Liquid chromatography/Fourier transform IR spectrometry interface flow cell
Johnson, Charles C.; Taylor, Larry T.
1986-01-01
A zero dead volume (ZDV) microbore high performance liquid chromatography (.mu.HPLC)/Fourier transform infrared (FTIR) interface flow cell includes an IR transparent crystal having a small diameter bore therein through which a sample liquid is passed. The interface flow cell further includes a metal holder in combination with a pair of inner, compressible seals for directly coupling the thus configured spectrometric flow cell to the outlet of a .mu.HPLC column end fitting to minimize the transfer volume of the effluents exiting the .mu.HPLC column which exhibit excellent flow characteristics due to the essentially unencumbered, open-flow design. The IR beam passes transverse to the sample flow through the circular bore within the IR transparent crystal, which is preferably comprised of potassium bromide (KBr) or calcium fluoride (CaF.sub.2), so as to minimize interference patterns and vignetting encountered in conventional parallel-plate IR cells. The long IR beam pathlength and lensing effect of the circular cross-section of the sample volume in combination with the refractive index differences between the solvent and the transparent crystal serve to focus the IR beam in enhancing sample detection sensitivity by an order of magnitude.
Pollier, Jacob; Morreel, Kris; Geelen, Danny; Goossens, Alain
2011-01-01
Triterpenes are one of the largest classes of plant natural products, with an enormous variety in structure and bioactivities. Here, triterpene saponins from hairy roots of the model legume Medicago truncatula were profiled with reversed-phase liquid chromatography coupled to negative-ion electrospray ionization Fourier transform ion cyclotron resonance mass spectrometry (LC ESI FT-ICR MS). Owing to the accuracy of the FT-ICR MS, reliable molecular formulas of the detected compounds could be ...
Generalized Fourier transforms classes
DEFF Research Database (Denmark)
Berntsen, Svend; Møller, Steen
2002-01-01
The Fourier class of integral transforms with kernels $B(\\omega r)$ has by definition inverse transforms with kernel $B(-\\omega r)$. The space of such transforms is explicitly constructed. A slightly more general class of generalized Fourier transforms are introduced. From the general theory foll...... follows that integral transform with kernels which are products of a Bessel and a Hankel function or which is of a certain general hypergeometric type have inverse transforms of the same structure....
Directory of Open Access Journals (Sweden)
Swati SONAWANE
2015-12-01
Full Text Available The Chlorococcalean microalgae Ankistrodesmus convolutes was found in fresh water Godawari reservoir, Ahmednagar district of Maharashtra State, India. Microalgae are modern biomass for the production of liquid biofuel due to its high solar cultivation efficiency. The collection, harvesting and drying processes were play vital role in converting algal biomass into energy liquid fuel. The oil extraction was the important step for the biodiesel synthesis. The fatty acid methyl ester (FAME synthesis was carried through base catalyzed transesterification method. The product was analyzed by using the hyphened techniques like Fourier Transform-Infrared spectroscopy (FT-IR and Gas Chromatography Mass Spectroscopy (GCMS. FT-IR Spectroscopy was results the ester as functional group of obtained product while the Gas Chromatography Mass Spectroscopy was results the six type of fatty acid methyl ester with different concentration. Ankistrodesmus convolutes biodiesel consist of 46.5% saturated and 49.14% unsaturated FAME.
Fourier transform mass spectrometry.
Scigelova, Michaela; Hornshaw, Martin; Giannakopulos, Anastassios; Makarov, Alexander
2011-07-01
This article provides an introduction to Fourier transform-based mass spectrometry. The key performance characteristics of Fourier transform-based mass spectrometry, mass accuracy and resolution, are presented in the view of how they impact the interpretation of measurements in proteomic applications. The theory and principles of operation of two types of mass analyzer, Fourier transform ion cyclotron resonance and Orbitrap, are described. Major benefits as well as limitations of Fourier transform-based mass spectrometry technology are discussed in the context of practical sample analysis, and illustrated with examples included as figures in this text and in the accompanying slide set. Comparisons highlighting the performance differences between the two mass analyzers are made where deemed useful in assisting the user with choosing the most appropriate technology for an application. Recent developments of these high-performing mass spectrometers are mentioned to provide a future outlook.
