the flexibility factor of a pipe bend terminated by a straight part or a flange. This analysis is restricted to the linear elastic deformation behaviour under in plane bending. Analytical solutions are given for the propagation of ovalization in the elbow and in the straight part. Considering the response of the piping structures, we note that the ovalization of the piping systems are reduced significantly when the straight parts or flanges effects are included. This results are presented in terms of global as well local flexibility factors. They have been compared to numerical results obtained by shell type finite elements method. A complete piping system is analyzed, for economical reasons, with a beam type approach. Also, we show how it is possible to take into account an elbow's flexibilities the straight parts effects by means of flexibilities factors introduced in a beam type elements. We have implemented this method in the computer program TEDEL. In some specific geometrical features, we compare solutions using shell type elements and our formulation. (orig.)
Accounting for straight parts effects on elbow's flexibilities in a beam type finite element program
International Nuclear Information System (INIS)
Millard, A.; Vaghi, H.; Ricard, A.
1983-08-01
An extension of Von Karman's theory is applied to the calculations of the flexibility factor of a pipe bend terminated by a straight part or a flange. This analysis is restricted to the linear elastic deformation behaviour under in plane bending. Analytical solutions are given for the propagation of ovalization in the elbow and in the straight part. Considering the response of the piping structures, we note that the ovalization of the piping systems are reduced significantly when the straight parts or flanges effects are included. The results are presented in terms of global as well local flexibility factors. They have been compared to numerical results obtained by shell type finite element method. A complete piping system is analyzed, for economical reasons, with a beam type approach. Also, we show how it is possible to take into account on elbow's flexibilities the straight parts effects by means of flexibilities factors introduced in a beam type element. We have implemented this method in the computer program TEDEL. In some specific geometrical features, we compare solutions using shell type elements and our formulation
Harijishnu, R.; Jayakumar, J. S.
2017-09-01
The main objective of this paper is to study the heat transfer rate of thermal radiation in participating media. For that, a generated collimated beam has been passed through a two dimensional slab model of flint glass with a refractive index 2. Both Polar and azimuthal angle have been varied to generate such a beam. The Temperature of the slab and Snells law has been validated by Radiation Transfer Equation (RTE) in OpenFOAM (Open Field Operation and Manipulation), a CFD software which is the major computational tool used in Industry and research applications where the source code is modified in which radiation heat transfer equation is added to the case and different radiation heat transfer models are utilized. This work concentrates on the numerical strategies involving both transparent and participating media. Since Radiation Transfer Equation (RTE) is difficult to solve, the purpose of this paper is to use existing solver buoyantSimlpeFoam to solve radiation model in the participating media by compiling the source code to obtain the heat transfer rate inside the slab by varying the Intensity of radiation. The Finite Volume Method (FVM) is applied to solve the Radiation Transfer Equation (RTE) governing the above said physical phenomena.
International Nuclear Information System (INIS)
Rah, Jeong-Eun; Kim, Siyong; Cheong, Kwang-Ho; Lee, Jeong-Woo; Chung, Jin-Beom; Shin, Dong-Oh; Suh, Tae-Suk
2009-01-01
A radiophotoluminescent glass rod dosimeter (GRD) system has recently become commercially available. In this study we evaluated whether the GRD would be suitable for external dosimetric audit program in radiotherapy. For this purpose, we introduced a methodology of the absorbed dose determination with the GRD by establishing calibration coefficient and various correction factors (non-linearity dose response, fading, energy dependence and angular dependence). A feasibility test of the GRD postal dose intercomparison was also performed for eight high photon beams by considering four radiotherapy centers in Korea. In the accuracy evaluation of the GRD dosimetry established in this study, we obtained within 1.5% agreements with the ionization chamber dosimetry for the 60 Co beam. It was also observed that, in the feasibility study, all the relative deviations were smaller than 3%. Based on these results, we believe that the new GRD system has considerable potential to be used for a postal dose audit program
Image acquisition and analysis for beam diagnostics, applications of the Taiwan photon source
International Nuclear Information System (INIS)
Liao, C.Y.; Chen, J.; Cheng, Y.S.; Hsu, K.T.; Hu, K.H.; Kuo, C.H.; Wu, C.Y.
2012-01-01
Design and implementation of image acquisition and analysis is in proceeding for the Taiwan Photon Source (TPS) diagnostic applications. The optical system contains screen, lens, and lighting system. A CCD camera with Gigabit Ethernet interface (GigE Vision) will be a standard image acquisition device. Image acquisition will be done on EPICS IOC via PV channel and analysis the properties by using Matlab tool to evaluate the beam profile (sigma), beam size position and tilt angle et al. The EPICS IOC integrated with Matlab as a data processing system is not only could be used in image analysis but also in many types of equipment data processing applications. Progress of the project will be summarized in this report. (authors)
Dosimetric characteristics of Thermo-Shield material for orthovoltage photon beams
International Nuclear Information System (INIS)
Bahmaid, Mohammad; Kim, Siyong; Liu, Chihray R.; Palta, Jatinder R.
2003-01-01
Conventionally, lead has been used for field shaping in orthovoltage radiation therapy. Recently, a compensator material named Thermo-Shield was presented for field shaping in electron beams. Thermo-Shield is composed of nontoxic, high atomic weight metal particles dispersed in a thermoplastic matrix. It is manually moldable and conforms to human anatomy or any shape at temperatures of 108-132 degree sign F. It is reusable and can be continuously reshaped to better fit the treatment field. Dosimetric characteristics of thermoplastic material were studied for Philips RT250 orthovoltage photon beams ranging from 75 to 250 kVp. It was found that Thermo-Shield should be four to five times thicker than lead to achieve the same transmission (less than 5%). However, it did not cause significant degradation in penumbra. Clinical procedures for use are discussed
International Nuclear Information System (INIS)
Kappas, K.
1986-01-01
This work is dedicated to the influence of the human body inhomogeneities on the dose distribution for high energy photons beams used in Radiotherapy. It consists in an experimental part and a theoretical analysis leading to original models of calculation. We study essentially, - the beam quality of the machines used and its influence on some basic dosimetric quantities and on the response of an ionization chamber. - The dose perturbation due to off-axis heterogeneous volumes at off-axis points of measurement; a model is suggested to take into account the perturbation of the multiple scatter. The perturbation of the dose in the transition region, between water equivalent medium and heterogeneous medium (air) is also investigated. The last part is devoted to computer applications of the proposed correction methods and to a comparison between the different computerized treatment planning systems which take into account of inhomogeneities [fr
New beam-position monitor system for upgraded Photon Factory storage ring.
Haga, K; Honda, T; Tadano, M; Obina, T; Kasuga, T
1998-05-01
Accompanying the brilliance-upgrading project at the Photon Factory storage ring, the beam-position monitor (BPM) system has been renovated. The new system was designed to enable precise and fast measurements to correct the closed-orbit distortion (COD), as well as to feed back the orbit position during user runs. There are 42 BPMs newly installed, amounting to a total of 65 BPMs. All of the BPMs are calibrated on the test bench using a coaxially strung metallic wire. The measured electrical offsets are typically 200 micro m in both directions, which is 1/2-1/3 of those of the old-type BPMs. In the signal-processing system, PIN diode switches are employed in order to improve reliability. In the fastest mode, this system is capable of measuring COD within about 10 ms; this fast acquisition will allow fast suppression of the beam movement for frequencies up to 50 Hz using a global feedback system.
International Nuclear Information System (INIS)
Choi, Tae Jin; Kim, Jin Hee; Kim, Ok Bae
1996-01-01
Purpose : To get an acute steepness of dose gradients at outside the target volume in intracranial lesion and a less limitation of beam selection avoiding the high dose at normal brain tissue, this Photon Knife Radiosurgery System was developed in order to provide the three-dimensional dose distribution through the reconstruction of CT scan and the combined stereotactic tranmultiarc beam mode based on linear accelerator photon beam. Methods and Materials : This stereotactic radiosurgery, Photon Knife based on linear accelerator photon beam was provided the non-coplanar multiarc and trans-multiarc irradiations. The stereotactic trans-multiarc beam mode can be obtained from the patient position in decubitus. This study has provided the 3-dimensional isodose curve and anatomical structures with the surface rendering technique. Results : In this study, it shows that the dose distributions of stereotactic beam mode are significantly depended on the selected couch and gantry angle in same collimator size. Practical dose distribution of combined stereotactic trans-multiarc beam has shown a more small rim thickness than that of the non-coplanar multiarc beam mode in axial, sagittal and coronal plane in our study. 3-Dimensional dose line displayed with surface rendering of irregular target shape is helpful to determine the target dose and to predict the prognosis in follow-up radiosurgery. Conclusion : 3-Dimensional dose line displayed with surface rendering of irregular target shape is essential in stereotactic radiosurgery. This combined stereotactic trans-multiarc beam has shown a less limitation of the selection couch and gantry beam angles for the target surrounding critical organs. It has shown that the dose distribution of combined trans-multiarc beam greatly depended on the couch and gantry angles. In our experiments. the absorbed dose has been decreased to 27 % /mm in maximum at the interval of 50% to 80% of isodose line
Electron beam collimation with a photon MLC for standard electron treatments
Mueller, S.; Fix, M. K.; Henzen, D.; Frei, D.; Frauchiger, D.; Loessl, K.; Stampanoni, M. F. M.; Manser, P.
2018-01-01
Standard electron treatments are currently still performed using standard or molded patient-specific cut-outs placed in the electron applicator. Replacing cut-outs and electron applicators with a photon multileaf collimator (pMLC) for electron beam collimation would make standard electron treatments more efficient and would facilitate advanced treatment techniques like modulated electron radiotherapy (MERT) and mixed beam radiotherapy (MBRT). In this work, a multiple source Monte Carlo beam model for pMLC shaped electron beams commissioned at a source-to-surface distance (SSD) of 70 cm is extended for SSDs of up to 100 cm and validated for several Varian treatment units with field sizes typically used for standard electron treatments. Measurements and dose calculations agree generally within 3% of the maximal dose or 2 mm distance to agreement. To evaluate the dosimetric consequences of using pMLC collimated electron beams for standard electron treatments, pMLC-based and cut-out-based treatment plans are created for a left and a right breast boost, a sternum, a testis and a parotid gland case. The treatment plans consist of a single electron field, either alone (1E) or in combination with two 3D conformal tangential photon fields (1E2X). For each case, a pMLC plan with similar treatment plan quality in terms of dose homogeneity to the target and absolute mean dose values to the organs at risk (OARs) compared to a cut-out plan is found. The absolute mean dose to an OAR is slightly increased for pMLC-based compared to cut-out-based 1E plans if the OAR is located laterally close to the target with respect to beam direction, or if a 6 MeV electron beam is used at an extended SSD. In conclusion, treatment plans using cut-out collimation can be replaced by plans of similar treatment plan quality using pMLC collimation with accurately calculated dose distributions.
Finite Element Simulation of GFRP Reinforced Concrete Beam Externally Strengthened With CFRP Plates
Directory of Open Access Journals (Sweden)
Salleh Norhafizah
2017-01-01
Full Text Available The construction technology now has become more and more advanced allowing the development of new technologies or material to replace the previous one and also solved some of the troubles confronted by construction experts. The Glass Fibre Reinforced Polymer (GFRP composite is an alternative to replace the current usage of steel as it is rust proof and stronger in terms of stiffness compared to steel. Furthermore, GFRP bars have a high strength-to-weight ratio, making them attractive as reinforcement for concrete structures. However, the tensile behavior of GFRP bars is characterized by a linear elastic stress–strain relationship up to failure and, therefore, concrete elements reinforced with GFRP reinforcement exhibit brittle failure without warning. Design codes encourage over-reinforced GFRP design since it is more progressive and leads to a less catastrophic failure with a higher degree of deformability. Moreover, because of GFRP low modulus of elasticity, GFRP reinforced concrete members exhibit larger deflections and wider cracks width than steel reinforced concrete. This aims of this paper is to developed 2D Finite Element (FE models that can accurately simulate the respond on an improvement in the deflection of GFRP reinforced concrete beam externally strengthened with CFRP plates on the tension part of beam. The prediction of flexural response according to RCCSA software was also discussed. It was observed that the predicted FE results are given similar result with the experimental measured test data. Base on this good agreement, a parametric study was the performed using the validation FE model to investigate the effect of flexural reinforcement ratio and arrangement of the beams strengthened with different regions of CFRP plates.
Ezzati, A O; Xiao, Y; Sohrabpour, M; Studenski, M T
2015-01-01
The aim of this study was to quantify the DNA damage induced in a clinical megavoltage photon beam at various depths in and out of the field. MCNPX was used to simulate 10 × 10 and 20 × 20 cm(2) 10-MV photon beams from a clinical linear accelerator. Photon and electron spectra were collected in a water phantom at depths of 2.5, 12.5 and 22.5 cm on the central axis and at off-axis points out to 10 cm. These spectra were used as an input to a validated microdosimetric Monte Carlo code, MCDS, to calculate the RBE of induced DSB in DNA at points in and out of the primary radiation field at three depths. There was an observable difference in the energy spectra for photons and electrons for points in the primary radiation field and those points out of field. In the out-of-field region, the mean energy for the photon and electron spectra decreased by a factor of about six and three from the in-field mean energy, respectively. Despite the differences in spectra and mean energy, the change in RBE was photon and electron spectra, these changes do not correlate with a change in RBE in a clinical MV photon beam as the electron spectra are dominated by electrons with energies >20 keV.
Poster — Thur Eve — 37: Respiratory gating with an Elekta flattening filter free photon beam
International Nuclear Information System (INIS)
Péloquin, S; Furstoss, C; Munger, P; Wierzbicki, W; Carrier, J-F
2014-01-01
In cases where surgery is not possible for lung cancer treatment, stereotactic body radiation therapy (SBRT) may be an option. One problem when treating this type of cancer is the motion of the lungs caused by the patient's respiration. It is possible to reduce the impact of this movement with the use of respiratory gating. By combining respiratory gating with a flattening filter free (FFF) photon beam linac, the increased treatment time caused by a reduced beam-on time of respiratory gating methods can be compensated by the inherent increased dose rate of FFF beams. This project's aim is to create hardware and software interfaces allowing free respiration gating on an Elekta Synergy-S linac specially modified to deliver 6 MV FFF photon beams. First, a printed circuit board was created for reading the signal from a Bellows Belt from Philips (a respiration monitor belt) and transmitting an On/Off signal to the accelerator. A software was also developed to visualize patient respiration. Secondly, a FFF model was created with the Pinnacle treatment planning system from Philips. Gamma (Γ) analysis (2%, 2 mm) was used to evaluate model. For fields going from 5.6 × 5.6 to 12 × 12 cm 2 , central axis depth dose model fitting shows an average gamma value of 0.2 and 100% of gamma values remain under the Γ = 1 limit. For smaller fields (0.8 × 0.8 and 1.6 × 1.6 cm 2 ), Pinnacle has more trouble trying to fit the measurements, overestimating dose in penumbra and buildup regions
Directory of Open Access Journals (Sweden)
Teeraphot Supaviriyakit
2017-11-01
Full Text Available This paper presents a nonlinear finite element analysis of non-seismically detailed RC beam column connections under reversed cyclic load. The test of half-scale nonductile reinforced concrete beam-column joints was conducted. The tested specimens represented those of the actual mid-rise reinforced concrete frame buildings designed according to the non-seismic provisions of the ACI building code. The test results show that specimens representing small and medium column tributary area failed in brittle joint shear while specimen representing large column tributary area failed by ductile flexure though no ductile reinforcement details were provided. The nonlinear finite element analysis was applied to simulate the behavior of the specimens. The finite element analysis employs the smeared crack approach for modeling beam, column and joint, and employs the discrete crack approach for modeling the interface between beam and joint face. The nonlinear constitutive models of reinforced concrete elements consist of coupled tension-compression model to model normal force orthogonal and parallel to the crack and shear transfer model to capture the shear sliding mechanism. The FEM shows good comparison with test results in terms of load-displacement relations, hysteretic loops, cracking process and the failure mode of the tested specimens. The finite element analysis clarifies that the joint shear failure was caused by the collapse of principal diagonal concrete strut.
Srinivasan-Rao, Triveni
2002-01-01
A photon generator includes an electron gun for emitting an electron beam, a laser for emitting a laser beam, and an interaction ring wherein the laser beam repetitively collides with the electron beam for emitting a high energy photon beam therefrom in the exemplary form of x-rays. The interaction ring is a closed loop, sized and configured for circulating the electron beam with a period substantially equal to the period of the laser beam pulses for effecting repetitive collisions.
Coherent beam control through inhomogeneous media in multi-photon microscopy
Paudel, Hari Prasad
Multi-photon fluorescence microscopy has become a primary tool for high-resolution deep tissue imaging because of its sensitivity to ballistic excitation photons in comparison to scattered excitation photons. The imaging depth of multi-photon microscopes in tissue imaging is limited primarily by background fluorescence that is generated by scattered light due to the random fluctuations in refractive index inside the media, and by reduced intensity in the ballistic focal volume due to aberrations within the tissue and at its interface. We built two multi-photon adaptive optics (AO) correction systems, one for combating scattering and aberration problems, and another for compensating interface aberrations. For scattering correction a MEMS segmented deformable mirror (SDM) was inserted at a plane conjugate to the objective back-pupil plane. The SDM can pre-compensate for light scattering by coherent combination of the scattered light to make an apparent focus even at a depths where negligible ballistic light remains (i.e. ballistic limit). This problem was approached by investigating the spatial and temporal focusing characteristics of a broad-band light source through strongly scattering media. A new model was developed for coherent focus enhancement through or inside the strongly media based on the initial speckle contrast. A layer of fluorescent beads under a mouse skull was imaged using an iterative coherent beam control method in the prototype two-photon microscope to demonstrate the technique. We also adapted an AO correction system to an existing in three-photon microscope in a collaborator lab at Cornell University. In the second AO correction approach a continuous deformable mirror (CDM) is placed at a plane conjugate to the plane of an interface aberration. We demonstrated that this "Conjugate AO" technique yields a large field-of-view (FOV) advantage in comparison to Pupil AO. Further, we showed that the extended FOV in conjugate AO is maintained over a
Smart x-ray beam position monitor system for the Advanced Photon Source
International Nuclear Information System (INIS)
Shu, D.; Kuzay, T.
1996-01-01
In third-generation synchrotron radiation sources, such as the Advanced Photon Source (APS), the sensitivity and reliability requirements for the x-ray beam position monitors (XBPMs) are much higher than for earlier systems. Noise and contamination signals caused by radiation emitted from the bending magnet become a major problem. The regular XBPM calibration process can only provide signal correction for one set of conditions for the insertion devices (ID). During normal operation, parameters affecting the ID-emitted beam, such as the gap of the ID magnets and the beam current, are the variables. A new smart x-ray beam position monitor system (SBPM) has been conceived and designed for the APS. It has a built in self-learning structure with EEPROM memory that is large enough to open-quote open-quote remember close-quote close-quote a complete set of calibration data covering all the possible operating conditions. During the self-learning mode, the monitor system initializes a series of automatic scan motions with information for different ID setups and records them into the database array. During normal operation, the SBPM corrects the normalized output according to the ID setup information and the calibration database. So that, with this novel system, the SBPM is always calibrating itself with the changing ID set up conditions. copyright 1996 American Institute of Physics
International Nuclear Information System (INIS)
Wu, G.M.; Tsai, B.H.; Kung, S.F.; Wu, C.F.
