Fast Electron Beam Simulation and Dose Calculation
Trindade, A; Peralta, L; Lopes, M C; Alves, C; Chaves, A
2003-01-01
A flexible multiple source model capable of fast reconstruction of clinical electron beams is presented in this paper. A source model considers multiple virtual sources emulating the effect of accelerator head components. A reference configuration (10 MeV and 10x10 cm2 field size) for a Siemens KD2 linear accelerator was simulated in full detail using GEANT3 Monte Carlo code. Our model allows the reconstruction of other beam energies and field sizes as well as other beam configurations for similar accelerators using only the reference beam data. Electron dose calculations were performed with the reconstructed beams in a water phantom and compared with experimental data. An agreement of 1-2% / 1-2 mm was obtained, equivalent to the accuracy of full Monte Carlo accelerator simulation. The source model reduces accelerator simulation CPU time by a factor of 7500 relative to full Monte Carlo approaches. The developed model was then interfaced with DPM, a fast radiation transport Monte Carlo code for dose calculati...
Calculation of surface dose in rotational total skin electron irradiation
Pla, C.; Heese, R.; Pla, M.; Podgorsak, E.B.
1984-07-01
A single-field rotational total skin electron irradiation technique has recently been developed at the McGill University for treatment of skin malignancies. The dose received by a given surface point during rotation in a uniform large electron field depends on the radius of rotation of the surface point, on the local radius of curvature of the contour in the vicinity of the point of interest, and on the shadows cast by limbs (arms upon trunk or head and neck, and legs upon each other). A method for calculating the surface dose distribution on a patient is presented accounting for the various parameters affecting the dose. A series of measurements were performed with polystyrene and a humanoid phantom, and an excellent agreement between measured and calculated dose distributions was obtained.
Considerations of beta and electron transport in internal dose calculations
Bolch, W.E.; Poston, J.W. Sr.
1990-12-01
Ionizing radiation has broad uses in modern science and medicine. These uses often require the calculation of energy deposition in the irradiated media and, usually, the medium of interest is the human body. Energy deposition from radioactive sources within the human body and the effects of such deposition are considered in the field of internal dosimetry. In July of 1988, a three-year research project was initiated by the Nuclear Engineering Department at Texas A M University under the sponsorship of the US Department of Energy. The main thrust of the research was to consider, for the first time, the detailed spatial transport of electron and beta particles in the estimation of average organ doses under the Medical Internal Radiation Dose (MIRD) schema. At the present time (December of 1990), research activities are continuing within five areas. Several are new initiatives begun within the second or third year of the current contract period. They include: (1) development of small-scale dosimetry; (2) development of a differential volume phantom; (3) development of a dosimetric bone model; (4) assessment of the new ICRP lung model; and (5) studies into the mechanisms of DNA damage. A progress report is given for each of these tasks within the Comprehensive Report. In each case, preliminary results are very encouraging and plans for further research are detailed within this document.
Monte Carlo dose calculation improvements for low energy electron beams using eMC.
Fix, Michael K; Frei, Daniel; Volken, Werner; Neuenschwander, Hans; Born, Ernst J; Manser, Peter
2010-08-21
The electron Monte Carlo (eMC) dose calculation algorithm in Eclipse (Varian Medical Systems) is based on the macro MC method and is able to predict dose distributions for high energy electron beams with high accuracy. However, there are limitations for low energy electron beams. This work aims to improve the accuracy of the dose calculation using eMC for 4 and 6 MeV electron beams of Varian linear accelerators. Improvements implemented into the eMC include (1) improved determination of the initial electron energy spectrum by increased resolution of mono-energetic depth dose curves used during beam configuration; (2) inclusion of all the scrapers of the applicator in the beam model; (3) reduction of the maximum size of the sphere to be selected within the macro MC transport when the energy of the incident electron is below certain thresholds. The impact of these changes in eMC is investigated by comparing calculated dose distributions for 4 and 6 MeV electron beams at source to surface distance (SSD) of 100 and 110 cm with applicators ranging from 6 x 6 to 25 x 25 cm(2) of a Varian Clinac 2300C/D with the corresponding measurements. Dose differences between calculated and measured absolute depth dose curves are reduced from 6% to less than 1.5% for both energies and all applicators considered at SSD of 100 cm. Using the original eMC implementation, absolute dose profiles at depths of 1 cm, d(max) and R50 in water lead to dose differences of up to 8% for applicators larger than 15 x 15 cm(2) at SSD 100 cm. Those differences are now reduced to less than 2% for all dose profiles investigated when the improved version of eMC is used. At SSD of 110 cm the dose difference for the original eMC version is even more pronounced and can be larger than 10%. Those differences are reduced to within 2% or 2 mm with the improved version of eMC. In this work several enhancements were made in the eMC algorithm leading to significant improvements in the accuracy of the dose
Doucet, R.; Olivares, M.; DeBlois, F.; Podgorsak, E. B.; Kawrakow, I.; Seuntjens, J.
2003-08-01
Calculations of dose distributions in heterogeneous phantoms in clinical electron beams, carried out using the fast voxel Monte Carlo (MC) system XVMC and the conventional MC code EGSnrc, were compared with measurements. Irradiations were performed using the 9 MeV and 15 MeV beams from a Varian Clinac-18 accelerator with a 10 × 10 cm2 applicator and an SSD of 100 cm. Depth doses were measured with thermoluminescent dosimetry techniques (TLD 700) in phantoms consisting of slabs of Solid WaterTM (SW) and bone and slabs of SW and lung tissue-equivalent materials. Lateral profiles in water were measured using an electron diode at different depths behind one and two immersed aluminium rods. The accelerator was modelled using the EGS4/BEAM system and optimized phase-space files were used as input to the EGSnrc and the XVMC calculations. Also, for the XVMC, an experiment-based beam model was used. All measurements were corrected by the EGSnrc-calculated stopping power ratios. Overall, there is excellent agreement between the corrected experimental and the two MC dose distributions. Small remaining discrepancies may be due to the non-equivalence between physical and simulated tissue-equivalent materials and to detector fluence perturbation effect correction factors that were calculated for the 9 MeV beam at selected depths in the heterogeneous phantoms.
Doucet, R [Medical Physics Unit, McGill University, Montreal General Hospital, 1650 Ave Cedar, Montreal H3G 1A4 (Canada); Olivares, M [Medical Physics Unit, McGill University, Montreal General Hospital, 1650 Ave Cedar, Montreal H3G 1A4 (Canada); DeBlois, F [Medical Physics Unit, McGill University, Montreal General Hospital, 1650 Ave Cedar, Montreal H3G 1A4 (Canada); Podgorsak, E B [Medical Physics Unit, McGill University, Montreal General Hospital, 1650 Ave Cedar, Montreal H3G 1A4 (Canada); Kawrakow, I [National Research Council Canada, Ionizing Radiation Standards Group, Ottawa K1A 0R6, Canada (Canada); Seuntjens, J [Medical Physics Unit, McGill University, Montreal General Hospital, 1650 Ave Cedar, Montreal H3G 1A4 (Canada)
2003-08-07
Calculations of dose distributions in heterogeneous phantoms in clinical electron beams, carried out using the fast voxel Monte Carlo (MC) system XVMC and the conventional MC code EGSnrc, were compared with measurements. Irradiations were performed using the 9 MeV and 15 MeV beams from a Varian Clinac-18 accelerator with a 10 x 10 cm{sup 2} applicator and an SSD of 100 cm. Depth doses were measured with thermoluminescent dosimetry techniques (TLD 700) in phantoms consisting of slabs of Solid Water{sup TM} (SW) and bone and slabs of SW and lung tissue-equivalent materials. Lateral profiles in water were measured using an electron diode at different depths behind one and two immersed aluminium rods. The accelerator was modelled using the EGS4/BEAM system and optimized phase-space files were used as input to the EGSnrc and the XVMC calculations. Also, for the XVMC, an experiment-based beam model was used. All measurements were corrected by the EGSnrc-calculated stopping power ratios. Overall, there is excellent agreement between the corrected experimental and the two MC dose distributions. Small remaining discrepancies may be due to the non-equivalence between physical and simulated tissue-equivalent materials and to detector fluence perturbation effect correction factors that were calculated for the 9 MeV beam at selected depths in the heterogeneous phantoms.
A deterministic partial differential equation model for dose calculation in electron radiotherapy
Duclous, R; Dubroca, B [CELIA and IMB Laboratories, Bordeaux University, 33405 Talence (France); Frank, M, E-mail: duclous@celia.u-bordeaux1.f, E-mail: dubroca@celia.u-bordeaux1.f, E-mail: frank@mathcces.rwth-aachen.d [Department of Mathematics and Center for Computational Engineering Science, RWTH Aachen University, Schinkelstr. 2, 52062 Aachen (Germany)
2010-07-07
High-energy ionizing radiation is a prominent modality for the treatment of many cancers. The approaches to electron dose calculation can be categorized into semi-empirical models (e.g. Fermi-Eyges, convolution-superposition) and probabilistic methods (e.g. Monte Carlo). A third approach to dose calculation has only recently attracted attention in the medical physics community. This approach is based on the deterministic kinetic equations of radiative transfer. We derive a macroscopic partial differential equation model for electron transport in tissue. This model involves an angular closure in the phase space. It is exact for the free streaming and the isotropic regime. We solve it numerically by a newly developed HLLC scheme based on Berthon et al (2007 J. Sci. Comput. 31 347-89) that exactly preserves the key properties of the analytical solution on the discrete level. We discuss several test cases taken from the medical physics literature. A test case with an academic Henyey-Greenstein scattering kernel is considered. We compare our model to a benchmark discrete ordinate solution. A simplified model of electron interactions with tissue is employed to compute the dose of an electron beam in a water phantom, and a case of irradiation of the vertebral column. Here our model is compared to the PENELOPE Monte Carlo code. In the academic example, the fluences computed with the new model and a benchmark result differ by less than 1%. The depths at half maximum differ by less than 0.6%. In the two comparisons with Monte Carlo, our model gives qualitatively reasonable dose distributions. Due to the crude interaction model, these so far do not have the accuracy needed in clinical practice. However, the new model has a computational cost that is less than one-tenth of the cost of a Monte Carlo simulation. In addition, simulations can be set up in a similar way as a Monte Carlo simulation. If more detailed effects such as coupled electron-photon transport, bremsstrahlung
A deterministic partial differential equation model for dose calculation in electron radiotherapy
Duclous, R.; Dubroca, B.; Frank, M.
2010-07-01
High-energy ionizing radiation is a prominent modality for the treatment of many cancers. The approaches to electron dose calculation can be categorized into semi-empirical models (e.g. Fermi-Eyges, convolution-superposition) and probabilistic methods (e.g. Monte Carlo). A third approach to dose calculation has only recently attracted attention in the medical physics community. This approach is based on the deterministic kinetic equations of radiative transfer. We derive a macroscopic partial differential equation model for electron transport in tissue. This model involves an angular closure in the phase space. It is exact for the free streaming and the isotropic regime. We solve it numerically by a newly developed HLLC scheme based on Berthon et al (2007 J. Sci. Comput. 31 347-89) that exactly preserves the key properties of the analytical solution on the discrete level. We discuss several test cases taken from the medical physics literature. A test case with an academic Henyey-Greenstein scattering kernel is considered. We compare our model to a benchmark discrete ordinate solution. A simplified model of electron interactions with tissue is employed to compute the dose of an electron beam in a water phantom, and a case of irradiation of the vertebral column. Here our model is compared to the PENELOPE Monte Carlo code. In the academic example, the fluences computed with the new model and a benchmark result differ by less than 1%. The depths at half maximum differ by less than 0.6%. In the two comparisons with Monte Carlo, our model gives qualitatively reasonable dose distributions. Due to the crude interaction model, these so far do not have the accuracy needed in clinical practice. However, the new model has a computational cost that is less than one-tenth of the cost of a Monte Carlo simulation. In addition, simulations can be set up in a similar way as a Monte Carlo simulation. If more detailed effects such as coupled electron-photon transport, bremsstrahlung
Deterministic Partial Differential Equation Model for Dose Calculation in Electron Radiotherapy
Duclous, Roland; Frank, Martin
2009-01-01
Treatment with high energy ionizing radiation is one of the main methods in modern cancer therapy that is in clinical use. During the last decades, two main approaches to dose calculation were used, Monte Carlo simulations and semi-empirical models based on Fermi-Eyges theory. A third way to dose calculation has only recently attracted attention in the medical physics community. This approach is based on the deterministic kinetic equations of radiative transfer. Starting from these, we derive a macroscopic partial differential equation model for electron transport in tissue. This model involves an angular closure in the phase space. It is exact for the free-streaming and the isotropic regime. We solve it numerically by a newly developed HLLC scheme based on [BerCharDub], that exactly preserves key properties of the analytical solution on the discrete level. Several numerical results for test cases from the medical physics literature are presented.
Fix, Michael K; Cygler, Joanna; Frei, Daniel; Volken, Werner; Neuenschwander, Hans; Born, Ernst J; Manser, Peter
2013-05-07
The electron Monte Carlo (eMC) dose calculation algorithm available in the Eclipse treatment planning system (Varian Medical Systems) is based on the macro MC method and uses a beam model applicable to Varian linear accelerators. This leads to limitations in accuracy if eMC is applied to non-Varian machines. In this work eMC is generalized to also allow accurate dose calculations for electron beams from Elekta and Siemens accelerators. First, changes made in the previous study to use eMC for low electron beam energies of Varian accelerators are applied. Then, a generalized beam model is developed using a main electron source and a main photon source representing electrons and photons from the scattering foil, respectively, an edge source of electrons, a transmission source of photons and a line source of electrons and photons representing the particles from the scrapers or inserts and head scatter radiation. Regarding the macro MC dose calculation algorithm, the transport code of the secondary particles is improved. The macro MC dose calculations are validated with corresponding dose calculations using EGSnrc in homogeneous and inhomogeneous phantoms. The validation of the generalized eMC is carried out by comparing calculated and measured dose distributions in water for Varian, Elekta and Siemens machines for a variety of beam energies, applicator sizes and SSDs. The comparisons are performed in units of cGy per MU. Overall, a general agreement between calculated and measured dose distributions for all machine types and all combinations of parameters investigated is found to be within 2% or 2 mm. The results of the dose comparisons suggest that the generalized eMC is now suitable to calculate dose distributions for Varian, Elekta and Siemens linear accelerators with sufficient accuracy in the range of the investigated combinations of beam energies, applicator sizes and SSDs.
Fix, Michael K.; Cygler, Joanna; Frei, Daniel; Volken, Werner; Neuenschwander, Hans; Born, Ernst J.; Manser, Peter
2013-05-01
The electron Monte Carlo (eMC) dose calculation algorithm available in the Eclipse treatment planning system (Varian Medical Systems) is based on the macro MC method and uses a beam model applicable to Varian linear accelerators. This leads to limitations in accuracy if eMC is applied to non-Varian machines. In this work eMC is generalized to also allow accurate dose calculations for electron beams from Elekta and Siemens accelerators. First, changes made in the previous study to use eMC for low electron beam energies of Varian accelerators are applied. Then, a generalized beam model is developed using a main electron source and a main photon source representing electrons and photons from the scattering foil, respectively, an edge source of electrons, a transmission source of photons and a line source of electrons and photons representing the particles from the scrapers or inserts and head scatter radiation. Regarding the macro MC dose calculation algorithm, the transport code of the secondary particles is improved. The macro MC dose calculations are validated with corresponding dose calculations using EGSnrc in homogeneous and inhomogeneous phantoms. The validation of the generalized eMC is carried out by comparing calculated and measured dose distributions in water for Varian, Elekta and Siemens machines for a variety of beam energies, applicator sizes and SSDs. The comparisons are performed in units of cGy per MU. Overall, a general agreement between calculated and measured dose distributions for all machine types and all combinations of parameters investigated is found to be within 2% or 2 mm. The results of the dose comparisons suggest that the generalized eMC is now suitable to calculate dose distributions for Varian, Elekta and Siemens linear accelerators with sufficient accuracy in the range of the investigated combinations of beam energies, applicator sizes and SSDs.
Carver, Robert L.; Hogstrom, Kenneth R. [Mary Bird Perkins Cancer Center, 4950 Essen Lane, Baton Rouge, Louisiana 70809 (United States); Department of Physics and Astronomy, Louisiana State University, Baton Rouge, Louisiana 70803 (United States); Chu, Connel; Fields, Robert S. [Mary Bird Perkins Cancer Center, 4950 Essen Lane, Baton Rouge, Louisiana 70809 (United States); Sprunger, Conrad P. [Department of Physics and Astronomy, Louisiana State University, Baton Rouge, Louisiana 70803 (United States)
2013-07-15
Purpose: The purpose of this study was to document the improved accuracy of the pencil beam redefinition algorithm (PBRA) compared to the pencil beam algorithm (PBA) for bolus electron conformal therapy using cylindrical patient phantoms based on patient computed tomography (CT) scans of retromolar trigone and nose cancer.Methods: PBRA and PBA electron dose calculations were compared with measured dose in retromolar trigone and nose phantoms both with and without bolus. For the bolus treatment plans, a radiation oncologist outlined a planning target volume (PTV) on the central axis slice of the CT scan for each phantom. A bolus was designed using the planning.decimal{sup Registered-Sign} (p.d) software (.decimal, Inc., Sanford, FL) to conform the 90% dose line to the distal surface of the PTV. Dose measurements were taken with thermoluminescent dosimeters placed into predrilled holes. The Pinnacle{sup 3} (Philips Healthcare, Andover, MD) treatment planning system was used to calculate PBA dose distributions. The PBRA dose distributions were calculated with an in-house C++ program. In order to accurately account for the phantom materials a table correlating CT number to relative electron stopping and scattering powers was compiled and used for both PBA and PBRA dose calculations. Accuracy was determined by comparing differences in measured and calculated dose, as well as distance to agreement for each measurement point.Results: The measured doses had an average precision of 0.9%. For the retromolar trigone phantom, the PBRA dose calculations had an average {+-}1{sigma} dose difference (calculated - measured) of -0.65%{+-} 1.62% without the bolus and -0.20%{+-} 1.54% with the bolus. The PBA dose calculation had an average dose difference of 0.19%{+-} 3.27% without the bolus and -0.05%{+-} 3.14% with the bolus. For the nose phantom, the PBRA dose calculations had an average dose difference of 0.50%{+-} 3.06% without bolus and -0.18%{+-} 1.22% with the bolus. The PBA
An OpenCL-based Monte Carlo dose calculation engine (oclMC) for coupled photon-electron transport
Tian, Zhen; Folkerts, Michael; Qin, Nan; Jiang, Steve B; Jia, Xun
2015-01-01
Monte Carlo (MC) method has been recognized the most accurate dose calculation method for radiotherapy. However, its extremely long computation time impedes clinical applications. Recently, a lot of efforts have been made to realize fast MC dose calculation on GPUs. Nonetheless, most of the GPU-based MC dose engines were developed in NVidia CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a fast cross-platform MC dose engine oclMC using OpenCL environment for external beam photon and electron radiotherapy in MeV energy range. Coupled photon-electron MC simulation was implemented with analogue simulations for photon transports and a Class II condensed history scheme for electron transports. To test the accuracy and efficiency of our dose engine oclMC, we compared dose calculation results of oclMC and gDPM, our previously developed GPU-based MC code, for a 15 MeV electron ...
Habib, B.; Poumarede, B.; Tola, F.; Barthe, J. [CEA, LIST, Dept Technol Capteur et Signal, F-91191 Gif Sur Yvette, (France)
2010-07-01
The aim of the present study is to demonstrate the potential of accelerated dose calculations, using the fast Monte Carlo (MC) code referred to as PENFAST, rather than the conventional MC code PENELOPE, without losing accuracy in the computed dose. For this purpose, experimental measurements of dose distributions in homogeneous and inhomogeneous phantoms were compared with simulated results using both PENELOPE and PENFAST. The simulations and experiments were performed using a Saturne 43 linac operated at 12 MV (photons), and at 18 MeV (electrons). Pre-calculated phase space files (PSFs) were used as input data to both the PENELOPE and PENFAST dose simulations. Since depth-dose and dose profile comparisons between simulations and measurements in water were found to be in good agreement (within {+-} 1% to 1 mm), the PSF calculation is considered to have been validated. In addition, measured dose distributions were compared to simulated results in a set of clinically relevant, inhomogeneous phantoms, consisting of lung and bone heterogeneities in a water tank. In general, the PENFAST results agree to within a 1% to 1 mm difference with those produced by PENELOPE, and to within a 2% to 2 mm difference with measured values. Our study thus provides a pre-clinical validation of the PENFAST code. It also demonstrates that PENFAST provides accurate results for both photon and electron beams, equivalent to those obtained with PENELOPE. CPU time comparisons between both MC codes show that PENFAST is generally about 9-21 times faster than PENELOPE. (authors)
Accuracy of the phase space evolution dose calculation model for clinical 25 MeV electron beams
Korevaar, Erik W. [Daniel den Hoed Cancer Center, University Hospital Rotterdam, PO Box 5201, 3008 AE Rotterdam (Netherlands). E-mail: korevaar at kfih.azr.nl; Akhiat, Abdelhafid; Heijmen, Ben J.M. [Daniel den Hoed Cancer Center, University Hospital Rotterdam, PO Box 5201, 3008 AE Rotterdam (Netherlands); Huizenga, Henk [Joint Center for Radiation Oncology Arnhem-Nijmegen, University Medical Center Nijmegen, PO Box 9101, 6500 HB Nijmegen (Netherlands)
2000-10-01
The phase space evolution (PSE) model is a dose calculation model for electron beams in radiation oncology developed with the aim of a higher accuracy than the commonly used pencil beam (PB) models and with shorter calculation times than needed for Monte Carlo (MC) calculations. In this paper the accuracy of the PSE model has been investigated for 25 MeV electron beams of a MM50 racetrack microtron (Scanditronix Medical AB, Sweden) and compared with the results of a PB model. Measurements have been performed for tests like non-standard SSD, irregularly shaped fields, oblique incidence and in phantoms with heterogeneities of air, bone and lung. MC calculations have been performed as well, to reveal possible errors in the measurements and/or possible inaccuracies in the interaction data used for the bone and lung substitute materials. Results show a good agreement between PSE calculated dose distributions and measurements. For all points the differences - in absolute dose - were generally well within 3% and 3 mm. However, the PSE model was found to be less accurate in large regions of low-density material and errors of up to 6% were found for the lung phantom. Results of the PB model show larger deviations, with differences of up to 6% and 6 mm and of up to 10% for the lung phantom; at shortened SSDs the dose was overestimated by up to 6%. The agreement between MC calculations and measurement was good. For the bone and the lung phantom maximum deviations of 4% and 3% were found, caused by uncertainties about the actual interaction data. In conclusion, using the phase space evolution model, absolute 3D dose distributions of 25 MeV electron beams can be calculated with sufficient accuracy in most cases. The accuracy is significantly better than for a pencil beam model. In regions of lung tissue, a Monte Carlo model yields more accurate results than the current implementation of the PSE model. (author)
Accuracy of the phase space evolution dose calculation model for clinical 25 MeV electron beams
Korevaar, Erik W.; Akhiat, Abdelhafid; Heijmen, Ben J. M.; Huizenga, Henk
2000-10-01
The phase space evolution (PSE) model is a dose calculation model for electron beams in radiation oncology developed with the aim of a higher accuracy than the commonly used pencil beam (PB) models and with shorter calculation times than needed for Monte Carlo (MC) calculations. In this paper the accuracy of the PSE model has been investigated for 25 MeV electron beams of a MM50 racetrack microtron (Scanditronix Medical AB, Sweden) and compared with the results of a PB model. Measurements have been performed for tests like non-standard SSD, irregularly shaped fields, oblique incidence and in phantoms with heterogeneities of air, bone and lung. MC calculations have been performed as well, to reveal possible errors in the measurements and/or possible inaccuracies in the interaction data used for the bone and lung substitute materials. Results show a good agreement between PSE calculated dose distributions and measurements. For all points the differences - in absolute dose - were generally well within 3% and 3 mm. However, the PSE model was found to be less accurate in large regions of low-density material and errors of up to 6% were found for the lung phantom. Results of the PB model show larger deviations, with differences of up to 6% and 6 mm and of up to 10% for the lung phantom; at shortened SSDs the dose was overestimated by up to 6%. The agreement between MC calculations and measurement was good. For the bone and the lung phantom maximum deviations of 4% and 3% were found, caused by uncertainties about the actual interaction data. In conclusion, using the phase space evolution model, absolute 3D dose distributions of 25 MeV electron beams can be calculated with sufficient accuracy in most cases. The accuracy is significantly better than for a pencil beam model. In regions of lung tissue, a Monte Carlo model yields more accurate results than the current implementation of the PSE model.
Horowitz, Y.S. [Ben Gurion Univ. of the Negev, Beersheva (Israel); Hirning, C.R. [Ontario Hydro, Whitby (Canada); Yuen, P.; Wong, P. [Chalk River Labs., Ontario (Canada)
1994-10-01
Monte Carlo calculations have been carried out for monoenergetic electrons from 0.1 to 4 MeV irradiating LiF chips in both perpendicular and isotropic geometry. This enabled the calculation of skin dose correction factors (beta factors) for typical beta energy spectra as measured with a beta-ray spectrometer at CANDU nuclear generating stations. The correction factors were estimated by averaging the depth dose distributions for the monoenergetic electrons over the experimentally measured beta-ray spectra. The calculations illustrate the large uncertainty in beta factors arising from the unknown angular distribution of the beta-ray radiation field and uncertainties in the shape of the beta-ray spectra below 500 keV. 28 refs., 8 figs., 2 tabs.
Considerations of beta and electron transport in internal dose calculations. Progress report
Bolch, W.E.
1994-11-01
The goal of this particular task is to consider, for the first time, the explicit transport of beta particles and photon-generated electrons in the series of six phantoms developed by Cristy and Eckerman (1987) at the Oak Ridge National Laboratory. In their report, ORNL/TM-8381, specific absorbed fractions of energy are reported for phantoms representing the newborn (3.4 kg), the one-year-old (9.8 kg), the five-year-old (19 kg), the ten-year-old (32 kg), the fifteen-year-old/adult female (55-58 kg), and the adult male (70 kg). Radiation transport calculations were performed with the Monte Carlo code ALGAMP which allows photon transport only. In subsequent calculations of radionuclide S values as is done in the MIRDOSE2 computer program, electron absorbed fractions are thus considered to be either unity or zero depending upon whether the source region does or does not equal the target region, respectively.
Mairani, A; Valente, M; Battistoni, G; Botta, F; Pedroli, G; Ferrari, A; Cremonesi, M; Di Dia, A; Ferrari, M; Fasso, A
2011-01-01
Purpose: The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, FLUKA Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, FLUKA has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. Methods: FLUKA DPKS have been calculated in both water and compact bone for monoenergetic electrons (10-3 MeV) and for beta emitting isotopes commonly used for therapy ((89)Sr, (90)Y, (131)I, (153)Sm, (177)Lu, (186)Re, and (188)Re). Point isotropic...
Behrens, R
2013-07-01
In this work, conversion coefficients from electron fluence to absorbed dose to the eye lens were calculated using Monte Carlo simulations based on a detailed stylised eye model and a very simple but whole body phantom. These data supersede and complement data published earlier based on the simulation of only a single stylised eye. The new data differ from the old ones by not more than 3, 4, 7 and 16 % for angles of radiation incidence of α=0°, 15°, 30° and 45°, respectively, due to the inclusion of the whole body phantom. The data presented in the present work also complement those of a recent report of the International Commission on Radiological Protection (ICRP) (ICRP Publication 116), where conversion coefficients from electron fluence to absorbed dose to the lens of the eye are shown for solely 0°, 180° and isotropic radiation incidence (but for a much broader range of energies). In this article, values are provided for angles of incidence of 0° up to 180° in steps of 15° and for rotational geometry; no systematic deviation was observed from the values given in ICRP Publication 116 for 0° (based on the application of a bare eye) and 180° (based on the application of a voxel whole body phantom). Data are given for monoenergetic electrons from 0.1 up to 10 MeV and for a broad parallel beam geometry in vacuum.
Botta, F; Mairani, A; Battistoni, G; Cremonesi, M; Di Dia, A; Fassò, A; Ferrari, A; Ferrari, M; Paganelli, G; Pedroli, G; Valente, M
2011-07-01
The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, FLUKA Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, FLUKA has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. FLUKA DPKS have been calculated in both water and compact bone for monoenergetic electrons (10-3 MeV) and for beta emitting isotopes commonly used for therapy (89Sr, 90Y, 131I 153Sm, 177Lu, 186Re, and 188Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. FLUKA outcomes have been compared to PENELOPE v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (ETRAN, GEANT4, MCNPX) has been done. Maximum percentage differences within 0.8.RCSDA and 0.9.RCSDA for monoenergetic electrons (RCSDA being the continuous slowing down approximation range) and within 0.8.X90 and 0.9.X90 for isotopes (X90 being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9.RCSDA and 0.9.X90 for electrons and isotopes, respectively. Concerning monoenergetic electrons, within 0.8.RCSDA (where 90%-97% of the particle energy is deposed), FLUKA and PENELOPE agree mostly within 7%, except for 10 and 20 keV electrons (12% in water, 8.3% in bone). The
Botta, F; Di Dia, A; Pedroli, G; Mairani, A; Battistoni, G; Fasso, A; Ferrari, A; Ferrari, M; Paganelli, G
2011-06-01
The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, fluka Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, fluka has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one.Methods: fluka DPKs have been calculated in both water and compact bone for monoenergetic electrons (10–3 MeV) and for beta emitting isotopes commonly used for therapy (89Sr, 90Y, 131I, 153Sm, 177Lu, 186Re, and 188Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. fluka outcomes have been compared to penelope v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (etran, geant4, mcnpx) has been done. Maximum percentage differences within 0.8·RCSDA and 0.9·RCSDA for monoenergetic electrons (RCSDA being the continuous slowing down approximation range) and within 0.8·X90 and 0.9·X90 for isotopes (X90 being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9·RCSDA and 0.9·X90 for electrons and isotopes, respectively.Results: Concerning monoenergetic electrons, within 0.8·RCSDA (where 90%–97% of the particle energy is deposed), fluka and penelope agree mostly within 7%, except for 10 and 20 keV electrons (12% in water, 8
Botta, F.; Mairani, A.; Battistoni, G.; Cremonesi, M.; Di Dia, A.; Fasso, A.; Ferrari, A.; Ferrari, M.; Paganelli, G.; Pedroli, G.; Valente, M. [Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); Istituto Nazionale di Fisica Nucleare (I.N.F.N.), Via Celoria 16, 20133 Milan (Italy); Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); Jefferson Lab, 12000 Jefferson Avenue, Newport News, Virginia 23606 (United States); CERN, 1211 Geneva 23 (Switzerland); Medical Physics Department, European Institute of Oncology, Milan (Italy); Nuclear Medicine Department, European Institute of Oncology, Via Ripamonti 435, 2014 Milan (Italy); Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); FaMAF, Universidad Nacional de Cordoba and CONICET, Cordoba, Argentina C.P. 5000 (Argentina)
2011-07-15
Purpose: The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, fluka Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, fluka has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. Methods: fluka DPKs have been calculated in both water and compact bone for monoenergetic electrons (10{sup -3} MeV) and for beta emitting isotopes commonly used for therapy ({sup 89}Sr, {sup 90}Y, {sup 131}I, {sup 153}Sm, {sup 177}Lu, {sup 186}Re, and {sup 188}Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. fluka outcomes have been compared to penelope v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (etran, geant4, mcnpx) has been done. Maximum percentage differences within 0.8{center_dot}R{sub CSDA} and 0.9{center_dot}R{sub CSDA} for monoenergetic electrons (R{sub CSDA} being the continuous slowing down approximation range) and within 0.8{center_dot}X{sub 90} and 0.9{center_dot}X{sub 90} for isotopes (X{sub 90} being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9{center_dot}R{sub CSDA} and 0.9{center_dot}X{sub 90} for electrons and isotopes, respectively. Results: Concerning monoenergetic electrons
Jin, L; Eldib, A; Li, J; Price, R; Ma, C [Fox Chase Cancer Center, Philadelphia, PA (United States)
2015-06-15
Purpose: Uneven nose surfaces and air cavities underneath and the use of bolus present complexity and dose uncertainty when using a single electron energy beam to plan treatments of nose skin with a pencil beam-based planning system. This work demonstrates more accurate dose calculation and more optimal planning using energy and intensity modulated electron radiotherapy (MERT) delivered with a pMLC. Methods: An in-house developed Monte Carlo (MC)-based dose calculation/optimization planning system was employed for treatment planning. Phase space data (6, 9, 12 and 15 MeV) were used as an input source for MC dose calculations for the linac. To reduce the scatter-caused penumbra, a short SSD (61 cm) was used. Our previous work demonstrates good agreement in percentage depth dose and off-axis dose between calculations and film measurement for various field sizes. A MERT plan was generated for treating the nose skin using a patient geometry and a dose volume histogram (DVH) was obtained. The work also shows the comparison of 2D dose distributions between a clinically used conventional single electron energy plan and the MERT plan. Results: The MERT plan resulted in improved target dose coverage as compared to the conventional plan, which demonstrated a target dose deficit at the field edge. The conventional plan showed higher dose normal tissue irradiation underneath the nose skin while the MERT plan resulted in improved conformity and thus reduces normal tissue dose. Conclusion: This preliminary work illustrates that MC-based MERT planning is a promising technique in treating nose skin, not only providing more accurate dose calculation, but also offering an improved target dose coverage and conformity. In addition, this technique may eliminate the necessity of bolus, which often produces dose delivery uncertainty due to the air gaps that may exist between the bolus and skin.
Electronics Environmental Benefits Calculator
U.S. Environmental Protection Agency — The Electronics Environmental Benefits Calculator (EEBC) was developed to assist organizations in estimating the environmental benefits of greening their purchase,...
Su, Lin; Du, Xining; Liu, Tianyu; Xu, X. George
2014-06-01
An electron-photon coupled Monte Carlo code ARCHER - Accelerated Radiation-transport Computations in Heterogeneous EnviRonments - is being developed at Rensselaer Polytechnic Institute as a software testbed for emerging heterogeneous high performance computers that utilize accelerators such as GPUs. This paper presents the preliminary code development and the testing involving radiation dose related problems. In particular, the paper discusses the electron transport simulations using the class-II condensed history method. The considered electron energy ranges from a few hundreds of keV to 30 MeV. For photon part, photoelectric effect, Compton scattering and pair production were modeled. Voxelized geometry was supported. A serial CPU code was first written in C++. The code was then transplanted to the GPU using the CUDA C 5.0 standards. The hardware involved a desktop PC with an Intel Xeon X5660 CPU and six NVIDIA Tesla™ M2090 GPUs. The code was tested for a case of 20 MeV electron beam incident perpendicularly on a water-aluminum-water phantom. The depth and later dose profiles were found to agree with results obtained from well tested MC codes. Using six GPU cards, 6x106 electron histories were simulated within 2 seconds. In comparison, the same case running the EGSnrc and MCNPX codes required 1645 seconds and 9213 seconds, respectively. On-going work continues to test the code for different medical applications such as radiotherapy and brachytherapy.
Andreo, Pedro
1983-06-01
A computer program has been developed to facilitate the calculation of the absorbed dose in photon and electron beams from measurements with an ionization chamber in a phantom. The generalized Bragg-Gray theory, introduced in the latest recommendations of the Nordic Association of Clinical Physics (NACP), is used throughout the code, including more updated parameter values than those included in the NACP protocol. The calibration factor of the ionization chamber in units of absorbed dose in the air of the cavity can be derived for most of the chambers available today by using experimental data or fitted relations to Monte Carlo results.
Mullins, J; Duan, X; Kruse, J; Herman, M [Mayo Clinic, Rochester, MN (United States); Bues, M [Mayo Clinic Arizona, Phoenix, AZ (United States)
2014-06-01
Purpose: To determine the suitability of dual-energy CT (DECT) to calculate relative electron density (RED) of tissues for accurate proton therapy dose calculation. Methods: DECT images of RED tissue surrogates were acquired at 80 and 140 kVp. Samples (RED=0.19−2.41) were imaged in a water-equivalent phantom in a variety of configurations. REDs were calculated using the DECT numbers and inputs of the high and low energy spectral weightings. DECT-derived RED was compared between geometric configurations and for variations in the spectral inputs to assess the sensitivity of RED accuracy versus expected values. Results: RED accuracy was dependent on accurate spectral input influenced by phantom thickness and radius from the phantom center. Material samples located at the center of the phantom generally showed the best agreement to reference RED values, but only when attenuation of the surrounding phantom thickness was accounted for in the calculation spectra. Calculated RED changed by up to 10% for some materials when the sample was located at an 11 cm radius from the phantom center. Calculated REDs under the best conditions still differed from reference values by up to 5% in bone and 14% in lung. Conclusion: DECT has previously been used to differentiate tissue types based on RED and Z for binary tissue-type segmentation. To improve upon the current standard of empirical conversion of CT number to RED for treatment planning dose calculation, DECT methods must be able to calculate RED to better than 3% accuracy throughout the image. The DECT method is sensitive to the accuracy of spectral inputs used for calculation, as well as to spatial position in the anatomy. Effort to address adjustments to the spectral calculation inputs based on position and phantom attenuation will be required before DECT-determined RED can achieve a consistent level of accuracy for application in dose calculation.
Wieslander, Elinore; Knöös, Tommy
2007-02-01
The introduction of Monte Carlo (MC) techniques for treatment planning and also for verification purposes will have considerable impact on the radiation therapy planning process. The aim of this work was to use a virtual accelerator to study the performance of a MC-based electron dose calculation algorithm, implemented in a commercial treatment planning system. The performance in phantoms containing air and bone as well as in patient-specific geometries (thorax wall, nose, parotid gland and spinal cord) has been studied. The agreement between the virtual accelerator and the MC dose calculation algorithm is generally very good. A gamma-evaluation with criteria of 0.03 Gy/3 mm (per Gy at the depth of maximum dose) shows that, even for the worst cases, only a small volume of about 1.5% has gamma>1.0. In the worst case, with the 0.02 Gy/2 mm criteria, about 92% of the volume receiving more than 0.85 Gy per 100 monitor units (MU) has gamma-values <1.0. The corresponding value for the volume receiving more than 0.10 Gy/100 MU is about 98%. For the 18 MeV spinal-cord case, where a 6 x 20 cm2 insert is used, the TPS underestimates the dose outside the primary field due to inadequate modelling of the insert. The possibility of dose calculations in typical patient cases makes the virtual accelerator a powerful tool for validation and evaluation of dose calculation algorithms present in treatment planning systems.
Calculational Tool for Skin Contamination Dose Assessment
Hill, R L
2002-01-01
Spreadsheet calculational tool was developed to automate the calculations preformed for dose assessment of skin contamination. This document reports on the design and testing of the spreadsheet calculational tool.
Entrance surface dose according to dose calculation: Head and wrist
Sung, Ho Jin [Dept. Radiology, Chonnam National University Hospital, Gwangju (Korea, Republic of); Han, Jae Bok; Song, Jong Nam; Choi, Nam Gil [Dept. of Radiological Science, Dongshin University, Naju (Korea, Republic of)
2016-09-15
This study were compared with the direct measurement and indirect dose methods through various dose calculation in head and wrist. And, the modified equation was proposed considering equipment type, setting conditions, tube voltage, inherent filter, added filter and its accompanied back scatter factor. As a result, it decreased the error of the direct measurement than the existing dose calculation. Accordingly, diagnostic radiography patient dose comparison would become easier and radiographic exposure control and evaluation will become more efficient. The study findings are expected to be useful in patients' effective dose rate evaluation and dose reduction.
Hindie, Elif; Moretti, Jean-Luc [Hopital Saint-Louis, Service de Medecine Nucleaire, Paris (France)]|[Universite Paris 7, Imagerie Moleculaire Diagnostique et Ciblage Therapeutique, Paris (France); Champion, Christophe [Universite Paul Verlaine, Laboratoire de Physique Moleculaire et des Collisions, Metz Institut de Physique, Metz (France); Zanotti-Fregonara, Paolo; Ravasi, Laura [Commissariat a l' Energie Atomique, DSV/I2BM/SHFJ/LIME, Orsay (France); Rubello, Domenico [Instituto Oncologico Veneto (IOV) - IRCCS, Department of Nuclear Medicine - PET Centre, Rovigo (Italy); Colas-Linhart, Nicole [Faculte de Medecine Xavier Bichat, Laboratoire de Biophysique, Paris (France)
2009-01-15
We used the Monte Carlo code ''CELLDOSE'' to assess the dose received by specific target cells from electron emissions in a complex environment. {sup 131}I in a simulated thyroid was used as a model. Thyroid follicles were represented by 170{mu}m diameter spherical units made of a lumen of 150{mu}m diameter containing colloidal matter and a peripheral layer of 10{mu}m thick thyroid cells. Neighbouring follicles are 4{mu}m apart. {sup 131}I was assumed to be homogeneously distributed in the lumen and absent in cells. We firstly assessed electron dose distribution in a single follicle. Then, we expanded the simulation by progressively adding neighbouring layers of follicles, so to reassess the electron dose to this single follicle implemented with the contribution of the added layers. Electron dose gradient around a point source showed that the {sup 131}I electron dose is close to zero after 2,100{mu}m. Therefore, we studied all contributions to the central follicle deriving from follicles within 12 orders of neighbourhood (15,624 follicles surrounding the central follicle). The dose to colloid of the single follicle was twice as high as the dose to thyroid cells. Even when all neighbours were taken into account, the dose in the central follicle remained heterogeneous. For a 1-Gy average dose to tissue, the dose to colloidal matter was 1.168 Gy, the dose to thyroid cells was 0.982 Gy, and the dose to the inter-follicular tissue was 0.895 Gy. Analysis of the different contributions to thyroid cell dose showed that 17.3% of the dose derived from the colloidal matter of their own follicle, while the remaining 82.7% was delivered by the surrounding follicles. On the basis of these data, it is shown that when different follicles contain different concentrations of {sup 131}I, the impact in terms of cell dose heterogeneity can be important. By means of {sup 131}I in the thyroid as a theoretical model, we showed how a Monte Carlo code can be used to map
Superficial dose evaluation of four dose calculation algorithms
Cao, Ying; Yang, Xiaoyu; Yang, Zhen; Qiu, Xiaoping; Lv, Zhiping; Lei, Mingjun; Liu, Gui; Zhang, Zijian; Hu, Yongmei
2017-08-01
Accurate superficial dose calculation is of major importance because of the skin toxicity in radiotherapy, especially within the initial 2 mm depth being considered more clinically relevant. The aim of this study is to evaluate superficial dose calculation accuracy of four commonly used algorithms in commercially available treatment planning systems (TPS) by Monte Carlo (MC) simulation and film measurements. The superficial dose in a simple geometrical phantom with size of 30 cm×30 cm×30 cm was calculated by PBC (Pencil Beam Convolution), AAA (Analytical Anisotropic Algorithm), AXB (Acuros XB) in Eclipse system and CCC (Collapsed Cone Convolution) in Raystation system under the conditions of source to surface distance (SSD) of 100 cm and field size (FS) of 10×10 cm2. EGSnrc (BEAMnrc/DOSXYZnrc) program was performed to simulate the central axis dose distribution of Varian Trilogy accelerator, combined with measurements of superficial dose distribution by an extrapolation method of multilayer radiochromic films, to estimate the dose calculation accuracy of four algorithms in the superficial region which was recommended in detail by the ICRU (International Commission on Radiation Units and Measurement) and the ICRP (International Commission on Radiological Protection). In superficial region, good agreement was achieved between MC simulation and film extrapolation method, with the mean differences less than 1%, 2% and 5% for 0°, 30° and 60°, respectively. The relative skin dose errors were 0.84%, 1.88% and 3.90%; the mean dose discrepancies (0°, 30° and 60°) between each of four algorithms and MC simulation were (2.41±1.55%, 3.11±2.40%, and 1.53±1.05%), (3.09±3.00%, 3.10±3.01%, and 3.77±3.59%), (3.16±1.50%, 8.70±2.84%, and 18.20±4.10%) and (14.45±4.66%, 10.74±4.54%, and 3.34±3.26%) for AXB, CCC, AAA and PBC respectively. Monte Carlo simulation verified the feasibility of the superficial dose measurements by multilayer Gafchromic films. And the rank
Therapeutic dose from a pyroelectric electron accelerator.
Fullem, T Z; Fazel, K C; Geuther, J A; Danon, Y
2009-11-01
Simple heating of pyroelectric crystals has been used as the basis for compact sources of X rays, electrons, ions and neutrons. We report on the evaluation of the feasibility of using a portable pyroelectric electron accelerator to deliver a therapeutic dose to tissue. Such a device could be mass produced as a handheld, battery-powered instrument. Experiments were conducted with several crystal sizes in which the crystal was heated inside a vacuum chamber and the emitted electrons were allowed to penetrate a thin beryllium window into the surrounding air. A Faraday cup was used to count the number of electrons that exited the window. The energy of these electrons was determined by measuring the energy spectrum of the X rays that resulted from the electron interactions with the Faraday cup. Based on these measurements, the dose that this source could deliver to tissue was calculated using Monte Carlo calculations. It was found that 10(13) electrons with a peak energy of the order of 100 keV were emitted from the beryllium window and could deliver a dose of 1664 Gy to a 2-cm-diameter, 110-microm-deep region of tissue located 1.5 cm from the window with air between the window and the tissue. This dose level is high enough to consider this technology for medical applications in which shallow energy deposition is beneficial.
Optimization of Monte Carlo dose calculations: The interface problem
Soudentas, Edward
1998-05-01
High energy photon beams are widely used for radiation treatment of deep-seated tumors. The human body contains many types of interfaces between dissimilar materials that affect dose distribution in radiation therapy. Experimentally, significant radiation dose perturbations has been observed at such interfaces. The EGS4 Monte Carlo code was used to calculate dose perturbations at boundaries between dissimilar materials (such as bone/water) for 60Co and 6 MeV linear accelerator beams using a UNIX workstation. A simple test of the reliability of a random number generator was also developed. A systematic study of the adjustable parameters in EGS4 was performed in order to minimize calculational artifacts at boundaries. Calculations of dose perturbations at boundaries between different materials showed that there is a 12% increase in dose at water/bone interface, and a 44% increase in dose at water/copper interface. with the increase mainly due to electrons produced in water and backscattered from the high atomic number material. The dependence of the dose increase on the atomic number was also investigated. The clinically important case of using two parallel opposed beams for radiation therapy was investigated where increased doses at boundaries has been observed. The Monte Carlo calculations can provide accurate dosimetry data under conditions of electronic non-equilibrium at tissue interfaces.
Validation of dose calculation programmes for recycling
Menon, Shankar [Menon Consulting, Nykoeping (Sweden); Brun-Yaba, Christine [Inst. de Radioprotection et Securite Nucleaire (France); Yu, Charley; Cheng, Jing-Jy [Argonne National Laboratory, IL (United States). Environmental Assessment Div.; Bjerler, Jan [Studsvik Stensand, Nykoeping (Sweden); Williams, Alexander [Dept. of Energy (United States). Office of Environmental Management
2002-12-01
This report contains the results from an international project initiated by the SSI in 1999. The primary purpose of the project was to validate some of the computer codes that are used to estimate radiation doses due to the recycling of scrap metal. The secondary purpose of the validation project was to give a quantification of the level of conservatism in clearance levels based on these codes. Specifically, the computer codes RESRAD-RECYCLE and CERISE were used to calculate radiation doses to individuals during the processing of slightly contaminated material, mainly in Studsvik, Sweden. Calculated external doses were compared with measured data from different steps of the process. The comparison of calculations and measurements shows that the computer code calculations resulted in both overestimations and underestimations of the external doses for different recycling activities. The SSI draws the conclusion that the accuracy is within one order of magnitude when experienced modellers use their programmes to calculate external radiation doses for a recycling process involving material that is mainly contaminated with cobalt-60. No errors in the codes themselves were found. Instead, the inaccuracy seems to depend mainly on the choice of some modelling parameters related to the receptor (e.g., distance, time, etc.) and simplifications made to facilitate modelling with the codes (e.g., object geometry). Clearance levels are often based on studies on enveloping scenarios that are designed to cover all realistic exposure pathways. It is obvious that for most practical cases, this gives a margin to the individual dose constraint (in the order of 10 micro sievert per year within the EC). This may be accentuated by the use of conservative assumptions when modelling the enveloping scenarios. Since there can obviously be a fairly large inaccuracy in the calculations, it seems reasonable to consider some degree of conservatism when establishing clearance levels based on
Mandour, M A; Harder, D
1978-08-01
The algorithn proposed in 1976 to allow for the influences of tissue inhomogeneities in electron beam dose distributions has been improved. A factor adopted from Sternheimer corrects for differences in lateral displacement by multiple scattering within the inhomogeneity resp. the equivalent layer of the homogeneous medium. The algorithm has been tested by direct comparison with Monte Carlo computations for the example of cylindrical cavities in carbon at Eo=10 MeV and resulted in a very close approximation.
Superposition dose calculation in lung for 10MV photons.
Hoban, P W; Murray, D C; Metcalfe, P E; Round, W H
1990-06-01
Currently available radiotherapy treatment planning systems employ scatter function models such as ETAR and Batho dSAR for dose calculation. Errors using these models for high energy photon irradiation occur in and beyond lung tissue for small fields. For larger fields, central axis dose is correctly predicted but penumbral broadening in lung is underestimated. The major source of error is the assumption that lateral electronic equilibrium is always established. A superposition algorithm has been developed for 10MV photons which calculates the dose by convolving the TERMA (Total Energy Released per unit MAss by primary photons) with a dose spread array formed using the EGS4 Monte Carlo code. TERMA and dose spread arrays are both generated using a 10 component photon energy spectrum. Dose in inhomogeneous media is calculated using dose spread arrays generated for different density media and by scaling dose spread arrays according to density variations. This method ensures that electronic disequilibrium is modelled in situations where it exists. Superposition results in a lung phantom for a 5 x 5 cm field agree with EGS4 Monte Carlo results to within 2% for p = 0.20 gcm-3 and p = 0.30 gcm-3 lung. Profiles generated by superposition for a 10 x 10 cm field at mid-lung and compared with film measurements show that penumbral broadening in low density material is also correctly predicted.
Agriculture-related radiation dose calculations
Furr, J.M.; Mayberry, J.J.; Waite, D.A.
1987-10-01
Estimates of radiation dose to the public must be made at each stage in the identification and qualification process leading to siting a high-level nuclear waste repository. Specifically considering the ingestion pathway, this paper examines questions of reliability and adequacy of dose calculations in relation to five stages of data availability (geologic province, region, area, location, and mass balance) and three methods of calculation (population, population/food production, and food production driven). Calculations were done using the model PABLM with data for the Permian and Palo Duro Basins and the Deaf Smith County area. Extra effort expended in gathering agricultural data at succeeding environmental characterization levels does not appear justified, since dose estimates do not differ greatly; that effort would be better spent determining usage of food types that contribute most to the total dose; and that consumption rate and the air dispersion factor are critical to assessment of radiation dose via the ingestion pathway. 17 refs., 9 figs., 32 tabs.
Dose calculations for intakes of ore dust
O`Brien, R.S
1998-08-01
This report describes a methodology for calculating the committed effective dose for mixtures of radionuclides, such as those which occur in natural radioactive ores and dusts. The formulae are derived from first principles, with the use of reasonable assumptions concerning the nature and behaviour of the radionuclide mixtures. The calculations are complicated because these `ores` contain a range of particle sizes, have different degrees of solubility in blood and other body fluids, and also have different biokinetic clearance characteristics from the organs and tissues in the body. The naturally occurring radionuclides also tend to occur in series, i.e. one is produced by the radioactive decay of another `parent` radionuclide. The formulae derived here can be used, in conjunction with a model such as LUDEP, for calculating total dose resulting from inhalation and/or ingestion of a mixture of radionuclides, and also for deriving annual limits on intake and derived air concentrations for these mixtures. 15 refs., 14 tabs., 3 figs.
Electronics reliability calculation and design
Dummer, Geoffrey W A; Hiller, N
1966-01-01
Electronics Reliability-Calculation and Design provides an introduction to the fundamental concepts of reliability. The increasing complexity of electronic equipment has made problems in designing and manufacturing a reliable product more and more difficult. Specific techniques have been developed that enable designers to integrate reliability into their products, and reliability has become a science in its own right. The book begins with a discussion of basic mathematical and statistical concepts, including arithmetic mean, frequency distribution, median and mode, scatter or dispersion of mea
External dose-rate conversion factors for calculation of dose to the public
1988-07-01
This report presents a tabulation of dose-rate conversion factors for external exposure to photons and electrons emitted by radionuclides in the environment. This report was prepared in conjunction with criteria for limiting dose equivalents to members of the public from operations of the US Department of Energy (DOE). The dose-rate conversion factors are provided for use by the DOE and its contractors in performing calculations of external dose equivalents to members of the public. The dose-rate conversion factors for external exposure to photons and electrons presented in this report are based on a methodology developed at Oak Ridge National Laboratory. However, some adjustments of the previously documented methodology have been made in obtaining the dose-rate conversion factors in this report. 42 refs., 1 fig., 4 tabs.
Development of a computational methodology for internal dose calculations
Yoriyaz, H
2000-01-01
A new approach for calculating internal dose estimates was developed through the use of a more realistic computational model of the human body and a more precise tool for the radiation transport simulation. The present technique shows the capability to build a patient-specific phantom with tomography data (a voxel-based phantom) for the simulation of radiation transport and energy deposition using Monte Carlo methods such as in the MCNP-4B code. In order to utilize the segmented human anatomy as a computational model for the simulation of radiation transport, an interface program, SCMS, was developed to build the geometric configurations for the phantom through the use of tomographic images. This procedure allows to calculate not only average dose values but also spatial distribution of dose in regions of interest. With the present methodology absorbed fractions for photons and electrons in various organs of the Zubal segmented phantom were calculated and compared to those reported for the mathematical phanto...
Phage therapy pharmacology: calculating phage dosing.
Abedon, Stephen
2011-01-01
Phage therapy, which can be described as a phage-mediated biocontrol of bacteria (or, simply, biocontrol), is the application of bacterial viruses-also bacteriophages or phages-to reduce densities of nuisance or pathogenic bacteria. Predictive calculations for phage therapy dosing should be useful toward rational development of therapeutic as well as biocontrol products. Here, I consider the theoretical basis of a number of concepts relevant to phage dosing for phage therapy including minimum inhibitory concentration (but also "inundation threshold"), minimum bactericidal concentration (but also "clearance threshold"), decimal reduction time (D value), time until bacterial eradication, threshold bacterial density necessary to support phage population growth ("proliferation threshold"), and bacterial density supporting half-maximal phage population growth rates (K(B)). I also address the concepts of phage killing titers, multiplicity of infection, and phage peak densities. Though many of the presented ideas are not unique to this chapter, I nonetheless provide variations on derivations and resulting formulae, plus as appropriate discuss relative importance. The overriding goal is to present a variety of calculations that are useful toward phage therapy dosing so that they may be found in one location and presented in a manner that allows facile appreciation, comparison, and implementation. The importance of phage density as a key determinant of the phage potential to eradicate bacterial targets is stressed throughout the chapter.
Nielsen, Tine B; Wieslander, Elinore; Fogliata, Antonella;
2011-01-01
To investigate differences in calculated doses and normal tissue complication probability (NTCP) values between different dose algorithms.......To investigate differences in calculated doses and normal tissue complication probability (NTCP) values between different dose algorithms....
Calculation of dose distribution above contaminated soil
Kuroda, Junya; Tenzou, Hideki; Manabe, Seiya; Iwakura, Yukiko
2017-07-01
The purpose of this study was to assess the relationship between altitude and the distribution of the ambient dose rate in the air over soil decontamination area by using PHITS simulation code. The geometry configuration was 1000 m ×1000 m area and 1m in soil depth and 100m in altitude from the ground to simulate the area of residences or a school grounds. The contaminated region is supposed to be uniformly contaminated by Cs-137 γ radiation sources. The air dose distribution and space resolution was evaluated for flux of the gamma rays at each altitude, 1, 5, 10, and 20m. The effect of decontamination was calculated by defining sharpness S. S was the ratio of an average flux and a flux at the center of denomination area in each altitude. The suitable flight altitude of the drone is found to be less than 15m above a residence and 31m above a school grounds to confirm the decontamination effect. The calculation results can be a help to determine a flight planning of a drone to minimize the clash risk.
Russell, Kellie R; Tedgren, Asa K Carlsson; Ahnesjö, Anders
2005-09-01
In brachytherapy, tissue heterogeneities, source shielding, and finite patient/phantom extensions affect both the primary and scatter dose distributions. The primary dose is, due to the short range of secondary electrons, dependent only on the distribution of material located on the ray line between the source and dose deposition site. The scatter dose depends on both the direct irradiation pattern and the distribution of material in a large volume surrounding the point of interest, i.e., a much larger volume must be included in calculations to integrate many small dose contributions. It is therefore of interest to consider different methods for the primary and the scatter dose calculation to improve calculation accuracy with limited computer resources. The algorithms in present clinical use ignore these effects causing systematic dose errors in brachytherapy treatment planning. In this work we review a primary and scatter dose separation formalism (PSS) for brachytherapy source characterization to support separate calculation of the primary and scatter dose contributions. We show how the resulting source characterization data can be used to drive more accurate dose calculations using collapsed cone superposition for scatter dose calculations. Two types of source characterization data paths are used: a direct Monte Carlo simulation in water phantoms with subsequent parameterization of the results, and an alternative data path built on processing of AAPM TG43 formatted data to provide similar parameter sets. The latter path is motivated of the large amounts of data already existing in the TG43 format. We demonstrate the PSS methods using both data paths for a clinical 192Ir source. Results are shown for two geometries: a finite but homogeneous water phantom, and a half-slab consisting of water and air. The dose distributions are compared to results from full Monte Carlo simulations and we show significant improvement in scatter dose calculations when the collapsed
Monte Carlo dose calculation in dental amalgam phantom.
Aziz, Mohd Zahri Abdul; Yusoff, A L; Osman, N D; Abdullah, R; Rabaie, N A; Salikin, M S
2015-01-01
It has become a great challenge in the modern radiation treatment to ensure the accuracy of treatment delivery in electron beam therapy. Tissue inhomogeneity has become one of the factors for accurate dose calculation, and this requires complex algorithm calculation like Monte Carlo (MC). On the other hand, computed tomography (CT) images used in treatment planning system need to be trustful as they are the input in radiotherapy treatment. However, with the presence of metal amalgam in treatment volume, the CT images input showed prominent streak artefact, thus, contributed sources of error. Hence, metal amalgam phantom often creates streak artifacts, which cause an error in the dose calculation. Thus, a streak artifact reduction technique was applied to correct the images, and as a result, better images were observed in terms of structure delineation and density assigning. Furthermore, the amalgam density data were corrected to provide amalgam voxel with accurate density value. As for the errors of dose uncertainties due to metal amalgam, they were reduced from 46% to as low as 2% at d80 (depth of the 80% dose beyond Zmax) using the presented strategies. Considering the number of vital and radiosensitive organs in the head and the neck regions, this correction strategy is suggested in reducing calculation uncertainties through MC calculation.
Monte carlo dose calculation in dental amalgam phantom
Mohd Zahri Abdul Aziz
2015-01-01
Full Text Available It has become a great challenge in the modern radiation treatment to ensure the accuracy of treatment delivery in electron beam therapy. Tissue inhomogeneity has become one of the factors for accurate dose calculation, and this requires complex algorithm calculation like Monte Carlo (MC. On the other hand, computed tomography (CT images used in treatment planning system need to be trustful as they are the input in radiotherapy treatment. However, with the presence of metal amalgam in treatment volume, the CT images input showed prominent streak artefact, thus, contributed sources of error. Hence, metal amalgam phantom often creates streak artifacts, which cause an error in the dose calculation. Thus, a streak artifact reduction technique was applied to correct the images, and as a result, better images were observed in terms of structure delineation and density assigning. Furthermore, the amalgam density data were corrected to provide amalgam voxel with accurate density value. As for the errors of dose uncertainties due to metal amalgam, they were reduced from 46% to as low as 2% at d80 (depth of the 80% dose beyond Zmax using the presented strategies. Considering the number of vital and radiosensitive organs in the head and the neck regions, this correction strategy is suggested in reducing calculation uncertainties through MC calculation.
Ramotowska, Anna
2017-01-01
Background. The ELKa system is composed of computer software, with a database of nutrients, and a dedicated USB kitchen scale. It was designed to automatize the everyday calculations of food exchanges and prandial insulin doses. Aim. To investigate the influence of the ELKa on metabolic control in children with type 1 diabetes mellitus (T1DM). Methods. A randomized, parallel, open-label clinical trial involved 106 patients aged 50% of meals achieved lower HbA1C levels (P = 0.002), lower basal insulin amounts (P = 0.049), and lower intrasubject standard deviation of blood glucose levels (P = 0.023) in comparison with the control. Moreover, in the intervention group, significant reduction of HbA1C level, by 0.55% point (P = 0.002), was noted. No intergroup differences were found in the hypoglycemic episodes, BMI-SDS, bolus insulin dosage, and total daily insulin dosage. Conclusions. The ELKa system improves metabolic control in children with T1DM under regular usage. The trial is registered at ClinicalTrials.gov, number NCT02194517.
Recommendations for Insulin Dose Calculator Risk Management
2014-01-01
Several studies have shown the usefulness of an automated insulin dose bolus advisor (BA) in achieving improved glycemic control for insulin-using diabetes patients. Although regulatory agencies have approved several BAs over the past decades, these devices are not standardized in their approach to dosage calculation and include many features that may introduce risk to patients. Moreover, there is no single standard of care for diabetes worldwide and no guidance documents for BAs, specifically. Given the emerging and more stringent regulations on software used in medical devices, the approval process is becoming more difficult for manufacturers to navigate, with some manufacturers opting to remove BAs from their products altogether. A comprehensive literature search was performed, including publications discussing: diabetes BA use and benefit, infusion pump safety and regulation, regulatory submissions, novel BAs, and recommendations for regulation and risk management of BAs. Also included were country-specific and international guidance documents for medical device, infusion pump, medical software, and mobile medical application risk management and regulation. No definitive worldwide guidance exists regarding risk management requirements for BAs, specifically. However, local and international guidance documents for medical devices, infusion pumps, and medical device software offer guidance that can be applied to this technology. In addition, risk management exercises that are algorithm-specific can help prepare manufacturers for regulatory submissions. This article discusses key issues relevant to BA use and safety, and recommends risk management activities incorporating current research and guidance. PMID:24876550
Electron dose rate and photon contamination in electron arc therapy
Pla, M.; Podgorsak, E.B.; Pla, C. (McGill Univ., Montreal, Quebec (Canada))
1989-09-01
The electron dose rate at the depth of dose maximum dmax and the photon contamination are discussed as a function of several parameters of the rotational electron beam. A pseudoarc technique with an angular increment of 10 degrees and a constant number of monitor units per each stationary electron field was used in our experiments. The electron dose rate is defined as the electron dose at a given point in phantom divided by the number of monitor units given for any one stationary electron beam. For a given depth of isocenter di the electron dose rates at dmax are linearly dependent on the nominal field width w, while for a given w the dose rates are inversely proportional to di. The dose rates for rotational electron beams with different di are related through the inverse square law provided that the two beams have (di,w) combinations which give the same characteristic angle beta. The photon dose at the isocenter depends on the arc angle alpha, field width w, and isocenter depth di. For constant w and di the photon dose at isocenter is proportional to alpha, for constant alpha and w it is proportional to di, and for constant alpha and di it is inversely proportional to w. The w and di dependence implies that for the same alpha the photon dose at the isocenter is inversely proportional to the electron dose rate at dmax.
Electronic compensation technique to deliver a total body dose
Lakeman, Tara E.
Purpose: Total body irradiation (TBI) uses large parallel-opposed radiation fields to suppress the patient's immune system and eradicate the residual cancer cells in preparation of recipient for bone marrow transplant. The manual placement of lead compensators has been conventionally used to compensate for the varying thickness throughout the body in large-field TBI. The goal of this study is to pursue utilizing the modern electronic compensation technique to more accurately and efficiently deliver dose to patients in need of TBI. Method: Treatment plans utilizing the electronic compensation to deliver a total body dose were created retrospectively for patients for whom CT data had been previously acquired. Each treatment plan includes two pair of parallel opposed fields. One pair of large fields is used to encompass the majority of the patient's anatomy. The other pair are very small open fields focused only on the thin bottom portion of the patient's anatomy, which requires much less radiation than the rest of the body to reach 100% of the prescribed dose. A desirable fluence pattern was manually painted within each of the larger fields for each patient to provide a more uniform distribution. Results: Dose-volume histograms (DVH) were calculated for evaluating the electronic compensation technique. In the electronically compensated plans, the maximum body doses calculated from the DVH were reduced from the conventionally-compensated plans by an average of 15%, indicating a more uniform dose. The mean body doses calculated from the electronically compensated DVH remained comparable to that of the conventionally-compensated plans, indicating an accurate delivery of the prescription dose using electronic compensation. All calculated monitor units were within clinically acceptable limits. Conclusion: Electronic compensation technique for TBI will not increase the beam on time beyond clinically acceptable limits while it can substantially reduce the compensator setup
Park, Jong Min; Park, So Yeon; Kim, Jung In; Kim, Jin Ho [Dept. of Radiation Oncology, Seoul National University Hospital, Seoul (Korea, Republic of); Wu, Hong Gyun [Dept. of Radiation Oncology, Seoul National University College of Medicine, Seoul (Korea, Republic of)
2015-10-15
Since those organs are small in volume, dose calculation for those organs seems to be more susceptible to the calculation grid size in the treatment planning system (TPS). Moreover, since they are highly radio-sensitive organs, especially eye lens, they should be considered carefully for radiotherapy. On the other hand, in the treatment of head and neck (H and N) cancer or brain tumor that generally involves radiation exposure to eye lens and optic apparatus, intensity modulated radiation therapy (IMRT) or volumetric modulated arc therapy (VMAT) techniques are frequently used because of the proximity of various radio-sensitive normal organs to the target volumes. Since IMRT and VMAT can deliver prescription dose to target volumes while minimizing dose to nearby organs at risk (OARs) by generating steep dose gradients near the target volumes, high dose gradient sometimes occurs near or at the eye lenses and optic apparatus. In this case, the effect of dose calculation resolution on the accuracy of calculated dose to eye lens and optic apparatus might be significant. Therefore, the effect of dose calculation grid size on the accuracy of calculated doses for each eye lens and optic apparatus was investigated in this study. If an inappropriate calculation resolution was applied for dose calculation of eye lens and optic apparatus, considerable errors can be occurred due to the volume averaging effect in high dose gradient region.
Methods of calculating radiation absorbed dose.
Wegst, A V
1987-01-01
The new tumoricidal radioactive agents being developed will require a careful estimate of radiation absorbed tumor and critical organ dose for each patient. Clinical methods will need to be developed using standard imaging or counting instruments to determine cumulated organ activities with tracer amounts before the therapeutic administration of the material. Standard MIRD dosimetry methods can then be applied.
Calculation of dose conversion factors for thoron decay products
Ishikawa, Tetsuo [National Institute of Radiological Sciences, 4-9-1 Anagawa, Inage-ku, Chiba, 263-8555 (Japan); Tokonami, Shinji [National Institute of Radiological Sciences, 4-9-1 Anagawa, Inage-ku, Chiba, 263-8555 (Japan); Nemeth, Csaba [Pannon University, 10 Egyetem St, 8201 Veszprem (Hungary)
2007-12-15
The dose conversion factors for short-lived thoron decay products were calculated using a dosimetric approach. The calculations were based on a computer program LUDEP, which implements the ICRP 66 respiratory tract model. The dose per equilibrium equivalent concentration for thoron (EETC) was calculated with respect to (1) equivalent dose to each region of the lung tissues (bronchial, bronchiolar and alveolar), (2) weighted equivalent dose to organs other than lung, and (3) effective dose. The calculations indicated that (1) the most exposed region of the lung tissues was the bronchial for the unattached fraction and the bronchiolar for the attached fraction, (2) the effective dose is dominated by the contribution of lung dose, and (3) the effective dose per EETC was about four times larger than the effective dose per equilibrium equivalent concentration for radon (EERC). The calculated dose conversion factors were applied to the comparative dosimetry for some thoron-enhanced areas where the EERC and EETC have been measured. In the case of a spa in Japan, the dose from thoron decay products was larger than the dose from radon decay products.
Calculation of dose conversion factors for thoron decay products.
Ishikawa, Tetsuo; Tokonami, Shinji; Nemeth, Csaba
2007-12-01
The dose conversion factors for short-lived thoron decay products were calculated using a dosimetric approach. The calculations were based on a computer program LUDEP, which implements the ICRP 66 respiratory tract model. The dose per equilibrium equivalent concentration for thoron (EETC) was calculated with respect to (1) equivalent dose to each region of the lung tissues (bronchial, bronchiolar and alveolar), (2) weighted equivalent dose to organs other than lung, and (3) effective dose. The calculations indicated that (1) the most exposed region of the lung tissues was the bronchial for the unattached fraction and the bronchiolar for the attached fraction, (2) the effective dose is dominated by the contribution of lung dose, and (3) the effective dose per EETC was about four times larger than the effective dose per equilibrium equivalent concentration for radon (EERC). The calculated dose conversion factors were applied to the comparative dosimetry for some thoron-enhanced areas where the EERC and EETC have been measured. In the case of a spa in Japan, the dose from thoron decay products was larger than the dose from radon decay products.