Fourier transformation for pedestrians
Butz, Tilman
2015-01-01
This book is an introduction to Fourier Transformation with a focus on signal analysis, based on the first edition. It is well suited for undergraduate students in physics, mathematics, electronic engineering as well as for scientists in research and development. It gives illustrations and recommendations when using existing Fourier programs and thus helps to avoid frustrations. Moreover, it is entertaining and you will learn a lot unconsciously. Fourier series as well as continuous and discrete Fourier transformation are discussed with particular emphasis on window functions. Filter effects of digital data processing are illustrated. Two new chapters are devoted to modern applications. The first deals with data streams and fractional delays and the second with the back-projection of filtered projections in tomography. There are many figures and mostly easy to solve exercises with solutions.
Rajiv, P.; Rajeshwari, Sivaraj; Venckatesh, Rajendran
2013-12-01
Fourier transform infrared spectroscopy (FT-IR) and Gas chromatography-mass spectroscopy have been carried out to investigate the chemical composition of Parthenium mediated vermicompost. Four different concentrations of Parthenium and cow dung mixtures were vermicomposted using the earthworms (Eudrilus eugeniae). FT-IR spectra reveal the absence of Parthenin toxin (sesquiterpene lactone) and phenols in vermicompost which was obtained from high concentration of cow dung mixed treatments. GC-MS analysis shows no phenolic compounds and predominant level of intermediate metabolites such as 4,8,12,16-Tetramethylheptadecan-4-olide (7.61%), 2-Pentadecanone, 6,10,14-trimethyl- (5.29%) and Methyl 16-methyl-heptadecanoate (4.69%) during the vermicomposting process. Spectral results indicated that Parthenin toxin and phenols can be eradicated via vermicomposting if mixed with appropriate quantity of cow dung.
International Nuclear Information System (INIS)
We report on a fully automated 9.4 tesla Fourier transform ion resonance cyclotron (FTICR) mass spectrometer coupled to reverse-phase chromatography for high-throughput proteomic studies. Modifications made to the front-end of a commercial FTICR instrument--a dual-ESI-emitter ion source; dual-channel electrodynamic ion funnel; and collisional-cooling, selection and accumulation quadrupoles--significantly improved the sensitivity, dynamic range and mass measurement accuracy of the mass spectrometer. A high-pressure capillary liquid chromatography (LC) system was incorporated with an autosampler that enabled 24 h/day operation. A novel method for accumulating ions in the ICR cell was also developed. Unattended operation of the instrument revealed the exceptional reproducibility (1-5% deviation in elution times for peptides from a bacterial proteome), repeatability (10-20% deviation in detected abundances for peptides from the same aliquot analyzed a few weeks apart) and robustness (high-throughput operation for 5 months without downtime) of the LC/FTICR system. When combined with modulated-ion-energy gated trapping, the internal calibration of FTICR mass spectra decreased dispersion of mass measurement errors for peptide identifications in conjunction with high resolution capillary LC separations to < 5 ppm over a dynamic range for each spectrum of 10 3
Nielen, M.W.F.; Engelen, M.C. van; Zuiderent, R.; Ramaker, R.
2007-01-01
An emerging trend is recognised in hormone and veterinary drug residue analysis from liquid chromatography tandem mass spectrometry (LC/MS/MS) based screening and confirmation towards accurate mass alternatives such as LC coupled with time-of-flight (TOF), Fourier transform ion cyclotron resonance (
Kamleh, A.; Barrett, M. P.; Wildridge, D.; Burchmore, R. J. S.; Scheltema, R. A.; Watson, D. G.
2008-01-01
It was shown that coupling hydrophilic interaction chromatography (HILIC) to Orbitrap Fourier transform mass spectrometery (FT-MS) provided an excellent tool for metabolic profiling, principally due to rapid elution of lipids in advance of most metabolites entering the mass spectrometer. We used in
Fourier transforms principles and applications
Hansen, Eric W
2014-01-01
Fourier Transforms: Principles and Applications explains transform methods and their applications to electrical systems from circuits, antennas, and signal processors-ably guiding readers from vector space concepts through the Discrete Fourier Transform (DFT), Fourier series, and Fourier transform to other related transform methods. Featuring chapter end summaries of key results, over two hundred examples and four hundred homework problems, and a Solutions Manual this book is perfect for graduate students in signal processing and communications as well as practicing engineers.
Generic Quantum Fourier Transforms
Moore, Cristopher; Russell, A; Moore, Cristopher; Rockmore, Daniel; Russell, Alexander
2003-01-01
The quantum Fourier transform (QFT) is the principal algorithmic tool underlying most efficient quantum algorithms. We present a generic framework for the construction of efficient quantum circuits for the QFT by ``quantizing'' the separation of variables technique that has been so successful in the study of classical Fourier transform computations. Specifically, this framework applies the existence of computable Bratteli diagrams, adapted factorizations, and Gel'fand-Tsetlin bases to offer efficient quantum circuits for the QFT over a wide variety a finite Abelian and non-Abelian groups, including all group families for which efficient QFTs are currently known and many new group families. Moreover, the method gives rise to the first subexponential-size quantum circuits for the QFT over the linear groups GL_k(q), SL_k(q), and the finite groups of Lie type, for any fixed prime power q.