2011-01-01
Nitride-based thin-film materials have become increasingly important for the high brightness light-emitting diode applications. The improvements in light extraction and lower power consumption are highly desired. Although the internal quantum efficiency of GaN-based LED has been relatively high, only a small fraction of light can be extracted. In this study, a new design of two-dimensional photonic crystal array has been prepared on the top transparent contact layer of indium-tin oxide film to improve the light extraction efficiency using focused ion beam. The acceleration voltage of the Ga dual-beam nanotechnology system SMI 3050 was 30 kV and the ion beam current was 100 pA. The cylindrical air holes had the diameter of 150 nm and depth of 100 nm. The micro photoluminescence analysis results showed that the light output intensity could be 1.5 times of that of the non-patterned control sample. In addition, the structural damage from the focused ion beam drilling of GaN step could be eliminated. The excellent I-V characteristics have been maintained, and the external light extraction efficiency would be still improved for the LED devices. (author)
Directory of Open Access Journals (Sweden)
Nahideh Gharehaghaji
2013-02-01
Full Text Available Introduction: Gold nanoparticles have been used as radiation dose enhancing materials in recent investigations. In the current study, dose enhancement effect of gold nanoparticles on tumor cells was evaluated using Monte Carlo (MC simulation. Methods: We used MCNPX code for MC modeling in the current study. A water phantom and a tumor region with a size of 1×1×1 cm3 loaded with gold nanoparticles were simulated. The macroscopic dose enhancement factor was calculated for gold nanoparticles with sizes of 30, 50, and 100 nm. Also, we simulated different photon beams including mono-energetic beams (50-120 keV, a Cobalt-60 beam, 6 & 18 MV photon beams of a conventional linear accelerator. Results: We found a dose enhancement factor (DEF of from 1.4 to 3.7 for monoenergetic kilovoltage beams, while the DEFs for megavoltage beams were negligible and less than 3% for all GNP sizes and concentrations. The optimum energy for higher DEF was found to be the 90 keV monoenergetic beam. The effect of GNP size was not considerable, but the GNP concentration had a substantial impact on achieved DEF in GNP-based radiation therapy. Conclusion: The results were in close agreement with some previous studies considering the effect of photon energy and GNP concentration on observed DEF. Application of GNP-based radiation therapy using kilovoltage beams is recommended.
Kuess, Peter; Georg, Dietmar; Palmans, Hugo; Lechner, Wolfgang
2016-08-01
beams, they differed by 6.4%. This shows that the DPBQS can provide a differentiation of photon beam characteristics that would remain hidden by the use of a single beam quality specifier, such as %dd(10)x or HVL.
International Nuclear Information System (INIS)
Kuess, Peter; Georg, Dietmar; Lechner, Wolfgang; Palmans, Hugo
2016-01-01
0.8% for 6 MV FF and 6 MV FFF M beams, while for 10 MV beams, they differed by 6.4%. This shows that the DPBQS can provide a differentiation of photon beam characteristics that would remain hidden by the use of a single beam quality specifier, such as %dd(10) x or HVL.
Neutron contamination of Varian Clinac iX 10 MV photon beam using Monte Carlo simulation
International Nuclear Information System (INIS)
Yani, S; Haryanto, F; Arif, I; Tursinah, R; Rhani, M F; Soh, R C X
2016-01-01
High energy medical accelerators are commonly used in radiotherapy to increase the effectiveness of treatments. As we know neutrons can be emitted from a medical accelerator if there is an incident of X-ray that hits any of its materials. This issue becomes a point of view of many researchers. The neutron contamination has caused many problems such as image resolution and radiation protection for patients and radio oncologists. This study concerns the simulation of neutron contamination emitted from Varian Clinac iX 10 MV using Monte Carlo code system. As neutron production process is very complex, Monte Carlo simulation with MCNPX code system was carried out to study this contamination. The design of this medical accelerator was modelled based on the actual materials and geometry. The maximum energy of photons and neutron in the scoring plane was 10.5 and 2.239 MeV, respectively. The number and energy of the particles produced depend on the depth and distance from beam axis. From these results, it is pointed out that the neutron produced by linac 10 MV photon beam in a typical treatment is not negligible. (paper)
Characterization of phenolic pellets for ESR dosimetry in photon beam radiotherapy
International Nuclear Information System (INIS)
Gallo, Salvatore; Veronese, Ivan; Iacoviello, Giuseppina; Panzeca, Salvatore; Bartolotta, Antonio; Longo, Anna; Dondi, Daniele; Gueli, Anna Maria; Loi, Gianfranco; Mones, Eleonora; Marrale, Maurizio
2017-01-01
This work deals with the dosimetric features of a particular phenolic compound (IRGANOX 1076 registered ) for dosimetry of clinical photon beams by using electron spin resonance (ESR) spectroscopy. After the optimization of the ESR readout parameters (namely modulation amplitude and microwave power) to maximise the signal without excessive spectrum distortions, basic dosimetric properties of laboratory-made phenolic dosimeters in pellet form, such as reproducibility, dose-response, sensitivity, linearity and dose rate dependence were investigated. The dosimeters were tested by measuring the depth dose profile of a 6 MV photon beam. A satisfactory intra-batch reproducibility of the ESR signal of the manufactured dosimeters was obtained. The ESR signal proved to increase linearly with increasing dose in the investigated dose range 1-13 Gy. The presence of an intrinsic background signal limits the minimum detectable dose to a value of approximately 0.6 Gy. Reliable and accurate assessment of the dose was achieved, independently of the dose rate. Such characteristics, together with the fact that IRGANOX 1076 registered is almost tissue-equivalent, and the stability of the ESR signal, make these dosimeters promising materials for ESR dosimetric applications in radiotherapy. (orig.)
Characterization of phenolic pellets for ESR dosimetry in photon beam radiotherapy
Energy Technology Data Exchange (ETDEWEB)
Gallo, Salvatore; Veronese, Ivan [Universita degli Studi di Milano, Department of Physics, Milan (Italy); Istituto Nazionale di Fisica Nucleare, Sezione di Milano (Italy); Iacoviello, Giuseppina [Hospital ARNAS-Civico, Medical Physics Department, Palermo (Italy); Panzeca, Salvatore [Universita degli Studi di Palermo, Department of Physics and Chemistry, Palermo (Italy); Istituto Nazionale di Fisica Nucleare-Sezione di Catania, Catania (Italy); Bartolotta, Antonio; Longo, Anna [Universita degli Studi di Palermo, Department of Physics and Chemistry, Palermo (Italy); Dondi, Daniele [Universita degli Studi di Pavia, Department of Chemistry, Pavia (Italy); Istituto Nazionale di Fisica Nucleare, Sezione di Pavia (Italy); Gueli, Anna Maria [Istituto Nazionale di Fisica Nucleare-Sezione di Catania, Catania (Italy); Universita degli Studi di Catania, Department of Physics and Astronomy, PH3DRA Laboratories, Catania (Italy); Loi, Gianfranco; Mones, Eleonora [Azienda Ospedaliero Universitaria Maggiore della Carita, Medical Physics Department, Novara (Italy); Marrale, Maurizio [Universita degli Studi di Palermo, Department of Physics and Chemistry, Palermo (Italy); Istituto Nazionale di Fisica Nucleare-Sezione di Catania, Catania (Italy); Universita degli Studi di Palermo, Advanced Technologies Network Center (ATeN Center), Palermo (Italy)
2017-11-15
This work deals with the dosimetric features of a particular phenolic compound (IRGANOX 1076 {sup registered}) for dosimetry of clinical photon beams by using electron spin resonance (ESR) spectroscopy. After the optimization of the ESR readout parameters (namely modulation amplitude and microwave power) to maximise the signal without excessive spectrum distortions, basic dosimetric properties of laboratory-made phenolic dosimeters in pellet form, such as reproducibility, dose-response, sensitivity, linearity and dose rate dependence were investigated. The dosimeters were tested by measuring the depth dose profile of a 6 MV photon beam. A satisfactory intra-batch reproducibility of the ESR signal of the manufactured dosimeters was obtained. The ESR signal proved to increase linearly with increasing dose in the investigated dose range 1-13 Gy. The presence of an intrinsic background signal limits the minimum detectable dose to a value of approximately 0.6 Gy. Reliable and accurate assessment of the dose was achieved, independently of the dose rate. Such characteristics, together with the fact that IRGANOX 1076 {sup registered} is almost tissue-equivalent, and the stability of the ESR signal, make these dosimeters promising materials for ESR dosimetric applications in radiotherapy. (orig.)
International Nuclear Information System (INIS)
Rosser, K.E.; Williams, A.J.
1999-01-01
At present, the primary standard of absorbed dose to water at NPL in high energy photon beams is a graphite calorimeter. However the quantity of interest in radiation dosimetry is absorbed dose to water. Therefore, a new absorbed dose to water standard based on water calorimetry is being developed at NPL. The calorimeter operates at 4 deg. C, with temperature control being provided by a combination of liquid and air cooling. The sealed glass inner vessel of the calorimeter has been designed to minimise the effect of non-water materials on the measurement of absorbed dose. Measurements of absorbed dose to water made in 6, 10 and 19 MV photon beams agreed within the measurement uncertainties with those determined using the primary standard graphite calorimeter. Also the absorbed dose to water measured using the water calorimeter agrees with that based on the air kerma standards for 60 Co γ-radiation within the uncertainties. The development of the water calorimeter will lead to a very robust dosimetry system at NPL, where the absorbed dose to water can be determined using three independent techniques. (author)
First doubly polarised photoproduction on 3He at the photon beam of MAMI
International Nuclear Information System (INIS)
Aguar Bartolome, Patricia
2010-11-01
A first experiment with a polarised 3 He target was carried out in July 2009 at the MAMI accelerator in Mainz in a photon energy range between 200 MeV and 800 MeV. The aim of this measurement was to investigate the Gerasimov-Drell-Hearn sum rule on the neutron. The use of the data obtained with the polarised 3 He target, compared to existing data on the deuteron, gives a complementary and more direct access to the neutron, due to the spin structure of the 3 He. The measurement of the helicity dependence of the inclusive total photoabsorption cross section required a beam of tagged circularly polarised photons incident on the longitudinally polarised 3 He target. The data were taken using the 4π Crystal Ball photon spectrometer in combination with TAPS as a forward wall and complemented by a threshold Cherenkov detector used to on-line suppress the background from electromagnetic events. The development and preparation of the different components of the 3 He experimental setup was an important part of this work and are described in detail in this thesis. The detector system and the analysis method were tested by the measurement of the unpolarised total inclusive photoabsorption cross section on liquid hydrogen. The results obtained are in good agreement with previous published data. Preliminary results of the unpolarised total photoabsorption cross section, as well as the helicity dependent photoabsorption cross section difference on 3 He compared with several theoretical models will also be presented. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Cheng, Chee-Wai; Das, Indra J.; Zhao, Li; Wolanski, Mark; Johnstone, Peter A.S.; Buchsbaum, Jeffrey C. [IU Health Proton Therapy Center, Bloomington (United States); Dept. of Radiation Oncology, Indiana Univ. School of Medicine, Indianapolis (United States)], e-mail: ccheng1@iuhealth.org; Srivastava, Shiv P. [Dept. of Radiation Oncology, Indiana Univ. School of Medicine, Indianapolis (United States); Dept. of Radiation Oncology, Reid Hospital, Richmond (United States); Simmons, Joseph [IU Health Proton Therapy Center, Bloomington (United States)
2013-04-15
Purpose: To investigate the moving gap region dosimetry in proton beam cranio-spinal irradiation (CSI) to provide optimal dose uniformity across the treatment volume. Material and methods: Proton beams of ranges 11.6 cm and 16 cm are used for the spine and the brain fields, respectively. Beam profiles for a 30 cm snout are first matched at the 50% level (hot match) on the computer. Feathering is simulated by shifting the dose profiles by a known distance two successive times to simulate a 2 x feathering scheme. The process is repeated for 2 mm and 4 mm gaps. Similar procedures are used to determine the dose profiles in the moving gap for a series of gap widths, 0-10 mm, and feathering step sizes, 4-10 mm, for a Varian iX 6MV beam. The proton and photon dose profiles in the moving gap region are compared. Results: The dose profiles in the moving gap exhibit valleys and peaks in both proton and photon beam CSI. The dose in the moving gap for protons is around 100% or higher for 0 mm gap, for both 5 and 10 mm feathering step sizes. When the field gap is comparable or larger than the penumbra, dose minima as low as 66% is obtained. The dosimetric characteristics for 6 MV photon beams can be made similar to those of the protons by appropriately combining gap width and feathering step size. Conclusion: The dose in the moving gap region is determined by the lateral penumbras, the width of the gap and the feathering step size. The dose decreases with increasing gap width or decreasing feathering step size. The dosimetric characteristics are similar for photon and proton beams. However, proton CSI has virtually no exit dose and is beneficial for pediatric patients, whereas with photon beams the whole lung and abdomen receive non-negligible exit dose.
A new approach to film dosimetry for high-energy photon beams using organic plastic scintillators
International Nuclear Information System (INIS)
Yeo, I.J.; Wang, C.-K.C.; Burch, S.E.
1999-01-01
Successful radiotherapy relies on accurate dose measurement. Traditional dosimeters such as ion chambers, TLDs and diodes have disadvantages such as relatively long measurement time and poor spatial resolution. These drawbacks become more serious problems for dynamic beams (i.e. with the use of dynamic wedges or even the intensity modulation technique). X-ray film, an integrating dosimeter, may not be associated with the above disadvantages and problems. However, there are several major issues regarding use of x-ray film for routine dosimetry, including the over-response of the film to low-energy photons, variations in the dose response curve (nonlinearity), lack of reproducibility due to variation in processing, etc. This paper addresses the first problem. That is, x-ray film over-responds to low-energy photons (energies below 400 keV), and thus generates unacceptably inaccurate dosimetric data compared with ion-chamber data. To overcome the over-response problem of x-ray film in a phantom, a scintillation method has been investigated. In this method, a film is sandwiched by two plastic scintillation screens to enhance the film response to upstream electrons, and therefore minimize the over-response caused by low-energy photons. The sandwiched system was tested with a 4 MV linac beam. The result shows that, depending on the uniformity of the scintillation screens, the depth-dose distribution obtained from the sandwich system can be made to agree well with that obtained from ion chambers. However, the required high degree of uniformity remains a challenge for the scintillation screen manufacturers. (author)
Energy Technology Data Exchange (ETDEWEB)
García-Garduño, O. A., E-mail: oagarciag@innn.edu.mx, E-mail: amanda.garcia.g@gmail.com [Laboratorio de Física Médica, Instituto Nacional de Neurología y Neurocirugía, Mexico City 14269, México and Centro de Investigación en Ciencia Aplicada y Tecnología Avanzada, Unidad Legaria, Instituto Politécnico Nacional, Legaria 694, México City 11500, México (Mexico); Rodríguez-Ponce, M. [Departamento de Biofísica, Instituto Nacional de Cancerología, Mexico City 14080, México (Mexico); Gamboa-deBuen, I. [Instituto de Ciencias Nucleares, Universidad Nacional Autónoma de México, Ciudad Universitaria, Mexico City 04510 (Mexico); Rodríguez-Villafuerte, M. [Instituto de Física, Universidad Nacional Autónoma de México, Ciudad Universitaria, Mexico City 04510 (Mexico); Galván de la Cruz, O. O. [Laboratorio de Física Médica, Instituto Nacional de Neurología y Neurocirugía, Mexico City 14269, México (Mexico); and others
2014-09-15
Purpose: To assess the impact of the detector used to commission small photon beams on the calculated dose distribution in stereotactic radiosurgery (SRS). Methods: In this study, six types of detectors were used to characterize small photon beams: three diodes [a silicon stereotactic field diode SFD, a silicon diode SRS, and a silicon diode E], an ionization chamber CC01, and two types of radiochromic film models EBT and EBT2. These detectors were used to characterize circular collimated beams that were generated by a Novalis linear accelerator. This study was conducted in two parts. First, the following dosimetric data, which are of particular interest in SRS, were compared for the different detectors: the total scatter factor (TSF), the tissue phantom ratios (TPRs), and the off-axis ratios (OARs). Second, the commissioned data sets were incorporated into the treatment planning system (TPS) to compare the calculated dose distributions and the dose volume histograms (DVHs) that were obtained using the different detectors. Results: The TSFs data measured by all of the detectors were in good agreement with each other within the respective statistical uncertainties: two exceptions, where the data were systematically below those obtained for the other detectors, were the CC01 results for all of the circular collimators and the EBT2 film results for circular collimators with diameters below 10.0 mm. The OAR results obtained for all of the detectors were in excellent agreement for all of the circular collimators. This observation was supported by the gamma-index test. The largest difference in the TPR data was found for the 4.0 mm circular collimator, followed by the 10.0 and 20.0 mm circular collimators. The results for the calculated dose distributions showed that all of the detectors passed the gamma-index test at 100% for the 3 mm/3% criteria. The aforementioned observation was true regardless of the size of the calculation grid for all of the circular collimators
A method for photon beam Monte Carlo multileaf collimator particle transport
Siebers, Jeffrey V.; Keall, Paul J.; Kim, Jong Oh; Mohan, Radhe
2002-09-01
Monte Carlo (MC) algorithms are recognized as the most accurate methodology for patient dose assessment. For intensity-modulated radiation therapy (IMRT) delivered with dynamic multileaf collimators (DMLCs), accurate dose calculation, even with MC, is challenging. Accurate IMRT MC dose calculations require inclusion of the moving MLC in the MC simulation. Due to its complex geometry, full transport through the MLC can be time consuming. The aim of this work was to develop an MLC model for photon beam MC IMRT dose computations. The basis of the MC MLC model is that the complex MLC geometry can be separated into simple geometric regions, each of which readily lends itself to simplified radiation transport. For photons, only attenuation and first Compton scatter interactions are considered. The amount of attenuation material an individual particle encounters while traversing the entire MLC is determined by adding the individual amounts from each of the simplified geometric regions. Compton scatter is sampled based upon the total thickness traversed. Pair production and electron interactions (scattering and bremsstrahlung) within the MLC are ignored. The MLC model was tested for 6 MV and 18 MV photon beams by comparing it with measurements and MC simulations that incorporate the full physics and geometry for fields blocked by the MLC and with measurements for fields with the maximum possible tongue-and-groove and tongue-or-groove effects, for static test cases and for sliding windows of various widths. The MLC model predicts the field size dependence of the MLC leakage radiation within 0.1% of the open-field dose. The entrance dose and beam hardening behind a closed MLC are predicted within +/-1% or 1 mm. Dose undulations due to differences in inter- and intra-leaf leakage are also correctly predicted. The MC MLC model predicts leaf-edge tongue-and-groove dose effect within +/-1% or 1 mm for 95% of the points compared at 6 MV and 88% of the points compared at 18 MV
A method for photon beam Monte Carlo multileaf collimator particle transport
Energy Technology Data Exchange (ETDEWEB)
Siebers, Jeffrey V. [Department of Radiation Oncology, Medical College of Virginia Hospitals, Virginia Commonwealth University, Richmond, VA (United States)]. E-mail: jsiebers@vcu.edu; Keall, Paul J.; Kim, Jong Oh; Mohan, Radhe [Department of Radiation Oncology, Medical College of Virginia Hospitals, Virginia Commonwealth University, Richmond, VA (United States)
2002-09-07
Monte Carlo (MC) algorithms are recognized as the most accurate methodology for patient dose assessment. For intensity-modulated radiation therapy (IMRT) delivered with dynamic multileaf collimators (DMLCs), accurate dose calculation, even with MC, is challenging. Accurate IMRT MC dose calculations require inclusion of the moving MLC in the MC simulation. Due to its complex geometry, full transport through the MLC can be time consuming. The aim of this work was to develop an MLC model for photon beam MC IMRT dose computations. The basis of the MC MLC model is that the complex MLC geometry can be separated into simple geometric regions, each of which readily lends itself to simplified radiation transport. For photons, only attenuation and first Compton scatter interactions are considered. The amount of attenuation material an individual particle encounters while traversing the entire MLC is determined by adding the individual amounts from each of the simplified geometric regions. Compton scatter is sampled based upon the total thickness traversed. Pair production and electron interactions (scattering and bremsstrahlung) within the MLC are ignored. The MLC model was tested for 6 MV and 18 MV photon beams by comparing it with measurements and MC simulations that incorporate the full physics and geometry for fields blocked by the MLC and with measurements for fields with the maximum possible tongue-and-groove and tongue-or-groove effects, for static test cases and for sliding windows of various widths. The MLC model predicts the field size dependence of the MLC leakage radiation within 0.1% of the open-field dose. The entrance dose and beam hardening behind a closed MLC are predicted within {+-}1% or 1 mm. Dose undulations due to differences in inter- and intra-leaf leakage are also correctly predicted. The MC MLC model predicts leaf-edge tongue-and-groove dose effect within {+-}1% or 1 mm for 95% of the points compared at 6 MV and 88% of the points compared at 18 MV
Evaluation of a 50-MV photon therapy beam from a racetrack microtron using MCNP4B Monte Carlo code
International Nuclear Information System (INIS)
Gudowska, I.; Svensson, R.