MLHD online : manual for the herbicide dose calculation module
PRI,; Kempenaar, C.
2004-01-01
MLHD is short for Minimum Lethal Herbicide Dose. MLHD is a new concept within chemical weed control. It supports effective weed control while herbicide doses are kept at minimum effective levels (minimum lethal doses). This manual describes how to use of the MLHD calculation module for users from ou
Block Tridiagonal Matrices in Electronic Structure Calculations
Petersen, Dan Erik
This thesis focuses on some of the numerical aspects of the treatment of the electronic structure problem, in particular that of determining the ground state electronic density for the non–equilibrium Green’s function formulation of two–probe systems and the calculation of transmission in the Lan...
Fast dose calculation in magnetic fields with GPUMCD
Hissoiny, S; Ozell, B [Ecole Polytechnique de Montreal, Departement de genie informatique et genie logiciel, 2500 Chemin de Polytechnique, Montreal, Quebec H3T 1J4 (Canada); Raaijmakers, A J E; Raaymakers, B W [Department of Radiotherapy, University Medical Center Utrecht, Heidelberglaan 100, 3584 CX, Utrecht (Netherlands); Despres, P, E-mail: sami.hissoiny@polymtl.ca [Departement de physique, Universite Laval, Quebec (Canada)
2011-08-21
A new hybrid imaging-treatment modality, the MRI-Linac, involves the irradiation of the patient in the presence of a strong magnetic field. This field acts on the charged particles, responsible for depositing dose, through the Lorentz force. These conditions require a dose calculation engine capable of taking into consideration the effect of the magnetic field on the dose distribution during the planning stage. Also in the case of a change in anatomy at the time of treatment, a fast online replanning tool is desirable. It is improbable that analytical solutions such as pencil beam calculations can be efficiently adapted for dose calculations within a magnetic field. Monte Carlo simulations have therefore been used for the computations but the calculation speed is generally too slow to allow online replanning. In this work, GPUMCD, a fast graphics processing unit (GPU)-based Monte Carlo dose calculation platform, was benchmarked with a new feature that allows dose calculations within a magnetic field. As a proof of concept, this new feature is validated against experimental measurements. GPUMCD was found to accurately reproduce experimental dose distributions according to a 2%-2 mm gamma analysis in two cases with large magnetic field-induced dose effects: a depth-dose phantom with an air cavity and a lateral-dose phantom surrounded by air. Furthermore, execution times of less than 15 s were achieved for one beam in a prostate case phantom for a 2% statistical uncertainty while less than 20 s were required for a seven-beam plan. These results indicate that GPUMCD is an interesting candidate, being fast and accurate, for dose calculations for the hybrid MRI-Linac modality.
Dosimetric accuracy of tomotherapy dose calculation in thorax lesions
Mangili Paola
2011-02-01
Full Text Available Abstract Background To analyse limits and capabilities in dose calculation of collapsed-cone-convolution (CCC algorithm implemented in helical tomotherapy (HT treatment planning system for thorax lesions. Methods The agreement between measured and calculated dose was verified both in homogeneous (Cheese Phantom and in a custom-made inhomogeneous phantom. The inhomogeneous phantom was employed to mimic a patient's thorax region with lung density encountered in extreme cases and acrylic inserts of various dimensions and positions inside the lung cavity. For both phantoms, different lung treatment plans (single or multiple metastases and targets in the mediastinum using HT technique were simulated and verified. Point and planar dose measurements, both with radiographic extended-dose-range (EDR2 and radiochromic external-beam-therapy (EBT2 films, were performed. Absolute point dose measurements, dose profile comparisons and quantitative analysis of gamma function distributions were analyzed. Results An excellent agreement between measured and calculated dose distributions was found in homogeneous media, both for point and planar dose measurements. Absolute dose deviations Conclusions Very acceptable accuracy was found for complex lung treatment plans calculated with CCC algorithm implemented in the tomotherapy TPS even in the heterogeneous phantom with very low lung-density.
Clinical implementation and evaluation of the Acuros dose calculation algorithm.
Yan, Chenyu; Combine, Anthony G; Bednarz, Greg; Lalonde, Ronald J; Hu, Bin; Dickens, Kathy; Wynn, Raymond; Pavord, Daniel C; Saiful Huq, M
2017-08-20
The main aim of this study is to validate the Acuros XB dose calculation algorithm for a Varian Clinac iX linac in our clinics, and subsequently compare it with the wildely used AAA algorithm. The source models for both Acuros XB and AAA were configured by importing the same measured beam data into Eclipse treatment planning system. Both algorithms were validated by comparing calculated dose with measured dose on a homogeneous water phantom for field sizes ranging from 6 cm × 6 cm to 40 cm × 40 cm. Central axis and off-axis points with different depths were chosen for the comparison. In addition, the accuracy of Acuros was evaluated for wedge fields with wedge angles from 15 to 60°. Similarly, variable field sizes for an inhomogeneous phantom were chosen to validate the Acuros algorithm. In addition, doses calculated by Acuros and AAA at the center of lung equivalent tissue from three different VMAT plans were compared to the ion chamber measured doses in QUASAR phantom, and the calculated dose distributions by the two algorithms and their differences on patients were compared. Computation time on VMAT plans was also evaluated for Acuros and AAA. Differences between dose-to-water (calculated by AAA and Acuros XB) and dose-to-medium (calculated by Acuros XB) on patient plans were compared and evaluated. For open 6 MV photon beams on the homogeneous water phantom, both Acuros XB and AAA calculations were within 1% of measurements. For 23 MV photon beams, the calculated doses were within 1.5% of measured doses for Acuros XB and 2% for AAA. Testing on the inhomogeneous phantom demonstrated that AAA overestimated doses by up to 8.96% at a point close to lung/solid water interface, while Acuros XB reduced that to 1.64%. The test on QUASAR phantom showed that Acuros achieved better agreement in lung equivalent tissue while AAA underestimated dose for all VMAT plans by up to 2.7%. Acuros XB computation time was about three times faster than AAA for VMAT plans, and
Methodology of dose calculation for the SRS SAR
Price, J.B.
1991-07-01
The Savannah River Site (SRS) Safety Analysis Report (SAR) covering K reactor operation assesses a spectrum of design basis accidents. The assessment includes estimation of the dose consequences from the analyzed accidents. This report discusses the methodology used to perform the dose analysis reported in the SAR and also includes the quantified doses. Doses resulting from postulated design basis reactor accidents in Chapter 15 of the SAR are discussed, as well as an accident in which three percent of the fuel melts. Doses are reported for both atmospheric and aqueous releases. The methodology used to calculate doses from these accidents as reported in the SAR is consistent with NRC guidelines and industry standards. The doses from the design basis accidents for the SRS reactors are below the limits set for commercial reactors by the NRC and also meet industry criteria. A summary of doses for various postulated accidents is provided.
Fluence-convolution broad-beam (FCBB) dose calculation.
Lu, Weiguo; Chen, Mingli
2010-12-07
IMRT optimization requires a fast yet relatively accurate algorithm to calculate the iteration dose with small memory demand. In this paper, we present a dose calculation algorithm that approaches these goals. By decomposing the infinitesimal pencil beam (IPB) kernel into the central axis (CAX) component and lateral spread function (LSF) and taking the beam's eye view (BEV), we established a non-voxel and non-beamlet-based dose calculation formula. Both LSF and CAX are determined by a commissioning procedure using the collapsed-cone convolution/superposition (CCCS) method as the standard dose engine. The proposed dose calculation involves a 2D convolution of a fluence map with LSF followed by ray tracing based on the CAX lookup table with radiological distance and divergence correction, resulting in complexity of O(N(3)) both spatially and temporally. This simple algorithm is orders of magnitude faster than the CCCS method. Without pre-calculation of beamlets, its implementation is also orders of magnitude smaller than the conventional voxel-based beamlet-superposition (VBS) approach. We compared the presented algorithm with the CCCS method using simulated and clinical cases. The agreement was generally within 3% for a homogeneous phantom and 5% for heterogeneous and clinical cases. Combined with the 'adaptive full dose correction', the algorithm is well suitable for calculating the iteration dose during IMRT optimization.
Dose Rate Calculations for Rotary Mode Core Sampling Exhauster
Foust, D J
2000-01-01
This document provides the calculated estimated dose rates for three external locations on the Rotary Mode Core Sampling (RMCS) exhauster HEPA filter housing, per the request of Characterization Field Engineering.
[CUDA-based fast dose calculation in radiotherapy].
Wang, Xianliang; Liu, Cao; Hou, Qing
2011-10-01
Dose calculation plays a key role in treatment planning of radiotherapy. Algorithms for dose calculation require high accuracy and computational efficiency. Finite size pencil beam (FSPB) algorithm is a method commonly adopted in the treatment planning system for radiotherapy. However, improvement on its computational efficiency is still desirable for such purpose as real time treatment planning. In this paper, we present an implementation of the FSPB, by which the most time-consuming parts in the algorithm are parallelized and ported on graphic processing unit (GPU). Compared with the FSPB completely running on central processing unit (CPU), the GPU-implemented FSPB can speed up the dose calculation for 25-35 times on a low price GPU (Geforce GT320) and for 55-100 times on a Tesla C1060, indicating that the GPU-implemented FSPB can provide fast enough dose calculations for real-time treatment planning.
Verification of calculated skin doses in postmastectomy helical tomotherapy.
Ito, Shima; Parker, Brent C; Levine, Renee; Sanders, Mary Ella; Fontenot, Jonas; Gibbons, John; Hogstrom, Kenneth
2011-10-01
To verify the accuracy of calculated skin doses in helical tomotherapy for postmastectomy radiation therapy (PMRT). In vivo thermoluminescent dosimeters (TLDs) were used to measure the skin dose at multiple points in each of 14 patients throughout the course of treatment on a TomoTherapy Hi·Art II system, for a total of 420 TLD measurements. Five patients were evaluated near the location of the mastectomy scar, whereas 9 patients were evaluated throughout the treatment volume. The measured dose at each location was compared with calculations from the treatment planning system. The mean difference and standard error of the mean difference between measurement and calculation for the scar measurements was -1.8% ± 0.2% (standard deviation [SD], 4.3%; range, -11.1% to 10.6%). The mean difference and standard error of the mean difference between measurement and calculation for measurements throughout the treatment volume was -3.0% ± 0.4% (SD, 4.7%; range, -18.4% to 12.6%). The mean difference and standard error of the mean difference between measurement and calculation for all measurements was -2.1% ± 0.2% (standard deviation, 4.5%: range, -18.4% to 12.6%). The mean difference between measured and calculated TLD doses was statistically significant at two standard deviations of the mean, but was not clinically significant (i.e., was <5%). However, 23% of the measured TLD doses differed from the calculated TLD doses by more than 5%. The mean of the measured TLD doses agreed with TomoTherapy calculated TLD doses within our clinical criterion of 5%. Copyright © 2011 Elsevier Inc. All rights reserved.
Gamma Knife radiosurgery with CT image-based dose calculation.
Xu, Andy Yuanguang; Bhatnagar, Jagdish; Bednarz, Greg; Niranjan, Ajay; Kondziolka, Douglas; Flickinger, John; Lunsford, L Dade; Huq, M Saiful
2015-11-08
The Leksell GammaPlan software version 10 introduces a CT image-based segmentation tool for automatic skull definition and a convolution dose calculation algorithm for tissue inhomogeneity correction. The purpose of this work was to evaluate the impact of these new approaches on routine clinical Gamma Knife treatment planning. Sixty-five patients who underwent CT image-guided Gamma Knife radiosurgeries at the University of Pittsburgh Medical Center in recent years were retrospectively investigated. The diagnoses for these cases include trigeminal neuralgia, meningioma, acoustic neuroma, AVM, glioma, and benign and metastatic brain tumors. Dose calculations were performed for each patient with the same dose prescriptions and the same shot arrangements using three different approaches: 1) TMR 10 dose calculation with imaging skull definition; 2) convolution dose calculation with imaging skull definition; 3) TMR 10 dose calculation with conventional measurement-based skull definition. For each treatment matrix, the total treatment time, the target coverage index, the selectivity index, the gradient index, and a set of dose statistics parameters were compared between the three calculations. The dose statistics parameters investigated include the prescription isodose volume, the 12 Gy isodose volume, the minimum, maximum and mean doses on the treatment targets, and the critical structures under consideration. The difference between the convolution and the TMR 10 dose calculations for the 104 treatment matrices were found to vary with the patient anatomy, location of the treatment shots, and the tissue inhomogeneities around the treatment target. An average difference of 8.4% was observed for the total treatment times between the convolution and the TMR algorithms. The maximum differences in the treatment times, the prescription isodose volumes, the 12 Gy isodose volumes, the target coverage indices, the selectivity indices, and the gradient indices from the convolution
Dose-Response Calculator for ArcGIS
Hanser, Steven E.; Aldridge, Cameron L.; Leu, Matthias; Nielsen, Scott E.
2011-01-01
The Dose-Response Calculator for ArcGIS is a tool that extends the Environmental Systems Research Institute (ESRI) ArcGIS 10 Desktop application to aid with the visualization of relationships between two raster GIS datasets. A dose-response curve is a line graph commonly used in medical research to examine the effects of different dosage rates of a drug or chemical (for example, carcinogen) on an outcome of interest (for example, cell mutations) (Russell and others, 1982). Dose-response curves have recently been used in ecological studies to examine the influence of an explanatory dose variable (for example, percentage of habitat cover, distance to disturbance) on a predicted response (for example, survival, probability of occurrence, abundance) (Aldridge and others, 2008). These dose curves have been created by calculating the predicted response value from a statistical model at different levels of the explanatory dose variable while holding values of other explanatory variables constant. Curves (plots) developed using the Dose-Response Calculator overcome the need to hold variables constant by using values extracted from the predicted response surface of a spatially explicit statistical model fit in a GIS, which include the variation of all explanatory variables, to visualize the univariate response to the dose variable. Application of the Dose-Response Calculator can be extended beyond the assessment of statistical model predictions and may be used to visualize the relationship between any two raster GIS datasets (see example in tool instructions). This tool generates tabular data for use in further exploration of dose-response relationships and a graph of the dose-response curve.
Local orbitals in electron scattering calculations*
Winstead, Carl L.; McKoy, Vincent
2016-05-01
We examine the use of local orbitals to improve the scaling of calculations that incorporate target polarization in a description of low-energy electron-molecule scattering. After discussing the improved scaling that results, we consider the results of a test calculation that treats scattering from a two-molecule system using both local and delocalized orbitals. Initial results are promising. Contribution to the Topical Issue "Advances in Positron and Electron Scattering", edited by Paulo Limao-Vieira, Gustavo Garcia, E. Krishnakumar, James Sullivan, Hajime Tanuma and Zoran Petrovic.
Calculated Electron Fluxes at Airplane Altitudes
Schaefer, R K; Stanev, T
1993-01-01
A precision measurement of atmospheric electron fluxes has been performed on a Japanese commercial airliner (Enomoto, {\\it et al.}, 1991). We have performed a monte carlo calculation of the cosmic ray secondary electron fluxes expected in this experiment. The monte carlo uses the hadronic portion of our neutrino flux cascade program combined with the electromagnetic cascade portion of the CERN library program GEANT. Our results give good agreement with the data, provided we boost the overall normalization of the primary cosmic ray flux by 12\\% over the normalization used in the neutrino flux calculation.
Study of dose calculation on breast brachytherapy using prism TPS
Fendriani, Yoza; Haryanto, Freddy [Nuclear Physics and Biophysics Research Division, FMIPA Institut Teknologi Bandung, Physics Buildings, Jl. Ganesha 10, Bandung 40132 (Indonesia)
2015-09-30
PRISM is one of non-commercial Treatment Planning System (TPS) and is developed at the University of Washington. In Indonesia, many cancer hospitals use expensive commercial TPS. This study aims to investigate Prism TPS which been applied to the dose distribution of brachytherapy by taking into account the effect of source position and inhomogeneities. The results will be applicable for clinical Treatment Planning System. Dose calculation has been implemented for water phantom and CT scan images of breast cancer using point source and line source. This study used point source and line source and divided into two cases. On the first case, Ir-192 seed source is located at the center of treatment volume. On the second case, the source position is gradually changed. The dose calculation of every case performed on a homogeneous and inhomogeneous phantom with dimension 20 × 20 × 20 cm{sup 3}. The inhomogeneous phantom has inhomogeneities volume 2 × 2 × 2 cm{sup 3}. The results of dose calculations using PRISM TPS were compared to literature data. From the calculation of PRISM TPS, dose rates show good agreement with Plato TPS and other study as published by Ramdhani. No deviations greater than ±4% for all case. Dose calculation in inhomogeneous and homogenous cases show similar result. This results indicate that Prism TPS is good in dose calculation of brachytherapy but not sensitive for inhomogeneities. Thus, the dose calculation parameters developed in this study were found to be applicable for clinical treatment planning of brachytherapy.
A decision tool to adjust the prescribed dose after change in the dose calculation algorithm
Abdulhamid Chaikh
2014-12-01
Full Text Available Purpose: This work aims to introduce a method to quantify and assess the differences in monitor unites MUs when changing to new dose calculation software that uses a different algorithm, and to evaluate the need and extent of adjustment of the prescribed dose to maintain the same clinical results. Methods: Doses were calculated using two classical algorithms based on the Pencil Beam Convolution PBC model, using 6 patients presenting lung cancers. For each patient, 3 treatment plans were generated: Plan 1 was calculated using reference algorithm PBC without heterogeneity correction, Plan 2 was calculated using test algorithm with heterogeneity correction, and in plan 3 the dose was recalculated using test algorithm and monitor unites MUs obtained from plan 1 as input. To assess the differences in the calculated MUs, isocenter dose, and spatial dose distributions using a gamma index were compared. Statistical analysis was based on a Wilcoxon signed rank test. Results: The test algorithm in plan 2 calculated significantly less MUs than reference algorithm in plan 1 by on average 5%, (p < 0.001. We also found underestimating dose for target volumes using 3D gamma index analysis. In this example, in order to obtain the same clinical outcomes with the two algorithms the prescribed dose should be adjusted by 5%.Conclusion: This method provides a quantitative evaluation of the differences between two dose calculation algorithms and the consequences on the prescribed dose. It could be used to adjust the prescribed dose when changing calculation software to maintain the same clinical results as obtained with the former software. In particular, the gamma evaluation could be applied to any situation where changes in the dose calculation occur in radiotherapy.
Widder, Joachim; Hollander, Miranda; Ubbels, Jan F.; Bolt, Rene A.; Langendijk, Johannes A.
2010-01-01
Purpose: To define a method of dose prescription employing Monte Carlo (MC) dose calculation in stereotactic body radiotherapy (SBRT) for lung tumours aiming at a dose as low as possible outside of the PTV. Methods and materials: Six typical T1 lung tumours - three small, three large - were construc
Calculation of the dose caused by internal radiation
NONE
2000-07-01
For the purposes of monitoring radiation exposure it is necessary to determine or to estimate the dose caused by both external and internal radiation. When comparing the value of exposure to the dose limits, account must be taken of the total dose incurred from different sources. This guide explains how to calculate the committed effective dose caused by internal radiation and gives the conversion factors required for the calculation. Application of the maximum values for radiation exposure is dealt with in ST guide 7.2, which also sets out the definitions of the quantities and concepts most commonly used in the monitoring of radiation exposure. The monitoring of exposure and recording of doses are dealt with in ST Guides 7.1 and 7.4.
Program Calculates Current Densities Of Electronic Designs
Cox, Brian
1996-01-01
PDENSITY computer program calculates current densities for use in calculating power densities of electronic designs. Reads parts-list file for given design, file containing current required for each part, and file containing size of each part. For each part in design, program calculates current density in units of milliamperes per square inch. Written by use of AWK utility for Sun4-series computers running SunOS 4.x and IBM PC-series and compatible computers running MS-DOS. Sun version of program (NPO-19588). PC version of program (NPO-19171).
Quantification of Proton Dose Calculation Accuracy in the Lung
Grassberger, Clemens, E-mail: Grassberger.Clemens@mgh.harvard.edu [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, Massachusetts (United States); Center for Proton Radiotherapy, Paul Scherrer Institute, Villigen (Switzerland); Daartz, Juliane; Dowdell, Stephen; Ruggieri, Thomas; Sharp, Greg; Paganetti, Harald [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, Massachusetts (United States)
2014-06-01
Purpose: To quantify the accuracy of a clinical proton treatment planning system (TPS) as well as Monte Carlo (MC)–based dose calculation through measurements and to assess the clinical impact in a cohort of patients with tumors located in the lung. Methods and Materials: A lung phantom and ion chamber array were used to measure the dose to a plane through a tumor embedded in the lung, and to determine the distal fall-off of the proton beam. Results were compared with TPS and MC calculations. Dose distributions in 19 patients (54 fields total) were simulated using MC and compared to the TPS algorithm. Results: MC increased dose calculation accuracy in lung tissue compared with the TPS and reproduced dose measurements in the target to within ±2%. The average difference between measured and predicted dose in a plane through the center of the target was 5.6% for the TPS and 1.6% for MC. MC recalculations in patients showed a mean dose to the clinical target volume on average 3.4% lower than the TPS, exceeding 5% for small fields. For large tumors, MC also predicted consistently higher V5 and V10 to the normal lung, because of a wider lateral penumbra, which was also observed experimentally. Critical structures located distal to the target could show large deviations, although this effect was highly patient specific. Range measurements showed that MC can reduce range uncertainty by a factor of ∼2: the average (maximum) difference to the measured range was 3.9 mm (7.5 mm) for MC and 7 mm (17 mm) for the TPS in lung tissue. Conclusion: Integration of Monte Carlo dose calculation techniques into the clinic would improve treatment quality in proton therapy for lung cancer by avoiding systematic overestimation of target dose and underestimation of dose to normal lung. In addition, the ability to confidently reduce range margins would benefit all patients by potentially lowering toxicity.
COMPARING MEASURED AND CALCULATED DOSES IN INTERVENTIONAL CARDIOLOGY PROCEDURES.
Oliveira da Silva, M W; Canevaro, L V; Hunt, J; Rodrigues, B B D
2017-03-16
Interventional cardiology requires complex procedures and can result in high doses and dose rates to the patient and medical staff. The many variables that influence the dose to the patient and staff include the beam position and angle, beam size, kVp, filtration, kerma-area product and focus-skin distance. A number of studies using the Monte Carlo method have been undertaken to obtain prospective dose assessments. In this paper, detailed irradiation scenarios were simulated mathematically and the resulting dose estimates were compared with real measurements made previously under very similar irradiation conditions and geometries. The real measurements and the calculated doses were carried out using or simulating an interventional cardiology system with a flat monoplane detector installed in a dedicated room with an Alderson phantom placed on the procedure table. The X-ray spectra, beam angles, focus-skin distance, measured kerma-area product and filtration were simulated, and the real dose measurements and calculated doses were compared. It was shown that the Monte Carlo method was capable of reproducing the real dose measurements within acceptable levels of uncertainty.
Napier, B.A.
1992-12-01
A series of scoping calculations has been undertaken to evaluate the absolute and relative contributions of different radionuclides and exposure pathways to doses that may have been received by individuals living in the vicinity of the Hanford Site. This scoping calculation (Calculation 004) examined the contributions of numerous radionuclides to cumulative dose via environmental exposures and accumulation in foods. Addressed in this calculation were the contributions to organ and effective dose of infants and adults from (1) air submersion and groundshine external dose, (2) inhalation, (3) ingestion of soil by humans, (4) ingestion of leafy vegetables, (5) ingestion of other vegetables and fruits, (6) ingestion of meat, (7) ingestion of eggs, and (8) ingestion of cows` milk from Feeding Regime 1, as described in calculation 002. This calculation specifically addresses cumulative radiation doses to infants and adults resulting from releases occurring over the period 1945 through 1972.
Polystyrene calorimeter for electron beam dose measurements
Miller, A.
1995-01-01
Calorimeters from polystrene have been constructed for dose measurement at 4-10 MeV electron accelerators. These calorimeters have been used successfully for a few years, and polystyrene calorimeters for use at energies down to 1 MeV and being tested. Advantage of polystyrene as the absorbing...
Limitations of analytical dose calculations for small field proton radiosurgery
Geng, Changran; Daartz, Juliane; Lam-Tin-Cheung, Kimberley; Bussiere, Marc; Shih, Helen A.; Paganetti, Harald; Schuemann, Jan
2017-01-01
The purpose of the work was to evaluate the dosimetric uncertainties of an analytical dose calculation engine and the impact on treatment plans using small fields in intracranial proton stereotactic radiosurgery (PSRS) for a gantry based double scattering system. 50 patients were evaluated including 10 patients for each of 5 diagnostic indications of: arteriovenous malformation (AVM), acoustic neuroma (AN), meningioma (MGM), metastasis (METS), and pituitary adenoma (PIT). Treatment plans followed standard prescription and optimization procedures for PSRS. We performed comparisons between delivered dose distributions, determined by Monte Carlo (MC) simulations, and those calculated with the analytical dose calculation algorithm (ADC) used in our current treatment planning system in terms of dose volume histogram parameters and beam range distributions. Results show that the difference in the dose to 95% of the target (D95) is within 6% when applying measured field size output corrections for AN, MGM, and PIT. However, for AVM and METS, the differences can be as great as 10% and 12%, respectively. Normalizing the MC dose to the ADC dose based on the dose of voxels in a central area of the target reduces the difference of the D95 to within 6% for all sites. The generally applied margin to cover uncertainties in range (3.5% of the prescribed range + 1 mm) is not sufficient to cover the range uncertainty for ADC in all cases, especially for patients with high tissue heterogeneity. The root mean square of the R90 difference, the difference in the position of distal falloff to 90% of the prescribed dose, is affected by several factors, especially the patient geometry heterogeneity, modulation and field diameter. In conclusion, implementation of Monte Carlo dose calculation techniques into the clinic can reduce the uncertainty of the target dose for proton stereotactic radiosurgery. If MC is not available for treatment planning, using MC dose distributions to
Multigrid Methods in Electronic Structure Calculations
Briggs, E L; Bernholc, J
1996-01-01
We describe a set of techniques for performing large scale ab initio calculations using multigrid accelerations and a real-space grid as a basis. The multigrid methods provide effective convergence acceleration and preconditioning on all length scales, thereby permitting efficient calculations for ill-conditioned systems with long length scales or high energy cut-offs. We discuss specific implementations of multigrid and real-space algorithms for electronic structure calculations, including an efficient multigrid-accelerated solver for Kohn-Sham equations, compact yet accurate discretization schemes for the Kohn-Sham and Poisson equations, optimized pseudo\\-potentials for real-space calculations, efficacious computation of ionic forces, and a complex-wavefunction implementation for arbitrary sampling of the Brillioun zone. A particular strength of a real-space multigrid approach is its ready adaptability to massively parallel computer architectures, and we present an implementation for the Cray-T3D with essen...
Tedgren, Åsa Carlsson; Carlsson, Gudrun Alm
2013-04-21
Model-based dose calculation algorithms (MBDCAs), recently introduced in treatment planning systems (TPS) for brachytherapy, calculate tissue absorbed doses. In the TPS framework, doses have hereto been reported as dose to water and water may still be preferred as a dose specification medium. Dose to tissue medium Dmed then needs to be converted into dose to water in tissue Dw,med. Methods to calculate absorbed dose to differently sized water compartments/cavities inside tissue, infinitesimal (used for definition of absorbed dose), small, large or intermediate, are reviewed. Burlin theory is applied to estimate photon energies at which cavity sizes in the range 1 nm-10 mm can be considered small or large. Photon and electron energy spectra are calculated at 1 cm distance from the central axis in cylindrical phantoms of bone, muscle and adipose tissue for 20, 50, 300 keV photons and photons from (125)I, (169)Yb and (192)Ir sources; ratios of mass-collision-stopping powers and mass energy absorption coefficients are calculated as applicable to convert Dmed into Dw,med for small and large cavities. Results show that 1-10 nm sized cavities are small at all investigated photon energies; 100 µm cavities are large only at photon energies <20 keV. A choice of an appropriate conversion coefficient Dw, med/Dmed is discussed in terms of the cavity size in relation to the size of important cellular targets. Free radicals from DNA bound water of nanometre dimensions contribute to DNA damage and cell killing and may be the most important water compartment in cells implying use of ratios of mass-collision-stopping powers for converting Dmed into Dw,med.
Impact of dose calculation algorithm on radiation therapy
Wen-Zhou; Chen; Ying; Xiao; Jun; Li
2014-01-01
The quality of radiation therapy depends on the ability to maximize the tumor control probability while minimizing the normal tissue complication probability.Both of these two quantities are directly related to the accuracy of dose distributions calculated by treatment planning systems.The commonly used dose calculation algorithms in the treatment planning systems are reviewed in this work.The accuracy comparisons among these algorithms are illustrated by summarizing the highly cited research papers on this topic.Further,the correlation between the algorithms and tumor control probability/normal tissue complication probability values are manifested by several recent studies from different groups.All the cases demonstrate that dose calculation algorithms play a vital role in radiation therapy.
Program Calculates Power Demands Of Electronic Designs
Cox, Brian
1995-01-01
CURRENT computer program calculates power requirements of electronic designs. For given design, CURRENT reads in applicable parts-list file and file containing current required for each part. Program also calculates power required for circuit at supply potentials of 5.5, 5.0, and 4.5 volts. Written by use of AWK utility for Sun4-series computers running SunOS 4.x and IBM PC-series and compatible computers running MS-DOS. Sun version of program (NPO-19590). PC version of program (NPO-19111).
Effects of the difference in tube voltage of the CT scanner on dose calculation
Rhee, Dong Joo; Moon, Young Min; Kim, Jung Ki; Jeong, Dong Hyeok
2015-01-01
Computed Tomography (CT) measures the attenuation coefficient of an object and converts the value assigned to each voxel into a CT number. In radiation therapy, CT number, which is directly proportional to the linear attenuation coefficient, is required to be converted to electron density for radiation dose calculation for cancer treatment. However, if various tube voltages were applied to take the patient CT image without applying the specific CT number to electron density conversion curve, the accuracy of dose calculation would be unassured. In this study, changes in CT numbers for different materials due to change in tube voltage were demonstrated and the dose calculation errors in percentage depth dose (PDD) and a clinical case were analyzed. The maximum dose difference in PDD from TPS dose calculation and Monte Carlo simulation were 1.3 % and 1.1 % respectively when applying the same CT number to electron density conversion curve to the 80 kVp and 140 kVp images. In the clinical case, the different CT nu...
Isogeometric analysis in electronic structure calculations
Cimrman, Robert; Kolman, Radek; Tůma, Miroslav; Vackář, Jiří
2016-01-01
In electronic structure calculations, various material properties can be obtained by means of computing the total energy of a system as well as derivatives of the total energy w.r.t. atomic positions. The derivatives, also known as Hellman-Feynman forces, require, because of practical computational reasons, the discretized charge density and wave functions having continuous second derivatives in the whole solution domain. We describe an application of isogeometric analysis (IGA), a spline modification of finite element method (FEM), to achieve the required continuity. The novelty of our approach is in employing the technique of B\\'ezier extraction to add the IGA capabilities to our FEM based code for ab-initio calculations of electronic states of non-periodic systems within the density-functional framework, built upon the open source finite element package SfePy. We compare FEM and IGA in benchmark problems and several numerical results are presented.
Smartphone apps for calculating insulin dose: a systematic assessment.
Huckvale, Kit; Adomaviciute, Samanta; Prieto, José Tomás; Leow, Melvin Khee-Shing; Car, Josip
2015-05-06
Medical apps are widely available, increasingly used by patients and clinicians, and are being actively promoted for use in routine care. However, there is little systematic evidence exploring possible risks associated with apps intended for patient use. Because self-medication errors are a recognized source of avoidable harm, apps that affect medication use, such as dose calculators, deserve particular scrutiny. We explored the accuracy and clinical suitability of apps for calculating medication doses, focusing on insulin calculators for patients with diabetes as a representative use for a prevalent long-term condition. We performed a systematic assessment of all English-language rapid/short-acting insulin dose calculators available for iOS and Android. Searches identified 46 calculators that performed simple mathematical operations using planned carbohydrate intake and measured blood glucose. While 59% (n = 27/46) of apps included a clinical disclaimer, only 30% (n = 14/46) documented the calculation formula. 91% (n = 42/46) lacked numeric input validation, 59% (n = 27/46) allowed calculation when one or more values were missing, 48% (n = 22/46) used ambiguous terminology, 9% (n = 4/46) did not use adequate numeric precision and 4% (n = 2/46) did not store parameters faithfully. 67% (n = 31/46) of apps carried a risk of inappropriate output dose recommendation that either violated basic clinical assumptions (48%, n = 22/46) or did not match a stated formula (14%, n = 3/21) or correctly update in response to changing user inputs (37%, n = 17/46). Only one app, for iOS, was issue-free according to our criteria. No significant differences were observed in issue prevalence by payment model or platform. The majority of insulin dose calculator apps provide no protection against, and may actively contribute to, incorrect or inappropriate dose recommendations that put current users at risk of both catastrophic overdose and more
Benchmarking analytical calculations of proton doses in heterogeneous matter.
Ciangaru, George; Polf, Jerimy C; Bues, Martin; Smith, Alfred R
2005-12-01
A proton dose computational algorithm, performing an analytical superposition of infinitely narrow proton beamlets (ASPB) is introduced. The algorithm uses the standard pencil beam technique of laterally distributing the central axis broad beam doses according to the Moliere scattering theory extended to slablike varying density media. The purpose of this study was to determine the accuracy of our computational tool by comparing it with experimental and Monte Carlo (MC) simulation data as benchmarks. In the tests, parallel wide beams of protons were scattered in water phantoms containing embedded air and bone materials with simple geometrical forms and spatial dimensions of a few centimeters. For homogeneous water and bone phantoms, the proton doses we calculated with the ASPB algorithm were found very comparable to experimental and MC data. For layered bone slab inhomogeneity in water, the comparison between our analytical calculation and the MC simulation showed reasonable agreement, even when the inhomogeneity was placed at the Bragg peak depth. There also was reasonable agreement for the parallelepiped bone block inhomogeneity placed at various depths, except for cases in which the bone was located in the region of the Bragg peak, when discrepancies were as large as more than 10%. When the inhomogeneity was in the form of abutting air-bone slabs, discrepancies of as much as 8% occurred in the lateral dose profiles on the air cavity side of the phantom. Additionally, the analytical depth-dose calculations disagreed with the MC calculations within 3% of the Bragg peak dose, at the entry and midway depths in the phantom. The distal depth-dose 20%-80% fall-off widths and ranges calculated with our algorithm and the MC simulation were generally within 0.1 cm of agreement. The analytical lateral-dose profile calculations showed smaller (by less than 0.1 cm) 20%-80% penumbra widths and shorter fall-off tails than did those calculated by the MC simulations. Overall
Patient-specific dose calculation methods for high-dose-rate iridium-192 brachytherapy
Poon, Emily S.
In high-dose-rate 192Ir brachytherapy, the radiation dose received by the patient is calculated according to the AAPM Task Group 43 (TG-43) formalism. This table-based dose superposition method uses dosimetry parameters derived with the radioactive 192Ir source centered in a water phantom. It neglects the dose perturbations caused by inhomogeneities, such as the patient anatomy, applicators, shielding, and radiographic contrast solution. In this work, we evaluated the dosimetric characteristics of a shielded rectal applicator with an endocavitary balloon injected with contrast solution. The dose distributions around this applicator were calculated by the GEANT4 Monte Carlo (MC) code and measured by ionization chamber and GAFCHROMIC EBT film. A patient-specific dose calculation study was then carried out for 40 rectal treatment plans. The PTRAN_CT MC code was used to calculate the dose based on computed tomography (CT) images. This study involved the development of BrachyGUI, an integrated treatment planning tool that can process DICOM-RT data and create PTRAN_CT input initialization files. BrachyGUI also comes with dose calculation and evaluation capabilities. We proposed a novel scatter correction method to account for the reduction in backscatter radiation near tissue-air interfaces. The first step requires calculating the doses contributed by primary and scattered photons separately, assuming a full scatter environment. The scatter dose in the patient is subsequently adjusted using a factor derived by MC calculations, which depends on the distances between the point of interest, the 192Ir source, and the body contour. The method was validated for multicatheter breast brachytherapy, in which the target and skin doses for 18 patient plans agreed with PTRAN_CT calculations better than 1%. Finally, we developed a CT-based analytical dose calculation method. It corrects for the photon attenuation and scatter based upon the radiological paths determined by ray tracing
Electron mobility calculation for graphene on substrates
Hirai, Hideki; Ogawa, Matsuto [Department of Electrical and Electronic Engineering, Graduate School of Engineering, Kobe University, 1-1, Rokko-dai, Nada-ku, Kobe 657-8501 (Japan); Tsuchiya, Hideaki, E-mail: tsuchiya@eedept.kobe-u.ac.jp [Department of Electrical and Electronic Engineering, Graduate School of Engineering, Kobe University, 1-1, Rokko-dai, Nada-ku, Kobe 657-8501 (Japan); Japan Science and Technology Agency, CREST, Chiyoda, Tokyo 102-0075 (Japan); Kamakura, Yoshinari; Mori, Nobuya [Japan Science and Technology Agency, CREST, Chiyoda, Tokyo 102-0075 (Japan); Division of Electrical, Electronic and Information Engineering, Graduate School of Engineering, Osaka University, Suita, Osaka 565-0871 (Japan)
2014-08-28
By a semiclassical Monte Carlo method, the electron mobility in graphene is calculated for three different substrates: SiO{sub 2}, HfO{sub 2}, and hexagonal boron nitride (h-BN). The calculations account for polar and non-polar surface optical phonon (OP) scatterings induced by the substrates and charged impurity (CI) scattering, in addition to intrinsic phonon scattering in pristine graphene. It is found that HfO{sub 2} is unsuitable as a substrate, because the surface OP scattering of the substrate significantly degrades the electron mobility. The mobility on the SiO{sub 2} and h-BN substrates decreases due to CI scattering. However, the mobility on the h-BN substrate exhibits a high electron mobility of 170 000 cm{sup 2}/(V·s) for electron densities less than 10{sup 12 }cm{sup −2}. Therefore, h-BN should be an appealing substrate for graphene devices, as confirmed experimentally.
GTV-based prescription in SBRT for lung lesions using advanced dose calculation algorithms.
Lacornerie, Thomas; Lisbona, Albert; Mirabel, Xavier; Lartigau, Eric; Reynaert, Nick
2014-10-16
The aim of current study was to investigate the way dose is prescribed to lung lesions during SBRT using advanced dose calculation algorithms that take into account electron transport (type B algorithms). As type A algorithms do not take into account secondary electron transport, they overestimate the dose to lung lesions. Type B algorithms are more accurate but still no consensus is reached regarding dose prescription. The positive clinical results obtained using type A algorithms should be used as a starting point. In current work a dose-calculation experiment is performed, presenting different prescription methods. Three cases with three different sizes of peripheral lung lesions were planned using three different treatment platforms. For each individual case 60 Gy to the PTV was prescribed using a type A algorithm and the dose distribution was recalculated using a type B algorithm in order to evaluate the impact of the secondary electron transport. Secondly, for each case a type B algorithm was used to prescribe 48 Gy to the PTV, and the resulting doses to the GTV were analyzed. Finally, prescriptions based on specific GTV dose volumes were evaluated. When using a type A algorithm to prescribe the same dose to the PTV, the differences regarding median GTV doses among platforms and cases were always less than 10% of the prescription dose. The prescription to the PTV based on type B algorithms, leads to a more important variability of the median GTV dose among cases and among platforms, (respectively 24%, and 28%). However, when 54 Gy was prescribed as median GTV dose, using a type B algorithm, the variability observed was minimal. Normalizing the prescription dose to the median GTV dose for lung lesions avoids variability among different cases and treatment platforms of SBRT when type B algorithms are used to calculate the dose. The combination of using a type A algorithm to optimize a homogeneous dose in the PTV and using a type B algorithm to prescribe the
Analytical probabilistic proton dose calculation and range uncertainties
Bangert, M.; Hennig, P.; Oelfke, U.
2014-03-01
We introduce the concept of analytical probabilistic modeling (APM) to calculate the mean and the standard deviation of intensity-modulated proton dose distributions under the influence of range uncertainties in closed form. For APM, range uncertainties are modeled with a multivariate Normal distribution p(z) over the radiological depths z. A pencil beam algorithm that parameterizes the proton depth dose d(z) with a weighted superposition of ten Gaussians is used. Hence, the integrals ∫ dz p(z) d(z) and ∫ dz p(z) d(z)2 required for the calculation of the expected value and standard deviation of the dose remain analytically tractable and can be efficiently evaluated. The means μk, widths δk, and weights ωk of the Gaussian components parameterizing the depth dose curves are found with least squares fits for all available proton ranges. We observe less than 0.3% average deviation of the Gaussian parameterizations from the original proton depth dose curves. Consequently, APM yields high accuracy estimates for the expected value and standard deviation of intensity-modulated proton dose distributions for two dimensional test cases. APM can accommodate arbitrary correlation models and account for the different nature of random and systematic errors in fractionated radiation therapy. Beneficial applications of APM in robust planning are feasible.
A simplified analytical random walk model for proton dose calculation
Yao, Weiguang; Merchant, Thomas E.; Farr, Jonathan B.
2016-10-01
We propose an analytical random walk model for proton dose calculation in a laterally homogeneous medium. A formula for the spatial fluence distribution of primary protons is derived. The variance of the spatial distribution is in the form of a distance-squared law of the angular distribution. To improve the accuracy of dose calculation in the Bragg peak region, the energy spectrum of the protons is used. The accuracy is validated against Monte Carlo simulation in water phantoms with either air gaps or a slab of bone inserted. The algorithm accurately reflects the dose dependence on the depth of the bone and can deal with small-field dosimetry. We further applied the algorithm to patients’ cases in the highly heterogeneous head and pelvis sites and used a gamma test to show the reasonable accuracy of the algorithm in these sites. Our algorithm is fast for clinical use.
A corrector for spacecraft calculated electron moments
J. Geach
2005-03-01
Full Text Available We present the application of a numerical method to correct electron moments calculated on-board spacecraft from the effects of potential broadening and energy range truncation. Assuming a shape for the natural distribution of the ambient plasma and employing the scalar approximation, the on-board moments can be represented as non-linear integral functions of the underlying distribution. We have implemented an algorithm which inverts this system successfully over a wide range of parameters for an assumed underlying drifting Maxwellian distribution. The outputs of the solver are the corrected electron plasma temperature T_{e}, density N_{e} and velocity vector V_{e}. We also make an estimation of the temperature anisotropy A of the distribution. We present corrected moment data from Cluster's PEACE experiment for a range of plasma environments and make comparisons with electron and ion data from other Cluster instruments, as well as the equivalent ground-based calculations using full 3-D distribution PEACE telemetry.
Internal dose conversion factors for calculation of dose to the public
1988-07-01
This publication contains 50-year committed dose equivalent factors, in tabular form. The document is intended to be used as the primary reference by the US Department of Energy (DOE) and its contractors for calculating radiation dose equivalents for members of the public, resulting from ingestion or inhalation of radioactive materials. Its application is intended specifically for such materials released to the environment during routine DOE operations, except in those instances where compliance with 40 CFR 61 (National Emission Standards for Hazardous Air Pollutants) requires otherwise. However, the calculated values may be equally applicable to unusual releases or to occupational exposures. The use of these committed dose equivalent tables should ensure that doses to members of the public from internal exposures are calculated in a consistent manner at all DOE facilities.
Atomic Reference Data for Electronic Structure Calculations
Kotochigova, S; Shirley, E L
We have generated data for atomic electronic structure calculations, to provide a standard reference for results of specified accuracy under commonly used approximations. Results are presented here for total energies and orbital energy eigenvalues for all atoms from H to U, at microHartree accuracy in the total energy, as computed in the local-density approximation (LDA) the local-spin-density approximation (LSD); the relativistic local-density approximation (RLDA); and scalar-relativistic local-density approximation (ScRLDA).
A convolution-superposition dose calculation engine for GPUs
Hissoiny, Sami; Ozell, Benoit; Despres, Philippe [Departement de genie informatique et genie logiciel, Ecole polytechnique de Montreal, 2500 Chemin de Polytechnique, Montreal, Quebec H3T 1J4 (Canada); Departement de radio-oncologie, CRCHUM-Centre hospitalier de l' Universite de Montreal, 1560 rue Sherbrooke Est, Montreal, Quebec H2L 4M1 (Canada)
2010-03-15
Purpose: Graphic processing units (GPUs) are increasingly used for scientific applications, where their parallel architecture and unprecedented computing power density can be exploited to accelerate calculations. In this paper, a new GPU implementation of a convolution/superposition (CS) algorithm is presented. Methods: This new GPU implementation has been designed from the ground-up to use the graphics card's strengths and to avoid its weaknesses. The CS GPU algorithm takes into account beam hardening, off-axis softening, kernel tilting, and relies heavily on raytracing through patient imaging data. Implementation details are reported as well as a multi-GPU solution. Results: An overall single-GPU acceleration factor of 908x was achieved when compared to a nonoptimized version of the CS algorithm implemented in PlanUNC in single threaded central processing unit (CPU) mode, resulting in approximatively 2.8 s per beam for a 3D dose computation on a 0.4 cm grid. A comparison to an established commercial system leads to an acceleration factor of approximately 29x or 0.58 versus 16.6 s per beam in single threaded mode. An acceleration factor of 46x has been obtained for the total energy released per mass (TERMA) calculation and a 943x acceleration factor for the CS calculation compared to PlanUNC. Dose distributions also have been obtained for a simple water-lung phantom to verify that the implementation gives accurate results. Conclusions: These results suggest that GPUs are an attractive solution for radiation therapy applications and that careful design, taking the GPU architecture into account, is critical in obtaining significant acceleration factors. These results potentially can have a significant impact on complex dose delivery techniques requiring intensive dose calculations such as intensity-modulated radiation therapy (IMRT) and arc therapy. They also are relevant for adaptive radiation therapy where dose results must be obtained rapidly.
Prenatal radiation exposure. Dose calculation; Praenatale Strahlenexposition. Dosisermittlung
Scharwaechter, C.; Schwartz, C.A.; Haage, P. [University Hospital Witten/Herdecke, Wuppertal (Germany). Dept. of Diagnostic and Interventional Radiology; Roeser, A. [University Hospital Witten/Herdecke, Wuppertal (Germany). Dept. of Radiotherapy and Radio-Oncology
2015-05-15
The unborn child requires special protection. In this context, the indication for an X-ray examination is to be checked critically. If thereupon radiation of the lower abdomen including the uterus cannot be avoided, the examination should be postponed until the end of pregnancy or alternative examination techniques should be considered. Under certain circumstances, either accidental or in unavoidable cases after a thorough risk assessment, radiation exposure of the unborn may take place. In some of these cases an expert radiation hygiene consultation may be required. This consultation should comprise the expected risks for the unborn while not perturbing the mother or the involved medical staff. For the risk assessment in case of an in-utero X-ray exposition deterministic damages with a defined threshold dose are distinguished from stochastic damages without a definable threshold dose. The occurrence of deterministic damages depends on the dose and the developmental stage of the unborn at the time of radiation. To calculate the risks of an in-utero radiation exposure a three-stage concept is commonly applied. Depending on the amount of radiation, the radiation dose is either estimated, roughly calculated using standard tables or, in critical cases, accurately calculated based on the individual event. The complexity of the calculation thereby increases from stage to stage. An estimation based on stage one is easily feasible whereas calculations based on stages two and especially three are more complex and often necessitate execution by specialists. This article demonstrates in detail the risks for the unborn child pertaining to its developmental phase and explains the three-stage concept as an evaluation scheme. It should be noted, that all risk estimations are subject to considerable uncertainties.
Calculation of fractional electron capture probabilities
Schoenfeld, E
1998-01-01
A 'Table of Radionuclides' is being prepared which will supersede the 'Table de Radionucleides' formerly issued by the LMRI/LPRI (France). In this effort it is desirable to have a uniform basis for calculating theoretical values of fractional electron capture probabilities. A table has been compiled which allows one to calculate conveniently and quickly the fractional probabilities P sub K , P sub L , P sub M , P sub N and P sub O , their ratios and the assigned uncertainties for allowed and non-unique first forbidden electron capture transitions of known transition energy for radionuclides with atomic numbers from Z=3 to 102. These results have been applied to a total of 28 transitions of 14 radionuclides ( sup 7 Be, sup 2 sup 2 Na, sup 5 sup 1 Cr, sup 5 sup 4 Mn, sup 5 sup 5 Fe, sup 6 sup 8 Ge , sup 6 sup 8 Ga, sup 7 sup 5 Se, sup 1 sup 0 sup 9 Cd, sup 1 sup 2 sup 5 I, sup 1 sup 3 sup 9 Ce, sup 1 sup 6 sup 9 Yb, sup 1 sup 9 sup 7 Hg, sup 2 sup 0 sup 2 Tl). The values are in reasonable agreement with measure...
Christiansen, Rasmus Lübeck; Jensen, Henrik R.; Brink, Carsten
2017-01-01
Background: Current state of the art radiotherapy planning of prostate cancer utilises magnetic resonance (MR) for soft tissue delineation and computed tomography (CT) to provide an electron density map for dose calculation. This dual scan workflow is prone to setup and registration error....... This study evaluates the feasibility of an MR-only workflow and the validity of dose calculation from an MR derived pseudo CT. Material and methods: Thirty prostate cancer patients were CT and MR scanned. Clinical treatment plans were generated on CT using a single 18 MV arc volumetric modulated arc therapy...
Modeling Considerations for Ingestion Pathway Dose Calculations Using CAP88.
Stuenkel, David
2017-04-01
The CAP88-PC computer model was developed by the U.S. Environmental Protection Agency to demonstrate compliance under the National Emission Standards for Hazardous Air Pollutants (NESHAPS). The program combines atmospheric transport models with the terrestrial food chain models in the U.S. Nuclear Regulatory Commission Regulatory Guide 1.109 to compute the radionuclide concentrations in the air, on ground surfaces and plants, and the concentrations in food to estimate the dose to individuals living in the area around a facility emitting radionuclides into the atmosphere. CAP88 allows the user to select the size of the assessment area and the receptor locations used to calculate the radionuclide concentrations in non-leafy vegetables, leafy vegetables, milk, and meat consumed by the receptors. Depending on the food scenario selected and the type of calculation ("Population" or "Individual") chosen, the annual effective dose from ingestion can depend on both the size of the assessment area and the location of the receptors. Illustrative examples demonstrate the effect of the choice of these input parameters on the annual effective dose from ingestion. An understanding of the model used in CAP88 and the differences between "Population" and "Individual" run types will enable the CAP88 user to better model the ingestion dose.
Intensity modulation with electrons: calculations, measurements and clinical applications.
Karlsson, M G; Karlsson, M; Zackrisson, B
1998-05-01
Intensity modulation of electron beams is one step towards truly conformal therapy. This can be realized with the MM50 racetrack microtron that utilizes a scanning beam technique. By adjusting the scan pattern it is possible to obtain arbitrary fluence distributions. Since the monitor chambers in the treatment head are segmented in both x- and y-directions it is possible to verify the fluence distribution to the patient at any time during the treatment. Intensity modulated electron beams have been measured with film and a plane parallel chamber and compared with calculations. The calculations were based on a pencil beam method. An intensity distribution at the multileaf collimator (MLC) level was calculated by superposition of measured pencil beams over scan patterns. By convolving this distribution with a Gaussian pencil beam, which has propagated from the MLC to the isocentre, a fluence distribution at isocentre level was obtained. The agreement between calculations and measurements was within 2% in dose or 1 mm in distance in the penumbra zones. A standard set of intensity modulated electron beams has been developed. These beams have been implemented in a treatment planning system and are used for manual optimization. A clinical example (prostate) of such an application is presented and compared with a standard irradiation technique.
Intensity modulation with electrons: calculations, measurements and clinical applications
Karlsson, Magnus G.; Karlsson, Mikael [Department of Radiation Physics, Umeaa University, S-901 85 Umeaa (Sweden); Zackrisson, Bjoern [Department of Oncology, Umeaa University, S-901 85 Umeaa (Sweden)
1998-05-01
Intensity modulation of electron beams is one step towards truly conformal therapy. This can be realized with the MM50 racetrack microtron that utilizes a scanning beam technique. By adjusting the scan pattern it is possible to obtain arbitrary fluence distributions. Since the monitor chambers in the treatment head are segmented in both x- and y-directions it is possible to verify the fluence distribution to the patient at any time during the treatment. Intensity modulated electron beams have been measured with film and a plane parallel chamber and compared with calculations. The calculations were based on a pencil beam method. An intensity distribution at the multileaf collimator (MLC) level was calculated by superposition of measured pencil beams over scan patterns. By convolving this distribution with a Gaussian pencil beam, which has propagated from the MLC to the isocentre, a fluence distribution at isocentre level was obtained. The agreement between calculations and measurements was within 2% in dose or 1 mm in distance in the penumbra zones. A standard set of intensity modulated electron beams has been developed. These beams have been implemented in a treatment planning system and are used for manual optimization. A clinical example (prostate) of such an application is presented and compared with a standard irradiation technique. (author)
Yang, Bo; Qiu, Rui; Li, JunLi; Lu, Wei; Wu, Zhen; Li, Chunyan
2017-02-01
When a strong laser beam irradiates a solid target, a hot plasma is produced and high-energy electrons are usually generated (the so-called "hot electrons"). These energetic electrons subsequently generate hard X-rays in the solid target through the Bremsstrahlung process. To date, only limited studies have been conducted on this laser-induced radiological protection issue. In this study, extensive literature reviews on the physics and properties of hot electrons have been conducted. On the basis of these information, the photon dose generated by the interaction between hot electrons and a solid target was simulated with the Monte Carlo code FLUKA. With some reasonable assumptions, the calculated dose can be regarded as the upper boundary of the experimental results over the laser intensity ranging from 1019 to 1021 W/cm2. Furthermore, an equation to estimate the photon dose generated from ultraintense laser-solid interactions based on the normalized laser intensity is derived. The shielding effects of common materials including concrete and lead were also studied for the laser-driven X-ray source. The dose transmission curves and tenth-value layers (TVLs) in concrete and lead were calculated through Monte Carlo simulations. These results could be used to perform a preliminary and fast radiation safety assessment for the X-rays generated from ultraintense laser-solid interactions.
In-air fluence profiles and water depth dose for uncollimated electron beams
Toutaoui Abdelkader
2008-01-01
Full Text Available Advanced electron beam dose calculation models for radiation treatment planning systems require the input of a phase space beam model to configure a clinical electron beam in a computer. This beam model is a distribution in position, energy, and direction of electrons and photons in a plane in front of the patient. The phase space beam model can be determined by Monte Carlo simulation of the treatment head or from a limited set of measurements. In the latter case, parameters of the electron phase space beam model are obtained by fitting measured to calculated dosimetric data. In the present work, data for air fluence profiles and water depth doses have been presented for electron beams without an applicator for a medical linear accelerator. These data are used to parameterize the electron phase space beam model to a Monte Carlo dose calculation module available in the first commercial (MDS Nordion, now Nucletron Monte Carlo treatment planning for electron beams.
In-air fluence profiles and water depth dose for uncollimated electron beams
Toutaoui, Abedelkadar; Aichouche, Amar Nassim; Adjidir, Kenza Adjidir; Chami, Ahmed Chafik
2008-01-01
Advanced electron beam dose calculation models for radiation treatment planning systems require the input of a phase space beam model to configure a clinical electron beam in a computer. This beam model is a distribution in position, energy, and direction of electrons and photons in a plane in front of the patient. The phase space beam model can be determined by Monte Carlo simulation of the treatment head or from a limited set of measurements. In the latter case, parameters of the electron phase space beam model are obtained by fitting measured to calculated dosimetric data. In the present work, data for air fluence profiles and water depth doses have been presented for electron beams without an applicator for a medical linear accelerator. These data are used to parameterize the electron phase space beam model to a Monte Carlo dose calculation module available in the first commercial (MDS Nordion, now Nucletron) Monte Carlo treatment planning for electron beams. PMID:19893707
A single-source photon source model of a linear accelerator for Monte Carlo dose calculation.
Nwankwo, Obioma; Glatting, Gerhard; Wenz, Frederik; Fleckenstein, Jens
2017-01-01
To introduce a new method of deriving a virtual source model (VSM) of a linear accelerator photon beam from a phase space file (PSF) for Monte Carlo (MC) dose calculation. A PSF of a 6 MV photon beam was generated by simulating the interactions of primary electrons with the relevant geometries of a Synergy linear accelerator (Elekta AB, Stockholm, Sweden) and recording the particles that reach a plane 16 cm downstream the electron source. Probability distribution functions (PDFs) for particle positions and energies were derived from the analysis of the PSF. These PDFs were implemented in the VSM using inverse transform sampling. To model particle directions, the phase space plane was divided into a regular square grid. Each element of the grid corresponds to an area of 1 mm2 in the phase space plane. The average direction cosines, Pearson correlation coefficient (PCC) between photon energies and their direction cosines, as well as the PCC between the direction cosines were calculated for each grid element. Weighted polynomial surfaces were then fitted to these 2D data. The weights are used to correct for heteroscedasticity across the phase space bins. The directions of the particles created by the VSM were calculated from these fitted functions. The VSM was validated against the PSF by comparing the doses calculated by the two methods for different square field sizes. The comparisons were performed with profile and gamma analyses. The doses calculated with the PSF and VSM agree to within 3% /1 mm (>95% pixel pass rate) for the evaluated fields. A new method of deriving a virtual photon source model of a linear accelerator from a PSF file for MC dose calculation was developed. Validation results show that the doses calculated with the VSM and the PSF agree to within 3% /1 mm.
Zhang, Aizhen; Wen, Ning; Nurushev, Teamour; Burmeister, Jay; Chetty, Indrin J
2013-03-04
A commercial electron Monte Carlo (eMC) dose calculation algorithm has become available in Eclipse treatment planning system. The purpose of this work was to evaluate the eMC algorithm and investigate the clinical implementation of this system. The beam modeling of the eMC algorithm was performed for beam energies of 6, 9, 12, 16, and 20 MeV for a Varian Trilogy and all available applicator sizes in the Eclipse treatment planning system. The accuracy of the eMC algorithm was evaluated in a homogeneous water phantom, solid water phantoms containing lung and bone materials, and an anthropomorphic phantom. In addition, dose calculation accuracy was compared between pencil beam (PB) and eMC algorithms in the same treatment planning system for heterogeneous phantoms. The overall agreement between eMC calculations and measurements was within 3%/2 mm, while the PB algorithm had large errors (up to 25%) in predicting dose distributions in the presence of inhomogeneities such as bone and lung. The clinical implementation of the eMC algorithm was investigated by performing treatment planning for 15 patients with lesions in the head and neck, breast, chest wall, and sternum. The dose distributions were calculated using PB and eMC algorithms with no smoothing and all three levels of 3D Gaussian smoothing for comparison. Based on a routine electron beam therapy prescription method, the number of eMC calculated monitor units (MUs) was found to increase with increased 3D Gaussian smoothing levels. 3D Gaussian smoothing greatly improved the visual usability of dose distributions and produced better target coverage. Differences of calculated MUs and dose distributions between eMC and PB algorithms could be significant when oblique beam incidence, surface irregularities, and heterogeneous tissues were present in the treatment plans. In our patient cases, monitor unit differences of up to 7% were observed between PB and eMC algorithms. Monitor unit calculations were also preformed
Kanematsu, Nobuyuki
2007-01-01
A simple and efficient variant of the pencil-beam algorithm for dose distribution calculation is proposed. Compared to the conventional pencil-beam algorithms, the new algorithm is intrinsically faster due to minimized computation within the convolution integral. Namely, computation for physical interaction is decoupled from the convolution integral and the convolution kernel is approximated by simple grid-to-grid correlation. Implementation to a treatment planning system for carbon-ion radiotherapy has enabled realistic beam blurring with marginal speed decrease from the broad-beam calculation. Evaluation of a modeled proton pencil beam exhibits inaccuracy within its spread at the Bragg peak when the beam incidence is angled to all the dose grid axes, which will be minimized in broad-beam formation and may be acceptable depending on its relative significance to the other sources of errors. The new algorithm will provide balanced accuracy and speed without technical difficulty for high-resolution dose distrib...
Monte Carlo calculations of the depth-dose distribution in skin contaminated by hot particles
Patau, J.-P. (Toulouse-3 Univ., 31 (France))
1991-01-01
Accurate computer programs were developed in order to calculate the spatial distribution of absorbed radiation doses in the skin, near high activity particles (''hot particles''). With a view to ascertaining the reliability of the codes the transport of beta particles was simulated in a complex configuration used for dosimetric measurements: spherical {sup 60}Co sources of 10-1000 {mu}m fastened to an aluminium support with a tissue-equivalent adhesive overlaid with 10 {mu}m thick aluminium foil. Behind it an infinite polystyrene medium including an extrapolation chamber was assumed. The exact energy spectrum of beta emission was sampled. Production and transport of secondary knock-on electrons were also simulated. Energy depositions in polystyrene were calculated with a high spatial resolution. Finally, depth-dose distributions were calculated for hot particles placed on the skin. The calculations will be continued for other radionuclides and for a configuration suited to TLD measurements. (author).
Plante, I.; Cucinotta, F. A.
2010-01-01
INTRODUCTION: The radiation track structure is of crucial importance to understand radiation damage to molecules and subsequent biological effects. Of a particular importance in radiobiology is the induction of double-strand breaks (DSBs) by ionizing radiation, which are caused by clusters of lesions in DNA, and oxidative damage to cellular constituents leading to aberrant signaling cascades. DSB can be visualized within cell nuclei with gamma-H2AX experiments. MATERIAL AND METHODS: In DSB induction models, the DSB probability is usually calculated by the local dose obtained from a radial dose profile of HZE tracks. In this work, the local dose imparted by HZE ions is calculated directly from the 3D Monte-Carlo simulation code RITRACKS. A cubic volume of 5 micron edge (Figure 1) is irradiated by a (Fe26+)-56 ion of 1 GeV/amu (LET approx.150 keV/micron) and by a fluence of 450 H+ ions, 300 MeV/amu (LET approx. 0.3 keV/micron). In both cases, the dose deposited in the volume is approx.1 Gy. The dose is then calculated into each 3D pixels (voxels) of 20 nm edge and visualized in 3D. RESULTS AND DISCUSSION: The dose is deposited uniformly in the volume by the H+ ions. The voxels which receive a high dose (orange) corresponds to electron track ends. The dose is deposited differently by the 56Fe26+ ion. Very high dose (red) is deposited in voxels with direct ion traversal. Voxels with electron track ends (orange) are also found distributed around the path of the track. In both cases, the appearance of the dose distribution looks very similar to DSBs seen in gammaH2AX experiments, particularly when the visualization threshold is applied. CONCLUSION: The refinement of the dose calculation to the nanometer scale has revealed important differences in the energy deposition between high- and low-LET ions. Voxels of very high dose are only found in the path of high-LET ions. Interestingly, experiments have shown that DSB induced by high-LET radiation are more difficult to
Dose-rate conversion factors for external exposure to photons and electrons
Kocher, D.C.
1981-08-01
Dose-rate conversion factors for external exposure to photons and electrons have been calculated for approximately 500 radionuclides of potential importance in environmental radiological assessments. The dose-rate factors were obtained using the DOSFACTER computer code. The results given in this report incorporate calculation of electron dose-rate factors for radiosensitive tissues of the skin, improved estimates of organ dose-rate factors for photons, based on organ doses for monoenergetic sources at the body surface of an exposed individual, and the spectra of scattered photons in air from monoenergetic sources in an infinite, uniformly contaminated atmospheric cloud, calculation of dose-rate factors for other radionuclides in addition to those of interest in the nuclear fuel cycle, and incorporation of updated radioactive decay data for all radionuclides. Dose-rate factors are calculated for three exposure modes - immersion in contaminated air, immersion in contaminated water, and exposure at a height of 1 m above a contaminated ground surface. The report presents the equations used to calculate the external dose-rate factors for photons and electrons, documentation of the revised DOSFACTER computer code, and a complete tabulation of the calculated dose-rate factors. 30 refs., 12 figs.
Effects of physics change in Monte Carlo code on electron pencil beam dose distributions
Toutaoui, Abdelkader, E-mail: toutaoui.aek@gmail.com [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Khelassi-Toutaoui, Nadia, E-mail: nadiakhelassi@yahoo.fr [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Brahimi, Zakia, E-mail: zsbrahimi@yahoo.fr [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Chami, Ahmed Chafik, E-mail: chafik_chami@yahoo.fr [Laboratoire de Sciences Nucleaires, Faculte de Physique, Universite des Sciences et de la Technologie Houari Boumedienne, BP 32 El Alia, Bab Ezzouar, Algiers (Algeria)
2012-01-15
Pencil beam algorithms used in computerized electron beam dose planning are usually described using the small angle multiple scattering theory. Alternatively, the pencil beams can be generated by Monte Carlo simulation of electron transport. In a previous work, the 4th version of the Electron Gamma Shower (EGS) Monte Carlo code was used to obtain dose distributions from monoenergetic electron pencil beam, with incident energy between 1 MeV and 50 MeV, interacting at the surface of a large cylindrical homogeneous water phantom. In 2000, a new version of this Monte Carlo code has been made available by the National Research Council of Canada (NRC), which includes various improvements in its electron-transport algorithms. In the present work, we were interested to see if the new physics in this version produces pencil beam dose distributions very different from those calculated with oldest one. The purpose of this study is to quantify as well as to understand these differences. We have compared a series of pencil beam dose distributions scored in cylindrical geometry, for electron energies between 1 MeV and 50 MeV calculated with two versions of the Electron Gamma Shower Monte Carlo Code. Data calculated and compared include isodose distributions, radial dose distributions and fractions of energy deposition. Our results for radial dose distributions show agreement within 10% between doses calculated by the two codes for voxels closer to the pencil beam central axis, while the differences are up to 30% for longer distances. For fractions of energy deposition, the results of the EGS4 are in good agreement (within 2%) with those calculated by EGSnrc at shallow depths for all energies, whereas a slightly worse agreement (15%) is observed at deeper distances. These differences may be mainly attributed to the different multiple scattering for electron transport adopted in these two codes and the inclusion of spin effect, which produces an increase of the effective range of
Implementation of spot scanning dose optimization and dose calculation for helium ions in Hyperion
Fuchs, Hermann, E-mail: hermann.fuchs@meduniwien.ac.at [Department of Radiation Oncology, Division of Medical Radiation Physics, Medical University of Vienna/AKH Vienna, Vienna 1090, Austria and Christian Doppler Laboratory for Medical Radiation Research for Radiation Oncology, Medical University of Vienna, Vienna 1090 (Austria); Alber, Markus [Department for Oncology, Aarhus University Hospital, Aarhus 8000 (Denmark); Schreiner, Thomas [PEG MedAustron, Wiener Neustadt 2700 (Austria); Georg, Dietmar [Department of Radiation Oncology, Division of Medical Radiation Physics, Medical University of Vienna/AKH Vienna, Vienna 1090 (Austria); Christian Doppler Laboratory for Medical Radiation Research for Radiation Oncology, Medical University of Vienna, Vienna 1090 (Austria); Comprehensive Cancer Center, Medical University of Vienna/AKH Vienna, Vienna 1090 (Austria)
2015-09-15
Purpose: Helium ions ({sup 4}He) may supplement current particle beam therapy strategies as they possess advantages in physical dose distribution over protons. To assess potential clinical advantages, a dose calculation module accounting for relative biological effectiveness (RBE) was developed and integrated into the treatment planning system Hyperion. Methods: Current knowledge on RBE of {sup 4}He together with linear energy transfer considerations motivated an empirical depth-dependent “zonal” RBE model. In the plateau region, a RBE of 1.0 was assumed, followed by an increasing RBE up to 2.8 at the Bragg-peak region, which was then kept constant over the fragmentation tail. To account for a variable proton RBE, the same model concept was also applied to protons with a maximum RBE of 1.6. Both RBE models were added to a previously developed pencil beam algorithm for physical dose calculation and included into the treatment planning system Hyperion. The implementation was validated against Monte Carlo simulations within a water phantom using γ-index evaluation. The potential benefits of {sup 4}He based treatment plans were explored in a preliminary treatment planning comparison (against protons) for four treatment sites, i.e., a prostate, a base-of-skull, a pediatric, and a head-and-neck tumor case. Separate treatment plans taking into account physical dose calculation only or using biological modeling were created for protons and {sup 4}He. Results: Comparison of Monte Carlo and Hyperion calculated doses resulted in a γ{sub mean} of 0.3, with 3.4% of the values above 1 and γ{sub 1%} of 1.5 and better. Treatment plan evaluation showed comparable planning target volume coverage for both particles, with slightly increased coverage for {sup 4}He. Organ at risk (OAR) doses were generally reduced using {sup 4}He, some by more than to 30%. Improvements of {sup 4}He over protons were more pronounced for treatment plans taking biological effects into account. All
Quantification of confounding factors in MRI-based dose calculations as applied to prostate IMRT
Maspero, Matteo; Seevinck, Peter R.; Schubert, Gerald; Hoesl, Michaela A. U.; van Asselen, Bram; Viergever, Max A.; Lagendijk, Jan J. W.; Meijer, Gert J.; van den Berg, Cornelis A. T.