On the Replica Fourier Transform
Carlucci, D. M.; De Dominicis, C.
1997-01-01
The Replica Fourier Transform introduced previously is related to the standard definition of Fourier transforms over a group. Its use is illustrated by block-diagonalizing the eigenvalue equation of a four-replica Parisi matrix.
International Nuclear Information System (INIS)
Gas chromatography/mass spectrometry(GC/MS) is commonly used for oil fingerprinting and provides investigators with good forensic data. However, new challenges face oil spill forensic chemistry with the growing use of biodiesel as well as the recycling and reprocessing of used oil, particularly lubricating oils. This paper demonstrated that Fourier transform infrared (FTIR) spectroscopy may be a fast, cost effective and complementary method for forensic analysis of biodiesels (fatty acid methyl esters) and lubricating oils. Attenuated total reflectance (ATR)-FTIR spectroscopy was shown to be an interesting analytic method because of its use in monitoring and quantifying minor chemical compounds in sample matrices and its ability to identify a broad range or organic compounds. Unlike chromatography, FTIR spectroscopy with ATR can provide results without compound separation or lengthy sample preparation steps. This study described the combined use of GC and ATR-FTIR in environmental oil spill identification through the matching of source lube oil samples with artificially weathered samples. Samples recovered from a biodiesel spill incident were also investigated. ATR-FTIR provided detailed spectral information for rapid lube oil differentiation. This study was part of a continuing effort to develop a methodology to deal with chemical spills of unknown origin, which is an important aspect in environmental protection and emergency preparedness. This method was only successfully applied to the short term artificially weathered and fresh lube oil characterization, and to limited cases of biodiesel spills. It was concluded that further validation tests are needed to determine if this method can be applied to real-world weather lube oil samples. 10 refs., 11 figs.
Zheng, Bei; Li, Wentao; Li, Hongyan; Liu, Lin; Lei, Pei; Ge, Xiaopeng; Yu, Zhiyong; Zhou, Yiqi
2016-08-01
The components for connecting high-performance liquid chromatography (HPLC) with Fourier-transform infrared spectroscopy (FTIR) were investigated to determine estrogen in the water environment, including heating for atomization, solvent removal, sample deposition, drive control, spectrum collection, chip swap, cleaning and drying. Results showed that when the atomization temperature was increased to 388 K, the interference of mobile phase components (methanol, H2O, acetonitrile, and NaH2PO4) were completely removed in the IR measurement of estrogen, with 0.999 of similarity between IR spectra obtained after separation and corresponding to the standard IR spectra. In experiments with varying HPLC injection volumes, high similarity for IR spectra was obtained at 20 ul injection volume at 0.01 mg/L BPA while a useful IR spectrum for 10 ng/L BPA was obtained at 80 ul injection volume. In addition, estrogen concentrations in the natural water samples were calculated semi-quantitatively from the peak intensities of IR spectrum in the mid-infrared region.