2001-01-01
High energy photon therapy beam from the 50 MV racetrack microtron has been evaluated using the Monte Carlo code MCNP4B. The spatial and energy distribution of photons, radial and depth dose distributions in the phantom are calculated for the stationary and scanned photon beams from different targets. The calculated dose distributions are compared to the experimental data using a silicon diode detector. Measured and calculated depth-dose distributions are in fairly good agreement, within 2-3% for the positions in the range 2-30 cm in the phantom, whereas the larger discrepancies up to 10% are observed in the dose build-up region. For the stationary beams the differences in the calculated and measured radial dose distributions are about 2-10%. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Abat, E; Arik, E [Bogazici University, Faculty of Sciences, Department of Physics, TR-80815 Bebek-Istanbul (Turkey); Abdallah, J M [Institut de Fisica d' Altes Energies, IFAE, Universitat Autonoma de Barcelona, Edifici Cn, ES-08193 Bellaterra, Barcelona (Spain); Addy, T N [Hampton University, Department of Physics, Hampton, VA 23668 (United States); Adragna, P [Queen Mary, University of London, Mile End Road, E1 4NS, London (United Kingdom); Aharrouche, M [Universitaet Mainz, Institut fuer Physik, Staudinger Weg 7, DE-55099 (Germany); Ahmad, A [Insitute of Physics, Academia Sinica, TW-Taipei 11529, Taiwan (China); Akesson, T P A [Lunds universitet, Naturvetenskapliga fakulteten, Fysiska institutionen, Box 118, SE-221 00, Lund (Sweden); Aleksa, M; Anghinolfi, F; Baron, S [European Laboratory for Particle Physics (CERN), CH-1211 Geneva 23 (Switzerland); Alexa, C [National Institute of Physics and Nuclear Engineering (Bucharest -IFIN-HH), P.O. Box MG-6, R-077125 Bucharest (Romania); Anderson, K [University of Chicago, Enrico Fermi Institute, 5640 S. Ellis Avenue, Chicago, IL 60637 (United States); Andreazza, A; Banfi, D [INFN Sezione di Milano, via Celoria 16, IT-20133 Milano (Italy); Antonaki, A; Arabidze, G [University of Athens, Nuclear and Particle Physics Department of Physics, Panepistimiopouli Zografou, GR 15771 Athens (Greece); Atkinson, T [School of Physics, University of Melbourne, AU-Parkvill, Victoria 3010 (Australia); Baines, J [Rutherford Appleton Laboratory, Science and Technology Facilities Council, Harwell Science and Innovation Campus, Didcot OX11 0QX (United Kingdom); Baker, O K, E-mail: stathes.paganis@cern.ch [Yale University, Department of Physics, PO Box 208121, New Haven, CT06520-8121 (United States)
2011-04-01
The reconstruction of photons in the ATLAS detector is studied with data taken during the 2004 Combined Test Beam, where a full slice of the ATLAS detector was exposed to beams of particles of known energy at the CERN SPS. The results presented show significant differences in the longitudinal development of the electromagnetic shower between converted and unconverted photons as well as in the total measured energy. The potential to use the reconstructed converted photons as a means to precisely map the material of the tracker in front of the electromagnetic calorimeter is also considered. All results obtained are compared with a detailed Monte-Carlo simulation of the test-beam setup which is based on the same simulation and reconstruction tools as those used for the ATLAS detector itself.
International Nuclear Information System (INIS)
Abat, E; Arik, E; Abdallah, J M; Addy, T N; Adragna, P; Aharrouche, M; Ahmad, A; Akesson, T P A; Aleksa, M; Anghinolfi, F; Baron, S; Alexa, C; Anderson, K; Andreazza, A; Banfi, D; Antonaki, A; Arabidze, G; Atkinson, T; Baines, J; Baker, O K
2011-01-01
The reconstruction of photons in the ATLAS detector is studied with data taken during the 2004 Combined Test Beam, where a full slice of the ATLAS detector was exposed to beams of particles of known energy at the CERN SPS. The results presented show significant differences in the longitudinal development of the electromagnetic shower between converted and unconverted photons as well as in the total measured energy. The potential to use the reconstructed converted photons as a means to precisely map the material of the tracker in front of the electromagnetic calorimeter is also considered. All results obtained are compared with a detailed Monte-Carlo simulation of the test-beam setup which is based on the same simulation and reconstruction tools as those used for the ATLAS detector itself.
The contribution to surface dose form air scatter in mega voltage photon beams
International Nuclear Information System (INIS)
Carolan, M.G.; Butson, M.; Metcalfe, P.
1996-01-01
Full text: The minimisation of surface dose is an important requirement in radiotherapy in order to avoid undesirable skin reactions. For this reason significant effort has been expended to avoid and understand photon and electron scatter in the heads of linear accelerators which may contribute to surface dose. In this study we have examined the contribution to surface dose which arises due to scatter in the air above the patient. Experimental investigations of air contributions are difficult to design and execute. Therefore we have used Monte Carlo calculations to determine the effect that the presence of air has on surface dose. Methods: The Los Alamos Monte Carlo Neutron and Photon transport code, MCNP4A which incorporates the ETRAN electron transport code from the Integrated TIGER Series of codes was used for our simulations. The geometry used in the model was a 30 cm cube of water. The dose was tallied in cylindrical elements of 7 cm diameter along the axis of the photon beam. For the first millimetre along the beam axis in the phantom, the dose was determined at 0.1 mm increments in 0.1 mm thick volumes. For depths between 1.0 mm and 15.0 mm the dose was determined every 1 mm in 1 mm cylindrical volumes. This yields a depth dose profile with fine spatial resolution near the phantom surface. Dose was also tallied at depths of 5.0, 10.0, 15.0 and 20.0 cm. The simulations were done assuming a 6 MV photon source with a diameter of 1.5 cm, a gaussian intensity profile and a photon energy spectrum based on Mohan et al. (Med. Phys. 12 (1985) 592). No accelerator head geometry was modelled. The field size was defined by virtual collimators which were simply thin regions of zero photon importance and therefore do not contribute to photon or electron scatter. All simulations were run for sufficient particle histories (∼2x10 7 - 5x10 7 source photons) to give statistical uncertainties of ≤ 10% and in most cases ≤ 5%. Fields of size 10, 15, 20 and 25 cm were used
International Nuclear Information System (INIS)
LIU, B; Zhu, T
2014-01-01
Purpose: The dose in the buildup region of a photon beam is usually determined by the transport of the primary secondary electrons and the contaminating electrons from accelerator head. This can be quantified by the electron disequilibrium factor, E, defined as the ratio between total dose and equilibrium dose (proportional to total kerma), E = 1 in regions beyond buildup region. Ecan be different among accelerators of different models and/or manufactures of the same machine. This study compares E in photon beams from different machine models/ Methods: Photon beam data such as fractional depth dose curve (FDD) and phantom scatter factors as a function of field size and phantom depth were measured for different Linac machines. E was extrapolated from these fractional depth dose data while taking into account inverse-square law. The ranges of secondary electron were chosen as 3 and 6 cm for 6 and 15 MV photon beams, respectively. The field sizes range from 2x2 to 40x40 cm 2 . Results: The comparison indicates the standard deviations of electron contamination among different machines are about 2.4 - 3.3% at 5 mm depth for 6 MV and 1.2 - 3.9% at 1 cm depth for 15 MV for the same field size. The corresponding maximum deviations are 3.0 - 4.6% and 2 - 4% for 6 and 15 MV, respectively. Both standard and maximum deviations are independent of field sizes in the buildup region for 6 MV photons, and slightly decreasing with increasing field size at depths up to 1 cm for 15 MV photons. Conclusion: The deviations of electron disequilibrium factor for all studied Linacs are less than 3% beyond the depth of 0.5 cm for the photon beams for the full range of field sizes (2-40 cm) so long as they are from the same manufacturer
Absorbed dose calibration factors for parallel-plate chambers in high energy photon beams
International Nuclear Information System (INIS)
McEwen, M.R.; Duane, S.; Thomas, R.A.S.
2002-01-01
An investigation was carried out into the performance of parallel-plate chambers in 60 Co and MV photon beams. The aim was to derive calibration factors, investigate chamber-to-chamber variability and provide much-needed information on the use of parallel-plate chambers in high-energy X-ray beams. A set of NE2561/NE2611 reference chambers, calibrated against the primary standard graphite calorimeter is used for the dissemination of absorbed dose to water. The parallel-plate chambers were calibrated by comparison with the NPL reference chambers in a water phantom. Two types of parallel-plate chamber were investigated - the NACP -02 and Roos and measurements were made at 60 C0 and 6 linac photon energies (6-19 MV). Calibration factors were derived together with polarity corrections. The standard uncertainty in the calibration of a chamber in terms of absorbed dose to water is estimated to be ±0.75%. The results of the polarity measurements were somewhat confusing. One would expect the correction to be small and previous measurements in electron beams have indicated that there is little variation between chambers of these types. However, some chambers gave unexpectedly large polarity corrections, up to 0.8%. By contrast the measured polarity correction for a NE2611 chamber was less than 0.13% at all energies. The reason for these large polarity corrections is not clear, but experimental error and linac variations have been ruled out. By combining the calibration data for the different chambers it was possible to obtain experimental k Q factors for the two chamber types. It would appear from the data that the variations between chambers of the same type are random and one can therefore define a generic curve for each chamber type. These are presented in Figure 1, together with equivalent data for two cylindrical chamber types - NE2561/NE2611 and NE2571. As can be seen, there is a clear difference between the curves for the cylindrical chambers and those for the
Czarnecki, Damian; Poppe, Björn; Zink, Klemens
2017-06-01
The impact of removing the flattening filter in clinical electron accelerators on the relationship between dosimetric quantities such as beam quality specifiers and the mean photon and electron energies of the photon radiation field was investigated by Monte Carlo simulations. The purpose of this work was to determine the uncertainties when using the well-known beam quality specifiers or energy-based beam specifiers as predictors of dosimetric photon field properties when removing the flattening filter. Monte Carlo simulations applying eight different linear accelerator head models with and without flattening filter were performed in order to generate realistic radiation sources and calculate field properties such as restricted mass collision stopping power ratios (L¯/ρ)airwater, mean photon and secondary electron energies. To study the impact of removing the flattening filter on the beam quality correction factors k Q , this factor for detailed ionization chamber models was calculated by Monte Carlo simulations. Stopping power ratios (L¯/ρ)airwater and k Q values for different ionization chambers as a function of TPR1020 and %dd(10) x were calculated. Moreover, mean photon energies in air and at the point of measurement in water as well as mean secondary electron energies at the point of measurement were calculated. The results revealed that removing the flattening filter led to a change within 0.3% in the relationship between %dd(10) x and (L¯/ρ)airwater, whereby the relationship between TPR1020 and (L¯/ρ)airwater changed up to 0.8% for high energy photon beams. However, TPR1020 was a good predictor of (L¯/ρ)airwater for both types of linear accelerator with energies filter within 1.1% and 1.6% was observed for TPR1020 and %dd(10) x respectively. The results of this study have shown that removing the flattening filter led to a change in the relationship between the well-known beam quality specifiers and dosimetric quantities at the point of measurement
International Nuclear Information System (INIS)
Indogo, V.
2014-07-01
The aim of this research was to observe dose distributions in the vicinity of titanium prosthetic implants during radiotherapy procedures. Data were obtained using a locally fabricated tissue equivalent phantom CT images, and in blue water phantom with titanium prosthesis which was irradiated with 60 Co gamma radiation and Elekta Platform photon beams. Images obtained were loaded into Prowess Panther and Oncentra treatment planning systems (TPSs) for dose simulations. Prowess TPS (1.25 MeV) estimated lesser errors whilst Oncentra (6 and 15 MV) dose simulations yielded large variations. Proximal ends of the metal recorded slight increase in doses as a rcsult of backscatter with dose increment below acceptable tolerance of ±3%. Doses measured decreases on the distal side of the prosthesis at a distance less than d max from the plate on each beam energy. Beyond certain depth along the axis, depth doses increased slightly mainly due to increase in electron fluence by portions receiving unperturbed dose. An increase in the plate thickness showed a corresponding decrease on percentage depth dose. A reduction in the above trend was also noticed with an increase in beam energy primarily because scattered photons are more forwardly directed. Prowess TPS (convolution superposition algorithm) was found to be better at reducing dose variation than OMP (collapse cone algorithm) when correction for artifact. Manual calculations on blue phantom data agree with results from Prowess. Oncentra is not capable of simulating dose around titanium prosthesis as its range of densities, 0.00121 to 2.83, excludes titanium density (rED for titanium is 3.74). (au)
Characterization of the phantom material virtual water in high-energy photon and electron beams.
McEwen, M R; Niven, D
2006-04-01
The material Virtual Water has been characterized in photon and electron beams. Range-scaling factors and fluence correction factors were obtained, the latter with an uncertainty of around 0.2%. This level of uncertainty means that it may be possible to perform dosimetry in a solid phantom with an accuracy approaching that of measurements in water. Two formulations of Virtual Water were investigated with nominally the same elemental composition but differing densities. For photon beams neither formulation showed exact water equivalence-the water/Virtual Water dose ratio varied with the depth of measurement with a difference of over 1% at 10 cm depth. However, by using a density (range) scaling factor very good agreement (water and Virtual Water at all depths was obtained. In the case of electron beams a range-scaling factor was also required to match the shapes of the depth dose curves in water and Virtual Water. However, there remained a difference in the measured fluence in the two phantoms after this scaling factor had been applied. For measurements around the peak of the depth-dose curve and the reference depth this difference showed some small energy dependence but was in the range 0.1%-0.4%. Perturbation measurements have indicated that small slabs of material upstream of a detector have a small (<0.1% effect) on the chamber reading but material behind the detector can have a larger effect. This has consequences for the design of experiments and in the comparison of measurements and Monte Carlo-derived values.
Proceeding of the workshop on gamma-ray spectroscopy utilizing heavy-ion, photon and RI beams
Energy Technology Data Exchange (ETDEWEB)
Oshima, Masumi; Sugita, Michiaki; Hayakawa, Takehito [eds.
1998-03-01
Three time since 1992, we have held the symposia entitled `Joint Spectroscopy Experiments Utilizing JAERI Tandem-Booster Accelerator` at the Tokai Research Establishment. In the symposia, we have mainly discussed the plans of experiments to be done in this joint program. The joint program started in 1994. Several experiments have been made since and some new results have already come up. This symposium `Gamma-ray Spectroscopy utilizing heavy-ion, Photon and RI beams` was held at Tokai Research Establishment of JAERI. Because this symposium is the first occasion after the program started, the first purpose of the symposium is to present and discuss the experimental results so far obtained using the JAERI Tandem-Booster. The second purpose of the symposium is to discuss new possibilities of gamma-ray spectroscopy using new resources such as RI-beam and Photon-beam. The participants from RIKEN, Tohoku University and JAERI Neutron Science Research Center presented the future plans of experiments with RI-beam at each facility. Compared with these nuclear beams, photon beam provides a completely new tool for the {gamma}-ray spectroscopy, which is achieved by inverse Compton scattering between high-energy electron and laser beams. The 23 of the presented papers are indexed individually. (J.P.N.)
International Nuclear Information System (INIS)
Tannenbaum, M.J.
1980-01-01
In the range of electron energies available at Fermilab, 100 GeV less than or equal to E less than or equal to 500 GeV, coherent Bremsstrahlung in crystals, particularly diamond, gives a huge enhancement to the equivalent photon spectrum at large values of x where x = k/E. The photons in this enhancement are polarized. Requirements on electron beam energy spread, angular divergence and spot size imposed by the use of a diamond as a radiator are discussed. The physics program emphasizes hard processes and tests of QCD using polarization
Electron and photon-beam induced reactions of adsorbed disilane: Low-temperature thin-film growth
International Nuclear Information System (INIS)
Bozso, F.; Avouris, Ph.
1991-01-01
Electrons and photons of sufficient energy can cause fragmentation and desorption of adsorbed molecules or fragments of them, by inducing electronic excitations to dissociative states. The surface species after such excitations are mostly of highly reactive radical character, which readily react with the substrate and with other molecular or radical species in the adsorbed layer. This paper discusses the adsorption, thermal and electron/photon-beam induced reactions of disilane, oxygen and ammonia on Si(111)-7x7, and the electron/photon-induced growth of silicon, silicon dioxide and silicon nitride films at 100K
Dosimetric characterization of the 18-MV photon beam from the Siemens Mevatron 77 linear accelerator
International Nuclear Information System (INIS)
Palta, J.R.; Meyer, J.A.; Hogstrom, K.R.
1984-01-01
A comprehensive set of dosimetric measurements has been made on the Mevatron 77.80.67 18-MV photon beam. Percentage depth dose, dose in the buildup region, field size dependence of output, transmission through lead, tray attenuation, and isodose curves for the open and wedged fields were measured using an ionization chamber in water and polystyrene phantoms. These dosimetric measurements sufficiently characterized the beam to permit clinical use. The depth dose at 10-cm depth for a 10 x 10 cm 2 field at 100-cm source-to-skin distance (SSD) is 80.9%, which meets design specifications. Central axis depth-dose data were fitted to within 0.5% by a set of polynomial equations utilizing a two-dimensional linear regression analysis. Tissue--maximum ratios calculated from depth-dose data agree with measured data to within 2%. Output differences as large as 2.5% were measured for rectangular fields depending on which collimator jaws defined the long dimension of the field. The field size dependence of output was fit to within +- 0.1% by a linear regression. The half-value thickness of the beam was measured to be 13 mm of lead
Dose determination in radiotherapy for photon beams modified by static intensity modulators
International Nuclear Information System (INIS)
Castellanos Lopez, M.E.
1998-01-01
The static intensity modulators, used in radiotherapy, modify the spectral composition of the beam and lead to specific problems of the dose calculation. The aim of this work was to establish a three dimensional calculation, global and accurate, adapted to the primary-diffused separation algorithm and valid for any static modulator type. A theoretical study, experimentally verified, allowed the evaluation of the primary fluence, resulting from metallic sheets placed between photons beams of 6 to 23 MV nominal energy. It has been showed that the diffused, coming from the modulators, could be neglected for weak thickness and for the relative dose variation. In return it leads to significant variations of many % on the absolute dose and must be take into account for the bigger thicknesses. Corrective methods for the primary fluence have been proposed. From the energy spectra of the beam, the metallic modulator influence has been studied on the primary and diffused components of the dose and improvements of the calculation method have been proposed. These improvements are based on the modulator representation as a transmission matrix and on semi-empirical corrective factors. (A.L.B.)
The curvature of sensitometric curves for Kodak XV-2 film irradiated with photon and electron beams.
van Battum, L J; Huizenga, H
2006-07-01
Sensitometric curves of Kodak XV-2 film, obtained in a time period of ten years with various types of equipment, have been analyzed both for photon and electron beams. The sensitometric slope in the dataset varies more than a factor of 2, which is attributed mainly to variations in developer conditions. In the literature, the single hit equation has been proposed as a model for the sensitometric curve, as with the parameters of the sensitivity and maximum optical density. In this work, the single hit equation has been translated into a polynomial like function as with the parameters of the sensitometric slope and curvature. The model has been applied to fit the sensitometric data. If the dataset is fitted for each single sensitometric curve separately, a large variation is observed for both fit parameters. When sensitometric curves are fitted simultaneously it appears that all curves can be fitted adequately with a sensitometric curvature that is related to the sensitometric slope. When fitting each curve separately, apparently measurement uncertainty hides this relation. This relation appears to be dependent only on the type of densitometer used. No significant differences between beam energies or beam modalities are observed. Using the intrinsic relation between slope and curvature in fitting sensitometric data, e.g., for pretreatment verification of intensity-modulated radiotherapy, will increase the accuracy of the sensitometric curve. A calibration at a single dose point, together with a predetermined densitometer-dependent parameter ODmax will be adequate to find the actual relation between optical density and dose.
Design and performance of the beam loss monitor system for the Advanced Photon Source
International Nuclear Information System (INIS)
Patterson, D.
1994-01-01
The design of the beam loss monitor system for the Argonne National Laboratory Advanced Photon Source is based on using a number of air dielectric coaxial cables as long ionization chambers. The coaxial cables are multiplexed into a high sensitivity DC current-to-voltage converter, which provides an output proportional to the average loss rate over the length of the multiplexed cable. Losses of sufficient amplitude generate measurable voltage pulses on the coaxial cable at a location near the loss point. Multiplexed pulse timing circuits determine the location of the losses by measuring the time at which these voltage pulses arrive at the beginning of the coaxial cable. The loss monitor system has been tested on the SPEAR accelerator at SSRL and was demonstrated to be as sensitive as the DCCT. Preliminary performance data from the APS injector show that the sensitivities of the current-to-voltage, converter circuit are about ten picoamperes of loss monitor signal per picocoulomb per second beam loss rate. The corresponding pulse sensitivity is about 28 μV pulse amplitude in the coaxial cable per picocoulomb of loss. Both these sensitivities are at 300-MeV beam energies. The loss monitor has proven useful in initial commissioning of the injector. Further data will be available as accelerator construction and commissioning continue
Advanced R ampersand D for electron and photon beams at Brookhaven National Laboratory
International Nuclear Information System (INIS)
Kirk, H.G.