2017-02-01
Magnetic resonance (MR)-only radiotherapy treatment planning requires pseudo-CT (pCT) images to enable MR-based dose calculations. To verify the accuracy of MR-based dose calculations, institutions interested in introducing MR-only planning will have to compare pCT-based and computer tomography (CT)-based dose calculations. However, interpreting such comparison studies may be challenging, since potential differences arise from a range of confounding factors which are not necessarily specific to MR-only planning. Therefore, the aim of this study is to identify and quantify the contribution of factors confounding dosimetric accuracy estimation in comparison studies between CT and pCT. The following factors were distinguished: set-up and positioning differences between imaging sessions, MR-related geometric inaccuracy, pCT generation, use of specific calibration curves to convert pCT into electron density information, and registration errors. The study comprised fourteen prostate cancer patients who underwent CT/MRI-based treatment planning. To enable pCT generation, a commercial solution (MRCAT, Philips Healthcare, Vantaa, Finland) was adopted. IMRT plans were calculated on CT (gold standard) and pCTs. Dose difference maps in a high dose region (CTV) and in the body volume were evaluated, and the contribution to dose errors of possible confounding factors was individually quantified. We found that the largest confounding factor leading to dose difference was the use of different calibration curves to convert pCT and CT into electron density (0.7%). The second largest factor was the pCT generation which resulted in pCT stratified into a fixed number of tissue classes (0.16%). Inter-scan differences due to patient repositioning, MR-related geometric inaccuracy, and registration errors did not significantly contribute to dose differences (0.01%). The proposed approach successfully identified and quantified the factors confounding accurate MRI-based dose calculation in
GPU-based fast Monte Carlo simulation for radiotherapy dose calculation
Jia, Xun; Graves, Yan Jiang; Folkerts, Michael; Jiang, Steve B
2011-01-01
Monte Carlo (MC) simulation is commonly considered to be the most accurate dose calculation method in radiotherapy. However, its efficiency still requires improvement for many routine clinical applications. In this paper, we present our recent progress towards the development a GPU-based MC dose calculation package, gDPM v2.0. It utilizes the parallel computation ability of a GPU to achieve high efficiency, while maintaining the same particle transport physics as in the original DPM code and hence the same level of simulation accuracy. In GPU computing, divergence of execution paths between threads can considerably reduce the efficiency. Since photons and electrons undergo different physics and hence attain different execution paths, we use a simulation scheme where photon transport and electron transport are separated to partially relieve the thread divergence issue. High performance random number generator and hardware linear interpolation are also utilized. We have also developed various components to hand...
Patient-dependent beam-modifier physics in Monte Carlo photon dose calculations.
Schach von Wittenau, A E; Bergstrom, P M; Cox, L J
2000-05-01
Model pencil-beam on slab calculations are used as well as a series of detailed calculations of photon and electron output from commercial accelerators to quantify level(s) of physics required for the Monte Carlo transport of photons and electrons in treatment-dependent beam modifiers, such as jaws, wedges, blocks, and multileaf collimators, in photon teletherapy dose calculations. The physics approximations investigated comprise (1) not tracking particles below a given kinetic energy, (2) continuing to track particles, but performing simplified collision physics, particularly in handling secondary particle production, and (3) not tracking particles in specific spatial regions. Figures-of-merit needed to estimate the effects of these approximations are developed, and these estimates are compared with full-physics Monte Carlo calculations of the contribution of the collimating jaws to the on-axis depth-dose curve in a water phantom. These figures of merit are next used to evaluate various approximations used in coupled photon/electron physics in beam modifiers. Approximations for tracking electrons in air are then evaluated. It is found that knowledge of the materials used for beam modifiers, of the energies of the photon beams used, as well as of the length scales typically found in photon teletherapy plans, allows a number of simplifying approximations to be made in the Monte Carlo transport of secondary particles from the accelerator head and beam modifiers to the isocenter plane.
Investigation of Nonuniform Dose Voxel Geometry in Monte Carlo Calculations.
Yuan, Jiankui; Chen, Quan; Brindle, James; Zheng, Yiran; Lo, Simon; Sohn, Jason; Wessels, Barry
2015-08-01
The purpose of this work is to investigate the efficacy of using multi-resolution nonuniform dose voxel geometry in Monte Carlo (MC) simulations. An in-house MC code based on the dose planning method MC code was developed in C++ to accommodate the nonuniform dose voxel geometry package since general purpose MC codes use their own coupled geometry packages. We devised the package in a manner that the entire calculation volume was first divided into a coarse mesh and then the coarse mesh was subdivided into nonuniform voxels with variable voxel sizes based on density difference. We name this approach as multi-resolution subdivision (MRS). It generates larger voxels in small density gradient regions and smaller voxels in large density gradient regions. To take into account the large dose gradients due to the beam penumbra, the nonuniform voxels can be further split using ray tracing starting from the beam edges. The accuracy of the implementation of the algorithm was verified by comparing with the data published by Rogers and Mohan. The discrepancy was found to be 1% to 2%, with a maximum of 3% at the interfaces. Two clinical cases were used to investigate the efficacy of nonuniform voxel geometry in the MC code. Applying our MRS approach, we started with the initial voxel size of 5 × 5 × 3 mm(3), which was further divided into smaller voxels. The smallest voxel size was 1.25 × 1.25 × 3 mm(3). We found that the simulation time per history for the nonuniform voxels is about 30% to 40% faster than the uniform fine voxels (1.25 × 1.25 × 3 mm(3)) while maintaining similar accuracy.
GPU-based fast Monte Carlo simulation for radiotherapy dose calculation.
Jia, Xun; Gu, Xuejun; Graves, Yan Jiang; Folkerts, Michael; Jiang, Steve B
2011-11-21
Monte Carlo (MC) simulation is commonly considered to be the most accurate dose calculation method in radiotherapy. However, its efficiency still requires improvement for many routine clinical applications. In this paper, we present our recent progress toward the development of a graphics processing unit (GPU)-based MC dose calculation package, gDPM v2.0. It utilizes the parallel computation ability of a GPU to achieve high efficiency, while maintaining the same particle transport physics as in the original dose planning method (DPM) code and hence the same level of simulation accuracy. In GPU computing, divergence of execution paths between threads can considerably reduce the efficiency. Since photons and electrons undergo different physics and hence attain different execution paths, we use a simulation scheme where photon transport and electron transport are separated to partially relieve the thread divergence issue. A high-performance random number generator and a hardware linear interpolation are also utilized. We have also developed various components to handle the fluence map and linac geometry, so that gDPM can be used to compute dose distributions for realistic IMRT or VMAT treatment plans. Our gDPM package is tested for its accuracy and efficiency in both phantoms and realistic patient cases. In all cases, the average relative uncertainties are less than 1%. A statistical t-test is performed and the dose difference between the CPU and the GPU results is not found to be statistically significant in over 96% of the high dose region and over 97% of the entire region. Speed-up factors of 69.1 ∼ 87.2 have been observed using an NVIDIA Tesla C2050 GPU card against a 2.27 GHz Intel Xeon CPU processor. For realistic IMRT and VMAT plans, MC dose calculation can be completed with less than 1% standard deviation in 36.1 ∼ 39.6 s using gDPM.
Assessing medical students’ competence in calculating drug doses
Catherine Harries
2013-09-01
Full Text Available Evidence suggests that healthcare professionals are not optimally able to calculate medicine doses and various strategies have been employed to improve these skills. In this study, the performance of third and fourth year medical students was assessed and the success of various educational interventions investigated. Students were given four types of dosing calculations typical of those required in an emergency setting. Full competence (at the 100% level was defined as correctly answering all four categories of calculation at any one time. Three categories correct meant competence at the 75% level. Interventions comprised an assignment with a model answer for self-assessment in the third year and a small group tutorial in the fourth year. The small groups provided opportunities for peer-assisted learning. A subgroup of 23 students received individual tuition from the lecturer prior to the start of the fourth year. Amongst the 364 eligible students, full competence rose from 23% at the beginning of the third year to 66% by the end of the fourth year. More students succeeded during the fourth than the third year of study. Success of small group tuition was assessed in a sample of 200 students who had formal assessments both before and after the fourth year tuition. Competence at the 75% level improved by 10% in attendees and decreased by 3% in non-attendees, providing evidence of the value of students receiving assistance from more able same-language peers. Good results were achieved with one-on-one tuition where individualised assistance allowed even struggling students to improve.
The calculation of radial dose from heavy ions: predictions of biological action cross sections
Katz, Robert; Cucinotta, Francis A.; Zhang, C. X.
1996-02-01
The track structure model of heavy ion cross sections was developed by Katz and co-workers in the 1960s. In this model the action cross section is evaluated by mapping the dose-response of a detector to γ rays (modeled from biological target theory) onto the radial dose distribution from δ rays about the path of the ion. This is taken to yield the radial distribution of probability for a "hit" (an interaction leading to an observable end-point). Radial integration of the probability yields the cross section. When different response from ions of different Z having the same stopping power is observed this model may be indicated. Since the 1960s there have been several developments in the computation of the radial dose distribution, in the measurement of these distributions, and in new radiobiological data against which to test the model. The earliest model, by Butts and Katz, made use of simplified δ ray distribution functions, of simplified electron range-energy relations, and neglected angular distributions. Nevertheless it made possible the calculation of cross sections for the inactivation of enzymes and viruses, and allowed extension to tracks in nuclear emulsions and other detectors and to biological cells. It set the pattern for models of observable effects in the matter through which the ion passed. Here we outline subsequent calculations of radial dose which make use of improved knowledge of the electron emission spectrum, the electron range-energy relation, the angular distribution, and some considerations of molecular excitation, of particular interest both close to the path of the ion and the outer limits of electron penetration. These are applied to the modeling of action cross sections for the inactivation of several strains of E-coli and B. subtilis spores where extensive measurements in the "thin-down" region have been made with heavy ion beams. Such calculations serve to test the radial dose calculations at the outer limit of electron penetration
Strenge, D.L.; Peloquin, R.A.
1981-04-01
The computer code HADOC (Hanford Acute Dose Calculations) is described and instructions for its use are presented. The code calculates external dose from air submersion and inhalation doses following acute radionuclide releases. Atmospheric dispersion is calculated using the Hanford model with options to determine maximum conditions. Building wake effects and terrain variation may also be considered. Doses are calculated using dose conversion factor supplied in a data library. Doses are reported for one and fifty year dose commitment periods for the maximum individual and the regional population (within 50 miles). The fractional contribution to dose by radionuclide and exposure mode are also printed if requested.
Limitations of the TG-43 formalism for skin high-dose-rate brachytherapy dose calculations
Granero, Domingo, E-mail: dgranero@eresa.com [Department of Radiation Physics, ERESA, Hospital General Universitario, 46014 Valencia (Spain); Perez-Calatayud, Jose [Radiotherapy Department, La Fe University and Polytechnic Hospital, Valencia 46026 (Spain); Vijande, Javier [Department of Atomic, Molecular and Nuclear Physics, University of Valencia, Burjassot 46100, Spain and IFIC (UV-CSIC), Paterna 46980 (Spain); Ballester, Facundo [Department of Atomic, Molecular and Nuclear Physics, University of Valencia, Burjassot 46100 (Spain); Rivard, Mark J. [Department of Radiation Oncology, Tufts University School of Medicine, Boston, Massachusetts 02111 (United States)
2014-02-15
Purpose: In skin high-dose-rate (HDR) brachytherapy, sources are located outside, in contact with, or implanted at some depth below the skin surface. Most treatment planning systems use the TG-43 formalism, which is based on single-source dose superposition within an infinite water medium without accounting for the true geometry in which conditions for scattered radiation are altered by the presence of air. The purpose of this study is to evaluate the dosimetric limitations of the TG-43 formalism in HDR skin brachytherapy and the potential clinical impact. Methods: Dose rate distributions of typical configurations used in skin brachytherapy were obtained: a 5 cm × 5 cm superficial mould; a source inside a catheter located at the skin surface with and without backscatter bolus; and a typical interstitial implant consisting of an HDR source in a catheter located at a depth of 0.5 cm. Commercially available HDR{sup 60}Co and {sup 192}Ir sources and a hypothetical {sup 169}Yb source were considered. The Geant4 Monte Carlo radiation transport code was used to estimate dose rate distributions for the configurations considered. These results were then compared to those obtained with the TG-43 dose calculation formalism. In particular, the influence of adding bolus material over the implant was studied. Results: For a 5 cm × 5 cm{sup 192}Ir superficial mould and 0.5 cm prescription depth, dose differences in comparison to the TG-43 method were about −3%. When the source was positioned at the skin surface, dose differences were smaller than −1% for {sup 60}Co and {sup 192}Ir, yet −3% for {sup 169}Yb. For the interstitial implant, dose differences at the skin surface were −7% for {sup 60}Co, −0.6% for {sup 192}Ir, and −2.5% for {sup 169}Yb. Conclusions: This study indicates the following: (i) for the superficial mould, no bolus is needed; (ii) when the source is in contact with the skin surface, no bolus is needed for either {sup 60}Co and {sup 192}Ir. For
Emergency Doses (ED) - Revision 3: A calculator code for environmental dose computations
Rittmann, P.D.
1990-12-01
The calculator program ED (Emergency Doses) was developed from several HP-41CV calculator programs documented in the report Seven Health Physics Calculator Programs for the HP-41CV, RHO-HS-ST-5P (Rittman 1984). The program was developed to enable estimates of offsite impacts more rapidly and reliably than was possible with the software available for emergency response at that time. The ED - Revision 3, documented in this report, revises the inhalation dose model to match that of ICRP 30, and adds the simple estimates for air concentration downwind from a chemical release. In addition, the method for calculating the Pasquill dispersion parameters was revised to match the GENII code within the limitations of a hand-held calculator (e.g., plume rise and building wake effects are not included). The summary report generator for printed output, which had been present in the code from the original version, was eliminated in Revision 3 to make room for the dispersion model, the chemical release portion, and the methods of looping back to an input menu until there is no further no change. This program runs on the Hewlett-Packard programmable calculators known as the HP-41CV and the HP-41CX. The documentation for ED - Revision 3 includes a guide for users, sample problems, detailed verification tests and results, model descriptions, code description (with program listing), and independent peer review. This software is intended to be used by individuals with some training in the use of air transport models. There are some user inputs that require intelligent application of the model to the actual conditions of the accident. The results calculated using ED - Revision 3 are only correct to the extent allowed by the mathematical models. 9 refs., 36 tabs.
Lee, Boram; Lee, Jungseok; Kang, Sangwon; Cho, Hyelim; Shin, Gwisoon; Lee, Jeong-Woo; Choi, Jonghak
2013-01-01
The objective of this study was to evaluate the patient effective dose and scattered dose from recently developed dental mobile equipment in Korea. The MCNPX 2.6 (Los Alamos National Laboratory, USA) was used in a Monte Carlo simulation to calculate both the effective and scattered doses. The MCNPX code was constructed identically as in the general use of equipment and the effective dose and scattered dose were calculated using the KTMAN-2 digital phantom. The effective dose was calculated as 906 μSv. The equivalent doses per organ were calculated via the MCNPX code, and were 32 174 and 19 μSv in the salivary gland and oesophagus, respectively. The scattered dose of 22.5-32.6 μSv of the tube side at 25 cm from the centre in anterior and posterior planes was measured as 1.4-3 times higher than the detector side of 10.5-16.0 μSv.
De Smedt, B; Fippel, M; Reynaert, N; Thierens, H
2006-06-01
In order to evaluate the performance of denoising algorithms applied to Monte Carlo calculated dose distributions, conventional evaluation methods (rms difference, 1% and 2% difference) can be used. However, it is illustrated that these evaluation methods sometimes underestimate the introduction of bias, since possible bias effects are averaged out over the complete dose distribution. In the present work, a new evaluation method is introduced based on a sliding window superimposed on a difference dose distribution (reference dose-noisy/denoised dose). To illustrate its importance, a new denoising technique (ANRT) is presented based upon a combination of the principles of bilateral filtering and Savitzky-Golay filters. This technique is very conservative in order to limit the introduction of bias in high dose gradient regions. ANRT is compared with IRON for three challenging cases, namely an electron and photon beam impinging on heterogeneous phantoms and two IMRT treatment plans of head-and-neck cancer patients to determine the clinical relevance of the obtained results. For the electron beam case, IRON outperforms ANRT concerning the smoothing capabilities, while no differences in systematic bias are observed. However, for the photon beam case, although ANRT and IRON perform equally well on the conventional evaluation tests (rms difference, 1% and 2% difference), IRON clearly introduces much more bias in the penumbral regions while ANRT seems to introduce no bias at all. When applied to the IMRT patient cases, both denoising methods perform equally well regarding smoothing and bias introduction. This is probably caused by the summation of a large set of different beam segments, decreasing dose gradients compared to a single beam. A reduction in calculation time without introducing large systematic bias can shorten a Monte Carlo treatment planning process considerably and is therefore very useful for the initial trial and error phase of the treatment planning
Response of cellulose nitrate track detectors to electron doses
Segovia, N; Moreno, A; Vazquez-Polo, G; Santamaría, T; Aranda, P; Hernández, A
1999-01-01
In order to study alternative dose determination methods, the bulk etching velocity and the latent track annealing of LR 115 track detectors was studied during electron irradiation runs from a Pelletron accelerator. For this purpose alpha irradiated and blank detectors were exposed to increasing electron doses from 10.5 to 317.5 kGy. After the irradiation with electrons the detectors were etched under routine conditions, except for the etching time, that was varied for each electron dose in order to reach a fixed residual thickness. The variation of the bulk etching velocity as a function of each one of the electron doses supplied, was interpolated in order to obtain dosimetric response curves. The observed annealing effect on the latent tracks is discussed as a function of the total electron doses supplied and the temperature.
None, None
2017-05-01
Operations of Sandia National Laboratories, Nevada (SNL/NV) at the Tonopah Test Range (TTR) resulted in no planned point radiological releases during 1996. Other releases from SNL/NV included diffuse transuranic sources consisting of the three Clean Slate sites. Air emissions from these sources result from wind resuspension of near-surface transuranic contaminated soil particulates. The total area of contamination has been estimated to exceed 20 million square meters. Soil contamination was documented in an aerial survey program in 1977 (EG&G 1979). Surface contamination levels were generally found to be below 400 pCi/g of combined plutonium-238, plutonium-239, plutonium-240, and americium-241 (i.e., transuranic) activity. Hot spot areas contain up to 43,000 pCi/g of transuranic activity. Recent measurements confirm the presence of significant levels of transuranic activity in the surface soil. An annual diffuse source term of 0.39 Ci of transuranic material was calculated for the cumulative release from all three Clean Slate sites. A maximally exposed individual dose of 1.1 mrem/yr at the TTR airport area was estimated based on the 1996 diffuse source release amounts and site-specific meteorological data. A population dose of 0.86 person-rem/yr was calculated for the local residents. Both dose values were attributable to inhalation of transuranic contaminated dust.
GPU-based fast Monte Carlo dose calculation for proton therapy.
Jia, Xun; Schümann, Jan; Paganetti, Harald; Jiang, Steve B
2012-12-07
Accurate radiation dose calculation is essential for successful proton radiotherapy. Monte Carlo (MC) simulation is considered to be the most accurate method. However, the long computation time limits it from routine clinical applications. Recently, graphics processing units (GPUs) have been widely used to accelerate computationally intensive tasks in radiotherapy. We have developed a fast MC dose calculation package, gPMC, for proton dose calculation on a GPU. In gPMC, proton transport is modeled by the class II condensed history simulation scheme with a continuous slowing down approximation. Ionization, elastic and inelastic proton nucleus interactions are considered. Energy straggling and multiple scattering are modeled. Secondary electrons are not transported and their energies are locally deposited. After an inelastic nuclear interaction event, a variety of products are generated using an empirical model. Among them, charged nuclear fragments are terminated with energy locally deposited. Secondary protons are stored in a stack and transported after finishing transport of the primary protons, while secondary neutral particles are neglected. gPMC is implemented on the GPU under the CUDA platform. We have validated gPMC using the TOPAS/Geant4 MC code as the gold standard. For various cases including homogeneous and inhomogeneous phantoms as well as a patient case, good agreements between gPMC and TOPAS/Geant4 are observed. The gamma passing rate for the 2%/2 mm criterion is over 98.7% in the region with dose greater than 10% maximum dose in all cases, excluding low-density air regions. With gPMC it takes only 6-22 s to simulate 10 million source protons to achieve ∼1% relative statistical uncertainty, depending on the phantoms and energy. This is an extremely high efficiency compared to the computational time of tens of CPU hours for TOPAS/Geant4. Our fast GPU-based code can thus facilitate the routine use of MC dose calculation in proton therapy.
Tian, Zhen; Folkerts, Michael; Shi, Feng; Jiang, Steve B; Jia, Xun
2015-01-01
Monte Carlo (MC) simulation is considered as the most accurate method for radiation dose calculations. Accuracy of a source model for a linear accelerator is critical for the overall dose calculation accuracy. In this paper, we presented an analytical source model that we recently developed for GPU-based MC dose calculations. A key concept called phase-space-ring (PSR) was proposed. It contained a group of particles that are of the same type and close in energy and radial distance to the center of the phase-space plane. The model parameterized probability densities of particle location, direction and energy for each primary photon PSR, scattered photon PSR and electron PSR. For a primary photon PSRs, the particle direction is assumed to be from the beam spot. A finite spot size is modeled with a 2D Gaussian distribution. For a scattered photon PSR, multiple Gaussian components were used to model the particle direction. The direction distribution of an electron PSRs was also modeled as a 2D Gaussian distributi...
Surface dose with grids in electron beam radiation therapy
Lin, K.-H.; Huang, C.-Y.; Lin, J.-P.; Chu, T.-C. E-mail: tcchu@mx.nthu.edu.tw
2002-03-01
This investigation attempts to solve the problem of the lack of skin-sparing effect in electron radiation therapy and to increase the tolerance of skin to radiation using the grid technique. Electron grid therapy involves the mounting of a Cerrobend grid in the electron cone. Film dosimetry was employed to measure the relative surface dose and the percentage depth dose profile of electron grid portals. Various grid hole diameters (d=0.45, 1.0, 1.5 cm) and grid hole spacings (s=0.4, 0.2 cm) were considered for electron beams from 6 to 14 MeV. Experimental results indicate that the electron grid technique can reduce the relative surface dose in electron radiation therapy. Degradations of the relative surface dose depend on the percentage of open area in the grid portal. A proper grid design allows the surface dose to be reduced and the range of nonhomogeneous doses to be limited to a depth at which the target volume can receive a homogeneous dose. The grid technique can lower the surface dose in electron radiation therapy.
High-resolution low-dose scanning transmission electron microscopy.
Buban, James P; Ramasse, Quentin; Gipson, Bryant; Browning, Nigel D; Stahlberg, Henning
2010-01-01
During the past two decades instrumentation in scanning transmission electron microscopy (STEM) has pushed toward higher intensity electron probes to increase the signal-to-noise ratio of recorded images. While this is suitable for robust specimens, biological specimens require a much reduced electron dose for high-resolution imaging. We describe here protocols for low-dose STEM image recording with a conventional field-emission gun STEM, while maintaining the high-resolution capability of the instrument. Our findings show that a combination of reduced pixel dwell time and reduced gun current can achieve radiation doses comparable to low-dose TEM.
Calculation of electron wave functions and refractive index of Ne
2008-01-01
The radial wave functions of inner electron shell and outer electron shell of a Ne atom were obtained by the approximate analytical method and tested by calculating the ground state energy of the Ne atom. The equivalent volume of electron cloud and the refractive index of Ne were calculated. The calculated refractive index agrees well with the experimental result. Relationship between the refractive index and the wave function of Ne was discovered.
SU-E-T-357: Electronic Compensation Technique to Deliver Total Body Dose
Lakeman, T [State University of New York at Buffalo, Buffalo, NY (United States); Wang, I; Podgorsak, M [State University of New York at Buffalo, Buffalo, NY (United States); Roswell Park Cancer Institute, Buffalo, NY (United States)
2015-06-15
Purpose: Total body irradiation (TBI) uses large parallel-opposed radiation fields to suppress the patient’s immune system and eradicate the residual cancer cells in preparation of recipient for bone marrow transplant. The manual placement of lead compensators has conventionally been used to compensate for the varying thickness through the entire body in large-field TBI. The goal of this study is to pursue utilizing the modern electronic compensation technique to more accurately and efficiently deliver dose to patients in need of TBI. Methods: Treatment plans utilizing electronic compensation to deliver a total body dose were created retrospectively for patients for whom CT data had been previously acquired. Each treatment plan includes two, specifically weighted, pair of opposed fields. One pair of open, large fields (collimator=45°), to encompass the patient’s entire anatomy, and one pair of smaller fields (collimator=0°) focused only on the thicker midsection of the patient. The optimal fluence for each one of the smaller fields was calculated at a patient specific penetration depth. Irregular surface compensators provide a more uniform dose distribution within the smaller opposed fields. Results: Dose-volume histograms (DVH) were calculated for the evaluating the electronic compensation technique. In one case, the maximum body doses calculated from the DVH were reduced from the non-compensated 195.8% to 165.3% in the electronically compensated plans, indicating a more uniform dose with the region of electronic compensation. The mean body doses calculated from the DVH were also reduced from the non-compensated 120.6% to 112.7% in the electronically compensated plans, indicating a more accurate delivery of the prescription dose. All calculated monitor units were well within clinically acceptable limits. Conclusion: Electronic compensation technique for TBI will not substantially increase the beam on time while it can significantly reduce the compensator
Absorbed Dose Calculations Using Mesh-based Human Phantoms And Monte Carlo Methods
Kramer, Richard
2011-08-01
Health risks attributable to the exposure to ionizing radiation are considered to be a function of the absorbed or equivalent dose to radiosensitive organs and tissues. However, as human tissue cannot express itself in terms of equivalent dose, exposure models have to be used to determine the distribution of equivalent dose throughout the human body. An exposure model, be it physical or computational, consists of a representation of the human body, called phantom, plus a method for transporting ionizing radiation through the phantom and measuring or calculating the equivalent dose to organ and tissues of interest. The FASH2 (Female Adult meSH) and the MASH2 (Male Adult meSH) computational phantoms have been developed at the University of Pernambuco in Recife/Brazil based on polygon mesh surfaces using open source software tools and anatomical atlases. Representing standing adults, FASH2 and MASH2 have organ and tissue masses, body height and body mass adjusted to the anatomical data published by the International Commission on Radiological Protection for the reference male and female adult. For the purposes of absorbed dose calculations the phantoms have been coupled to the EGSnrc Monte Carlo code, which can transport photons, electrons and positrons through arbitrary media. This paper reviews the development of the FASH2 and the MASH2 phantoms and presents dosimetric applications for X-ray diagnosis and for prostate brachytherapy.
500-MeV electron beam bench-mark experiments and calculations
Farley, E.; Crase, K.; Selway, D.
1979-12-01
Experiments measuring the energy deposited by electron beams were performed to provide bench marks against which to evaluate our HANDYL76 electron beam computer code. The experiments, done at Stanford's Mk III accelerator, measured dose vs depth and dose vs radius profiles induced in layered aluminum targets by 500-MeV electrons. The dose was measured by passive thermoluminescence and photographic film placed between aluminum plates. The calculations predict a dose vs radius profile that forward-peaks on axis after the beam passes through a 200-cm air gap; the experimental measurements do not show this peak. This discrepancy indicates there may be a problem in using HANDYL76 to calculate deep penetration of a target with a large gap.
Electronic structure calculations on helical conducting polymers.
Ripoll, Juan D; Serna, Andrei; Guerra, Doris; Restrepo, Albeiro
2010-10-21
We present a study of the electronic structure and derived properties of polyfurane (PFu), polypyrrol (PPy), and polythiophene (PTh). Two spatial arrangements are considered: trans chain (tc-PFu, tc-PPy, tc-PTh) and cis α-helical (α-PFu, α-PPy, α-PTh). Even at the small sizes considered here, helical conformations appear to be stable. Band gaps of pure, undoped oligomers fall into the semiconductor range. Density of states (DOS) analysis suggest dense valence and conduction bands. Bond length alternation analysis predicts almost complete delocalization of the π clouds in all spatial arrangements. Doping with electron donors or electron-withdrawing impurities reduces all band gaps close to the metallic regime in addition to increasing the DOS for the valence and conduction bands.
Knöös, Tommy; Wieslander, Elinore; Cozzi, Luca; Brink, Carsten; Fogliata, Antonella; Albers, Dirk; Nyström, Håkan; Lassen, Søren
2006-11-21
A study of the performance of five commercial radiotherapy treatment planning systems (TPSs) for common treatment sites regarding their ability to model heterogeneities and scattered photons has been performed. The comparison was based on CT information for prostate, head and neck, breast and lung cancer cases. The TPSs were installed locally at different institutions and commissioned for clinical use based on local procedures. For the evaluation, beam qualities as identical as possible were used: low energy (6 MV) and high energy (15 or 18 MV) x-rays. All relevant anatomical structures were outlined and simple treatment plans were set up. Images, structures and plans were exported, anonymized and distributed to the participating institutions using the DICOM protocol. The plans were then re-calculated locally and exported back for evaluation. The TPSs cover dose calculation techniques from correction-based equivalent path length algorithms to model-based algorithms. These were divided into two groups based on how changes in electron transport are accounted for ((a) not considered and (b) considered). Increasing the complexity from the relatively homogeneous pelvic region to the very inhomogeneous lung region resulted in less accurate dose distributions. Improvements in the calculated dose have been shown when models consider volume scatter and changes in electron transport, especially when the extension of the irradiated volume was limited and when low densities were present in or adjacent to the fields. A Monte Carlo calculated algorithm input data set and a benchmark set for a virtual linear accelerator have been produced which have facilitated the analysis and interpretation of the results. The more sophisticated models in the type b group exhibit changes in both absorbed dose and its distribution which are congruent with the simulations performed by Monte Carlo-based virtual accelerator.
Block Tridiagonal Matrices in Electronic Structure Calculations
in the Landauer–Büttiker ballistic transport regime. These calculations concentrate on determining the so– called Green’s function matrix, or portions thereof, which is the inverse of a block tridiagonal general complex matrix. To this end, a sequential algorithm based on Gaussian elimination named Sweeps...
BENCHMARKING UPGRADED HOTSPOT DOSE CALCULATIONS AGAINST MACCS2 RESULTS
Brotherton, Kevin
2009-04-30
The radiological consequence of interest for a documented safety analysis (DSA) is the centerline Total Effective Dose Equivalent (TEDE) incurred by the Maximally Exposed Offsite Individual (MOI) evaluated at the 95th percentile consequence level. An upgraded version of HotSpot (Version 2.07) has been developed with the capabilities to read site meteorological data and perform the necessary statistical calculations to determine the 95th percentile consequence result. These capabilities should allow HotSpot to join MACCS2 (Version 1.13.1) and GENII (Version 1.485) as radiological consequence toolbox codes in the Department of Energy (DOE) Safety Software Central Registry. Using the same meteorological data file, scenarios involving a one curie release of {sup 239}Pu were modeled in both HotSpot and MACCS2. Several sets of release conditions were modeled, and the results compared. In each case, input parameter specifications for each code were chosen to match one another as much as the codes would allow. The results from the two codes are in excellent agreement. Slight differences observed in results are explained by algorithm differences.
Effect of Tissue Composition on Dose Distribution in Electron Beam Radiotherapy
Ghorbani, M.; Tabatabaei, Z. S.; Vejdani Noghreiyan, A.; Vosoughi, H.; Knaup, C.
2015-01-01
Objective The aim of this study is to evaluate the effect of tissue composition on dose distribution in electron beam radiotherapy. Methods A Siemens Primus linear accelerator and a phantom were simulated using MCNPX Monte Carlo code. In a homogeneous cylindrical phantom, six types of soft tissue and three types of tissue-equivalent materials were investigated. The tissues included muscle (skeletal), adipose tissue, blood (whole), breast tissue, soft tissue (9-components) and soft tissue (4-component). The tissue-equivalent materials were water, A-150 tissue-equivalent plastic and perspex. Electron dose relative to dose in 9-component soft tissue at various depths on the beam’s central axis was determined for 8, 12, and 14 MeV electron energies. Results The results of relative electron dose in various materials relative to dose in 9-component soft tissue were reported for 8, 12 and 14 MeV electron beams as tabulated data. While differences were observed between dose distributions in various soft tissues and tissue-equivalent materials, which vary with the composition of material, electron energy and depth in phantom, they can be ignored due to the incorporated uncertainties in Monte Carlo calculations. Conclusion Based on the calculations performed, differences in dose distributions in various soft tissues and tissue-equivalent materials are not significant. However, due to the difference in composition of various materials, further research in this field with lower uncertainties is recommended. PMID:25973407
Effect of Tissue Composition on Dose Distribution in Electron Beam Radiotherapy
Ghorbani M.
2015-03-01
Full Text Available Objective: The aim of this study is to evaluate the effect of tissue composition on dose distribution in electron beam radiotherapy. Methods: A Siemens Primus linear accelerator and a phantom were simulated using MCNPX Monte Carlo code. In a homogeneous cylindrical phantom, six types of soft tissue and three types of tissue-equivalent materials were investigated. The tissues included muscle (skeletal, adipose tissue, blood (whole, breast tissue, soft tissue (9-components and soft tissue (4-component. The tissue-equivalent materials were water, A-150 tissue-equivalent plastic and perspex. Electron dose relative to dose in 9-component soft tissue at various depths on the beam’s central axis was determined for 8, 12, and 14 MeV electron energies. Results: The results of relative electron dose in various materials relative to dose in 9-component soft tissue were reported for 8, 12 and 14 MeV electron beams as tabulated data. While differences were observed between dose distributions in various soft tissues and tissue-equivalent materials, which vary with the composition of material, electron energy and depth in phantom, they can be ignored due to the incorporated uncertainties in Monte Carlo calculations. Conclusion: Based on the calculations performed, differences in dose distributions in various soft tissues and tissue-equivalent materials are not significant. However, due to the difference in composition of various materials, further research in this field with lower uncertainties is recommended.
Hassan Ranjbar
2010-06-01
Full Text Available Introduction: Secondary radiation such as photoelectrons, Auger electrons and characteristic radiations cause a local boost in dose for a tumor when irradiated with an external X-ray beam after being loaded with elements capable of activating the tumor, e.g.; I and Gd. Materials and Methods: In this investigation, the MCNPX code was used for simulation and calculation of dose enhancement factor for a tumor loaded with activating elements. The designed model comprised the X-ray source, phantom (target tissue and loaded tumor with activating agent, detector, interactions modeling and results. The source was defined as monochromatic and plane surface situated at 50 cm (z = 50. Phantom geometry was a 10 × 10 × 10 cm3 cube centered at (0, 0, 0 with a 2.2 × 2.2 × 2.2 cm3 cubic tumor with a center located at 3 cm depth inside the phantom Results: Dose enhancement factor and optimum energy in radiotherapy are evaluated using the photon activation therapy method. Result show that the dose enhancement factor increases with activating concentration in the tumor. The maximum dose enhancement factor for iodine in the tumor occurs for photons in the energy range of 50-60 keV. Dose uniformity is less for lower energy photons within the activated region inside the tumor. Results indicate that the dose enhancement factor varies linearly with the activating concentration agent. Discussion and Conclusion: In this study, the obtained results point out a considerable enhancement in dose in the presence of activating agents in the tumor regions.
Basis functions for electronic structure calculations on spheres
Gill, Peter M W; Agboola, Davids
2014-01-01
We introduce a new basis function (the spherical gaussian) for electronic structure calculations on spheres of any dimension $D$. We find \\alert{general} expressions for the one- and two-electron integrals and propose an efficient computational algorithm incorporating the Cauchy-Schwarz bound. Using numerical calculations for the $D = 2$ case, we show that spherical gaussians are more efficient than spherical harmonics when the electrons are strongly localized.
Albright, N; Bergstrom, P M; Daly, T P; Descalle, M; Garrett, D; House, R K; Knapp, D K; May, S; Patterson, R W; Siantar, C L; Verhey, L; Walling, R S; Welczorek, D
1999-07-01
PEREGRINE is a 3D Monte Carlo dose calculation system designed to serve as a dose calculation engine for clinical radiation therapy treatment planning systems. Taking advantage of recent advances in low-cost computer hardware, modern multiprocessor architectures and optimized Monte Carlo transport algorithms, PEREGRINE performs mm-resolution Monte Carlo calculations in times that are reasonable for clinical use. PEREGRINE has been developed to simulate radiation therapy for several source types, including photons, electrons, neutrons and protons, for both teletherapy and brachytherapy. However the work described in this paper is limited to linear accelerator-based megavoltage photon therapy. Here we assess the accuracy, reliability, and added value of 3D Monte Carlo transport for photon therapy treatment planning. Comparisons with clinical measurements in homogeneous and heterogeneous phantoms demonstrate PEREGRINE's accuracy. Studies with variable tissue composition demonstrate the importance of material assignment on the overall dose distribution. Detailed analysis of Monte Carlo results provides new information for radiation research by expanding the set of observables.
Kusano, Maggie; Caldwell, Curtis B
2014-07-01
A primary goal of nuclear medicine facility design is to keep public and worker radiation doses As Low As Reasonably Achievable (ALARA). To estimate dose and shielding requirements, one needs to know both the dose equivalent rate constants for soft tissue and barrier transmission factors (TFs) for all radionuclides of interest. Dose equivalent rate constants are most commonly calculated using published air kerma or exposure rate constants, while transmission factors are most commonly calculated using published tenth-value layers (TVLs). Values can be calculated more accurately using the radionuclide's photon emission spectrum and the physical properties of lead, concrete, and/or tissue at these energies. These calculations may be non-trivial due to the polyenergetic nature of the radionuclides used in nuclear medicine. In this paper, the effects of dose equivalent rate constant and transmission factor on nuclear medicine dose and shielding calculations are investigated, and new values based on up-to-date nuclear data and thresholds specific to nuclear medicine are proposed. To facilitate practical use, transmission curves were fitted to the three-parameter Archer equation. Finally, the results of this work were applied to the design of a sample nuclear medicine facility and compared to doses calculated using common methods to investigate the effects of these values on dose estimates and shielding decisions. Dose equivalent rate constants generally agreed well with those derived from the literature with the exception of those from NCRP 124. Depending on the situation, Archer fit TFs could be significantly more accurate than TVL-based TFs. These results were reflected in the sample shielding problem, with unshielded dose estimates agreeing well, with the exception of those based on NCRP 124, and Archer fit TFs providing a more accurate alternative to TVL TFs and a simpler alternative to full spectral-based calculations. The data provided by this paper should assist
Hanford Site Annual Report Radiological Dose Calculation Upgrade Evaluation
Snyder, Sandra F.
2010-02-28
Operations at the Hanford Site, Richland, Washington, result in the release of radioactive materials to offsite residents. Site authorities are required to estimate the dose to the maximally exposed offsite resident. Due to the very low levels of exposure at the residence, computer models, rather than environmental samples, are used to estimate exposure, intake, and dose. A DOS-based model has been used in the past (GENII version 1.485). GENII v1.485 has been updated to a Windows®-based software (GENII version 2.08). Use of the updated software will facilitate future dose evaluations, but must be demonstrated to provide results comparable to those of GENII v1.485. This report describes the GENII v1.485 and GENII v2.08 dose exposure, intake, and dose estimates for the maximally exposed offsite resident reported for calendar year 2008. The GENII v2.08 results reflect updates to implemented algorithms. No two environmental models produce the same results, as was again demonstrated in this report. The aggregated dose results from 2008 Hanford Site airborne and surface water exposure scenarios provide comparable dose results. Therefore, the GENII v2.08 software is recommended for future offsite resident dose evaluations.
Photonuclear dose calculations for high-energy photon beams from Siemens and Varian linacs.
Chibani, Omar; Ma, Chang-Ming Charlie
2003-08-01
The dose from photon-induced nuclear particles (neutrons, protons, and alpha particles) generated by high-energy photon beams from medical linacs is investigated. Monte Carlo calculations using the MCNPX code are performed for three different photon beams from two different machines: Siemens 18 MV, Varian 15 MV, and Varian 18 MV. The linac head components are simulated in detail. The dose distributions from photons, neutrons, protons, and alpha particles are calculated in a tissue-equivalent phantom. Neutrons are generated in both the linac head and the phantom. This study includes (a) field size effects, (b) off-axis dose profiles, (c) neutron contribution from the linac head, (d) dose contribution from capture gamma rays, (e) phantom heterogeneity effects, and (f) effects of primary electron energy shift. Results are presented in terms of absolute dose distributions and also in terms of DER (dose equivalent ratio). The DER is the maximum dose from the particle (neutron, proton, or alpha) divided by the maximum photon dose, multiplied by the particle quality factor and the modulation scaling factor. The total DER including neutrons, protons, and alphas is about 0.66 cSv/Gy for the Siemens 18 MV beam (10 cm x 10 cm). The neutron DER decreases with decreasing field size while the proton (or alpha) DER does not vary significantly except for the 1 cm x 1 cm field. Both Varian beams (15 and 18 MV) produce more neutrons, protons, and alphas particles than the Siemens 18 MV beam. This is mainly due to their higher primary electron energies: 15 and 18.3 MeV, respectively, vs 14 MeV for the Siemens 18 MV beam. For all beams, neutrons contribute more than 75% of the total DER, except for the 1 cm x 1 cm field (approximately 50%). The total DER is 1.52 and 2.86 cSv/Gy for the 15 and 18 MV Varian beams (10 cm x 10 cm), respectively. Media with relatively high-Z elements like bone may increase the dose from heavy charged particles by a factor 4. The total DER is sensitive to
New Approach to Total Dose Specification for Spacecraft Electronics
Xapsos, Michael
2017-01-01
Variability of the space radiation environment is investigated with regard to total dose specification for spacecraft electronics. It is shown to have a significant impact. A new approach is developed for total dose requirements that replaces the radiation design margin concept with failure probability during a mission.
Abdulhamid Chaikh
2015-12-01
Full Text Available Purpose: the objective of this study is to validate a procedure based on a statistical method to assess the agreement and the correlation between measured and calculated dose in the process of quality assurance (QA for Intensity-Modulated Radiation Therapy (IMRT.Patients and methods: 10 patients including 56 fields for head and neck cancer treatment were analyzed. For each patient, one treatment plan was generated using Eclipse TPS®. To compare the calculated dose with measured dose a CT-scan of solid water slabs (30 × 30 × 15 cm3 was used. The measurements were done for absolute dose by a pinpoint ionization chamber and 2D dose distributions using electronic portal imaging device dosimetry. Six criteria levels were applied for each case (3%, 3 mm, (4%, 3 mm, (5%, 3 mm, (4%, 4 mm, (5%, 4 mm and (5%, 5 mm. The normality of the data and the variance homogeneity were tested using Shapiro-Wilks test and Levene’s test, respectively. Wilcoxon signed-rank paired test was used to calculate p-value. Bland-Altman method was used to calculate the limit of agreement between calculated and measured doses and to draw a scatter plot. The correlation between calculated and measured doses was assessed using Spearman’s rank test.Results: The statistical tests indicate that the data do not fulfill normal distribution, p < 0.001 and had a homogenous variance, p = 0.85. The upper and lower limit of agreements for absolute dose measurements were 6.44% and -6.40%, respectively. Wilcoxon test indicated a significance difference between calculated and measured dose with ionization chamber, p = 0.01. Spearman’s test indicated a strong correlation between calculated and absolute measured dose, ρ = 0.99. Therefore, there is a lack of correlation between dose difference for absolute dose measurements and gamma passing rates for 2D dose measurements.Conclusion: the statistical tests showed that the common acceptance criteria’s using gamma evaluation are not able
Effects of CT based Voxel Phantoms on Dose Distribution Calculated with Monte Carlo Method
Chen Chaobin; Huang Qunying; Wu Yican
2005-01-01
A few CT-based voxel phantoms were produced to investigate the sensitivity of Monte Carlo simulations of X-ray beam and electron beam to the proportions of elements and the mass densities of the materials used to express the patient's anatomical structure. The human body can be well outlined by air, lung, adipose, muscle, soft bone and hard bone to calculate the dose distribution with Monte Carlo method. The effects of the calibration curves established by using various CT scanners are not clinically significant based on our investigation. The deviation from the values of cumulative dose volume histogram derived from CT-based voxel phantoms is less than 1% for the given target.
Monte Carlo PENRADIO software for dose calculation in medical imaging
Adrien, Camille; Lòpez Noriega, Mercedes; Bonniaud, Guillaume; Bordy, Jean-Marc; Le Loirec, Cindy; Poumarede, Bénédicte
2014-06-01
The increase on the collective radiation dose due to the large number of medical imaging exams has led the medical physics community to deeply consider the amount of dose delivered and its associated risks in these exams. For this purpose we have developed a Monte Carlo tool, PENRADIO, based on a modified version of PENELOPE code 2006 release, to obtain an accurate individualized radiation dose in conventional and interventional radiography and in computed tomography (CT). This tool has been validated showing excellent agreement between the measured and simulated organ doses in the case of a hip conventional radiography and a coronography. We expect the same accuracy in further results for other localizations and CT examinations.
Komemushi, Atsushi; Tanigawa, Noboru; Kariya, Shuji; Yagi, Rie; Nakatani, Miyuki; Suzuki, Satoshi; Sano, Akira; Ikeda, Koshi; Utsunomiya, Keita; Harima, Yoko; Sawada, Satoshi
2012-01-01
Purpose. To assess differences in dose distribution of a vertebral body injected with bone cement as calculated by radiation treatment planning system (RTPS) and actual dose distribution. Methods. We prepared two water-equivalent phantoms with cement, and the other two phantoms without cement. The bulk density of the bone cement was imported into RTPS to reduce error from high CT values. A dose distribution map for the phantoms with and without cement was calculated using RTPS with clinical setting and with the bulk density importing. Actual dose distribution was measured by the film density. Dose distribution as calculated by RTPS was compared to the dose distribution measured by the film dosimetry. Results. For the phantom with cement, dose distribution was distorted for the areas corresponding to inside the cement and on the ventral side of the cement. However, dose distribution based on film dosimetry was undistorted behind the cement and dose increases were seen inside cement and around the cement. With the equivalent phantom with bone cement, differences were seen between dose distribution calculated by RTPS and that measured by the film dosimetry. Conclusion. The dose distribution of an area containing bone cement calculated using RTPS differs from actual dose distribution.
Analysis of offsite dose calculation methodology for a nuclear power reactor
Moser, Donna Smith [Univ. of North Carolina, Chapel Hill, NC (United States)
1995-01-01
This technical study reviews the methodology for calculating offsite dose estimates as described in the offsite dose calculation manual (ODCM) for Pennsylvania Power and Light - Susquehanna Steam Electric Station (SSES). An evaluation of the SSES ODCM dose assessment methodology indicates that it conforms with methodology accepted by the US Nuclear Regulatory Commission (NRC). Using 1993 SSES effluent data, dose estimates are calculated according to SSES ODCM methodology and compared to the dose estimates calculated according to SSES ODCM and the computer model used to produce the reported 1993 dose estimates. The 1993 SSES dose estimates are based on the axioms of Publication 2 of the International Commission of Radiological Protection (ICRP). SSES Dose estimates based on the axioms of ICRP Publication 26 and 30 reveal the total body estimates to be the most affected.
A GPU OpenCL based cross-platform Monte Carlo dose calculation engine (goMC)
Tian, Zhen; Shi, Feng; Folkerts, Michael; Qin, Nan; Jiang, Steve B.; Jia, Xun
2015-09-01
Monte Carlo (MC) simulation has been recognized as the most accurate dose calculation method for radiotherapy. However, the extremely long computation time impedes its clinical application. Recently, a lot of effort has been made to realize fast MC dose calculation on graphic processing units (GPUs). However, most of the GPU-based MC dose engines have been developed under NVidia’s CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a GPU OpenCL based cross-platform MC dose engine named goMC with coupled photon-electron simulation for external photon and electron radiotherapy in the MeV energy range. Compared to our previously developed GPU-based MC code named gDPM (Jia et al 2012 Phys. Med. Biol. 57 7783-97), goMC has two major differences. First, it was developed under the OpenCL environment for high code portability and hence could be run not only on different GPU cards but also on CPU platforms. Second, we adopted the electron transport model used in EGSnrc MC package and PENELOPE’s random hinge method in our new dose engine, instead of the dose planning method employed in gDPM. Dose distributions were calculated for a 15 MeV electron beam and a 6 MV photon beam in a homogenous water phantom, a water-bone-lung-water slab phantom and a half-slab phantom. Satisfactory agreement between the two MC dose engines goMC and gDPM was observed in all cases. The average dose differences in the regions that received a dose higher than 10% of the maximum dose were 0.48-0.53% for the electron beam cases and 0.15-0.17% for the photon beam cases. In terms of efficiency, goMC was ~4-16% slower than gDPM when running on the same NVidia TITAN card for all the cases we tested, due to both the different electron transport models and the different development environments. The code portability of our new dose engine goMC was validated by
A GPU OpenCL based cross-platform Monte Carlo dose calculation engine (goMC).
Tian, Zhen; Shi, Feng; Folkerts, Michael; Qin, Nan; Jiang, Steve B; Jia, Xun
2015-10-07
Monte Carlo (MC) simulation has been recognized as the most accurate dose calculation method for radiotherapy. However, the extremely long computation time impedes its clinical application. Recently, a lot of effort has been made to realize fast MC dose calculation on graphic processing units (GPUs). However, most of the GPU-based MC dose engines have been developed under NVidia's CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a GPU OpenCL based cross-platform MC dose engine named goMC with coupled photon-electron simulation for external photon and electron radiotherapy in the MeV energy range. Compared to our previously developed GPU-based MC code named gDPM (Jia et al 2012 Phys. Med. Biol. 57 7783-97), goMC has two major differences. First, it was developed under the OpenCL environment for high code portability and hence could be run not only on different GPU cards but also on CPU platforms. Second, we adopted the electron transport model used in EGSnrc MC package and PENELOPE's random hinge method in our new dose engine, instead of the dose planning method employed in gDPM. Dose distributions were calculated for a 15 MeV electron beam and a 6 MV photon beam in a homogenous water phantom, a water-bone-lung-water slab phantom and a half-slab phantom. Satisfactory agreement between the two MC dose engines goMC and gDPM was observed in all cases. The average dose differences in the regions that received a dose higher than 10% of the maximum dose were 0.48-0.53% for the electron beam cases and 0.15-0.17% for the photon beam cases. In terms of efficiency, goMC was ~4-16% slower than gDPM when running on the same NVidia TITAN card for all the cases we tested, due to both the different electron transport models and the different development environments. The code portability of our new dose engine goMC was validated by
Ability of medical students to calculate drug doses in children after their paediatric attachment
Oshikoya KA
2008-12-01
Full Text Available Dose calculation errors constitute a significant part of prescribing errors which might have resulted from informal teaching of the topic in medical schools. Objectives: To determine adequacy of knowledge and skills of drug dose calculations in children acquired by medical students during their clinical attachment in paediatrics.Methods: Fifty two 5th year medical students of the Lagos State University College of Medicine (LASUCOM, Ikeja were examined on drug dose calculations from a vial and ampoules of injections, syrup and suspension, and tablet formulation. The examination was with a structured questionnaire mostly in the form of multiple choice questions.Results: Thirty-six (69.2% and 30 (57.7% students were taught drug dose calculation in neonatal posting and during ward rounds/ bed-side teaching, respectively. Less than 50% of the students were able to calculate the correct doses of each of adrenaline, gentamicin, chloroquine and sodium bicarbonate injections required by the patient. Dose calculation was however relatively better with adrenalin when compared with the other injections. The proportion of female students that calculated the correct doses of quinine syrup and cefuroxime suspension were significantly higher than those of their male counterparts (p<0.05 and p<0.01, respectively; Chi-square test. When doses calculated in mg/dose and mL/dose was compared for adrenalin injection and each of quinine syrup and cefuroxime suspension, there were significant differences (adrenaline and quinine, p=0.005; adrenaline and cefuroxime, p=0.003: Fischer’s exact test. Dose calculation errors of similar magnitude to injections, syrup and suspension were also observed with tablet formulation.Conclusions: LASUCOM medical students lacked the basic knowledge of paediatric drug dose calculations but were willing to learn if the topic was formally taught. Drug dose calculations should be given a prominent consideration in the undergraduate medical
Dose calculation and in-phantom measurement in BNCT using response matrix method.
Rahmani, Faezeh; Shahriari, Majid
2011-12-01
In-phantom measurement of physical dose distribution is very important for Boron Neutron Capture Therapy (BNCT) planning validation. If any changes take place in therapeutic neutron beam due to the beam shaping assembly (BSA) change, the dose will be changed so another group of simulations should be carried out for dose calculation. To avoid this time consuming procedure and speed up the dose calculation to help patients not wait for a long time, response matrix method was used. This procedure was performed for neutron beam of the optimized BSA as a reference beam. These calculations were carried out using the MCNPX, Monte Carlo code. The calculated beam parameters were measured for a SNYDER head phantom placed 10 cm away from beam the exit of the BSA. The head phantom can be assumed as a linear system and neutron beam and dose distribution can be assumed as an input and a response of this system (head phantom), respectively. Neutron spectrum energy was digitized into 27 groups. Dose response of each group was calculated. Summation of these dose responses is equal to a total dose of the whole neutron/gamma spectrum. Response matrix is the double dimension matrix (energy/dose) in which each parameter represents a depth-dose resulted from specific energy. If the spectrum is changed, response of each energy group may be differed. By considering response matrix and energy vector, dose response can be calculated. This method was tested for some BSA, and calculations show statistical errors less than 10%.
User Guide for GoldSim Model to Calculate PA/CA Doses and Limits
Smith, F. [Savannah River Site (SRS), Aiken, SC (United States). Savannah River National Lab. (SRNL)
2016-10-31
A model to calculate doses for solid waste disposal at the Savannah River Site (SRS) and corresponding disposal limits has been developed using the GoldSim commercial software. The model implements the dose calculations documented in SRNL-STI-2015-00056, Rev. 0 “Dose Calculation Methodology and Data for Solid Waste Performance Assessment (PA) and Composite Analysis (CA) at the Savannah River Site”.
Imaging and Measuring Electron Beam Dose Distributions Using Holographic Interferometry
Miller, Arne; McLaughlin, W. L.
1975-01-01
Holographic interferometry was used to image and measure ionizing radiation depth-dose and isodose distributions in transparent liquids. Both broad and narrowly collimated electron beams from accelerators (2–10 MeV) provided short irradiation times of 30 ns to 0.6 s. Holographic images and measur......Holographic interferometry was used to image and measure ionizing radiation depth-dose and isodose distributions in transparent liquids. Both broad and narrowly collimated electron beams from accelerators (2–10 MeV) provided short irradiation times of 30 ns to 0.6 s. Holographic images...... and measurements of absorbed dose distributions were achieved in liquids of various densities and thermal properties and in water layers thinner than the electron range and with backings of materials of various densities and atomic numbers. The lowest detectable dose in some liquids was of the order of a few k...
Radial dose distributions from protons of therapeutic energies calculated with Geant4-DNA
Wang, He; Vassiliev, Oleg N.
2014-07-01
Models based on the amorphous track structure approximation have been successful in predicting the biological effects of heavy charged particles. Development of such models remains an active area of research that includes applications to hadrontherapy. In such models, the radial distribution of the dose deposited by delta electrons and directly by the particle is the main characteristic of track structure. We calculated these distributions with Geant4-DNA Monte Carlo code for protons in the energy range from 10 to 100 MeV. These results were approximated by a simple formula that combines the well-known inverse square distance dependence with two factors that eliminate the divergence of the radial dose integral at both small and large distances. A clear physical interpretation is given to the asymptotic behaviour of the radial dose distribution resulting from these two factors. The proposed formula agrees with the Monte Carlo data within 10% for radial distances of up to 10 μm, which corresponds to a dose range covering over eight orders of magnitude. Differences between our results and those of previously published analytical models are discussed.
A comparison of Monte Carlo dose calculation denoising techniques
El Naqa, I.; Kawrakow, I.; Fippel, M.; Siebers, J. V.; Lindsay, P. E.; Wickerhauser, M. V.; Vicic, M.; Zakarian, K.; Kauffmann, N.; Deasy, J. O.
2005-03-01
Recent studies have demonstrated that Monte Carlo (MC) denoising techniques can reduce MC radiotherapy dose computation time significantly by preferentially eliminating statistical fluctuations ('noise') through smoothing. In this study, we compare new and previously published approaches to MC denoising, including 3D wavelet threshold denoising with sub-band adaptive thresholding, content adaptive mean-median-hybrid (CAMH) filtering, locally adaptive Savitzky-Golay curve-fitting (LASG), anisotropic diffusion (AD) and an iterative reduction of noise (IRON) method formulated as an optimization problem. Several challenging phantom and computed-tomography-based MC dose distributions with varying levels of noise formed the test set. Denoising effectiveness was measured in three ways: by improvements in the mean-square-error (MSE) with respect to a reference (low noise) dose distribution; by the maximum difference from the reference distribution and by the 'Van Dyk' pass/fail criteria of either adequate agreement with the reference image in low-gradient regions (within 2% in our case) or, in high-gradient regions, a distance-to-agreement-within-2% of less than 2 mm. Results varied significantly based on the dose test case: greater reductions in MSE were observed for the relatively smoother phantom-based dose distribution (up to a factor of 16 for the LASG algorithm); smaller reductions were seen for an intensity modulated radiation therapy (IMRT) head and neck case (typically, factors of 2-4). Although several algorithms reduced statistical noise for all test geometries, the LASG method had the best MSE reduction for three of the four test geometries, and performed the best for the Van Dyk criteria. However, the wavelet thresholding method performed better for the head and neck IMRT geometry and also decreased the maximum error more effectively than LASG. In almost all cases, the evaluated methods provided acceleration of MC results towards statistically more accurate
A comparison of Monte Carlo dose calculation denoising techniques
Naqa, I El [Washington University, St Louis, MO (United States); Kawrakow, I [National Research Council of Canada, Ottawa, Ontario, Canada (Canada); Fippel, M [Univ Tuebingen, Tuebingen (Germany); Siebers, J V [Virginia Commonwealth University, Richmond, VA (United States); Lindsay, P E [Washington University, St Louis, MO (United States); Wickerhauser, M V [Washington University, St Louis, MO (United States); Vicic, M [Washington University, St Louis, MO (United States); Zakarian, K [Washington University, St Louis, MO (United States); Kauffmann, N [Ecole Polytechnique, Palaiseau (France); Deasy, J O [Washington University, St Louis, MO (United States)
2005-03-07
Recent studies have demonstrated that Monte Carlo (MC) denoising techniques can reduce MC radiotherapy dose computation time significantly by preferentially eliminating statistical fluctuations ('noise') through smoothing. In this study, we compare new and previously published approaches to MC denoising, including 3D wavelet threshold denoising with sub-band adaptive thresholding, content adaptive mean-median-hybrid (CAMH) filtering, locally adaptive Savitzky-Golay curve-fitting (LASG), anisotropic diffusion (AD) and an iterative reduction of noise (IRON) method formulated as an optimization problem. Several challenging phantom and computed-tomography-based MC dose distributions with varying levels of noise formed the test set. Denoising effectiveness was measured in three ways: by improvements in the mean-square-error (MSE) with respect to a reference (low noise) dose distribution; by the maximum difference from the reference distribution and by the 'Van Dyk' pass/fail criteria of either adequate agreement with the reference image in low-gradient regions (within 2% in our case) or, in high-gradient regions, a distance-to-agreement-within-2% of less than 2 mm. Results varied significantly based on the dose test case: greater reductions in MSE were observed for the relatively smoother phantom-based dose distribution (up to a factor of 16 for the LASG algorithm); smaller reductions were seen for an intensity modulated radiation therapy (IMRT) head and neck case (typically, factors of 2-4). Although several algorithms reduced statistical noise for all test geometries, the LASG method had the best MSE reduction for three of the four test geometries, and performed the best for the Van Dyk criteria. However, the wavelet thresholding method performed better for the head and neck IMRT geometry and also decreased the maximum error more effectively than LASG. In almost all cases, the evaluated methods provided acceleration of MC results towards
A GPU implementation of a track-repeating algorithm for proton radiotherapy dose calculations
Yepes, Pablo P; Taddei, Phillip J
2010-01-01
An essential component in proton radiotherapy is the algorithm to calculate the radiation dose to be delivered to the patient. The most common dose algorithms are fast but they are approximate analytical approaches. However their level of accuracy is not always satisfactory, especially for heterogeneous anatomic areas, like the thorax. Monte Carlo techniques provide superior accuracy, however, they often require large computation resources, which render them impractical for routine clinical use. Track-repeating algorithms, for example the Fast Dose Calculator, have shown promise for achieving the accuracy of Monte Carlo simulations for proton radiotherapy dose calculations in a fraction of the computation time. We report on the implementation of the Fast Dose Calculator for proton radiotherapy on a card equipped with graphics processor units (GPU) rather than a central processing unit architecture. This implementation reproduces the full Monte Carlo and CPU-based track-repeating dose calculations within 2%, w...
Vianello, Elizabeth A.; Almeida, Carlos E. de [Instituto Nacional do Cancer, Rio de Janeiro, RJ (Brazil); Biaggio, Maria F. de [Universidade do Estado, Rio de Janeiro, RJ (Brazil)
1998-09-01
This paper describes a manual method for dose calculation in brachytherapy of gynecological tumors, which allows the calculation of the doses at any plane or point of clinical interest. This method uses basic principles of vectorial algebra and the simulating orthogonal films taken from the patient with the applicators and dummy sources in place. The results obtained with method were compared with the values calculated with the values calculated with the treatment planning system model Theraplan and the agreement was better than 5% in most cases. The critical points associated with the final accuracy of the proposed method is related to the quality of the image and the appropriate selection of the magnification factors. This method is strongly recommended to the radiation oncology centers where are no treatment planning systems available and the dose calculations are manually done. (author) 10 refs., 5 figs.
Konstanty, Ewelina; Malicki, Julian; Łagodowska, Katarzyna; Kowalik, Anna
2017-01-01
Treatment of proliferative diseases of the hematopoietic system involves, in most cases, chemotherapy combined with radiation therapy, which is intended to provide adequate immunosuppressant. Conventionally, total body irradiation (TBI) was used; however, total marrow irradiation (TMI) performed with helical tomotherapy (HT) has been proposed as an alternative, with the aim of delivering the highest dose in the target area (skeleton bone). The purpose of this study is to evaluate the accuracy of the dose calculation algorithm for the lung in TMI delivered with HT. Thermoluminescent detectors (TLD-100 Harshaw) were used to measure delivered doses. Doses were calculated for 95 selected points in the central lung (53 TLDs) and near the rib bones (42 TLDs) in the anthropomorphic phantom. A total of 12 Gy were delivered (6 fractions of 2 Gy/fraction). HT-TMI technique reduces the dose delivered to the lungs in a phantom model to levels that are much lower than those reported for TBI delivered by a conventional linear accelerator. The mean calculated lung dose was 5.6 Gy versus a mean measured dose of 5.7 ± 2.4 Gy. The maximum and minimum measured doses were, respectively, 11.3 Gy (chest wall) and 2.8 Gy (central lung). At most of the 95 points, the measured dose was lower than the calculated dose, with the largest differences observed in the region located between the target volume and the adjacent lung tissue. The mean measured dose was lower than the calculated dose in both primary locations: -3.7% in the 42 rib-adjacent detectors and -3.0% in the 53 central lung TLDs. Our study has shown that the measured doses may be lower than those calculated by the HT-TMI calculation algorithm. Although these differences between calculated and measured doses are not clinically relevant, this finding merits further investigation.
Monte Carlo calculations for reporting patient organ doses from interventional radiology
Huo, Wanli; Feng, Mang; Pi, Yifei; Chen, Zhi; Gao, Yiming; Xu, X. George
2017-09-01
This paper describes a project to generate organ dose data for the purposes of extending VirtualDose software from CT imaging to interventional radiology (IR) applications. A library of 23 mesh-based anthropometric patient phantoms were involved in Monte Carlo simulations for database calculations. Organ doses and effective doses of IR procedures with specific beam projection, filed of view (FOV) and beam quality for all parts of body were obtained. Comparing organ doses for different beam qualities, beam projections, patients' ages and patient's body mass indexes (BMIs) which generated by VirtualDose-IR, significant discrepancies were observed. For relatively long time exposure, IR doses depend on beam quality, beam direction and patient size. Therefore, VirtualDose-IR, which is based on the latest anatomically realistic patient phantoms, can generate accurate doses for IR treatment. It is suitable to apply this software in clinical IR dose management as an effective tool to estimate patient doses and optimize IR treatment plans.
Recommended environmental dose calculation methods and Hanford-specific parameters
Schreckhise, R.G.; Rhoads, K.; Napier, B.A.; Ramsdell, J.V. (Pacific Northwest Lab., Richland, WA (United States)); Davis, J.S. (Westinghouse Hanford Co., Richland, WA (United States))
1993-03-01
This document was developed to support the Hanford Environmental Dose overview Panel (HEDOP). The Panel is responsible for reviewing all assessments of potential doses received by humans and other biota resulting from the actual or possible environmental releases of radioactive and other hazardous materials from facilities and/or operations belonging to the US Department of Energy on the Hanford Site in south-central Washington. This document serves as a guide to be used for developing estimates of potential radiation doses, or other measures of risk or health impacts, to people and other biota in the environs on and around the Hanford Site. It provides information to develop technically sound estimates of exposure (i.e., potential or actual) to humans or other biotic receptors that could result from the environmental transport of potentially harmful materials that have been, or could be, released from Hanford operations or facilities. Parameter values and information that are specific to the Hanford environs as well as other supporting material are included in this document.
Ballester, Facundo; Granero, Domingo; Perez-Calatayud, Jose; Melhus, Christopher S.; Rivard, Mark J. [Department of Atomic, Molecular and Nuclear Physics, University of Valencia, C/Dr. Moliner 50, E-46100 Burjassot (Spain) and IFIC, CSIC-University of Valencia, C/Dr. Moliner 50, E-46100 Burjassot (Spain); Department of Radiation Physics, ERESA, Hospital General Universitario, Avenida Tres Cruces, 2, E-46014 Valencia (Spain); Department of Radiation Oncology, La Fe University Hospital, Avenida Campanar 21, E-46009 Valencia (Spain); Department of Radiation Oncology, Tufts University School of Medicine, Boston, Massachusetts 02111 (United States)
2009-09-15
Purpose: The region of electronic disequilibrium near photon-emitting brachytherapy sources of high-energy radionuclides ({sup 60}Co, {sup 137}Cs, {sup 192}Ir, and {sup 169}Yb) and contributions to total dose from emitted electrons were studied using the GEANT4 and PENELOPE Monte Carlo codes. Methods: Hypothetical sources with active and capsule materials mimicking those of actual sources but with spherical shape were examined. Dose contributions due to source photons, x rays, and bremsstrahlung; source {beta}{sup -}, Auger electrons, and internal conversion electrons; and water collisional kerma were scored. To determine if conclusions obtained for electronic equilibrium conditions and electron dose contribution to total dose for the representative spherical sources could be applied to actual sources, the {sup 192}Ir mHDR-v2 source model (Nucletron B.V., Veenendaal, The Netherlands) was simulated for comparison to spherical source results and to published data. Results: Electronic equilibrium within 1% is reached for {sup 60}Co, {sup 137}Cs, {sup 192}Ir, and {sup 169}Yb at distances greater than 7, 3.5, 2, and 1 mm from the source center, respectively, in agreement with other published studies. At 1 mm from the source center, the electron contributions to total dose are 1.9% and 9.4% for {sup 60}Co and {sup 192}Ir, respectively. Electron emissions become important (i.e., >0.5%) within 3.3 mm of {sup 60}Co and 1.7 mm of {sup 192}Ir sources, yet are negligible over all distances for {sup 137}Cs and {sup 169}Yb. Electronic equilibrium conditions along the transversal source axis for the mHDR-v2 source are comparable to those of the spherical sources while electron dose to total dose contribution are quite different. Conclusions: Electronic equilibrium conditions obtained for spherical sources could be generalized to actual sources while electron contribution to total dose depends strongly on source dimensions, material composition, and electron spectra.
Low Energy Scanned Electron-Beam Dose Distribution in Thin Layers
McLaughlin, W. L.; Hjortenberg, P. E.; Pedersen, Walther Batsberg
1975-01-01
Thin radiochromic dye film dosimeters, calibrated by means of calorimetry, make possible the determination of absorbed-dose distributions due to low-energy scanned electron beam penetrations in moderately thin coatings and laminar media. For electrons of a few hundred keV, calibrated dosimeters...... of about 30–60 μm thickness may be used in stacks or interleaved between layers of materials of interest and supply a sufficient number of experimental data points throughout the depth of penetration of electrons to provide a depth-dose curve. Depth doses may be resolved in various polymer layers...... on different backings (wood, aluminum, and iron) for scanned electron beams (Emax = 400 keV) having a broad energy spectrum and diffuse incidence, such as those used in radiation curing of coatings, textiles, plastics, etc. Theoretical calculations of such distributions of energy depositions are relatively...