Zheng, Bei; Li, Wentao; Li, Hongyan; Liu, Lin; Lei, Pei; Ge, Xiaopeng; Yu, Zhiyong; Zhou, Yiqi
2016-01-01
The components for connecting high-performance liquid chromatography (HPLC) with Fourier-transform infrared spectroscopy (FTIR) were investigated to determine estrogen in the water environment, including heating for atomization, solvent removal, sample deposition, drive control, spectrum collection, chip swap, cleaning and drying. Results showed that when the atomization temperature was increased to 388 K, the interference of mobile phase components (methanol, H2O, acetonitrile, and NaH2PO4) were completely removed in the IR measurement of estrogen, with 0.999 of similarity between IR spectra obtained after separation and corresponding to the standard IR spectra. In experiments with varying HPLC injection volumes, high similarity for IR spectra was obtained at 20 ul injection volume at 0.01 mg/L BPA while a useful IR spectrum for 10 ng/L BPA was obtained at 80 ul injection volume. In addition, estrogen concentrations in the natural water samples were calculated semi-quantitatively from the peak intensities of IR spectrum in the mid-infrared region. PMID:27577974
Suzuki, Hideyuki; Sasaki, Ryosuke; Ogata, Yoshiyuki; Nakamura, Yukiko; Sakurai, Nozomu; Kitajima, Mariko; Takayama, Hiromitsu; Kanaya, Shigehiko; Aoki, Koh; Shibata, Daisuke; Saito, Kazuki
2008-01-01
Flavonoids detected from a model legume plant, Lotus japonicus accessions Miyakojima MG-20 and Gifu B-129, were profiled using liquid chromatography Fourier transform ion cyclotron resonance mass spectrometry (LC-FTICR/MS). Five flavonols and two anthocyanidins were detected as aglycones. LC-FTICR/MS facilitated simultaneous detection of 61 flavonoids including compounds that have not been reported previously. Chemical information of the peaks such as retention time, lambdamax, m/z value of the quasi-molecular ion, m/z value of MS/MS fragment ions, and relative intensity of MS/MS fragments was obtained, along with the molecular formulas and conjugate structures. Fourteen were completely identified by comparison with authentic compounds. The high accuracy of m/z values, being 0.081 ppm between observed and theoretical values, allowed prediction of molecular formulas of unknown compounds with the help of isotope peak information for determination of chemical composition. Based on a predicted elemental composition, the presence of a novel nitrogen-containing flavonoid was proposed. A comparison of flavonoid profiles in flowers, stems, and leaves demonstrated that the flowers yielded the most complex profile, containing 30 flower-specific flavonoids including gossypetin glycosides and isorhamnetin glycosides. A comparison of flavonoid profiles between MG-20 and B-129 grown under the same conditions revealed that the accumulation of anthocyanins was higher in B-129 than MG-20, particularly in the stem. Developmental changes in the flavonoid profiles demonstrated that kaempferol glycosides increased promptly after germination. In contrast, quercetin glycosides, predominant flavonoids in the seeds, were not detectable in growing leaves.
Fourier Transform Spectrometer System
Campbell, Joel F. (Inventor)
2014-01-01
A Fourier transform spectrometer (FTS) data acquisition system includes an FTS spectrometer that receives a spectral signal and a laser signal. The system further includes a wideband detector, which is in communication with the FTS spectrometer and receives the spectral signal and laser signal from the FTS spectrometer. The wideband detector produces a composite signal comprising the laser signal and the spectral signal. The system further comprises a converter in communication with the wideband detector to receive and digitize the composite signal. The system further includes a signal processing unit that receives the composite signal from the converter. The signal processing unit further filters the laser signal and the spectral signal from the composite signal and demodulates the laser signal, to produce velocity corrected spectral data.
Wavelet-fractional Fourier transforms
Institute of Scientific and Technical Information of China (English)
Yuan Lin
2008-01-01
This paper extends the definition of fractional Fourier transform (FRFT) proposed by Namias V by using other orthonormal bases for L2 (R) instead of Hermite-Ganssian functions.The new orthonormal basis is gained indirectly from multiresolution analysis and orthonormal wavelets. The so defined FRFT is called wavelets-fractional Fourier transform.
Institute of Scientific and Technical Information of China (English)
Cai Sheng Wu; Zhi Xin Jia; Bao Ming Ning; Jin Lan Zhang; Song Wu
2012-01-01
In this paper,a high-performance liquid chromatography coupled with ultraviolet detection and Fourier transform-ion cyclotron resonance mass spectrometry (HPLC-UV/FTICRMS) method was described for the investigation of impurity profile in moxifloxacin (MOX) drug substance and chemical reference substance.Ten impurities were detected by HPLC-UV,while eight impurities were identified by using the high accurate molecular mass combined with multiple-stage mass spectrometric data and fragmentation rules.In addition,to our knowledge,five impurities were founded for the first time in MOX drug substance.
Kok, S.J.; Wold, C.A.; Hankemeier, Th.; Schoenmakers, P.J.
2003-01-01
Two commercial liquid chromatography-Fourier-transform infrared spectroscopy interfaces (LC-FTIR), viz. a flow cell and a solvent-elimination interface have been assessed for use in size-exclusion chromatography (SEC) with respect to their chromatographic integrity (i.e. peak asymmetry, chromatograp
Energy Technology Data Exchange (ETDEWEB)
Hembree, D.M.; Garrison, A.A.; Crocombe, R.A.; Yokley, R.A.; Wehry, E.L.; Mamantov, G.
1981-10-01
A detailed description of a system for gas chromatographic (GC) detection of matrix isolation Fourier transform infrared (FTIR) spectrometry is presented. Substances eluting from a support-coated open tubular column are deposited directly on individual faces of a 12-sided movable gold-plated copper disk mounted within the cold head of a closed-cycle cryostat. The same gas is used as GC carrier and spectroscopic matrix gas. An optical interface, comprising beam condensing and rod optics, permits the modified cryostat head to be mounted in proximity to the sample chamber of an FTIR spectrometer. Applications of the technique to identification of polycyclic aromatic hydrocarbon constituents of coal-derived materials are demonstrated, and quantitative aspects of the procedure are considered.