1989-08-01
The Brookhaven Accelerator Test Facility consists of a 50-MeV linear accelerator and a laser system capable of generating short (a few picoseconds) laser pulses at both UV (266 nm) and infrared (10 μm) wavelengths. With these systems in place, the ATF has unique capabilities for the study of fundamental interactions between charged-particle beams and intense electromagnetic radiation. The principal research goals of the Accelerator Test Facility (ATF) axe the following. Laser Acceleration Program: We wig study the principles and techniques of particle acceleration at ultra-high frequencies (up to 30 THz) and with very high acceleration gradients (up to 1 GV/m). Production of Coherent Radiation: We wish to develop the next generation of photon sources with features like (a) short pulses (picoseconds or less), (b) coherence, and (c) high peak power. All of these attributes can be provided by free-electron lasers. High-brightness sources: A common denominator for the above programs is the need for electron beams with very small transverse and longitudinal emittances. We will devote a substantial amount of our resources to the production and understanding of electron beams that have these attributes. We will build advanced electron sources such as switched-power devices and rf guns with photocathodes. Important applications of this line of research include the development of high-luminosity linear colliders and free-electron lasers in the XUV regime
Charged-particle beam diagnostics for the Advanced Photon Source (APS)
International Nuclear Information System (INIS)
Lumpkin, A.H.; Decker, G.; Kahana, E.; Patterson, D.; Sellyey, W.; Wang, X.; Chung, Y.
1992-01-01
Plans, prototypes, and initial test results for the charged-particle beam (e - , e + ) diagnostic systems on the injector rings, their transport lines, and the storage ring for the Advanced Photon Source (APS) are presented. The APS will be a synchrotron radiation user facility with one of the world's brightest x-ray sources in the 10-keV to 100-keV regime. Its 200-MeV electron linac, 450-MeV positron linac, positron accumulator ring, 7-GeV booster synchrotron, 7-GeV storage ring, and undulator test lines will also demand the development and demonstration of key particle-beam characterization techniques over a wide range of parameter space. Some of these parameter values overlap or approach those projected for fourth generation light sources (linac-driven FELs and high brightness storage rings) as described at a recent workshop. Initial results from the diagnostics prototypes on the linac test stand operating at 45-MeV include current monitor data, beam loss monitor data, and video digitization using VME architecture
An ultrasound system for simultaneous ultrasound hyperthermia and photon beam irradiation
International Nuclear Information System (INIS)
Straube, William L.; Moros, Eduardo G.; Low, Daniel A.; Klein, Eric E.; Willcut, Virgil M.; Myerson, Robert J.
1996-01-01
Purpose: An existing ultrasound system has been adapted for simultaneous use with external photon beam irradiation. The system is being used to investigate the potential for increased biological benefit of simultaneously combined hyperthermia and external beam irradiation with currently achievable temperature distributions. Methods and Materials: An existing clinical ultrasound system has been modified for simultaneous operation with a 60 Co teletherapy machine. The generator, thermometry system, computer, and applicators are located inside the treatment room, while the monitor and system control are located at the control console. Two approaches have been used clinically to combine the two modalities. In the first approach, an en-face setup is used in which the ultrasound beam and the photon beam travel through the same window of entry to the tumor. This is achieved by a reflecting system designed to deflect the ultrasound to the tumor while positioning the ultrasound transducer outside the radiation beam. The reflecting system consists of water and water-equivalent materials except for a 1 mm sheet of polished brass that is used as the reflector. The relative pressure fields were measured in water at the same distance from the ultrasound source using a scanning hydrophone with and without the reflector at the two operating frequencies of the device (1.0 and 3.4 MHz) for two applicators. Radiation dosimetry measurements were performed to determine the relationship between 60 Co irradiation through the reflector and absorbed dose. In the second approach the ultrasound and the radiation beam travel into the tumor from different windows of entry such that the radiation beam passes through no portion of the water bolus prior to entering the patient. We have termed this approach the orthogonal approach. For both approaches, the radiation fraction is given in the middle of an uninterrupted 60-min hyperthermia treatment. Results: The system modifications did not impair
Measuring linac photon beam energy through EPID image analysis of physically wedged fields
Energy Technology Data Exchange (ETDEWEB)
Dawoud, S. M., E-mail: samir.dawoud@leedsth.nhs.uk; Weston, S. J.; Bond, I.; Ward, G. C.; Rixham, P. A.; Mason, J.; Huckle, A. [Department of Medical Physics and Engineering, St. James Institute of Oncology, St. James University Hospital, Leeds LS9 7TF (United Kingdom); Sykes, J. R. [Institute of Medical Physics, School of Physics, The University of Sydney, New South Wales 2006, Australia and Department of Medical Physics and Engineering, St. James Institute of Oncology, St. James University Hospital, Leeds LS9 7TF (United Kingdom)
2014-02-15
Purpose: Electronic portal imaging devices (EPIDs) have proven to be useful tools for measuring several parameters of interest in linac quality assurance (QA). However, a method for measuring linac photon beam energy using EPIDs has not previously been reported. In this report, such a method is devised and tested, based on fitting a second order polynomial to the profiles of physically wedged beams, where the metric of interest is the second order coefficientα. The relationship between α and the beam quality index [percentage depth dose at 10 cm depth (PDD{sub 10})] is examined to produce a suitable calibration curve between these two parameters. Methods: Measurements were taken in a water-tank for beams with a range of energies representative of the local QA tolerances about the nominal value 6 MV. In each case, the beam quality was found in terms of PDD{sub 10} for 100 × 100 mm{sup 2} square fields. EPID images of 200 × 200 mm{sup 2} wedged fields were then taken for each beam and the wedge profile was fitted in MATLAB 2010b (The MathWorks, Inc., Natick, MA). α was then plotted against PDD{sub 10} and fitted with a linear relation to produce the calibration curve. The uncertainty in α was evaluated by taking five repeat EPID images of the wedged field for a beam of 6 MV nominal energy. The consistency of measuring α was found by taking repeat measurements on a single linac over a three month period. The method was also tested at 10 MV by repeating the water-tank crosscalibration for a range of energies centered approximately about a 10 MV nominal value. Finally, the calibration curve from the test linac and that from a separate clinical machine were compared to test consistency of the method across machines in a matched fleet. Results: The relationship betweenα and PDD{sub 10} was found to be strongly linear (R{sup 2} = 0.979) while the uncertainty in α was found to be negligible compared to that associated with measuring PDD{sub 10} in the water-tank (
Radical distributions in ammonium tartrate single crystals exposed to photon and neutron beams
International Nuclear Information System (INIS)
Marrale, M.; Longo, A.; Brai, M.; Barbon, A.; Brustolon, M.
2014-01-01
The radiation therapy carried out by means of heavy charged particles (such as carbon ions) and neutrons is rapidly becoming widespread worldwide. The success of these radiation therapies relies on the high density of energy released by these particles or by secondary particles produced after primary interaction with matter. The biological damages produced by ionising radiations in tissues and cells depend more properly on the energy released per unit pathlength, which is the linear energy transfer and which determines the radiation quality. To improve the therapy effectiveness, it is necessary to grasp the mechanisms of free radical production and distribution after irradiation with these particles when compared with the photon beams. In this work some preliminary results on the analysis of the spatial distributions of the free radicals produced after exposure of ammonium tartrate crystals to various radiation beams ( 60 Co gamma photons and thermal neutrons) were reported. Electron spin resonance analyses were performed by the electron spin echo technique, which allows the determination of local spin concentrations and by double electron-electron resonance technique, which is able to measure the spatial distance distribution (range 1.5-8 nm) among pairs of radicals in solids. The results of these analyses are discussed on the basis of the different distributions of free radicals produced by the two different radiation beams used. This paper extends to the single crystal case, a similar work done on AT powder irradiated with different beams, with assessment of microscopic radical concentration by determining the amount of ID contribution and obtaining the inter-radical distance distributions by double microwave irradiation. In this paper single crystals of AT have been exposed to 60 Co photons and neutrons. The results confirm that advanced pulse EPR techniques allow the direct measurement of the local free radical concentration and provide information about the
Investigation of active structural intensity control in finite beams: theory and experiment
Audrain; Masson; Berry
2000-08-01
An investigation of structural intensity control is presented in this paper. As opposed to previous work, the instantaneous intensity is completely taken into account in the control algorithm, i.e., all the terms are considered in the real-time control process and, in particular, the evanescent waves are considered in this approach. A finite difference approach using five accelerometers is used as the sensing scheme. A feedforward filtered-X least mean square algorithm is adapted to this energy-based control problem, involving a nonpositive definite quadratic form in general. In this respect, the approach is limited to cases where the geometry is such that the intensity component will have the same sign for the control source and the primary disturbance. Results from numerical simulations are first presented to illustrate the benefit of using a cost function based on structural intensity. Experimental validation of the approach is conducted on a free-free beam covered with viscoelastic material. A comparison is made between classical acceleration control and structural intensity control and the performance of both approaches is presented. These results confirm that using intensity control allows the error sensors to be placed closer to the control source and the primary disturbance, while preserving a good control performance.
Non-material finite element modelling of large vibrations of axially moving strings and beams
Vetyukov, Yury
2018-02-01
We present a new mathematical model for the dynamics of a beam or a string, which moves in a given axial direction across a particular domain. Large in-plane vibrations are coupled with the gross axial motion, and a Lagrangian (material) form of the equations of structural mechanics becomes inefficient. The proposed mixed Eulerian-Lagrangian description features mechanical fields as functions of a spatial coordinate in the axial direction. The material travels across a finite element mesh, and the boundary conditions are applied in fixed nodes. Beginning with the variational equation of virtual work in its material form, we analytically derive the Lagrange's equations of motion of the second kind for the considered case of a discretized non-material control domain and for geometrically exact kinematics. The dynamic analysis is straightforward as soon as the strain and the kinetic energies of the control domain are available. In numerical simulations we demonstrate the rapid mesh convergence of the model, the effect of the bending stiffness and the dynamic instability when the axial velocity gets high. We also show correspondence to the results of fully Lagrangian benchmark solutions.
Rotational Stiffness of Precast Beam-Column Connection using Finite Element Method
Hashim, N.; Agarwal, J.
2018-04-01
Current design practice in structural analysis is to assume the connection as pinned or rigid, however this cannot be relied upon for safety against collapse because during services the actual connection reacts differently where the connection has rotated in relevance. This situation may lead to different reactions and consequently affect design results and other frame responses. In precast concrete structures, connections play an important part in ensuring the safety of the whole structure. Thus, investigates on the actual connection behavior by construct the moment-rotation relationship is significant. Finite element (FE) method is chosen for modeling a 3-dimensional beam-column connection. The model is built in symmetry to reduce analysis time. Results demonstrate that precast billet connection is categorized as semi-rigid connection with Sini of 23,138kNm/rad. This is definitely different from the assumption of pinned or rigid connection used in design practice. Validation were made by comparing with mathematical equation and small differences were achieved that led to the conclusion where precast billet connection using FE method is acceptable.
Energy Technology Data Exchange (ETDEWEB)
Casar, Bozidar [Institute of Oncology, Ljubljana (Slovenia). Dept. of Radiation Physics; Pasler, Marlies [Lake Constance Radiation Oncology Center, Singen and Friedrichshafen (Germany); Wegener, Sonja [Wuerzburg Univ. (Germany). Dept. of Radiation Oncology; and others
2017-10-01
The influence of the Integral Quality Monitor (IQM) transmission detector on photon beam properties was evaluated in a preclinical phase, using data from nine participating centres: (i) the change of beam quality (beam hardening), (ii) the influence on surface dose, and (iii) the attenuation of the IQM detector. For 6 different nominal photon energies (4 standard, 2 FFF) and square field sizes from 1 x 1 cm{sup 2} to 20 x 20 cm{sup 2}, the effect of IQM on beam quality was assessed from the PDD{sub 20,10} values obtained from the percentage dose depth (PDD) curves, measured with and without IQM in the beam path. The change in surface dose with/without IQM was assessed for all available energies and field sizes from 4 x 4 cm{sup 2} to 20 x 20 cm{sup 2}. The transmission factor was calculated by means of measured absorbed dose at 10 cm depth for all available energies and field sizes. (i) A small (0.11-0.53%) yet statistically significant beam hardening effect was observed, depending on photon beam energy. (ii) The increase in surface dose correlated with field size (p < 0.01) for all photon energies except for 18 MV. The change in surface dose was smaller than 3.3% in all cases except for the 20 x 20 cm{sup 2} field and 10 MV FFF beam, where it reached 8.1%. (iii) For standard beams, transmission of the IQM showed a weak dependence on the field size, and a pronounced dependence on the beam energy (0.9412 for 6 MV to 0.9578 for 18 MV and 0.9440 for 6 MV FFF; 0.9533 for 10 MV FFF). The effects of the IQM detector on photon beam properties were found to be small yet statistically significant. The magnitudes of changes which were found justify treating IQM either as tray factors within the treatment planning system (TPS) for a particular energy or alternatively as modified outputs for specific beam energy of linear accelerators, which eases the introduction of the IQM into clinical practice.
Dosimetry quality audit of high energy photon beams in greek radiotherapy centers
International Nuclear Information System (INIS)
Hourdakis, Constantine J.; Boziari, A.
2008-01-01
Background and purpose: Dosimetry quality audits and intercomparisons in radiotherapy centers is a useful tool in order to enhance the confidence for an accurate therapy and to explore and dissolve discrepancies in dose delivery. This is the first national comprehensive study that has been carried out in Greece. During 2002 - 2006 the Greek Atomic Energy Commission performed a dosimetry quality audit of high energy external photon beams in all (23) Greek radiotherapy centers, where 31 linacs and 13 Co-60 teletherapy units were assessed in terms of their mechanical performance characteristics and relative and absolute dosimetry. Materials and Methods: The quality audit in dosimetry of external photon beams took place by means of on-site visits, where certain parameters of the photon beams were measured, calculated and assessed according to a specific protocol and the IAEA TRS 398 dosimetry code of practice. In each radiotherapy unit (Linac or Co-60), certain functional parameters were measured and the results were compared to tolerance values and limits. Doses in water under reference and non reference conditions were measured and compared to the stated values. Also, the treatment planning systems (TPS) were evaluated with respect to irradiation time calculations. Results: The results of the mechanical tests, dosimetry measurements and TPS evaluation have been presented in this work and discussed in detail. This study showed that Co-60 units had worse performance mechanical characteristics than linacs. 28% of all irradiation units (23% of linacs and 42% of Co-60 units) exceeded the acceptance limit at least in one mechanical parameter. Dosimetry accuracy was much worse in Co60 units than in linacs. 61% of the Co60 units exhibited deviations outside ±3% and 31% outside ±5%. The relevant percentages for the linacs were 24% and 7% respectively. The results were grouped for each hospital and the sources of errors (functional and human) have been investigated and
Dosimetry quality audit of high energy photon beams in greek radiotherapy centers.
Hourdakis, Constantine J; Boziari, A
2008-04-01
Dosimetry quality audits and intercomparisons in radiotherapy centers is a useful tool in order to enhance the confidence for an accurate therapy and to explore and dissolve discrepancies in dose delivery. This is the first national comprehensive study that has been carried out in Greece. During 2002--2006 the Greek Atomic Energy Commission performed a dosimetry quality audit of high energy external photon beams in all (23) Greek radiotherapy centers, where 31 linacs and 13 Co-60 teletherapy units were assessed in terms of their mechanical performance characteristics and relative and absolute dosimetry. The quality audit in dosimetry of external photon beams took place by means of on-site visits, where certain parameters of the photon beams were measured, calculated and assessed according to a specific protocol and the IAEA TRS 398 dosimetry code of practice. In each radiotherapy unit (Linac or Co-60), certain functional parameters were measured and the results were compared to tolerance values and limits. Doses in water under reference and non reference conditions were measured and compared to the stated values. Also, the treatment planning systems (TPS) were evaluated with respect to irradiation time calculations. The results of the mechanical tests, dosimetry measurements and TPS evaluation have been presented in this work and discussed in detail. This study showed that Co-60 units had worse performance mechanical characteristics than linacs. 28% of all irradiation units (23% of linacs and 42% of Co-60 units) exceeded the acceptance limit at least in one mechanical parameter. Dosimetry accuracy was much worse in Co60 units than in linacs. 61% of the Co60 units exhibited deviations outside +/-3% and 31% outside +/-5%. The relevant percentages for the linacs were 24% and 7% respectively. The results were grouped for each hospital and the sources of errors (functional and human) have been investigated and discussed in details. This quality audit proved to be a
Hodges, Robert V.; Nixon, Mark W.; Rehfield, Lawrence W.
1987-01-01
A methodology was developed for the structural analysis of composite rotor blades. This coupled-beam analysis is relatively simple to use compared with alternative analysis techniques. The beam analysis was developed for thin-wall single-cell rotor structures and includes the effects of elastic coupling. This paper demonstrates the effectiveness of the new composite-beam analysis method through comparison of its results with those of an established baseline analysis technique. The baseline analysis is an MSC/NASTRAN finite-element model built up from anisotropic shell elements. Deformations are compared for three linear static load cases of centrifugal force at design rotor speed, applied torque, and lift for an ideal rotor in hover. A D-spar designed to twist under axial loading is the subject of the analysis. Results indicate the coupled-beam analysis is well within engineering accuracy.
International Nuclear Information System (INIS)
Stucki, G.; Muench, W.; Quintel, H.
2002-01-01
Full text: The Swiss Federal Office of Metrology and Accreditation (METAS) provides an absorbed dose to water calibration service for reference dosimeters using 60 Co γ radiation, ten X-ray beam qualities between TPR 20,10 =0.639 and 0.802 and ten electron beam qualities between R 50 =1.75 gcm -2 and 8.54 gcm -2 . A 22 MeV microtron accelerator with a conventional treatment head is used as radiation source for the high energy photon and electron beams. The treatment head produces clinical beams. The METAS absorbed dose calibration service for high energy photons is based on a primary standard sealed water calorimeter of the Domen type, that is used to calibrate several METAS transfer standards of type NE2611A and NE2571A in terms of absorbed dose to water in the energy range from 60 Co to TPR 20,10 = 0.802. User reference dosimeters are compared with the transfer standards to give calibration factors in absorbed dose to water with an uncertainty of 1.0% for 60 Co γ radiation and 1.4% for higher energies (coverage factor k=2). The calibration service was launched in 1997. The calibration factors measured by METAS have been compared with those derived from the Code of Practice of the International Atomic Energy Agency using the calculated k Q factors listed in table 14. The comparison showed a maximum difference of 0.8% for the NE25611A and NE 2571A chambers. At 60 Co γ radiation the METAS primary standard of absorbed dose to water was bilaterally compared with the primary standards of the Bureau International des Poids et Mesures BIPM (Sevres) as well as of the National Research Council NRC (Canada). In either case the standards were in agreement within the comparison uncertainties. The METAS absorbed dose calibration service for high energy electron beams is based on a primary standard chemical dosimeter. A monoenergetic electron beam of precisely known particle energy and beam charge is totally absorbed in Fricke solution (ferrous ammonium sulphate) of a given
Directory of Open Access Journals (Sweden)
Kazuhisa Nakajima
2014-12-01
Full Text Available We present a new external-beam radiation therapy system using very-high-energy (VHE electron/photon beams generated by a centimeter-scale laser plasma accelerator built in a robotic system. Most types of external-beam radiation therapy are delivered using a machine called a medical linear accelerator driven by radio frequency (RF power amplifiers, producing electron beams with an energy range of 6–20 MeV, in conjunction with modern radiation therapy technologies for effective shaping of three-dimensional dose distributions and spatially accurate dose delivery with imaging verification. However, the limited penetration depth and low quality of the transverse penumbra at such electron beams delivered from the present RF linear accelerators prevent the implementation of advanced modalities in current cancer treatments. These drawbacks can be overcome if the electron energy is increased to above 50 MeV. To overcome the disadvantages of the present RF-based medical accelerators, harnessing recent advancement of laser-driven plasma accelerators capable of producing 1-GeV electron beams in a 1-cm gas cell, we propose a new embodiment of the external-beam radiation therapy robotic system delivering very high-energy electron/photon beams with an energy of 50–250 MeV; it is more compact, less expensive, and has a simpler operation and higher performance in comparison with the current radiation therapy system.