Output calculation of electron therapy at extended SSD using an improved LBR method
Alkhatib, Hassaan A.; Gebreamlak, Wondesen T., E-mail: wondtassew@gmail.com; Wright, Ben W.; Neglia, William J. [South Carolina Oncology Associates, Columbia, South Carolina 29210 (United States); Tedeschi, David J. [Department of Physics and Astronomy, University of South Carolina, Columbia, South Carolina 29208 (United States); Mihailidis, Dimitris [CAMC Cancer Center and Alliance Oncology, Charleston, West Virginia 25304 (United States); Sobash, Philip T. [The Medical University of South Carolina, Charleston, South Carolina 29425 (United States); Fontenot, Jonas D. [Department of Physics, Mary Bird Perkins Cancer Center, Baton Rouge, Louisiana 70809 (United States)
2015-02-15
Purpose: To calculate the output factor (OPF) of any irregularly shaped electron beam at extended SSD. Methods: Circular cutouts were prepared from 2.0 cm diameter to the maximum possible size for 15 × 15 applicator cone. In addition, two irregular cutouts were prepared. For each cutout, percentage depth dose (PDD) at the standard SSD and doses at different SSD values were measured using 6, 9, 12, and 16 MeV electron beam energies on a Varian 2100C LINAC and the distance at which the central axis electron fluence becomes independent of cutout size was determined. The measurements were repeated with an ELEKTA Synergy LINAC using 14 × 14 applicator cone and electron beam energies of 6, 9, 12, and 15 MeV. The PDD measurements were performed using a scanning system and two diodes—one for the signal and the other a stationary reference outside the tank. The doses of the circular cutouts at different SSDs were measured using PTW 0.125 cm{sup 3} Semiflex ion-chamber and EDR2 films. The electron fluence was measured using EDR2 films. Results: For each circular cutout, the lateral buildup ratio (LBR) was calculated from the measured PDD curve using the open applicator cone as the reference field. The effective SSD (SSD{sub eff}) of each circular cutout was calculated from the measured doses at different SSD values. Using the LBR value and the radius of the circular cutout, the corresponding lateral spread parameter [σ{sub R}(z)] was calculated. Taking the cutout size dependence of σ{sub R}(z) into account, the PDD curves of the irregularly shaped cutouts at the standard SSD were calculated. Using the calculated PDD curve of the irregularly shaped cutout along with the LBR and SSD{sub eff} values of the circular cutouts, the output factor of the irregularly shaped cutout at extended SSD was calculated. Finally, both the calculated PDD curves and output factor values were compared with the measured values. Conclusions: The improved LBR method has been generalized to
Uncertainties in Monte Carlo-based absorbed dose calculations for an experimental benchmark.
Renner, F; Wulff, J; Kapsch, R-P; Zink, K
2015-10-01
There is a need to verify the accuracy of general purpose Monte Carlo codes like EGSnrc, which are commonly employed for investigations of dosimetric problems in radiation therapy. A number of experimental benchmarks have been published to compare calculated values of absorbed dose to experimentally determined values. However, there is a lack of absolute benchmarks, i.e. benchmarks without involved normalization which may cause some quantities to be cancelled. Therefore, at the Physikalisch-Technische Bundesanstalt a benchmark experiment was performed, which aimed at the absolute verification of radiation transport calculations for dosimetry in radiation therapy. A thimble-type ionization chamber in a solid phantom was irradiated by high-energy bremsstrahlung and the mean absorbed dose in the sensitive volume was measured per incident electron of the target. The characteristics of the accelerator and experimental setup were precisely determined and the results of a corresponding Monte Carlo simulation with EGSnrc are presented within this study. For a meaningful comparison, an analysis of the uncertainty of the Monte Carlo simulation is necessary. In this study uncertainties with regard to the simulation geometry, the radiation source, transport options of the Monte Carlo code and specific interaction cross sections are investigated, applying the general methodology of the Guide to the expression of uncertainty in measurement. Besides studying the general influence of changes in transport options of the EGSnrc code, uncertainties are analyzed by estimating the sensitivity coefficients of various input quantities in a first step. Secondly, standard uncertainties are assigned to each quantity which are known from the experiment, e.g. uncertainties for geometric dimensions. Data for more fundamental quantities such as photon cross sections and the I-value of electron stopping powers are taken from literature. The significant uncertainty contributions are identified as
Carver, R [Mary Bird Perkins Cancer Center, Baton Rouge, LA (United States); Popple, R; Benhabib, S [UniversityAlabama Birmingham, Birmingham, AL (United Kingdom); Antolak, J [Mayo Clinic, Rochester, MN (United States); Sprunger, C [Louisiana State University, Baton Rouge, LA (United States); Hogstrom, K [Mary Bird Perkins Cancer Center, Baton Rouge, LA (United States); Louisiana State University, Baton Rouge, LA (United States)
2014-06-01
Purpose: To evaluate the accuracy of electron dose distribution calculated by the Varian Eclipse electron Monte Carlo (eMC) algorithm for use with recent commercially available bolus electron conformal therapy (ECT). Methods: eMC-calculated electron dose distributions for bolus ECT have been compared to those previously measured for cylindrical phantoms (retromolar trigone and nose), whose axial cross sections were based on the mid-PTV CT anatomy for each site. The phantoms consisted of SR4 muscle substitute, SR4 bone substitute, and air. The bolus ECT treatment plans were imported into the Eclipse treatment planning system and calculated using the maximum allowable histories (2×10{sup 9}), resulting in a statistical error of <0.2%. Smoothing was not used for these calculations. Differences between eMC-calculated and measured dose distributions were evaluated in terms of absolute dose difference as well as distance to agreement (DTA). Results: Results from the eMC for the retromolar trigone phantom showed 89% (41/46) of dose points within 3% dose difference or 3 mm DTA. There was an average dose difference of −0.12% with a standard deviation of 2.56%. Results for the nose phantom showed 95% (54/57) of dose points within 3% dose difference or 3 mm DTA. There was an average dose difference of 1.12% with a standard deviation of 3.03%. Dose calculation times for the retromolar trigone and nose treatment plans were 15 min and 22 min, respectively, using 16 processors (Intel Xeon E5-2690, 2.9 GHz) on a Varian Eclipse framework agent server (FAS). Results of this study were consistent with those previously reported for accuracy of the eMC electron dose algorithm and for the .decimal, Inc. pencil beam redefinition algorithm used to plan the bolus. Conclusion: These results show that the accuracy of the Eclipse eMC algorithm is suitable for clinical implementation of bolus ECT.
Poster — Thur Eve — 14: Improving Tissue Segmentation for Monte Carlo Dose Calculation using DECT
Di Salvio, A.; Bedwani, S.; Carrier, J-F. [Centre hospitalier de l' Université de Montréal (Canada); Bouchard, H. [National Physics Laboratory, Teddington (United Kingdom)
2014-08-15
Purpose: To improve Monte Carlo dose calculation accuracy through a new tissue segmentation technique with dual energy CT (DECT). Methods: Electron density (ED) and effective atomic number (EAN) can be extracted directly from DECT data with a stoichiometric calibration method. Images are acquired with Monte Carlo CT projections using the user code egs-cbct and reconstructed using an FDK backprojection algorithm. Calibration is performed using projections of a numerical RMI phantom. A weighted parameter algorithm then uses both EAN and ED to assign materials to voxels from DECT simulated images. This new method is compared to a standard tissue characterization from single energy CT (SECT) data using a segmented calibrated Hounsfield unit (HU) to ED curve. Both methods are compared to the reference numerical head phantom. Monte Carlo simulations on uniform phantoms of different tissues using dosxyz-nrc show discrepancies in depth-dose distributions. Results: Both SECT and DECT segmentation methods show similar performance assigning soft tissues. Performance is however improved with DECT in regions with higher density, such as bones, where it assigns materials correctly 8% more often than segmentation with SECT, considering the same set of tissues and simulated clinical CT images, i.e. including noise and reconstruction artifacts. Furthermore, Monte Carlo results indicate that kV photon beam depth-dose distributions can double between two tissues of density higher than muscle. Conclusions: A direct acquisition of ED and the added information of EAN with DECT data improves tissue segmentation and increases the accuracy of Monte Carlo dose calculation in kV photon beams.
Castro, Robson C. de; Silva, Ademir X. da; Crispim, Verginia R. [Universidade Federal, Rio de Janeiro, RJ (Brazil). Coordenacao dos Programas de Pos-graduacao de Engenharia. Programa de Engenharia Nuclear]. E-mail: rcastro@con.ufrj.br; Facure, Alessandro; Falcao, Rossana C. [Comissao Nacional de Energia Nuclear (CNEN), Rio de Janeiro, RJ (Brazil)]. E-mail: afsoares@cnen.gov.br; Lima, Marco A.F. [Universidade Federal Fluminense (UFF), Niteroi, RJ (Brazil). Dept. de Biologia Geral. Lab. de Radiobiologia e Radiometria]. E-mail: egbakel@vm.uff.br
2005-07-01
Radiotherapy with photon and electron beams still represents the most technique to control and treat tumour diseases. To increase the treatment efficiency of this technique is linked to the increase of beam energy, resulting in fast neutrons in the radiotherapic beams that contribute with an undesired dose to the patient. In this work has been calculated, using the MCNP4B computer code radiation of transport and an mathematical anthropomorphic phantom, the equivalent doses in organs originated from generated photoneutrons from heads of linear accelerators of medical use, that operates in the 15 MV, 18 MV, 20 MV and 25 MV. The calculated values for the equivalent doses in organs established by the 74 publication of ICRP has show variations between 0.11 mSv.n Gy{sup -1} and 7.03 mSv.n Gy{sup -1}, for the accelerator that uses 18 MV therapic beams, showing good agreement with existing values in the literature. (author)
Penfold, S; Miller, A [University of Adelaide, Adelaide, SA (Australia)
2015-06-15
Purpose: Stoichiometric calibration of Hounsfield Units (HUs) for conversion to proton relative stopping powers (RStPs) is vital for accurate dose calculation in proton therapy. However proton dose distributions are not only dependent on RStP, but also on relative scattering power (RScP) of patient tissues. RScP is approximated from material density but a stoichiometric calibration of HU-density tables is commonly neglected. The purpose of this work was to quantify the difference in calculated dose of a commercial TPS when using HU-density tables based on tissue substitute materials and stoichiometric calibrated ICRU tissues. Methods: Two HU-density calibration tables were generated based on scans of the CIRS electron density phantom. The first table was based directly on measured HU and manufacturer quoted density of tissue substitute materials. The second was based on the same CT scan of the CIRS phantom followed by a stoichiometric calibration of ICRU44 tissue materials. The research version of Pinnacle{sup 3} proton therapy was used to compute dose in a patient CT data set utilizing both HU-density tables. Results: The two HU-density tables showed significant differences for bone tissues; the difference increasing with increasing HU. Differences in density calibration table translated to a difference in calculated RScP of −2.5% for ICRU skeletal muscle and 9.2% for ICRU femur. Dose-volume histogram analysis of a parallel opposed proton therapy prostate plan showed that the difference in calculated dose was negligible when using the two different HU-density calibration tables. Conclusion: The impact of HU-density calibration technique on proton therapy dose calculation was assessed. While differences were found in the calculated RScP of bony tissues, the difference in dose distribution for realistic treatment scenarios was found to be insignificant.
Napier, B.A.; Kennedy, W.E. Jr.; Soldat, J.K.
1980-03-01
A computer program, PABLM, was written to facilitate the calculation of internal radiation doses to man from radionuclides in food products and external radiation doses from radionuclides in the environment. This report contains details of mathematical models used and calculational procedures required to run the computer program. Radiation doses from radionuclides in the environment may be calculated from deposition on the soil or plants during an atmospheric or liquid release, or from exposure to residual radionuclides in the environment after the releases have ended. Radioactive decay is considered during the release of radionuclides, after they are deposited on the plants or ground, and during holdup of food after harvest. The radiation dose models consider several exposure pathways. Doses may be calculated for either a maximum-exposed individual or for a population group. The doses calculated are accumulated doses from continuous chronic exposure. A first-year committed dose is calculated as well as an integrated dose for a selected number of years. The equations for calculating internal radiation doses are derived from those given by the International Commission on Radiological Protection (ICRP) for body burdens and MPC's of each radionuclide. The radiation doses from external exposure to contaminated water and soil are calculated using the basic assumption that the contaminated medium is large enough to be considered an infinite volume or plane relative to the range of the emitted radiations. The equations for calculations of the radiation dose from external exposure to shoreline sediments include a correction for the finite width of the contaminated beach.
Kinetics and dose calculations of amikacin in the newborn
Sardemann, H; Colding, H; Hendel, J;
1976-01-01
compartment model. The absorption was evaluated in 8 of the infants after intramuscular injection of 7.5 mg amikacin per kilogram of body weight. The absorption rate, estimated by the tmax, was significantly faster than reported in adults. The total body clearance and apparent volume of distribution were...... weight. The volume of distribution per kilogram was significantly greater than in adults. On the basis of the derived kinetic parameters, a dose schedule is presented. In 5 children there was a reasonable agreement between the measured and predicted serum levels....
Characterizing a Proton Beam Scanning System for Monte Carlo Dose Calculation in Patients
Grassberger, C; Lomax, Tony; Paganetti, H
2015-01-01
The presented work has two goals. First, to demonstrate the feasibility of accurately characterizing a proton radiation field at treatment head exit for Monte Carlo dose calculation of active scanning patient treatments. Second, to show that this characterization can be done based on measured depth dose curves and spot size alone, without consideration of the exact treatment head delivery system. This is demonstrated through calibration of a Monte Carlo code to the specific beam lines of two institutions, Massachusetts General Hospital (MGH) and Paul Scherrer Institute (PSI). Comparison of simulations modeling the full treatment head at MGH to ones employing a parameterized phase space of protons at treatment head exit reveals the adequacy of the method for patient simulations. The secondary particle production in the treatment head is typically below 0.2% of primary fluence, except for low–energy electrons (protons), whose contribution to skin dose is negligible. However, there is significant difference between the two methods in the low-dose penumbra, making full treatment head simulations necessary to study out-of field effects such as secondary cancer induction. To calibrate the Monte Carlo code to measurements in a water phantom, we use an analytical Bragg peak model to extract the range-dependent energy spread at the two institutions, as this quantity is usually not available through measurements. Comparison of the measured with the simulated depth dose curves demonstrates agreement within 0.5mm over the entire energy range. Subsequently, we simulate three patient treatments with varying anatomical complexity (liver, head and neck and lung) to give an example how this approach can be employed to investigate site-specific discrepancies between treatment planning system and Monte Carlo simulations. PMID:25549079
Sakamoto, Y
2002-01-01
In the prevention of nuclear disaster, there needs the information on the dose equivalent rate distribution inside and outside the site, and energy spectra. The three dimensional radiation transport calculation code is a useful tool for the site specific detailed analysis with the consideration of facility structures. It is important in the prediction of individual doses in the future countermeasure that the reliability of the evaluation methods of dose equivalent rate distribution and energy spectra by using of Monte Carlo radiation transport calculation code, and the factors which influence the dose equivalent rate distribution outside the site are confirmed. The reliability of radiation transport calculation code and the influence factors of dose equivalent rate distribution were examined through the analyses of critical accident at JCO's uranium processing plant occurred on September 30, 1999. The radiation transport calculations including the burn-up calculations were done by using of the structural info...
GMC: a GPU implementation of a Monte Carlo dose calculation based on Geant4.
Jahnke, Lennart; Fleckenstein, Jens; Wenz, Frederik; Hesser, Jürgen
2012-03-07
We present a GPU implementation called GMC (GPU Monte Carlo) of the low energy (CUDA programming interface. The classes for electron and photon interactions as well as a new parallel particle transport engine were implemented. The way a particle is processed is not in a history by history manner but rather by an interaction by interaction method. Every history is divided into steps that are then calculated in parallel by different kernels. The geometry package is currently limited to voxelized geometries. A modified parallel Mersenne twister was used to generate random numbers and a random number repetition method on the GPU was introduced. All phantom results showed a very good agreement between GPU and CPU simulation with gamma indices of >97.5% for a 2%/2 mm gamma criteria. The mean acceleration on one GTX 580 for all cases compared to Geant4 on one CPU core was 4860. The mean number of histories per millisecond on the GPU for all cases was 658 leading to a total simulation time for one intensity-modulated radiation therapy dose distribution of 349 s. In conclusion, Geant4-based Monte Carlo dose calculations were significantly accelerated on the GPU.
Yoo, Hyunjun; Kim, Chankyu; Kim, Yewon; Kim, Giyoon; Cho, Gyuseong [Korea Advanced Institute of Science and Technology, Daejeon (Korea, Republic of)
2014-05-15
There are two categories in personal dosimeters, one is passive type dosimeter such as TLD (thermoluminescence dosimeter) and the other is active type dosimeter such as electronic dosimeter can show radiation dose immediately while TLD needs long time to readout its data by heating process. For improving the reliability of measuring dose for any energy of radiations, electronic dosimeter uses energy filter by metal packaging its detector using aluminum or copper, but measured dose of electronic dosimeter with energy filter cannot be completely compensated in wide radiation energy region. So, in this paper, we confirmed the accuracy of dose measurement of two types of commercial EPDs using Geiger Muller tube and PIN diode with CsI(Tl) scintillator in three different energy of radiation field. The experiment results for Cs-137 was almost similar with calculation value in the results of both electronic dosimeters, but, the other experiment values with Na-22 and Co-60 had higher error comparing with Cs-137. These results were caused by optimization of their energy filters. The optimization was depending on its thickness of energy filter. So, the electronic dosimeters have to optimizing the energy filter for increasing the accuracy of dose measurement or the electronic dosimeter using PIN diode with CsI(Tl) scintillator uses the multi-channel discriminator for using its energy information.
The calculation of vibrational intensities in forbidden electronic transitions.
Johnson, Philip M; Xu, Haifeng; Sears, Trevor J
2006-10-28
A method is described for the use of electronic structure and Franck-Condon factor programs in the calculation of the vibrational intensities in forbidden electronic transitions. Using the B 2B2-X 2B1 electronic transition of benzonitrile cation as a test case, transition moments were calculated using the symmetry adapted cluster/configuration interaction method at various points along the normal mode displacements of the molecule, from which transition moment derivatives were obtained. The transition moments were found to vary almost linearly with respect to the normal mode displacements. Using these, along with Franck-Condon factors, an expansion of the transition moment with respect to the normal coordinates provides a measure of vibrational intensities, including the effects of geometry change and Duschinsky rotation [Acta Physicochim. URSS 7, 551 (1937)]. Second order terms in the moment expansion are calculated, and it is determined that they must be included if the intensity of combination bands is to be properly obtained.
Bevelacqua, J J
2011-07-01
Conservatism in the calculation of the effective dose following an airborne release from an accident involving a fuel reprocessing waste tank is examined. Within the regulatory constraints at the Hanford Site, deterministic effective dose calculations are conservative by at least an order of magnitude. Deterministic calculations should be used with caution in reaching decisions associated with required safety systems and mitigation philosophy related to the accidental release of airborne radioactive material to the environment.
Rapid low dose electron tomography using a direct electron detection camera
Migunov, V.; Ryll, H.; Zhuge, X.; Simson, M.; Strüder, L.; Batenburg, K.J.; Houben, L.; Dunin-Borkowski, R.E.
2015-01-01
We demonstrate the ability to record a tomographic tilt series containing 3487 images in only 3.5 s by using a direct electron detector in a transmission electron microscope. The electron dose is lower by at least one order of magnitude when compared with that used to record a conventional tilt seri
SU-E-T-164: Evaluation of Electron Dose Distribution Using Two Algorithms
Liu, D; Li, Z; Shang, K; Jing, Z; Wang, J; Miao, M [The Fourth Hospital of Hebei Medical University, Shijiazhuang Hebei, CN (China); Yang, J [Cangzhou People' s Hospital, Cangzhou, Hebei (China)
2014-06-01
Purpose: To appreciate the difference of electron dose distributions calculated from the Monte Carlo and Electron 3D algorithms of radiotherapy in a heterogeneous phantom. Methods: A phantom consisted of two different materials (lungs mimicked by low-density cork and others by polystyrene) with an 11x16 cm field size (SSD = 100 cm) was utilized to estimate the two-dimensional dose distributions under 6 and 18 MeV beams. On behalf of two different types of tissue, the heterogeneous phantom was comprised of 3 identical slabs in the longitudinal direction with a thickness of 1 cm for each slab and 2 with a thickness of 2.5 cm. The Monte Carlo/MCTP application package constituted of five codes was performed to simulate the electron beams of a Varian Clinac 23IX. A 20x20 cm2 type III (open walled) applicator was used in these simulations. It has been shown elsewhere that the agreement of the phase space data between the calculation results of MCTP application package and the measured data were within 2% on depth-dose and transverse profiles, as well as output factor calculations. The electron 3D algorithm owned by Pinnacle 8.0m and the MCTP application package were applied for the two-dimensional dose distributions calculation. The curves at 50% and 100%-prescribed dose were observed for 6 and 18 MeV beams, respectively. Results: The MC calculations results were compared with the electron 3D calculations in terms of two-dimensional dose distributions for 6 and 18 MeV beams showed excellent agreement except in distal boundary where it was the very junction of the high and low-density region. Conclusions: The Monte Carlo/MCTP method could be used to better reflect the dose variation caused by heterogeneous tissues. Conclusion: A case study showed that the Monte Carlo/MCTP method could be used to better reflect the dose variation caused by heterogeneous tissues.
Dose conversion coefficients for electron exposure of the human eye lens
Behrens, R [Physikalisch-Technische Bundesanstalt, Bundesallee 100, D-38116 Braunschweig (Germany); Dietze, G [Paracelsusstrasse 7, D-38116 Braunschweig (Germany); Zankl, M [Helmholtz Zentrum Muenchen, Ingolstaedter Landstrasse 1, D-85764 Neuherberg (Germany)], E-mail: rolf.behrens@ptb.de
2009-07-07
Recent epidemiological studies suggest a rather low dose threshold (below 0.5 Gy) for the induction of a cataract of the eye lens. Some other studies even assume that there is no threshold at all. Therefore, protection measures have to be optimized and current dose limits for the eye lens may be reduced in the future. Two questions arise from this situation: first, which dose quantity is related to the risk of developing a cataract, and second, which personal dose equivalent quantity is appropriate for monitoring this dose quantity. While the dose equivalent quantity H{sub p}(0.07) has often been seen as being sufficiently accurate for monitoring the dose to the lens of the eye, this would be questionable in the case when the dose limits were reduced and, thus, it may be necessary to generally use the dose equivalent quantity H{sub p}(3) for this purpose. The basis for a decision, however, must be the knowledge of accurate conversion coefficients from fluence to equivalent dose to the lens. This is especially important for low-penetrating radiation, for example, electrons. Formerly published values of conversion coefficients are based on quite simple models of the eye. In this paper, quite a sophisticated model of the eye including the inner structure of the lens was used for the calculations and precise conversion coefficients for electrons with energies between 0.2 MeV and 12 MeV, and for angles of radiation incidence between 0 deg. and 45 deg. are presented. Compared to the values adopted in 1996 by the International Commission on Radiological Protection (ICRP), the new values are up to 1000 times smaller for electron energies below 1 MeV, nearly equal at 1 MeV and above 4 MeV, and by a factor of 1.5 larger at about 1.5 MeV electron energy.
Dose conversion coefficients for electron exposure of the human eye lens
Behrens, R.; Dietze, G.; Zankl, M.
2009-07-01
Recent epidemiological studies suggest a rather low dose threshold (below 0.5 Gy) for the induction of a cataract of the eye lens. Some other studies even assume that there is no threshold at all. Therefore, protection measures have to be optimized and current dose limits for the eye lens may be reduced in the future. Two questions arise from this situation: first, which dose quantity is related to the risk of developing a cataract, and second, which personal dose equivalent quantity is appropriate for monitoring this dose quantity. While the dose equivalent quantity Hp(0.07) has often been seen as being sufficiently accurate for monitoring the dose to the lens of the eye, this would be questionable in the case when the dose limits were reduced and, thus, it may be necessary to generally use the dose equivalent quantity Hp(3) for this purpose. The basis for a decision, however, must be the knowledge of accurate conversion coefficients from fluence to equivalent dose to the lens. This is especially important for low-penetrating radiation, for example, electrons. Formerly published values of conversion coefficients are based on quite simple models of the eye. In this paper, quite a sophisticated model of the eye including the inner structure of the lens was used for the calculations and precise conversion coefficients for electrons with energies between 0.2 MeV and 12 MeV, and for angles of radiation incidence between 0° and 45° are presented. Compared to the values adopted in 1996 by the International Commission on Radiological Protection (ICRP), the new values are up to 1000 times smaller for electron energies below 1 MeV, nearly equal at 1 MeV and above 4 MeV, and by a factor of 1.5 larger at about 1.5 MeV electron energy.
Bourguignon, Laurent; Goutelle, Sylvain; Gérard, Cécile; Guillermet, Anne; Burdin de Saint Martin, Julie; Maire, Pascal; Ducher, Michel
2009-01-01
The use of amikacin is difficult because of its toxicity and its pharmacokinetic variability. This variability is almost ignored in adult standard dosage regimens since only the weight is used in the dose calculation. Our objective is to test if the pharmacokinetic of amikacin can be regarded as homogenous, and if the method for calculating the dose according to patients' weight is appropriate. From a cohort of 580 patients, five groups of patients were created by statistical data partitioning. A population pharmacokinetic analysis was performed in each group. The adult population is not homogeneous in term of pharmacokinetics. The doses required to achieve a maximum concentration of 60 mg/L are strongly different (585 to 1507 mg) between groups. The exclusive use of the weight to calculate the dose of amikacine appears inappropriate for 80% of the patients, showing the limits of the formulae for calculating doses of aminoglycosides.
Multiconfiguration calculations of electronic isotope shift factors in Al I
Filippin, Livio; Ekman, Jörgen; Fritzsche, Stephan; Godefroid, Michel; Jönsson, Per
2016-01-01
The present work reports results from systematic multiconfiguration Dirac-Hartree-Fock calculations of electronic isotope shift factors for a set of transitions between low-lying states in neutral aluminium. These electronic quantities together with observed isotope shifts between different pairs of isotopes provide the changes in mean-square charge radii of the atomic nuclei. Two computational approaches are adopted for the estimation of the mass- and field shift factors. Within these approaches, different models for electron correlation are explored in a systematic way to determine a reliable computational strategy and estimate theoretical uncertainties of the isotope shift factors.
NONE
2000-07-01
The guide sets out the mathematical definitions and principles involved in the calculation of the equivalent dose and the effective dose, and the instructions concerning the application of the maximum values of these quantities. further, for monitoring the dose caused by internal radiation, the guide defines the limits derived from annual dose limits (the Annual Limit on Intake and the Derived Air Concentration). Finally, the guide defines the operational quantities to be used in estimating the equivalent dose and the effective dose, and also sets out the definitions of some other quantities and concepts to be used in monitoring radiation exposure. The guide does not include the calculation of patient doses carried out for the purposes of quality assurance.
Tian, Zhen; Li, Yongbao; Hassan-Rezaeian, Nima; Jiang, Steve B; Jia, Xun
2017-03-01
We have previously developed a GPU-based Monte Carlo (MC) dose engine on the OpenCL platform, named goMC, with a built-in analytical linear accelerator (linac) beam model. In this paper, we report our recent improvement on goMC to move it toward clinical use. First, we have adapted a previously developed automatic beam commissioning approach to our beam model. The commissioning was conducted through an optimization process, minimizing the discrepancies between calculated dose and measurement. We successfully commissioned six beam models built for Varian TrueBeam linac photon beams, including four beams of different energies (6 MV, 10 MV, 15 MV, and 18 MV) and two flattening-filter-free (FFF) beams of 6 MV and 10 MV. Second, to facilitate the use of goMC for treatment plan dose calculations, we have developed an efficient source particle sampling strategy. It uses the pre-generated fluence maps (FMs) to bias the sampling of the control point for source particles already sampled from our beam model. It could effectively reduce the number of source particles required to reach a statistical uncertainty level in the calculated dose, as compared to the conventional FM weighting method. For a head-and-neck patient treated with volumetric modulated arc therapy (VMAT), a reduction factor of ~2.8 was achieved, accelerating dose calculation from 150.9 s to 51.5 s. The overall accuracy of goMC was investigated on a VMAT prostate patient case treated with 10 MV FFF beam. 3D gamma index test was conducted to evaluate the discrepancy between our calculated dose and the dose calculated in Varian Eclipse treatment planning system. The passing rate was 99.82% for 2%/2 mm criterion and 95.71% for 1%/1 mm criterion. Our studies have demonstrated the effectiveness and feasibility of our auto-commissioning approach and new source sampling strategy for fast and accurate MC dose calculations for treatment plans. © 2017 The Authors. Journal of Applied Clinical Medical Physics published by
First Principles Calculations of Electronic Excitations in 2D Materials
Rasmussen, Filip Anselm
-thin electronics and high efficiency solar cells. Contrary to many other nano-materials, methods for large scale fabrication and patterning have already been demonstrated and the first real technological applications have already be showcased. Still the technology is very young and the number of well-studied 2D...... materials are few. However as the list of 2D materials is growing it is necessary to investigate their fundamental structural, electronic and optical properties. These are determined by the atomic and electronic structure of the materials that can quite accurately predicted by computational quantum...... as if it is being screened by the electrons in the material. This method has been very successful for calculating quasiparticle energies of bulk materials but results have been more varying for 2D materials. The reason is that the 2D confined electrons are less able to screen the added charge and some...
Ribeiro, Rosane Moreira; Souza-Santos, Denison
2017-10-01
A comparison between neutron physics lists given by GEANT4, is made in the calculation of the ambient dose equivalent, and ambient absorbed dose, per fluence conversion coefficients (H* (10) / ϕ and D* (10) / ϕ) for neutrons in the range of 10-9 MeV to 15 MeV. Physics processes are included for neutrons, photons and charged particles, and calculations are made for neutrons and secondary particles. Results obtained for QBBC, QGSP_BERT, QGSP_BIC and Neutron High Precision physics lists are compared with values published in ICRP 74 and previously published articles. Neutron high precision physics lists showed the best results in the studied energy range.
Han, C; Schultheiss, T [City of Hope National Medical Center, Duarte, CA (United States)
2015-06-15
Purpose: In this study, we aim to evaluate the effect of dose grid size on the accuracy of calculated dose for small lesions in intracranial stereotactic radiosurgery (SRS), and to verify dose calculation accuracy with radiochromic film dosimetry. Methods: 15 intracranial lesions from previous SRS patients were retrospectively selected for this study. The planning target volume (PTV) ranged from 0.17 to 2.3 cm{sup 3}. A commercial treatment planning system was used to generate SRS plans using the volumetric modulated arc therapy (VMAT) technique using two arc fields. Two convolution-superposition-based dose calculation algorithms (Anisotropic Analytical Algorithm and Acuros XB algorithm) were used to calculate volume dose distribution with dose grid size ranging from 1 mm to 3 mm with 0.5 mm step size. First, while the plan monitor units (MU) were kept constant, PTV dose variations were analyzed. Second, with 95% of the PTV covered by the prescription dose, variations of the plan MUs as a function of dose grid size were analyzed. Radiochomic films were used to compare the delivered dose and profile with the calculated dose distribution with different dose grid sizes. Results: The dose to the PTV, in terms of the mean dose, maximum, and minimum dose, showed steady decrease with increasing dose grid size using both algorithms. With 95% of the PTV covered by the prescription dose, the total MU increased with increasing dose grid size in most of the plans. Radiochromic film measurements showed better agreement with dose distributions calculated with 1-mm dose grid size. Conclusion: Dose grid size has significant impact on calculated dose distribution in intracranial SRS treatment planning with small target volumes. Using the default dose grid size could lead to under-estimation of delivered dose. A small dose grid size should be used to ensure calculation accuracy and agreement with QA measurements.
A centralized dose calculation system for radiation therapy.
Xiao, Y; Galvin, J
2000-05-01
Centralization of treatment planning in a radiation therapy department is a realistic strategy to achieve an integrated and quality-controlled planning system, especially for institutions with numerous affiliations. The rapid evolution of computer hardware and software technology makes this a distinct possibility. However, the procedure of three-dimensional treatment planning involves a number of steps, such as: (1) input of patient computed tomography (CT) images and contour information; (2) interactions with local devices such as a film digitizer; and (3) output of beam information to be integrated with the record and verify the system. A full-fledged realization of the web-based centralized three-dimensional treatment planning system will require an extensive commercial development effort. We have developed and incorporated a web-based Timer/Monitor Unit (MU) program as a first step towards the full implementation of a centralized treatment planning system. The software application was developed in JAVA language. It uses the internet server and client technology. With one server that can handle multiple threads, it is a simple process to access the application anywhere on the network with an internet browser. Both the essential data needed for the calculation and the results are stored on the server, which centralizes the maintenance of the software and the storage of patient information.
Monte Carlo calculation of skyshine'' neutron dose from ALS (Advanced Light Source)
Moin-Vasiri, M.
1990-06-01
This report discusses the following topics on skyshine'' neutron dose from ALS: Sources of radiation; ALS modeling for skyshine calculations; MORSE Monte-Carlo; Implementation of MORSE; Results of skyshine calculations from storage ring; and Comparison of MORSE shielding calculations.
Structural and electronic properties of perylene from first principles calculations.
Fedorov, I A; Zhuravlev, Y N; Berveno, V P
2013-03-07
The electronic structure of crystalline perylene has been investigated within the framework of density functional theory including van der Waals interactions. The computations of the lattice parameters and cohesive energy have good agreement with experimental values. We have also calculated the binding distance and energy of perylene dimers, using different schemes, which include van der Waals interactions.
Three real-space discretization techniques in electronic structure calculations
Torsti, T; Eirola, T; Enkovaara, J; Hakala, T; Havu, P; Havu, [No Value; Hoynalanmaa, T; Ignatius, J; Lyly, M; Makkonen, [No Value; Rantala, TT; Ruokolainen, J; Ruotsalainen, K; Rasanen, E; Saarikoski, H; Puska, MJ
2006-01-01
A characteristic feature of the state-of-the-art of real-space methods in electronic structure calculations is the diversity of the techniques used in the discretization of the relevant partial differential equations. In this context, the main approaches include finite-difference methods, various ty
Simmer, Gregor
2012-04-11
Due to secondary cosmic radiation (SCR), pilots and flight attendants receive elevated effective doses at flight altitudes. For this reason, since 2003 aircrew members are considered as occupationally exposed, in Germany. This work deals with the calculation of dose conversion coefficients (DCC) for protons, neutrons, electrons, positrons, photons and myons, which are crucial for estimation of effective dose from SCR. For the first time, calculations were performed combining Geant4 - a Monte Carlo code developed at CERN - with the voxel phantoms for the reference female and male published in 2008 by ICRP and ICRU. Furthermore, measurements of neutron fluence spectra - which contribute the major part to the effective dose of SCR - were carried out at the Environmental Research Station Schneefernerhaus (UFS) located at 2650 m above sea level nearby the Zugspitze mountain, Germany. These measured neutron spectra, and additionally available calculated spectra, were then folded with the DCC calculated in this work, and effective dose rates for different heights were calculated.
Real-time feedback from iterative electronic structure calculations
Vaucher, Alain C; Reiher, Markus
2015-01-01
Real-time feedback from iterative electronic structure calculations requires to mediate between the inherently unpredictable execution times of the iterative algorithm employed and the necessity to provide data in fixed and short time intervals for real-time rendering. We introduce the concept of a mediator as a component able to deal with infrequent and unpredictable reference data to generate reliable feedback. In the context of real-time quantum chemistry, the mediator takes the form of a surrogate potential that has the same local shape as the first-principles potential and can be evaluated efficiently to deliver atomic forces as real-time feedback. The surrogate potential is updated continuously by electronic structure calculations and guarantees to provide a reliable response to the operator for any molecular structure. To demonstrate the application of iterative electronic structure methods in real-time reactivity exploration, we implement self-consistent semi-empirical methods as the data source and a...
Electron impact double ionization of helium from classical trajectory calculations
Geyer, T
2004-01-01
With a recently proposed quasiclassical ansatz [Geyer and Rost, J. Phys. B 35 (2002) 1479] it is possible to perform classical trajectory ionization calculations on many electron targets. The autoionization of the target is prevented by a M\\o{}ller type backward--forward propagation scheme and allows to consider all interactions between all particles without additional stabilization. The application of the quasiclassical ansatz for helium targets is explained and total and partially differential cross sections for electron impact double ionization are calculated. In the high energy regime the classical description fails to describe the dominant TS1 process, which leads to big deviations, whereas for low energies the total cross section is reproduced well. Differential cross sections calculated at 250 eV await their experimental confirmation.
Implementation of spot scanning dose optimization and dose calculation for helium ions in Hyperion
Fuchs, Hermann; Alber, Markus; Schreiner, Thomas
2015-01-01
and integrated into the treatment planning system Hyperion. METHODS: Current knowledge on RBE of (4)He together with linear energy transfer considerations motivated an empirical depth-dependent "zonal" RBE model. In the plateau region, a RBE of 1.0 was assumed, followed by an increasing RBE up to 2...... doses resulted in a γ mean of 0.3, with 3.4% of the values above 1 and γ 1% of 1.5 and better. Treatment plan evaluation showed comparable planning target volume coverage for both particles, with slightly increased coverage for (4)He. Organ at risk (OAR) doses were generally reduced using (4)He, some...
Validation of Monte Carlo calculated surface doses for megavoltage photon beams.
Abdel-Rahman, Wamied; Seuntjens, Jan P; Verhaegen, Frank; Deblois, François; Podgorsak, Ervin B
2005-01-01
Recent work has shown that there is significant uncertainty in measuring build-up doses in mega-voltage photon beams especially at high energies. In this present investigation we used a phantom-embedded extrapolation chamber (PEEC) made of Solid Water to validate Monte Carlo (MC)-calculated doses in the dose build-up region for 6 and 18 MV x-ray beams. The study showed that the percentage depth ionizations (PDIs) obtained from measurements are higher than the percentage depth doses (PDDs) obtained with Monte Carlo techniques. To validate the MC-calculated PDDs, the design of the PEEC was incorporated into the simulations. While the MC-calculated and measured PDIs in the dose build-up region agree with one another for the 6 MV beam, a non-negligible difference is observed for the 18 MV x-ray beam. A number of experiments and theoretical studies of various possible effects that could be the source of this discrepancy were performed. The contribution of contaminating neutrons and protons to the build-up dose region in the 18 MV x-ray beam is negligible. Moreover, the MC calculations using the XCOM photon cross-section database and the NIST bremsstrahlung differential cross section do not explain the discrepancy between the MC calculations and measurement in the dose build-up region for the 18 MV. A simple incorporation of triplet production events into the MC dose calculation increases the calculated doses in the build-up region but does not fully account for the discrepancy between measurement and calculations for the 18 MV x-ray beam.
Calculation of dose in homogeneous phantoms for partially attenuated photon beams
El-Khatib, E.; Podgorsak, E.B.; Pla, C.
1988-03-01
Measured and calculated dose distributions under attenuators, which are of smaller cross-sectional dimensions than the radiation field, are presented. The study was performed on a 4-MV linac at a source--surface distance of 120 cm on the beam central axis in a water phantom for several thicknesses and cross sections of lead attenuators. Dose correction factors, which are used to multiply the open beam data to get dose distributions under partial attenuators, depend strongly on attenuator parameters and on depths in phantom. A method to calculate dose correction factors for any combination of attenuator parameters and any phantom depth is presented. The calculated dose distributions under partial attenuators agree well with measured data, which indicates that the method can be applied in clinical situations.
Influence of polarization and a source model for dose calculation in MRT
Bartzsch, Stefan, E-mail: s.bartzsch@dkfz.de; Oelfke, Uwe [The Institute of Cancer Research, 15 Cotswold Road, Belmont, Sutton, Surrey SM2 5NG, United Kingdom and Deutsches Krebsforschungszentrum, Im Neuenheimer Feld 280, D-69120 Heidelberg (Germany); Lerch, Michael; Petasecca, Marco [Centre for Medical Radiation Physics, University of Wollongong, Northfields Avenue, Wollongong 2522 (Australia); Bräuer-Krisch, Elke [European Synchrotron Radiation Facility, 6 Rue Jules Horowitz, 38000 Grenoble (France)
2014-04-15
Purpose: Microbeam Radiation Therapy (MRT), an alternative preclinical treatment strategy using spatially modulated synchrotron radiation on a micrometer scale, has the great potential to cure malignant tumors (e.g., brain tumors) while having low side effects on normal tissue. Dose measurement and calculation in MRT is challenging because of the spatial accuracy required and the arising high dose differences. Dose calculation with Monte Carlo simulations is time consuming and their accuracy is still a matter of debate. In particular, the influence of photon polarization has been discussed in the literature. Moreover, it is controversial whether a complete knowledge of phase space trajectories, i.e., the simulation of the machine from the wiggler to the collimator, is necessary in order to accurately calculate the dose. Methods: With Monte Carlo simulations in the Geant4 toolkit, the authors investigate the influence of polarization on the dose distribution and the therapeutically important peak to valley dose ratios (PVDRs). Furthermore, the authors analyze in detail phase space information provided byMartínez-Rovira et al. [“Development and commissioning of a Monte Carlo photon model for the forthcoming clinical trials in microbeam radiation therapy,” Med. Phys. 39(1), 119–131 (2012)] and examine its influence on peak and valley doses. A simple source model is developed using parallel beams and its applicability is shown in a semiadjoint Monte Carlo simulation. Results are compared to measurements and previously published data. Results: Polarization has a significant influence on the scattered dose outside the microbeam field. In the radiation field, however, dose and PVDRs deduced from calculations without polarization and with polarization differ by less than 3%. The authors show that the key consequences from the phase space information for dose calculations are inhomogeneous primary photon flux, partial absorption due to inclined beam incidence outside
A Mass-Conserving 4D XCAT Phantom for Dose Calculation and Accumulation
Williams, Christopher L; Seco, Joao; James, Sara St; Mak, Raymond H; Berbeco, Ross I; Lewis, John H
2013-01-01
The XCAT phantom is a realistic 4D digital torso phantom that is widely used in imaging and therapy research. However, lung mass is not conserved between respiratory phases of the phantom, making detailed dosimetric simulations and dose accumulation unphysical. A framework is developed to correct this issue by enforcing local mass conservation in the XCAT lung. Dose calculations are performed to assess the implications of neglecting mass conservation, and to demonstrate an application of the phantom to calculate the accumulated delivered dose in an irregularly breathing patient. Monte Carlo methods are used to simulate conventional and SBRT treatment delivery. The spatial distribution of the lung dose was qualitatively changed by the use of mass conservation; however the corresponding DVH did not change significantly. Comparison of the delivered dose with 4DCT-based predictions shows similar lung metric results, however dose differences of 10% can be seen in some spatial regions. Using this tool to simulate p...
A CT-based analytical dose calculation method for HDR {sup 192}Ir brachytherapy
Poon, Emily; Verhaegen, Frank [Medical Physics Unit, McGill University, 1650 Cedar Avenue, Montreal, Quebec H3G 1A4 (Canada); Medical Physics Unit, McGill University, 1650 Cedar Avenue, Montreal, Quebec H3G 1A4 (Canada) and Department of Radiation Oncology (MAASTRO), GROW, University Hospital Maastricht, Maastricht 6229ET (Netherlands)
2009-09-15
Purpose: This article presents an analytical dose calculation method for high-dose-rate {sup 192}Ir brachytherapy, taking into account the effects of inhomogeneities and reduced photon backscatter near the skin. The adequacy of the Task Group 43 (TG-43) two-dimensional formalism for treatment planning is also assessed. Methods: The proposed method uses material composition and density data derived from computed tomography images. The primary and scatter dose distributions for each dwell position are calculated first as if the patient is an infinite water phantom. This is done using either TG-43 or a database of Monte Carlo (MC) dose distributions. The latter can be used to account for the effects of shielding in water. Subsequently, corrections for photon attenuation, scatter, and spectral variations along medium- or low-Z inhomogeneities are made according to the radiological paths determined by ray tracing. The scatter dose is then scaled by a correction factor that depends on the distances between the point of interest, the body contour, and the source position. Dose calculations are done for phantoms with tissue and lead inserts, as well as patient plans for head-and-neck, esophagus, and MammoSite balloon breast brachytherapy treatments. Gamma indices are evaluated using a dose-difference criterion of 3% and a distance-to-agreement criterion of 2 mm. PTRAN{sub C}T MC calculations are used as the reference dose distributions. Results: For the phantom with tissue and lead inserts, the percentages of the voxels of interest passing the gamma criteria (P{sub {gamma}{>=}1}) are 100% for the analytical calculation and 91% for TG-43. For the breast patient plan, TG-43 overestimates the target volume receiving the prescribed dose by 4% and the dose to the hottest 0.1 cm{sup 3} of the skin by 9%, whereas the analytical and MC results agree within 0.4%. P{sub {gamma}{>=}1} are 100% and 48% for the analytical and TG-43 calculations, respectively. For the head-and-neck and
A CT-based analytical dose calculation method for HDR 192Ir brachytherapy.
Poon, Emily; Verhaegen, Frank
2009-09-01
This article presents an analytical dose calculation method for high-dose-rate 192Ir brachytherapy, taking into account the effects of inhomogeneities and reduced photon backscatter near the skin. The adequacy of the Task Group 43 (TG-43) two-dimensional formalism for treatment planning is also assessed. The proposed method uses material composition and density data derived from computed tomography images. The primary and scatter dose distributions for each dwell position are calculated first as if the patient is an infinite water phantom. This is done using either TG-43 or a database of Monte Carlo (MC) dose distributions. The latter can be used to account for the effects of shielding in water. Subsequently, corrections for photon attenuation, scatter, and spectral variations along medium- or low-Z inhomogeneities are made according to the radiological paths determined by ray tracing. The scatter dose is then scaled by a correction factor that depends on the distances between the point of interest, the body contour, and the source position. Dose calculations are done for phantoms with tissue and lead inserts, as well as patient plans for head-and-neck, esophagus, and MammoSite balloon breast brachytherapy treatments. Gamma indices are evaluated using a dose-difference criterion of 3% and a distance-to-agreement criterion of 2 mm. PTRAN_CT MC calculations are used as the reference dose distributions. For the phantom with tissue and lead inserts, the percentages of the voxels of interest passing the gamma criteria (Pgamma > or = 1) are 100% for the analytical calculation and 91% for TG-43. For the breast patient plan, TG-43 overestimates the target volume receiving the prescribed dose by 4% and the dose to the hottest 0.1 cm3 of the skin by 9%, whereas the analytical and MC results agree within 0.4%. Pgamma > or = 1 are 100% and 48% for the analytical and TG-43 calculations, respectively. For the head-and-neck and esophagus patient plans, Pgamma > or = 1 are > or
Tholomier, M.; Vicario, E.; Doghmane, N.
1987-10-01
The contribution of backscattered electrons to Auger electrons yield was studied with a multiple scattering Monte-Carlo simulation. The Auger backscattering factor has been calculated in the 5 keV-60 keV energy range. The dependence of the Auger backscattering factor on the primary energy and the beam incidence angle were determined. Spatial distributions of backscattered electrons and Auger electrons are presented for a point incident beam. Correlations between these distributions are briefly investigated.
SU-E-T-481: In Vivo and Post Mortem Animal Irradiation: Measured Vs. Calculated Doses
Heintz, P [Univ New Mexico Radiology Dept., Albuquerque, NM (United States); Heintz, B [Texas Oncology, PA, Southlake, TX (United States); Sandoval, D [University of New Mexico, Albuquerque, NM (United States); Weber, W; Melo, D; Guilmette, R [Lovelace Respiratory Research Institute, Albuquerque, NM (United States)
2015-06-15
Purpose: Computerized radiation therapy treatment planning is performed on almost all patients today. However it is seldom used for laboratory irradiations. The first objective is to assess whether modern radiation therapy treatment planning (RTP) systems accurately predict the subject dose by comparing in vivo and decedent dose measurements to calculated doses. The other objective is determine the importance of using a RTP system for laboratory irradiations. Methods: 5 MOSFET radiation dosimeters were placed enterically in each subject (2 sedated Rhesus Macaques) to measure the absorbed dose at 5 levels (carina, lung, heart, liver and rectum) during whole body irradiation. The subjects were treated with large opposed lateral fields and extended distances to cover the entire subject using a Varian 600C linac. CT simulation was performed ante-mortem (AM) and post-mortem (PM). To compare AM and PM doses, calculation points were placed at the location of each dosimeter in the treatment plan. The measured results were compared to the results using Varian Eclipse and Prowess Panther RTP systems. Results: The Varian and Prowess treatment planning system agreed to within in +1.5% for both subjects. However there were significant differences between the measured and calculated doses. For both animals the calculated central axis dose was higher than prescribed by 3–5%. This was caused in part by inaccurate measurement of animal thickness at the time of irradiation. For one subject the doses ranged from 4% to 7% high and the other subject the doses ranged 7% to 14% high when compared to the RTP doses. Conclusions: Our results suggest that using proper CT RTP system can more accurately deliver the prescribed dose to laboratory subjects. It also shows that there is significant dose variation in such subjects when inhomogeneities are not considered in the planning process.
Method for calculating dose when lung tissue lies in the treatment field
McDonald, S.C.; Keller, B.E.; Rubin, P.
1976-07-01
The absorbed dose in lung and beyond lung as a result of increased lung transmission of x and ..gamma.. irradiation is described. The correction factor used to calculate the absorbed dose is a function of beam energy, field area, lung density, and lung and soft tissue depth. Agreement between measurements and calculations in the Alderson phantom is within 3%. An example of how this technique can be used is described. (AIP)
X-ray tube output based calculation of patient entrance surface dose: validation of the method
Harju, O.; Toivonen, M.; Tapiovaara, M.; Parviainen, T. [Radiation and Nuclear Safety Authority, Helsinki (Finland)
2003-06-01
X-ray departments need methods to monitor the doses delivered to the patients in order to be able to compare their dose level to established reference levels. For this purpose, patient dose per radiograph is described in terms of the entrance surface dose (ESD) or dose-area product (DAP). The actual measurement is often made by using a DAP-meter or thermoluminescent dosimeters (TLD). The third possibility, the calculation of ESD from the examination technique factors, is likely to be a common method for x-ray departments that do not have the other methods at their disposal or for examinations where the dose may be too low to be measured by the other means (e.g. chest radiography). We have developed a program for the determination of ESD by the calculation method and analysed the accuracy that can be achieved by this indirect method. The program calculates the ESD from the current time product, x-ray tube voltage, beam filtration and focus- to-skin distance (FSD). Additionally, for calibrating the dose calculation method and thereby improving the accuracy of the calculation, the x-ray tube output should be measured for at least one x-ray tube voltage value in each x-ray unit. The aim of the present work is to point out the restrictions of the method and details of its practical application. The first experiences from the use of the method will be summarised. (orig.)
First-Principles Calculations of Electron Transfer in Organic Molecules
Pati, Ranjit; Karna, Shashi P.
2000-03-01
Suitably tailored organic structures are considered potential candidates as components in molecular electronic devices. A common molecular architecture for electronics consists of an electron donor (D) and an electron acceptor (A) moiety bonded together by a chemically inert bridging moiety, called spacer (S). The D-S-A combination constitutes the basic component equivalent of a solid state capacitor. A useful physical property that determines the applicability of molecular structures in moletronics is the electron transfer (ET) rate, which is related, in a two-state approximation, to the coupling matrix between the two electronic states representing the localization of electrons. In an effort to model potential organic structures, we have calculated the ET coupling matrix elements in a number of D-, S-, and A-type organic molecules with the use of ab initio Hartree-Fock method and two different basis sets, namely an STO-3G and a double zeta plus polarization (DZP). A number of important findings have emerged from this study: (i) The ET coupling matrix strongly depends upon the geometrical arrangement of the molecular fragment(s) in the architecture. (ii) In an oligomeric chain, the ET matrix decreases exponentially with molecular length (number of monomer units). (iii) In cyclic alkanes, the magnitude of the ET coupling matrix decreases with increasing size of fused rings.
Graves, Yan Jiang; Jia, Xun; Jiang, Steve B
2013-03-21
The γ-index test has been commonly adopted to quantify the degree of agreement between a reference dose distribution and an evaluation dose distribution. Monte Carlo (MC) simulation has been widely used for the radiotherapy dose calculation for both clinical and research purposes. The goal of this work is to investigate both theoretically and experimentally the impact of the MC statistical fluctuation on the γ-index test when the fluctuation exists in the reference, the evaluation, or both dose distributions. To the first order approximation, we theoretically demonstrated in a simplified model that the statistical fluctuation tends to overestimate γ-index values when existing in the reference dose distribution and underestimate γ-index values when existing in the evaluation dose distribution given the original γ-index is relatively large for the statistical fluctuation. Our numerical experiments using realistic clinical photon radiation therapy cases have shown that (1) when performing a γ-index test between an MC reference dose and a non-MC evaluation dose, the average γ-index is overestimated and the gamma passing rate decreases with the increase of the statistical noise level in the reference dose; (2) when performing a γ-index test between a non-MC reference dose and an MC evaluation dose, the average γ-index is underestimated when they are within the clinically relevant range and the gamma passing rate increases with the increase of the statistical noise level in the evaluation dose; (3) when performing a γ-index test between an MC reference dose and an MC evaluation dose, the gamma passing rate is overestimated due to the statistical noise in the evaluation dose and underestimated due to the statistical noise in the reference dose. We conclude that the γ-index test should be used with caution when comparing dose distributions computed with MC simulation.
Convergent close-coupling calculations of electron-helium scattering
Fursa, D.V.; Bray, I. [Flinders Univ. of South Australia, Adelaide, SA (Australia). Electronic Structure of Materials Centre
1996-11-01
We present a review of the recent electron-helium calculations and experiments concentrating on the extensive application of the convergent close-coupling (CCC) method. Elastic, excitation, and ionization processes are considered, as well as excitation of the metastable states. The present status of agreement between theory and experiment for elastic and discrete excitations of the ground state is, in our view, quite satisfactory. However, discrepancies for excitation of the metastable states are substantial and invite urgent attention. Application of the CCC method to the calculation of differential ionization cross sections is encouraging, but also shows some fundamental difficulties. (authors). 92 refs., 15 figs.
Convergent close-coupling calculations of electron-helium scattering
Fursa, D.V.; Bray, I. [Flinders Univ. of South Australia, Adelaide, SA (Australia). Electronic Structure of Materials Centre
1996-11-01
We present a review of the recent electron-helium calculations and experiments concentrating on the extensive application of the convergent close-coupling (CCC) method. Elastic, excitation, and ionization processes are considered, as well as excitation of the metastable states. The present status of agreement between theory and experiment for elastic and discrete excitations of the ground state is, in our view, quite satisfactory. However, discrepancies for excitation of the metastable states are substantial and invite urgent attention. Application of the CCC method to the calculation of differential ionization cross sections is encouraging, but also shows some fundamental difficulties. (authors). 92 refs., 15 figs.
Chibani, Omar, E-mail: omar.chibani@fccc.edu; C-M Ma, Charlie [Fox Chase Cancer Center, Philadelphia, Pennsylvania 19111 (United States)
2014-05-15
Purpose: To present a new accelerated Monte Carlo code for CT-based dose calculations in high dose rate (HDR) brachytherapy. The new code (HDRMC) accounts for both tissue and nontissue heterogeneities (applicator and contrast medium). Methods: HDRMC uses a fast ray-tracing technique and detailed physics algorithms to transport photons through a 3D mesh of voxels representing the patient anatomy with applicator and contrast medium included. A precalculated phase space file for the{sup 192}Ir source is used as source term. HDRM is calibrated to calculated absolute dose for real plans. A postprocessing technique is used to include the exact density and composition of nontissue heterogeneities in the 3D phantom. Dwell positions and angular orientations of the source are reconstructed using data from the treatment planning system (TPS). Structure contours are also imported from the TPS to recalculate dose-volume histograms. Results: HDRMC was first benchmarked against the MCNP5 code for a single source in homogenous water and for a loaded gynecologic applicator in water. The accuracy of the voxel-based applicator model used in HDRMC was also verified by comparing 3D dose distributions and dose-volume parameters obtained using 1-mm{sup 3} versus 2-mm{sup 3} phantom resolutions. HDRMC can calculate the 3D dose distribution for a typical HDR cervix case with 2-mm resolution in 5 min on a single CPU. Examples of heterogeneity effects for two clinical cases (cervix and esophagus) were demonstrated using HDRMC. The neglect of tissue heterogeneity for the esophageal case leads to the overestimate of CTV D90, CTV D100, and spinal cord maximum dose by 3.2%, 3.9%, and 3.6%, respectively. Conclusions: A fast Monte Carlo code for CT-based dose calculations which does not require a prebuilt applicator model is developed for those HDR brachytherapy treatments that use CT-compatible applicators. Tissue and nontissue heterogeneities should be taken into account in modern HDR
Sensitivity of MOS transistors to gamma and electron doses
Petr, I. (Ceske Vysoke Uceni Technicke, Prague (Czechoslovakia). Fakulta Jaderna a Fysikalne Inzenyrska); Gilar, O. (Tesla, Premysleni (Czechoslovakia). Vyzkumny Ustav Pristroju Jaderne Techniky)
1985-03-01
The threshold shift has been as a function of the properties of the SiO/sub 2/ layer, viz. oxide thickness and diffusion length of the charge carriers formed in the oxide layer. Experimental results obtained from gamma, fast electron, and fast neutron irradiation of MOS transistors having different oxide thickness are given. The theoretical results are compared with the measured data. The sensitivity of MOS transistors to gamma rays and fast electrons is given together with the energy dependence of radiation doses. The application of MOS transistors as dosemeters is discussed.
王岩国; 刘红荣; 杨奇斌; 张泽
2003-01-01
Off-axis electron holography in a field emission gun transmission-electron microscope and electron dynamic calculation are used to determine the absorption coefficient and inelastic mean free path (IMFP) of copper.Dependence of the phase shift of the exit electron wave on the specimen thickness is established by electron dynamic simulation. The established relationship makes it possible to determine the specimen thickness with the calculated phase shift by match of the phase shift measured in the reconstructed phase image. Based on the measured amplitudes in reconstructed exit electron wave and reference wave in the vacuum, the examined IMFP of electron with energy of 200kV in Cu is obtained to be 96nm.
GPUMCD: a new GPU-oriented Monte Carlo dose calculation platform
Hissoiny, Sami; Ozell, Benoît; Després, Philippe
2011-01-01
Purpose: Monte Carlo methods are considered the gold standard for dosimetric computations in radiotherapy. Their execution time is however still an obstacle to the routine use of Monte Carlo packages in a clinical setting. To address this problem, a completely new, and designed from the ground up for the GPU, Monte Carlo dose calculation package for voxelized geometries is proposed: GPUMCD. Method : GPUMCD implements a coupled photon-electron Monte Carlo simulation for energies in the range 0.01 MeV to 20 MeV. An analogue simulation of photon interactions is used and a Class II condensed history method has been implemented for the simulation of electrons. A new GPU random number generator, some divergence reduction methods as well as other optimization strategies are also described. GPUMCD was run on a NVIDIA GTX480 while single threaded implementations of EGSnrc and DPM were run on an Intel Core i7 860. Results : Dosimetric results obtained with GPUMCD were compared to EGSnrc. In all but one test case, 98% o...
Molinelli, Silvia; Mairani, Andrea; Matsufuji, Naruhiro; Kanematsu, Nobuyuki; Inaniwa, Taku; Mirandola, Alfredo; Russo, Stefania; Mastella, Edoardo; Hasegawa, Azusa; Tsuji, Hiroshi; Yamada, Shigeru; Vischioni, Barbara; Vitolo, Viviana; Ferrari, Alfredo; Ciocca, Mario; Kamada, Tadashi; Tsujii, Hirohiko; Orecchia, Roberto; Fossati, Piero
2016-01-01
Background and purpose: In carbon ion radiotherapy (CIRT), the use of different relative biological effectiveness (RBE) models in the RBE-weighted dose $(D_{RBE})$ calculation can lead to deviations in the physical dose $(D_{phy})$ delivered to the patient. Our aim is to reduce target $D_{phy}$ deviations by converting prescription dose values. Material and methods: Planning data of patients treated at the National Institute of Radiological Sciences (NIRS) were collected, with prescribed doses per fraction ranging from 3.6 Gy (RBE) to 4.6 Gy (RBE), according to the Japanese semi-empirical model. The $D_{phy}$ was Monte Carlo (MC) re-calculated simulating the NIRS beamline. The local effect model (LEM)_I was then applied to estimate $D_{RBE}$. Target median $D_{RBE}$ ratios between MC + LEM_I and NIRS plans determined correction factors for the conversion of prescription doses. Plans were re-optimized in a LEM_I-based commercial system, prescribing the NIRS uncorrected and corrected $D_{RBE}$. Results: The MC ...
Using electron microscopy to calculate optical properties of biological samples
Wu, Wenli; Radosevich, Andrew J.; Eshein, Adam; Nguyen, The-Quyen; Yi, Ji; Cherkezyan, Lusik; Roy, Hemant K.; Szleifer, Igal; Backman, Vadim
2016-01-01
The microscopic structural origins of optical properties in biological media are still not fully understood. Better understanding these origins can serve to improve the utility of existing techniques and facilitate the discovery of other novel techniques. We propose a novel analysis technique using electron microscopy (EM) to calculate optical properties of specific biological structures. This method is demonstrated with images of human epithelial colon cell nuclei. The spectrum of anisotropy...
Assessment of doses caused by electrons in thin layers of tissue-equivalent materials, using MCNP.
Heide, Bernd
2013-10-01
Absorbed doses caused by electron irradiation were calculated with Monte Carlo N-Particle transport code (MCNP) for thin layers of tissue-equivalent materials. The layers were so thin that the calculation of energy deposition was on the border of the scope of MCNP. Therefore, in this article application of three different methods of calculation of energy deposition is discussed. This was done by means of two scenarios: in the first one, electrons were emitted from the centre of a sphere of water and also recorded in that sphere; and in the second, an irradiation with the PTB Secondary Standard BSS2 was modelled, where electrons were emitted from an (90)Sr/(90)Y area source and recorded inside a cuboid phantom made of tissue-equivalent material. The speed and accuracy of the different methods were of interest. While a significant difference in accuracy was visible for one method in the first scenario, the difference in accuracy of the three methods was insignificant for the second one. Considerable differences in speed were found for both scenarios. In order to demonstrate the need for calculating the dose in thin small zones, a third scenario was constructed and simulated as well. The third scenario was nearly equal to the second one, but a pike of lead was assumed to be inside the phantom in addition. A dose enhancement (caused by the pike of lead) of ∼113 % was recorded for a thin hollow cylinder at a depth of 0.007 cm, which the basal-skin layer is referred to in particular. Dose enhancements between 68 and 88 % were found for a slab with a radius of 0.09 cm for all depths. All dose enhancements were hardly noticeable for a slab with a cross-sectional area of 1 cm(2), which is usually applied to operational radiation protection.
TH-A-19A-09: Towards Sub-Second Proton Dose Calculation On GPU
Silva, J da [University of Cambridge, Cambridge, Cambridgeshire (United Kingdom)
2014-06-15
Purpose: To achieve sub-second dose calculation for clinically relevant proton therapy treatment plans. Rapid dose calculation is a key component of adaptive radiotherapy, necessary to take advantage of the better dose conformity offered by hadron therapy. Methods: To speed up proton dose calculation, the pencil beam algorithm (PBA; clinical standard) was parallelised and implemented to run on a graphics processing unit (GPU). The implementation constitutes the first PBA to run all steps on GPU, and each part of the algorithm was carefully adapted for efficiency. Monte Carlo (MC) simulations obtained using Fluka of individual beams of energies representative of the clinical range impinging on simple geometries were used to tune the PBA. For benchmarking, a typical skull base case with a spot scanning plan consisting of a total of 8872 spots divided between two beam directions of 49 energy layers each was provided by CNAO (Pavia, Italy). The calculations were carried out on an Nvidia Geforce GTX680 desktop GPU with 1536 cores running at 1006 MHz. Results: The PBA reproduced within ±3% of maximum dose results obtained from MC simulations for a range of pencil beams impinging on a water tank. Additional analysis of more complex slab geometries is currently under way to fine-tune the algorithm. Full calculation of the clinical test case took 0.9 seconds in total, with the majority of the time spent in the kernel superposition step. Conclusion: The PBA lends itself well to implementation on many-core systems such as GPUs. Using the presented implementation and current hardware, sub-second dose calculation for a clinical proton therapy plan was achieved, opening the door for adaptive treatment. The successful parallelisation of all steps of the calculation indicates that further speedups can be expected with new hardware, brightening the prospects for real-time dose calculation. This work was funded by ENTERVISION, European Commission FP7 grant 264552.
Bahreyni Toossi, Mohammad Taghi; Ghorbani, Mahdi; Akbari, Fatemeh; Mehrpouyan, Mohammad; Sobhkhiz Sabet, Leila
2016-03-01
The aim of this study is to evaluate the effect of tooth and dental restoration materials on electron dose distribution and photon contamination production in electron beams of a medical linac. This evaluation was performed on 8, 12 and 14 MeV electron beams of a Siemens Primus linac. MCNPX Monte Carlo code was utilized and a 10 × 10 cm(2) applicator was simulated in the cases of tooth and combinations of tooth and Ceramco C3 ceramic veneer, tooth and Eclipse alloy and tooth and amalgam restoration materials in a soft tissue phantom. The relative electron and photon contamination doses were calculated for these materials. The presence of tooth and dental restoration material changed the electron dose distribution and photon contamination in phantom, depending on the type of the restoration material and electron beam's energy. The maximum relative electron dose was 1.07 in the presence of tooth including amalgam for 14 MeV electron beam. When 100.00 cGy was prescribed for the reference point, the maximum absolute electron dose was 105.10 cGy in the presence of amalgam for 12 MeV electron beam and the maximum absolute photon contamination dose was 376.67 μGy for tooth in 14 MeV electron beam. The change in electron dose distribution should be considered in treatment planning, when teeth are irradiated in electron beam radiotherapy. If treatment planning can be performed in such a way that the teeth are excluded from primary irradiation, the potential errors in dose delivery to the tumour and normal tissues can be avoided.
Fast Monte Carlo Simulation for Patient-specific CT/CBCT Imaging Dose Calculation
Jia, Xun; Gu, Xuejun; Jiang, Steve B
2011-01-01
Recently, X-ray imaging dose from computed tomography (CT) or cone beam CT (CBCT) scans has become a serious concern. Patient-specific imaging dose calculation has been proposed for the purpose of dose management. While Monte Carlo (MC) dose calculation can be quite accurate for this purpose, it suffers from low computational efficiency. In response to this problem, we have successfully developed a MC dose calculation package, gCTD, on GPU architecture under the NVIDIA CUDA platform for fast and accurate estimation of the x-ray imaging dose received by a patient during a CT or CBCT scan. Techniques have been developed particularly for the GPU architecture to achieve high computational efficiency. Dose calculations using CBCT scanning geometry in a homogeneous water phantom and a heterogeneous Zubal head phantom have shown good agreement between gCTD and EGSnrc, indicating the accuracy of our code. In terms of improved efficiency, it is found that gCTD attains a speed-up of ~400 times in the homogeneous water ...
Villeneuve, David; Brothers, Adam; Harvey, Eric; Kemna, Mariska; Law, Yuk; Nemeth, Thomas; Gantt, Soren
2013-02-01
Pediatric valganciclovir dosing recommendations have not been extensively validated for prevention or treatment for CMV infection. As such, we performed a pharmacokinetic study to compare different valganciclovir dosing regimens and the potential benefits of individualized dose adjustments in children following organ transplantation. Ganciclovir AUCs were calculated from four plasma drug levels in pediatric SOT recipients aged six months through three yr receiving valganciclovir suspension by mouth. Of the 28 ganciclovir AUC calculations performed, 11 (39%) were outside the therapeutic target range of 40-60 mcg h/L leading to a valganciclovir dose adjustment. Current manufacturer-recommended dosing based on BSA and CrCl was estimated to result in therapeutic AUCs in fewer patients than the simple weight-based formula used in our institution (4 vs. 13; p = 0.017). An AUC calculation using only the two- and five-h measurements was strongly correlated with the AUC using all four time measurements (R(2) = 0.846; p < 0.001). A simple weight-based dosing approach gives a higher probability for therapeutic AUCs compared to the manufacturer-recommended dosing in pediatric transplant patients aged six months through three yr with normal renal function. An AUC calculated using two sample times might allow for fewer blood draws in the future. © 2012 John Wiley & Sons A/S.
Electronic structure calculations of ESR parameters of melanin units.
Batagin-Neto, Augusto; Bronze-Uhle, Erika Soares; Graeff, Carlos Frederico de Oliveira
2015-03-21
Melanins represent an important class of natural pigments present in plants and animals that are currently considered to be promising materials for applications in optic and electronic devices. Despite their interesting properties, some of the basic features of melanins are not satisfactorily understood, including the origin of their intrinsic paramagnetism. A number of experiments have been performed to investigate the electron spin resonance (ESR) response of melanin derivatives, but until now, there has been no consensus regarding the real structure of the paramagnetic centers involved. In this work, we have employed electronic structure calculations to evaluate the ESR parameters of distinct melanin monomers and dimers in order to identify the possible structures associated with unpaired spins in this biopolymer. The g-factors and hyperfine constants of the cationic, anionic and radicalar structures were investigated. The results confirm the existence of at least two distinct paramagnetic centers in melanin structure, identifying the chemical species associated with them and their roles in electrical conductivity.