Fast Numerical Nonlinear Fourier Transforms
Wahls, Sander
2014-01-01
The nonlinear Fourier transform, which is also known as the forward scattering transform, decomposes a periodic signal into nonlinearly interacting waves. In contrast to the common Fourier transform, these waves no longer have to be sinusoidal. Physically relevant waveforms are often available for the analysis instead. The details of the transform depend on the waveforms underlying the analysis, which in turn are specified through the implicit assumption that the signal is governed by a certain evolution equation. For example, water waves generated by the Korteweg-de Vries equation can be expressed in terms of cnoidal waves. Light waves in optical fiber governed by the nonlinear Schr\\"dinger equation (NSE) are another example. Nonlinear analogs of classic problems such as spectral analysis and filtering arise in many applications, with information transmission in optical fiber, as proposed by Yousefi and Kschischang, being a very recent one. The nonlinear Fourier transform is eminently suited to address them ...
Slice Fourier transform and convolutions
Cnudde, Lander; De Bie, Hendrik
2015-01-01
Recently the construction of various integral transforms for slice monogenic functions has gained a lot of attention. In line with these developments, the article at hand introduces the slice Fourier transform. In the first part, the kernel function of this integral transform is constructed using the Mehler formula. An explicit expression for the integral transform is obtained and allows for the study of its properties. In the second part, two kinds of corresponding convolutions are examined:...
Static Fourier transform infrared spectrometer.
Schardt, Michael; Murr, Patrik J; Rauscher, Markus S; Tremmel, Anton J; Wiesent, Benjamin R; Koch, Alexander W
2016-04-01
Fourier transform spectroscopy has established itself as the standard method for spectral analysis of infrared light. Here we present a robust and compact novel static Fourier transform spectrometer design without any moving parts. The design is well suited for measurements in the infrared as it works with extended light sources independent of their size. The design is experimentally evaluated in the mid-infrared wavelength region between 7.2 μm and 16 μm. Due to its large etendue, its low internal light loss, and its static design it enables high speed spectral analysis in the mid-infrared.
Directory of Open Access Journals (Sweden)
Jan Zuber
2016-01-01
Full Text Available Pyrolysis oil from the slow pyrolysis of German brown coal from Schöningen, obtained at a temperature of 500°C, was separated and analyzed using hyphenation of gas chromatography with an atmospheric pressure chemical ionization source operated in negative ion mode and Fourier transform ion cyclotron resonance mass spectrometry (GC-APCI-FT-ICR-MS. Development of this ultrahigh-resolving analysis method is described, that is, optimization of specific GC and APCI parameters and performed data processing. The advantages of GC-APCI-FT-ICR-MS hyphenation, for example, soft ionization, ultrahigh-resolving detection, and most important isomer separation, were demonstrated for the sample liquid. For instance, it was possible to separate and identify nine different propylphenol, ethylmethylphenol, and trimethylphenol isomers. Furthermore, homologous series of different acids, for example, alkyl and alkylene carboxylic acids, were verified, as well as homologous series of alkyl phenols, alkyl dihydroxy benzenes, and alkoxy alkyl phenols.
Zuber, Jan; Kroll, Marius M; Rathsack, Philipp; Otto, Matthias
2016-01-01
Pyrolysis oil from the slow pyrolysis of German brown coal from Schöningen, obtained at a temperature of 500°C, was separated and analyzed using hyphenation of gas chromatography with an atmospheric pressure chemical ionization source operated in negative ion mode and Fourier transform ion cyclotron resonance mass spectrometry (GC-APCI-FT-ICR-MS). Development of this ultrahigh-resolving analysis method is described, that is, optimization of specific GC and APCI parameters and performed data processing. The advantages of GC-APCI-FT-ICR-MS hyphenation, for example, soft ionization, ultrahigh-resolving detection, and most important isomer separation, were demonstrated for the sample liquid. For instance, it was possible to separate and identify nine different propylphenol, ethylmethylphenol, and trimethylphenol isomers. Furthermore, homologous series of different acids, for example, alkyl and alkylene carboxylic acids, were verified, as well as homologous series of alkyl phenols, alkyl dihydroxy benzenes, and alkoxy alkyl phenols. PMID:27066076
A More Accurate Fourier Transform
Courtney, Elya
2015-01-01
Fourier transform methods are used to analyze functions and data sets to provide frequencies, amplitudes, and phases of underlying oscillatory components. Fast Fourier transform (FFT) methods offer speed advantages over evaluation of explicit integrals (EI) that define Fourier transforms. This paper compares frequency, amplitude, and phase accuracy of the two methods for well resolved peaks over a wide array of data sets including cosine series with and without random noise and a variety of physical data sets, including atmospheric $\\mathrm{CO_2}$ concentrations, tides, temperatures, sound waveforms, and atomic spectra. The FFT uses MIT's FFTW3 library. The EI method uses the rectangle method to compute the areas under the curve via complex math. Results support the hypothesis that EI methods are more accurate than FFT methods. Errors range from 5 to 10 times higher when determining peak frequency by FFT, 1.4 to 60 times higher for peak amplitude, and 6 to 10 times higher for phase under a peak. The ability t...