Dependence of alanine gel dosimeter response as a function of photon clinical beams dose rate
International Nuclear Information System (INIS)
Silva, Cleber Feijo; Campos, Leticia Lucente
2013-01-01
Gel dosimetry is a new area developed by Gore, it is ery useful for application in radiotherapy because using NMR imaging as evaluation technique is possible to evaluate three dimensional absorbed dose distribution. The measure technique is based on difference of ferrous (Fe 2+ ) and ferric (Fe 3+ ) ) ions concentration that can be measured also by spectrophotometry technique. The Alanine gel dosimeter was developed at IPEN. The alanine is an amino acid and tissue equivalent material that presents significant improvement on previous alanine dosimetry systems. The addition of Alanine increases the production of ferric ions in the solution. This work aims to study the dose rate dependence of photon clinical beams radiation on the alanine gel dosimeter optical response, as well as the response repeatability and gel production reproducibility, since this property is very important for characterization and standardization of any dosimeter. (author)
International Nuclear Information System (INIS)
Ay, F; Iñurrategui, I; Geskus, D; Aravazhi, S; Pollnau, M
2011-01-01
Deeply etched Bragg gratings were fabricated by focused ion beam (FIB) milling in KGd x Lu 1-x (WO 4 ) 2 :Yb 3+ to obtain photonic cavity structures. By optimizing parameters such as dose per area, dwell time and pixel resolution the redeposition effects were minimized and grating structures more than 4 μm in depth with an improved sidewall angle of ∼ 5° were achieved. Fabry-Perot microcavities were defined and used to assess the optical performance of the grating structures at ∼ 1530 nm. An on-chip integrated laser cavity at ∼ 980 nm was achieved by defining a FIB reflective grating and FIB polished waveguide end-facet. With this cavity, an on-chip integrated waveguide laser in crystalline potassium double tungstate was demonstrated
Molecular beam epitaxy grown Ge/Si pin layer sequence for photonic devices
International Nuclear Information System (INIS)
Schulze, J.; Oehme, M.; Werner, J.
2012-01-01
A key challenge to obtain a convergence of classical Si-based microelectronics and optoelectronics is the manufacturing of photonic integrated circuits integrable into classical Si-based integrated circuits. This integration would be greatly enhanced if similar facilities and technologies could be used. Therefore one approach is the development of optoelectronic components and devices made from group-IV-based materials such as SiGe, Ge or Ge:Sn. In this paper the optoelectronic performances of a pin diode made from a Ge/Si heterostructure pin layer sequence grown by molecular beam epitaxy are discussed. After a detailed description of the layer sequence growth and the device manufacturing process it will be shown that – depending on the chosen operating point and device design – the diode serves as a broadband high speed photo detector, Franz–Keldysh effect modulator or light emitting diode.
Molecular beam epitaxy grown Ge/Si pin layer sequence for photonic devices
Energy Technology Data Exchange (ETDEWEB)
Schulze, J., E-mail: schulze@iht.uni-stuttgart.de; Oehme, M.; Werner, J.
2012-02-01
A key challenge to obtain a convergence of classical Si-based microelectronics and optoelectronics is the manufacturing of photonic integrated circuits integrable into classical Si-based integrated circuits. This integration would be greatly enhanced if similar facilities and technologies could be used. Therefore one approach is the development of optoelectronic components and devices made from group-IV-based materials such as SiGe, Ge or Ge:Sn. In this paper the optoelectronic performances of a pin diode made from a Ge/Si heterostructure pin layer sequence grown by molecular beam epitaxy are discussed. After a detailed description of the layer sequence growth and the device manufacturing process it will be shown that - depending on the chosen operating point and device design - the diode serves as a broadband high speed photo detector, Franz-Keldysh effect modulator or light emitting diode.
Determination Of Output Factor For Photon Beam Of The Mitsubishi EXL-14 Linear Accelerator
International Nuclear Information System (INIS)
Nurman R; Sri-Inang S; Dani
2003-01-01
This paper describes the determination of output factor for 6 MV photon beam of The Mitsubishi EXL-14 linear accelerator teletherapy unit. Determination of percentage depth dose curve has been done using Wallhofer dosemeter at source to surface distance, SSD of 100 cm and field size of 10 cm x 10 cm. Measurement of output has been carried out using a 0.6 cc ionization chamber inside a water phantom at depth of 5 cm with source to surface distance, SSD of 100 cm for square fields ranging in size 4 cm x 4 cm up to 10 cm x 10 cm. Output for rectangular fields which equal to the equivalent square fields were also determined. The results obtained shows that the deviations of the output for 12 cm x 3 cm and 19 cm x 7 cm were higher than ±2% to the output of the equivalent square fields. (author)
First doubly polarised photoproduction on {sup 3}He at the photon beam of MAMI
Energy Technology Data Exchange (ETDEWEB)
Aguar Bartolome, Patricia
2010-11-15
A first experiment with a polarised {sup 3}He target was carried out in July 2009 at the MAMI accelerator in Mainz in a photon energy range between 200 MeV and 800 MeV. The aim of this measurement was to investigate the Gerasimov-Drell-Hearn sum rule on the neutron. The use of the data obtained with the polarised {sup 3}He target, compared to existing data on the deuteron, gives a complementary and more direct access to the neutron, due to the spin structure of the {sup 3}He. The measurement of the helicity dependence of the inclusive total photoabsorption cross section required a beam of tagged circularly polarised photons incident on the longitudinally polarised {sup 3}He target. The data were taken using the 4{pi} Crystal Ball photon spectrometer in combination with TAPS as a forward wall and complemented by a threshold Cherenkov detector used to on-line suppress the background from electromagnetic events. The development and preparation of the different components of the {sup 3}He experimental setup was an important part of this work and are described in detail in this thesis. The detector system and the analysis method were tested by the measurement of the unpolarised total inclusive photoabsorption cross section on liquid hydrogen. The results obtained are in good agreement with previous published data. Preliminary results of the unpolarised total photoabsorption cross section, as well as the helicity dependent photoabsorption cross section difference on {sup 3}He compared with several theoretical models will also be presented. (orig.)
EPR dosimetry of radiotherapy photon beams in inhomogeneous media using alanine films
Energy Technology Data Exchange (ETDEWEB)
Oesteraas, Bjoern Helge [Department of Radiation Biology, Institute for Cancer Research, Norwegian Radium Hospital, Montebello, N-0310 Oslo (Norway); Hole, Eli Olaug [Department of Physics, University of Oslo, PO Box 1048 Blindern, N-0316 Oslo (Norway); Olsen, Dag Rune [Department of Radiation Biology, Institute for Cancer Research, The Norwegian Radium Hospital, Montebello, N-0310 Oslo (Norway); Malinen, Eirik [Department of Radiation Biology, Institute for Cancer Research, The Norwegian Radium Hospital, Montebello, N-0310 Oslo (Norway)
2006-12-21
In the current work, EPR (electron paramagnetic resonance) dosimetry using alanine films (134 {mu}m thick) was utilized for dose measurements in inhomogeneous phantoms irradiated with radiotherapy photon beams. The main phantom material was PMMA, while either Styrofoam or aluminium was introduced as an inhomogeneity. The phantoms were irradiated to a maximum dose of about 30 Gy with 6 or 15 MV photons. The performance of the alanine film dosimeters was investigated and compared to results from ion chamber dosimetry, Monte Carlo simulations and radiotherapy treatment planning calculations. It was found that the alanine film dosimeters had a linear dose response above approximately 5 Gy, while a background signal obscured the response at lower dose levels. For doses between 5 and 60 Gy, the standard deviation of single alanine film dose estimates was about 2%. The alanine film dose estimates yielded results comparable to those from the Monte Carlo simulations and the ion chamber measurements, with absolute differences between estimates in the order of 1-15%. The treatment planning calculations exhibited limited applicability. The current work shows that alanine film dosimetry is a method suitable for estimating radiotherapeutical doses and for dose measurements in inhomogeneous media.
EPR dosimetry of radiotherapy photon beams in inhomogeneous media using alanine films
International Nuclear Information System (INIS)
Oesteraas, Bjoern Helge; Hole, Eli Olaug; Olsen, Dag Rune; Malinen, Eirik
2006-01-01
In the current work, EPR (electron paramagnetic resonance) dosimetry using alanine films (134 μm thick) was utilized for dose measurements in inhomogeneous phantoms irradiated with radiotherapy photon beams. The main phantom material was PMMA, while either Styrofoam or aluminium was introduced as an inhomogeneity. The phantoms were irradiated to a maximum dose of about 30 Gy with 6 or 15 MV photons. The performance of the alanine film dosimeters was investigated and compared to results from ion chamber dosimetry, Monte Carlo simulations and radiotherapy treatment planning calculations. It was found that the alanine film dosimeters had a linear dose response above approximately 5 Gy, while a background signal obscured the response at lower dose levels. For doses between 5 and 60 Gy, the standard deviation of single alanine film dose estimates was about 2%. The alanine film dose estimates yielded results comparable to those from the Monte Carlo simulations and the ion chamber measurements, with absolute differences between estimates in the order of 1-15%. The treatment planning calculations exhibited limited applicability. The current work shows that alanine film dosimetry is a method suitable for estimating radiotherapeutical doses and for dose measurements in inhomogeneous media
Delivery of modulated electron beams with conventional photon multi-leaf collimators
International Nuclear Information System (INIS)
Klein, Eric E; Mamalui-Hunter, Maria; Low, Daniel A
2009-01-01
Electron beam radiotherapy is an accepted method to treat shallow tumors. However, modulation of electrons to customize dose distributions has not readily been achieved. Studies of bolus and tertiary collimation systems have been met with limitations. We pursue the use of photon multi-leaf collimators (MLC) for modulated electron radiotherapy (MERT) to achieve customized distributions for potential clinical use. As commercial planning systems do not support the use of MLC with electrons, planning was conducted using Monte Carlo calculations. Segmented and dynamic modulated delivery of multiple electron segments was configured, calculated and delivered for validation. Delivery of electrons with segmented or dynamic leaf motion was conducted. A phantom possessing an idealized stepped target was planned and optimized with subsequent validation by measurements. Finally, clinical treatment plans were conducted for post-mastectomy and cutaneous lymphoma of the scalp using forward optimization techniques. Comparison of calculations and measurements was successful with agreement of ±2%/2 mm for the energies, segment sizes, depths tested for delivered segments for the dynamic and segmented delivery. Clinical treatment plans performed provided optimal dose coverage of the target while sparing distal organs at risk. Execution of plans using an anthropomorphic phantom to ensure safe and efficient delivery was conducted. Our study validates that MERT is not only possible using the photon MLC, but the efficient and safe delivery inherent with the dynamic delivery provides an ideal technique for shallow tumor treatment.
Modelling lateral beam quality variations in pencil kernel based photon dose calculations
International Nuclear Information System (INIS)
Nyholm, T; Olofsson, J; Ahnesjoe, A; Karlsson, M
2006-01-01
Standard treatment machines for external radiotherapy are designed to yield flat dose distributions at a representative treatment depth. The common method to reach this goal is to use a flattening filter to decrease the fluence in the centre of the beam. A side effect of this filtering is that the average energy of the beam is generally lower at a distance from the central axis, a phenomenon commonly referred to as off-axis softening. The off-axis softening results in a relative change in beam quality that is almost independent of machine brand and model. Central axis dose calculations using pencil beam kernels show no drastic loss in accuracy when the off-axis beam quality variations are neglected. However, for dose calculated at off-axis positions the effect should be considered, otherwise errors of several per cent can be introduced. This work proposes a method to explicitly include the effect of off-axis softening in pencil kernel based photon dose calculations for arbitrary positions in a radiation field. Variations of pencil kernel values are modelled through a generic relation between half value layer (HVL) thickness and off-axis position for standard treatment machines. The pencil kernel integration for dose calculation is performed through sampling of energy fluence and beam quality in sectors of concentric circles around the calculation point. The method is fully based on generic data and therefore does not require any specific measurements for characterization of the off-axis softening effect, provided that the machine performance is in agreement with the assumed HVL variations. The model is verified versus profile measurements at different depths and through a model self-consistency check, using the dose calculation model to estimate HVL values at off-axis positions. A comparison between calculated and measured profiles at different depths showed a maximum relative error of 4% without explicit modelling of off-axis softening. The maximum relative error
International Nuclear Information System (INIS)
Agababyan, K.Sh.; Adamyan, F.V.; Ajrapetyan, A.V.
1985-01-01
The experimental method for measuring the asymmetry of the γd → pn photodisintegration reaction on the linearly polarized photon beam of the Erevan synchrotron is described. The results of Monte Carlo calculations, the calibration of apparatus, the procedure of measurements and experimental data processing are repored
Energy Technology Data Exchange (ETDEWEB)
Gibbons, John P., E-mail: john.gibbons@marybird.com [Department of Physics, Mary Bird Perkins Cancer Center, Baton Rouge, Louisiana 70809 (United States); Antolak, John A. [Department of Radiation Oncology, Mayo Clinic, Rochester, Minnesota 55905 (United States); Followill, David S. [Department of Radiation Physics, UT M.D. Anderson Cancer Center, Houston, Texas 77030 (United States); Huq, M. Saiful [Department of Radiation Oncology, University of Pittsburgh Cancer Institute, Pittsburgh, Pennsylvania 15232 (United States); Klein, Eric E. [Department of Radiation Oncology, Washington University School of Medicine, St. Louis, Missouri 63110 (United States); Lam, Kwok L. [Department of Radiation Oncology, University of Michigan, Ann Arbor, Michigan 48109 (United States); Palta, Jatinder R. [Department of Radiation Oncology, Virginia Commonwealth University, Richmond, Virginia 23298 (United States); Roback, Donald M. [Department of Radiation Oncology, Cancer Centers of North Carolina, Raleigh, North Carolina 27607 (United States); Reid, Mark [Department of Medical Physics, Fletcher-Allen Health Care, Burlington, Vermont 05401 (United States); Khan, Faiz M. [Department of Radiation Oncology, University of Minnesota, Minneapolis, Minnesota 55455 (United States)
2014-03-15
A protocol is presented for the calculation of monitor units (MU) for photon and electron beams, delivered with and without beam modifiers, for constant source-surface distance (SSD) and source-axis distance (SAD) setups. This protocol was written by Task Group 71 of the Therapy Physics Committee of the American Association of Physicists in Medicine (AAPM) and has been formally approved by the AAPM for clinical use. The protocol defines the nomenclature for the dosimetric quantities used in these calculations, along with instructions for their determination and measurement. Calculations are made using the dose per MU under normalization conditions, D{sub 0}{sup ′}, that is determined for each user's photon and electron beams. For electron beams, the depth of normalization is taken to be the depth of maximum dose along the central axis for the same field incident on a water phantom at the same SSD, where D{sub 0}{sup ′} = 1 cGy/MU. For photon beams, this task group recommends that a normalization depth of 10 cm be selected, where an energy-dependent D{sub 0}{sup ′} ≤ 1 cGy/MU is required. This recommendation differs from the more common approach of a normalization depth of d{sub m}, with D{sub 0}{sup ′} = 1 cGy/MU, although both systems are acceptable within the current protocol. For photon beams, the formalism includes the use of blocked fields, physical or dynamic wedges, and (static) multileaf collimation. No formalism is provided for intensity modulated radiation therapy calculations, although some general considerations and a review of current calculation techniques are included. For electron beams, the formalism provides for calculations at the standard and extended SSDs using either an effective SSD or an air-gap correction factor. Example tables and problems are included to illustrate the basic concepts within the presented formalism.
Casar, Bozidar; Pasler, Marlies; Wegener, Sonja; Hoffman, David; Talamonti, Cinzia; Qian, Jianguo; Mendez, Ignasi; Brojan, Denis; Perrin, Bruce; Kusters, Martijn; Canters, Richard; Pallotta, Stefania; Peterlin, Primoz
2017-09-01
The influence of the Integral Quality Monitor (IQM) transmission detector on photon beam properties was evaluated in a preclinical phase, using data from nine participating centres: (i) the change of beam quality (beam hardening), (ii) the influence on surface dose, and (iii) the attenuation of the IQM detector. For 6 different nominal photon energies (4 standard, 2 FFF) and square field sizes from 1×1cm 2 to 20×20cm 2 , the effect of IQM on beam quality was assessed from the PDD 20,10 values obtained from the percentage dose depth (PDD) curves, measured with and without IQM in the beam path. The change in surface dose with/without IQM was assessed for all available energies and field sizes from 4×4cm 2 to 20×20cm 2 . The transmission factor was calculated by means of measured absorbed dose at 10cm depth for all available energies and field sizes. (i) A small (0.11-0.53%) yet statistically significant beam hardening effect was observed, depending on photon beam energy. (ii) The increase in surface dose correlated with field size (pphoton energies except for 18MV. The change in surface dose was smaller than 3.3% in all cases except for the 20×20cm 2 field and 10MV FFF beam, where it reached 8.1%. (iii) For standard beams, transmission of the IQM showed a weak dependence on the field size, and a pronounced dependence on the beam energy (0.9412 for 6MV to 0.9578 for 18MV and 0.9440 for 6MV FFF; 0.9533 for 10MV FFF). The effects of the IQM detector on photon beam properties were found to be small yet statistically significant. The magnitudes of changes which were found justify treating IQM either as tray factors within the treatment planning system (TPS) for a particular energy or alternatively as modified outputs for specific beam energy of linear accelerators, which eases the introduction of the IQM into clinical practice. Copyright © 2017. Published by Elsevier GmbH.
Medin, Joakim; Ross, Carl K; Klassen, Norman V; Palmans, Hugo; Grusell, Erik; Grindborg, Jan-Erik
2006-03-21
Absorbed doses determined with a sealed water calorimeter operated at 4 degrees C are compared with the results obtained using ionization chambers and the IAEA TRS-398 code of practice in a 10 MV photon beam (TPR(20,10) = 0.734) and a 175 MeV proton beam (at a depth corresponding to the residual range, R(res) = 14.7 cm). Three NE 2571 and two FC65-G ionization chambers were calibrated in terms of absorbed-dose-to-water in (60)Co at the Swedish secondary standard dosimetry laboratory, directly traceable to the BIPM. In the photon beam quality, calorimetry was found to agree with ionometry within 0.3%, confirming the k(Q) values tabulated in TRS-398. In contrast, a 1.8% deviation was found in the proton beam at 6 g cm(-2) depth, suggesting that the TRS-398 tabulated k(Q) values for these two ionization chamber types are too high. Assuming no perturbation effect in the proton beam for the ionization chambers, a value for (w(air)/e)(Q) of 33.6 J C(-1) +/- 1.7% (k = 1) can be derived from these measurements. An analytical evaluation of the effect from non-elastic nuclear interactions in the ionization chamber wall indicates a perturbation effect of 0.6%. Including this estimated result in the proton beam would increase the determined (w(air)/e)(Q) value by the same amount.
International Nuclear Information System (INIS)
Ceberg, Crister P.; Bjaerngard, Bengt E.