Wilson, J. W.; Khandelwal, G. S.
1976-01-01
Calculational methods for estimation of dose from external proton exposure of arbitrary convex bodies are briefly reviewed. All the necessary information for the estimation of dose in soft tissue is presented. Special emphasis is placed on retaining the effects of nuclear reaction, especially in relation to the dose equivalent. Computer subroutines to evaluate all of the relevant functions are discussed. Nuclear reaction contributions for standard space radiations are in most cases found to be significant. Many of the existing computer programs for estimating dose in which nuclear reaction effects are neglected can be readily converted to include nuclear reaction effects by use of the subroutines described herein.
FOOD: an interactive code to calculate internal radiation doses from contaminated food products
Baker, D.A.; Hoenes, G.R.; Soldat, J.K.
1976-01-01
An interactive code, FOOD, has been written in BASIC for the UNIVAC 1108 to facilitate calculation of internal radiation doses to man from radionuclides in food products. In the dose model, vegetation may be contaminated by either air or irrigation water containing radionuclides. The model considers two mechanisms for radionuclide contamination of vegetation: direct deposition on leaves and uptake from soil through the root system. The user may select up to 14 food categories with corresponding consumption rates, growing periods and either irrigation rates or atmospheric deposition rates. These foods include various kinds of produce, grains and animal products. At present, doses may be calculated for the skin, total body and five internal organs from 190 radionuclides. Dose summaries can be displayed at the local terminal. Further details on percent contribution to dose by nuclide and by food type are available from an auxiliary high-speed printer. This output also includes estimated radionuclide concentrations in soil, plants and animal products.
Dose calculation using a numerical method based on Haar wavelets integration
Belkadhi, K., E-mail: khaled.belkadhi@ult-tunisie.com [Unité de Recherche de Physique Nucléaire et des Hautes Énergies, Faculté des Sciences de Tunis, Université Tunis El-Manar (Tunisia); Manai, K. [Unité de Recherche de Physique Nucléaire et des Hautes Énergies, Faculté des Sciences de Tunis, Université Tunis El-Manar (Tunisia); College of Science and Arts, University of Bisha, Bisha (Saudi Arabia)
2016-03-11
This paper deals with the calculation of the absorbed dose in an irradiation cell of gamma rays. Direct measurement and simulation have shown that they are expensive and time consuming. An alternative to these two operations is numerical methods, a quick and efficient way can furnish an estimation of the absorbed dose by giving an approximation of the photon flux at a specific point of space. To validate the numerical integration method based on the Haar wavelet for absorbed dose estimation, a study with many configurations was performed. The obtained results with the Haar wavelet method showed a very good agreement with the simulation highlighting good efficacy and acceptable accuracy. - Highlights: • A numerical integration method using Haar wavelets is detailed. • Absorbed dose is estimated with Haar wavelets method. • Calculated absorbed dose using Haar wavelets and Monte Carlo simulation using Geant4 are compared.
Pla, C.; Podgorsak, E.B.; El-Khatib, E.
1988-07-01
Measured and calculated dose profiles under partial attenuators which cover only part of the radiation beam are presented. The study was performed for x-ray beams generated with a 4-MV linear accelerator at a source--surface distance of 120 cm in a water phantom for lead attenuators of arbitrary shape but constant thickness. Dose correction factors, which are used to multiply the open beam data to predict doses under partial attenuators, depend strongly on attenuator parameters, such as its thickness, lateral dimensions, and distance from phantom or patient surface, in addition to depending on depths in the phantom. The dose correction factors are calculated with Clarkson sector integration techniques, and the results, in spite of the simplifying assumptions used in the algorithm, generally agree with measured data to within 3%. The calculational method therefore may be applied to general clinical situations in which partial attenuators are used.
Klüter, Sebastian, E-mail: sebastian.klueter@med.uni-heidelberg.de; Schubert, Kai; Lissner, Steffen; Sterzing, Florian; Oetzel, Dieter; Debus, Jürgen [Department of Radiation Oncology, Heidelberg University Hospital, Im Neuenheimer Feld 400, 69120 Heidelberg, Germany, and Heidelberg Institute for Radiation Oncology (HIRO), Im Neuenheimer Feld 400, 69120 Heidelberg, Germany, and German Consortium for Translational Cancer Research (DKTK), Im Neuenheimer Feld 400, 69120 Heidelberg (Germany); Schlegel, Wolfgang [German Cancer Research Center (DKFZ), Im Neuenheimer Feld 280, 69120 Heidelberg (Germany); Oelfke, Uwe [German Cancer Research Center (DKFZ), Im Neuenheimer Feld 280, 69120 Heidelberg, Germany and Joint Department of Physics at The Institute of Cancer Research and The Royal Marsden NHS Foundation Trust, London SM2 5NG (United Kingdom); Nill, Simeon [Joint Department of Physics at The Institute of Cancer Research and The Royal Marsden NHS Foundation Trust, London SM2 5NG (United Kingdom)
2014-08-15
Purpose: The dosimetric verification of treatment plans in helical tomotherapy usually is carried out via verification measurements. In this study, a method for independent dose calculation of tomotherapy treatment plans is presented, that uses a conventional treatment planning system with a pencil kernel dose calculation algorithm for generation of verification dose distributions based on patient CT data. Methods: A pencil beam algorithm that directly uses measured beam data was configured for dose calculation for a tomotherapy machine. Tomotherapy treatment plans were converted into a format readable by an in-house treatment planning system by assigning each projection to one static treatment field and shifting the calculation isocenter for each field in order to account for the couch movement. The modulation of the fluence for each projection is read out of the delivery sinogram, and with the kernel-based dose calculation, this information can directly be used for dose calculation without the need for decomposition of the sinogram. The sinogram values are only corrected for leaf output and leaf latency. Using the converted treatment plans, dose was recalculated with the independent treatment planning system. Multiple treatment plans ranging from simple static fields to real patient treatment plans were calculated using the new approach and either compared to actual measurements or the 3D dose distribution calculated by the tomotherapy treatment planning system. In addition, dose–volume histograms were calculated for the patient plans. Results: Except for minor deviations at the maximum field size, the pencil beam dose calculation for static beams agreed with measurements in a water tank within 2%/2 mm. A mean deviation to point dose measurements in the cheese phantom of 0.89% ± 0.81% was found for unmodulated helical plans. A mean voxel-based deviation of −0.67% ± 1.11% for all voxels in the respective high dose region (dose values >80%), and a mean local
A dose optimization method for electron radiotherapy using randomized aperture beams.
Engel, Konrad; Gauer, Tobias
2009-09-01
The present paper describes the entire optimization process of creating a radiotherapy treatment plan for advanced electron irradiation. Special emphasis is devoted to the selection of beam incidence angles and beam energies as well as to the choice of appropriate subfields generated by a refined version of intensity segmentation and a novel random aperture approach. The algorithms have been implemented in a stand-alone programme using dose calculations from a commercial treatment planning system. For this study, the treatment planning system Pinnacle from Philips has been used and connected to the optimization programme using an ASCII interface. Dose calculations in Pinnacle were performed by Monte Carlo simulations for a remote-controlled electron multileaf collimator (MLC) from Euromechanics. As a result, treatment plans for breast cancer patients could be significantly improved when using randomly generated aperture beams. The combination of beams generated through segmentation and randomization achieved the best results in terms of target coverage and sparing of critical organs. The treatment plans could be further improved by use of a field reduction treatment plans could be further improved by use of a field reduction algorithm. Without a relevant loss in dose distribution, the total number of MLC fields and monitor units could be reduced by up to 20%. In conclusion, using randomized aperture beams is a promising new approach in radiotherapy and exhibits potential for further improvements in dose optimization through a combination of randomized electron and photon aperture beams.
Fast pencil beam dose calculation for proton therapy using a double-Gaussian beam model
Joakim eda Silva
2015-12-01
Full Text Available The highly conformal dose distributions produced by scanned proton pencil beams are more sensitive to motion and anatomical changes than those produced by conventional radiotherapy. The ability to calculate the dose in real time as it is being delivered would enable, for example, online dose monitoring, and is therefore highly desirable. We have previously described an implementation of a pencil beam algorithm running on graphics processing units (GPUs intended specifically for online dose calculation. Here we present an extension to the dose calculation engine employing a double-Gaussian beam model to better account for the low-dose halo. To the best of our knowledge, it is the first such pencil beam algorithm for proton therapy running on a GPU. We employ two different parametrizations for the halo dose, one describing the distribution of secondary particles from nuclear interactions found in the literature and one relying on directly fitting the model to Monte Carlo simulations of pencil beams in water. Despite the large width of the halo contribution, we show how in either case the second Gaussian can be included whilst prolonging the calculation of the investigated plans by no more than 16%, or the calculation of the most time-consuming energy layers by about 25%. Further, the calculation time is relatively unaffected by the parametrization used, which suggests that these results should hold also for different systems. Finally, since the implementation is based on an algorithm employed by a commercial treatment planning system, it is expected that with adequate tuning, it should be able to reproduce the halo dose from a general beam line with sufficient accuracy.
Patient-specific dose calculations for pediatric CT of the chest, abdomen and pelvis
Kost, Susan D.; Carver, Diana E.; Stabin, Michael G. [Vanderbilt University, Physics and Astronomy Department, Nashville, TN (United States); Vanderbilt University Medical Center, Department of Radiology and Radiological Sciences, Nashville, TN (United States); Fraser, Nicholas D.; Pickens, David R.; Price, Ronald R.; Hernanz-Schulman, Marta [Vanderbilt University Medical Center, Department of Radiology and Radiological Sciences, Nashville, TN (United States)
2015-11-15
Organ dose is essential for accurate estimates of patient dose from CT. To determine organ doses from a broad range of pediatric patients undergoing diagnostic chest-abdomen-pelvis CT and investigate how these relate to patient size. We used a previously validated Monte Carlo simulation model of a Philips Brilliance 64 multi-detector CT scanner (Philips Healthcare, Best, The Netherlands) to calculate organ doses for 40 pediatric patients (M:F = 21:19; range 0.6-17 years). Organ volumes and positions were determined from the images using standard segmentation techniques. Non-linear regression was performed to determine the relationship between volume CT dose index (CTDI{sub vol})-normalized organ doses and abdominopelvic diameter. We then compared results with values obtained from independent studies. We found that CTDI{sub vol}-normalized organ dose correlated strongly with exponentially decreasing abdominopelvic diameter (R{sup 2} > 0.8 for most organs). A similar relationship was determined for effective dose when normalized by dose-length product (R{sup 2} = 0.95). Our results agreed with previous studies within 12% using similar scan parameters (e.g., bowtie filter size, beam collimation); however results varied up to 25% when compared to studies using different bowtie filters. Our study determined that organ doses can be estimated from measurements of patient size, namely body diameter, and CTDI{sub vol} prior to CT examination. This information provides an improved method for patient dose estimation. (orig.)
Estimates of Columbia River radionuclide concentrations: Data for Phase 1 dose calculations
Richmond, M.C.; Walters, W.H.
1991-05-01
Pacific Northwest Laboratory is conducting the Hanford Environmental Dose Reconstruction Project to estimate the radiation doses people may have received from historical Hanford Site operations. Under the direction of an independent Technical Steering Panel, the project is being conducted in phases. The objective of the first phase is to assess the feasibility of the project-wide technical approach for acquiring data and developing models needed to calculate potential radiation doses. This report summarizes data that were generated for the Phase 1 dose calculations. These included monthly average concentrations of specific radionuclides in Columbia River water and sediments between Priest Rapids Dam and McNary Dam for the years 1964 to 1966. Nine key radionuclides were selected for analysis based on estimation of their contribution to dose. Concentrations of these radionuclides in the river were estimated using existing measurements and hydraulic calculations based on the simplifying assumption that dilution and decay were the primary processes controlling the fate of radionuclides released to the river. Five sub-reaches between Priest Rapids Dam and McNary Dam, corresponding to population centers and tributary confluences, were identified and monthly average radionuclide concentrations were calculated for each sub-reach. The hydraulic calculations were performed to provide radionuclide concentration estimates for time periods and geographic locations where measured data were not available. The validity of the calculation method will be evaluated in Phase 2. 12 refs., 13 figs., 49 tabs.
Model-based dose calculations for {sup 125}I lung brachytherapy
Sutherland, J. G. H.; Furutani, K. M.; Garces, Y. I.; Thomson, R. M. [Carleton Laboratory for Radiotherapy Physics, Department of Physics, Carleton University, Ottawa, Ontario K1S 5B6 (Canada); Department of Radiation Oncology, Mayo Clinic College of Medicine, Rochester, Minnesota 55905 (United States); Carleton Laboratory for Radiotherapy Physics, Department of Physics, Carleton University, Ottawa, Ontario K1S 5B6 (Canada)
2012-07-15
Purpose: Model-baseddose calculations (MBDCs) are performed using patient computed tomography (CT) data for patients treated with intraoperative {sup 125}I lung brachytherapy at the Mayo Clinic Rochester. Various metallic artifact correction and tissue assignment schemes are considered and their effects on dose distributions are studied. Dose distributions are compared to those calculated under TG-43 assumptions. Methods: Dose distributions for six patients are calculated using phantoms derived from patient CT data and the EGSnrc user-code BrachyDose. {sup 125}I (GE Healthcare/Oncura model 6711) seeds are fully modeled. Four metallic artifact correction schemes are applied to the CT data phantoms: (1) no correction, (2) a filtered back-projection on a modified virtual sinogram, (3) the reassignment of CT numbers above a threshold in the vicinity of the seeds, and (4) a combination of (2) and (3). Tissue assignment is based on voxel CT number and mass density is assigned using a CT number to mass density calibration. Three tissue assignment schemes with varying levels of detail (20, 11, and 5 tissues) are applied to metallic artifact corrected phantoms. Simulations are also performed under TG-43 assumptions, i.e., seeds in homogeneous water with no interseed attenuation. Results: Significant dose differences (up to 40% for D{sub 90}) are observed between uncorrected and metallic artifact corrected phantoms. For phantoms created with metallic artifact correction schemes (3) and (4), dose volume metrics are generally in good agreement (less than 2% differences for all patients) although there are significant local dose differences. The application of the three tissue assignment schemes results in differences of up to 8% for D{sub 90}; these differences vary between patients. Significant dose differences are seen between fully modeled and TG-43 calculations with TG-43 underestimating the dose (up to 36% in D{sub 90}) for larger volumes containing higher proportions of
SU-E-I-28: Evaluating the Organ Dose From Computed Tomography Using Monte Carlo Calculations
Ono, T; Araki, F [Faculty of Life Sciences, Kumamoto University, Kumamoto (Japan)
2014-06-01
Purpose: To evaluate organ doses from computed tomography (CT) using Monte Carlo (MC) calculations. Methods: A Philips Brilliance CT scanner (64 slice) was simulated using the GMctdospp (IMPS, Germany) based on the EGSnrc user code. The X-ray spectra and a bowtie filter for MC simulations were determined to coincide with measurements of half-value layer (HVL) and off-center ratio (OCR) profile in air. The MC dose was calibrated from absorbed dose measurements using a Farmer chamber and a cylindrical water phantom. The dose distribution from CT was calculated using patient CT images and organ doses were evaluated from dose volume histograms. Results: The HVLs of Al at 80, 100, and 120 kV were 6.3, 7.7, and 8.7 mm, respectively. The calculated HVLs agreed with measurements within 0.3%. The calculated and measured OCR profiles agreed within 3%. For adult head scans (CTDIvol) =51.4 mGy), mean doses for brain stem, eye, and eye lens were 23.2, 34.2, and 37.6 mGy, respectively. For pediatric head scans (CTDIvol =35.6 mGy), mean doses for brain stem, eye, and eye lens were 19.3, 24.5, and 26.8 mGy, respectively. For adult chest scans (CTDIvol=19.0 mGy), mean doses for lung, heart, and spinal cord were 21.1, 22.0, and 15.5 mGy, respectively. For adult abdominal scans (CTDIvol=14.4 mGy), the mean doses for kidney, liver, pancreas, spleen, and spinal cord were 17.4, 16.5, 16.8, 16.8, and 13.1 mGy, respectively. For pediatric abdominal scans (CTDIvol=6.76 mGy), mean doses for kidney, liver, pancreas, spleen, and spinal cord were 8.24, 8.90, 8.17, 8.31, and 6.73 mGy, respectively. In head scan, organ doses were considerably different from CTDIvol values. Conclusion: MC dose distributions calculated by using patient CT images are useful to evaluate organ doses absorbed to individual patients.
SU-E-T-464: Implementation and Validation of 4D Acuros XB Dose Calculations
Thomas, S; Yuen, C [BC Cancer Agency, Vancouver, BC (Canada); Huang, V [BC Cancer Agency, Surrey, BC (United Kingdom); Milette, M; Teke, T [BC Cancer Agency-Southern Interior, Kelowna, BC (Canada)
2015-06-15
Purpose: In this abstract we implement and validate a 4D VMAT Acuros XB dose calculation using Gafchromic film. Special attention is paid to the physical material assignment in the CT dataset and to reported dose to water and dose to medium. Methods: A QUASAR phantom with a 3 cm sinusoidal tumor motion and 5 second period was scanned using 4D computed tomography. A CT was also obtained of the static QUASAR phantom with the tumor at the central position. A VMAT plan was created on the average CT dataset and was delivered on a Varian TrueBeam linear accelerator. The trajectory log file from this treatment was acquired and used to create 10 VMAT subplans (one for each portion of the breathing cycle). Motion for each subplan was simulated by moving the beam isocentre in the superior/inferior direction in the Treatment Planning System on the static CT scan. The 10 plans were calculated (both dose to medium and dose to water) and summed for 1) the original HU values from the static CT scan and 2) the correct physical material assignment in the CT dataset. To acquire a breathing phase synchronized film measurements the trajectory log was used to create a VMAT delivery plan which includes dynamic couch motion using the Developer Mode. Three different treatment start phases were investigated (mid inhalation, full inhalation and full exhalation). Results: For each scenario the coronal dose distributions were measured using Gafchromic film and compared to the corresponding calculation with Film QA Pro Software using a Gamma test with a 3%/3mm distance to agreement criteria. Good agreement was found between calculation and measurement. No statistically significant difference in agreement was found between calculations to original HU values vs calculations to over-written (material-assigned) HU values. Conclusion: The investigated 4D dose calculation method agrees well with measurement.
A general model for stray dose calculation of static and intensity-modulated photon radiation
Hauri, Pascal, E-mail: pascal.hauri2@uzh.ch; Schneider, Uwe [Faculty of Science, University of Zurich, Zurich 8057, Switzerland and Radiotherapy Hirslanden, Hirslanden Medical Center, Aarau 5000 (Switzerland); Hälg, Roger A.; Besserer, Jürgen [Radiotherapy Hirslanden, Hirslanden Medical Center, Aarau 5000 (Switzerland)
2016-04-15
Purpose: There is an increasing number of cancer survivors who are at risk of developing late effects caused by ionizing radiation such as induction of second tumors. Hence, the determination of out-of-field dose for a particular treatment plan in the patient’s anatomy is of great importance. The purpose of this study was to analytically model the stray dose according to its three major components. Methods: For patient scatter, a mechanistic model was developed. For collimator scatter and head leakage, an empirical approach was used. The models utilize a nominal beam energy of 6 MeV to describe two linear accelerator types of a single vendor. The parameters of the models were adjusted using ionization chamber measurements registering total absorbed dose in simple geometries. Whole-body dose measurements using thermoluminescent dosimeters in an anthropomorphic phantom for static and intensity-modulated treatment plans were compared to the 3D out-of-field dose distributions calculated by a combined model. Results: The absolute mean difference between the whole-body predicted and the measured out-of-field dose of four different plans was 11% with a maximum difference below 44%. Computation time of 36 000 dose points for one field was around 30 s. By combining the model-calculated stray dose with the treatment planning system dose, the whole-body dose distribution can be viewed in the treatment planning system. Conclusions: The results suggest that the model is accurate, fast and can be used for a wide range of treatment modalities to calculate the whole-body dose distribution for clinical analysis. For similar energy spectra, the mechanistic patient scatter model can be used independently of treatment machine or beam orientation.
Suresh Rana
2013-01-01
Full Text Available Purpose: The purpose of this study is to assess the dosimetric impact of Acuros XB dose calculation algorithm (AXB, in comparisons with Anisotropic Analytical Algorithm (AAA calculations in prostate cancer treatment using RapidArc. Materials and Methods: A computed tomography (CT dataset of low-risk prostate cancer patients treated at Arizona Center for Cancer Care was selected and contoured for prostate, seminal vesicles, and organs at risk (OARs(rectum, bladder, and femur heads. Plans were created for 6 MV photon beam using RapidArc technique in Eclipse treatment planning system. Dose calculations were performed with AAA and AXB for same number of monitor units and identical beam setup. Mean and maximum doses to planning target volume (PTV and OARs were analyzed. Additionally, minimum dose to PTV and V100 was analyzed. Finally, point-dose difference between planar dose distributions of AAA and AXB plans was investigated. Results: The highest dose difference was up to 0.43% (range: 0.05−0.43%, P> 0.05 for PTV and 1.98% (range: 0.22−1.98%, P> 0.05 for OARs with AAA predicting higher dose than AXB. The V100 values of AAA plans (95 % and AXB plans (range: 93.1−97.9 % had an average difference of 0.89±1.47% with no statistical significance (P = 0.25411. The point-dose difference analysis showed that AAA predicted higher dose than AXB at significantly higher percentage (in average 94.15 of total evaluated points. Conclusion: The dosimetric results of this study suggest that the AXB can perform the dose computation comparable to AAA in RapidArc prostate cancer treatment plans that are generated by a partial single-arc technique.
Calculation of Ambient (H*(10)) and Personal (Hp(10)) Dose Equivalent from a 252Cf Neutron Source
Traub, Richard J.
2010-03-26
The purpose of this calculation is to calculate the neutron dose factors for the Sr-Cf-3000 neutron source that is located in the 318 low scatter room (LSR). The dose factors were based on the dose conversion factors published in ICRP-21 Appendix 6, and the Ambient dose equivalent (H*(10)) and Personal dose equivalent (Hp(10)) dose factors published in ICRP Publication 74.
Famulari, Gabriel; Pater, Piotr; Enger, Shirin A.
2017-07-01
The aim of this study was to calculate microdosimetric distributions for low energy electrons simulated using the Monte Carlo track structure code Geant4-DNA. Tracks for monoenergetic electrons with kinetic energies ranging from 100 eV to 1 MeV were simulated in an infinite spherical water phantom using the Geant4-DNA extension included in Geant4 toolkit version 10.2 (patch 02). The microdosimetric distributions were obtained through random sampling of transfer points and overlaying scoring volumes within the associated volume of the tracks. Relative frequency distributions of energy deposition f(>E)/f(>0) and dose mean lineal energy (\\bar{y}D ) values were calculated in nanometer-sized spherical and cylindrical targets. The effects of scoring volume and scoring techniques were examined. The results were compared with published data generated using MOCA8B and KURBUC. Geant4-DNA produces a lower frequency of higher energy deposits than MOCA8B. The \\bar{y}D values calculated with Geant4-DNA are smaller than those calculated using MOCA8B and KURBUC. The differences are mainly due to the lower ionization and excitation cross sections of Geant4-DNA for low energy electrons. To a lesser extent, discrepancies can also be attributed to the implementation in this study of a new and fast scoring technique that differs from that used in previous studies. For the same mean chord length (\\bar{l} ), the \\bar{y}D calculated in cylindrical volumes are larger than those calculated in spherical volumes. The discrepancies due to cross sections and scoring geometries increase with decreasing scoring site dimensions. A new set of \\bar{y}D values has been presented for monoenergetic electrons using a fast track sampling algorithm and the most recent physics models implemented in Geant4-DNA. This dataset can be combined with primary electron spectra to predict the radiation quality of photon and electron beams.
Seyed Mohsen Hosseini Daghigh
2012-03-01
Full Text Available Introduction Interaluminal brachytherapy is one of the important methods of esophageal cancer treatment. The effect of applicator attenuation is not considered in dose calculation method released by AAPM-TG43. In this study, the effect of High-Dose Rate (HDR brachytherapy esophageal applicator on dose distribution was surveyed in HDR brachytherapy. Materials and Methods A cylindrical PMMA phantom was built in order to be inserted by various sizes of esophageal applicators. EDR2 films were placed at 33 mm from Ir-192 source and irradiated with 1.5 Gy after planning using treatment planning system for all applicators. Results The results of film dosimetry in reference point for 6, 8, 10, and 20 mm applicators were 1.54, 1.53, 1.48, and 1.50 Gy, respectively. The difference between practical and treatment planning system results was 0.023 Gy (
Impact of dose calculation accuracy during optimization on lung IMRT plan quality.
Li, Ying; Rodrigues, Anna; Li, Taoran; Yuan, Lulin; Yin, Fang-Fang; Wu, Q Jackie
2015-01-08
The purpose of this study was to evaluate the effect of dose calculation accuracy and the use of an intermediate dose calculation step during the optimization of intensity-modulated radiation therapy (IMRT) planning on the final plan quality for lung cancer patients. This study included replanning for 11 randomly selected free-breathing lung IMRT plans. The original plans were optimized using a fast pencil beam convolution algorithm. After optimization, the final dose calculation was performed using the analytical anisotropic algorithm (AAA). The Varian Treatment Planning System (TPS) Eclipse v11, includes an option to perform intermediate dose calculation during optimization using the AAA. The new plans were created using this intermediate dose calculation during optimization with the same planning objectives and dose constraints as in the original plan. Differences in dosimetric parameters for the planning target volume (PTV) dose coverage, organs-at-risk (OARs) dose sparing, and the number of monitor units (MU) between the original and new plans were analyzed. Statistical significance was determined with a p-value of less than 0.05. All plans were normalized to cover 95% of the PTV with the prescription dose. Compared with the original plans, the PTV in the new plans had on average a lower maximum dose (69.45 vs. 71.96Gy, p = 0.005), a better homogeneity index (HI) (0.08 vs. 0.12, p = 0.002), and a better conformity index (CI) (0.69 vs. 0.59, p = 0.003). In the new plans, lung sparing was increased as the volumes receiving 5, 10, and 30 Gy were reduced when compared to the original plans (40.39% vs. 42.73%, p = 0.005; 28.93% vs. 30.40%, p = 0.001; 14.11%vs. 14.84%, p = 0.031). The volume receiving 20 Gy was not significantly lower (19.60% vs. 20.38%, p = 0.052). Further, the mean dose to the lung was reduced in the new plans (11.55 vs. 12.12 Gy, p = 0.024). For the esophagus, the mean dose, the maximum dose, and the volumes receiving 20 and 60 Gy were lower in
SU-E-T-27: A Tool for Routine Quality Assurance of Radiotherapy Dose Calculation Software
Popple, R; Cardan, R; Duan, J; Wu, X; Shen, S; Brezovich, I [The University of Alabama at Birmingham, Birmingham, AL (United States)
2014-06-01
Purpose: Dose calculation software is thoroughly evaluated when it is commissioned; however, evaluation of periodic software updates is typically limited in scope due to staffing constraints and the need to quickly return the treatment planning system to clinical service. We developed a tool for quickly and comprehensively testing and documenting dose calculation software against measured data. Methods: A tool was developed using MatLab (The MathWorks, Natick, MA) for evaluation of dose calculation algorithms against measured data. Inputs to the tool are measured data, reference DICOM RT PLAN files describing the measurements, and dose calculations in DICOM format. The tool consists of a collection of extensible modules that can perform analysis of point dose, depth dose curves, and profiles using dose difference, distance-to-agreement, and the gamma-index. Each module generates a report subsection that is incorporated into a master template, which is converted to final form in portable document format (PDF). Results: After each change to the treatment planning system, a report can be generated in approximately 90 minutes. The tool has been in use for more than 5 years, spanning 5 versions of the eMC and 4 versions of the AAA. We have detected changes to the algorithms that affected clinical practice once during this period. Conclusion: Our tool provides an efficient method for quality assurance of dose calculation software, providing a complete set of tests for an update. Future work includes the addition of plan level tests, allowing incorporation of, for example, the TG-119 test suite for IMRT, and integration with the treatment planning system via an application programming interface. Integration with the planning system will permit fully-automated testing and reporting at scheduled intervals.
R-matrix calculations of electron impact electronic excitation of BeH
Darby-Lewis, Daniel; Mašín, Zdeněk; Tennyson, Jonathan
2017-09-01
The R-matrix method is used to perform high-level calculations of electron collisions with beryllium mono-hydride at its equilibrium geometry with a particular emphasis on electron impact electronic excitation. Several target and scattering models are considered. The calculations were performed using (1) the UKRMol suite which relies on the use of Gaussian type orbitals (GTOs) to represent the continuum and (2) using the new UKRMol+ suite which allows the inclusion of B-spline type orbitals in the basis for the continuum. The final close-coupling scattering models used the UKRMol+ code and a frozen core, valence full configuration interaction, method based on a diffuse GTO atomic basis set. The calculated electronic properties of the molecule are in very good agreement with state-of-the-art electronic structure calculations. The use of the UKRMol+ suite proved critical since it allowed the use of a large R-matrix sphere (35 Bohr), necessary to contain the diffuse electronic states of the molecule. The corresponding calculations using UKRMol are not possible due to numerical problems associated with the combination of GTO-only continuum and a large R-matrix sphere. This work provides the first demonstration of the utility and numerical stability of the new UKRMol+ code. The inelastic cross sections obtained here present a significant improvement over the results of earlier studies on BeH.
García-Garduño, O. A., E-mail: oagarciag@innn.edu.mx, E-mail: amanda.garcia.g@gmail.com [Laboratorio de Física Médica, Instituto Nacional de Neurología y Neurocirugía, Mexico City 14269, México and Centro de Investigación en Ciencia Aplicada y Tecnología Avanzada, Unidad Legaria, Instituto Politécnico Nacional, Legaria 694, México City 11500, México (Mexico); Rodríguez-Ponce, M. [Departamento de Biofísica, Instituto Nacional de Cancerología, Mexico City 14080, México (Mexico); Gamboa-deBuen, I. [Instituto de Ciencias Nucleares, Universidad Nacional Autónoma de México, Ciudad Universitaria, Mexico City 04510 (Mexico); Rodríguez-Villafuerte, M. [Instituto de Física, Universidad Nacional Autónoma de México, Ciudad Universitaria, Mexico City 04510 (Mexico); Galván de la Cruz, O. O. [Laboratorio de Física Médica, Instituto Nacional de Neurología y Neurocirugía, Mexico City 14269, México (Mexico); and others
2014-09-15
Purpose: To assess the impact of the detector used to commission small photon beams on the calculated dose distribution in stereotactic radiosurgery (SRS). Methods: In this study, six types of detectors were used to characterize small photon beams: three diodes [a silicon stereotactic field diode SFD, a silicon diode SRS, and a silicon diode E], an ionization chamber CC01, and two types of radiochromic film models EBT and EBT2. These detectors were used to characterize circular collimated beams that were generated by a Novalis linear accelerator. This study was conducted in two parts. First, the following dosimetric data, which are of particular interest in SRS, were compared for the different detectors: the total scatter factor (TSF), the tissue phantom ratios (TPRs), and the off-axis ratios (OARs). Second, the commissioned data sets were incorporated into the treatment planning system (TPS) to compare the calculated dose distributions and the dose volume histograms (DVHs) that were obtained using the different detectors. Results: The TSFs data measured by all of the detectors were in good agreement with each other within the respective statistical uncertainties: two exceptions, where the data were systematically below those obtained for the other detectors, were the CC01 results for all of the circular collimators and the EBT2 film results for circular collimators with diameters below 10.0 mm. The OAR results obtained for all of the detectors were in excellent agreement for all of the circular collimators. This observation was supported by the gamma-index test. The largest difference in the TPR data was found for the 4.0 mm circular collimator, followed by the 10.0 and 20.0 mm circular collimators. The results for the calculated dose distributions showed that all of the detectors passed the gamma-index test at 100% for the 3 mm/3% criteria. The aforementioned observation was true regardless of the size of the calculation grid for all of the circular collimators
Real-time feedback from iterative electronic structure calculations.
Vaucher, Alain C; Haag, Moritz P; Reiher, Markus
2016-04-05
Real-time feedback from iterative electronic structure calculations requires to mediate between the inherently unpredictable execution times of the iterative algorithm used and the necessity to provide data in fixed and short time intervals for real-time rendering. We introduce the concept of a mediator as a component able to deal with infrequent and unpredictable reference data to generate reliable feedback. In the context of real-time quantum chemistry, the mediator takes the form of a surrogate potential that has the same local shape as the first-principles potential and can be evaluated efficiently to deliver atomic forces as real-time feedback. The surrogate potential is updated continuously by electronic structure calculations and guarantees to provide a reliable response to the operator for any molecular structure. To demonstrate the application of iterative electronic structure methods in real-time reactivity exploration, we implement self-consistent semiempirical methods as the data source and apply the surrogate-potential mediator to deliver reliable real-time feedback.
Ab initio calculations of yttrium nitride: structural and electronic properties
Zerroug, S.; Ali Sahraoui, F. [Universite Ferhat Abbas, Laboratoire d' Optoelectronique et Composants, Departement de Physique, Setif (Algeria); Bouarissa, N. [King Khalid University, Department of Physics, Faculty of Science, P.O. Box 9004, Abha (Saudi Arabia)
2009-11-15
Using first principles total energy calculations within the full-potential linearized augmented plane wave method, we have studied the structural and electronic properties of yttrium nitride (YN) in the three phases, namely wurtzite, caesium chloride and rocksalt structures. The calculations are performed at zero and under hydrostatic pressure. In agreement with previous findings, it is found that the favored phase for YN is the rocksalt-like structure. We predict that at zero pressure YN in the rocksalt structure is a semiconductor with an indirect bandgap of 0.8 eV. A phase transition from a rocksalt to a caesium chloride structure is found to occur at {proportional_to}134 GPa. Besides, a transition from an indirect ({gamma}-X) bandgap semiconductor to a direct (X-X) one is predicted at pressure of {proportional_to}84 GPa. For the electron effective mass of rocksalt YN, these are the first results, to our knowledge. The information derived from the present study may be useful for the use of YN as an active layer in electronic devices such as diodes and transistors. (orig.)
Cardenas, Carlos E; Nitsch, Paige L; Kudchadker, Rajat J; Howell, Rebecca M; Kry, Stephen F
2016-07-08
Out-of-field doses from radiotherapy can cause harmful side effects or eventually lead to secondary cancers. Scattered doses outside the applicator field, neutron source strength values, and neutron dose equivalents have not been broadly investigated for high-energy electron beams. To better understand the extent of these exposures, we measured out-of-field dose characteristics of electron applicators for high-energy electron beams on two Varian 21iXs, a Varian TrueBeam, and an Elekta Versa HD operating at various energy levels. Out-of-field dose profiles and percent depth-dose curves were measured in a Wellhofer water phantom using a Farmer ion chamber. Neutron dose was assessed using a combination of moderator buckets and gold activation foils placed on the treatment couch at various locations in the patient plane on both the Varian 21iX and Elekta Versa HD linear accelerators. Our findings showed that out-of-field electron doses were highest for the highest electron energies. These doses typically decreased with increasing distance from the field edge but showed substantial increases over some distance ranges. The Elekta linear accelerator had higher electron out-of-field doses than the Varian units examined, and the Elekta dose profiles exhibited a second dose peak about 20 to 30 cm from central-axis, which was found to be higher than typical out-of-field doses from photon beams. Electron doses decreased sharply with depth before becoming nearly constant; the dose was found to decrease to a depth of approximately E(MeV)/4 in cm. With respect to neutron dosimetry, Q values and neutron dose equivalents increased with electron beam energy. Neutron contamination from electron beams was found to be much lower than that from photon beams. Even though the neutron dose equivalent for electron beams represented a small portion of neutron doses observed under photon beams, neutron doses from electron beams may need to be considered for special cases.
The denoising of Monte Carlo dose distributions using convolution superposition calculations.
El Naqa, I; Cui, J; Lindsay, P; Olivera, G; Deasy, J O
2007-09-07
Monte Carlo (MC) dose calculations can be accurate but are also computationally intensive. In contrast, convolution superposition (CS) offers faster and smoother results but by making approximations. We investigated MC denoising techniques, which use available convolution superposition results and new noise filtering methods to guide and accelerate MC calculations. Two main approaches were developed to combine CS information with MC denoising. In the first approach, the denoising result is iteratively updated by adding the denoised residual difference between the result and the MC image. Multi-scale methods were used (wavelets or contourlets) for denoising the residual. The iterations are initialized by the CS data. In the second approach, we used a frequency splitting technique by quadrature filtering to combine low frequency components derived from MC simulations with high frequency components derived from CS components. The rationale is to take the scattering tails as well as dose levels in the high-dose region from the MC calculations, which presumably more accurately incorporates scatter; high-frequency details are taken from CS calculations. 3D Butterworth filters were used to design the quadrature filters. The methods were demonstrated using anonymized clinical lung and head and neck cases. The MC dose distributions were calculated by the open-source dose planning method MC code with varying noise levels. Our results indicate that the frequency-splitting technique for incorporating CS-guided MC denoising is promising in terms of computational efficiency and noise reduction.
The denoising of Monte Carlo dose distributions using convolution superposition calculations
El Naqa, I [Department of Radiation Oncology, Washington University School of Medicine, St Louis, MO (United States); Cui, J [Department of Radiation Oncology, Washington University School of Medicine, St Louis, MO (United States); Lindsay, P [MD Anderson, Houston, TX (United States); Olivera, G [Tomotherapy Inc., Madison, WI (United States); Deasy, J O [Department of Radiation Oncology, Washington University School of Medicine, St Louis, MO (United States)
2007-09-07
Monte Carlo (MC) dose calculations can be accurate but are also computationally intensive. In contrast, convolution superposition (CS) offers faster and smoother results but by making approximations. We investigated MC denoising techniques, which use available convolution superposition results and new noise filtering methods to guide and accelerate MC calculations. Two main approaches were developed to combine CS information with MC denoising. In the first approach, the denoising result is iteratively updated by adding the denoised residual difference between the result and the MC image. Multi-scale methods were used (wavelets or contourlets) for denoising the residual. The iterations are initialized by the CS data. In the second approach, we used a frequency splitting technique by quadrature filtering to combine low frequency components derived from MC simulations with high frequency components derived from CS components. The rationale is to take the scattering tails as well as dose levels in the high-dose region from the MC calculations, which presumably more accurately incorporates scatter; high-frequency details are taken from CS calculations. 3D Butterworth filters were used to design the quadrature filters. The methods were demonstrated using anonymized clinical lung and head and neck cases. The MC dose distributions were calculated by the open-source dose planning method MC code with varying noise levels. Our results indicate that the frequency-splitting technique for incorporating CS-guided MC denoising is promising in terms of computational efficiency and noise reduction. (note)
NOTE: The denoising of Monte Carlo dose distributions using convolution superposition calculations
El Naqa, I.; Cui, J.; Lindsay, P.; Olivera, G.; Deasy, J. O.
2007-09-01
Monte Carlo (MC) dose calculations can be accurate but are also computationally intensive. In contrast, convolution superposition (CS) offers faster and smoother results but by making approximations. We investigated MC denoising techniques, which use available convolution superposition results and new noise filtering methods to guide and accelerate MC calculations. Two main approaches were developed to combine CS information with MC denoising. In the first approach, the denoising result is iteratively updated by adding the denoised residual difference between the result and the MC image. Multi-scale methods were used (wavelets or contourlets) for denoising the residual. The iterations are initialized by the CS data. In the second approach, we used a frequency splitting technique by quadrature filtering to combine low frequency components derived from MC simulations with high frequency components derived from CS components. The rationale is to take the scattering tails as well as dose levels in the high-dose region from the MC calculations, which presumably more accurately incorporates scatter; high-frequency details are taken from CS calculations. 3D Butterworth filters were used to design the quadrature filters. The methods were demonstrated using anonymized clinical lung and head and neck cases. The MC dose distributions were calculated by the open-source dose planning method MC code with varying noise levels. Our results indicate that the frequency-splitting technique for incorporating CS-guided MC denoising is promising in terms of computational efficiency and noise reduction.
Park, S; Lee, J K; Lee, C; Lee, C
2008-01-01
Absorbed fractions (AFs) and specific absorbed fractions (SAFs) for internally deposited electron were calculated using a Korean reference adult stylised phantom, where a total of 15 internal organ volumes and external body dimension were designed to match average Korean adult male. The walls of oesophagus, stomach, colon and urinary bladder were additionally divided into the mucosal layer and residual wall to accommodate dose calculation for weakly penetrating electron. The mucosal wall thicknesses were determined by the data reported in the International Commission on Radiological Protection Publication 89 and other literature resources and by direct measurements. The Monte Carlo transport code MCNPX (version 2.5.0) was employed to calculate the electron energy deposited. The SAFs and AFs for monoenergetic electrons with the energies ranging from 10 keV to 2 MeV were calculated. The results were compared with those of the revised Oak Ridge National Laboratory phantoms and showed considerable differences up to 150% in SAFs, whereas no substantial differences were observed in the AFs.
Takashi Kadowaki
Full Text Available BACKGROUND: Pragmatic methods for dose optimization are required for the successful basal management in daily clinical practice. To derive a useful formula for calculating recommended glargine doses, we analyzed data from the Add-on Lantus® to Oral Hypoglycemic Agents (ALOHA study, a 24-week observation of Japanese type 2 diabetes patients. METHODOLOGY/PRINCIPAL FINDINGS: The patients who initiated insulin glargine in basal-supported oral therapy (BOT regimen (n = 3506 were analyzed. The correlations between average changes in glargine dose and HbA1c were calculated, and its regression formula was estimated from grouped data categorized by baseline HbA1c levels. Starting doses of the background-subgroup achieving the HbA1c target with a last-observed dose above the average were compared to an assumed optimal starting dose of 0.15 U/kg/day. The difference in regression lines between background-subgroups was examined. A formula for determining the optimal starting and titration doses was thereby derived. The correlation coefficient between changes in dose and HbA1c was -0.9043. The estimated regression line formula was -0.964 × change in HbA1c+2.000. A starting dose of 0.15 U/kg/day was applicable to all background-subgroups except for patients with retinopathy (0.120 U/kg/day and/or with eGFR<60 mL/min/1.73 m(2 (0.114 U/kg/day. Additionally, women (0.135 U/kg/day and patients with sulfonylureas (0.132 U/kg/day received a slightly decreased starting dose. CONCLUSIONS/SIGNIFICANCE: We suggest a simplified and pragmatic dose calculation formula for type 2 diabetes patients starting glargine BOT optimal daily dose at 24 weeks = starting dose (0.15×weight + incremental dose (baseline HbA1c - target HbA1c+2. This formula should be further validated using other samples in a prospective follow-up, especially since several patient groups required lower starting doses.
Endo, Akira; Yamaguchi, Yasuhiro [Japan Atomic Energy Research Inst., Tokai, Ibaraki (Japan). Tokai Research Establishment; Tamura, Tsutomu
1999-07-01
Nuclear decay data used for dose calculations were compiled for 162 nuclides with half-lives greater than or equal to 10 min that are not listed in ICRP Publication 38 (Publ. 38) and their 28 daughter nuclides. Additional 14 nuclides that are considered to be important in fusion reactor facilities were also included. The data were compiled using decay data sets of the Evaluated Nuclear Structure Data File (ENSDF), the latest version in August 1997. Investigations of the data sets were performed to check their consistency by referring to recent literature and NUBASE, the database for nuclear and decay properties of nuclides, and by using the utility programs of ENSDF. Possible revisions of the data sets were made for their format and syntax errors, level schemes, normalization records, and so on. The revised data sets were processed by EDISTR in order to calculate the energies and intensities of {alpha} particles, {beta} particles, {gamma} rays including annihilation photons, internal conversion electrons, X rays, and Auger electrons emitted in nuclear transformations of the radionuclides. For spontaneously fissioning nuclides, the average energies and intensities of neutrons, fission fragments, prompt {gamma} rays, delayed {gamma} rays, and {beta} particles were also calculated. The compiled data were presented in two types of format; Publ. 38 and NUCDECAY formats. This report provides the decay data in the Publ. 38 format along with decay scheme drawings. The data will be widely used for internal and external dose calculations in radiation protection. (author)
SU-E-T-202: Impact of Monte Carlo Dose Calculation Algorithm On Prostate SBRT Treatments
Venencia, C; Garrigo, E; Cardenas, J; Castro Pena, P [Instituto de Radioterapia - Fundacion Marie Curie, Cordoba (Argentina)
2014-06-01
Purpose: The purpose of this work was to quantify the dosimetric impact of using Monte Carlo algorithm on pre calculated SBRT prostate treatment with pencil beam dose calculation algorithm. Methods: A 6MV photon beam produced by a Novalis TX (BrainLAB-Varian) linear accelerator equipped with HDMLC was used. Treatment plans were done using 9 fields with Iplanv4.5 (BrainLAB) and dynamic IMRT modality. Institutional SBRT protocol uses a total dose to the prostate of 40Gy in 5 fractions, every other day. Dose calculation is done by pencil beam (2mm dose resolution), heterogeneity correction and dose volume constraint (UCLA) for PTV D95%=40Gy and D98%>39.2Gy, Rectum V20Gy<50%, V32Gy<20%, V36Gy<10% and V40Gy<5%, Bladder V20Gy<40% and V40Gy<10%, femoral heads V16Gy<5%, penile bulb V25Gy<3cc, urethra and overlap region between PTV and PRV Rectum Dmax<42Gy. 10 SBRT treatments plans were selected and recalculated using Monte Carlo with 2mm spatial resolution and mean variance of 2%. DVH comparisons between plans were done. Results: The average difference between PTV doses constraints were within 2%. However 3 plans have differences higher than 3% which does not meet the D98% criteria (>39.2Gy) and should have been renormalized. Dose volume constraint differences for rectum, bladder, femoral heads and penile bulb were les than 2% and within tolerances. Urethra region and overlapping between PTV and PRV Rectum shows increment of dose in all plans. The average difference for urethra region was 2.1% with a maximum of 7.8% and for the overlapping region 2.5% with a maximum of 8.7%. Conclusion: Monte Carlo dose calculation on dynamic IMRT treatments could affects on plan normalization. Dose increment in critical region of urethra and PTV overlapping region with PTV could have clinical consequences which need to be studied. The use of Monte Carlo dose calculation algorithm is limited because inverse planning dose optimization use only pencil beam.
Calculation of depth-dose distribution of intermediate energy heavy-ion beams
无
2002-01-01
Based on the characteristics of the interactions between intermediate energy heavy-ion beam and target matter, a method to calculate the depth-dose distribution of heavy-ion beams with intermediate energy (10 -100 MeV/u) is presented. By comparing high energy beams where projectile fragmentation is overwhelm ing with lowenergies where energy straggling is the sole factor instead, a crescent energy spread with increasing depth and a simple fragmentation assumption were included for the depth-dose calculation of the intermediate energy beam. Rel ative depth-dose curves of carbon and oxygen ion beams with intermediate energie s were computed according to the method here. Comparisons between the calculated relative doses and measurements are shown. The calculated Bragg curves, especially the upstream and downstream Bragg peaks, agree with the measured data. Differences between the two results appear only around the peak regions because of th e limitations of the calculation and experimental conditions, but the calculated curves generally reproduce the measured data within the experimental errors. Th e reasons for the divergences were analyzed carefully and the magnitudes of the deviations are given.
Electronic Structure of Silicon Nanowires Matrix from Ab Initio Calculations.
Monastyrskii, Liubomyr S; Boyko, Yaroslav V; Sokolovskii, Bogdan S; Potashnyk, Vasylyna Ya
2016-12-01
An investigation of the model of porous silicon in the form of periodic set of silicon nanowires has been carried out. The electronic energy structure was studied using a first-principle band method-the method of pseudopotentials (ultrasoft potentials in the basis of plane waves) and linearized mode of the method of combined pseudopotentials. Due to the use of hybrid exchange-correlation potentials (B3LYP), the quantitative agreement of the calculated value of band gap in the bulk material with experimental data is achieved. The obtained results show that passivation of dangling bonds with hydrogen atoms leads to substantial transformation of electronic energy structure. At complete passivation of the dangling silicon bonds by hydrogen atoms, the band gap value takes the magnitude which substantially exceeds that for bulk silicon. The incomplete passivation gives rise to opposite effect when the band gap value decreases down the semimetallic range.
Correlated Electron Calculations with Hartree-Fock Scaling
Gebauer, Ralph; Car, Roberto
2013-01-01
We introduce an energy functional for ground-state electronic structure calculations with fundamental variables the natural spin orbitals and their joint occupation probabilities in an implied many-body trial wave function. We use a controlled approximation for the two-particle density matrix that greatly extends the accuracy compared to current functionals of the one-particle density matrix only. Algebraic scaling of computational cost with electron number is achieved in general, and Hartree-Fock scaling in the seniority-zero version of the theory. We present results obtained with the latter version for saturated small molecular systems for which highly accurate quantum chemical computations are available for comparison. The results are variational, capturing most of the correlation energy from equilibrium to dissociation.
Relativistic collision rate calculations for electron-air interactions
Graham, G. [EG and G Energy Measurements, Inc., Los Alamos, NM (United States); Roussel-Dupre, R. [Los Alamos National Lab., NM (United States). Space Science and Technologies
1992-12-16
The most recent data available on differential cross sections for electron-air interactions are used to calculate the avalanche, momentum transfer, and energy loss rates that enter into the fluid equations. Data for the important elastic, inelastic, and ionizing processes are generally available out to electron energies of 1--10 kev. Prescriptions for extending these cross sections to the relativistic regime are presented. The angular dependence of the cross sections is included where data is available as is the doubly differential cross section for ionizing collisions. The collision rates are computed by taking moments of the Boltzmann collision integrals with the assumption that the electron momentum distribution function is given by the Juettner distribution function which satisfies the relativistic H- theorem and which reduces to the familiar Maxwellian velocity distribution in the nonrelativistic regime. The distribution function is parameterized in terms of the electron density, mean momentum, and thermal energy and the rates are therefore computed on a two-dimensional grid as a function of mean kinetic energy and thermal energy.
Using electron microscopy to calculate optical properties of biological samples.
Wu, Wenli; Radosevich, Andrew J; Eshein, Adam; Nguyen, The-Quyen; Yi, Ji; Cherkezyan, Lusik; Roy, Hemant K; Szleifer, Igal; Backman, Vadim
2016-11-01
The microscopic structural origins of optical properties in biological media are still not fully understood. Better understanding these origins can serve to improve the utility of existing techniques and facilitate the discovery of other novel techniques. We propose a novel analysis technique using electron microscopy (EM) to calculate optical properties of specific biological structures. This method is demonstrated with images of human epithelial colon cell nuclei. The spectrum of anisotropy factor g, the phase function and the shape factor D of the nuclei are calculated. The results show strong agreement with an independent study. This method provides a new way to extract the true phase function of biological samples and provides an independent validation for optical property measurement techniques.
An Emphasis of Electron Energy Calculation in Quantum Wells
GAOShao-Wen; CAOJun-Cheng; FENGSong-Lin
2004-01-01
We investigate various methods for the calculation of the electron energy in semiconductor quantum wells and focus on a matrix algorithm method. The results show better fitness of the factor -h2/2 э/эz 1/m*（z） э/эz than that of -h2/2 1/m*（z） э2/эz2 in the first part of the Schroedinger equation. The effect of nonparabolicity in the conduction band is also discussed.
Panthere V2: Multipurpose Simulation Software for 3D Dose Rate Calculations
Penessot, Gaël; Bavoil, Éléonore; Wertz, Laurent; Malouch, Fadhel; Visonneau, Thierry; Dubost, Julien
2017-09-01
PANTHERE is a multipurpose radiation protection software developed by EDF to calculate gamma dose rates in complex 3D environments. PANTHERE takes a key role in the EDF ALARA process, enabling to predict dose rates and to organize and optimize operations in high radiation environments. PANTHERE is also used for nuclear waste characterization, transport of nuclear materials, etc. It is used in most of the EDF engineering units and their design service providers and industrial partners.
Impact of temporal probability in 4D dose calculation for lung tumors.
Rouabhi, Ouided; Ma, Mingyu; Bayouth, John; Xia, Junyi
2015-11-08
The purpose of this study was to evaluate the dosimetric uncertainty in 4D dose calculation using three temporal probability distributions: uniform distribution, sinusoidal distribution, and patient-specific distribution derived from the patient respiratory trace. Temporal probability, defined as the fraction of time a patient spends in each respiratory amplitude, was evaluated in nine lung cancer patients. Four-dimensional computed tomography (4D CT), along with deformable image registration, was used to compute 4D dose incorporating the patient's respiratory motion. First, the dose of each of 10 phase CTs was computed using the same planning parameters as those used in 3D treatment planning based on the breath-hold CT. Next, deformable image registration was used to deform the dose of each phase CT to the breath-hold CT using the deformation map between the phase CT and the breath-hold CT. Finally, the 4D dose was computed by summing the deformed phase doses using their corresponding temporal probabilities. In this study, 4D dose calculated from the patient-specific temporal probability distribution was used as the ground truth. The dosimetric evaluation matrix included: 1) 3D gamma analysis, 2) mean tumor dose (MTD), 3) mean lung dose (MLD), and 4) lung V20. For seven out of nine patients, both uniform and sinusoidal temporal probability dose distributions were found to have an average gamma passing rate > 95% for both the lung and PTV regions. Compared with 4D dose calculated using the patient respiratory trace, doses using uniform and sinusoidal distribution showed a percentage difference on average of -0.1% ± 0.6% and -0.2% ± 0.4% in MTD, -0.2% ± 1.9% and -0.2% ± 1.3% in MLD, 0.09% ± 2.8% and -0.07% ± 1.8% in lung V20, -0.1% ± 2.0% and 0.08% ± 1.34% in lung V10, 0.47% ± 1.8% and 0.19% ± 1.3% in lung V5, respectively. We concluded that four-dimensional dose computed using either a uniform or sinusoidal temporal probability distribution can
Calculation of absorbed doses to water pools in severe accident sequences
Weber, C.F. [Oak Ridge National Lab., TN (United States)
1991-12-01
A methodology is presented for calculating the radiation dose to a water pool from the decay of uniformly distributed nuclides in that pool. Motivated by the need to accurately model radiolysis reactions of iodine, direct application is made to fission product sources dissolved or suspended in containment sumps or pools during a severe nuclear reactor accident. Two methods of calculating gamma absorption are discussed - one based on point-kernal integration and the other based on Monte Carlo techniques. Using least-squares minimization, the computed results are used to obtain a correlation that relates absorbed dose to source energy and surface-to-volume ratio of the pool. This correlation is applied to most relevant fission product nuclides and used to actually calculate transient sump dose rate in a pressurized-water reactor (PWR) severe accident sequence.
Ramounet-le Gall, B.; Fritsch, P.; Abram, M.C.; Rateau, G.; Grillon, G.; Guillet, K. [Lab. de Radiotoxicologie, CEA/DSV/DRR/SRCA, Bruyeres le Chatel (France); Baude, S. [Lab. de Mesures Specifiques Gaz, CEA/DAM/DASE/SRCE, Bruyeres le Chatel (France); Berard, P. [Cabinet du Conseiller medical du CEA, CEA/DEN/DPS/LABM Saclay, Gif sur Yvette (France); Ansoborlo, E. [CEA/DEN/DRCP/CETAMA, Bagnols sur Ceze (France); Delforge, J. [Lab. de Radiotoxicologie, CEA/DSV/DRR/SRCA, Bruyeres le Chatel (France)
2002-07-01
A review on specific parameter measurements to calculate doses per unit of incorporation according to recommendations of the International Commission of Radiological Protection has been performed for inhaled actinide oxides. Alpha activity distribution of the particles can be obtained by autoradiography analysis using aerosol sampling filters at the work places. This allows us to characterize granulometric parameters of 'pure' actinide oxides, but complementary analysis by scanning electron microscopy is needed for complex aerosols. Dissolution parameters with their standard deviation are obtained after rat inhalation exposure, taking into account both mechanical lung clearance and actinide transfer to the blood estimated from bone retention. In vitro experiments suggest that the slow dissolution rate might decrease as a function of time following exposure. Dose calculation software packages have been developed to take into account granulometry and dissolution parameters as well as specific physiological parameters of exposed individuals. In the case of poorly soluble actinide oxides, granulometry and physiology appear as the main parameters controlling dose value, whereas dissolution only alters dose distribution. Validation of these software packages are in progress. (author)
Zhang, Y; Zhang, J; Hu, Q; Tie, J; Wu, H [Key Laboratory of Carcinogenesis and Translational Research (Ministry of Education), Department of Radiotherapy, Peking University Cancer Hospital ' Institute, Beijing (China); Deng, J [Department of Therapeutic Radiology, Yale University, New Haven, CT (United States)
2014-06-01
Purpose: To investigate the possibility of applying optimized scanning protocols for pediatric CT simulation by quantifying the dosimetric inaccuracy introduced by using a fixed HU to density conversion. Methods: The images of a CIRS electron density reference phantom (Model 062) were acquired by a Siemens CT simulator (Sensation Open) using the following settings of tube voltage and beam current: 120 kV/190mA (the reference protocol used to calibrate CT for our treatment planning system (TPS)); Fixed 190mA combined with all available kV: 80, 100, and 140; fixed 120 kV and various current from 37 to 444 mA (scanner extremes) with interval of 30 mA. To avoid the HU uncertainty of point sampling in the various inserts of known electron densities, the mean CT numbers of the central cylindrical volume were calculated using DICOMan software. The doses per 100 MU to the reference point (SAD=100cm, Depth=10cm, Field=10X10cm, 6MV photon beam) in a virtual cubic phantom (30X30X30cm) were calculated using Eclipse TPS (calculation model: AcurosXB-11031) by assigning the CT numbers to HU of typical materials acquired by various protocols. Results: For the inserts of densities less than muscle, CT number fluctuations of all protocols were within the tolerance of 10 HU as accepted by AAPM-TG66. For more condensed materials, fixed kV yielded stable HU with any mA combination where largest disparities were found in 1750mg/cc insert: HU{sub reference}=1801(106.6cGy), HU{sub minimum}=1799 (106.6cGy, error{sub dose}=0.00%), HU{sub maximum}=1815 (106.8cGy, error{sub dose}=0.19%). Yet greater disagreements were observed with increasing density when kV was modified: HU{sub minimum}=1646 (104.5cGy, error{sub dose}=- 1.97%), HU{sub maximum}=2487 (116.4cGy, error{sub dose}=9.19%) in 1750mg/cc insert. Conclusion: Without affecting treatment dose calculation, personalized mA optimization of CT simulator can be conducted by fixing kV for a better cost-effectiveness of imaging dose and quality
Polednik Martin
2006-11-01
Full Text Available Abstract Background The aim of this study was to compare and to validate different dose calculation algorithms for the use in radiation therapy of small lung lesions and to optimize the treatment planning using accurate dose calculation algorithms. Methods A 9-field conformal treatment plan was generated on an inhomogeneous phantom with lung mimics and a soft tissue equivalent insert, mimicking a lung tumor. The dose distribution was calculated with the Pencil Beam and Collapsed Cone algorithms implemented in Masterplan (Nucletron and the Monte Carlo system XVMC and validated using Gafchromic EBT films. Differences in dose distribution were evaluated. The plans were then optimized by adding segments to the outer shell of the target in order to increase the dose near the interface to the lung. Results The Pencil Beam algorithm overestimated the dose by up to 15% compared to the measurements. Collapsed Cone and Monte Carlo predicted the dose more accurately with a maximum difference of -8% and -3% respectively compared to the film. Plan optimization by adding small segments to the peripheral parts of the target, creating a 2-step fluence modulation, allowed to increase target coverage and homogeneity as compared to the uncorrected 9 field plan. Conclusion The use of forward 2-step fluence modulation in radiotherapy of small lung lesions allows the improvement of tumor coverage and dose homogeneity as compared to non-modulated treatment plans and may thus help to increase the local tumor control probability. While the Collapsed Cone algorithm is closer to measurements than the Pencil Beam algorithm, both algorithms are limited at tissue/lung interfaces, leaving Monte-Carlo the most accurate algorithm for dose prediction.
Dose calculation method with 60-cobalt gamma rays in total body irradiation
Scaff, L A M
2001-01-01
Physical factors associated to total body irradiation using sup 6 sup 0 Co gamma rays beams, were studied in order to develop a calculation method of the dose distribution that could be reproduced in any radiotherapy center with good precision. The method is based on considering total body irradiation as a large and irregular field with heterogeneities. To calculate doses, or doses rates, of each area of interest (head, thorax, thigh, etc.), scattered radiation is determined. It was observed that if dismagnified fields were considered to calculate the scattered radiation, the resulting values could be applied on a projection to the real size to obtain the values for dose rate calculations. In a parallel work it was determined the variation of the dose rate in the air, for the distance of treatment, and for points out of the central axis. This confirm that the use of the inverse square law is not valid. An attenuation curve for a broad beam was also determined in order to allow the use of absorbers. In this wo...
Improved patient size estimates for accurate dose calculations in abdomen computed tomography
Lee, Chang-Lae
2017-07-01
The radiation dose of CT (computed tomography) is generally represented by the CTDI (CT dose index). CTDI, however, does not accurately predict the actual patient doses for different human body sizes because it relies on a cylinder-shaped head (diameter : 16 cm) and body (diameter : 32 cm) phantom. The purpose of this study was to eliminate the drawbacks of the conventional CTDI and to provide more accurate radiation dose information. Projection radiographs were obtained from water cylinder phantoms of various sizes, and the sizes of the water cylinder phantoms were calculated and verified using attenuation profiles. The effective diameter was also calculated using the attenuation of the abdominal projection radiographs of 10 patients. When the results of the attenuation-based method and the geometry-based method shown were compared with the results of the reconstructed-axial-CT-image-based method, the effective diameter of the attenuation-based method was found to be similar to the effective diameter of the reconstructed-axial-CT-image-based method, with a difference of less than 3.8%, but the geometry-based method showed a difference of less than 11.4%. This paper proposes a new method of accurately computing the radiation dose of CT based on the patient sizes. This method computes and provides the exact patient dose before the CT scan, and can therefore be effectively used for imaging and dose control.
Noblet, C.; Chiavassa, S.; Smekens, F.; Sarrut, D.; Passal, V.; Suhard, J.; Lisbona, A.; Paris, F.; Delpon, G.
2016-05-01
In preclinical studies, the absorbed dose calculation accuracy in small animals is fundamental to reliably investigate and understand observed biological effects. This work investigated the use of the split exponential track length estimator (seTLE), a new kerma based Monte Carlo dose calculation method for preclinical radiotherapy using a small animal precision micro irradiator, the X-RAD 225Cx. Monte Carlo modelling of the irradiator with GATE/GEANT4 was extensively evaluated by comparing measurements and simulations for half-value layer, percent depth dose, off-axis profiles and output factors in water and water-equivalent material for seven circular fields, from 20 mm down to 1 mm in diameter. Simulated and measured dose distributions in cylinders of water obtained for a 360° arc were also compared using dose, distance-to-agreement and gamma-index maps. Simulations and measurements agreed within 3% for all static beam configurations, with uncertainties estimated to 1% for the simulation and 3% for the measurements. Distance-to-agreement accuracy was better to 0.14 mm. For the arc irradiations, gamma-index maps of 2D dose distributions showed that the success rate was higher than 98%, except for the 0.1 cm collimator (92%). Using the seTLE method, MC simulations compute 3D dose distributions within minutes for realistic beam configurations with a clinically acceptable accuracy for beam diameter as small as 1 mm.
Applying graphics processor units to Monte Carlo dose calculation in radiation therapy
Bakhtiari M
2010-01-01
Full Text Available We investigate the potential in using of using a graphics processor unit (GPU for Monte-Carlo (MC-based radiation dose calculations. The percent depth dose (PDD of photons in a medium with known absorption and scattering coefficients is computed using a MC simulation running on both a standard CPU and a GPU. We demonstrate that the GPU′s capability for massive parallel processing provides a significant acceleration in the MC calculation, and offers a significant advantage for distributed stochastic simulations on a single computer. Harnessing this potential of GPUs will help in the early adoption of MC for routine planning in a clinical environment.
Oliveira, C
2001-01-01
A systematic study of isodose distributions and dose uniformity in sample carriers of the Portuguese Gamma Irradiation Facility was carried out using the MCNP code. The absorbed dose rate, gamma flux per energy interval and average gamma energy were calculated. For comparison purposes, boxes filled with air and 'dummy' boxes loaded with layers of folded and crumpled newspapers to achieve a given value of density were used. The magnitude of various contributions to the total photon spectra, including source-dependent factors, irradiator structures, sample material and other origins were also calculated.
[Calculation of the first dose of amikacine: evaluation of the current dosage recommendations].
Jean-Bart, E; Debeurme, G; Ducher, M; Bourguignon, L
2013-01-01
Aminoglycosides, including amikacin, are antibiotics with major interest in the management of sepsis, but with a high potential toxicity. The French national recommendations revised in 2011 recommend a dose of amikacin ranging from 15 to 30 mg/kg. The objective was to assess if such a dose interval allows reaching the efficiency target concentrations of 64 mg/L without exceeding the toxic threshold of 2.5mg/L. From a cohort of 100 patients treated with amikacin, the individual pharmacokinetic parameters were estimated using pharmacokinetic software (MM-USCPACK). Peak and residual concentrations obtained after simulated doses ranging from 15 to 30 mg/kg were estimated and compared with the effective and toxic thresholds. The optimum dose to achieve precisely the efficiency target was calculated for each patient. Patients studied had a mean age of 79 years, mean weight of 58 kg, and mean creatinine clearance of 45 mL/min. The dose of 30 mg/kg allows the achievement of an effective peak in 98.7% of patients, but led to a potentially toxic through for 72.4% of them. The optimal dose was at mean of 1264 mg, significantly different than doses calculated with weight (P<0.0001). A weak correlation was found between weight and the optimal dose. A fixed dose of 30 mg/kg seems to be effective for most patients, but often excessive and leads to a toxic residual to 72% of patients, whereas 15 mg/kg was insufficient for most patients. The low correlation between optimal dose and patient weight shows that weight does not explain fully the interindividual variability.
Range and modulation dependencies for proton beam dose per monitor unit calculations
Hsi, Wen C.; Schreuder, Andries N.; Moyers, Michael F.; Allgower, Chris E.; Farr, Jonathan B.; Mascia, Anthony E.
2009-01-01
Calculations of dose per monitor unit (D∕MU) are required in addition to measurements to increase patient safety in the clinical practice of proton radiotherapy. As in conventional photon and electron therapy, the D∕MU depends on several factors. This study focused on obtaining range and modulation dependence factors used in D∕MU calculations for the double scattered proton beam line at the Midwest Proton Radiotherapy Institute. Three dependencies on range and one dependency on modulation were found. A carefully selected set of measurements was performed to discern these individual dependencies. Dependencies on range were due to: (1) the stopping power of the protons passing through the monitor chamber; (2) the reduction of proton fluence due to nuclear interactions within the patient; and (3) the variation of proton fluence passing through the monitor chamber due to different source-to-axis distances (SADs) for different beam ranges. Different SADs are produced by reconfigurations of beamline elements to provide different field sizes and ranges. The SAD effect on the D∕MU varies smoothly as the beam range is varied, except at the beam range for which the first scatterers are exchanged and relocated to accommodate low and high beam ranges. A geometry factor was devised to model the SAD variation effect on the D∕MU. The measured D∕MU variation as a function of range can be predicted within 1% using the three modeled dependencies on range. Investigation of modulated beams showed that an analytical formula can predict the D∕MU dependency as a function of modulation to within 1.5%. Special attention must be applied when measuring the D∕MU dependence on modulation to avoid interplay between range and SAD effects. PMID:19292004
Range and modulation dependencies for proton beam dose per monitor unit calculations
Hsi, Wen C.; Schreuder, Andries N.; Moyers, Michael F.; Allgower, Chris E.; Farr, Jonathan B.; Mascia, Anthony E. [Midwest Proton Radiotherapy Institute, Bloomington, Indiana 47408 and University Florida Proton Therapy Institute, Jacksonville, Florida 32206 (United States); ProCure Treatment Centers, Inc., Bloomington, Indiana 47404 (United States); Proton Therapy, Inc., Colton, California 92324 (United States); Midwest Proton Radiotherapy Institute, Bloomington, Indiana 47408 (United States); Midwest Proton Radiotherapy Institute, Bloomington, Indiana 47408 and Westdeutsches Protonentherapiezentrum, Universitaetsklinikum, Hufelandstrasse 55, 45147 Essen (Germany); Midwest Proton Radiotherapy Institute, Bloomington, Indiana 47408 (United States)
2009-02-15
Calculations of dose per monitor unit (D/MU) are required in addition to measurements to increase patient safety in the clinical practice of proton radiotherapy. As in conventional photon and electron therapy, the D/MU depends on several factors. This study focused on obtaining range and modulation dependence factors used in D/MU calculations for the double scattered proton beam line at the Midwest Proton Radiotherapy Institute. Three dependencies on range and one dependency on modulation were found. A carefully selected set of measurements was performed to discern these individual dependencies. Dependencies on range were due to: (1) the stopping power of the protons passing through the monitor chamber; (2) the reduction of proton fluence due to nuclear interactions within the patient; and (3) the variation of proton fluence passing through the monitor chamber due to different source-to-axis distances (SADs) for different beam ranges. Different SADs are produced by reconfigurations of beamline elements to provide different field sizes and ranges. The SAD effect on the D/MU varies smoothly as the beam range is varied, except at the beam range for which the first scatterers are exchanged and relocated to accommodate low and high beam ranges. A geometry factor was devised to model the SAD variation effect on the D/MU. The measured D/MU variation as a function of range can be predicted within 1% using the three modeled dependencies on range. Investigation of modulated beams showed that an analytical formula can predict the D/MU dependency as a function of modulation to within 1.5%. Special attention must be applied when measuring the D/MU dependence on modulation to avoid interplay between range and SAD effects.