Kong, H.; Wang, M.; Venema, K.; Maathuis, A.; Heijden, R. van der; Greef, J. van der; Xu, G.; Hankemeier, T.
2009-01-01
A high-performance liquid chromatography-high resolution Fourier transform ion cyclotron resonance mass spectrometry (HPLC-FTICR-MS) method was developed to investigate the metabolism of ginsenosides in in vitro models of the gastro-intestinal tract. The metabolites were identified by high-resolutio
Logarithm of the Discrete Fourier Transform
Directory of Open Access Journals (Sweden)
Michael Aristidou
2007-01-01
Full Text Available The discrete Fourier transform defines a unitary matrix operator. The logarithm of this operator is computed, along with the projection maps onto its eigenspaces. A geometric interpretation of the discrete Fourier transform is also given.
Logarithm of the Discrete Fourier Transform
Michael Aristidou; Jason Hanson
2007-01-01
The discrete Fourier transform defines a unitary matrix operator. The logarithm of this operator is computed, along with the projection maps onto its eigenspaces. A geometric interpretation of the discrete Fourier transform is also given.
Fourier-transform optical microsystems
Collins, S. D.; Smith, R. L.; Gonzalez, C.; Stewart, K. P.; Hagopian, J. G.; Sirota, J. M.
1999-01-01
The design, fabrication, and initial characterization of a miniature single-pass Fourier-transform spectrometer (FTS) that has an optical bench that measures 1 cm x 5 cm x 10 cm is presented. The FTS is predicated on the classic Michelson interferometer design with a moving mirror. Precision translation of the mirror is accomplished by microfabrication of dovetailed bearing surfaces along single-crystal planes in silicon. Although it is miniaturized, the FTS maintains a relatively high spectral resolution, 0.1 cm-1, with adequate optical throughput.
JPL Fourier transform ultraviolet spectrometer
Cageao, R. P.; Friedl, R. R.; Sander, Stanley P.; Yung, Y. L.
1994-01-01
The Fourier Transform Ultraviolet Spectrometer (FTUVS) is a new high resolution interferometric spectrometer for multiple-species detection in the UV, visible and near-IR. As an OH sensor, measurements can be carried out by remote sensing (limb emission and column absorption), or in-situ sensing (long-path absorption or laser-induced fluorescence). As a high resolution detector in a high repetition rate (greater than 10 kHz) LIF system, OH fluorescence can be discriminated against non-resonant background emission and laser scatter, permitting (0, 0) excitation.
Novel Micro Fourier Transform Spectrometers
Institute of Scientific and Technical Information of China (English)
KONG Yan-mei; LIANG Jing-qiu; LIANG Zhong-zhu; WANG-Bo; ZHANG Jun
2008-01-01
The miniaturization of spectrometer opens a new application area with real-time and on-site measurements. The Fourier transform spectrometer(FTS) is much attractive considering its particular advantages among the approaches. This paper reviews the current status of micro FTS in worldwide and describes its developments; In addition, analyzed are the key problems in designing and fabricating FTS to be settled during the miniaturization. Finally, a novel model of micro FTS with no moving parts is proposed and analyzed, which may provide new concepts for the design of spectrometers.
Fourier-Transform Infrared Spectrometer
Schindler, R. A.
1986-01-01
Fourier-transform spectrometer provides approximately hundredfold increase in luminosity at detector plane over that achievable with older instruments of this type. Used to analyze such weak sources as pollutants and other low-concentration substances in atmosphere. Interferometer creates fringe patterns on two distinct arrays of light detectors, which observe different wavelength bands. Objective lens focuses scene on image plane, which contains optical chopper. To make instrument less susceptible to variations in scene under observation, field and detector lenses focus entrance aperture, rather that image, onto detector array.