1995-01-01
An approximate experimental determination of photon pencil-beams can be based on the reciprocity theorem. The scatter part of the pencil-beam is then essentially the derivative with respect to the field radius of measured scatter-to-primary ratios in circular fields. Obtained in this way, however, the pencil-beam implicitly carries the influence from the lateral fluence and beam quality variations of the incident photons, as well as the effects of the divergence of the beam. In this work we show how these effects can be corrected for. The procedure was to calculate scatter-to-primary ratios using an analytical expression for the pencil-beam. By disregarding one by one the effects of the divergence and the fluence and beam quality variations, the influence of these effects were separated and quantified. For instance, for a 6 MV beam of 20x20 cm 2 field size, at 20 cm depth and a source distance of 100 cm, the total effect was 3.9%; 2.0% was due to the non-uniform incident profile, 1.0% due to the non-uniform beam quality, and 0.9% due to the divergence of the beam. At a source distance of 400 cm, all these effects were much lower, adding up to a total of 0.3 %. Using calculated correction factors like these, measured scatter-to-primary ratios were then stripped from the effects of non-uniform and non-parallel measurement conditions, and the scatter part of the pencil-beam was determined using the reciprocity theorem without approximations
Glaser, Adam K; Andreozzi, Jacqueline M; Zhang, Rongxiao; Pogue, Brian W; Gladstone, David J
2015-07-01
To test the use of a three-dimensional (3D) optical cone beam computed tomography reconstruction algorithm, for estimation of the imparted 3D dose distribution from megavoltage photon beams in a water tank for quality assurance, by imaging the induced Cherenkov-excited fluorescence (CEF). An intensified charge-coupled device coupled to a standard nontelecentric camera lens was used to tomographically acquire two-dimensional (2D) projection images of CEF from a complex multileaf collimator (MLC) shaped 6 MV linear accelerator x-ray photon beam operating at a dose rate of 600 MU/min. The resulting projections were used to reconstruct the 3D CEF light distribution, a potential surrogate of imparted dose, using a Feldkamp-Davis-Kress cone beam back reconstruction algorithm. Finally, the reconstructed light distributions were compared to the expected dose values from one-dimensional diode scans, 2D film measurements, and the 3D distribution generated from the clinical Varian ECLIPSE treatment planning system using a gamma index analysis. A Monte Carlo derived correction was applied to the Cherenkov reconstructions to account for beam hardening artifacts. 3D light volumes were successfully reconstructed over a 400 × 400 × 350 mm(3) volume at a resolution of 1 mm. The Cherenkov reconstructions showed agreement with all comparative methods and were also able to recover both inter- and intra-MLC leaf leakage. Based upon a 3%/3 mm criterion, the experimental Cherenkov light measurements showed an 83%-99% pass fraction depending on the chosen threshold dose. The results from this study demonstrate the use of optical cone beam computed tomography using CEF for the profiling of the imparted dose distribution from large area megavoltage photon beams in water.
Dosimetry study comparing NCS report-2 versus IAEA TRS-398 protocol for high energy photon beams
International Nuclear Information System (INIS)
Attalaa, E.M.; Khaled, N.E.; Abou Elenein, H.S.; Elsayed, A.A.
2005-01-01
In this work a dosimetry study is presented in which the results of absorbed dose determined at reference condition according to the IAEA TRS-398 protocol and the NCS report-2 are compared. The IAEA TRS-398 protocol for absorbed dose calibration is based on ionization chamber having absorbed dose to water calibration factor N d w, while the NCS-2 dosimetry report for absorbed dose calibration is based on an ionization chamber having air- kerma calibration factor N k . This study shows that the absorbed dose which is calculated with The IAEA TRS-398 formalisms is higher than that calculated with NCS report-2 formalisms within range from 0.4 to 0.9% in cobalt-60 beam as sensed by different ionization chambers, and from 0.2 to 1.1% for different higher energy photon beams of 6, 8 and 18 MV. The chambers used are PTW 30001, 30004, and NE-2571; which have calibration factors N k and N d w traceable to the BIPM (Bureau International des Poids et Mesures)
International Nuclear Information System (INIS)
Lumpkin, A.H.
1993-01-01
The Advanced Photon Source (APS) will be a third-generation synchrotron radiation (SR) user facility in the hard x-ray regime (10--100 keV). The design objectives for the 7-GeV storage ring include a positron beam natural emittance of 8 x 10 -9 m-rad at an average current of 100 mA. Proposed methods for measuring the transverse and longitudinal profiles will be described. Additionally, a research and development effort using an rf gun as a low-emittance source of electrons for injection into the 200- to 650-MeV linac subsystem is underway. This latter system is projected to produce electron beams with a normalized, rms emittance of ∼2 π mm-mrad at peak currents of near one hundred amps. This interesting characterization problem will also be briefly discussed. The combination of both source types within one laboratory facility will stimulate the development of diagnostic techniques in these parameter spaces
Radiation dose response of N channel MOSFET submitted to filtered X-ray photon beam
Gonçalves Filho, Luiz C.; Monte, David S.; Barros, Fabio R.; Santos, Luiz A. P.
2018-01-01
MOSFET can operate as a radiation detector mainly in high-energy photon beams, which are normally used in cancer treatments. In general, such an electronic device can work as a dosimeter from threshold voltage shift measurements. The purpose of this article is to show a new way for measuring the dose-response of MOSFETs when they are under X-ray beams generated from 100kV potential range, which is normally used in diagnostic radiology. Basically, the method consists of measuring the MOSFET drain current as a function of the radiation dose. For this the type of device, it has to be biased with a high value resistor aiming to see a substantial change in the drain current after it has been irradiated with an amount of radiation dose. Two types of N channel device were used in the experiment: a signal transistor and a power transistor. The delivered dose to the device was varied and the electrical curves were plotted. Also, a sensitivity analysis of the power MOSFET response was made, by varying the tube potential of about 20%. The results show that both types of devices have responses very similar, the shift in the electrical curve is proportional to the radiation dose. Unlike the power MOSFET, the signal transistor does not provide a linear function between the dose rate and its drain current. We also have observed that the variation in the tube potential of the X-ray equipment produces a very similar dose-response.
Analysis of ionic mobilities in liquid isooctane with low dose radiotherapy pulsed photon beams
International Nuclear Information System (INIS)
Pardo-Montero, J; Gago-Arias, A; González-Castaño, D M; Gómez, F; Tegami, S; Holzscheiter, M H
2012-01-01
In this work we present a model of signal temporal development in ionization chambers and we use it to determine ionic mobilities and relative densities of charge carriers in non-ultrapure liquid isooctane using a liquid-filled ionization chamber dosimeter. The detector has been irradiated with a low dose rate, short pulsed photon beam generated with a medical LINAC. Ionic mobilities have been obtained by studying the temporal development of the readout signal and fitting it to a model for low dose rate beams where recombination is negligible. The best fit has been obtained for 3 ionic species with mobilities k 1 = (2.22±0.22) × 10 −8 , k 2 = (3.37±0.43) × 10 −8 , k 3 = (19.69±2.59) × 10 −8 m 2 V −1 s −1 and relative densities n 1 = 0.5 (n 1 is not a fitting parameter), n 2 = 0.23±0.03 and n 3 = 0.27±0.03.
Sensitivity and offset calibration for the beam position monitors at the Advanced Photon Source
International Nuclear Information System (INIS)
Chung, Y.; Barr, D.; Decker, G.; Evans, K. Jr.; Kahana, E.
1995-01-01
The beam position monitors (BPMs) play a critically important role in commissioning and operation of accelerators. Accurate determination of the offsets relative to the magnetic axis and sensitivities of individual BPMs is thus needed. We will describe in this paper the schemes for calibrating all of the 360 BPMs for sensitivity and offset in the 7-GeV Advanced Photon Source (APS) storage ring and the results. For the sensitivity calibration, a 2-dimensional map of the BPM response in the aluminum vacuum chamber is obtained theoretically, which is combined with the measured nonlinear response of the BPM electronics. A set of 2-dimensional polynomial coefficients is then obtained to approximate the result analytically. The offset calibration of the BPMs is done relative to the magnetic axis of the quadrupoles using the beam. This avoids the problem arising from various mechanical sources as well as the offset in the processing electronics. The measurement results for the resolution and long-term drift of the BPM electronics shows 0.06-μm/√Hz resolution and 2-μm/hr drift over a period of 1.5 hrs
International Nuclear Information System (INIS)
Frolov, A.M.; Maisheev, V.A.; Arakelyan, E.A.; Armaganyan, A.A.; Avakyan, R.O.; Bayatyan, G.L.; Grigoryan, N.K.; Kechechyan, A.O.; Knyazyan, S.G.; Margaryan, A.T.
1980-01-01
Tagged photons coherently emitted in a silicon crystal by the 31 GeV electron beam of intensity 4 x 10 4 ppp and beam pulse duration of up to 1.7 s have been obtained at the Serpukhov proton accelerator. The photon intensities were I approx. 10 -1 - 10 -2 γ/e - in five almost equal energy bins within the total range k = (8.2-24.2) GeV. The calculated linear polarizations were P approx. 50-20%, respectively. Narrow peaks in the radiation intensity were observed when varying the orientation of a silicon crystal which could not be explained. The method for the experimental alignment of a crystal in electron beams at the proton accelerator has been described. (orig.)
Tahmasebibirgani, Mohammad Javad; Maskani, Reza; Behrooz, Mohammad Ali; Zabihzadeh, Mansour; Shahbazian, Hojatollah; Fatahiasl, Jafar; Chegeni, Nahid
2017-04-01
In radiotherapy, megaelectron volt (MeV) electrons are employed for treatment of superficial cancers. Magnetic fields can be used for deflection and deformation of the electron flow. A magnetic field is composed of non-uniform permanent magnets. The primary electrons are not mono-energetic and completely parallel. Calculation of electron beam deflection requires using complex mathematical methods. In this study, a device was made to apply a magnetic field to an electron beam and the path of electrons was simulated in the magnetic field using finite element method. A mini-applicator equipped with two neodymium permanent magnets was designed that enables tuning the distance between magnets. This device was placed in a standard applicator of Varian 2100 CD linear accelerator. The mini-applicator was simulated in CST Studio finite element software. Deflection angle and displacement of the electron beam was calculated after passing through the magnetic field. By determining a 2 to 5cm distance between two poles, various intensities of transverse magnetic field was created. The accelerator head was turned so that the deflected electrons became vertical to the water surface. To measure the displacement of the electron beam, EBT2 GafChromic films were employed. After being exposed, the films were scanned using HP G3010 reflection scanner and their optical density was extracted using programming in MATLAB environment. Displacement of the electron beam was compared with results of simulation after applying the magnetic field. Simulation results of the magnetic field showed good agreement with measured values. Maximum deflection angle for a 12 MeV beam was 32.9° and minimum deflection for 15 MeV was 12.1°. Measurement with the film showed precision of simulation in predicting the amount of displacement in the electron beam. A magnetic mini-applicator was made and simulated using finite element method. Deflection angle and displacement of electron beam were calculated. With
Rahman, N.; Alam, M. N.
2018-02-01
Vibration response analysis of a hybrid beam with surface mounted patch piezoelectric layer is presented in this work. A one dimensional finite element (1D-FE) model based on efficient layerwise (zigzag) theory is used for the analysis. The beam element has eight mechanical and a variable number of electrical degrees of freedom. The beams are also modelled in 2D-FE (ABAQUS) using a plane stress piezoelectric quadrilateral element for piezo layers and a plane stress quadrilateral element for the elastic layers of hybrid beams. Results are presented to assess the effect of size of piezoelectric patch layer on the free and forced vibration responses of thin and moderately thick beams under clamped-free and clamped-clamped configurations. The beams are subjected to unit step loading and harmonic loading to obtain the forced vibration responses. The vibration control using in phase actuation potential on piezoelectric patches is also studied. The 1D-FE results are compared with the 2D-FE results.
A fixed angle double mirror filter for preparing a pink undulator beam at the Advanced Photon Source
International Nuclear Information System (INIS)
Dufresne, E.; Sanchez, T.; Nurushev, T.; Clarke, R.; Dierker, S.B.
2000-01-01
Recent advances in X-ray Photon Correlation Spectroscopy (XPCS) use the full bandwidth of an undulator harmonic in order to maximize the coherent flux for small angle X-ray scattering experiments. X-ray mirrors and filters are typically used to select a given harmonic of the spectrum. At the University of Michigan/Howard University/Lucent Technologies, Bell Labs, Collaborative Access Team (MHATT-CAT) undulator beamline of the Advanced Photon Source, we have designed a fixed-angle Double Mirror Filter which will provide a 'pink beam' (i.e., 2-3% bandwidth) for XPCS experiments. This device uses two small mirrors which vertically reflect a 0.1 mmx0.1 mm white beam in a symmetric geometry. The doubly reflected beam propagates parallel to the incident white beam, but is offset vertically by 35 mm. Using the standard offset of the APS allows one to stop the white beam with a standard APS beam stop. In this report, we will describe our design considerations for this instrument. We also report the results of preliminary tests of the performance. The mirrors preserve the transverse coherence of the source, and filter the undulator spectrum as expected
International Nuclear Information System (INIS)
Azadbakht, B.; Hadad, K.; Zahmatkesh, M. H.
2010-01-01
In this work, the investigation of the normoxic Pagan polymer-gel dosimeter percentage depth dose and it's dose rate dependence has been made. Using MRI, the formulation to give the maximum change in the transverse relaxation rate R2 was determined to be 4.5% N,N'-methylene-bis-acrylamide(bis), 4.5% acrylamide, 5% gelatine, 5 m M tetrakis (hydroxymethyl) phosphonium chloride, 0.01 m M hydroquinone and 86% HPLC(Water). Irradiation of vials was performed using photon beams of Co-60 therapy unit and an Elec ta linear accelerator. Gel dosimeters were imaged in a Siemens Symphony 1.5 Tesla clinical MRI scanner using a head coil. Post-manufacture irradiation and post imaging times were both selected to be 1 day. For determining the percentage depth dose of the Pagan gel it was found that at the depth of 21 cm, the percentage depth dose for 1.25 MeV γ-ray photons of 60 Co and for 4,6 and 18 MV x-ray photons of Elec ta linear accelerator, are 48%, 52%, 57.3% and 59.73%, respectively. Thus, in the case of the higher energy photon beams, a higher dose can he delivered. to deep-seated tumors. The dose rate dependence of percentage depth dose was studied for 6 MV x-ray photons with the use of dose rates of 80, 160, 240, 320, 400 and 480 c Gy/min. No trend in polymer-gel dosimeter 1/T 2 dependence was found on the mean dose rate and energy for the photon beams.
Energy Technology Data Exchange (ETDEWEB)
Pasler, Marlies; Wirtz, Holger; Lutterbach, Johannes [Lake Constance Radiation Oncology Center Singen-Friedrichshafen, Singen (Germany); Georg, Dietmar [Medical Univ. Vienna (Austria). Dept. of Radiotherapy
2011-12-15
The goal of the research was to evaluate treatment plan quality and dosimetric accuracy of volumetric modulated arc therapy (VMAT) and intensity-modulated radiotherapy (IMRT) plans using 6, 10, and 15 MV photon beams for prostate cancer including lymph nodes. In this retrospective study, VMAT and IMRT plans were generated with the Pinnacle {sup copyright} treatment planning system (TPS) (V9.0) for 10 prostate cancer cases. Each plan consisted of two target volumes: PTV{sub B} included the prostate bed, PTV{sub PC+LN} contained PTV{sub B} and lymph nodes. For plan evaluation statistics, the homogeneity index, conformity index, mean doses, and near-max doses to organs at risk (OAR) were analyzed. Treatment time and number of monitor units were assessed to compare delivery efficiency. Dosimetric plan verification was performed with a 2D ionization chamber array placed in a full scatter phantom. Results: No differences were found for target and OAR parameters in low and high energy photon beam plans for both VMAT and IMRT. A slightly higher low dose volume was detected for 6 MV VMAT plans (normal tissue: D{sub mean} = 16.47 Gy) compared to 10 and 15 MV VMAT plans (D{sub mean} = 15.90 Gy and 15.74 Gy, respectively), similar to the findings in IMRT. In VMAT, > 96% of detector points passed the 3%/ 3 mm {gamma} criterion; marginally better accuracy was found in IMRT (> 97%). Conclusion: For static and rotational IMRT, 15 MV photons did not show advantages over 6 and 10 MV high energy photon beams in large volume pelvic plans. For the investigated TPS and linac combination, 10 MV photon beams can be used as the general purpose energy for intensity modulation.
International Nuclear Information System (INIS)
Silva, Cleber F.
2011-01-01
The DL-Alanine (C 3 H 7 NO 2 ) is an amino acid tissue equivalent traditionally used as standard dosimetric material in EPR dosimetry. Recently, it has been studied to be applied in gel dosimetry, considering that the addition of Alanine in the Fricke gel solution improves the production of ferric ions radiation induced. The spectrophotometric evaluation technique can be used comparing the two spectrum wavelengths bands: 457 nm band that corresponds to ferrous ions and 588 nm band that corresponds to ferric ions concentration to evaluate the dosimetric properties of this material. The performance of the Alanine gel solution developed at IPEN has been firstly studied using the spectrophotometric technique aiming to apply this material to 3D clinical doses evaluations using MRI technique. In this work, the optical and the energy dependent response of this solution submitted to clinical photons and electrons beams were studied. Different batches of gel solutions were prepared and maintained at low temperature during 12 h to solidification. Before irradiation, the samples were maintained during 1 h at room temperature. The photons and electrons irradiations were carried out using a Varian 2100C Medical Linear Accelerator of the Radiotherapy Department of the Hospital das Clinicas of the University of Sao Paulo with absorbed doses between 1 and 40 Gy; radiation field of 10 x 10 cm 2 ; photon energies of 6 MeV and 15 MeV; and electron with energies between 6 and 15 MeV. The obtained results indicate that signal response dependence for clinical photons and electrons beams, to the same doses, for Alanine gel dosimeter is better than 3.6 % (1σ), and the energy dependence response, to the same doses, is better 3% (1σ) for both beams. These results indicate that the same calibration factor can be used and the optical response is energy independent in the studied dose range and clinical photons and electrons beams energies. (author)
Modification of H2O adsorbed Si(100)-(2 x 1) surface by photon and electron beam
International Nuclear Information System (INIS)
Moon, S.W.; Chung, S.M.; Hwang, C.C.; Ihm, K.W.; Kang, T.-H.; Chen, C.H.; Park, C.-Y.
2004-01-01
Full text: Oxidation of silicon has been the subject of intense scientific and technological interest due to the several uses of thin oxide films as insulating layers in microelectronic devices. The great strides have been made in understanding about the formation and thermal evolution of the Si/SiO 2 interface. In this presentation, we provide synchrotron radiation photoemission spectroscopy (SRPES) and photoemission electron microscope (PEEM) results, showing how a H 2 O adsorbed Si(100) surface evolves into an ultra-thin silicon oxide m when exposed to monochromatized synchrotron radiation and electron beam at room temperature. All SRPES, PEEM experiments have been performed at the beam line, 4B1, of Pohang Light Source (PLS) in Korea. Water dissociates into OH(a) and H(a) species upon adsorption on the Si(100)-(2 - 1) at room temperature. The bonding (b 2 ) and antibonding (a 1 ) OH orbital and the oxygen lone pair orbital (b 1 ) from the dissociated OH and H species has been identified in ultraviolet photoemission spectra (UPS). These structures gradually changed and a new silicon oxide peak appeared with the photon/E-beam irradiation. This indicates that the H 2 O adsorbed on Si surface transforms into a thin silicon oxide film by photon/E-beam irradiation. We have shown in our PEEM images that one can make micro-patterns on silicon surface by using the photon induced surface modification. The fabricated patterns can be clearly identified through the inverse contrast images between photon exposed region and unexposed one. The near edge x-ray absorption fine structure (NEXAFS) results revealed that the OH adsorbed Si surface transforms into a thin silicon oxide film by photon irradiation
Energy Technology Data Exchange (ETDEWEB)
Plessis, F C P du; Leal, A; Stathakis, S; Xiong, W; Ma, C-M [Department of Radiation Oncology, Fox Chase Cancer Center, Philadelphia, PA 19111 (United States)
2006-04-21
A study is presented that characterizes megavoltage electron beams delivered through an existing double-focused photon multi-leaf collimator (pMLC) using film measurements in a solid water phantom. Machine output stability and linearity were evaluated as well as the effect of source-to-surface distance (SSD) and field size on the penumbra for electron energies between 6 and 18 MeV over an SSD range of 60-100 cm. Penumbra variations as a function of field size, depth of measurement and the influence of the jaws were also studied. Field abutment, field flatness and target coverage for segmented beams were also addressed. The measured field size for electrons transported through the pMLC was the same as that for an x-ray beam up to SSDs of 70 cm. At larger SSD, the lower energy electron fields deviated from the projected field. Penumbra data indicated that 60 cm SSD was the most favourable treatment distance. Backprojection of P{sub 20-80} penumbra data yielded a virtual source position located at 98.9 cm from the surface for 18 MeV electrons. For 6 MeV electrons, the virtual source position was at a distance of 82.6 cm. Penumbra values were smaller for small beam slits and reached a near-constant value for field widths larger than 5 cm. The influence of the jaws had a small effect on the penumbra. The R{sub 90} values ranged from 1.4 to 4.8 cm between 6 and 21 MeV as measured at 60 cm SSD for a 9 x 9 cm{sup 2} field. Uniformity and penumbra improvement could be demonstrated using weighted abutted fields especially useful for small segments. No detectable electron leakage through the pMLC was observed. Bremsstrahlung measurements taken at 60 cm SSD for a 9 x 9 cm{sup 2} field as shaped by the pMLC compared within 1% to bremsstrahlung measurements taken at 100 cm SSD for a 10 x 10 cm{sup 2} electron applicator field at 100 cm SSD.