Photodissociation of CCH: Classical trajectory calculations involving seven electronic states
Apaydın, Gökşin; Fink, William H.; Jackson, William M.
2004-11-01
The photodissociation dynamics of ethynyl radical, C2H, involving seven electronic states is studied by classical trajectory calculations. Initial values of the trajectories are selected based on relative absorption intensities calculated by Mebel et al. The energies and the derivatives are interpolated by three-dimensional cubic spline interpolator using an extended data pool. Mean square errors and standard deviations in interpolation of energies for 450 data points are found to be in the range 3.1×10-6-1.4×10-5 and 1.7×10-3-3.8×10-3 hartrees, respectively. The photofragments of C2 and H are produced mainly in the X 1Σg+, a 3Πu, b 3Σg-, c 3Σu+, A 1Πu, B 1Δg electronic states of C2 as product. The avoided crossings do not appear to be in the main dissociation pathways. The internal distributions are in good accord with the experimental results where comparison is possible, suggesting that the fragmentation mechanism of C2H2 into C2 and H is a two step process involving C2H radical as an intermediate with a life time long enough to allow complete collection of the phase space in the experiments.
DFT calculation of core-electron binding energies
Takahata, Yuji; Chong, Delano P. E-mail: chong@chem.ubc.ca
2003-11-01
A total of 59 core-electron binding energies (CEBEs) were studied with the Amsterdam Density Functional Program (ADF) program and compared with the observed values. The results indicate that a polarized triple-zeta basis set of Slater-type orbitals is adequate for routine assessment of the performance of each method of computation. With such a basis set, seven density functionals were tested. In addition, the performance of 21 energy density functionals were computed from the density calculated with the statistical average of orbital potentials (SAOP). Among all the choices tested, the best density functional for core-electron binding energies of C to F turns out to be the combination of Perdew-Wang (1986) functional for exchange and the Perdew-Wang (1991) functional for correlation, confirming earlier studies based on contracted Gaussian-type orbitals. For this best functional, five Slater-type orbital basis sets were examined, ranging from polarized double-zeta quality to the largest set available in the ADF package. For the best functional with the best basis set, the average absolute deviation (AAD) of the calculated value from experiment is only 0.16 eV.
Efficient Execution of Electronic Structure Calculations on SMP Clusters
Ustemirov, Nurzhan [Iowa State Univ., Ames, IA (United States)
2006-01-01
Applications augmented with adaptive capabilities are becoming common in parallel computing environments. For large-scale scientific applications, dynamic adjustments to a computationally-intensive part may lead to a large pay-off in facilitating efficient execution of the entire application while aiming at avoiding resource contention. Application-specific knowledge, often best revealed during the run-time, is required to initiate and time these adjustments. In particular, General Atomic and Molecular Electronic Structure System (GAMESS) is a program for ab initio quantum chemistry that places significant demands on the high-performance computing platforms. Certain electronic structure calculations are characterized by high consumption of a particular resource, such as CPU, main memory, or disk I/O. This may lead to resource contention among concurrent GAMESS jobs and other programs in the dynamically changing environment. Thus, it is desirable to improve GAMESS calculations by means of dynamic adaptations. In this thesis, we show how an application- or algorithm-specific knowledge may play a significant role in achieving this goal. The choice of implementation is facilitated by a module-driven middleware easily integrated with GAMESS that assesses resource consumption and invokes GAMESS adaptations to the system environment. We show that the throughput of GAMESS jobs may be improved greatly as a result of such adaptations.
An electronic application for rapidly calculating Charlson comorbidity score
Jani Ashesh B
2004-12-01
Full Text Available Abstract Background Uncertainty regarding comorbid illness, and ability to tolerate aggressive therapy has led to minimal enrollment of elderly cancer patients into clinical trials and often substandard treatment. Increasingly, comorbid illness scales have proven useful in identifying subgroups of elderly patients who are more likely to tolerate and benefit from aggressive therapy. Unfortunately, the use of such scales has yet to be widely integrated into either clinical practice or clinical trials research. Methods This article reviews evidence for the validity of the Charlson Comorbidity Index (CCI in oncology and provides a Microsoft Excel (MS Excel Macro for the rapid and accurate calculation of CCI score. The interaction of comorbidity and malignant disease and the validation of the Charlson Index in oncology are discussed. Results The CCI score is based on one year mortality data from internal medicine patients admitted to an inpatient setting and is the most widely used comorbidity index in oncology. An MS Excel Macro file was constructed for calculating the CCI score using Microsoft Visual Basic. The Macro is provided for download and dissemination. The CCI has been widely used and validated throughout the oncology literature and has demonstrated utility for most major cancers. The MS Excel CCI Macro provides a rapid method for calculating CCI score with or without age adjustments. The calculator removes difficulty in score calculation as a limitation for integration of the CCI into clinical research. The simple nature of the MS Excel CCI Macro and the CCI itself makes it ideal for integration into emerging electronic medical records systems. Conclusions The increasing elderly population and concurrent increase in oncologic disease has made understanding the interaction between age and comorbid illness on life expectancy increasingly important. The MS Excel CCI Macro provides a means of increasing the use of the CCI scale in clinical
Paripas, B. [Department of Physics, University of Miskolc, Miskolc-Egyetemvaros (Hungary)]. E-mail: fizpari@gold.uni-miskolc.hu; Vitez, G. [Department of Physics, University of Miskolc, Miskolc-Egyetemvaros (Hungary); Vikor, Gy. [Department of Atomic Physics, Stockholm University, Stockholm (Sweden); Tokesi, K. [Institute of Nuclear Research of the Hungarian Academy of Sciences (ATOMKI), Debrecen (Hungary); Institute of Theoretical Physics, Vienna University of Technology, Vienna (Austria); Gulyas, L. [Institute of Nuclear Research of the Hungarian Academy of Sciences (ATOMKI), Debrecen (Hungary)
2001-08-28
The distortion effects of the post-collision interaction on argon L{sub 2,3}M{sub 2,3}M{sub 2,3} Auger-electron lineshapes have been calculated for 300 eV, 500 eV and 2 keV electron impact inner-shell ionization. The calculations were based on the model of Kuchiev and Sheinerman (Kuchiev M.Yu. and Sheinerman S.A. 1987 Sov. Phys.-Tech. Phys. 32 879) applying our cross section data for secondary electrons, determined by a classical trajectory Monte Carlo method and in addition a continuum distorted wave method for 2 keV. The asymmetry parameters of the calculated peak shapes have been examined as a function of the projectile energy and the Auger emission angle and for some cases compared with our previous experimental results. (author)
Monte Carlo Calculations of Dose to Medium and Dose to Water for Carbon Ion Beams in Various Media
Herrmann, Rochus; Petersen, Jørgen B.B.; Jäkel, Oliver
. The dose to medium (Dm ) may however differ from Dw , due to the different particle spectrum and stopping power found herein. Monte Carlo particle transport codes are capable of directly calculating dose to medium (Dm ), and was for instance recently investigated by Paganetti 2009 for various proton...... treatment plans. Here, we quantisize the effect of dose to water vs. dose to medium for a series of typical target materials found in medical physics. 2 Material and Methods The Monte Carlo code FLUKA [Battistioni et al. 2007] is used to simulate the particle fluence spectrum in a series of target...... the PSTAR, ASTAR stopping power routines available at NIST1 and MSTAR2 provided by H. Paul et al. 3 Results For a pristine carbon ion beam we encountered a maximum deviation between Dw and Dm up to 8% for bone. In addition we investigate spread out Bragg peak configurations which dilutes the effect...
Badkul, R; Nejaiman, S; Pokhrel, D; Jiang, H; Kumar, P [University of Kansas Medical Center, Kansas City, KS (United States)
2015-06-15
Purpose: Skin dose can be the limiting factor and fairly common reason to interrupt the treatment, especially for treating head-and-neck with Intensity-modulated-radiation-therapy(IMRT) or Volumetrically-modulated - arc-therapy (VMAT) and breast with tangentially-directed-beams. Aim of this study was to investigate accuracy of near-surface dose predicted by Eclipse treatment-planning-system (TPS) using Anisotropic-Analytic Algorithm (AAA)with varying calculation grid-size and comparing with metal-oxide-semiconductor-field-effect-transistors(MOSFETs)measurements for a range of clinical-conditions (open-field,dynamic-wedge, physical-wedge, IMRT,VMAT). Methods: QUASAR™-Body-Phantom was used in this study with oval curved-surfaces to mimic breast, chest wall and head-and-neck sites.A CT-scan was obtained with five radio-opaque markers(ROM) placed on the surface of phantom to mimic the range of incident angles for measurements and dose prediction using 2mm slice thickness.At each ROM, small structure(1mmx2mm) were contoured to obtain mean-doses from TPS.Calculations were performed for open-field,dynamic-wedge,physical-wedge,IMRT and VMAT using Varian-21EX,6&15MV photons using twogrid-sizes:2.5mm and 1mm.Calibration checks were performed to ensure that MOSFETs response were within ±5%.Surface-doses were measured at five locations and compared with TPS calculations. Results: For 6MV: 2.5mm grid-size,mean calculated doses(MCD)were higher by 10%(±7.6),10%(±7.6),20%(±8.5),40%(±7.5),30%(±6.9) and for 1mm grid-size MCD were higher by 0%(±5.7),0%(±4.2),0%(±5.5),1.2%(±5.0),1.1% (±7.8) for open-field,dynamic-wedge,physical-wedge,IMRT,VMAT respectively.For 15MV: 2.5mm grid-size,MCD were higher by 30%(±14.6),30%(±14.6),30%(±14.0),40%(±11.0),30%(±3.5)and for 1mm grid-size MCD were higher by 10% (±10.6), 10%(±9.8),10%(±8.0),30%(±7.8),10%(±3.8) for open-field, dynamic-wedge, physical-wedge, IMRT, VMAT respectively.For 6MV, 86% and 56% of all measured values
Cristy, M.
1980-01-01
Mathematical phantoms of the human body at various ages are employed with Monte Carlo radiation transport codes for calculation of photon specific absorbed fractions. The author has developed a pediatric phantom series based on the design of the adult phantom, but with explicit equations for each organ so that organ sizes and marrow distributions could be assigned properly. Since the phantoms comprise simple geometric shapes, predictive dose capability is limited when geometry is critical to the calculation. Hence, there is a demand for better phantom design in situations where geometry is critical, such as for external irradiation or for internal emitters with low energy photons. Recent advances in computerized axial tomography (CAT) present the potential for derivation of anatomical information, which is so critical to development of phantoms, and ongoing developmental work on compuer architecture to handle large arrays for Monte Carlo calculations should make complex-geometry dose calculations economically feasible within this decade.
Radiation doses inside industrial irradiation installation with linear electron accelerator
Lima, Alexandre R., E-mail: alexandre.lima@cnen.gov.br [Comissao Nacional de Energia Nuclear (CNEN), Rio de Janeiro, RJ (Brazil); Pelegrineli, Samuel Q.; Alo, Gabriel F., E-mail: samuelfisica@yahoo.com.br, E-mail: gabriel.alo@aceletron.com.br [Aceletron Irradiacao Industrial, Aceletrica Comercio e Representacoes Ltda, Rio de Janeiro, RJ (Brazil); Silva, Francisco C.A. Da, E-mail: dasilva@ird.gov.br [Instituto de Radioprotecao e Dosimetria (IRD/CNEN-RJ), Rio de Janeiro, RJ (Brazil)
2015-07-01
Aceletron Industrial Irradiation Company is the unique installation in South America to provide industrial irradiation service using two linear electron accelerators of 18 kW and 10 MeV energy. The electron beam technology allows using electrons to irradiate many goods and materials, such as hospital and medical equipment, cosmetics, herbal products, polymers, peat, gemstones and food. Aceletron Company uses a concrete bunker with 3.66 m of thickness to provide the necessary occupational and environmental radiation protection of X-rays produced. The bunker is divided in main four areas: irradiation room, maze, tower and pit. Inside the irradiation room the x-rays radiation rates are measured in two ways: direct beam and 90 deg C. The rates produced in the conveyor system using 10 MeV energy are 500 Gy/min/mA and 15 Gy/min/mA, respectively. For a 1.8 mA current, the rates produced are 900 Gy/min and 27 Gy/min, respectively. Outside the bunker the radiation rate is at background level, but in the tower door and modulation room the radiation rate is 10 μSv/h. In 2014, during a routine operation, an effective dose of 30.90 mSv was recorded in a monthly individual dosimeter. After the investigation, it was concluded that the dose was only in the dosimeter because it felt inside the irradiation room. As Aceletron Company follows the principles of safety culture, it was decided to perform the radiation isodose curves, inside the four areas of the installation, to know exactly the hotspots positions, exposure times and radiation doses. Five hotspots were chosen taking into account worker's routes and possible operational places. The first experiment was done using a package with three TLD and OSLD dosimeters to obtain better statistical results. The first results for the five hotspots near the accelerator machine showed that the radiation dose rates were between 26 Gy/h and 31 Gy/h. The final measurements were performed using a package with one TLD and one OSLD
Influence of the jaw tracking technique on the dose calculation accuracy of small field VMAT plans.
Swinnen, Ans C C; Öllers, Michel C; Roijen, Erik; Nijsten, Sebastiaan M; Verhaegen, Frank
2017-01-01
The aim of this study was to evaluate experimentally the accuracy of the dose calculation algorithm AcurosXB in small field highly modulated Volumetric Modulated Arc Therapy (VMAT). The 1000SRS detector array inserted in the rotational Octavius 4D phantom (PTW) was used for 3D dose verification of VMAT treatments characterized by small to very small targets. Clinical treatment plans (n = 28) were recalculated on the phantom CT data set in the Eclipse TPS. All measurements were done on a Varian TrueBeamSTx, which can provide the jaw tracking technique (JTT). The effect of disabling the JTT, thereby fixing the jaws at static field size of 3 × 3 cm(2) and applying the MLC to shape the smallest apertures, was investigated for static fields between 0.5 × 0.5-3 × 3 cm(2) and for seven VMAT patients with small brain metastases. The dose calculation accuracy has been evaluated by comparing the measured and calculated dose outputs and dose distributions. The dosimetric agreement has been presented by a local gamma evaluation criterion of 2%/2 mm. Regarding the clinical plans, the mean ± SD of the volumetric gamma evaluation scores considering the dose levels for evaluation of 10%, 50%, 80% and 95% are (96.0 ± 6.9)%, (95.2 ± 6.8)%, (86.7 ± 14.8)% and (56.3 ± 42.3)% respectively. For the smallest field VMAT treatments, discrepancies between calculated and measured doses up to 16% are obtained. The difference between the 1000SRS central chamber measurements compared to the calculated dose outputs for static fields 3 × 3, 2 × 2, 1 × 1 and 0.5 × 0.5 cm(2) collimated with MLC whereby jaws are fixed at 3 × 3 cm(2) and for static fields shaped with the collimator jaws only (MLC retracted), is on average respectively, 0.2%, 0.8%, 6.8%, 5.7% (6 MV) and 0.1%, 1.3%, 11.7%, 21.6% (10 MV). For the seven brain mets patients was found that the smaller the target volumes, the higher the improvement in agreement between measured and calculated doses after disabling the JTT
Egashira, Yusuke; Nishio, Teiji; Matsuura, Taeko; Kameoka, Satoru; Uesaka, Mitsuru
2012-07-01
In proton therapy, pencil-beam algorithms (PBAs) are the most widely used dose calculation methods. However, the PB calculations that employ one-dimensional density scaling neglect the effects of lateral density heterogeneity on the dose distributions, whereas some particles included in such pencil beams could overextend beyond the interface of the density heterogeneity. We have simplified a pencil-beam redefinition algorithm (PBRA), which was proposed for electron therapy, by a spatial resampling technique toward an application for proton therapy. The purpose of this study is to evaluate the calculation results of the spatial resampling technique in terms of lateral density heterogeneity by comparison with the dose distributions that were measured in heterogeneous slab phantoms. The pencil beams are characterized for multiple residual-range (i.e., proton energy) bins. To simplify the PBRA, the given pencil beams are resampled on one or two transport planes, in which smaller sub-beams that are parallel to each other are generated. We addressed the problem of lateral density heterogeneity comparing the calculation results to the dose distributions measured at different depths in heterogeneous slab phantoms using a two-dimensional detector. Two heterogeneity slab phantoms, namely, phantoms A and B, were designed for the measurements and calculations. In phantom A, the heterogeneity slab was placed close to the surface. On the other hand, in phantom B, it was placed close to the Bragg peak in the mono-energetic proton beam. In measurements, lateral dose profiles showed a dose reduction and increment in the vicinity ofx = 0 mm in both phantoms at depths z = 142 and 161 mm due to lateral particle disequilibrium. In phantom B, these dose reduction/increment effects were higher/lower, respectively, than those in phantom A. This is because a longer distance from the surface to the heterogeneous slab increases the strength of proton scattering. Sub-beams, which were
SU-E-T-639: Proton Dose Calculation for Irregular Motion Using a Sliding Interface
Phillips, J; Gueorguiev, G; Grassberger, C; Paganetti, H; Sharp, G [Massachusetts General Hospital, Boston, MA (United States)
2015-06-15
Purpose: While many techniques exist to evaluate dose to regularly moving lung targets, there are few available to calculate dose at tumor positions not present in the 4DCT. We have previously developed a method that extrapolates an existing dose to a new tumor location. In this abstract, we present a novel technique that accounts for relative anatomical shifts at the chest wall interface. We also utilize this procedure to simulate breathing motion functions on a cohort of eleven patients. Amplitudes exceeding the original range of motion were used to evaluate coverage using several aperture and smearing beam settings. Methods: The water-equivalent depth (WED) technique requires an initial dose and CT image at the corresponding tumor position. Each dose volume was converted from its Cartesian geometry into a beam-specific radiological depth space. The sliding chest wall interface was determined by converting the lung contour into this same space. Any dose proximal to the initial boundary of the warped lung contour was held fixed, while the remaining distal dose was moved in the direction of motion along the interface. Results: V95 coverage was computed for each patient using the updated algorithm. Incorporation of the sliding motion yielded large dose differences, with gamma pass rates as low as 69.7% (3mm, 3%) and V95 coverage differences up to 2.0%. Clinical coverage was maintained for most patients with 5 mm excess simulated breathing motion, and up to 10 mm of excess motion was tolerated for a subset of patients and beam settings. Conclusion: We have established a method to determine the maximum allowable excess breathing motion for a given plan on a patient-by-patient basis. By integrating a sliding chest wall interface into our dose calculation technique, we have analyzed the robustness of breathing patterns that differ during treatment from at the time of 4DCT acquisition.
Kanematsu, Nobuyuki
2011-04-01
This work addresses computing techniques for dose calculations in treatment planning with proton and ion beams, based on an efficient kernel-convolution method referred to as grid-dose spreading (GDS) and accurate heterogeneity-correction method referred to as Gaussian beam splitting. The original GDS algorithm suffered from distortion of dose distribution for beams tilted with respect to the dose-grid axes. Use of intermediate grids normal to the beam field has solved the beam-tilting distortion. Interplay of arrangement between beams and grids was found as another intrinsic source of artifact. Inclusion of rectangular-kernel convolution in beam transport, to share the beam contribution among the nearest grids in a regulatory manner, has solved the interplay problem. This algorithmic framework was applied to a tilted proton pencil beam and a broad carbon-ion beam. In these cases, while the elementary pencil beams individually split into several tens, the calculation time increased only by several times with the GDS algorithm. The GDS and beam-splitting methods will complementarily enable accurate and efficient dose calculations for radiotherapy with protons and ions. Copyright © 2010 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.
Calculation of organ doses in x-ray examinations of premature babies.
Smans, Kristien; Tapiovaara, Markku; Cannie, Mieke; Struelens, Lara; Vanhavere, Filip; Smet, Marleen; Bosmans, Hilde
2008-02-01
Lung disease represents one of the most life-threatening conditions in prematurely born children. In the evaluation of the neonatal chest, the primary and most important diagnostic study is the chest radiograph. Since prematurely born children are very sensitive to radiation, those radiographs may lead to a significant radiation detriment. Knowledge of the radiation dose is therefore necessary to justify the exposures. To calculate doses in the entire body and in specific organs, computational models of the human anatomy are needed. Using medical imaging techniques, voxel phantoms have been developed to achieve a representation as close as possible to the anatomical properties. In this study two voxel phantoms, representing prematurely born babies, were created from computed tomography- and magnetic resonance images: Phantom 1 (1910 g) and Phantom 2 (590 g). The two voxel phantoms were used in Monte Carlo calculations (MCNPX) to assess organ doses. The results were compared with the commercially available software package PCXMC in which the available mathematical phantoms can be downsized toward the prematurely born baby. The simple phantom-scaling method used in PCXMC seems to be sufficient to calculate doses for organs within the radiation field. However, one should be careful in specifying the irradiation geometry. Doses in organs that are wholly or partially outside the primary radiation field depend critically on the irradiation conditions and the phantom model.
Wu, Vincent W C; Tse, Teddy K H; Ho, Cola L M; Yeung, Eric C Y
2013-01-01
Monte Carlo (MC) simulation is currently the most accurate dose calculation algorithm in radiotherapy planning but requires relatively long processing time. Faster model-based algorithms such as the anisotropic analytical algorithm (AAA) by the Eclipse treatment planning system and multigrid superposition (MGS) by the XiO treatment planning system are 2 commonly used algorithms. This study compared AAA and MGS against MC, as the gold standard, on brain, nasopharynx, lung, and prostate cancer patients. Computed tomography of 6 patients of each cancer type was used. The same hypothetical treatment plan using the same machine and treatment prescription was computed for each case by each planning system using their respective dose calculation algorithm. The doses at reference points including (1) soft tissues only, (2) bones only, (3) air cavities only, (4) soft tissue-bone boundary (Soft/Bone), (5) soft tissue-air boundary (Soft/Air), and (6) bone-air boundary (Bone/Air), were measured and compared using the mean absolute percentage error (MAPE), which was a function of the percentage dose deviations from MC. Besides, the computation time of each treatment plan was recorded and compared. The MAPEs of MGS were significantly lower than AAA in all types of cancers (pplans was significantly lower than that of the MGS (palgorithms demonstrated dose deviations of less than 4.0% in most clinical cases and their performance was better in homogeneous tissues than at tissue boundaries. In general, MGS demonstrated relatively smaller dose deviations than AAA but required longer computation time.
Accuracy of out-of-field dose calculations by a commercial treatment planning system
Howell, Rebecca M; Scarboro, Sarah B; Kry, S F; Yaldo, Derek Z, E-mail: Rhowell@mdanderson.or [University of Texas Health Science Center Houston, Graduate School of Biomedical Sciences, Houston, TX 77030 (United States)
2010-12-07
The dosimetric accuracy of treatment planning systems (TPSs) decreases for locations outside the treatment field borders. However, the true accuracy of specific TPSs for locations beyond the treatment field borders is not well documented. Our objective was to quantify the accuracy of out-of-field dose predicted by the commercially available Eclipse version 8.6 TPS (Varian Medical Systems, Palo Alto, CA) for a clinical treatment delivered on a Varian Clinac 2100. We calculated (in the TPS) and determined (with thermoluminescent dosimeters) doses at a total of 238 points of measurement (with distance from the field edge ranging from 3.75 to 11.25 cm). Our comparisons determined that the Eclipse TPS underestimated out-of-field doses by an average of 40% over the range of distances examined. As the distance from the treatment field increased, the TPS underestimated the dose with increasing magnitude--up to 55% at 11.25 cm from the treatment field border. These data confirm that accuracy beyond the treatment border is inadequate, and out-of-field data from TPSs should be used only with a clear understanding of this limitation. Studies that require accurate out-of-field dose should use other dose reconstruction methods, such as direct measurements or Monte Carlo calculations.
Accuracy of out-of-field dose calculations by a commercial treatment planning system
Howell, Rebecca M.; Scarboro, Sarah B.; Kry, S. F.; Yaldo, Derek Z.
2010-12-01
The dosimetric accuracy of treatment planning systems (TPSs) decreases for locations outside the treatment field borders. However, the true accuracy of specific TPSs for locations beyond the treatment field borders is not well documented. Our objective was to quantify the accuracy of out-of-field dose predicted by the commercially available Eclipse version 8.6 TPS (Varian Medical Systems, Palo Alto, CA) for a clinical treatment delivered on a Varian Clinac 2100. We calculated (in the TPS) and determined (with thermoluminescent dosimeters) doses at a total of 238 points of measurement (with distance from the field edge ranging from 3.75 to 11.25 cm). Our comparisons determined that the Eclipse TPS underestimated out-of-field doses by an average of 40% over the range of distances examined. As the distance from the treatment field increased, the TPS underestimated the dose with increasing magnitude--up to 55% at 11.25 cm from the treatment field border. These data confirm that accuracy beyond the treatment border is inadequate, and out-of-field data from TPSs should be used only with a clear understanding of this limitation. Studies that require accurate out-of-field dose should use other dose reconstruction methods, such as direct measurements or Monte Carlo calculations.
Mazonakis, Michalis; Berris, Theocharris; Lyraraki, Efrossyni; Damilakis, John
2015-03-01
This study was conducted to calculate the peripheral dose to critical structures and assess the radiation risks from modern radiotherapy for stage IIA/IIB testicular seminoma. A Monte Carlo code was used for treatment simulation on a computational phantom representing an average adult. The initial treatment phase involved anteroposterior and posteroanaterior modified dog-leg fields exposing para-aortic and ipsilateral iliac lymph nodes followed by a cone-down phase for nodal mass irradiation. Peripheral doses were calculated using different modified dog-leg field dimensions and an extended conventional dog-leg portal. The risk models of the BEIR-VII report and ICRP-103 were combined with dosimetric calculations to estimate the probability of developing stochastic effects. Radiotherapy for stage IIA seminoma with a target dose of 30 Gy resulted in a range of 23.0-603.7 mGy to non-targeted peripheral tissues and organs. The corresponding range for treatment of stage IIB disease to a cumulative dose of 36 Gy was 24.2-633.9 mGy. A dose variation of less than 13% was found by altering the field dimensions. Radiotherapy with the conventional instead of the modern modified dog-leg field increased the peripheral dose up to 8.2 times. The calculated heart doses of 589.0-632.9 mGy may increase the risk for developing cardiovascular diseases whereas the testicular dose of more than 231.9 mGy may lead to a temporary infertility. The probability of birth abnormalities in the offspring of cancer survivors was below 0.13% which is much lower than the spontaneous mutation rate. Abdominoplevic irradiation may increase the lifetime intrinsic risk for the induction of secondary malignancies by 0.6-3.9% depending upon the site of interest, patient’s age and tumor dose. Radiotherapy for stage IIA/IIB seminoma with restricted fields and low doses is associated with an increased morbidity. These data may allow the definition of a risk-adapted follow-up scheme for long
A brief look at model-based dose calculation principles, practicalities, and promise.
Sloboda, Ron S; Morrison, Hali; Cawston-Grant, Brie; Menon, Geetha V
2017-02-01
Model-based dose calculation algorithms (MBDCAs) have recently emerged as potential successors to the highly practical, but sometimes inaccurate TG-43 formalism for brachytherapy treatment planning. So named for their capacity to more accurately calculate dose deposition in a patient using information from medical images, these approaches to solve the linear Boltzmann radiation transport equation include point kernel superposition, the discrete ordinates method, and Monte Carlo simulation. In this overview, we describe three MBDCAs that are commercially available at the present time, and identify guidance from professional societies and the broader peer-reviewed literature intended to facilitate their safe and appropriate use. We also highlight several important considerations to keep in mind when introducing an MBDCA into clinical practice, and look briefly at early applications reported in the literature and selected from our own ongoing work. The enhanced dose calculation accuracy offered by a MBDCA comes at the additional cost of modelling the geometry and material composition of the patient in treatment position (as determined from imaging), and the treatment applicator (as characterized by the vendor). The adequacy of these inputs and of the radiation source model, which needs to be assessed for each treatment site, treatment technique, and radiation source type, determines the accuracy of the resultant dose calculations. Although new challenges associated with their familiarization, commissioning, clinical implementation, and quality assurance exist, MBDCAs clearly afford an opportunity to improve brachytherapy practice, particularly for low-energy sources.
A brief look at model-based dose calculation principles, practicalities, and promise
Ron S. Sloboda
2017-02-01
Full Text Available Model-based dose calculation algorithms (MBDCAs have recently emerged as potential successors to the highly practical, but sometimes inaccurate TG-43 formalism for brachytherapy treatment planning. So named for their capacity to more accurately calculate dose deposition in a patient using information from medical images, these approaches to solve the linear Boltzmann radiation transport equation include point kernel superposition, the discrete ordinates method, and Monte Carlo simulation. In this overview, we describe three MBDCAs that are commercially available at the present time, and identify guidance from professional societies and the broader peer-reviewed literature intended to facilitate their safe and appropriate use. We also highlight several important considerations to keep in mind when introducing an MBDCA into clinical practice, and look briefly at early applications reported in the literature and selected from our own ongoing work. The enhanced dose calculation accuracy offered by a MBDCA comes at the additional cost of modelling the geometry and material composition of the patient in treatment position (as determined from imaging, and the treatment applicator (as characterized by the vendor. The adequacy of these inputs and of the radiation source model, which needs to be assessed for each treatment site, treatment technique, and radiation source type, determines the accuracy of the resultant dose calculations. Although new challenges associated with their familiarization, commissioning, clinical implementation, and quality assurance exist, MBDCAs clearly afford an opportunity to improve brachytherapy practice, particularly for low-energy sources.
Tolerances for the accuracy of photon beam dose calculations of treatment planning systems
Venselaar, J; Welleweerd, H; Mijnheer, B
2001-01-01
Background and purpose: To design a consistent set of criteria for acceptability of photon beam dose calculations of treatment planning systems. The set should be applicable in combination with a test package used for evaluation of a treatment planning system, such as the ones proposed by the AAPM T
Knöös, Tommy; Wieslander, Elinore; Cozzi, Luca
2006-01-01
to the fields. A Monte Carlo calculated algorithm input data set and a benchmark set for a virtual linear accelerator have been produced which have facilitated the analysis and interpretation of the results. The more sophisticated models in the type b group exhibit changes in both absorbed dose and its...
Ulmer, W
2010-01-01
We have developed a model for proton depth dose and lateral distributions based on Monte Carlo calculations (GEANT4) and an integration procedure of the Bethe-Bloch equation (BBE). The model accounts for the transport of primary and secondary protons, the creation of recoil protons and heavy recoil nuclei as well as lateral scattering of these contributions. The buildup, which is experimentally observed in higher energy depth dose curves, is modeled by inclusion of two different origins: 1. Secondary reaction protons with a contribution of ca. 65 % of the buildup (for monoenergetic protons). 2. Landau tails as well as Gaussian type of fluctuations for range straggling effects. All parameters of the model for initially monoenergetic proton beams have been obtained from Monte Carlo calculations or checked by them. Furthermore, there are a few parameters, which can be obtained by fitting the model to measured depth dose curves in order to describe individual characteristics of the beamline - the most important b...
Modulation index for VMAT considering both mechanical and dose calculation uncertainties
Park, Jong Min; Park, So Yeon; Kim, Jung In [Dept. of Radiation Oncology, Seoul National University Hospital, Seoul (Korea, Republic of); Ye, Sung Joon [Program in Biomedical Radiation Sciences, Department of Transdisciplinary Studies, Seoul National University Graduate School of Convergence Science and Technology, Seoul (Korea, Republic of); Wu, Hong Gyun [Dept. of Radiation Oncology, Seoul National University College of Medicine, Seoul (Korea, Republic of); Kim, Hyoung Nyoun [Graduate School of Information, Yonsei University, Seoul (Korea, Republic of)
2015-10-15
The mechanical uncertainty of multi-leaf collimator (MLC) movements, gantry rotations and beam control systems as well as inaccurate dose calculations of small or irregular fields result in discrepancies between planned dose distributions as intended to be delivered to the patient, and the actual delivery to the patient. In this study, we designed a weighting factor which considers the size and irregularity of field apertures at each control point (CP) by utilizing the thinning algorithm, an image processing technique. After that, we combined this weighting factor with the previously suggested MIt, which considers the mechanical uncertainty of VMAT. In doing so, we attempted to design a modulation index which considers both mechanical and dose calculation uncertainties due to excessive modulation of VMAT plans. The MI{sub c} (f = 0.5) demonstrated considerable power to predict VMAT delivery accuracy showing strong correlations to various measures of VMAT delivery accuracy.
A dose-response curve for biodosimetry from a 6 MV electron linear accelerator
Lemos-Pinto, M.M.P.; Cadena, M.; Santos, N.; Fernandes, T.S.; Borges, E.; Amaral, A., E-mail: marcelazoo@yahoo.com.br [Universidade Federal de Pernambuco (UFPE), Recife, PE (Brazil). Departamento de Energia Nuclear
2015-10-15
Biological dosimetry (biodosimetry) is based on the investigation of radiation-induced biological effects (biomarkers), mainly dicentric chromosomes, in order to correlate them with radiation dose. To interpret the dicentric score in terms of absorbed dose, a calibration curve is needed. Each curve should be constructed with respect to basic physical parameters, such as the type of ionizing radiation characterized by low or high linear energy transfer (LET) and dose rate. This study was designed to obtain dose calibration curves by scoring of dicentric chromosomes in peripheral blood lymphocytes irradiated in vitro with a 6 MV electron linear accelerator (Mevatron M, Siemens, USA). Two software programs, CABAS (Chromosomal Aberration Calculation Software) and Dose Estimate, were used to generate the curve. The two software programs are discussed; the results obtained were compared with each other and with other published low LET radiation curves. Both software programs resulted in identical linear and quadratic terms for the curve presented here, which was in good agreement with published curves for similar radiation quality and dose rates. (author)
A dose-response curve for biodosimetry from a 6 MV electron linear accelerator
Lemos-Pinto, M.M.P.; Cadena, M.; Santos, N.; Fernandes, T.S.; Borges, E.; Amaral, A.
2015-01-01
Biological dosimetry (biodosimetry) is based on the investigation of radiation-induced biological effects (biomarkers), mainly dicentric chromosomes, in order to correlate them with radiation dose. To interpret the dicentric score in terms of absorbed dose, a calibration curve is needed. Each curve should be constructed with respect to basic physical parameters, such as the type of ionizing radiation characterized by low or high linear energy transfer (LET) and dose rate. This study was designed to obtain dose calibration curves by scoring of dicentric chromosomes in peripheral blood lymphocytes irradiated in vitro with a 6 MV electron linear accelerator (Mevatron M, Siemens, USA). Two software programs, CABAS (Chromosomal Aberration Calculation Software) and Dose Estimate, were used to generate the curve. The two software programs are discussed; the results obtained were compared with each other and with other published low LET radiation curves. Both software programs resulted in identical linear and quadratic terms for the curve presented here, which was in good agreement with published curves for similar radiation quality and dose rates. PMID:26445334
A dose-response curve for biodosimetry from a 6 MV electron linear accelerator.
Lemos-Pinto, M M P; Cadena, M; Santos, N; Fernandes, T S; Borges, E; Amaral, A
2015-10-01
Biological dosimetry (biodosimetry) is based on the investigation of radiation-induced biological effects (biomarkers), mainly dicentric chromosomes, in order to correlate them with radiation dose. To interpret the dicentric score in terms of absorbed dose, a calibration curve is needed. Each curve should be constructed with respect to basic physical parameters, such as the type of ionizing radiation characterized by low or high linear energy transfer (LET) and dose rate. This study was designed to obtain dose calibration curves by scoring of dicentric chromosomes in peripheral blood lymphocytes irradiated in vitro with a 6 MV electron linear accelerator (Mevatron M, Siemens, USA). Two software programs, CABAS (Chromosomal Aberration Calculation Software) and Dose Estimate, were used to generate the curve. The two software programs are discussed; the results obtained were compared with each other and with other published low LET radiation curves. Both software programs resulted in identical linear and quadratic terms for the curve presented here, which was in good agreement with published curves for similar radiation quality and dose rates.
A dose-response curve for biodosimetry from a 6 MV electron linear accelerator
M.M.P. Lemos-Pinto
2015-01-01
Full Text Available Biological dosimetry (biodosimetry is based on the investigation of radiation-induced biological effects (biomarkers, mainly dicentric chromosomes, in order to correlate them with radiation dose. To interpret the dicentric score in terms of absorbed dose, a calibration curve is needed. Each curve should be constructed with respect to basic physical parameters, such as the type of ionizing radiation characterized by low or high linear energy transfer (LET and dose rate. This study was designed to obtain dose calibration curves by scoring of dicentric chromosomes in peripheral blood lymphocytes irradiated in vitro with a 6 MV electron linear accelerator (Mevatron M, Siemens, USA. Two software programs, CABAS (Chromosomal Aberration Calculation Software and Dose Estimate, were used to generate the curve. The two software programs are discussed; the results obtained were compared with each other and with other published low LET radiation curves. Both software programs resulted in identical linear and quadratic terms for the curve presented here, which was in good agreement with published curves for similar radiation quality and dose rates.
Individual Dose Calculations with Use of the Revised Techa River Dosimetry System TRDS-2009D
Degteva, M. O.; Shagina, N. B.; Tolstykh, E. I.; Vorobiova, M. I.; Anspaugh, L. R.; Napier, Bruce A.
2009-10-23
An updated deterministic version of the Techa River Dosimetry System (TRDS-2009D) has been developed to estimate individual doses from external exposure and intake of radionuclides for residents living on the Techa River contaminated as a result of radioactive releases from the Mayak plutonium facility in 1949–1956. The TRDS-2009D is designed as a flexible system that uses, depending on the input data for an individual, various elements of system databases to provide the dosimetric variables requested by the user. Several phases are included in the computation schedule. The first phase includes calculations with use of a common protocol for all cohort members based on village-average-intake functions and external dose rates; individual data on age, gender and history of residence are included in the first phase. This phase results in dose estimates similar to those obtained with system TRDS-2000 used previously to derive risks of health effects in the Techa River Cohort. The second phase includes refinement of individual internal doses for those persons who have had body-burden measurements or exposure parameters specific to the household where he/she lived on the Techa River. The third phase includes summation of individual doses from environmental exposure and from radiological examinations. The results of TRDS-2009D dose calculations have demonstrated for the ETRC members on average a moderate increase in RBM dose estimates (34%) and a minor increase (5%) in estimates of stomach dose. The calculations for the members of the ETROC indicated similar small changes for stomach, but significant increase in RBM doses (400%). Individual-dose assessments performed with use of TRDS-2009D have been provided to epidemiologists for exploratory risk analysis in the ETRC and ETROC. These data provide an opportunity to evaluate the possible impact on radiogenic risk of such factors as confounding exposure (environmental and medical), changes in the Techa River source
Suitability of point kernel dose calculation techniques in brachytherapy treatment planning
Lakshminarayanan Thilagam
2010-01-01
Full Text Available Brachytherapy treatment planning system (TPS is necessary to estimate the dose to target volume and organ at risk (OAR. TPS is always recommended to account for the effect of tissue, applicator and shielding material heterogeneities exist in applicators. However, most brachytherapy TPS software packages estimate the absorbed dose at a point, taking care of only the contributions of individual sources and the source distribution, neglecting the dose perturbations arising from the applicator design and construction. There are some degrees of uncertainties in dose rate estimations under realistic clinical conditions. In this regard, an attempt is made to explore the suitability of point kernels for brachytherapy dose rate calculations and develop new interactive brachytherapy package, named as BrachyTPS, to suit the clinical conditions. BrachyTPS is an interactive point kernel code package developed to perform independent dose rate calculations by taking into account the effect of these heterogeneities, using two regions build up factors, proposed by Kalos. The primary aim of this study is to validate the developed point kernel code package integrated with treatment planning computational systems against the Monte Carlo (MC results. In the present work, three brachytherapy applicators commonly used in the treatment of uterine cervical carcinoma, namely (i Board of Radiation Isotope and Technology (BRIT low dose rate (LDR applicator and (ii Fletcher Green type LDR applicator (iii Fletcher Williamson high dose rate (HDR applicator, are studied to test the accuracy of the software. Dose rates computed using the developed code are compared with the relevant results of the MC simulations. Further, attempts are also made to study the dose rate distribution around the commercially available shielded vaginal applicator set (Nucletron. The percentage deviations of BrachyTPS computed dose rate values from the MC results are observed to be within plus/minus 5
Calculation of residence times and radiation doses using the standard PC software Excel.
Herzog, H; Zilken, H; Niederbremer, A; Friedrich, W; Müller-Gärtner, H W
1997-12-01
We developed a program which aims to facilitate the calculation of radiation doses to single organs and the whole body. IMEDOSE uses Excel to include calculations, graphical displays, and interactions with the user in a single general-purpose PC software tool. To start the procedure the input data are copied into a spreadsheet. They must represent percentage uptake values of several organs derived from measurements in animals or humans. To extrapolate these data up to seven half-lives of the radionuclide, fitting to one or two exponentional functions is included and can be checked by the user. By means of the approximate time-activity information the cumulated activity or residence times are calculated. Finally these data are combined with the absorbed fraction doses (S-values) given by MIRD pamphlet No. 11 to yield radiation doses, the effective dose equivalent and the effective dose. These results are presented in a final table. Interactions are realized with push-buttons and drop-down menus. Calculations use the Visual Basic tool of Excel. In order to test our program, biodistribution data of fluorine-18 fluorodeoxyglucose were taken from the literature (Meija et al., J Nucl Med 1991; 32:699-706). For a 70-kg adult the resulting radiation doses of all target organs listed in MIRD 11 were different from the ICRP 53 values by 1%+/-18% on the average. When the residence times were introduced into MIRDOSE3 (Stabin, J Nucl Med 1996; 37:538-546) the mean difference between our results and those of MIRDOSE3 was -3%+/-6%. Both outcomes indicate the validity of the present approach.
Tang, Grace; Earl, Matthew A.; Luan, Shuang; Wang, Chao; Cao, Daliang; Yu, Cedric X.; Naqvi, Shahid A.
2008-09-01
Dose calculations for radiation arc therapy are traditionally performed by approximating continuous delivery arcs with multiple static beams. For 3D conformal arc treatments, the shape and weight variation per degree is usually small enough to allow arcs to be approximated by static beams separated by 5°-10°. But with intensity-modulated arc therapy (IMAT), the variation in shape and dose per degree can be large enough to require a finer angular spacing. With the increase in the number of beams, a deterministic dose calculation method, such as collapsed-cone convolution/superposition, will require proportionally longer computational times, which may not be practical clinically. We propose to use a homegrown Monte Carlo kernel-superposition technique (MCKS) to compute doses for rotational delivery. The IMAT plans were generated with 36 static beams, which were subsequently interpolated into finer angular intervals for dose calculation to mimic the continuous arc delivery. Since MCKS uses random sampling of photons, the dose computation time only increased insignificantly for the interpolated-static-beam plans that may involve up to 720 beams. Ten past IMRT cases were selected for this study. Each case took approximately 15-30 min to compute on a single CPU running Mac OS X using the MCKS method. The need for a finer beam spacing is dictated by how fast the beam weights and aperture shapes change between the adjacent static planning beam angles. MCKS, however, obviates the concern by allowing hundreds of beams to be calculated in practically the same time as for a few beams. For more than 43 beams, MCKS usually takes less CPU time than the collapsed-cone algorithm used by the Pinnacle3 planning system.
Calculation of mean central dose in interstitial brachytherapy using Delaunay triangulation.
Astrahan, M A; Streeter, O E; Jozsef, G
2001-06-01
In 1997 the ICRU published Report 58 "Dose and Volume Specification for Reporting Interstitial Therapy" with the objective of addressing the problem of absorbed dose specification for reporting contemporary interstitial therapy. One of the concepts proposed in that report is "mean central dose." The fundamental goal of the mean central dose (MCD) calculation is to obtain a single, readily reportable and intercomparable value which is representative of dose in regions of the implant "where the dose gradient approximates a plateau." Delaunay triangulation (DT) is a method used in computational geometry to partition the space enclosed by the convex hull of a set of distinct points P into a set of nonoverlapping cells. In the three-dimensional case, each point of P becomes a vertex of a tetrahedron and the result of the DT is a set of tetrahedra. All treatment planning for interstitial brachytherapy inherently requires that the location of the radioactive sources, or dwell positions in the case of HDR, be known or digitized. These source locations may be regarded as a set of points representing the implanted volume. Delaunay triangulation of the source locations creates a set of tetrahedra without manual intervention. The geometric centers of these tetrahedra define a new set of points which lie "in between" the radioactive sources and which are distributed uniformly over the volume of the implant. The arithmetic mean of the dose at these centers is a three dimensional analog of the two-dimensional triangulation and inspection methods proposed for calculating MCD in ICRU 58. We demonstrate that DT can be successfully incorporated into a computerized treatment planning system and used to calculate the MCD.
Calculation of Electron Beam Potential Energy from RF Photocathode Gun
Liu Wan Ming; Power, John G; Wang, Haitao
2005-01-01
In this paper, we consider the contribution of potential energy to beam dynamics as simulated by PARMELA at low energies (10 - 30MeV). We have developed a routine to calculate the potential energy of the relativistic electron beam using the static coulomb potential in the rest frame (first order approximation as in PARMELA). We found that the potential energy contribution to the beam dynamics could be very significant, particularly with high charge beams generated by an RF photocathode gun. Our results show that when the potential energy is counted correctly and added to the kinetic energy from PARMELA, the total energy is conserved. Simulation results of potential and kinetic energies for short beams (~1 mm) at various charges (1 - 100 nC) generated by a high current RF photocathode gun are presented.
Electronic structure calculations toward new potentially AChE inhibitors
de Paula, A. A. N.; Martins, J. B. L.; Gargano, R.; dos Santos, M. L.; Romeiro, L. A. S.
2007-10-01
The main purpose of this study was the use of natural non-isoprenoid phenolic lipid of cashew nut shell liquid from Anacardium occidentale as lead material for generating new potentially candidates of acetylcholinesterase inhibitors. Therefore, we studied the electronic structure of 15 molecules derivatives from the cardanol using the following groups: methyl, acetyl, N, N-dimethylcarbamoyl, N, N-dimethylamine, N, N-diethylamine, piperidine, pyrrolidine, and N-benzylamine. The calculations were performed at RHF level using 6-31G, 6-31G(d), 6-31+G(d) and 6-311G(d,p) basis functions. Among the proposed compounds we found that the structures with substitution by acetyl, N, N-dimethylcarbamoyl, N, N-dimethylamine, and pyrrolidine groups were better correlated to rivastigmine indicating possible activity.
Gradient type optimization methods for electronic structure calculations
Zhang, Xin; Wen, Zaiwen; Zhou, Aihui
2013-01-01
The density functional theory (DFT) in electronic structure calculations can be formulated as either a nonlinear eigenvalue or direct minimization problem. The most widely used approach for solving the former is the so-called self-consistent field (SCF) iteration. A common observation is that the convergence of SCF is not clear theoretically while approaches with convergence guarantee for solving the latter are often not competitive to SCF numerically. In this paper, we study gradient type methods for solving the direct minimization problem by constructing new iterations along the gradient on the Stiefel manifold. Global convergence (i.e., convergence to a stationary point from any initial solution) as well as local convergence rate follows from the standard theory for optimization on manifold directly. A major computational advantage is that the computation of linear eigenvalue problems is no longer needed. The main costs of our approaches arise from the assembling of the total energy functional and its grad...
Neutron spectra and dose equivalents calculated in tissue for high-energy radiation therapy
Kry, Stephen F.; Howell, Rebecca M.; Salehpour, Mohammad; Followill, David S. [Department of Radiation Physics, University of Texas M. D. Anderson Cancer Center, 1515 Holcombe Boulevard, Houston, Texas 77030 (United States)
2009-04-15
Neutrons are by-products of high-energy radiation therapy and a source of dose to normal tissues. Thus, the presence of neutrons increases a patient's risk of radiation-induced secondary cancer. Although neutrons have been thoroughly studied in air, little research has been focused on neutrons at depths in the patient where radiosensitive structures may exist, resulting in wide variations in neutron dose equivalents between studies. In this study, we characterized properties of neutrons produced during high-energy radiation therapy as a function of their depth in tissue and for different field sizes and different source-to-surface distances (SSD). We used a previously developed Monte Carlo model of an accelerator operated at 18 MV to calculate the neutron fluences, energy spectra, quality factors, and dose equivalents in air and in tissue at depths ranging from 0.1 to 25 cm. In conjunction with the sharply decreasing dose equivalent with increased depth in tissue, the authors found that the neutron energy spectrum changed drastically as a function of depth in tissue. The neutron fluence decreased gradually as the depth increased, while the average neutron energy decreased sharply with increasing depth until a depth of approximately 7.5 cm in tissue, after which it remained nearly constant. There was minimal variation in the quality factor as a function of depth. At a given depth in tissue, the neutron dose equivalent increased slightly with increasing field size and decreasing SSD; however, the percentage depth-dose equivalent curve remained constant outside the primary photon field. Because the neutron dose equivalent, fluence, and energy spectrum changed substantially with depth in tissue, we concluded that when the neutron dose equivalent is being determined at a depth within a patient, the spectrum and quality factor used should be appropriate for depth rather than for in-air conditions. Alternately, an appropriate percent depth-dose equivalent curve
Kim, Min Joo; Lee, Seu Ran; Suh, Tae Suk [Dept. of Biomedical Engineering, The Catholic University of Korea, Bucheon (Korea, Republic of)
2011-11-15
Modern radiation therapy techniques, such as Image-guided radiation therapy (IGRT), Adaptive radiation therapy (ART) has become a routine clinical practice on linear accelerators for the increase the tumor dose conformity and improvement of normal tissue sparing at the same time. For these highly developed techniques, megavoltage cone beam computed tomography (MVCBCT) system produce volumetric images at just one rotation of the x-ray beam source and detector on the bottom of conventional linear accelerator for real-time application of patient condition into treatment planning. MV CBCT image scan be directly registered to a reference CT data set which is usually kilo-voltage fan-beam computed tomography (kVFBCT) on treatment planning system and the registered image scan be used to adjust patient set-up error. However, to use MV CBCT images in radiotherapy, reliable electron density (ED) distribution are required. Patients scattering, beam hardening and softening effect caused by different energy application between kVCT, MV CBCT can cause cupping artifacts in MV CBCT images and distortion of Houns field Unit (HU) to ED conversion. The goal of this study, for reliable application of MV CBCT images into dose calculation, MV CBCT images was modified to correct distortion of HU to ED using the relationship of HU and ED from kV FBCT and MV CBCT images. The HU-density conversion was performed on MV CBCT image set using Dose difference map was showing in Figure 1. Finally, percentage differences above 3% were reduced depending on applying density calibration method. As a result, total error co uld be reduced to under 3%. The present study demonstrates that dose calculation accuracy using MV CBCT image set can be improved my applying HU-density conversion method. The dose calculation and comparison of dose distribution from MV CBCT image set with/without HU-density conversion method was performed. An advantage of this study compared to other approaches is that HU
Zhu, Jinhan; Chen, Lixin; Chen, Along; Luo, Guangwen; Deng, Xiaowu; Liu, Xiaowei
2015-04-11
To use a graphic processing unit (GPU) calculation engine to implement a fast 3D pre-treatment dosimetric verification procedure based on an electronic portal imaging device (EPID). The GPU algorithm includes the deconvolution and convolution method for the fluence-map calculations, the collapsed-cone convolution/superposition (CCCS) algorithm for the 3D dose calculations and the 3D gamma evaluation calculations. The results of the GPU-based CCCS algorithm were compared to those of Monte Carlo simulations. The planned and EPID-based reconstructed dose distributions in overridden-to-water phantoms and the original patients were compared for 6 MV and 10 MV photon beams in intensity-modulated radiation therapy (IMRT) treatment plans based on dose differences and gamma analysis. The total single-field dose computation time was less than 8 s, and the gamma evaluation for a 0.1-cm grid resolution was completed in approximately 1 s. The results of the GPU-based CCCS algorithm exhibited good agreement with those of the Monte Carlo simulations. The gamma analysis indicated good agreement between the planned and reconstructed dose distributions for the treatment plans. For the target volume, the differences in the mean dose were less than 1.8%, and the differences in the maximum dose were less than 2.5%. For the critical organs, minor differences were observed between the reconstructed and planned doses. The GPU calculation engine was used to boost the speed of 3D dose and gamma evaluation calculations, thus offering the possibility of true real-time 3D dosimetric verification.
Adaptations in Electronic Structure Calculations in Heterogeneous Environments
Talamudupula, Sai [Iowa State Univ., Ames, IA (United States)
2011-01-01
Modern quantum chemistry deals with electronic structure calculations of unprecedented complexity and accuracy. They demand full power of high-performance computing and must be in tune with the given architecture for superior e ciency. To make such applications resourceaware, it is desirable to enable their static and dynamic adaptations using some external software (middleware), which may monitor both system availability and application needs, rather than mix science with system-related calls inside the application. The present work investigates scienti c application interlinking with middleware based on the example of the computational chemistry package GAMESS and middleware NICAN. The existing synchronous model is limited by the possible delays due to the middleware processing time under the sustainable runtime system conditions. Proposed asynchronous and hybrid models aim at overcoming this limitation. When linked with NICAN, the fragment molecular orbital (FMO) method is capable of adapting statically and dynamically its fragment scheduling policy based on the computing platform conditions. Signi cant execution time and throughput gains have been obtained due to such static adaptations when the compute nodes have very di erent core counts. Dynamic adaptations are based on the main memory availability at run time. NICAN prompts FMO to postpone scheduling certain fragments, if there is not enough memory for their immediate execution. Hence, FMO may be able to complete the calculations whereas without such adaptations it aborts.
Dosimetric evaluation of photon dose calculation under jaw and MLC shielding
Fogliata, A.; Clivio, A.; Vanetti, E.; Nicolini, G.; Belosi, M. F.; Cozzi, L. [Oncology Institute of Southern Switzerland, Medical Physics Unit, Bellinzona 6500 (Switzerland)
2013-10-15
Purpose: The accuracy of photon dose calculation algorithms in out-of-field regions is often neglected, despite its importance for organs at risk and peripheral dose evaluation. The present work has assessed this for the anisotropic analytical algorithm (AAA) and the Acuros-XB algorithms implemented in the Eclipse treatment planning system. Specifically, the regions shielded by the jaw, or the MLC, or both MLC and jaw for flattened and unflattened beams have been studied.Methods: The accuracy in out-of-field dose under different conditions was studied for two different algorithms. Measured depth doses out of the field, for different field sizes and various distances from the beam edge were compared with the corresponding AAA and Acuros-XB calculations in water. Four volumetric modulated arc therapy plans (in the RapidArc form) were optimized in a water equivalent phantom, PTW Octavius, to obtain a region always shielded by the MLC (or MLC and jaw) during the delivery. Doses to different points located in the shielded region and in a target-like structure were measured with an ion chamber, and results were compared with the AAA and Acuros-XB calculations. Photon beams of 6 and 10 MV, flattened and unflattened were used for the tests.Results: Good agreement between calculated and measured depth doses was found using both algorithms for all points measured at depth greater than 3 cm. The mean dose differences (±1SD) were −8%± 16%, −3%± 15%, −16%± 18%, and −9%± 16% for measurements vs AAA calculations and −10%± 14%, −5%± 12%, −19%± 17%, and −13%± 14% for Acuros-XB, for 6X, 6 flattening-filter free (FFF), 10X, and 10FFF beams, respectively. The same figures for dose differences relative to the open beam central axis dose were: −0.1%± 0.3%, 0.0%± 0.4%, −0.3%± 0.3%, and −0.1%± 0.3% for AAA and −0.2%± 0.4%, −0.1%± 0.4%, −0.5%± 0.5%, and −0.3%± 0.4% for Acuros-XB. Buildup dose was overestimated with AAA, while Acuros-XB gave
Dosimetric evaluation of photon dose calculation under jaw and MLC shielding.
Fogliata, A; Clivio, A; Vanetti, E; Nicolini, G; Belosi, M F; Cozzi, L
2013-10-01
The accuracy of photon dose calculation algorithms in out-of-field regions is often neglected, despite its importance for organs at risk and peripheral dose evaluation. The present work has assessed this for the anisotropic analytical algorithm (AAA) and the Acuros-XB algorithms implemented in the Eclipse treatment planning system. Specifically, the regions shielded by the jaw, or the MLC, or both MLC and jaw for flattened and unflattened beams have been studied. The accuracy in out-of-field dose under different conditions was studied for two different algorithms. Measured depth doses out of the field, for different field sizes and various distances from the beam edge were compared with the corresponding AAA and Acuros-XB calculations in water. Four volumetric modulated arc therapy plans (in the RapidArc form) were optimized in a water equivalent phantom, PTW Octavius, to obtain a region always shielded by the MLC (or MLC and jaw) during the delivery. Doses to different points located in the shielded region and in a target-like structure were measured with an ion chamber, and results were compared with the AAA and Acuros-XB calculations. Photon beams of 6 and 10 MV, flattened and unflattened were used for the tests. Good agreement between calculated and measured depth doses was found using both algorithms for all points measured at depth greater than 3 cm. The mean dose differences (± 1SD) were -8% ± 16%, -3% ± 15%, -16% ± 18%, and -9% ± 16% for measurements vs AAA calculations and -10% ± 14%, -5% ± 12%, -19% ± 17%, and -13% ± 14% for Acuros-XB, for 6X, 6 flattening-filter free (FFF), 10X, and 10FFF beams, respectively. The same figures for dose differences relative to the open beam central axis dose were: -0.1% ± 0.3%, 0.0% ± 0.4%, -0.3% ± 0.3%, and -0.1% ± 0.3% for AAA and -0.2% ± 0.4%, -0.1% ± 0.4%, -0.5% ± 0.5%, and -0.3% ± 0.4% for Acuros-XB. Buildup dose was overestimated with AAA, while Acuros-XB gave results more consistent with
Neutron dose calculation at the maze entrance of medical linear accelerator rooms.
Falcão, R C; Facure, A; Silva, A X
2007-01-01
Currently, teletherapy machines of cobalt and caesium are being replaced by linear accelerators. The maximum photon energy in these machines can vary from 4 to 25 MeV, and one of the great advantages of these equipments is that they do not have a radioactive source incorporated. High-energy (E > 10 MV) medical linear accelerators offer several physical advantages over lower energy ones: the skin dose is lower, the beam is more penetrating, and the scattered dose to tissues outside the target volume is smaller. Nevertheless, the contamination of undesirable neutrons in the therapeutic beam, generated by the high-energy photons, has become an additional problem as long as patient protection and occupational doses are concerned. The treatment room walls are shielded to attenuate the primary and secondary X-ray fluence, and this shielding is generally adequate to attenuate the neutrons. However, these neutrons are scattered through the treatment room maze and may result in a radiological problem at the door entrance, a high occupancy area in a radiotherapy facility. In this article, we used MCNP Monte Carlo simulation to calculate neutron doses in the maze of radiotherapy rooms and we suggest an alternative method to the Kersey semi-empirical model of neutron dose calculation at the entrance of mazes. It was found that this new method fits better measured values found in literature, as well as our Monte Carlo simulated ones.
PCXMC, a Monte Carlo program for calculating patient doses in medical x-ray examinations
Tapiovaara, M.; Siiskonen, T.
2008-11-15
PCXMC is a Monte Carlo program for calculating patients' organ doses and effective doses in medical x-ray examinations. The organs and tissues considered in the program are: active bone marrow, adrenals, brain, breasts, colon (upper and lower large intestine), extrathoracic airways, gall bladder, heart, kidneys, liver, lungs, lymph nodes, muscle, oesophagus, oral mucosa, ovaries, pancreas, prostate, salivary glands, skeleton, skin, small intestine, spleen, stomach, testicles, thymus, thyroid, urinary bladder and uterus. The program calculates the effective dose with both the present tissue weighting factors of ICRP Publication 103 (2007) and the old tissue weighting factors of ICRP Publication 60 (1991). The anatomical data are based on the mathematical hermaphrodite phantom models of Cristy and Eckerman (1987), which describe patients of six different ages: new-born, 1, 5, 10, 15-year-old and adult patients. Some changes are made to these phantoms in order to make them more realistic for external irradiation conditions and to enable the calculation of the effective dose according to the new ICRP Publication 103 tissue weighting factors. The phantom sizes are adjustable to mimic patients of an arbitrary weight and height. PCXMC allows a free adjustment of the x-ray beam projection and other examination conditions of projection radiography and fluoroscopy
Tissue decomposition from dual energy CT data for MC based dose calculation in particle therapy
Hünemohr, Nora, E-mail: n.huenemohr@dkfz.de; Greilich, Steffen [Medical Physics in Radiation Oncology, German Cancer Research Center, 69120 Heidelberg (Germany); Paganetti, Harald; Seco, Joao [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, Massachusetts 02114 (United States); Jäkel, Oliver [Medical Physics in Radiation Oncology, German Cancer Research Center, 69120 Heidelberg, Germany and Department of Radiation Oncology and Radiation Therapy, University Hospital of Heidelberg, 69120 Heidelberg (Germany)
2014-06-15
Purpose: The authors describe a novel method of predicting mass density and elemental mass fractions of tissues from dual energy CT (DECT) data for Monte Carlo (MC) based dose planning. Methods: The relative electron density ϱ{sub e} and effective atomic number Z{sub eff} are calculated for 71 tabulated tissue compositions. For MC simulations, the mass density is derived via one linear fit in the ϱ{sub e} that covers the entire range of tissue compositions (except lung tissue). Elemental mass fractions are predicted from the ϱ{sub e} and the Z{sub eff} in combination. Since particle therapy dose planning and verification is especially sensitive to accurate material assignment, differences to the ground truth are further analyzed for mass density, I-value predictions, and stopping power ratios (SPR) for ions. Dose studies with monoenergetic proton and carbon ions in 12 tissues which showed the largest differences of single energy CT (SECT) to DECT are presented with respect to range uncertainties. The standard approach (SECT) and the new DECT approach are compared to reference Bragg peak positions. Results: Mean deviations to ground truth in mass density predictions could be reduced for soft tissue from (0.5±0.6)% (SECT) to (0.2±0.2)% with the DECT method. Maximum SPR deviations could be reduced significantly for soft tissue from 3.1% (SECT) to 0.7% (DECT) and for bone tissue from 0.8% to 0.1%. MeanI-value deviations could be reduced for soft tissue from (1.1±1.4%, SECT) to (0.4±0.3%) with the presented method. Predictions of elemental composition were improved for every element. Mean and maximum deviations from ground truth of all elemental mass fractions could be reduced by at least a half with DECT compared to SECT (except soft tissue hydrogen and nitrogen where the reduction was slightly smaller). The carbon and oxygen mass fraction predictions profit especially from the DECT information. Dose studies showed that most of the 12 selected tissues would
Calculation of electron trajectory and energy deposition in no screening region
Kia, Mohammad Reza, E-mail: m_r_kia@aut.ac.ir; Noshad, Houshyar, E-mail: hnoshad@aut.ac.ir
2016-01-01
The probability density function (PDF) of energy for inelastic collision is obtained by solving the integro-differential form of the quantity equation with the Bhabha differential cross section for particles with spin 1/2. Hence, the total PDF in no screening region is determined by folding theory with the following two assumptions: (1) the electron loses energy by collision and radiation and (2) the electron velocity does not change with a thin absorber. Therefore, a set of coupled stochastic differential equations based on the deviation and energy loss PDFs for electron is presented to obtain the electron trajectory inside the target. The energy PDFs for an electron beam with incident energy of 15.7 MeV inside aluminum and copper are calculated. Besides, the dose distributions for an electron beam with incident energies of 20, 10.2, 6, and 0.5 MeV in water are obtained. The results are in excellent agreement with the experimental data reported in the literature.
Development of environmental dose calculation program TEDII-60 and its application
Kang, Sang Ho; Chung, Chan Young [KOPEC, Yongin City (Korea, Republic of)
1999-07-01
Korean nuclear regulations have introduced new laws regarding dose limits to the public. These laws have adopted dose limits and radiological protection guides as prescribed in the International Commission on Radiation Protection Publication 60 (ICRP-60). These new public dose limits should be applied and implemented to the design of nuclear plants by using radiation protection data provided in ICRP-60. As a result of these new regulations, KOPEC developed a new offsite dose assessment computer code called TEDII-60 (Totally Visualized Environmental Dose Assessment Code System Incorporating ICRP-60 Recommendations). While ensuring compliance with ICRP-60 recommendations, this new code integrated all the related processes in the code such as; meteorological data processing, atmospheric and oceanographic dispersion analysis, and expected annual effluent estimation. TEDII-60 uses graphic I/O display to do offsite dose analyses. Pathway analyses are executed for a maximum of 27 tissues or organs of 6 age groups for both maximally exposed individuals including a population within 80 km of the site. This code uses dose coefficients from recent ICRP publications and US EPA reports that include about 800 radionuclides. TEDII-60 can assess doses from all the chemical forms and particle sizes from the 0.001-10 {mu}m AMAD of any nuclide. This code has been verified and benchmarked against other similar codes used in the nuclear industry. TEDII-60 will be used to prepare calculations dealing with offsite dose assessments to be incorporated into safety analysis reports and environmental reports for new NPPs; and plants that are currently in operation when administrative procedures are implemented. (author). 24 refs., 5 tabs., 13 figs.
A fourier analysis on the maximum acceptable grid size for discrete proton beam dose calculation.
Li, Haisen S; Romeijn, H Edwin; Dempsey, James F
2006-09-01
We developed an analytical method for determining the maximum acceptable grid size for discrete dose calculation in proton therapy treatment plan optimization, so that the accuracy of the optimized dose distribution is guaranteed in the phase of dose sampling and the superfluous computational work is avoided. The accuracy of dose sampling was judged by the criterion that the continuous dose distribution could be reconstructed from the discrete dose within a 2% error limit. To keep the error caused by the discrete dose sampling under a 2% limit, the dose grid size cannot exceed a maximum acceptable value. The method was based on Fourier analysis and the Shannon-Nyquist sampling theorem as an extension of our previous analysis for photon beam intensity modulated radiation therapy [J. F. Dempsey, H. E. Romeijn, J. G. Li, D. A. Low, and J. R. Palta, Med. Phys. 32, 380-388 (2005)]. The proton beam model used for the analysis was a near monoenergetic (of width about 1% the incident energy) and monodirectional infinitesimal (nonintegrated) pencil beam in water medium. By monodirection, we mean that the proton particles are in the same direction before entering the water medium and the various scattering prior to entrance to water is not taken into account. In intensity modulated proton therapy, the elementary intensity modulation entity for proton therapy is either an infinitesimal or finite sized beamlet. Since a finite sized beamlet is the superposition of infinitesimal pencil beams, the result of the maximum acceptable grid size obtained with infinitesimal pencil beam also applies to finite sized beamlet. The analytic Bragg curve function proposed by Bortfeld [T. Bortfeld, Med. Phys. 24, 2024-2033 (1997)] was employed. The lateral profile was approximated by a depth dependent Gaussian distribution. The model included the spreads of the Bragg peak and the lateral profiles due to multiple Coulomb scattering. The dependence of the maximum acceptable dose grid size on the
Development of CT scanner models for patient organ dose calculations using Monte Carlo methods
Gu, Jianwei
CT scanner models in this dissertation were versatile and accurate tools for estimating dose to different patient phantoms undergoing various CT procedures. The organ doses from kV and MV CBCT were also calculated. This dissertation finally summarizes areas where future research can be performed including MV CBCT further validation and application, dose reporting software and image and dose correlation study.
Patient-specific Monte Carlo dose calculations for 103Pd breast brachytherapy
Miksys, N.; Cygler, J. E.; Caudrelier, J. M.; Thomson, R. M.