Kuo, Su-Ching; Chuang, Shien-Kai; Lin, Ho-Yang; Wang, Lai-Hao
2009-10-19
The purpose of this study was to develop and test a diffuse reflectance infrared Fourier transform spectroscopy (DRIFTS) method, a fast and non-destructive method without extraction, and compare it with the standard gas chromatography (GC) method currently used. A micro-orifice uniform deposit impactor (MOUDI) was used to sample all the size distributions of the aerosol particles of essential oils to investigate the relation between size distributions and the indoor concentration distributions of ylang essential oils. Correlation coefficients for DRIFTS and GC were 0.9904, 0.9910, 0.9913, and 0.9983 for eugenol, isoeugenol, methyl ether, and eugenyl acetate, respectively. The results showed that the concentrations of the four eugenol derivatives of smoke were approximately three times higher than those of mist. Additionally, the major size distributions of aerosol were 0.19 microm and 1.8 microm for the smoke and mist methods, respectively. Because these two methods produce similar results, DRIFTS is a practical method for assessing these fragrances in aerosols. PMID:19800479
Han, Fei; Li, Yanting; Mao, Xinjuan; Xu, Rui; Yin, Ran
2016-05-01
In this work, an approach using high-performance liquid chromatography coupled with diode-array detection and Fourier-transform ion cyclotron resonance mass spectrometer (HPLC-FT-ICR MS) for the identification and profiling of chemical constituents in Rhodiola crenulata was developed for the first time. The chromatographic separation was achieved on an Inertsil ODS-3 column (150 mm × 4.6 mm,3 µm) using a gradient elution program, and the detection was performed on a Bruker Solarix 7.0 T mass spectrometer equipped with electrospray ionization source in both positive and negative modes. Under the optimized conditions, a total of 48 chemical compounds, including 26 alcohols and their glycosides, 12 flavonoids and their glycosides, 5 flavanols and gallic acid derivatives, 4 organic acids and 1 cyanogenic glycoside were identified or tentatively characterized. The results indicated that the developed HPLC-FT-ICR MS method with ultra-high sensitivity and resolution is suitable for identifying and characterizing the chemical constituents in R. crenulata. And it provides a helpful chemical basis for further research on R. crenulata. Copyright © 2016 John Wiley & Sons, Ltd.
Tyburczy, Cynthia; Mossoba, Magdi M; Fardin-Kia, Ali Reza; Rader, Jeanne I
2012-08-01
Current interest by the food industry in exploring reformulation options that lower the content of trans fat in edible fats and oils requires methods to accurately measure low levels of trans fat. In the present study, the quantitation of trans fat in 25 edible fat and oil samples was evaluated using two current analytical approaches, attenuated total reflection-Fourier transform infrared spectroscopy (ATR-FTIR), and gas chromatography with flame ionization detection (GC-FID) according to Official Methods of the American Oil Chemists' Society. Significant differences between the ATR-FTIR and reference GC-FID quantitations were found for samples with a trans fat content fat. These discrepancies could be explained, in part, by the presence of certain oil constituents (e.g., vitamins, carotenoids, high levels of saturated fat) that produced absorbance bands at or near 966 cm(-1) in the ATR-FTIR spectra, a region that was previously identified as being characteristic of isolated trans double bonds. Results demonstrate that the natural content of such oil constituents could result in significant overestimations of trans fat when ATR-FTIR is used to analyze edible fats and oils with a trans fat content fat.
Nakabayashi, Ryo; Tsugawa, Hiroshi; Kitajima, Mariko; Takayama, Hiromitsu; Saito, Kazuki
2015-01-01
In metabolomics, the analysis of product ions in tandem mass spectrometry (MS/MS) is noteworthy to chemically assign structural information. However, the development of relevant analytical methods are less advanced. Here, we developed a method to boost sensitivity in liquid chromatography-Fourier transform ion cyclotron resonance-tandem mass spectrometry analysis (MS/MS boost analysis). To verify the MS/MS boost analysis, both quercetin and uniformly labeled (13)C quercetin were analyzed, revealing that the origin of the product ions is not the instrument, but the analyzed compounds resulting in sensitive product ions. Next, we applied this method to the analysis of monoterpene indole alkaloids (MIAs). The comparative analyses of MIAs having indole basic skeleton (ajmalicine, catharanthine, hirsuteine, and hirsutine) and oxindole skeleton (formosanine, isoformosanine, pteropodine, isopteropodine, rhynchophylline, isorhynchophylline, and mitraphylline) identified 86 and 73 common monoisotopic ions, respectively. The comparative analyses of the three pairs of stereoisomers showed more than 170 common monoisotopic ions in each pair. This method was also applied to the targeted analysis of MIAs in Catharanthus roseus and Uncaria rhynchophylla to profile indole and oxindole compounds using the product ions. This analysis is suitable for chemically assigning features of the metabolite groups, which contributes to targeted metabolome analysis. PMID:26734034
Fourier Analysis and Structure Determination: Part I: Fourier Transforms.