International Nuclear Information System (INIS)
Lucherini, V.; Bianchi, N.; De Sanctis, E.; Guaraldo, C.; Levi Sandri, P.; Muccifora, V.; Polli, E.; Reolon, A.R.; Aiello, S.; De Filippo, E.; Lanzano', G.; Lo Nigro, S.; Milone, C.; Pagano, A.; Botvina, A.S.; Iljinov, A.S.; Mebel, M.V.
1990-01-01
The mechanisms of excitation with subsequent fission of heavy nuclei can be conveniently studied by means of photons, since this probe is able to interact deeply inside the nucleus. We propose the use of the (200-1200 MeV) tagged photon beam from the ADONE Jet Target in order to study the mass-energy and total momentum distributions of fission fragments, to obtain experimental information on the configurations (excitation energy and nucleonic composition) of produced compound nuclei and on their decay channels
Negative refraction at infrared wavelengths in a two-dimensional photonic crystal
International Nuclear Information System (INIS)
Berrier, A.; Mulot, M.; Swillo, M.; Qiu, M.; Thylen, L.; Anand, S.; Talneau, A.
2004-01-01
We report on the first experimental evidence of negative refraction at telecommunication wavelengths by a two-dimensional photonic crystal field. Samples were fabricated by chemically assisted ion beam etching in the InP-based low-index constrast system. Experiments of beam imaging and light collection show light focusing by the photonic crystal field. Finite-difference time-domain simulations confirm that the observed focusing is due to negative refraction in the photonic crystal area
SU-F-T-491: Photon Beam Matching Analysis at Multiple Sites Up to Twelve Years Post Installation
Energy Technology Data Exchange (ETDEWEB)
Able, C [Florida Cancer Specialists - New Port Richey, New Port Richey, FL (United States); Zakikhani, R [Florida Cancer Specialists - Largo, Largo, FL (United States); Yan, K [Florida Cancer Specialists- Tampa, Tampa, FL (United States); Sha, D [Florida Cancer Specialist - New Port Richey, New Port Richey, FL (United States); Chopra, A [Florida Cancer Specialist - Brooksville, Spring Hill, FL (United States)
2016-06-15
Purpose: To determine if the photon beams associated with several models of accelerators are matched with ‘Golden Beam’ data (VGBD) to assess treatment planning modeling and delivery. Methods: Six accelerators’ photon beams were evaluated to determine if they matched the manufacturer’s (Varian Medical Systems, Inc.) VGBD. Additional direct comparisons of the 6X and 18X beams using the manufacturer’s specification of Basic and Fine beam matching were also performed. The Cseries accelerator models were 21 EX (3), IX (2), and a IX Trilogy, ranging from three to twelve years post installation. Computerized beam scanning was performed (IBA Blue Phantom 2) with 2 CC13 ion chambers in water at 100 cm SSD. Dmax (10 cm2 field size), percentage depth dose (6 cm2, 10 cm2, 20 cm2, and 30 cm2 field sizes) and beam uniformity (10 cm2, 30 cm2 and 40 cm2 field sizes) were evaluated. Results: When comparing the beams with VGBD using the ‘Basic’ matching criteria, all beams were within the specifications ( 1.5mm at dmax, 1% PDD, and 2% Profiles). When considering the “Fine” matching criteria ( 1.5mm at dmax, 0.5% PDD, and 2% Profiles), only three of six 6MV beams and two of six high energy (five 18MV & one 15MV) beams passed. Direct comparisons between accelerators using the Clinac IX (installed 2012) as the reference beam datasets resulted in all 6 MV and 18MV beams meeting both the “Basic” and “Fine” criterion with the exception of two accelerators. Conclusion: Linear accelerators installed up to nine years apart are capable of meeting the manufacturers beam matching criteria for “Basic” matching. Without any adjustments most beams, when evaluated, may meet the “Fine” match criteria. The use of a single dataset (VGBD or designated accelerator reference data) for treatment planning commissioning is acceptable and can provide quality treatment delivery.
International Nuclear Information System (INIS)
Sereno, Nicholas S.
1998-01-01
The Advanced Photon Source (APS) linac beam position monitors can be used to monitor the position of a beam containing both positrons and electrons. To accomplish this task, both the signal at the bunching frequency of 2856 MHz and the signal at 2x2856 MHz are acquired and processed for each stripline. The positron beam position is obtained by forming a linear combination of both 2856 and 5712 MHz signals for each stripline and then performing the standard difference over sum computation. The required linear combination of the 2856 and 5712 MHz signals depends on the electrical calibration of each stripline/cable combination. In this paper, the calibration constants for both 2856 MHz and 5712 MHz signals for each stripline are determined using a pure beam of electrons. The calibration constants are obtained by measuring the 2856 and 5712 MHz stripline signals at various electron beam currents and positions. Finally, the calibration constants measured using electrons are used to determine positron beam position for the mixed beam case
Ravichandran, Ramamoorthy; Binukumar, John Pichy; Al Amri, Iqbal; Davis, Cheriyathmanjiyil Antony
2016-03-08
Diamond detectors (DD) are preferred in small field dosimetry of radiation beams because of small dose profile penumbras, better spatial resolution, and tissue-equivalent properties. We investigated a commercially available 'microdiamond' detector in realizing absorbed dose from first principles. A microdiamond detector, type TM 60019 with tandem electrometer is used to measure absorbed doses in water, nylon, and PMMA phantoms. With sensitive volume 0.004 mm3, radius 1.1mm, thickness 1 x10(-3) mm, the nominal response is 1 nC/Gy. It is assumed that the diamond detector could collect total electric charge (nC) developed during irradiation at 0 V bias. We found that dose rate effect is less than 0.7% for changing dose rate by 500 MU/min. The reproducibility in obtaining readings with diamond detector is found to be ± 0.17% (1 SD) (n = 11). The measured absorbed doses for 6 MV and 15 MV photons arrived at using mass energy absorption coefficients and stop-ping power ratios compared well with Nd, water calibrated ion chamber measured absorbed doses within 3% in water, PMMA, and nylon media. The calibration factor obtained for diamond detector confirmed response variation is due to sensitivity due to difference in manufacturing process. For electron beams, we had to apply ratio of electron densities of water to carbon. Our results qualify diamond dosimeter as a transfer standard, based on long-term stability and reproducibility. Based on micro-dimensions, we recommend these detectors for pretreatment dose verifications in small field irradiations like stereotactic treatments with image guidance.
International Nuclear Information System (INIS)
Hayashi, Naoki; Shibamoto, Yuta; Obata, Yasunori; Kimura, Takashi; Nakazawa, Hisato; Hagiwara, Masahiro; Hashizume, Chisa I.; Mori, Yoshimasa; Kobayashi, Tatsuya
2010-01-01
The purpose of this study was to evaluate the effect of megavoltage photon beam attenuation (PBA) by couch tops and to propose a method for correction of PBA. Four series of phantom measurements were carried out. First, PBA by the exact couch top (ECT, Varian) and Imaging Couch Top (ICT, BrainLAB) was evaluated using a water-equivalent phantom. Second, PBA by Type-S system (Med-Tec), ECT and ICT was compared with a spherical phantom. Third, percentage depth dose (PDD) after passing through ICT was measured to compare with control data of PDD. Forth, the gantry angle dependency of PBA by ICT was evaluated. Then, an equation for PBA correction was elaborated and correction factors for PBA at isocenter were obtained. Finally, this method was applied to a patient with hepatoma. PBA of perpendicular beams by ICT was 4.7% on average. With the increase in field size, the measured values became higher. PBA by ICT was greater than that by Type-S system and ECT. PBA increased significantly as the angle of incidence increased, ranging from 4.3% at 180 deg to 11.2% at 120 deg. Calculated doses obtained by the equation and correction factors agreed quite well with the measured doses between 120 deg and 180 deg of angles of incidence. Also in the patient, PBA by ICT was corrected quite well by the equation and correction factors. In conclusion, PBA and its gantry angle dependency by ICT were observed. This simple method using the equation and correction factors appeared useful to correct the isocenter dose when the PBA effect cannot be corrected by a treatment planning system. (author)
Directory of Open Access Journals (Sweden)
AHMED W. AL-ZAND
2017-01-01
Full Text Available Nonlinear finite element (FE models are prepared to investigate the behaviour of concrete-filled steel tube (CFST beams strengthened by carbon fibre reinforced polymer (CFRP sheets. The beams are strengthened from the bottom side only by varied sheet lengths (full and partial beam lengths and then subjected to ultimate flexural loads. Three surface interaction techniques are used to implement the bonding behaviour between the steel tube and the CFRP sheet, namely, full tie interaction (TI, cohesive element (CE and cohesive behaviour (CB techniques using ABAQUS software. Results of the comparison between the FE analysis and existing experimental study confirm that the FE models with the TI technique could be applicable for beams strengthened by CFRP sheets with a full wrapping length; the technique could not accurately implement the CFRP delamination failure, which occurred for beams with a partial wrapping length. Meanwhile, the FE models with the CE and CB techniques are applicable in the implementation of both CFRP failures (rapture and delamination for both full and partial wrapping lengths, respectively. Where, the ultimate loads' ratios achieved by the FE models using TI, CE and CB techniques about 1.122, 1.047 and 1.045, respectively, comparing to the results of existing experimental tests.
Sukrawa, Made
2017-11-01
Experimental and analytical researches on the effect of web opening in steel beams have been repeatedly reported in literature because of the advantages gain from the many function of the opening. Most of the research on this area, however, did not consider deformation and stress in the beam due to axial force. In seismic design of steel structure, the axial force in the beam could be significantly high and therefore worth considering. In this study a beam extracted from a braced frame structure was analyzed using finite element models to investigate the effect of combined bending and axial forces on the deformation and stresses in the vicinity of the opening. Large size of square, rectangular, and circular openings of the same depth were reinforced and placed in pair, symmetrical to the concentrated load at mid span of the beam. Four types of reinforcement were used, all around (AA), short horizontal (SH), long horizontal (LH), and doubler plate (DP). The effect of axial load was also investigated using rigid frame model loaded vertically and laterally. Validation of the modelling technique was done prior to the parametric study. It was revealed that the axial force significantly contributes to the stress concentration near the hole. Stiffener of circular shape was effective to improve the stress distribution around the circular opening. For square and rectangular openings, however, the horizontal stiffener, extended beyond the edge of opening, performed better than the other type of stiffeners.
Energy Technology Data Exchange (ETDEWEB)
Singh, Ashutosh, E-mail: asingh.rs.ece@iitbhu.ac.in [Faculty of Physical Sciences, Institute of Natural Sciences and Humanities, Shri Ramswaroop Memorial University, Lucknow-Deva Road, Uttar Pradesh 225003 (India); Center of Research in Microwave Tubes, Department of Electronics Engineering, Indian Institute of Technology (Banaras Hindu University), Varanasi 221005 (India); Jain, P. K. [Center of Research in Microwave Tubes, Department of Electronics Engineering, Indian Institute of Technology (Banaras Hindu University), Varanasi 221005 (India)
2015-09-15
In this paper, the effects of electron beam parameters and velocity spread on the RF behavior of a metallic photonic band gap (PBG) cavity gyrotron operating at 35 GHz with TE{sub 041}–like mode have been theoretically demonstrated. PBG cavity is used here to achieve a single mode operation of the overmoded cavity. The nonlinear time-dependent multimode analysis has been used to observe the beam-wave interaction behavior of the PBG cavity gyrotron, and a commercially available PIC code “CST Particle Studio” has been reconfigured to obtain 3D simulation results in order to validate the analytical values. The output power for this typical PBG gyrotron has been obtained ∼108 kW with ∼15.5% efficiency in a well confined TE{sub 041}–like mode, while all other competing modes have significantly low values of power output. The output power and efficiency of a gyrotron depend highly on the electron beam parameters and velocity spread. The influence of several electron beam parameters, e.g., beam voltage, beam current, beam velocity pitch factor, and DC magnetic field, on the PBG gyrotron operations has been investigated. This study would be helpful in optimising the electron beam parameters and estimating accurate RF output power of the high frequency PBG cavity based gyrotron oscillators.
International Nuclear Information System (INIS)
Fan Zhen-Kai; Li Shu-Guang; Fan Yu-Qiu; Zhang Wan; An Guo-Wen; Bao Ya-Jie
2014-01-01
We design a novel kind of polarization beam splitter based on a gold-filled dual-core photonic crystal fiber (DC-PCF). Owing to filling with two gold wires in this DC-PCF, its coupling characteristics can be changed greatly by the second-order surface plasmon polariton (SPP) and the resonant coupling between the surface plasmon modes and the fiber-core guided modes can enhance the directional power transfer in the two fiber-cores. Numerical results by using the finite element method show the extinction ratio at the wavethlengths of 1.327 μm and 1.55 μm can reach −58 dB and −60 dB and the bandwidths as the extinction ratio better than −12 dB are about 54 nm and 47 nm, respectively. Compared with the gold-unfilled DC-PCF, a 1.746-mm-long gold-filled DC-PCF is better applied to the polarization beam splitter in the two communication bands of λ = 1.327 μm and 1.55 μm. (electromagnetism, optics, acoustics, heat transfer, classical mechanics, and fluid dynamics)
Finite Element Formulation for Stability and Free Vibration Analysis of Timoshenko Beam
Directory of Open Access Journals (Sweden)
Abbas Moallemi-Oreh
2013-01-01
Full Text Available A two-node element is suggested for analyzing the stability and free vibration of Timoshenko beam. Cubic displacement polynomial and quadratic rotational fields are selected for this element. Moreover, it is assumed that shear strain of the element has the constant value. Interpolation functions for displacement field and beam rotation are exactly calculated by employing total beam energy and its stationing to shear strain. By exploiting these interpolation functions, beam elements' stiffness matrix is also examined. Furthermore, geometric stiffness matrix and mass matrix of the proposed element are calculated by writing governing equation on stability and beam free vibration. At last, accuracy and efficiency of proposed element are evaluated through numerical tests. These tests show high accuracy of the element in analyzing beam stability and finding its critical load and free vibration analysis.
Finite Element Models for Electron Beam Freeform Fabrication Process, Phase II
National Aeronautics and Space Administration — This Small Business Innovation Research Phase II proposal offers to develop a comprehensive computer simulation methodology based on the finite element method for...
Dose-calculation algorithms in the context of inhomogeneity corrections for high energy photon beams
International Nuclear Information System (INIS)
Papanikolaou, Niko; Stathakis, Sotirios
2009-01-01
Radiation therapy has witnessed a plethora of innovations and developments in the past 15 years. Since the introduction of computed tomography for treatment planning there has been a steady introduction of new methods to refine treatment delivery. Imaging continues to be an integral part of the planning, but also the delivery, of modern radiotherapy. However, all the efforts of image guided radiotherapy, intensity-modulated planning and delivery, adaptive radiotherapy, and everything else that we pride ourselves in having in the armamentarium can fall short, unless there is an accurate dose-calculation algorithm. The agreement between the calculated and delivered doses is of great significance in radiation therapy since the accuracy of the absorbed dose as prescribed determines the clinical outcome. Dose-calculation algorithms have evolved greatly over the years in an effort to be more inclusive of the effects that govern the true radiation transport through the human body. In this Vision 20/20 paper, we look back to see how it all started and where things are now in terms of dose algorithms for photon beams and the inclusion of tissue heterogeneities. Convolution-superposition algorithms have dominated the treatment planning industry for the past few years. Monte Carlo techniques have an inherent accuracy that is superior to any other algorithm and as such will continue to be the gold standard, along with measurements, and maybe one day will be the algorithm of choice for all particle treatment planning in radiation therapy.
Extended calibration range for prompt photon emission in ion beam irradiation
Bellini, F.
2014-01-01
Monitoring the dose delivered during proton and carbon ion therapy is still a matter of research. Among the possible solutions, several exploit the measurement of the single photon emission from nuclear decays induced by the irradiation. To fully characterize such emission the detectors need development, since the energy spectrum spans the range above the MeV that is not traditionally used in medical applications. On the other hand, a deeper understanding of the reactions involving gamma production is needed in order to improve the physic models of Monte Carlo codes, relevant for an accurate prediction of the prompt-gamma energy spectrum.This paper describes a calibration technique tailored for the range of energy of interest and reanalyzes the data of the interaction of a 80MeV/u fully stripped carbon ion beam with a Poly-methyl methacrylate target. By adopting the FLUKA simulation with the appropriate calibration and resolution a significant improvement in the agreement between data and simulation is report...
Modrzynski, Pawel; Knapitsch, Arno; Kunicki, Piotr; Lecoq, Paul; Moczala, Magdalena; Papakonstantinou, Ioannis; Auffray, Etiennette
2016-01-01
“Photonic Crystals (PhC)” have been used in a variety of fields as a structure for improving the light extraction efficiency from materials with high index of refraction. In previous work we already showed the light extraction improvement of several PhC covered LYSO crystals in computer simulations and practical measurements. In this work, new samples are made using different materials and techniques which allows further efficiency improvements. For rapid prototyping of PhC patterns on scintillators we tested a new method using “Focused Ion Beam (FIB)” patterning. The FIB machine is a device similar to a “Scanning Electron Microscope (SEM)”, but it uses ions (mainly gallium) instead of electrons for the imaging of the samples' surface. The additional feature of FIB devices is the option of surface patterning in nano-scale which was exploited for our samples. Three samples using FIB patterning have been produced. One of them is a direct patterning of the extraction face of a 0.8×0.8×10 $mm^3$ LYS...
Extended calibration range for prompt photon emission in ion beam irradiation
International Nuclear Information System (INIS)
Bellini, F.; Boehlen, T.T.; Chin, M.P.W.; Collamati, F.; De Lucia, E.; Faccini, R.; Ferrari, A.; Lanza, L.; Mancini-Terracciano, C.; Marafini, M.; Mattei, I.; Morganti, S.; Ortega, P.G.; Patera, V.; Piersanti, L.; Russomando, A.; Sala, P.R.
2014-01-01
Monitoring the dose delivered during proton and carbon ion therapy is still a matter of research. Among the possible solutions, several exploit the measurement of the single photon emission from nuclear decays induced by the irradiation. To fully characterize such emission the detectors need development, since the energy spectrum spans the range above the MeV that is not traditionally used in medical applications. On the other hand, a deeper understanding of the reactions involving gamma production is needed in order to improve the physic models of Monte Carlo codes, relevant for an accurate prediction of the prompt-gamma energy spectrum. This paper describes a calibration technique tailored for the range of energy of interest and reanalyzes the data of the interaction of a 80 MeV/u fully stripped carbon ion beam with a Poly-methyl methacrylate target. By adopting the FLUKA simulation with the appropriate calibration and resolution a significant improvement in the agreement between data and simulation is reported
Extended calibration range for prompt photon emission in ion beam irradiation
Energy Technology Data Exchange (ETDEWEB)
Bellini, F. [Dipartimento di Fisica, Sapienza Università di Roma, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Boehlen, T.T.; Chin, M.P.W. [CERN, Geneva (Switzerland); Collamati, F. [Dipartimento di Fisica, Sapienza Università di Roma, Roma (Italy); INFN Sezione di Roma, Roma (Italy); De Lucia, E. [Laboratori Nazionali di Frascati dell' INFN, Frascati (Italy); Faccini, R., E-mail: riccardo.faccini@roma1.infn.it [Dipartimento di Fisica, Sapienza Università di Roma, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Ferrari, A. [CERN, Geneva (Switzerland); Lanza, L. [Dipartimento di Fisica, Sapienza Università di Roma, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Mancini-Terracciano, C. [CERN, Geneva (Switzerland); Dipartimento di Fisica, Università Roma Tre, Roma (Italy); Marafini, M. [Museo Storico della Fisica e Centro Studi e Ricerche “E. Fermi”, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Mattei, I. [Dipartimento di Fisica, Università Roma Tre, Roma (Italy); Laboratori Nazionali di Frascati dell' INFN, Frascati (Italy); Morganti, S. [INFN Sezione di Roma, Roma (Italy); Ortega, P.G. [CERN, Geneva (Switzerland); Patera, V. [Dipartimento di Scienze di Base e Applicate per Ingegneria, Sapienza Università di Roma, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Piersanti, L. [Dipartimento di Scienze di Base e Applicate per Ingegneria, Sapienza Università di Roma, Roma (Italy); Laboratori Nazionali di Frascati dell' INFN, Frascati (Italy); Russomando, A. [Center for Life Nano Science@Sapienza, Istituto Italiano di Tecnologia, Roma (Italy); INFN Sezione di Roma, Roma (Italy); Sala, P.R. [INFN Sezione di Milano, Milano (Italy); and others
2014-05-01
Monitoring the dose delivered during proton and carbon ion therapy is still a matter of research. Among the possible solutions, several exploit the measurement of the single photon emission from nuclear decays induced by the irradiation. To fully characterize such emission the detectors need development, since the energy spectrum spans the range above the MeV that is not traditionally used in medical applications. On the other hand, a deeper understanding of the reactions involving gamma production is needed in order to improve the physic models of Monte Carlo codes, relevant for an accurate prediction of the prompt-gamma energy spectrum. This paper describes a calibration technique tailored for the range of energy of interest and reanalyzes the data of the interaction of a 80 MeV/u fully stripped carbon ion beam with a Poly-methyl methacrylate target. By adopting the FLUKA simulation with the appropriate calibration and resolution a significant improvement in the agreement between data and simulation is reported.