2016-04-01
This work retrospectively investigates patient-specific Monte Carlo (MC) dose calculations for 103Pd permanent implant breast brachytherapy, exploring various necessary assumptions for deriving virtual patient models: post-implant CT image metallic artifact reduction (MAR), tissue assignment schemes (TAS), and elemental tissue compositions. Three MAR methods (thresholding, 3D median filter, virtual sinogram) are applied to CT images; resulting images are compared to each other and to uncorrected images. Virtual patient models are then derived by application of different TAS ranging from TG-186 basic recommendations (mixed adipose and gland tissue at uniform literature-derived density) to detailed schemes (segmented adipose and gland with CT-derived densities). For detailed schemes, alternate mass density segmentation thresholds between adipose and gland are considered. Several literature-derived elemental compositions for adipose, gland and skin are compared. MC models derived from uncorrected CT images can yield large errors in dose calculations especially when used with detailed TAS. Differences in MAR method result in large differences in local doses when variations in CT number cause differences in tissue assignment. Between different MAR models (same TAS), PTV {{D}90} and skin {{D}1~\\text{c{{\\text{m}}3}}} each vary by up to 6%. Basic TAS (mixed adipose/gland tissue) generally yield higher dose metrics than detailed segmented schemes: PTV {{D}90} and skin {{D}1~\\text{c{{\\text{m}}3}}} are higher by up to 13% and 9% respectively. Employing alternate adipose, gland and skin elemental compositions can cause variations in PTV {{D}90} of up to 11% and skin {{D}1~\\text{c{{\\text{m}}3}}} of up to 30%. Overall, AAPM TG-43 overestimates dose to the PTV ({{D}90} on average 10% and up to 27%) and underestimates dose to the skin ({{D}1~\\text{c{{\\text{m}}3}}} on average 29% and up to 48%) compared to the various MC models derived using the post-MAR CT images studied
Modulation index for VMAT considering both mechanical and dose calculation uncertainties
Park, Jong Min; Park, So-Yeon; Kim, Hyoungnyoun
2015-09-01
The aim of this study is to present a modulation index considering both mechanical and dose calculation uncertainties for volumetric modulated arc therapy (VMAT). As a modulation index considering only mechanical uncertainty of VMAT, MIt has been previously suggested. In this study, we developed a weighting factor which represents dose calculation uncertainty based on the aperture shapes of fluence maps at every control point of VMAT plans. In order to calculate the weighting factor, the thinning algorithm of image processing techniques was applied to measure field aperture irregularity. By combining this weighting factor with the previously suggested modulation index, MIt, comprehensive modulation index (MIc) was designed. To evaluate the performance of MIc, gamma passing rates, differences in mechanical parameters between plans and log files and differences in dose-volume parameters between plans and the plans reconstructed from log files were acquired with a total of 52 VMAT plans. Spearman’s correlation coefficients (rs) between the values of MIc and measures of VMAT delivery accuracy were calculated. The rs values of MIc (f = 0.5) to global gamma passing rates with 2%/2 mm, 1%/2 mm and 2%/1 mm were -0.728,-0.847 and -0.617, respectively (p VMAT delivery accuracy showing strong correlations to various measures of VMAT delivery accuracy.
Results of 1 year of clinical experience with independent dose calculation software for VMAT fields
Juan Fernando Mata Colodro
2014-01-01
Full Text Available It is widely accepted that a redundant independent dose calculation (RIDC must be included in any treatment planning verification procedure. Specifically, volumetric modulated arc therapy (VMAT technique implies a comprehensive quality assurance (QA program in which RIDC should be included. In this paper, the results obtained in 1 year of clinical experience are presented. Eclipse from Varian is the treatment planning system (TPS, here in use. RIDC were performed with the commercial software; Diamond ® (PTW which is capable of calculating VMAT fields. Once the plan is clinically accepted, it is exported via Digital Imaging and Communications in Medicine (DICOM to RIDC, together with the body contour, and then a point dose calculation is performed, usually at the isocenter. A total of 459 plans were evaluated. The total average deviation was -0.3 ± 1.8% (one standard deviation (1SD. For higher clearance the plans were grouped by location in: Prostate, pelvis, abdomen, chest, head and neck, brain, stereotactic radiosurgery, lung stereotactic body radiation therapy, and miscellaneous. The highest absolute deviation was -0.8 ± 1.5% corresponding to the prostate. A linear fit between doses calculated by RIDC and by TPS produced a correlation coefficient of 0.9991 and a slope of 1.0023. These results are very close to those obtained in the validation process. This agreement led us to consider this RIDC software as a valuable tool for QA in VMAT plans.
Results of 1 year of clinical experience with independent dose calculation software for VMAT fields.
Colodro, Juan Fernando Mata; Berna, Alfredo Serna; Puchades, Vicente Puchades; Amores, David Ramos; Baños, Miguel Alcaraz
2014-10-01
It is widely accepted that a redundant independent dose calculation (RIDC) must be included in any treatment planning verification procedure. Specifically, volumetric modulated arc therapy (VMAT) technique implies a comprehensive quality assurance (QA) program in which RIDC should be included. In this paper, the results obtained in 1 year of clinical experience are presented. Eclipse from Varian is the treatment planning system (TPS), here in use. RIDC were performed with the commercial software; Diamond(®) (PTW) which is capable of calculating VMAT fields. Once the plan is clinically accepted, it is exported via Digital Imaging and Communications in Medicine (DICOM) to RIDC, together with the body contour, and then a point dose calculation is performed, usually at the isocenter. A total of 459 plans were evaluated. The total average deviation was -0.3 ± 1.8% (one standard deviation (1SD)). For higher clearance the plans were grouped by location in: Prostate, pelvis, abdomen, chest, head and neck, brain, stereotactic radiosurgery, lung stereotactic body radiation therapy, and miscellaneous. The highest absolute deviation was -0.8 ± 1.5% corresponding to the prostate. A linear fit between doses calculated by RIDC and by TPS produced a correlation coefficient of 0.9991 and a slope of 1.0023. These results are very close to those obtained in the validation process. This agreement led us to consider this RIDC software as a valuable tool for QA in VMAT plans.
Clouvas, A; Antonopoulos-Domis, M; Silva, J
2000-01-01
The dose rate conversion factors D/sub CF/ (absorbed dose rate in air per unit activity per unit of soil mass, nGy h/sup -1/ per Bq kg/sup -1/) are calculated 1 m above ground for photon emitters of natural radionuclides uniformly distributed in the soil. Three Monte Carlo codes are used: 1) The MCNP code of Los Alamos; 2) The GEANT code of CERN; and 3) a Monte Carlo code developed in the Nuclear Technology Laboratory of the Aristotle University of Thessaloniki. The accuracy of the Monte Carlo results is tested by the comparison of the unscattered flux obtained by the three Monte Carlo codes with an independent straightforward calculation. All codes and particularly the MCNP calculate accurately the absorbed dose rate in air due to the unscattered radiation. For the total radiation (unscattered plus scattered) the D/sub CF/ values calculated from the three codes are in very good agreement between them. The comparison between these results and the results deduced previously by other authors indicates a good ag...
Faught, Austin M; Davidson, Scott E; Fontenot, Jonas; Kry, Stephen F; Etzel, Carol; Ibbott, Geoffrey S; Followill, David S
2017-09-01
The Imaging and Radiation Oncology Core Houston (IROC-H) (formerly the Radiological Physics Center) has reported varying levels of agreement in their anthropomorphic phantom audits. There is reason to believe one source of error in this observed disagreement is the accuracy of the dose calculation algorithms and heterogeneity corrections used. To audit this component of the radiotherapy treatment process, an independent dose calculation tool is needed. Monte Carlo multiple source models for Elekta 6 MV and 10 MV therapeutic x-ray beams were commissioned based on measurement of central axis depth dose data for a 10 × 10 cm(2) field size and dose profiles for a 40 × 40 cm(2) field size. The models were validated against open field measurements consisting of depth dose data and dose profiles for field sizes ranging from 3 × 3 cm(2) to 30 × 30 cm(2) . The models were then benchmarked against measurements in IROC-H's anthropomorphic head and neck and lung phantoms. Validation results showed 97.9% and 96.8% of depth dose data passed a ±2% Van Dyk criterion for 6 MV and 10 MV models respectively. Dose profile comparisons showed an average agreement using a ±2%/2 mm criterion of 98.0% and 99.0% for 6 MV and 10 MV models respectively. Phantom plan comparisons were evaluated using ±3%/2 mm gamma criterion, and averaged passing rates between Monte Carlo and measurements were 87.4% and 89.9% for 6 MV and 10 MV models respectively. Accurate multiple source models for Elekta 6 MV and 10 MV x-ray beams have been developed for inclusion in an independent dose calculation tool for use in clinical trial audits. © 2017 American Association of Physicists in Medicine.
Chetty, Indrin J; Moran, Jean M; McShan, Daniel L; Fraass, Benedick A; Wilderman, Scott J; Bielajew, Alex F
2002-06-01
A comprehensive set of measurements and calculations has been conducted to investigate the accuracy of the Dose Planning Method (DPM) Monte Carlo code for dose calculations from 10 and 50 MeV scanned electron beams produced from a racetrack microtron. Central axis depth dose measurements and a series of profile scans at various depths were acquired in a water phantom using a Scanditronix type RK ion chamber. Source spatial distributions for the Monte Carlo calculations were reconstructed from in-air ion chamber measurements carried out across the two-dimensional beam profile at 100 cm downstream from the source. The in-air spatial distributions were found to have full width at half maximum of 4.7 and 1.3 cm, at 100 cm from the source, for the 10 and 50 MeV beams, respectively. Energy spectra for the 10 and 50 MeV beams were determined by simulating the components of the microtron treatment head using the code MCNP4B. DPM calculations are on average within +/- 2% agreement with measurement for all depth dose and profile comparisons conducted in this study. The accuracy of the DPM code illustrated in this work suggests that DPM may be used as a valuable tool for electron beam dose calculations.
Mizutani, Shohei; Takada, Yoshihisa; Kohno, Ryosuke; Hotta, Kenji; Tansho, Ryohei; Akimoto, Tetsuo
2016-03-01
Full Monte Carlo (FMC) calculation of dose distribution has been recognized to have superior accuracy, compared with the pencil beam algorithm (PBA). However, since the FMC methods require long calculation time, it is difficult to apply them to routine treatment planning at present. In order to improve the situation, a simplified Monte Carlo (SMC) method has been introduced to the dose kernel calculation applicable to dose optimization procedure for the proton pencil beam scanning. We have evaluated accuracy of the SMC calculation by comparing a result of the dose kernel calculation using the SMC method with that using the FMC method in an inhomogeneous phantom. The dose distribution obtained by the SMC method was in good agreement with that obtained by the FMC method. To assess the usefulness of SMC calculation in clinical situations, we have compared results of the dose calculation using the SMC with those using the PBA method for three clinical cases of tumor treatment. The dose distributions calculated with the PBA dose kernels appear to be homogeneous in the planning target volumes (PTVs). In practice, the dose distributions calculated with the SMC dose kernels with the spot weights optimized with the PBA method show largely inhomogeneous dose distributions in the PTVs, while those with the spot weights optimized with the SMC method have moderately homogeneous distributions in the PTVs. Calculation using the SMC method is faster than that using the GEANT4 by three orders of magnitude. In addition, the graphic processing unit (GPU) boosts the calculation speed by 13 times for the treatment planning using the SMC method. Thence, the SMC method will be applicable to routine clinical treatment planning for reproduction of the complex dose distribution more accurately than the PBA method in a reasonably short time by use of the GPU-based calculation engine. PACS number(s): 87.55.Gh.
Bagher Farhood
2016-09-01
Full Text Available Introduction Wedge modifiers are commonly applied in external beam radiotherapy to change the dose distribution corresponding to the body contour and to obtain a uniform dose distribution within the target volume. Since the radiation dose delivered to the target must be within ±5% of the prescribed dose, accurate dose calculation by a treatment planning system (TPS is important. The objective of the present study was to quantify the dose calculation accuracy of TiGRT TPS for physical wedged fields in radiotherapy. Materials and Methods A Semiﬂex™ ionization chamber was used for dose measurements in a water phantom; TiGRT TPS was also applied for dose calculations. The central axis (i.e., high dose-small dose gradient, build-up (i.e., high dose-large dose gradient, off-axis (i.e., high dose-small dose gradient, and out-of-field (i.e., low dose-small dose gradient regions were evaluated in this study. Finally, the confidence limit values were obtained to quantify the dose calculation accuracy of TPS in these regions. Results The confidence limit values for the central axis, build-up, off-axis, and out-of-field regions were 1.01, 8.62, 1.79, and 55.24, respectively. Furthermore, the results showed that TiGRT TPS underestimated the dose of build-up and out-of-field regions for most points. Conclusion According to the results of the present study, it can be concluded that the dose calculation accuracy of TiGRT TPS for physical wedged ﬁelds in the central axis, build-up, and off-axis regions is adequate, while it is insufficient for out-of-field regions.
Takashima, Y.; Oki, S.; Sugiyama, H.; Kobayakawa, H.
2005-10-01
The spatial distribution of the effective dose of photons, electrons and positrons caused by beam loss around a small electron storage ring in a synchrotron radiation source is calculated. We propose a simple formula applicable to calculate the effective dose for storage rings for beam energies ranging from 200 MeV to 5 GeV. The formula is derived from Monte Carlo calculations of radiation flux using the simulation code EGS4. We apply the formula to estimate the effective dose distribution in a small synchrotron radiation facility planned by the Nagoya University.
Electron paramagnetic resonance measurements of absorbed dose in teeth from citizens of Ozyorsk
Wieser, A.; Semiochkina, N. [Helmholtz Zentrum Muenchen - German Research Center for Environmental Health, Institute of Radiation Protection, Neuherberg (Germany); Vasilenko, E.; Aladova, E.; Smetanin, M. [Southern Urals Biophysics Institute, Ozyorsk (Russian Federation); Fattibene, P. [Istituto Superiore di Sanita, Rome (Italy)
2014-05-15
In 1945, within the frame of the Uranium Project for the production of nuclear weapons, the Mayak nuclear facilities were constructed at the Lake Irtyash in the Southern Urals, Russia. The nuclear workers of the Mayak Production Association (MPA), who lived in the city of Ozyorsk, are the focus of epidemiological studies for the assessment of health risks due to protracted exposure to ionising radiation. Electron paramagnetic resonance measurements of absorbed dose in tooth enamel have already been used in the past, in an effort to validate occupational external doses that were evaluated in the Mayak Worker Dosimetry System. In the present study, 229 teeth of Ozyorsk citizens not employed at MPA were investigated for the assessment of external background exposure in Ozyorsk. The annually absorbed dose in tooth enamel from natural background radiation was estimated to be (0.7 ± 0.3) mGy. For citizens living in Ozyorsk during the time of routine noble gas releases of the MPA, which peaked in 1953, the average excess absorbed dose in enamel above natural background was (36 ± 29) mGy, which is consistent with the gamma dose obtained by model calculations. In addition, there were indications of possible accidental gaseous MPA releases that affected the population of Ozyorsk, during the early and late MPA operation periods, before 1951 and after 1960. (orig.)
The Evaluation of Skin Toxicity during Brain Tumor Irradiation Dose Calculation
Oxana A. Pashkovskaya
2013-12-01
Full Text Available Background: Radiotherapy is the keystone in brain tumor treatment, including posterior fossa tumors, and can achieve better patient health-related quality of life. Radiation exposure can be associated with the risk of skin radiation injuries. Accurate tumor and critical structure delineations and precise dose planning may improve the outcomes and decrease radiation complications. The objective of this study was to compare the influence of the headrest and treatment couch during dose planning, on the dose distributions and skin injury post irradiation. Material and Methods: Treatment planning calculations were performed for 14 brain tumor patients using the volumetric modulated arc therapy (VMAT to study the dose distribution and dose-volume histograms (DVH. We compared the following three cases of general patient contours: patient body contour alone, body contour including the headrest, and body contour with headrest, couch and immobilization mask. The same configuration beams were used in all these cases; general patient contours alone were altered. Results: For dose estimations, the skin was delineated as a 2 mm layer beneath the patient’s body contour. The comparisons showed that the average dose on the skin, among all the patients included in this study, in the case of body contour alone is 3.3 Gy, whereas in the case of body contour with headrest, it is 6.3 Gy and in the case of body contour with headrest, couch and immobilization mask it is 9.4 Gy. Conclusion: For brain tumors, located in the posterior fossa and near the patient’s skin, the skin needed to be included as a critical structure. The skin dose should be considered when evaluating treatment plans, taking into account the bolus effect of the headrest and couch.
Dose estimation and shielding calculation for X-ray hazard at high intensity laser facilities
Qiu, Rui; Zhang, Hui; Yang, Bo; James, C. Liu; Sayed, H. Rokni; Michael, B. Woods; Li, Jun-Li
2014-12-01
An ionizing radiation hazard produced from the interaction between high intensity lasers and solid targets has been observed. Laser-plasma interactions create “hot” electrons, which generate bremsstrahlung X-rays when they interact with ions in the target. However, up to now only limited studies have been conducted on this laser-induced radiological protection issue. In this paper, the physical process and characteristics of the interaction between high intensity lasers and solid targets are analyzed. The parameters of the radiation sources are discussed, including the energy conversion efficiency from laser to hot electrons, hot electron energy spectrum and electron temperature, and the bremsstrahlung X-ray energy spectrum produced by hot electrons. Based on this information, the X-ray dose generated with high-Z targets for laser intensities between 1014 and 1020 W/cm2 is estimated. The shielding effects of common shielding items such as the glass view port, aluminum chamber wall and concrete wall are also studied using the FLUKA Monte Carlo code. This study provides a reference for the dose estimation and the shielding design of high intensity laser facilities.
Endo, Akira; Yamaguchi, Yasuhiro [Japan Atomic Energy Research Inst., Tokai, Ibaraki (Japan). Tokai Research Establishment
2001-03-01
New nuclear decay data used for dose calculation have been compiled for 817 radionuclides that are listed in ICRP Publication 38 (Publ. 38) and for 6 additional isomers. The decay data were prepared using decay data sets from the Evaluated Nuclear Structure Data File (ENSDF), the latest version in August 1997. Basic nuclear properties in the decay data sets that are particularly important for calculating energies and intensities of emissions were examined and updated by referring to NUBASE, the database for nuclear and decay properties of nuclides. The reviewed and updated data were half-life, decay mode and its branching ratio, spin and parity of the ground and isomeric states, excitation energy of isomers, and Q value. In addition, possible revisions of partial and incomplete decay data sets were done for their format and syntax errors, level schemes, normalization records, and so on. After that, the decay data sets were processed by EDISTR in order to compute the energies and intensities of {alpha} particles, {beta} particles, {gamma} rays, internal conversion electrons, X rays, and Auger electrons emitted in nuclear transformation. For spontaneously fissioning nuclides, the average energies and intensities of neutrons, fission fragments, prompt {gamma} rays, delayed {gamma} rays, and {beta} particles were also calculated. The compiled data were prepared in two different types of format: Publ. 38 and NUCDECAY formats. Comparison of the compiled decay data with those in Publ. 38 was also presented. The decay data will be widely used for internal and external dose calculations in radiation protection and will be beneficial to a future revision of ICRP Publ. 38. (author)
Fogliata, Antonella; Nicolini, Giorgia; Clivio, Alessandro; Vanetti, Eugenio; Cozzi, Luca [Oncology Institute of Southern Switzerland, Medical Physics Unit, Bellinzona (Switzerland); Mancosu, Pietro, E-mail: afc@iosi.ch [Istituto Clinico Humanitas, Radio-Oncology Department, Milan-Rozzano (Italy)
2011-03-21
A new algorithm, Acuros (registered) XB Advanced Dose Calculation, has been introduced by Varian Medical Systems in the Eclipse planning system for photon dose calculation in external radiotherapy. Acuros XB is based on the solution of the linear Boltzmann transport equation (LBTE). The LBTE describes the macroscopic behaviour of radiation particles as they travel through and interact with matter. The implementation of Acuros XB in Eclipse has not been assessed; therefore, it is necessary to perform these pre-clinical validation tests to determine its accuracy. This paper summarizes the results of comparisons of Acuros XB calculations against measurements and calculations performed with a previously validated dose calculation algorithm, the Anisotropic Analytical Algorithm (AAA). The tasks addressed in this paper are limited to the fundamental characterization of Acuros XB in water for simple geometries. Validation was carried out for four different beams: 6 and 15 MV beams from a Varian Clinac 2100 iX, and 6 and 10 MV 'flattening filter free' (FFF) beams from a TrueBeam linear accelerator. The TrueBeam FFF are new beams recently introduced in clinical practice on general purpose linear accelerators and have not been previously reported on. Results indicate that Acuros XB accurately reproduces measured and calculated (with AAA) data and only small deviations were observed for all the investigated quantities. In general, the overall degree of accuracy for Acuros XB in simple geometries can be stated to be within 1% for open beams and within 2% for mechanical wedges. The basic validation of the Acuros XB algorithm was therefore considered satisfactory for both conventional photon beams as well as for FFF beams of new generation linacs such as the Varian TrueBeam.
A comparison of measured and calculated organ doses from CT examinations
Calzado, A.; Ruiz Sanz, S.; Melchor, M.; Vano, E. [Universidad Complutense, Madrid (Spain). Facultad de Medicina
1995-12-31
Organ doses from a set of frequent CT examinations have been estimated from measurements in a physical anthropomorphic phantom (Remab system) by using thermoluminescence dosemeters. For the same examination techniques, organ dose coefficients (taken from the literature) obtained by Monte Carlo techniques and using mathematical phantoms. The results arrived at by the two methods are compared, trying to explain the most significant differences and their influence on the estimated values of effective dose. The experimental and calculated outcomes from such simulations are also compared to the mean dosimetric results on patients from a 1991 regional survey of CT practice in the area of Madrid. Some comments about the complementary use of information coming from both methods are made. (Author).
Estimation of Electron Dose Delivered by a 0.4 MeV Accelerator from Bremsstrahlung Dose Measurements
Karadjov, A. G.; Hansen, Jørgen-Walther
1980-01-01
Determination of a 0.4 MeV electron dose from a bremsstrahlung dose measurement using a converter-detector system is considered. The detector used is a Frickle dosimeter, and the converters are aluminum, copper and lead foils. Optimal converter thickness is ascertained experimentally for each mat...
Tajaldeen, A [RMIT university, Docklands, Vic (Australia); Ramachandran, P [Peter MacCallum Cancer Centre, Bendigo (Australia); Geso, M [RMIT University, Bundoora, Melbourne (Australia)
2015-06-15
Purpose: The purpose of this study was to investigate and quantify the variation in dose distributions in small field lung cancer radiotherapy using seven different dose calculation algorithms. Methods: The study was performed in 21 lung cancer patients who underwent Stereotactic Ablative Body Radiotherapy (SABR). Two different methods (i) Same dose coverage to the target volume (named as same dose method) (ii) Same monitor units in all algorithms (named as same monitor units) were used for studying the performance of seven different dose calculation algorithms in XiO and Eclipse treatment planning systems. The seven dose calculation algorithms include Superposition, Fast superposition, Fast Fourier Transform ( FFT) Convolution, Clarkson, Anisotropic Analytic Algorithm (AAA), Acurous XB and pencil beam (PB) algorithms. Prior to this, a phantom study was performed to assess the accuracy of these algorithms. Superposition algorithm was used as a reference algorithm in this study. The treatment plans were compared using different dosimetric parameters including conformity, heterogeneity and dose fall off index. In addition to this, the dose to critical structures like lungs, heart, oesophagus and spinal cord were also studied. Statistical analysis was performed using Prism software. Results: The mean±stdev with conformity index for Superposition, Fast superposition, Clarkson and FFT convolution algorithms were 1.29±0.13, 1.31±0.16, 2.2±0.7 and 2.17±0.59 respectively whereas for AAA, pencil beam and Acurous XB were 1.4±0.27, 1.66±0.27 and 1.35±0.24 respectively. Conclusion: Our study showed significant variations among the seven different algorithms. Superposition and AcurosXB algorithms showed similar values for most of the dosimetric parameters. Clarkson, FFT convolution and pencil beam algorithms showed large differences as compared to superposition algorithms. Based on our study, we recommend Superposition and AcurosXB algorithms as the first choice of
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources
Townson, Reid W.; Jia, Xun; Tian, Zhen; Jiang Graves, Yan; Zavgorodni, Sergei; Jiang, Steve B.
2013-06-01
A novel phase-space source implementation has been designed for graphics processing unit (GPU)-based Monte Carlo dose calculation engines. Short of full simulation of the linac head, using a phase-space source is the most accurate method to model a clinical radiation beam in dose calculations. However, in GPU-based Monte Carlo dose calculations where the computation efficiency is very high, the time required to read and process a large phase-space file becomes comparable to the particle transport time. Moreover, due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel source implementation utilizing pre-processed patient-independent phase-spaces that are sorted by particle type, energy and position. Position bins located outside a rectangular region of interest enclosing the treatment field are ignored, substantially decreasing simulation time with little effect on the final dose distribution. The three methods were validated in absolute dose against BEAMnrc/DOSXYZnrc and compared using gamma-index tests (2%/2 mm above the 10% isodose). It was found that the PSL method has the optimal balance between accuracy and efficiency and thus is used as the default method in gDPM v3.0. Using the PSL method, open fields of 4 × 4, 10 × 10 and 30 × 30 cm
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources.
Townson, Reid W; Jia, Xun; Tian, Zhen; Graves, Yan Jiang; Zavgorodni, Sergei; Jiang, Steve B
2013-06-21
A novel phase-space source implementation has been designed for graphics processing unit (GPU)-based Monte Carlo dose calculation engines. Short of full simulation of the linac head, using a phase-space source is the most accurate method to model a clinical radiation beam in dose calculations. However, in GPU-based Monte Carlo dose calculations where the computation efficiency is very high, the time required to read and process a large phase-space file becomes comparable to the particle transport time. Moreover, due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel source implementation utilizing pre-processed patient-independent phase-spaces that are sorted by particle type, energy and position. Position bins located outside a rectangular region of interest enclosing the treatment field are ignored, substantially decreasing simulation time with little effect on the final dose distribution. The three methods were validated in absolute dose against BEAMnrc/DOSXYZnrc and compared using gamma-index tests (2%/2 mm above the 10% isodose). It was found that the PSL method has the optimal balance between accuracy and efficiency and thus is used as the default method in gDPM v3.0. Using the PSL method, open fields of 4 × 4, 10 × 10 and 30 × 30 cm
Mikell, Justin; Cheenu Kappadath, S; Wareing, Todd; Erwin, William D; Titt, Uwe; Mourtada, Firas
2016-06-21
To evaluate the 3D Grid-based Boltzmann Solver (GBBS) code ATTILA (®) for coupled electron and photon transport in the nuclear medicine energy regime for electron (beta, Auger and internal conversion electrons) and photon (gamma, x-ray) sources. Codes rewritten based on ATTILA are used clinically for both high-energy photon teletherapy and (192)Ir sealed source brachytherapy; little information exists for using the GBBS to calculate voxel-level absorbed doses in nuclear medicine. We compared DOSXYZnrc Monte Carlo (MC) with published voxel-S-values to establish MC as truth. GBBS was investigated for mono-energetic 1.0, 0.1, and 0.01 MeV electron and photon sources as well as (131)I and (90)Y radionuclides. We investigated convergence of GBBS by analyzing different meshes ([Formula: see text]), energy group structures ([Formula: see text]) for each radionuclide component, angular quadrature orders ([Formula: see text], and scattering order expansions ([Formula: see text]-[Formula: see text]); higher indices imply finer discretization. We compared GBBS to MC in (1) voxel-S-value geometry for soft tissue, lung, and bone, and (2) a source at the interface between combinations of lung, soft tissue, and bone. Excluding Auger and conversion electrons, MC agreed within ≈5% of published source voxel absorbed doses. For the finest discretization, most GBBS absorbed doses in the source voxel changed by less than 1% compared to the next finest discretization along each phase space variable indicating sufficient convergence. For the finest discretization, agreement with MC in the source voxel ranged from -3% to -20% with larger differences at lower energies (-3% for 1 MeV electron in lung to -20% for 0.01 MeV photon in bone); similar agreement was found for the interface geometries. Differences between GBBS and MC in the source voxel for (90)Y and (131)I were -6%. The GBBS ATTILA was benchmarked against MC in the nuclear medicine regime. GBBS can be a
Mikell, Justin; Cheenu Kappadath, S.; Wareing, Todd; Erwin, William D.; Titt, Uwe; Mourtada, Firas
2016-06-01
To evaluate the 3D Grid-based Boltzmann Solver (GBBS) code ATTILA ® for coupled electron and photon transport in the nuclear medicine energy regime for electron (beta, Auger and internal conversion electrons) and photon (gamma, x-ray) sources. Codes rewritten based on ATTILA are used clinically for both high-energy photon teletherapy and 192Ir sealed source brachytherapy; little information exists for using the GBBS to calculate voxel-level absorbed doses in nuclear medicine. We compared DOSXYZnrc Monte Carlo (MC) with published voxel-S-values to establish MC as truth. GBBS was investigated for mono-energetic 1.0, 0.1, and 0.01 MeV electron and photon sources as well as 131I and 90Y radionuclides. We investigated convergence of GBBS by analyzing different meshes ({{M}0},{{M}1},{{M}2} ), energy group structures ({{E}0},{{E}1},{{E}2} ) for each radionuclide component, angular quadrature orders (≤ft. {{S}4},{{S}8},{{S}16}\\right) , and scattering order expansions ({{P}0} -{{P}6} ); higher indices imply finer discretization. We compared GBBS to MC in (1) voxel-S-value geometry for soft tissue, lung, and bone, and (2) a source at the interface between combinations of lung, soft tissue, and bone. Excluding Auger and conversion electrons, MC agreed within ≈5% of published source voxel absorbed doses. For the finest discretization, most GBBS absorbed doses in the source voxel changed by less than 1% compared to the next finest discretization along each phase space variable indicating sufficient convergence. For the finest discretization, agreement with MC in the source voxel ranged from -3% to -20% with larger differences at lower energies (-3% for 1 MeV electron in lung to -20% for 0.01 MeV photon in bone); similar agreement was found for the interface geometries. Differences between GBBS and MC in the source voxel for 90Y and 131I were -6%. The GBBS ATTILA was benchmarked against MC in the nuclear medicine regime. GBBS can be a viable
Methodology for calculation of radiation doses in the environs from nuclear fuel cycle facilities
Soldat, J.K.
1976-08-01
Comparison of the impacts of various nuclear fuel cycle alternatives includes the evaluation of the radiological impacts. To evaluate the radiological impacts of fuel cycle alternatives, exposure to man must first be identified. The pathways of consequence by which man can be exposed to radiation from a nuclear facility are listed and are grouped into those associated with gaseous effluents, those associated with liquid effluents, and those involving exposure to direct radiation from the facility or from transportation of radioactive materials to or from the facility. Calculations for each pathway were made for those selected organs which could potentially receive the highest radiation dose. Some of the programs developed for calculating radiation doses from radionuclides in the environment are described. (CH)
Technical Note: Radiotherapy dose calculations using GEANT4 and the Amazon Elastic Compute Cloud
Poole, Christopher M; Trapp, Jamie V; Langton, Christian M
2011-01-01
Cloud computing allows for vast computational resources to be leveraged quickly and easily in bursts as and when required. Using the Amazon Elastic Compute Cloud and the Amazon Simple Storage Solution, we describe a technique that allows for Monte Carlo radiotherapy dose calculations to be performed using GEANT4 and executed in the cloud. Simulation cost and completion time was evaluated as a function of instance count using compute instances acquired via biding on the Elastic Compute Cloud spot market. Bidding for instances on the instance spot market was found to be 35-60% of the cost of on-demand instances of the same type. Using the technique, we demonstrate the potential usefulness of cloud computing as a solution for rapid Monte Carlo simulation for radiotherapy dose calculation.
Calculation of conversion factors for effective dose for various interventional radiology procedures
Compagnone, Gaetano; Giampalma, Emanuela; Domenichelli, Sara; Renzulli, Matteo; Golfieri, Rita [Medical Physics Department, S. Orsola-Malpighi University Hospital, Via Massarenti 9, 40138 Bologna (Italy); Radiology Department, S. Orsola-Malpighi University Hospital, Via Massarenti 9, 40138 Bologna (Italy); Medical Physics Department, S. Orsola-Malpighi University Hospital, Via Massarenti 9, 40138 Bologna (Italy); Radiology Department, S. Orsola-Malpighi University Hospital, Via Massarenti 9, 40138 Bologna (Italy)
2012-05-15
Purpose: To provide dose-area-product (DAP) to effective dose (E) conversion factors for complete interventional procedures, based on in-the-field clinical measurements of DAP values and using tabulated E/DAP conversion factors for single projections available from the literature. Methods: Nine types of interventional procedures were performed on 84 patients with two angiographic systems. Different calibration curves (with and without patient table attenuation) were calculated for each DAP meter. Clinical and dosimetric parameters were recorded in-the-field for each projection and for all patients, and a conversion factor linking DAP and effective doses was derived for each complete procedure making use of published, Monte Carlo calculated conversion factors for single static projections. Results: Fluoroscopy time and DAP values for the lowest-dose procedure (biliary drainage) were approximately 3-fold and 13-fold lower, respectively, than those for the highest-dose examination (transjugular intrahepatic portosystemic shunt, TIPS). Median E/DAP conversion factors from 0.12 (abdominal percutaneous transluminal angioplasty) to 0.25 (Nephrostomy) mSvGy{sup -1} cm{sup -2} were obtained and good correlations between E and DAP were found for all procedures, with R{sup 2} coefficients ranging from 0.80 (abdominal angiography) to 0.99 (biliary stent insertion, Nephrostomy and TIPS). The DAP values obtained in this study showed general consistency with the values provided in the literature and median E values ranged from 4.0 mSv (biliary drainage) to 49.6 mSv (TIPS). Conclusions: Values of E/DAP conversion factors were derived for each procedure from a comprehensive analysis of projection and dosimetric data: they could provide a good evaluation for the stochastic effects. These results can be obtained by means of a close cooperation between different interventional professionals involved in patient care and dose optimization.
Gomes B, W. O., E-mail: wilsonottobatista@gmail.com [Instituto Federal da Bahia, Rua Emidio dos Santos s/n, Barbalho 40301-015, Salvador de Bahia (Brazil)
2016-10-15
This study aimed to develop a geometry of irradiation applicable to the software PCXMC and the consequent calculation of effective dose in applications of the Computed Tomography Cone Beam (CBCT). We evaluated two different CBCT equipment s for dental applications: Care stream Cs 9000 3-dimensional tomograph; i-CAT and GENDEX GXCB-500. Initially characterize each protocol measuring the surface kerma input and the product kerma air-area, P{sub KA}, with solid state detectors RADCAL and PTW transmission chamber. Then we introduce the technical parameters of each preset protocols and geometric conditions in the PCXMC software to obtain the values of effective dose. The calculated effective dose is within the range of 9.0 to 15.7 μSv for 3-dimensional computer 9000 Cs; within the range 44.5 to 89 μSv for GXCB-500 equipment and in the range of 62-111 μSv for equipment Classical i-CAT. These values were compared with results obtained dosimetry using TLD implanted in anthropomorphic phantom and are considered consistent. Os effective dose results are very sensitive to the geometry of radiation (beam position in mathematical phantom). This factor translates to a factor of fragility software usage. But it is very useful to get quick answers to regarding process optimization tool conclusions protocols. We conclude that use software PCXMC Monte Carlo simulation is useful assessment protocols for CBCT tests in dental applications. (Author)
Methodology for calculation of doses to man and implementation in Pandora
Avila, Rodolfo [Facilia AB, Bromma (Sweden); Bergstroem, Ulla [Swepro Project Management AB, Solna (Sweden)
2006-07-15
This report describes methods and data for calculation of doses to man to be used in safety assessments of repositories for nuclear fuel. The methods are based on the latest recommendations from the ICRP; the EU and the national radiation protection authorities. Equations are given for calculation of doses from ingestion of contaminated water and food, inhalation of contaminated air and external exposure from radionuclides in the ground. With the exception of the exposure from food ingestion, the equations are the same used in previous safety assessments. A general equation is suggested for estimation of the exposure from food ingestion, in which the annual demand of carbon is used instead of the annual ingestion of different food-stuffs, which was earlier applied. The report contains tables with recommended values for physiological characteristics such as water intake, food intake and inhalation rates, based on information summarised in an Appendix. Furthermore, tables are given with recommended age dependent dose conversion factors for ingestion and inhalation for a number of nuclides of interest for safety assessments. The most recently published dose conversion factors for external exposure from contaminated ground are also given. An overview of the implementation of the methodology in Pandora, which is the tool that SKB and Posiva currently use for biosphere modelling, is also provided. The work presented in the report is a result from a joint project commissioned by SKB and Posiva.
Fast CPU-based Monte Carlo simulation for radiotherapy dose calculation.
Ziegenhein, Peter; Pirner, Sven; Ph Kamerling, Cornelis; Oelfke, Uwe
2015-08-07
Monte-Carlo (MC) simulations are considered to be the most accurate method for calculating dose distributions in radiotherapy. Its clinical application, however, still is limited by the long runtimes conventional implementations of MC algorithms require to deliver sufficiently accurate results on high resolution imaging data. In order to overcome this obstacle we developed the software-package PhiMC, which is capable of computing precise dose distributions in a sub-minute time-frame by leveraging the potential of modern many- and multi-core CPU-based computers. PhiMC is based on the well verified dose planning method (DPM). We could demonstrate that PhiMC delivers dose distributions which are in excellent agreement to DPM. The multi-core implementation of PhiMC scales well between different computer architectures and achieves a speed-up of up to 37[Formula: see text] compared to the original DPM code executed on a modern system. Furthermore, we could show that our CPU-based implementation on a modern workstation is between 1.25[Formula: see text] and 1.95[Formula: see text] faster than a well-known GPU implementation of the same simulation method on a NVIDIA Tesla C2050. Since CPUs work on several hundreds of GB RAM the typical GPU memory limitation does not apply for our implementation and high resolution clinical plans can be calculated.
Kuo, Nathanael, E-mail: nkuo8@jhmi.edu; Prince, Jerry L. [Department of Electrical and Computer Engineering, Johns Hopkins University, Baltimore, Maryland 21218 (United States); Dehghan, Ehsan [Philips Research North America, Briarcliff Manor, New York 10510 (United States); Deguet, Anton [Department of Computer Science, Johns Hopkins University, Baltimore, Maryland 21218 (United States); Mian, Omar Y.; Le, Yi; Song, Danny Y. [Department of Radiation Oncology and Molecular Radiation Sciences, Johns Hopkins University, Baltimore, Maryland 21231 (United States); Burdette, E. Clif [Acoustic MedSystems Inc., Savoy, Illinois 61974 (United States); Fichtinger, Gabor [School of Computing, Queen' s University, Kingston, Ontario K7L3N6 (Canada); Lee, Junghoon [Department of Electrical and Computer Engineering, Johns Hopkins University, Baltimore, Maryland 21218 and Department of Radiation Oncology and Molecular Radiation Sciences, Johns Hopkins University, Baltimore, Maryland 21231 (United States)
2014-09-15
Purpose: Brachytherapy is a standard option of care for prostate cancer patients but may be improved by dynamic dose calculation based on localized seed positions. The American Brachytherapy Society states that the major current limitation of intraoperative treatment planning is the inability to localize the seeds in relation to the prostate. An image-guidance system was therefore developed to localize seeds for dynamic dose calculation. Methods: The proposed system is based on transrectal ultrasound (TRUS) and mobile C-arm fluoroscopy, while using a simple fiducial with seed-like markers to compute pose from the nonencoded C-arm. Three or more fluoroscopic images and an ultrasound volume are acquired and processed by a pipeline of algorithms: (1) seed segmentation, (2) fiducial detection with pose estimation, (3) seed matching with reconstruction, and (4) fluoroscopy-to-TRUS registration. Results: The system was evaluated on ten phantom cases, resulting in an overall mean error of 1.3 mm. The system was also tested on 37 patients and each algorithm was evaluated. Seed segmentation resulted in a 1% false negative rate and 2% false positive rate. Fiducial detection with pose estimation resulted in a 98% detection rate. Seed matching with reconstruction had a mean error of 0.4 mm. Fluoroscopy-to-TRUS registration had a mean error of 1.3 mm. Moreover, a comparison of dose calculations between the authors’ intraoperative method and an independent postoperative method shows a small difference of 7% and 2% forD{sub 90} and V{sub 100}, respectively. Finally, the system demonstrated the ability to detect cold spots and required a total processing time of approximately 1 min. Conclusions: The proposed image-guidance system is the first practical approach to dynamic dose calculation, outperforming earlier solutions in terms of robustness, ease of use, and functional completeness.
Karlsson Mikael
2010-06-01
Full Text Available Abstract Background Because of superior soft tissue contrast, the use of magnetic resonance imaging (MRI as a complement to computed tomography (CT in the target definition procedure for radiotherapy is increasing. To keep the workflow simple and cost effective and to reduce patient dose, it is natural to strive for a treatment planning procedure based entirely on MRI. In the present study, we investigate the dose calculation accuracy for different treatment regions when using bulk density assignments on MRI data and compare it to treatment planning that uses CT data. Methods MR and CT data were collected retrospectively for 40 patients with prostate, lung, head and neck, or brain cancers. Comparisons were made between calculations on CT data with and without inhomogeneity corrections and on MRI or CT data with bulk density assignments. The bulk densities were assigned using manual segmentation of tissue, bone, lung, and air cavities. Results The deviations between calculations on CT data with inhomogeneity correction and on bulk density assigned MR data were small. The maximum difference in the number of monitor units required to reach the prescribed dose was 1.6%. This result also includes effects of possible geometrical distortions. Conclusions The dose calculation accuracy at the investigated treatment sites is not significantly compromised when using MRI data when adequate bulk density assignments are made. With respect to treatment planning, MRI can replace CT in all steps of the treatment workflow, reducing the radiation exposure to the patient, removing any systematic registration errors that may occur when combining MR and CT, and decreasing time and cost for the extra CT investigation.
Computational method for general multicenter electronic structure calculations.
Batcho, P F
2000-06-01
Here a three-dimensional fully numerical (i.e., chemical basis-set free) method [P. F. Batcho, Phys. Rev. A 57, 6 (1998)], is formulated and applied to the calculation of the electronic structure of general multicenter Hamiltonian systems. The numerical method is presented and applied to the solution of Schrödinger-type operators, where a given number of nuclei point singularities is present in the potential field. The numerical method combines the rapid "exponential" convergence rates of modern spectral methods with the multiresolution flexibility of finite element methods, and can be viewed as an extension of the spectral element method. The approximation of cusps in the wave function and the formulation of multicenter nuclei singularities are efficiently dealt with by the combination of a coordinate transformation and a piecewise variational spectral approximation. The complete system can be efficiently inverted by established iterative methods for elliptical partial differential equations; an application of the method is presented for atomic, diatomic, and triatomic systems, and comparisons are made to the literature when possible. In particular, local density approximations are studied within the context of Kohn-Sham density functional theory, and are presented for selected subsets of atomic and diatomic molecules as well as the ozone molecule.
GPU-based ultra fast dose calculation using a finite pencil beam model
Gu, Xuejun; Men, Chunhua; Pan, Hubert; Majumdar, Amitava; Jiang, Steve B
2009-01-01
Online adaptive radiation therapy (ART) is an attractive concept that promises the ability to deliver an optimal treatment in response to the inter-fraction variability in patient anatomy. However, it has yet to be realized due to technical limitations. Fast dose deposit coefficient calculation is a critical component of the online planning process that is required for plan optimization of intensity modulated radiation therapy (IMRT). Computer graphics processing units (GPUs) are well-suited to provide the requisite fast performance for the data-parallel nature of dose calculation. In this work, we develop a dose calculation engine based on a finite-size pencil beam (FSPB) algorithm and a GPU parallel computing framework. The developed framework can accommodate any FSPB model. We test our implementation on a case of a water phantom and a case of a prostate cancer patient with varying beamlet and voxel sizes. All testing scenarios achieved speedup ranging from 200~400 times when using a NVIDIA Tesla C1060 card...
Camplin, W C; Brownless, G P; Round, G D; Winpenny, K; Hunt, G J [Centre for Environment, Fisheries and Aquaculture Science, CEFAS Laboratory, Lowestoft (United Kingdom)
2002-12-01
A new method for estimating radiation doses to UK critical groups is proposed for discussion. Amongst others, the Food Standards Agency (FSA) and the Scottish Environment Protection Agency (SEPA) undertake surveillance of UK food and the environment as a check on the effect of discharges of radioactive wastes. Discharges in gaseous and liquid form are made under authorisation by the Environment Agency and SEPA under powers in the Radioactive Substance Act. Results of surveillance by the FSA and SEPA are published in the Radioactivity in Food and the Environment (RIFE) report series. In these reports, doses to critical groups are normally estimated separately for gaseous and liquid discharge pathways. Simple summation of these doses would tend to overestimate doses actually received. Three different methods of combining the effects of both types of discharge in an integrated assessment are considered and ranked according to their ease of application, transparency, scientific rigour and presentational issues. A single integrated assessment method is then chosen for further study. Doses are calculated for surveillance data for the calendar year 2000 and compared with those from the existing RIFE method.
Dose distribution calculation for in-vivo X-ray fluorescence scanning
Figueroa, R. G. [Universidad de la Frontera, Departamento de Ciencias Fisicas, Av. Francisco Salazar 1145, Temuco 4811230, Araucania (Chile); Lozano, E. [Instituto Nacional del Cancer, Unidad de Fisica Medica, Av. Profesor Zanartu 1010, Santiago (Chile); Valente, M., E-mail: figueror@ufro.cl [Consejo Nacional de Investigaciones Cientificas y Tecnicas, Av. Ravadavia 1917, C1033AAJ, Buenos Aires (Argentina)
2013-08-01
In-vivo X-ray fluorescence constitutes a useful and accurate technique, worldwide established for constituent elementary distribution assessment. Actually, concentration distributions of arbitrary user-selected elements can be achieved along sample surface with the aim of identifying and simultaneously quantifying every constituent element. The method is based on the use of a collimated X-ray beam reaching the sample. However, one common drawback for considering the application of this technique for routine clinical examinations was the lack of information about associated dose delivery. This work presents a complete study of the dose distribution resulting from an in-vivo X-ray fluorescence scanning for quantifying biohazard materials on human hands. Absorbed dose has been estimated by means of dosimetric models specifically developed to this aim. In addition, complete dose distributions have been obtained by means of full radiation transport calculations in based on stochastic Monte Carlo techniques. A dedicated subroutine has been developed using the Penelope 2008 main code also integrated with dedicated programs -Mat Lab supported- for 3 dimensional dose distribution visualization. The obtained results show very good agreement between approximate analytical models and full descriptions by means of Monte Carlo simulations. (Author)
Almatani, Turki; Hugtenburg, Richard P; Lewis, Ryan D; Barley, Susan E; Edwards, Mark A
2016-10-01
Cone beam CT (CBCT) images contain more scatter than a conventional CT image and therefore provide inaccurate Hounsfield units (HUs). Consequently, CBCT images cannot be used directly for radiotherapy dose calculation. The aim of this study is to enable dose calculations to be performed with the use of CBCT images taken during radiotherapy and evaluate the necessity of replanning. A patient with prostate cancer with bilateral metallic prosthetic hip replacements was imaged using both CT and CBCT. The multilevel threshold (MLT) algorithm was used to categorize pixel values in the CBCT images into segments of homogeneous HU. The variation in HU with position in the CBCT images was taken into consideration. This segmentation method relies on the operator dividing the CBCT data into a set of volumes where the variation in the relationship between pixel values and HUs is small. An automated MLT algorithm was developed to reduce the operator time associated with the process. An intensity-modulated radiation therapy plan was generated from CT images of the patient. The plan was then copied to the segmented CBCT (sCBCT) data sets with identical settings, and the doses were recalculated and compared. Gamma evaluation showed that the percentage of points in the rectum with γ algorithms, respectively. Compared with the planning CT (pCT) plan, the MLT algorithm showed -0.46% dose difference with 8 h operator time while the automated MLT algorithm showed -1.3%, which are both considered to be clinically acceptable, when using collapsed cone algorithm. The segmentation of CBCT images using the method in this study can be used for dose calculation. For a patient with prostate cancer with bilateral hip prostheses and the associated issues with CT imaging, the MLT algorithms achieved a sufficient dose calculation accuracy that is clinically acceptable. The automated MLT algorithm reduced the operator time associated with implementing the MLT algorithm to achieve clinically
Strange, D. L.; Bander, T. J.
1981-04-01
The MILDOS Computer Code estimates impacts from radioactive emissions from uranium milling facilities. These impacts are presented as dose commitments to individuals and the regional population within an 80 km radius of the facility. Only airborne releases of radioactive materials are considered: releases to surface water and to groundwater are not addressed in MILDOS. This code is multi-purposed and can be used to evaluate population doses for NEPA assessments, maximum individual doses for predictive 40 CFR 190 compliance evaluations, or maximum offsite air concentrations for predictive evaluations of 10 CFR 20 compliance. Emissions of radioactive materials from fixed point source locations and from area sources are modeled using a sector-averaged Gaussian plume dispersion model, which utilizes user-provided wind frequency data. Mechanisms such as deposition of particulates, resuspension. radioactive decay and ingrowth of daughter radionuclides are included in the transport model. Annual average air concentrations are computed, from which subsequent impacts to humans through various pathways are computed. Ground surface concentrations are estimated from deposition buildup and ingrowth of radioactive daughters. The surface concentrations are modified by radioactive decay, weathering and other environmental processes. The MILDOS Computer Code allows the user to vary the emission sources as a step function of time by adjustinq the emission rates. which includes shutting them off completely. Thus the results of a computer run can be made to reflect changing processes throughout the facility's operational lifetime. The pathways considered for individual dose commitments and for population impacts are: • Inhalation • External exposure from ground concentrations • External exposure from cloud immersion • Ingestioo of vegetables • Ingestion of meat • Ingestion of milk • Dose commitments are calculated using dose conversion factors, which are ultimately based
Beaulieu, Luc [Departement de Radio-Oncologie et Centre de Recherche en Cancerologie de l' Universite Laval, Centre hospitalier universitaire de Quebec, Quebec, Quebec G1R 2J6 (Canada) and Departement de Physique, de Genie Physique et d' Optique, Universite Laval, Quebec, Quebec G1R 2J6 (Canada); Carlsson Tedgren, Asa [Department of Medical and Health Sciences (IMH), Radiation Physics, Faculty of Health Sciences, Linkoeping University, SE-581 85 Linkoeping (Sweden) and Swedish Radiation Safety Authority, SE-171 16 Stockholm (Sweden); Carrier, Jean-Francois [Departement de radio-oncologie, CRCHUM, Centre hospitalier de l' Universite de Montreal, Montreal, Quebec H2L 4M1 (Canada) and Departement de physique, Universite de Montreal, Montreal, Quebec H3C 3J7 (Canada); and others
2012-10-15
the local medium be reported along with the TG-43 calculated doses. Assignments of voxel-by-voxel cross sections represent a particular challenge. Electron density information is readily extracted from CT imaging, but cannot be used to distinguish between different materials having the same density. Therefore, a recommendation is made to use a number of standardized materials to maintain uniformity across institutions. Sensitivity analysis shows that this recommendation offers increased accuracy over TG-43. MBDCA commissioning will share commonalities with current TG-43-based systems, but in addition there will be algorithm-specific tasks. Two levels of commissioning are recommended: reproducing TG-43 dose parameters and testing the advanced capabilities of MBDCAs. For validation of heterogeneity and scatter conditions, MBDCAs should mimic the 3D dose distributions from reference virtual geometries. Potential changes in BT dose prescriptions and MBDCA limitations are discussed. When data required for full MBDCA implementation are insufficient, interim recommendations are made and potential areas of research are identified. Application of TG-186 guidance should retain practice uniformity in transitioning from the TG-43 to the MBDCA approach.
Towards real-time photon Monte Carlo dose calculation in the cloud.
Ziegenhein, Peter; Kozin, Igor; Kamerling, Cornelis Philippus; Oelfke, Uwe
2017-01-31
Near real-time application of Monte Carlo (MC) dose calculation in clinic and research is hindered by long computational runtimes of established software. Currently, fast MC software solutions are available utilising accelerators such as GPUs or clusters of central processing units (CPU)-based system. Both platforms are expensive in terms of purchase costs and maintenance and, in case of the GPU, provide only limited scalability. In this work we propose a cloud-based MC solution, which offers high scalability of accurate photon dose calculations. The MC simulations run on a private virtual supercomputer that forms in the cloud. Computational resources can be provisioned dynamically at low costs without upfront investment in expensive hardware. A client-server software solution has been developed which controls the simulations and efficiently transports data to and from the cloud. The client application integrates seamlessly into a Treatment Planning System (TPS). It runs the MC simulation workflow automatically and securely exchanges simulation data with the server side application that controls the virtual supercomputer. The Advanced Encryption Standard (AES) was used to add an addition security layer which encrypts and decrypts patient data on-the-fly at the processor register level. We could show that our cloud-based MC framework enables near real-time dose computation. It delivers excellent linear scaling for high-resolution datasets with absolute runtimes of 1.1 to 10.9 seconds for simulating a clinical prostate and liver case up to 1\\% statistical uncertainty. The computation times include the data transportation processes with the cloud as well as process scheduling and synchronisation overhead. Cloud based MC simulations offer a fast, affordable and easily accessible alternative for near real-time accurate dose calculations to currently used GPU or cluster solutions.
Towards real-time photon Monte Carlo dose calculation in the cloud
Ziegenhein, Peter; Kozin, Igor N.; Kamerling, Cornelis Ph; Oelfke, Uwe
2017-06-01
Near real-time application of Monte Carlo (MC) dose calculation in clinic and research is hindered by the long computational runtimes of established software. Currently, fast MC software solutions are available utilising accelerators such as graphical processing units (GPUs) or clusters based on central processing units (CPUs). Both platforms are expensive in terms of purchase costs and maintenance and, in case of the GPU, provide only limited scalability. In this work we propose a cloud-based MC solution, which offers high scalability of accurate photon dose calculations. The MC simulations run on a private virtual supercomputer that is formed in the cloud. Computational resources can be provisioned dynamically at low cost without upfront investment in expensive hardware. A client-server software solution has been developed which controls the simulations and transports data to and from the cloud efficiently and securely. The client application integrates seamlessly into a treatment planning system. It runs the MC simulation workflow automatically and securely exchanges simulation data with the server side application that controls the virtual supercomputer. Advanced encryption standards were used to add an additional security layer, which encrypts and decrypts patient data on-the-fly at the processor register level. We could show that our cloud-based MC framework enables near real-time dose computation. It delivers excellent linear scaling for high-resolution datasets with absolute runtimes of 1.1 seconds to 10.9 seconds for simulating a clinical prostate and liver case up to 1% statistical uncertainty. The computation runtimes include the transportation of data to and from the cloud as well as process scheduling and synchronisation overhead. Cloud-based MC simulations offer a fast, affordable and easily accessible alternative for near real-time accurate dose calculations to currently used GPU or cluster solutions.
Diego Azcona, Juan; Barbés, Benigno; Wang, Lilie; Burguete, Javier
2016-01-01
This paper presents a method to obtain the pencil-beam kernels that characterize a megavoltage photon beam generated in a flattening filter free (FFF) linear accelerator (linac) by deconvolution from experimental measurements at different depths. The formalism is applied to perform independent dose calculations in modulated fields. In our previous work a formalism was developed for ideal flat fluences exiting the linac’s head. That framework could not deal with spatially varying energy fluences, so any deviation from the ideal flat fluence was treated as a perturbation. The present work addresses the necessity of implementing an exact analysis where any spatially varying fluence can be used such as those encountered in FFF beams. A major improvement introduced here is to handle the actual fluence in the deconvolution procedure. We studied the uncertainties associated to the kernel derivation with this method. Several Kodak EDR2 radiographic films were irradiated with a 10 MV FFF photon beam from two linacs from different vendors, at the depths of 5, 10, 15, and 20cm in polystyrene (RW3 water-equivalent phantom, PTW Freiburg, Germany). The irradiation field was a 50mm diameter circular field, collimated with a lead block. The 3D kernel for a FFF beam was obtained by deconvolution using the Hankel transform. A correction on the low dose part of the kernel was performed to reproduce accurately the experimental output factors. Error uncertainty in the kernel derivation procedure was estimated to be within 0.2%. Eighteen modulated fields used clinically in different treatment localizations were irradiated at four measurement depths (total of fifty-four film measurements). Comparison through the gamma-index to their corresponding calculated absolute dose distributions showed a number of passing points (3%, 3mm) mostly above 99%. This new procedure is more reliable and robust than the previous one. Its ability to perform accurate independent dose calculations was
SU-E-T-496: A Study of Two Commercial Dose Calculation Algorithms in Low Density Phantom
Lim, S; Lovelock, D; Yorke, E; Kuo, L; LoSasso, T [Memorial Sloan- Kettering Cancer Center, NY, NY (United States)
2014-06-01
Purpose: Some lung cancer patients have very low lung density due to comorbidities. We investigate calculation accuracy of Eclipse AAA and Acuros(AXB) using a phantom that simulates this situation. Methods: A 2.5 x 5.0 x 5 cm (long) solid water inhomogeneity positioned 10 cm deep in a Balsa lung phantom (density 0.099 gm/cc) was irradiated with an off-center field such that the central axis was parallel to one side of the inhomogeneity. Radiochromic films were placed at 2.5cm(S1) and 5cm(S2) depths. After CT scanning, Hounsfield Units(HU) were converted to electron(ρe) and mass(ρm) density using in-house(IH) and vendor-supplied(V) calibration curves. IH electron densities were generated using a commercial electron density phantom. The phantom was exposed to 6 MV 3x3 and 20x20 fields. Dose distributions were calculated using the AAA and AXB algorithms. Results: The HU of BW is -910±40 which translates to ρe of 0.088±0.050(IH) and 0.090±0.050(V), and ρm of 0.101±0.045(IH) and 0.103±0.039(V). Both ρe(V) and ρm(V) are higher than ρe(IH) and ρm(IH) respectively by 1.4-5.3% and 0.5-12.3%. The average calculated dose inside the solid water ‘tumor’ are within 3.7% and 2.4% of measurements for both calibrations and field sizes using AAA and AXB. Within 10mm outside the ‘tumor’, AAA on average underestimates by 18.3% and 17.0% respectively for 3x3 using IH and V. AXB underestimates by 5.9%(S1)-6.6%(S2) and 13.1%(S1)-16.0%(S2) respectively using IH and V. For 20x20, AAA and AXB underestimate by 2.8%(S1)-4.4%(S2) and 0.3%(S1)-1.4%(S2) respectively with either calibration. Conclusion: The difference in the HU calibration between V and IH is not of clinical significance in normal field sizes. In the low density region of small fields, the calculations from both algorithms differ significantly from measurements. This may be attributed to the insufficient lateral electron transport modeled by two algorithms resulting in the over-estimation in penumbra
vantVeld, AA
1997-01-01
Accuracy in dose and position, defined as complementary criteria, was determined for blocked photon field calculations with a pencil beam based treatment planning system. The concept of field accuracy has been defined as a combination of deviations in dose and position. Absolute dose deviations were
Shielding calculations for industrial 5/7.5MeV electron accelerators using the MCNP Monte Carlo Code
Peri, Eyal; Orion, Itzhak
2017-09-01
High energy X-rays from accelerators are used to irradiate food ingredients to prevent growth and development of unwanted biological organisms in food, and by that extend the shelf life of the products. The production of X-rays is done by accelerating 5 MeV electrons and bombarding them into a heavy target (high Z). Since 2004, the FDA has approved using 7.5 MeV energy, providing higher production rates with lower treatments costs. In this study we calculated all the essential data needed for a straightforward concrete shielding design of typical food accelerator rooms. The following evaluation is done using the MCNP Monte Carlo code system: (1) Angular dependence (0-180°) of photon dose rate for 5 MeV and 7.5 MeV electron beams bombarding iron, aluminum, gold, tantalum, and tungsten targets. (2) Angular dependence (0-180°) spectral distribution simulations of bremsstrahlung for gold, tantalum, and tungsten bombarded by 5 MeV and 7.5 MeV electron beams. (3) Concrete attenuation calculations in several photon emission angles for the 5 MeV and 7.5 MeV electron beams bombarding a tantalum target. Based on the simulation, we calculated the expected increase in dose rate for facilities intending to increase the energy from 5 MeV to 7.5 MeV, and the concrete width needed to be added in order to keep the existing dose rate unchanged.
Cochran, Donna J.; Boutte, Alvin J.; Chen, Dakai; Pellish, Jonathan A.; Ladbury, Raymond L.; Casey, Megan C.; Campola, Michael J.; Wilcox, Edward P.; Obryan, Martha V.; LaBel, Kenneth A.;
2012-01-01
Vulnerability of a variety of candidate spacecraft electronics to total ionizing dose and displacement damage is studied. Devices tested include optoelectronics, digital, analog, linear, and hybrid devices.
Dose conversion coefficients for monoenergetic electrons incident on a realistic human eye model with different lens cell populations
Nogueira, P; Vaz, P [Technological and Nuclear Institute, Estrada Nacional No 10, 2686-953 Sacavem (Portugal); Zankl, M; Schlattl, H, E-mail: pedro.nogueira@helmholtz-muenchen.de [Helmholtz Zentrum Muenchen-German Research Center for Environmental Health, Research Unit Medical Radiation Physics and Diagnostics, Ingolstaedter Landstrasse 1, D-85764 Neuherberg (Germany)
2011-11-07
The radiation-induced posterior subcapsular cataract has long been generally accepted to be a deterministic effect that does not occur at doses below a threshold of at least 2 Gy. Recent epidemiological studies indicate that the threshold for cataract induction may be much lower or that there may be no threshold at all. A thorough study of this subject requires more accurate dose estimates for the eye lens than those available in ICRP Publication 74. Eye lens absorbed dose per unit fluence conversion coefficients for electron irradiation were calculated using a geometrical model of the eye that takes into account different cell populations of the lens epithelium, together with the MCNPX Monte Carlo radiation transport code package. For the cell population most sensitive to ionizing radiation-the germinative cells-absorbed dose per unit fluence conversion coefficients were determined that are up to a factor of 4.8 higher than the mean eye lens absorbed dose conversion coefficients for electron energies below 2 MeV. Comparison of the results with previously published values for a slightly different eye model showed generally good agreement for all electron energies. Finally, the influence of individual anatomical variability was quantified by positioning the lens at various depths below the cornea. A depth difference of 2 mm between the shallowest and the deepest location of the germinative zone can lead to a difference between the resulting absorbed doses of up to nearly a factor of 5000 for electron energy of 0.7 MeV.
Comparison of CT number calibration techniques for CBCT-based dose calculation
Dunlop, Alex [The Royal Marsden NHS Foundation Trust, Joint Department of Physics, Institute of Cancer Research, London (United Kingdom); The Royal Marsden Hospital, Sutton, Surrey, Downs Road (United Kingdom); McQuaid, Dualta; Nill, Simeon; Hansen, Vibeke N.; Oelfke, Uwe [The Royal Marsden NHS Foundation Trust, Joint Department of Physics, Institute of Cancer Research, London (United Kingdom); Murray, Julia; Bhide, Shreerang; Harrington, Kevin [The Royal Marsden Hospital, Sutton, Surrey, Downs Road (United Kingdom); The Institute of Cancer Research, London (United Kingdom); Poludniowski, Gavin [Karolinska University Hospital, Department of Medical Physics, Stockholm (Sweden); Nutting, Christopher [The Institute of Cancer Research, London (United Kingdom); Newbold, Kate [The Royal Marsden Hospital, Sutton, Surrey, Downs Road (United Kingdom)
2015-12-15
The aim of this work was to compare and validate various computed tomography (CT) number calibration techniques with respect to cone beam CT (CBCT) dose calculation accuracy. CBCT dose calculation accuracy was assessed for pelvic, lung, and head and neck (H and N) treatment sites for two approaches: (1) physics-based scatter correction methods (CBCT{sub r}); (2) density override approaches including assigning water density to the entire CBCT (W), assignment of either water or bone density (WB), and assignment of either water or lung density (WL). Methods for CBCT density assignment within a commercially available treatment planning system (RS{sub auto}), where CBCT voxels are binned into six density levels, were assessed and validated. Dose-difference maps and dose-volume statistics were used to compare the CBCT dose distributions with the ground truth of a planning CT acquired the same day as the CBCT. For pelvic cases, all CTN calibration methods resulted in average dose-volume deviations below 1.5 %. RS{sub auto} provided larger than average errors for pelvic treatments for patients with large amounts of adipose tissue. For H and N cases, all CTN calibration methods resulted in average dose-volume differences below 1.0 % with CBCT{sub r} (0.5 %) and RS{sub auto} (0.6 %) performing best. For lung cases, WL and RS{sub auto} methods generated dose distributions most similar to the ground truth. The RS{sub auto} density override approach is an attractive option for CTN adjustments for a variety of anatomical sites. RS{sub auto} methods were validated, resulting in dose calculations that were consistent with those calculated on diagnostic-quality CT images, for CBCT images acquired of the lung, for patients receiving pelvic RT in cases without excess adipose tissue, and for H and N cases. (orig.) [German] Ziel dieser Arbeit ist der Vergleich und die Validierung mehrerer CT-Kalibrierungsmethoden zur Dosisberechnung auf der Grundlage von Kegelstrahlcomputertomographie
SU-F-BRD-06: Robust Dose Calculation in Intensity Modulated Proton Therapy
Brosch, R [ASU, Tempe, AZ (United States); Liu, W [Mayo Clinic Arizona, Phoenix, AZ (United States)
2015-06-15
Purpose: Commissioning data for intensity modulated proton therapy (IMPT) must be post-processed by fits to ad-hoc functions to derive the dose calculation kernel parameters in a treatment planning system (TPS). Whether from experimental measurement or Monte Carlo simulation, the limited and noisy nature of such data makes this task very challenging. We present a method to improve the modeling of the lateral dose distribution of clinical energy proton beams in water to commission an in-house IMPT dose calculation engine. Methods: A linear sum of three Gaussian distribution functions was fitted to the lateral dose data in logarithmic scale. Starting values of fitting solutions were determined from the Generalized Highland Approximation. We exhaustively optimized the combinations of data weights with upper bounds of the fitting solutions to minimize confidence intervals of the fitting solutions while maintaining the coefficient of determination (R{sup 2}). Results: Across all energies, average confidence bounds improved 72.88% [Max: 88.28%, Min: 55.05%] for small angle coulomb scattering, 114.25% [409.13%, 66.72%,] for nuclear scattering, and 68.66% [141.09%, 33.27%] for large angle coulomb scattering, while the coefficients of determination of the fits (R{sup 2}) remained comparable. On average R {sup 2} only changed 0.18% and were very close to 1 (approx. 0.999). Wilcoxon signed rank tests comparing unweighted/unbounded fits with weighted/bounded fits averaged 0.0146 (Max: 0.177, Min: 7.05×10−{sup 7}) for small angle Coulomb, 0.0903 (0.945, 7.05×10−{sup 7}) for nuclear, and 0.254 (0.871, 1.86×10−{sup 6}) for large angle Coulomb scattering. This allows rejection of the null hypothesis for small angle Coulomb scattering at the 0.015 level and nuclear interaction at the 0.1 level. Conclusion: Optimal weights assigned to IMPT lateral dose data minimized fitting to stochastic noise in the tail region. Optimizing the upper bounds of fitting parameters improved
Wignall, Jessica A.; Shapiro, Andrew J.; Wright, Fred A.; Woodruff, Tracey J.; Chiu, Weihsueh A.; Guyton, Kathryn Z.
2014-01-01
Background: Benchmark dose (BMD) modeling computes the dose associated with a prespecified response level. While offering advantages over traditional points of departure (PODs), such as no-observed-adverse-effect-levels (NOAELs), BMD methods have lacked consistency and transparency in application, interpretation, and reporting in human health assessments of chemicals. Objectives: We aimed to apply a standardized process for conducting BMD modeling to reduce inconsistencies in model fitting and selection. Methods: We evaluated 880 dose–response data sets for 352 environmental chemicals with existing human health assessments. We calculated benchmark doses and their lower limits [10% extra risk, or change in the mean equal to 1 SD (BMD/L10/1SD)] for each chemical in a standardized way with prespecified criteria for model fit acceptance. We identified study design features associated with acceptable model fits. Results: We derived values for 255 (72%) of the chemicals. Batch-calculated BMD/L10/1SD values were significantly and highly correlated (R2 of 0.95 and 0.83, respectively, n = 42) with PODs previously used in human health assessments, with values similar to reported NOAELs. Specifically, the median ratio of BMDs10/1SD:NOAELs was 1.96, and the median ratio of BMDLs10/1SD:NOAELs was 0.89. We also observed a significant trend of increasing model viability with increasing number of dose groups. Conclusions: BMD/L10/1SD values can be calculated in a standardized way for use in health assessments on a large number of chemicals and critical effects. This facilitates the exploration of health effects across multiple studies of a given chemical or, when chemicals need to be compared, providing greater transparency and efficiency than current approaches. Citation: Wignall JA, Shapiro AJ, Wright FA, Woodruff TJ, Chiu WA, Guyton KZ, Rusyn I. 2014. Standardizing benchmark dose calculations to improve science-based decisions in human health assessments. Environ Health
Influence of metallic dental implants and metal artefacts on dose calculation accuracy
Maerz, Manuel; Koelbl, Oliver; Dobler, Barbara [Regensburg University Medical Center, Department of Radiotherapy, Regensburg (Germany)
2014-10-31
Metallic dental implants cause severe streaking artefacts in computed tomography (CT) data, which inhibit the correct representation of shape and density of the metal and the surrounding tissue. The aim of this study was to investigate the impact of dental implants on the accuracy of dose calculations in radiation therapy planning and the benefit of metal artefact reduction (MAR). A second aim was to determine the treatment technique which is less sensitive to the presence of metallic implants in terms of dose calculation accuracy. Phantoms consisting of homogeneous water equivalent material surrounding dental implants were designed. Artefact-containing CT data were corrected using the correct density information. Intensity-modulated radiation therapy (IMRT) and volumetric modulated arc therapy (VMAT) plans were calculated on corrected and uncorrected CT data and compared to 2-dimensional dose measurements using GafChromic trademark EBT2 films. For all plans the accuracy of dose calculations is significantly higher if performed on corrected CT data (p = 0.015). The agreement of calculated and measured dose distributions is significantly higher for VMAT than for IMRT plans for calculations on uncorrected CT data (p = 0.011) as well as on corrected CT data (p = 0.029). For IMRT and VMAT the application of metal artefact reduction significantly increases the agreement of dose calculations with film measurements. VMAT was found to provide the highest accuracy on corrected as well as on uncorrected CT data. VMAT is therefore preferable over IMRT for patients with metallic implants, if plan quality is comparable for the two techniques. (orig.) [German] Zahnimplantate aus Metall verursachen in Computertomographiedaten (CT) streifenfoermige Artefakte. Diese verhindern eine korrekte Zuordnung von Form und Dichteeigenschaften des Metalls und des umgebenden Gewebes. Ziel dieser Studie war es, den Einfluss von Zahnimplantaten auf die Genauigkeit der Dosisberechnung in der
Cheng-Ting Shih
Full Text Available Polymer gels are regarded as a potential dosimeter for independent validation of absorbed doses in clinical radiotherapy. Several imaging modalities have been used to convert radiation-induced polymerization to absorbed doses from a macro-scale viewpoint. This study developed a novel dose conversion mechanism by texture analysis of scanning electron microscopy (SEM images. The modified N-isopropyl-acrylamide (NIPAM gels were prepared under normoxic conditions, and were administered radiation doses from 5 to 20 Gy. After freeze drying, the gel samples were sliced for SEM scanning with 50×, 500×, and 3500× magnifications. Four texture indices were calculated based on the gray level co-occurrence matrix (GLCM. The results showed that entropy and homogeneity were more suitable than contrast and energy as dose indices for higher linearity and sensitivity of the dose response curves. After parameter optimization, an R (2 value of 0.993 can be achieved for homogeneity using 500× magnified SEM images with 27 pixel offsets and no outlier exclusion. For dose verification, the percentage errors between the prescribed dose and the measured dose for 5, 10, 15, and 20 Gy were -7.60%, 5.80%, 2.53%, and -0.95%, respectively. We conclude that texture analysis can be applied to the SEM images of gel dosimeters to accurately convert micro-scale structural features to absorbed doses. The proposed method may extend the feasibility of applying gel dosimeters in the fields of diagnostic radiology and radiation protection.
Shih, Cheng-Ting; Hsu, Jui-Ting; Han, Rou-Ping; Hsieh, Bor-Tsung; Chang, Shu-Jun; Wu, Jay
2013-01-01
Polymer gels are regarded as a potential dosimeter for independent validation of absorbed doses in clinical radiotherapy. Several imaging modalities have been used to convert radiation-induced polymerization to absorbed doses from a macro-scale viewpoint. This study developed a novel dose conversion mechanism by texture analysis of scanning electron microscopy (SEM) images. The modified N-isopropyl-acrylamide (NIPAM) gels were prepared under normoxic conditions, and were administered radiation doses from 5 to 20 Gy. After freeze drying, the gel samples were sliced for SEM scanning with 50×, 500×, and 3500× magnifications. Four texture indices were calculated based on the gray level co-occurrence matrix (GLCM). The results showed that entropy and homogeneity were more suitable than contrast and energy as dose indices for higher linearity and sensitivity of the dose response curves. After parameter optimization, an R (2) value of 0.993 can be achieved for homogeneity using 500× magnified SEM images with 27 pixel offsets and no outlier exclusion. For dose verification, the percentage errors between the prescribed dose and the measured dose for 5, 10, 15, and 20 Gy were -7.60%, 5.80%, 2.53%, and -0.95%, respectively. We conclude that texture analysis can be applied to the SEM images of gel dosimeters to accurately convert micro-scale structural features to absorbed doses. The proposed method may extend the feasibility of applying gel dosimeters in the fields of diagnostic radiology and radiation protection.
Schach von Wittenau, A.E.; Cox, L.J.; Bergstrom, P.M. Jr.; Hornstein, S.M. [Lawrence Livermore National Lab., CA (United States); Mohan, R.; Libby, B.; Wu, Q. [Medical Coll. of Virginia, Richmond, VA (United States); Lovelock, D.M.J. [Memorial Sloan-Kettering Cancer Center, New York, NY (United States)
1997-03-01
The goal of the PEREGRINE Monte Carlo Dose Calculation Project is to deliver a Monte Carlo package that is both accurate and sufficiently fast for routine clinical use. One of the operational requirements for photon-treatment plans is a fast, accurate method of describing the photon phase-space distribution at the surface of the patient. The open-field case is computationally the most tractable; we know, a priori, for a given machine and energy, the locations and compositions of the relevant accelerator components (i.e., target, primary collimator, flattening filter, and monitor chamber). Therefore, we can precalculate and store the expected photon distributions. For any open-field treatment plan, we then evaluate these existing photon phase-space distributions at the patient`s surface, and pass the obtained photons to the dose calculation routines within PEREGRINE. We neglect any effect of the intervening air column, including attenuation of the photons and production of contaminant electrons. In principle, for treatment plans requiring jaws, blocks, and wedges, we could precalculate and store photon phase-space distributions for various combinations of field sizes and wedges. This has the disadvantage that we would have to anticipate those combinations and that subsequently PEREGRINE would not be able to treat other plans. Therefore, PEREGRINE tracks photons through the patient-dependent beam modifiers. The geometric and physics methods used to do this are described here. 4 refs., 8 figs.
Lung Dose Calculation With SPECT/CT for {sup 90}Yittrium Radioembolization of Liver Cancer
Yu, Naichang, E-mail: yun@ccf.org [Department of Radiation Oncology, Cleveland Clinic, Cleveland, OH (United States); Srinivas, Shaym M.; DiFilippo, Frank P.; Shrikanthan, Sankaran [Department of Nuclear Medicine, Cleveland Clinic, Cleveland, OH (United States); Levitin, Abraham; McLennan, Gordon; Spain, James [Department of Interventional Radiology, Cleveland Clinic, Cleveland, OH (United States); Xia, Ping; Wilkinson, Allan [Department of Radiation Oncology, Cleveland Clinic, Cleveland, OH (United States)
2013-03-01
Purpose: To propose a new method to estimate lung mean dose (LMD) using technetium-99m labeled macroaggregated albumin ({sup 99m}Tc-MAA) single photon emission CT (SPECT)/CT for {sup 90}Yttrium radioembolization of liver tumors and to compare the LMD estimated using SPECT/CT with clinical estimates of LMD using planar gamma scintigraphy (PS). Methods and Materials: Images of 71 patients who had SPECT/CT and PS images of {sup 99m}Tc-MAA acquired before TheraSphere radioembolization of liver cancer were analyzed retrospectively. LMD was calculated from the PS-based lung shunt assuming a lung mass of 1 kg and 50 Gy per GBq of injected activity shunted to the lung. For the SPECT/CT-based estimate, the LMD was calculated with the activity concentration and lung volume derived from SPECT/CT. The effect of attenuation correction and the patient's breathing on the calculated LMD was studied with the SPECT/CT. With these effects correctly taken into account in a more rigorous fashion, we compared the LMD calculated with SPECT/CT with the LMD calculated with PS. Results: The mean dose to the central region of the lung leads to a more accurate estimate of LMD. Inclusion of the lung region around the diaphragm in the calculation leads to an overestimate of LMD due to the misregistration of the liver activity to the lung from the patient's breathing. LMD calculated based on PS is a poor predictor of the actual LMD. For the subpopulation with large lung shunt, the mean overestimation from the PS method for the lung shunt was 170%. Conclusions: A new method of calculating the LMD for TheraSphere and SIR-Spheres radioembolization of liver cancer based on {sup 99m}Tc-MAA SPECT/CT is presented. The new method provides a more accurate estimate of radiation risk to the lungs. For patients with a large lung shunt calculated from PS, a recalculation of LMD based on SPECT/CT is recommended.
Shi, Chengyu
Assessment of radiation dose and possible risk to a pregnant woman and her fetus is an important task in radiation protection. Although stylized models for male and female patients of different ages have been developed, tomographic models for pregnant women have not been developed to date. This dissertation presents an effort to construct a partial-body model of a pregnant woman from a set of CT images. The patient was 30-weeks pregnant, and the CT scan covered the portion of the body between the lower breast and the upper thigh in 70 slices, each 7 mm thick. The image resolution was 512 x 512 pixels in a 48 cm x 48 cm field. The images were carefully segmented to identify 34 organs and tissues, It has been found that the masses are different from the Reference Woman. The characteristics of the resulting model is discussed and compared with one existing stylized mathematical model for pregnant women. Based on this tomographic model, a Monte Carlo code, EGS4-VLSI, was used to derive Specific Absorbed Fractions. Monoenergetic and isotropic photon and electron emitters distributed in different source organs were assumed and the energies ranged from 10 keV to 4 MeV for photons and from 100 keV to 4 MeV for electrons. The results for high energy (>50 keV) photons showed general agreement with previous studies, however, the results for lower energy (women are satisfactory for a very large size patient for most of the photon energies (between 50 keV and 4 MeV). However, a tomographic model has to be used to obtain acceptable dose assessments for electrons. The newly calculated SAF data set can provide the nuclear medicine dosimetry field a new perspective involving internal electrons.
Westerly, David C. [Department of Radiation Oncology, University of Colorado School of Medicine, Aurora, Colorado 80045 (United States); Mo Xiaohu; DeLuca, Paul M. Jr. [Department of Medical Physics, School of Medicine and Public Health, University of Wisconsin, Madison, Wisconsin 53705 (United States); Tome, Wolfgang A. [Department of Medical Physics, School of Medicine and Public Health, University of Wisconsin, Madison, Wisconsin 53705 and Institute of Onco-Physics, Albert Einstein College of Medicine and Division of Medical Physics, Department of Radiation Oncology, Montefiore Medical Center, Bronx, New York 10461 (United States); Mackie, Thomas R. [Department of Medical Physics, School of Medicine and Public Health, University of Wisconsin, Madison, Wisconsin 53705 and Department of Human Oncology, School of Medicine and Public Health, University of Wisconsin, Madison, Wisconsin 53792 (United States)
2013-06-15
Purpose: Pencil beam algorithms are commonly used for proton therapy dose calculations. Szymanowski and Oelfke ['Two-dimensional pencil beam scaling: An improved proton dose algorithm for heterogeneous media,' Phys. Med. Biol. 47, 3313-3330 (2002)] developed a two-dimensional (2D) scaling algorithm which accurately models the radial pencil beam width as a function of depth in heterogeneous slab geometries using a scaled expression for the radial kernel width in water as a function of depth and kinetic energy. However, an assumption made in the derivation of the technique limits its range of validity to cases where the input expression for the radial kernel width in water is derived from a local scattering power model. The goal of this work is to derive a generalized form of 2D pencil beam scaling that is independent of the scattering power model and appropriate for use with any expression for the radial kernel width in water as a function of depth. Methods: Using Fermi-Eyges transport theory, the authors derive an expression for the radial pencil beam width in heterogeneous slab geometries which is independent of the proton scattering power and related quantities. The authors then perform test calculations in homogeneous and heterogeneous slab phantoms using both the original 2D scaling model and the new model with expressions for the radial kernel width in water computed from both local and nonlocal scattering power models, as well as a nonlocal parameterization of Moliere scattering theory. In addition to kernel width calculations, dose calculations are also performed for a narrow Gaussian proton beam. Results: Pencil beam width calculations indicate that both 2D scaling formalisms perform well when the radial kernel width in water is derived from a local scattering power model. Computing the radial kernel width from a nonlocal scattering model results in the local 2D scaling formula under-predicting the pencil beam width by as much as 1.4 mm (21%) at
A BrachyPhantom for verification of dose calculation of HDR brachytherapy planning system
Austerlitz, C. [Clinica Diana Campos, Recife, PE 52020-030 (Brazil); Campos, C. A. T. [Pontifícia Universidade Católica do Rio de Janeiro, RJ 22451-900 (Brazil)
2013-11-15
Purpose: To develop a calibration phantom for {sup 192}Ir high dose rate (HDR) brachytherapy units that renders possible the direct measurement of absorbed dose to water and verification of treatment planning system.Methods: A phantom, herein designated BrachyPhantom, consists of a Solid Water™ 8-cm high cylinder with a diameter of 14 cm cavity in its axis that allows the positioning of an A1SL ionization chamber with its reference measuring point at the midheight of the cylinder's axis. Inside the BrachyPhantom, at a 3-cm radial distance from the chamber's reference measuring point, there is a circular channel connected to a cylindrical-guide cavity that allows the insertion of a 6-French flexible plastic catheter from the BrachyPhantom surface. The PENELOPE Monte Carlo code was used to calculate a factor, P{sub sw}{sup lw}, to correct the reading of the ionization chamber to a full scatter condition in liquid water. The verification of dose calculation of a HDR brachytherapy treatment planning system was performed by inserting a catheter with a dummy source in the phantom channel and scanning it with a CT. The CT scan was then transferred to the HDR computer program in which a multiple treatment plan was programmed to deliver a total dose of 150 cGy to the ionization chamber. The instrument reading was then converted to absorbed dose to water using the N{sub gas} formalism and the P{sub sw}{sup lw} factor. Likewise, the absorbed dose to water was calculated using the source strength, S{sub k}, values provided by 15 institutions visited in this work.Results: A value of 1.020 (0.09%, k= 2) was found for P{sub sw}{sup lw}. The expanded uncertainty in the absorbed dose assessed with the BrachyPhantom was found to be 2.12% (k= 1). To an associated S{sub k} of 27.8 cGy m{sup 2} h{sup −1}, the total irradiation time to deliver 150 cGy to the ionization chamber point of reference was 161.0 s. The deviation between the absorbed doses to water assessed with
Takahashi, F; Endo, A
2007-01-01
A system utilising radiation transport codes has been developed to derive accurate dose distributions in a human body for radiological accidents. A suitable model is quite essential for a numerical analysis. Therefore, two tools were developed to setup a 'problem-dependent' input file, defining a radiation source and an exposed person to simulate the radiation transport in an accident with the Monte Carlo calculation codes-MCNP and MCNPX. Necessary resources are defined by a dialogue method with a generally used personal computer for both the tools. The tools prepare human body and source models described in the input file format of the employed Monte Carlo codes. The tools were validated for dose assessment in comparison with a past criticality accident and a hypothesized exposure.
Calculation of fluence and absorbed dose in head tissues due to different photon energies.
Azorín, C; Vega-Carrillo, H R; Rivera, T; Azorín, J
2014-01-01
Calculations of fluence and absorbed dose in head tissues due to different photon energies were carried out using the MCNPX code, to simulate two models of a patient's head: one spherical and another more realistic ellipsoidal. Both head models had concentric shells to describe the scalp skin, the cranium and the brain. The tumor was located at the center of the head and it was a 1 cm-radius sphere. The MCNPX code was run for different energies. Results showed that the fluence decreases as the photons pass through the different head tissues. It can be observed that, although the fluence into the tumor is different for both head models, absorbed dose is the same.
Pacilio, M.; Lanconelli, N.; Lo Meo, S.; Betti, M.; Montani, L.; Torres Aroche, L. A.; Coca Perez, M. A. [Department of Medical Physics, Azienda Ospedaliera S. Camillo Forlanini, Piazza Forlanini 1, Rome 00151 (Italy); Department of Physics, Alma Mater Studiorum University of Bologna, Viale Berti-Pichat 6/2, Bologna 40127 (Italy); Department of Medical Physics, Azienda Ospedaliera S. Camillo Forlanini, Piazza Forlanini 1, Rome 00151 (Italy); Department of Medical Physics, Azienda Ospedaliera Sant' Andrea, Via di Grotarossa 1035, Rome 00189 (Italy); Department of Medical Physics, Center for Clinical Researches, Calle 34 North 4501, Havana 11300 (Cuba)
2009-05-15
Several updated Monte Carlo (MC) codes are available to perform calculations of voxel S values for radionuclide targeted therapy. The aim of this work is to analyze the differences in the calculations obtained by different MC codes and their impact on absorbed dose evaluations performed by voxel dosimetry. Voxel S values for monoenergetic sources (electrons and photons) and different radionuclides ({sup 90}Y, {sup 131}I, and {sup 188}Re) were calculated. Simulations were performed in soft tissue. Three general-purpose MC codes were employed for simulating radiation transport: MCNP4C, EGSnrc, and GEANT4. The data published by the MIRD Committee in Pamphlet No. 17, obtained with the EGS4 MC code, were also included in the comparisons. The impact of the differences (in terms of voxel S values) among the MC codes was also studied by convolution calculations of the absorbed dose in a volume of interest. For uniform activity distribution of a given radionuclide, dose calculations were performed on spherical and elliptical volumes, varying the mass from 1 to 500 g. For simulations with monochromatic sources, differences for self-irradiation voxel S values were mostly confined within 10% for both photons and electrons, but with electron energy less than 500 keV, the voxel S values referred to the first neighbor voxels showed large differences (up to 130%, with respect to EGSnrc) among the updated MC codes. For radionuclide simulations, noticeable differences arose in voxel S values, especially in the bremsstrahlung tails, or when a high contribution from electrons with energy of less than 500 keV is involved. In particular, for {sup 90}Y the updated codes showed a remarkable divergence in the bremsstrahlung region (up to about 90% in terms of voxel S values) with respect to the EGS4 code. Further, variations were observed up to about 30%, for small source-target voxel distances, when low-energy electrons cover an important part of the emission spectrum of the radionuclide
Training software using virtual-reality technology and pre-calculated effective dose data.
Ding, Aiping; Zhang, Di; Xu, X George
2009-05-01
This paper describes the development of a software package, called VR Dose Simulator, which aims to provide interactive radiation safety and ALARA training to radiation workers using virtual-reality (VR) simulations. Combined with a pre-calculated effective dose equivalent (EDE) database, a virtual radiation environment was constructed in VR authoring software, EON Studio, using 3-D models of a real nuclear power plant building. Models of avatars representing two workers were adopted with arms and legs of the avatar being controlled in the software to simulate walking and other postures. Collision detection algorithms were developed for various parts of the 3-D power plant building and avatars to confine the avatars to certain regions of the virtual environment. Ten different camera viewpoints were assigned to conveniently cover the entire virtual scenery in different viewing angles. A user can control the avatar to carry out radiological engineering tasks using two modes of avatar navigation. A user can also specify two types of radiation source: Cs and Co. The location of the avatar inside the virtual environment during the course of the avatar's movement is linked to the EDE database. The accumulative dose is calculated and displayed on the screen in real-time. Based on the final accumulated dose and the completion status of all virtual tasks, a score is given to evaluate the performance of the user. The paper concludes that VR-based simulation technologies are interactive and engaging, thus potentially useful in improving the quality of radiation safety training. The paper also summarizes several challenges: more streamlined data conversion, realistic avatar movement and posture, more intuitive implementation of the data communication between EON Studio and VB.NET, and more versatile utilization of EDE data such as a source near the body, etc., all of which needs to be addressed in future efforts to develop this type of software.
Investigation of the usability of conebeam CT data sets for dose calculation
Wilbert Jürgen
2008-12-01
Full Text Available Abstract Background To investigate the feasibility and accuracy of dose calculation in cone beam CT (CBCT data sets. Methods Kilovoltage CBCT images were acquired with the Elekta XVI system, CT studies generated with a conventional multi-slice CT scanner (Siemens Somatom Sensation Open served as reference images. Material specific volumes of interest (VOI were defined for commercial CT Phantoms (CATPhan® and Gammex RMI® and CT values were evaluated in CT and CBCT images. For CBCT imaging, the influence of image acquisition parameters such as tube voltage, with or without filter (F1 or F0 and collimation on the CT values was investigated. CBCT images of 33 patients (pelvis n = 11, thorax n = 11, head n = 11 were compared with corresponding planning CT studies. Dose distributions for three different treatment plans were calculated in CT and CBCT images and differences were evaluated. Four different correction strategies to match CT values (HU and density (D in CBCT images were analysed: standard CT HU-D table without adjustment for CBCT; phantom based HU-D tables; patient group based HU-D tables (pelvis, thorax, head; and patient specific HU-D tables. Results CT values in the CBCT images of the CATPhan® were highly variable depending on the image acquisition parameters: a mean difference of 564 HU ± 377 HU was calculated between CT values determined from the planning CT and CBCT images. Hence, two protocols were selected for CBCT imaging in the further part of the study and HU-D tables were always specific for these protocols (pelvis and thorax with M20F1 filter, 120 kV; head S10F0 no filter, 100 kV. For dose calculation in real patient CBCT images, the largest differences between CT and CBCT were observed for the standard CT HU-D table: differences were 8.0% ± 5.7%, 10.9% ± 6.8% and 14.5% ± 10.4% respectively for pelvis, thorax and head patients using clinical treatment plans. The use of patient and group based HU-D tables resulted in
Maerz, Manuel; Mittermair, Pia; Koelbl, Oliver; Dobler, Barbara [Regensburg University Medical Center, Department of Radiotherapy, Regensburg (Germany); Krauss, Andreas [Siemens Healthcare GmbH, Forchheim (Germany)
2016-06-15
Metallic dental implants cause severe streaking artifacts in computed tomography (CT) data, which affect the accuracy of dose calculations in radiation therapy. The aim of this study was to investigate the benefit of the metal artifact reduction algorithm iterative metal artifact reduction (iMAR) in terms of correct representation of Hounsfield units (HU) and dose calculation accuracy. Heterogeneous phantoms consisting of different types of tissue equivalent material surrounding metallic dental implants were designed. Artifact-containing CT data of the phantoms were corrected using iMAR. Corrected and uncorrected CT data were compared to synthetic CT data to evaluate accuracy of HU reproduction. Intensity-modulated radiation therapy (IMRT) and volumetric modulated arc therapy (VMAT) plans were calculated in Oncentra v4.3 on corrected and uncorrected CT data and compared to Gafchromic trademark EBT3 films to assess accuracy of dose calculation. The use of iMAR increased the accuracy of HU reproduction. The average deviation of HU decreased from 1006 HU to 408 HU in areas including metal and from 283 HU to 33 HU in tissue areas excluding metal. Dose calculation accuracy could be significantly improved for all phantoms and plans: The mean passing rate for gamma evaluation with 3 % dose tolerance and 3 mm distance to agreement increased from 90.6 % to 96.2 % if artifacts were corrected by iMAR. The application of iMAR allows metal artifacts to be removed to a great extent which leads to a significant increase in dose calculation accuracy. (orig.) [German] Metallische Implantate verursachen streifenfoermige Artefakte in CT-Bildern, welche die Dosisberechnung beeinflussen. In dieser Studie soll der Nutzen des iterativen Metall-Artefakt-Reduktions-Algorithmus iMAR hinsichtlich der Wiedergabetreue von Hounsfield-Werten (HU) und der Genauigkeit von Dosisberechnungen untersucht werden. Es wurden heterogene Phantome aus verschiedenen Arten gewebeaequivalenten Materials mit
Kim, Sejoong; Lee, Hyun-Woo
2006-05-01
A pioneering experiment [E. Schuster, E. Buks, M. Heiblum, D. Mahalu, V. Umansky, and Hadas Shtrikman, Nature 385, 417 (1997)] reported the measurement of the transmission phase of an electron traversing a quantum dot and found the intriguing feature of a sudden phase drop in the conductance valleys. Based on the Friedel sum rule for a spinless effective one-dimensional system, it has been previously argued [H.-W. Lee, Phys. Rev. Lett. 82, 2358 (1999)] that the sudden phase drop should be accompanied by the vanishing of the transmission amplitude, or transmission zero. Here we address roles of strong electron-electron interactions on the electron transport through a two-level quantum dot where one level couples with the leads much more strongly than the other level does [P. G. Silvestrov and Y. Imry, Phys. Rev. Lett. 85, 2565 (2000)]. We perform a perturbative conductance calculation with an explicit account of large charging energy and verify that the resulting conductance exhibits transmission zero, in agreement with the analysis based on the Friedel sum rule.
Scherpelz, R. I.; Borst, F. J.; Hoenes, G. R.
1980-12-01
WRAITH is a FORTRAN computer code which calculates the doses received by a standard man exposed to an accidental release of radioactive material. The movement of the released material through the atmosphere is calculated using a bivariate straight-line Gaussian distribution model, with Pasquill values for standard deviations. The quantity of material in the released cloud is modified during its transit time to account for radioactive decay and daughter production. External doses due to exposure to the cloud can be calculated using a semi-infinite cloud approximation. In situations where the semi-infinite cloud approximation is not a good one, the external dose can be calculated by a "finite plume" three-dimensional point-kernel numerical integration technique. Internal doses due to acute inhalation are cal.culated using the ICRP Task Group Lung Model and a four-segmented gastro-intestinal tract model. Translocation of the material between body compartments and retention in the body compartments are calculated using multiple exponential retention functions. Internal doses to each organ are calculated as sums of cross-organ doses, with each target organ irradiated by radioactive material in a number of source organs. All doses are calculated in rads, with separate values determined for high-LET and low-LET radiation.
Kang, Sei-Kwon; Yoon, Jai-Woong; Hwang, Taejin; Park, Soah; Cheong, Kwang-Ho; Jin Han, Tae; Kim, Haeyoung; Lee, Me-Yeon; Ju Kim, Kyoung, E-mail: kjkim@hallym.or.kr; Bae, Hoonsik
2015-10-01
A metallic contact eye shield has sometimes been used for eyelid treatment, but dose distribution has never been reported for a patient case. This study aimed to show the shield-incorporated CT-based dose distribution using the Pinnacle system and Monte Carlo (MC) calculation for 3 patient cases. For the artifact-free CT scan, an acrylic shield machined as the same size as that of the tungsten shield was used. For the MC calculation, BEAMnrc and DOSXYZnrc were used for the 6-MeV electron beam of the Varian 21EX, in which information for the tungsten, stainless steel, and aluminum material for the eye shield was used. The same plan was generated on the Pinnacle system and both were compared. The use of the acrylic shield produced clear CT images, enabling delineation of the regions of interest, and yielded CT-based dose calculation for the metallic shield. Both the MC and the Pinnacle systems showed a similar dose distribution downstream of the eye shield, reflecting the blocking effect of the metallic eye shield. The major difference between the MC and the Pinnacle results was the target eyelid dose upstream of the shield such that the Pinnacle system underestimated the dose by 19 to 28% and 11 to 18% for the maximum and the mean doses, respectively. The pattern of dose difference between the MC and the Pinnacle systems was similar to that in the previous phantom study. In conclusion, the metallic eye shield was successfully incorporated into the CT-based planning, and the accurate dose calculation requires MC simulation.
Dose calculations using MARS for Bremsstrahlung beam stops and collimators in APS beamline stations.
Dooling, J.; Accelerator Systems Division (APS)
2010-11-01
The Monte Carlo radiation transport code MARS is used to model the generation of gas bremsstrahlung (GB) radiation from 7-GeV electrons which scatter from residual gas atoms in undulator straight sections within the Advanced Photon Source (APS) storage ring. Additionally, MARS is employed to model the interactions of the GB radiation with components along the x-ray beamlines and then determine the expected radiation dose-rates that result. In this manner, MARS can be used to assess the adequacy of existing shielding or the specifications for new shielding when required. The GB radiation generated in the 'thin-target' of an ID straight section will consist only of photons in a 1/E-distribution up to the full energy of the stored electron beam. Using this analytical model, the predicted GB power for a typical APS 15.38-m insertion device (ID) straight section is 4.59 x 10{sup -7} W/nTorr/mA, assuming a background gas composed of air (Z{sub eff} = 7.31) at room temperature (293K). The total GB power provides a useful benchmark for comparisons between analytical and numerical approaches. We find good agreement between MARS and analytical estimates for total GB power. The extended straight section 'target' creates a radial profile of GB, which is highly peaked centered on the electron beam. The GB distribution reflects the size of the electron beam that creates the radiation. Optimizing the performance of MARS in terms of CPU time per incident trajectory requires the use of a relatively short, high-density gas target (air); in this report, the target density is {rho}L = 2.89 x 10{sup -2} g/cm{sup 2} over a length of 24 cm. MARS results are compared with the contact dose levels reported in TB-20, which used EGS4 for radiation transport simulations. Maximum dose-rates in 1 cc of tissue phantom form the initial basis for comparison. MARS and EGS4 results are approximately the same for maximum 1-cc dose-rates and attenuation in the photon
Biological shielding assessment and dose rate calculation for a neutron inspection portal
Donzella, A.; Bonomi, G.; Giroletti, E.; Zenoni, A.
2012-04-01
With reference to the prototype of neutron inspection portal built and successfully tested in the Rijeka seaport (Croatia) within the EURITRACK (EURopean Illicit Trafficking Countermeasures Kit) project, an assessment of the biological shielding in different set-up configurations of a future portal has been calculated with MCNP Monte Carlo code in the frame of the Eritr@C (European Riposte against Illicit TR@ffiCking) project. In the configurations analyzed the compliance with the dose limits for workers and the population stated by the European legislation is provided by appropriate shielding of the neutron sources and by the delimitation of a controlled area.
Nunes, J.C.
1991-01-01
This work quantifies the changes effected in electron absorbed dose to a soft-tissue equivalent medium when part of this medium is replaced by a material that is not soft-tissue equivalent. That is, heterogeneous dosimetry is addressed. Radionuclides which emit beta particles are the electron sources of primary interest. They are used in brachytherapy and in nuclear medicine: for example, beta-ray applicators made with strontium-90 are employed in certain ophthalmic treatments and iodine-131 is used to test thyroid function. More recent medical procedures under development and which involve beta radionuclides include radiommunotherapy and radiation synovectomy; the first is a cancer modality and the second deals with the treatment of rheumatoid arthritis. In addition, the possibility of skin surface contamination exists whenever there is handling of radioactive material. Determination of absorbed doses in the examples of the preceding paragraph requires considering boundaries of interfaces. Whilst the Monte Carlo method can be applied to boundary calculations, for routine work such as in clinical situations, or in other circumstances where doses need to be determined quickly, analytical dosimetry would be invaluable. Unfortunately, few analytical methods for boundary beta dosimetry exist. Furthermore, the accuracy of results from both Monte Carlo and analytical methods had to be assessed. Although restricted to one radionuclide, phosphorus-32, the experimental data obtained in this work serve several purposes, one of which is to provide standards against which calculated results can be tested. The experimental data may be useful in developing analytical boundary dosimetry methodology. The first application of the experimental data is demonstrated. Results from two Monte Carlo codes and two analytical methods are compared with experimental data. Monte Carlo results compare satisfactory with experimental results for the boundaries considered.
Deleuze, M.S.; Pickup, B.T.; Wilton, D.J.
2000-04-05
The authors present the theory of the electron propagator perturbed by an external electric field and show how it can be used to calculate a variety of one-electron linear response properties that are accurate through second order in electron correlation. Some illustrative calculations are discussed.
Application of GEANT4 radiation transport toolkit to dose calculations in anthropomorphic phantoms
Rodrigues, P; Peralta, L; Alves, C; Chaves, A; Lopes, M C
2003-01-01
In this paper we present the implementation of a dose calculation application, based on the GEANT4 Monte Carlo toolkit. Validation studies were performed with an homogeneous water phantom and an Alderson--Rando anthropomorphic phantom both irradiated with high--energy photon beams produced by a clinical linear accelerator. As input, this tool requires computer tomography images for automatic codification of voxel based geometries and phase space distributions to characterize the incident radiation field. Simulation results were compared with ionization chamber, thermoluminescent dosimetry data and commercial treatment planning system calculations. In homogeneous water phantom, overall agreement with measurements were within 1--2%. For anthropomorphic simulated setups (thorax and head irradiation) mean differences between GEANT4 and TLD measurements were less than 2%. Significant differences between GEANT4 and a semi--analytical algorithm implemented in the treatment planning system, were found in low density ...
A model of the circulating blood for use in radiation dose calculations
Hui, T.E.; Poston, J.W. Sr.
1987-01-01
Over the last few years there has been a significant increase in the use of radionuclides in leukocyte, platelet, and erythrocyte imaging procedures. Radiopharmaceutical used in these procedures are confined primarily to the blood, have short half-lives, and irradiate the body as they move through the circulatory system. There is a need for a model, to describe the circulatory system in an adult human, which can be used to provide radiation absorbed dose estimates for these procedures. A simplified model has been designed assuming a static circulatory system and including major organs of the body. The model has been incorporated into the MIRD phantom and calculations have been completed for a number of exposure situations and radionuclides of clinical importance. The model will be discussed in detail and results of calculations using this model will be presented.
A model of the circulating blood for use in radiation dose calculations
Hui, T.E.; Poston, J.W. Sr.
1987-12-31
Over the last few years there has been a significant increase in the use of radionuclides in leukocyte, platelet, and erythrocyte imaging procedures. Radiopharmaceutical used in these procedures are confined primarily to the blood, have short half-lives, and irradiate the body as they move through the circulatory system. There is a need for a model, to describe the circulatory system in an adult human, which can be used to provide radiation absorbed dose estimates for these procedures. A simplified model has been designed assuming a static circulatory system and including major organs of the body. The model has been incorporated into the MIRD phantom and calculations have been completed for a number of exposure situations and radionuclides of clinical importance. The model will be discussed in detail and results of calculations using this model will be presented.
Animesh
2005-01-01
Full Text Available The complexity of interactions and the nature of the approximations made in the formulation of the algorithm require that the user be familiar with the limitations of various models. As computer power keeps growing, calculation algorithms are tending more towards physically based models. The nature and quantity of the data required varies according to the model which may be either measurement based models or physical based models. Multiple dose calculation algorithm support found in XiO Treatment Planning System can be used to advantage when choice is to be made between speed and accuracy. Thus XiO allows end users generate plans accurately and quickly to optimize the delivery of radiation therapy.
Yepes, Pablo; Randeniya, Sharmalee; Taddei, Phillip J; Newhauser, Wayne D
2009-01-07
The Monte Carlo method is used to provide accurate dose estimates in proton radiation therapy research. While it is more accurate than commonly used analytical dose calculations, it is computationally intense. The aim of this work was to characterize for a clinical setup the fast dose calculator (FDC), a Monte Carlo track-repeating algorithm based on GEANT4. FDC was developed to increase computation speed without diminishing dosimetric accuracy. The algorithm used a database of proton trajectories in water to calculate the dose of protons in heterogeneous media. The extrapolation from water to 41 materials was achieved by scaling the proton range and the scattering angles. The scaling parameters were obtained by comparing GEANT4 dose distributions with those calculated with FDC for homogeneous phantoms. The FDC algorithm was tested by comparing dose distributions in a voxelized prostate cancer patient as calculated with well-known Monte Carlo codes (GEANT4 and MCNPX). The track-repeating approach reduced the CPU time required for a complete dose calculation in a voxelized patient anatomy by more than two orders of magnitude, while on average reproducing the results from the Monte Carlo predictions within 2% in terms of dose and within 1 mm in terms of distance.
Fotina, Irina; Kragl, Gabriele; Kroupa, Bernhard; Trausmuth, Robert; Georg, Dietmar [Medical Univ. Vienna (Austria). Division of Medical Radiation Physics, Dept. of Radiotherapy
2011-07-15
Comparison of the dosimetric accuracy of the enhanced collapsed cone (eCC) algorithm with the commercially available Monte Carlo (MC) dose calculation for complex treatment techniques. A total of 8 intensity-modulated radiotherapy (IMRT) and 2 stereotactic body radiotherapy (SBRT) lung cases were calculated with eCC and MC algorithms with the treatment planning systems (TPS) Oncentra MasterPlan 3.2 (Nucletron) and Monaco 2.01 (Elekta/CMS). Fluence optimization as well as sequencing of IMRT plans was primarily performed using Monaco. Dose prediction errors were calculated using MC as reference. The dose-volume histrogram (DVH) analysis was complemented with 2D and 3D gamma evaluation. Both algorithms were compared to measurements using the Delta4 system (Scandidos). Recalculated with eCC IMRT plans resulted in lower planned target volume (PTV) coverage, as well as in lower organs-at-risk (OAR) doses up to 8%. Small deviations between MC and eCC in PTV dose (1-2%) were detected for IMRT cases, while larger deviations were observed for SBRT (up to 5%). Conformity indices of both calculations were similar; however, the homogeneity of the eCC calculated plans was slightly better. Delta4 measurements confirmed high dosimetric accuracy of both TPS. Mean dose prediction errors < 3% for PTV suggest that both algorithms enable highly accurate dose calculations under clinical conditions. However, users should be aware of slightly underestimated OAR doses using the eCC algorithm. (orig.)
CALCULATION STUDIES OF SPATIAL DISTRIBUTION OF THE ABSORBED DOSE RATE FOR VARIOUS SEEDS
N. A. Nerozin
2015-01-01
Full Text Available Purpose. Conducting computational studies of dosimetric characteristics of microsources with the radionuclide I‑125, pilot production of which is established in the research and production complex of isotope and radiopharmaceuticals, JSC “State Scientific Centre of the Russian Federation — Institute for Physics and Power Engineering named after A. I. Leypunsky” (SSC RF IPPE. Sources of production IPPE are similar to the model 6711 of the company Nicomed Amersham, dosimetric characteristics of which are standardized in accordance with the TG43 AAPM formalism.Materials and methods. Microsourse «SEED No. 6711» (model of the company Nicomed Amersham is hermetically sealed in a titanium capsule silver rod covered with a thin layer of radioactive I‑125. The half-life of iodine‑125 is 59,43 days. In the process of decay of I‑125 is converted into the Te‑125.Calculation of parameters of microsources and their comparison with the standard model 6711 is carried out with use of the computer code MCNP.Results. The method of calculation of the basic dosimetric characteristics of the microsourse SSC RF-IPPE in accordance with the TG43 formalism is developed. A comparative analysis of experimental data and calculated results by MCNP code, which allowed to identify possible reasons for differences, is performed. The estimated dose characteristics and recommended standard data for dose characteristics of micro «SEED No. 6711» are compared.Conclusions. There are two possible reasons for the differences between experimental and calculated results. The first one may be the roughness of the surface of a silver rod or diffusion of radioactive iodine in silver. The second reason might be the difference of the cross sections of the characteristic radiation of silver used in MCNP code. In the comparison of calculated dose characteristics and recommended standard the role of the application activity is very important. In compliance with the standard
Electron Beam Dose Distribution in the Presence of Non-Uniform Magnetic Field
Mohamad Javad Tahmasebi-Birgani
2014-04-01
Full Text Available Introduction Magnetic fields are capable of altering the trajectory of electron beams andcan be used in radiation therapy.Theaim of this study was to produce regions with dose enhancement and reduction in the medium. Materials and Methods The NdFeB permanent magnets were arranged on the electron applicator in several configurations. Then, after the passage of the electron beams (9 and 15 MeV Varian 2100C/D through the non-uniform magnetic field, the Percentage Depth Dose(PDDs on central axis and dose profiles in three depths for each energy were measured in a 3D water phantom. Results For all magnet arrangements and for two different energies, the surface dose increment and shift in depth of maximum dose (dmax were observed. In addition, the pattern of dose distribution in buildup region was changed. Measurement of dose profile showed dose localization and spreading in some other regions. Conclusion The results of this study confirms that using magnetic field can alter the dose deposition patterns and as a result can produce dose enhancement as well as dose reduction in the medium using high-energy electron beams. These effects provide dose distribution with arbitrary shapes for use in radiation therapy.
Petillion, Saskia; Swinnen, Ans; Defraene, Gilles; Verhoeven, Karolien; Weltens, Caroline; Van den Heuvel, Frank
2014-07-08
The comparison of the pencil beam dose calculation algorithm with modified Batho heterogeneity correction (PBC-MB) and the analytical anisotropic algorithm (AAA) and the mutual comparison of advanced dose calculation algorithms used in breast radiotherapy have focused on the differences between the physical dose distributions. Studies on the radiobiological impact of the algorithm (both on the tumor control and the moderate breast fibrosis prediction) are lacking. We, therefore, investigated the radiobiological impact of the dose calculation algorithm in whole breast radiotherapy. The clinical dose distributions of 30 breast cancer patients, calculated with PBC-MB, were recalculated with fixed monitor units using more advanced algorithms: AAA and Acuros XB. For the latter, both dose reporting modes were used (i.e., dose-to-medium and dose-to-water). Next, the tumor control probability (TCP) and the normal tissue complication probability (NTCP) of each dose distribution were calculated with the Poisson model and with the relative seriality model, respectively. The endpoint for the NTCP calculation was moderate breast fibrosis five years post treatment. The differences were checked for significance with the paired t-test. The more advanced algorithms predicted a significantly lower TCP and NTCP of moderate breast fibrosis then found during the corresponding clinical follow-up study based on PBC calculations. The differences varied between 1% and 2.1% for the TCP and between 2.9% and 5.5% for the NTCP of moderate breast fibrosis. The significant differences were eliminated by determination of algorithm-specific model parameters using least square fitting. Application of the new parameters on a second group of 30 breast cancer patients proved their appropriateness. In this study, we assessed the impact of the dose calculation algorithms used in whole breast radiotherapy on the parameters of the radiobiological models. The radiobiological impact was eliminated by
ZHOU Chun-Mei; WU Zhen-Dong; HUANG Xiao-Long
2005-01-01
Calculations of energies and absolute intensities of Auger electron and X-ray arising from electron capture are introduced briefly. The calculation codes and main process are also presented. The application is also given by taking 55Fe ε decay as an example.
Critical groups vs. representative person: dose calculations due to predicted releases from USEXA
Ferreira, N.L.D., E-mail: nelson.luiz@ctmsp.mar.mil.br [Centro Tecnologico da Marinha (CTM/SP), Sao Paulo, SP (Brazil); Rochedo, E.R.R., E-mail: elainerochedo@gmail.com [Instituto de Radiprotecao e Dosimetria (lRD/CNEN-RJ), Rio de Janeiro, RJ (Brazil); Mazzilli, B.P., E-mail: mazzilli@ipen.br [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2013-07-01
The critical group cf Centro Experimental Aramar (CEA) site was previously defined based 00 the effluents releases to the environment resulting from the facilities already operational at CEA. In this work, effective doses are calculated to members of the critical group considering the predicted potential uranium releases from the Uranium Hexafluoride Production Plant (USEXA). Basically, this work studies the behavior of the resulting doses related to the type of habit data used in the analysis and two distinct situations are considered: (a) the utilization of average values obtained from official institutions (IBGE, IEA-SP, CNEN, IAEA) and from the literature; and (b) the utilization of the 95{sup tb} percentile of the values derived from distributions fit to the obtained habit data. The first option corresponds to the way that data was used for the definition of the critical group of CEA done in former assessments, while the second one corresponds to the use of data in deterministic assessments, as recommended by ICRP to estimate doses to the so--called 'representative person' . (author)
Jingyu Zhang
2016-01-01
Full Text Available In water-cooled reactor, the dominant radioactive source term under normal operation is activated corrosion products (ACPs, which have an important impact on reactor inspection and maintenance. A three-node transport model of ACPs was introduced into the new version of ACPs source term code CATE in this paper, which makes CATE capable of theoretically simulating the variation and the distribution of ACPs in a water-cooled reactor and suitable for more operating conditions. For code testing, MIT PWR coolant chemistry loop was simulated, and the calculation results from CATE are close to the experimental results from MIT, which means CATE is available and credible on ACPs analysis of water-cooled reactor. Then ACPs in the blanket cooling loop of water-cooled fusion reactor ITER under construction were analyzed using CATE and the results showed that the major contributors are the short-life nuclides, especially Mn-56. At last a point kernel integration code ARShield was coupled with CATE, and the dose rate around ITER blanket cooling loop was calculated. Results showed that after shutting down the reactor only for 8 days, the dose rate decreased nearly one order of magnitude, which was caused by the rapid decay of the short-life ACPs.
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources
Townson, Reid; Tian, Zhen; Graves, Yan Jiang; Zavgorodni, Sergei; Jiang, Steve B
2013-01-01
A novel phase-space source implementation has been designed for GPU-based Monte Carlo dose calculation engines. Due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel strategy to pre-process patient-independent phase-spaces and bin particles by type, energy and position. Position bins l...
A Comparison of Model Calculation and Measurement of Absorbed Dose for Proton Irradiation. Chapter 5
Zapp, N.; Semones, E.; Saganti, P.; Cucinotta, F.
2003-01-01
With the increase in the amount of time spent EVA that is necessary to complete the construction and subsequent maintenance of ISS, it will become increasingly important for ground support personnel to accurately characterize the radiation exposures incurred by EVA crewmembers. Since exposure measurements cannot be taken within the organs of interest, it is necessary to estimate these exposures by calculation. To validate the methods and tools used to develop these estimates, it is necessary to model experiments performed in a controlled environment. This work is such an effort. A human phantom was outfitted with detector equipment and then placed in American EMU and Orlan-M EVA space suits. The suited phantom was irradiated at the LLUPTF with proton beams of known energies. Absorbed dose measurements were made by the spaceflight operational dosimetrist from JSC at multiple sites in the skin, eye, brain, stomach, and small intestine locations in the phantom. These exposures are then modeled using the BRYNTRN radiation transport code developed at the NASA Langley Research Center, and the CAM (computerized anatomical male) human geometry model of Billings and Yucker. Comparisons of absorbed dose calculations with measurements show excellent agreement. This suggests that there is reason to be confident in the ability of both the transport code and the human body model to estimate proton exposure in ground-based laboratory experiments.
Dose calculation algorithm accuracy for small fields in non-homogeneous media: The lung SBRT case.
Fogliata, Antonella; Cozzi, Luca
2016-11-24
This review addresses the theme of dose calculation accuracy in the case of the stereotactic treatment of lung lesions. Based on the classical categories of type "a", "b" and "c" algorithms (according to their degree of complexity in the management of charged particle transport), a summary of findings from literature is reported. Two main critical areas have been identified: the use of small fields and the presence of low density medium. Concerning the latter point, the algorithm accuracy is intrinsic of the algorithm core, and, notwithstanding the materials discretization and their chemical composition knowledge, type "c" are, at the most, able to reproduce the actual physical dose distribution in heterogeneous media. For what concerns the small field management, the final accuracy could be strongly related to the beam configuration appropriateness in the TPS (as well for MC this relates to the proper linac head description). As a very crude summary, type "a" should be considered as unsuitable for this kind of treatment calculations (with differences of the order of 20-30%), while type "b" and "c" could keep their accuracy approximatively within 10 and 5%, respectively. Copyright Â© 2016. Published by Elsevier Ltd.
A GPU-based Monte Carlo dose calculation code for photon transport in a voxel phantom
Bellezzo, M.; Do Nascimento, E.; Yoriyaz, H., E-mail: mbellezzo@gmail.br [Instituto de Pesquisas Energeticas e Nucleares / CNEN, Av. Lineu Prestes 2242, Cidade Universitaria, 05508-000 Sao Paulo (Brazil)
2014-08-15
As the most accurate method to estimate absorbed dose in radiotherapy, Monte Carlo method has been widely used in radiotherapy treatment planning. Nevertheless, its efficiency can be improved for clinical routine applications. In this paper, we present the CUBMC code, a GPU-based Mc photon transport algorithm for dose calculation under the Compute Unified Device Architecture platform. The simulation of physical events is based on the algorithm used in Penelope, and the cross section table used is the one generated by the Material routine, als present in Penelope code. Photons are transported in voxel-based geometries with different compositions. To demonstrate the capabilities of the algorithm developed in the present work four 128 x 128 x 128 voxel phantoms have been considered. One of them is composed by a homogeneous water-based media, the second is composed by bone, the third is composed by lung and the fourth is composed by a heterogeneous bone and vacuum geometry. Simulations were done considering a 6 MeV monoenergetic photon point source. There are two distinct approaches that were used for transport simulation. The first of them forces the photon to stop at every voxel frontier, the second one is the Woodcock method, where the photon stop in the frontier will be considered depending on the material changing across the photon travel line. Dose calculations using these methods are compared for validation with Penelope and MCNP5 codes. Speed-up factors are compared using a NVidia GTX 560-Ti GPU card against a 2.27 GHz Intel Xeon CPU processor. (Author)
Dose conversion coefficients for high-energy photons, electrons, neutrons and protons
Sakamoto, Y; Sato, O; Tanaka, S I; Tsuda, S; Yamaguchi, Y; Yoshizawa, N
2003-01-01
In the International Commission on Radiological Protection (ICRP) 1990 Recommendations, radiation weighting factors were introduced in the place of quality factors, the tissue weighting factors were revised, and effective doses and equivalent doses of each tissues and organs were defined as the protection quantities. Dose conversion coefficients for photons, electrons and neutrons based on new ICRP recommendations were cited in the ICRP Publication 74, but the energy ranges of theses data were limited and there are no data for high energy radiations produced in accelerator facilities. For the purpose of designing the high intensity proton accelerator facilities at JAERI, the dose evaluation code system of high energy radiations based on the HERMES code was developed and the dose conversion coefficients of effective dose were evaluated for photons, neutrons and protons up to 10 GeV, and electrons up to 100 GeV. The dose conversion coefficients of effective dose equivalent were also evaluated using quality fact...
Herraiz Lblanca, M. D.; Diaz Romero, F.; Casares Magaz, O.; Garrido Breton, C.; Catalan Acosta, A.; Hernandez Armas, J.
2013-07-01
This paper proposes a method that allows you to calculate the effective dose in any interventional radiology procedure using an anthropomorphic mannequin Alderson RANDO and dosimeters TLD 100 chip. This method has been applied to an angio Radiology procedure: the biliary drainage. The objectives that have been proposed are: to) put together a method that, on an experimental basis, allows to know dosis en organs to calculate effective dose in complex procedures and b) apply the method to the calculation of the effective dose of biliary drainage. (Author)
Element-specific and constant parameters used for dose calculations in SR-Site
Norden, Sara (Svensk Kaernbraenslehantering AB (Sweden)); Avila, Rodolfo; De la Cruz, Idalmis; Stenberg, Kristofer; Grolander, Sara (Facilia AB (Sweden))
2010-12-15
The report presents Best Estimate (BE) values and Probability Distribution Functions (PDFs) of Concentration Ratios (CR) for different types of terrestrial and aquatic biota and distribution coefficients (K{sub d}) for organic and inorganic deposits, as well as for suspended matter in freshwater and marine ecosystems. The BE values have been used in deterministic simulations for derivation of Landscape Dose Factors (LDF) applied for dose assessments in SR-Site. The PDFs have been used in probabilistic simulations for uncertainty and sensitivity analysis of the LDFs. The derivation of LDFs for SR-Site is described in /Avila et al. 2010/. The CR and K{sub d} values have been derived using both site-specific data measured at Laxemar and Forsmark during the site investigation program and literature data. These two data sources have been combined using Bayesian updating methods, which are described in detail in an Appendix, along with the input data used in the statistical analyses and the results obtained. The report also describes a kinetic-allometric model that was applied for deriving values of CR for terrestrial herbivores in cases when site and literature data for an element were missing. In addition, the report presents values for a number of other parameters used in the SR-Site Radionuclide Model for the biosphere: radionuclide decay-ingrowth data, elemental diffusivities, fractions of element content released during decomposition processes, ingestion of food, water and soil by cattle, elements retention fraction on plant surfaces during irrigation. The report also presents parameter values used in calculation of doses to a reference man: dose coefficients for inhalation, ingestion and external exposure, inhalation rates, ingestion rates of food and water
Grassberger, C; Daartz, J; Dowdell, S; Ruggieri, T; Sharp, G; Paganetti, H [Massachusetts General Hospital and Harvard Medical School, Boston, MA (United States)
2014-06-15
Purpose: Evaluate Monte Carlo (MC) dose calculation and the prediction of the treatment planning system (TPS) in a lung phantom and compare them in a cohort of 20 lung patients treated with protons. Methods: A 2-dimensional array of ionization chambers was used to evaluate the dose across the target in a lung phantom. 20 lung cancer patients on clinical trials were re-simulated using a validated Monte Carlo toolkit (TOPAS) and compared to the TPS. Results: MC increases dose calculation accuracy in lung compared to the clinical TPS significantly and predicts the dose to the target in the phantom within ±2%: the average difference between measured and predicted dose in a plane through the center of the target is 5.6% for the TPS and 1.6% for MC. MC recalculations in patients show a mean dose to the clinical target volume on average 3.4% lower than the TPS, exceeding 5% for small fields. The lower dose correlates significantly with aperture size and the distance of the tumor to the chest wall (Spearman's p=0.0002/0.004). For large tumors MC also predicts consistently higher V{sub 5} and V{sub 10} to the normal lung, due to a wider lateral penumbra, which was also observed experimentally. Critical structures located distal to the target can show large deviations, though this effect is very patient-specific. Conclusion: Advanced dose calculation techniques, such as MC, would improve treatment quality in proton therapy for lung cancer by avoiding systematic overestimation of target dose and underestimation of dose to normal lung. This would increase the accuracy of the relationships between dose and effect, concerning tumor control as well as normal tissue toxicity. As the role of proton therapy in the treatment of lung cancer continues to be evaluated in clinical trials, this is of ever-increasing importance. This work was supported by National Cancer Institute Grant R01CA111590.
Yamamoto, Hideo; Yoneda, Tarou; Satou, Shuji; Ishikawa, Toru; Hara, Misako
2009-12-20
By input of the actual dose of a drug given into a radiology information system, the system converting with an accounting system into a cost of the drug from the actual dose in the electronic medical record was built. In the drug master, the first unit was set as the cost of the drug, and we set the second unit as the actual dose. The second unit in the radiology information system was received by the accounting system through electronic medical record. In the accounting system, the actual dose was changed into the cost of the drug using the dose of conversion to the first unit. The actual dose was recorded on a radiology information system and electronic medical record. The actual dose was indicated on the accounting system, and the cost for the drug was calculated. About the actual dose of drug, cooperation of the information in a radiology information system and electronic medical record were completed. It was possible to decide the volume of drug from the correct dose of drug at the previous inspection. If it is necessary for the patient to have another treatment of medicine, it is important to know the actual dose of drug given. Moreover, authenticity of electronic medical record based on a statute has also improved.
Quasiparticle GW calculations within the GPAW electronic structure code
Hüser, Falco
The GPAW electronic structure code, developed at the physics department at the Technical University of Denmark, is used today by researchers all over the world to model the structural, electronic, optical and chemical properties of materials. They address fundamental questions in material science...... and use their knowledge to design new materials for a vast range of applications. Todays hottest topics are, amongst many others, better materials for energy conversion (e.g. solar cells), energy storage (batteries) and catalysts for the removal of environmentally dangerous exhausts. The mentioned...... properties are to a large extent governed by the physics on the atomic scale, that means pure quantum mechanics. For many decades, Density Functional Theory has been the computational method of choice, since it provides a fairly easy and yet accurate way of determining electronic structures and related...
Jacob, Dayee; Lamberto, Melissa; DeSouza Lawrence, Lana; Mourtada, Firas
To retrospectively compare clinical dosimetry of CT-based tandem-ring treatment plans using a model-based dose calculation algorithm (MBDCA) with the standard TG-43-based dose formalism. A cohort of 10 cervical cancer cohorts treated using the tandem and ring high-dose-rate applicators were evaluated. The original treatment plans were created using the department CT-based volume optimization clinical standards. All plans originally calculated with TG-43 dose calculation formalism were recalculated using the MBDCA algorithm. The gross target volume and organs at risk (OARs) were contoured on each data set along with significant heterogeneities like air in cavity and high-density plastic tandem and ring components. The patient tissue was modeled as homogenous liquid water. D90, D95, and D100 for gross target volume, D0.1cm(3), D1.0cm(3), and D2.0cm(3) for bladder, rectum, and sigmoid were extracted from dose-volume histograms for TG-43 and MBDCA calculated plans. Mean absolute difference ± 2σ in the above metrics was calculated for each plan. Using the manual applicator contouring method, MBDCA plans (n = 10) showed 2.1 ± 1.1% reduction in dose to Point A average, 2.6 ± 0.9% reduction in Target D90 dose, and 2.1 ± 0.3% dose reduction to OARs. Results from plans using vendor supplied solid applicator models (n = 5) showed 2.2 ± 1.10% reduction in dose to Point A average, 2.7 ± 0.2% reduction in Target D90 dose, and 2.7 ± 1.0% dose reduction on average to OARs. For unshielded plastic gynecologic applicators, minimal dosimetric changes (<5%) were found using MBDCA relative to standard TG-43. Use of solid applicator model is more efficient than manual applicator contouring and also yielded similar MBDCA dosimetric results. Currently, TG-186 dose calculations should be reported along TG-43 until we obtain studies with larger cohorts to fully realize the potential of MBDCA dosimetry. Copyright © 2017 American Brachytherapy Society. Published by
Mitrikas, V G
2015-01-01
Monitoring of the radiation loading on cosmonauts requires calculation of absorbed dose dynamics with regard to the stay of cosmonauts in specific compartments of the space vehicle that differ in shielding properties and lack means of radiation measurement. The paper discusses different aspects of calculation modeling of radiation effects on human body organs and tissues and reviews the effective dose estimates for cosmonauts working in one or another compartment over the previous period of the International space station operation. It was demonstrated that doses measured by a real or personal dosimeters can be used to calculate effective dose values. Correct estimation of accumulated effective dose can be ensured by consideration for time course of the space radiation quality factor.
Krstic, D; Nikezic, D
2009-10-01
In this paper the effective dose in the age-dependent ORNL phantoms series, due to naturally occurring radionuclides in building materials, was calculated. The absorbed doses for various organs or human tissues have been calculated. The MCNP-4B computer code was used for this purpose. The effective dose was calculated according to ICRP Publication 74. The obtained values of dose conversion factors for a standard room are: 1.033, 0.752 and 0.0538 nSv h-1 per Bq kg-1 for elements of the U and Th decay series and for the K isotope, respectively. The values of effective dose agreed generally with those found in the literature, although the values estimated here for elements of the U series were higher in some cases.
Walters, Robert; Summers, Geoffrey P.; Warmer. Keffreu J/; Messenger, Scott; Lorentzen, Justin R.; Morton, Thomas; Taylor, Stephen J.; Evans, Hugh; Heynderickx, Daniel; Lei, Fan
2007-01-01
This paper presents a method for using the SPENVIS on-line computational suite to implement the displacement damage dose (D(sub d)) methodology for calculating end-of-life (EOL) solar cell performance for a specific space mission. This paper builds on our previous work that has validated the D(sub d) methodology against both measured space data [1,2] and calculations performed using the equivalent fluence methodology developed by NASA JPL [3]. For several years, the space solar community has considered general implementation of the D(sub d) method, but no computer program exists to enable this implementation. In a collaborative effort, NRL, NASA and OAI have produced the Solar Array Verification and Analysis Tool (SAVANT) under NASA funding, but this program has not progressed beyond the beta-stage [4]. The SPENVIS suite with the Multi Layered Shielding Simulation Software (MULASSIS) contains all of the necessary components to implement the Dd methodology in a format complementary to that of SAVANT [5]. NRL is currently working with ESA and BIRA to include the Dd method of solar cell EOL calculations as an integral part of SPENVIS. This paper describes how this can be accomplished.
Gibbons, John P., E-mail: john.gibbons@marybird.com [Department of Physics, Mary Bird Perkins Cancer Center, Baton Rouge, Louisiana 70809 (United States); Antolak, John A. [Department of Radiation Oncology, Mayo Clinic, Rochester, Minnesota 55905 (United States); Followill, David S. [Department of Radiation Physics, UT M.D. Anderson Cancer Center, Houston, Texas 77030 (United States); Huq, M. Saiful [Department of Radiation Oncology, University of Pittsburgh Cancer Institute, Pittsburgh, Pennsylvania 15232 (United States); Klein, Eric E. [Department of Radiation Oncology, Washington University School of Medicine, St. Louis, Missouri 63110 (United States); Lam, Kwok L. [Department of Radiation Oncology, University of Michigan, Ann Arbor, Michigan 48109 (United States); Palta, Jatinder R. [Department of Radiation Oncology, Virginia Commonwealth University, Richmond, Virginia 23298 (United States); Roback, Donald M. [Department of Radiation Oncology, Cancer Centers of North Carolina, Raleigh, North Carolina 27607 (United States); Reid, Mark [Department of Medical Physics, Fletcher-Allen Health Care, Burlington, Vermont 05401 (United States); Khan, Faiz M. [Department of Radiation Oncology, University of Minnesota, Minneapolis, Minnesota 55455 (United States)
2014-03-15
A protocol is presented for the calculation of monitor units (MU) for photon and electron beams, delivered with and without beam modifiers, for constant source-surface distance (SSD) and source-axis distance (SAD) setups. This protocol was written by Task Group 71 of the Therapy Physics Committee of the American Association of Physicists in Medicine (AAPM) and has been formally approved by the AAPM for clinical use. The protocol defines the nomenclature for the dosimetric quantities used in these calculations, along with instructions for their determination and measurement. Calculations are made using the dose per MU under normalization conditions, D{sub 0}{sup ′}, that is determined for each user's photon and electron beams. For electron beams, the depth of normalization is taken to be the depth of maximum dose along the central axis for the same field incident on a water phantom at the same SSD, where D{sub 0}{sup ′} = 1 cGy/MU. For photon beams, this task group recommends that a normalization depth of 10 cm be selected, where an energy-dependent D{sub 0}{sup ′} ≤ 1 cGy/MU is required. This recommendation differs from the more common approach of a normalization depth of d{sub m}, with D{sub 0}{sup ′} = 1 cGy/MU, although both systems are acceptable within the current protocol. For photon beams, the formalism includes the use of blocked fields, physical or dynamic wedges, and (static) multileaf collimation. No formalism is provided for intensity modulated radiation therapy calculations, although some general considerations and a review of current calculation techniques are included. For electron beams, the formalism provides for calculations at the standard and extended SSDs using either an effective SSD or an air-gap correction factor. Example tables and problems are included to illustrate the basic concepts within the presented formalism.
Gibbons, John P; Antolak, John A; Followill, David S; Huq, M Saiful; Klein, Eric E; Lam, Kwok L; Palta, Jatinder R; Roback, Donald M; Reid, Mark; Khan, Faiz M
2014-03-01
A protocol is presented for the calculation of monitor units (MU) for photon and electron beams, delivered with and without beam modifiers, for constant source-surface distance (SSD) and source-axis distance (SAD) setups. This protocol was written by Task Group 71 of the Therapy Physics Committee of the American Association of Physicists in Medicine (AAPM) and has been formally approved by the AAPM for clinical use. The protocol defines the nomenclature for the dosimetric quantities used in these calculations, along with instructions for their determination and measurement. Calculations are made using the dose per MU under normalization conditions, D'0, that is determined for each user's photon and electron beams. For electron beams, the depth of normalization is taken to be the depth of maximum dose along the central axis for the same field incident on a water phantom at the same SSD, where D'0 = 1 cGy/MU. For photon beams, this task group recommends that a normalization depth of 10 cm be selected, where an energy-dependent D'0 ≤ 1 cGy/MU is required. This recommendation differs from the more common approach of a normalization depth of dm, with D'0 = 1 cGy/MU, although both systems are acceptable within the current protocol. For photon beams, the formalism includes the use of blocked fields, physical or dynamic wedges, and (static) multileaf collimation. No formalism is provided for intensity modulated radiation therapy calculations, although some general considerations and a review of current calculation techniques are included. For electron beams, the formalism provides for calculations at the standard and extended SSDs using either an effective SSD or an air-gap correction factor. Example tables and problems are included to illustrate the basic concepts within the presented formalism.
Zahra Anjomani
2011-03-01
Full Text Available Introduction: Nowadays new radiochromic films have an essential role in radiotherapy dosimetry. Properties such as high sensitivity, good reproducibility, high spatial resolution, easy readout and portability have made them attractive for dosimetry, especially in high-dose-gradient regions. Material and Methods: In this study, electron-beam dose distributions in homogenous and heterogeneous phantoms were calculated using the MCNPX Monte Carlo code and compared with experimental measurements obtained by GAFCHROMIC® EBT film and p-type silicon diode dosimetry. Irradiation was carried out using an Elekta linear accelerator at two different electron energies (8 and 15 MeV, with a 10×10 cm2 applicator and at 100 cm source-to-surface distance. Results: The results show good agreement (within 2% between radiochromic film measurements and MCNP results. Conclusions: The results show that the new radiochromic films can be used in electron dosimetry and that they are also reliable in presence of heterogeneous media.
Zhang, J; Zhang, W; Lu, J [Cancer Hospital of Shantou University Medical College, Shantou, Guangdong (China)
2015-06-15
Purpose: To investigate the accuracy and feasibility of dose calculations using kilovoltage cone beam computed tomography in cervical cancer radiotherapy using a correction algorithm. Methods: The Hounsfield units (HU) and electron density (HU-density) curve was obtained for both planning CT (pCT) and kilovoltage cone beam CT (CBCT) using a CIRS-062 calibration phantom. The pCT and kV-CBCT images have different HU values, and if the HU-density curve of CBCT was directly used to calculate dose in CBCT images may have a deviation on dose distribution. It is necessary to normalize the different HU values between pCT and CBCT. A HU correction algorithm was used for CBCT images (cCBCT). Fifteen intensity-modulated radiation therapy (IMRT) plans of cervical cancer were chosen, and the plans were transferred to the pCT and cCBCT data sets without any changes for dose calculations. Phantom and patient studies were carried out. The dose differences and dose distributions were compared between cCBCT plan and pCT plan. Results: The HU number of CBCT was measured by several times, and the maximum change was less than 2%. To compare with pCT, the CBCT and cCBCT has a discrepancy, the dose differences in CBCT and cCBCT images were 2.48%±0.65% (range: 1.3%∼3.8%) and 0.48%±0.21% (range: 0.1%∼0.82%) for phantom study, respectively. For dose calculation in patient images, the dose differences were 2.25%±0.43% (range: 1.4%∼3.4%) and 0.63%±0.35% (range: 0.13%∼0.97%), respectively. And for the dose distributions, the passing rate of cCBCT was higher than the CBCTs. Conclusion: The CBCT image for dose calculation is feasible in cervical cancer radiotherapy, and the correction algorithm offers acceptable accuracy. It will become a useful tool for adaptive radiation therapy.
Sampson, Andrew; Le Yi; Williamson, Jeffrey F. [Department of Radiation Oncology, Virginia Commonwealth University, Richmond, Virginia 23298 (United States)
2012-02-15
Purpose: To demonstrate potential of correlated sampling Monte Carlo (CMC) simulation to improve the calculation efficiency for permanent seed brachytherapy (PSB) implants without loss of accuracy. Methods: CMC was implemented within an in-house MC code family (PTRAN) and used to compute 3D dose distributions for two patient cases: a clinical PSB postimplant prostate CT imaging study and a simulated post lumpectomy breast PSB implant planned on a screening dedicated breast cone-beam CT patient exam. CMC tallies the dose difference, {Delta}D, between highly correlated histories in homogeneous and heterogeneous geometries. The heterogeneous geometry histories were derived from photon collisions sampled in a geometrically identical but purely homogeneous medium geometry, by altering their particle weights to correct for bias. The prostate case consisted of 78 Model-6711 {sup 125}I seeds. The breast case consisted of 87 Model-200 {sup 103}Pd seeds embedded around a simulated lumpectomy cavity. Systematic and random errors in CMC were unfolded using low-uncertainty uncorrelated MC (UMC) as the benchmark. CMC efficiency gains, relative to UMC, were computed for all voxels, and the mean was classified in regions that received minimum doses greater than 20%, 50%, and 90% of D{sub 90}, as well as for various anatomical regions. Results: Systematic errors in CMC relative to UMC were less than 0.6% for 99% of the voxels and 0.04% for 100% of the voxels for the prostate and breast cases, respectively. For a 1 x 1 x 1 mm{sup 3} dose grid, efficiency gains were realized in all structures with 38.1- and 59.8-fold average gains within the prostate and breast clinical target volumes (CTVs), respectively. Greater than 99% of the voxels within the prostate and breast CTVs experienced an efficiency gain. Additionally, it was shown that efficiency losses were confined to low dose regions while the largest gains were located where little difference exists between the homogeneous and
Soldat, J.K.
1989-10-01
This report compares the results of the calculation of potential radiation doses to the public by two different environmental dosimetric systems for the years 1983 through 1987. Both systems project the environmental movement of radionuclides released with effluents from Hanford operations; their concentrations in air, water, and foods; the intake of radionuclides by ingestion and inhalation; and, finally, the potential radiation doses from radionuclides deposited in the body and from external sources. The first system, in use for the past decade at Hanford, calculates radiation doses in terms of 50-year cumulative dose equivalents to body organs and to the whole body, based on the methodology defined in ICRP Publication 2. This system uses a suite of three computer codes: PABLM, DACRIN, and KRONIC. In the new system, 50-year committed doses are calculated in accordance with the recommendations of the ICRP Publications 26 and 30, which were adopted by the US Department of Energy (DOE) in 1985. This new system calculates dose equivalent (DE) to individual organs and effective dose equivalent (EDE). The EDE is a risk-weighted DE that is designed to be an indicator of the potential health effects arising from the radiation dose. 16 refs., 1 fig., 38 tabs.