Steinetz, Bruce M.; Benyo, Theresa L.; Pines, Vladimir; Pines, Marianna; Forsley, Lawrence P.; Westmeyer, Paul A.; Chait, Arnon; Becks, Michael D.; Martin, Richard E.; Hendricks, Robert C.;
2017-01-01
Exposure of highly deuterated materials to a low-energy (nom. 2 MeV) photon beam resulted in nuclear activity of both the parent metals of hafnium and erbium and a witness material (molybdenum) mixed with the reactants. Gamma spectral analysis of all deuterated materials, ErD2.8+C36D74+Mo and HfD2+C36D74+Mo, showed that nuclear processes had occurred as shown by unique gamma signatures. For the deuterated erbium specimens, posttest gamma spectra showed evidence of radioisotopes of erbium ((163)Er and (171)Er) and of molybdenum ((99)Mo and (101)Mo) and by beta decay, technetium (99mTc and 101Tc). For the deuterated hafnium specimens, posttest gamma spectra showed evidence of radioisotopes of hafnium (180mHf and 181Hf) and molybdenum ((99)Mo and (101)Mo), and by beta decay, technetium ((99m)Tc and (101)Tc). In contrast, when either the hydrogenated or non-gas-loaded erbium or hafnium materials were exposed to the gamma flux, the gamma spectra revealed no new isotopes. Neutron activation materials showed evidence of thermal and epithermal neutrons. CR-39 solid-state nuclear track detectors showed evidence of fast neutrons with energies between 1.4 and 2.5 MeV and several instances of triple tracks, indicating (is) greater than 10 MeV neutrons. Further study is required to determine the mechanism causing the nuclear activity.
Tissue classifications in Monte Carlo simulations of patient dose for photon beam tumor treatments
Lin, Mu-Han; Chao, Tsi-Chian; Lee, Chung-Chi; Tung-Chieh Chang, Joseph; Tung, Chuan-Jong
2010-07-01
The purpose of this work was to study the calculated dose uncertainties induced by the material classification that determined the interaction cross-sections and the water-to-material stopping-power ratios. Calculations were made for a head- and neck-cancer patient treated with five intensity-modulated radiotherapy fields using 6 MV photon beams. The patient's CT images were reconstructed into two voxelized patient phantoms based on different CT-to-material classification schemes. Comparisons of the depth-dose curve of the anterior-to-posterior field and the dose-volume-histogram of the treatment plan were used to evaluate the dose uncertainties from such schemes. The results indicated that any misassignment of tissue materials could lead to a substantial dose difference, which would affect the treatment outcome. To assure an appropriate material assignment, it is desirable to have different conversion tables for various parts of the body. The assignment of stopping-power ratio should be based on the chemical composition and the density of the material.
Penumbra characteristics of square photon beams delimited by a GEMS multi-leaf collimator
Energy Technology Data Exchange (ETDEWEB)
Briot, E; Julia, F [Centre de Lutte Contre le Cancer Gustave-Roussy, 94 - Villejuif (France)
1995-12-01
A multi-leaf collimator (MLC) has been designed to replace directly the standard collimator of a SATURNE IV Series linac. It consists of 2 x 32 tungsten leaves and one set of upper block jaws. Isodose curves and dose profiles were measured for symmetric fields at the depth of the maximum and the reference depths for 6 MV, 10 MV, 18 MV photon beams. The penumbra (80%-20%) corresponding to the face and the side of the leaves have been compared with the standard collimators. Along with the X direction, the field delimitation is performed primarily with the leaves which are continuously variable in position. Along the Y direction, the field is initially approximated by the closure of opposite leaf pairs; then the Y upper jaws produce the exact size of the required field. As the leaves move linearly the penumbra (80%-20%) corresponding to the leaf ends is minimized and held constant at all positions by curvature of their faces. Penumbra obtained with the superposition of leaves and Y jaws depend on their relative position. The penumbra is minimum when the leaf side and the Y jaw edge coincide and the comparison of the measurement values with the conventional collimator shows that the differences are within 1 mm. When the leaves delineating the field are not entirely covered by the Y block upper jaws, the penumbra increases, and the junction of the opposing leaves, a width increase up to 3.5 mm has been measured.
The practical application of scintillation dosimetry in small-field photon-beam radiotherapy
Energy Technology Data Exchange (ETDEWEB)
Burke, Elisa; Poppinga, Daniela; Schoenfeld, Andreas A.; Poppe, Bjoern; Looe, Hui Khee [Oldenburg Univ. (Germany). Univ. Clinic for Medical Radiation Physics; Harder, Dietrich [Goettingen Univ. (Germany). Medical Physics and Biophysics
2017-07-01
Plastic scintillation detectors are a new instrument of stereotactic photon-beam dosimetry. The clinical application of the plastic scintillation detector Exradin W1 at the Siemens Artiste and Elekta Synergy accelerators is a matter of current interest. In order to reduce the measurement uncertainty, precautions have to be taken with regard to the geometrical arrangement of the scintillator, the light-guide fiber and the photodiode in the radiation field. To determine the ''Cerenkov light ratio'' CLR with a type A uncertainty below 1%, the Cerenkov calibration procedure for small-field measurements based on the two-channel spectral method was used. Output factors were correctly measured with the W1 for field sizes down to 0.5 x 0.5 cm{sup 2} with a type A uncertainty of 1.8%. Measurements of small field dose profiles and percentage depth dose curves were carried out with the W1 using automated water phantom profile scans, and a type A uncertainty for dose maxima of 1.4% was achieved. The agreement with a synthetic diamond detector (microDiamond, PTW Freiburg) and a plane parallel ionization chamber (Roos chamber, PTW Freiburg) in relative dose measurements was excellent. In oversight of all results, the suitability of the plastic scintillation detector Exradin W1 for clinical dosimetry under stereotactic conditions, in particular the tried and tested procedures for CLR determination, output factor measurement and automated dose profile scans in water phantoms, have been confirmed.
An Analytical-empirical Calculation of Linear Attenuation Coefficient of Megavoltage Photon Beams.
Seif, F; Tahmasebi-Birgani, M J; Bayatiani, M R
2017-09-01
In this study, a method for linear attenuation coefficient calculation was introduced. Linear attenuation coefficient was calculated with a new method that base on the physics of interaction of photon with matter, mathematical calculation and x-ray spectrum consideration. The calculation was done for Cerrobend as a common radiotherapy modifier and Mercury. The values of calculated linear attenuation coefficient with this new method are in acceptable range. Also, the linear attenuation coefficient decreases slightly as the thickness of attenuating filter (Cerrobend or mercury) increased, so the procedure of linear attenuation coefficient variation is in agreement with other documents. The results showed that the attenuation ability of mercury was about 1.44 times more than Cerrobend. The method that was introduced in this study for linear attenuation coefficient calculation is general enough to treat beam modifiers with any shape or material by using the same formalism; however, calculating was made only for mercury and Cerrobend attenuator. On the other hand, it seems that this method is suitable for high energy shields or protector designing.
Development of a system to verify the programs used for planning of photon beams teletherapy
International Nuclear Information System (INIS)
Ocariz Ayala, Victor Daniel
2004-12-01
The main objective of radiotherapy is to deliver to the tumor the radiation dose prescribed by the physician, in the most possible accurate form, to save, as much as possible, the healthy tissues located in the neighborhood of the tumor. In order to reach these objectives, it is necessary to carry out a treatment planning and the more the used technologies and therapeutical procedures are sophisticated, the more the planning will be sophisticated. The most sophisticated planning systems use computer programs and are able to determine dose distributions in three dimensions. However, since they work using mathematical models, they may fail and it is necessary to evaluate their performances in order to be considered reliable. Therefore, the availability of a system capable to evaluate the performance of planning systems employed in oncological teletherapy, using ionizing radiation, becomes important. In this work, a data file to be used in radiotherapy planning system quality control (Algorithm accuracy and dose distribution) was developed and it is able to be sent by mail to the radiotherapy services that work with photon beams. (author)
International Nuclear Information System (INIS)
Almeida, Laila G.; Amaral, Leonardo L.; Oliveira, Harley F.; Maia, Ana F.
2012-01-01
The virtual wedge is useful tool in the radiation treatment planning since it has series of advantages over the hard wedge. Quality control tests ensure correct performance of the planning done in treatment planning systems (TPS). This study aimed to compare doses calculated by TPS and doses measured by ionization chamber (CI) and an ionization chambers array in virtual wedge photon beams of 6 MV. Measures carried out in Primus linear accelerator with a solid water phantom and dosimeter positioned at 10 cm depth with gantry at 0° in many fields sizes and angles in the virtual wedge. Measurements on the central axis used as dosimeter an IC and on off-axis used an IC array. The simulation in CMS-XiO used the CT images of the phantom in the same configuration of the irradiation. Maximum and minimum values of the percentage differences between the doses provided by TPS and measurements with ionization chamber on the central axis were 1.43 and -0.10%, respectively, with average percentage difference of 0.08% and confidence limit of Δ=1.72%. In the region off-axis, the average percentage difference was 0.04%, with a maximum of 1.9%, minimum of 0% and confidence limit of Δ=1.91%. All values for dose percentage differences were below 2% and lower confidence limit of 3% are thus, according to the recommendations of the Technical Report Series - TRS-430. (author)
The Study of Tissue Dose Perturbation by Air Cavity with 6MV Photon Beam
International Nuclear Information System (INIS)
Shin, Byung Chul; Yoo, Myung Jin; Moon, Chang Woo; Jeung, Tae Sig; Yum, Ha Yong
1995-01-01
Purpose : To determine the perturbation effect in the tissue downstream from surface layers of lesions located in the air/tumor-tissue interface of larynx using 6MV photon beam. Materials and Methods : Thermoluminescent dosimeters(TLDs). Were embedded at 3 measurement locations in slab no.7 of a humanoid phantom and exposed to forward and backward direction using various field sizes(4 X 4cm 2 - 15 X 15 cm 2 ). Results : At the air/tissue interface, forward dose perturbation factor(FDPF) is about 1.085 with 4 X 4 cm 2 , 1.05 with 7 X 7 cm 2 , 1.048 with 10 X 10 cm 2 , and 1.041 with 15 X 15 cm 2 . Backscatter dose perturbation factor(BDPF) is about 0.99 with 4 X 4 cm 2 , 0.981 with 7 X 7 cm 2 , 0.956 with 10 X 10 cm 2 and 0.97 with 15 X 15 cm 2 . Conclusion : FDPF is greater as field size is smaller. And FDPF is smaller as the distance is further from the air/tissue interface
Preliminary design of the beam loss monitor system for the Advanced Photon Source
International Nuclear Information System (INIS)
Patterson, D.R.
1992-01-01
The preliminary design of the beam loss monitor for the ANL Advanced Photon Source is based on the use of an air dielectric coaxial cable as a long ionization chamber. Each coaxial cable section uses a high sensitivity DC current-to-voltage converter with both linear and integrating ranges. Pulse timing circuits determine the positions of individual losses by measuring the time at which the resulting voltage pulses arrive at the beginning of the coaxial ionization chamber. A possible timing ambiguity can be removed by correlating the particle bunch timing with the resulting voltage pulse timing. Measurements have shown that pulse rise times less than 15 nanoseconds can be obtained, so that determining loss locations to better than 7 feet may be possible. Best performance may be obtained when 500 VDC is applied to a 50-ohm, 7/8-inch air dielectric coaxial cable filled with approximately 8 psig of a 95% argon, 5% carbon dioxide gas mixture. Cable lengths will be between 100 and 300 feet long, depending on the part of the accelerator being monitored
Tissue classifications in Monte Carlo simulations of patient dose for photon beam tumor treatments
International Nuclear Information System (INIS)
Lin, Mu-Han; Chao, Tsi-Chian; Lee, Chung-Chi; Tung-Chieh Chang, Joseph; Tung, Chuan-Jong
2010-01-01
The purpose of this work was to study the calculated dose uncertainties induced by the material classification that determined the interaction cross-sections and the water-to-material stopping-power ratios. Calculations were made for a head- and neck-cancer patient treated with five intensity-modulated radiotherapy fields using 6 MV photon beams. The patient's CT images were reconstructed into two voxelized patient phantoms based on different CT-to-material classification schemes. Comparisons of the depth-dose curve of the anterior-to-posterior field and the dose-volume-histogram of the treatment plan were used to evaluate the dose uncertainties from such schemes. The results indicated that any misassignment of tissue materials could lead to a substantial dose difference, which would affect the treatment outcome. To assure an appropriate material assignment, it is desirable to have different conversion tables for various parts of the body. The assignment of stopping-power ratio should be based on the chemical composition and the density of the material.
Pradeep, K. R.; Thomas, A. M.; Basker, V. T.
2018-03-01
Structural health monitoring (SHM) is an essential component of futuristic civil, mechanical and aerospace structures. It detects the damages in system or give warning about the degradation of structure by evaluating performance parameters. This is achieved by the integration of sensors and actuators into the structure. Study of damage detection process in piezoelectric sensor and actuator integrated sandwich cantilever beam is carried out in this paper. Possible skin-core debond at the root of the cantilever beam is simulated and compared with undamaged case. The beam is actuated using piezoelectric actuators and performance differences are evaluated using Polyvinylidene fluoride (PVDF) sensors. The methodology utilized is the voltage/strain response of the damaged versus undamaged beam against transient actuation. Finite element model of piezo-beam is simulated in ANSYSTM using 8 noded coupled field element, with nodal degrees of freedoms are translations in the x, y directions and voltage. An aluminium sandwich beam with a length of 800mm, thickness of core 22.86mm and thickness of skin 0.3mm is considered. Skin-core debond is simulated in the model as unmerged nodes. Reduction in the fundamental frequency of the damaged beam is found to be negligible. But the voltage response of the PVDF sensor under transient excitation shows significantly visible change indicating the debond. Piezo electric based damage detection system is an effective tool for the damage detection of aerospace and civil structural system having inaccessible/critical locations and enables online monitoring possibilities as the power requirement is minimal.
International Nuclear Information System (INIS)
Barreras Caballero, A. A.; Hernandez Garcia, J.J.; Alfonso Laguardia, R.
2009-01-01
Were directly determined correction factors depending on the type camera beam quality, k, Q, and kQ, Qo, instead of the product (w, air p) Q, for three type cylindrical ionization chambers Pinpoint and divergent monoenergetic beams of photons in a wide range of energies (4-20 MV). The method of calculation used dispenses with the approaches taken in the classic procedure considered independent of braking power ratios and the factors disturbance of the camera. A detailed description of the geometry and materials chambers were supplied by the manufacturer and used as data input for the system 2006 of PENELOPE Monte Carlo calculation using a User code that includes correlated sampling, and forced interactions division of particles. We used a photon beam Co-60 as beam reference for calculating the correction factors for beam quality. No data exist for the cameras PTW 31014, 31015 and 31016 in the TRS-398 at they do not compare the results with data calculated or determined experimentally by other authors. (author)
Technical Report: Evaluation of peripheral dose for flattening filter free photon beams
Energy Technology Data Exchange (ETDEWEB)
Covington, E. L.; Moran, J. M.; Owrangi, A. M.; Prisciandaro, J. I., E-mail: joannp@med.umich.edu [Department of Radiation Oncology, University of Michigan, Ann Arbor, Michigan 48109 (United States); Ritter, T. A. [Department of Radiation Oncology, University of Michigan, Ann Arbor, Michigan 48109 and Department of Radiation Oncology, Veterans Affairs Ann Arbor Healthcare System, Ann Arbor, Michigan 48105 (United States)
2016-08-15
Purpose: To develop a comprehensive peripheral dose (PD) dataset for the two unflattened beams of nominal energy 6 and 10 MV for use in clinical care. Methods: Measurements were made in a 40 × 120 × 20 cm{sup 3} (width × length × depth) stack of solid water using an ionization chamber at varying depths (dmax, 5, and 10 cm), field sizes (3 × 3 to 30 × 30 cm{sup 2}), and distances from the field edge (5–40 cm). The effects of the multileaf collimator (MLC) and collimator rotation were also evaluated for a 10 × 10 cm{sup 2} field. Using the same phantom geometry, the accuracy of the analytic anisotropic algorithm (AAA) and Acuros dose calculation algorithm was assessed and compared to the measured values. Results: The PDs for both the 6 flattening filter free (FFF) and 10 FFF photon beams were found to decrease with increasing distance from the radiation field edge and the decreasing field size. The measured PD was observed to be higher for the 6 FFF than for the 10 FFF for all field sizes and depths. The impact of collimator rotation was not found to be clinically significant when used in conjunction with MLCs. AAA and Acuros algorithms both underestimated the PD with average errors of −13.6% and −7.8%, respectively, for all field sizes and depths at distances of 5 and 10 cm from the field edge, but the average error was found to increase to nearly −69% at greater distances. Conclusions: Given the known inaccuracies of peripheral dose calculations, this comprehensive dataset can be used to estimate the out-of-field dose to regions of interest such as organs at risk, electronic implantable devices, and a fetus. While the impact of collimator rotation was not found to significantly decrease PD when used in conjunction with MLCs, results are expected to be machine model and beam energy dependent. It is not recommended to use a treatment planning system to estimate PD due to the underestimation of the out-of-field dose and the inability to calculate dose
International Nuclear Information System (INIS)
Gur, D.; Bukovitz, A.G.; Rosen, J.C.; Holmes, B.G.
1979-01-01
Fast neutron contamination in photon beams in the 20 MV range have been reported in recent years. In order to determine if the variations were due mainly to differences in measurement procedures, or inherent in the design of the accelerators, three different 18-MV (BJR) photon beams were compared using identical analytical techniques. The units studied were a Philips SL/75-20 and a Siemens Mevatron-20 linear accelerators and a Schimadzu betatron. Gamma spectroscopy of an activated aluminum foil was the method used. By comparing the relative amounts of neutron contamination, errors associated with absolute measurements such as detector efficiency and differences in activation foils were eliminated. Fast neutron contaminations per rad of x rays in a ratio of 6.7:3.7:1 were found for the Philips, Schimadzu and Siemens accelerators, respectively
International Nuclear Information System (INIS)
Chung, Y.; Barr, D.; Decker, G.; Galayda, J.; Kirchman, J.; Lenkszus, F.; Lumpkin, A.; Votaw, A.J.
1995-01-01
The Advanced Photon Source (APS) is implementing an extensive beam position feedback system for both global and local stabilization of particle and photon beams based on digital signal processing. The description and operational experience of the system will be given in this paper. In particular, we will discuss the underlying fundamental principles, hardware layout, controls interface, and automatic software generation for multiple digital signal processors (DSPS) distributed in 20 VME crates around the ring. The feedback system runs at 4-kHz sampling frequency in order to achieve the correction bandwidth of approximately 100 Hz. For the maximum correction efficiency and resolution of conflicts among multiple local feedback systems due to the local bump closure error, the global and local feedback systems are combined into a single unified system. This novel approach is made possible through data sharing among the global and local systems via the fiber-optically networked reflective memories
Arakcheev, A.; Aulchenko, V.; Kudashkin, D.; Shekhtman, L.; Tolochko, B.; Zhulanov, V.
2017-06-01
Time-resolved experiments on the diffraction of synchrotron radiation (SR) from crystalline materials provide information on the evolution of a material structure after a heat, electron beam or plasma interaction with a sample under study. Changes in the material structure happen within a microsecond scale and a detector with corresponding parameters is needed. The SR channel 8 of the VEPP-4M storage ring provides radiation from the 7-pole wiggler that allows to rea