Chesick, John P.
1989-01-01
Provides a brief introduction with some definitions and properties of Fourier transforms. Shows relations, ways of understanding the mathematics, and applications. Notes proofs are not included but references are given. First of three part series. (MVL)
The Fourier Transform on Quantum Euclidean Space
Directory of Open Access Journals (Sweden)
Kevin Coulembier
2011-05-01
Full Text Available We study Fourier theory on quantum Euclidean space. A modified version of the general definition of the Fourier transform on a quantum space is used and its inverse is constructed. The Fourier transforms can be defined by their Bochner's relations and a new type of q-Hankel transforms using the first and second q-Bessel functions. The behavior of the Fourier transforms with respect to partial derivatives and multiplication with variables is studied. The Fourier transform acts between the two representation spaces for the harmonic oscillator on quantum Euclidean space. By using this property it is possible to define a Fourier transform on the entire Hilbert space of the harmonic oscillator, which is its own inverse and satisfies the Parseval theorem.
The multiple-parameter fractional Fourier transform
Institute of Scientific and Technical Information of China (English)
LANG Jun; TAO Ran; RAN QiWen; WANG Yue
2008-01-01
The fractional Fourier transform (FRFT) has multiplicity, which is intrinsic in frac-tional operator. A new source for the multiplicity of the weight-type fractional Fou-rier transform (WFRFT) is proposed, which can generalize the weight coefficients of WFRFT to contain two vector parameters MN,∈ZM. Therefore a generalized frac-tional Fourier transform can be defined, which is denoted by the multiple-parameter fractional Fourier transform (MPFRFT). It enlarges the multiplicity of the FRFT, which not only includes the conventional FRFT and general multi-fractional Fourier transform as special cases, but also introduces new fractional Fourier transforms. It provides a unified framework for the FRFT, and the method is also available for fractionalizing other linear operators. In addition, numerical simulations of the MPFRFT on the Hermite-Gaussian and rectangular functions have been performed as a simple application of MPFRFT to signal processing.
Matrix Fourier transform with discontinuous coefficients
Yaremko, O.; Zhuravleva, E.
2013-01-01
The explicit construction of direct and inverse Fourier's vector transform with discontinuous coefficients is presented. The technique of applying Fourier's vector transform with discontinuous coefficients for solving problems of mathematical physics.Multidimensional integral transformations with non-separated variables for problems with discontinuous coefficients are constructed in this work. The coefficient discontinuities focused on the of parallel hyperplanes. In this work explicit formul...
Two modified discrete chirp Fourier transform schemes
Institute of Scientific and Technical Information of China (English)
樊平毅; 夏香根
2001-01-01
This paper presents two modified discrete chirp Fourier transform (MDCFT) schemes.Some matched filter properties such as the optimal selection of the transform length, and its relationship to analog chirp-Fourier transform are studied. Compared to the DCFT proposed previously, theoretical and simulation results have shown that the two MDCFTs can further improve the chirp rate resolution of the detected signals.
Facchini, Laura; Losito, Ilario; Cataldi, Tommaso R I; Palmisano, Francesco
2016-09-01
Hydrophilic interaction liquid chromatography coupled to electrospray ionization-Fourier transform mass spectrometry was employed to study ceramide lipids occurring in mussels of sp. Mytilus galloprovincialis. Lipid extracts from alive mussels and mussels deliberately subjected to specific thermal treatments were analyzed. In particular, single and tandem MS measurements were performed on a hybrid quadrupole-Orbitrap mass spectrometer and then complemented by MS(n) analyses (n = 2, 3) achieved by a linear ion trap mass spectrometer. This approach enabled the characterization of 66 ceramide lipids, encompassing ceramide phosphoethanolamines (CPE), ceramide aminoethylphosphonates (CAEP) and N-monomethylated CAEP. The sphingoid and acyl chains of each ceramide lipid could be distinctly recognized in terms of numbers of carbon atoms and C=C bonds, and indications on the possible location of the latter on the sphingoid chain could be often inferred from fragmentation patterns. The occurrence of several species hydroxylated on the α carbon of the acyl chain was also discovered. On the other hand, the sphingoid chain of ceramide lipids was never found to be involved in oxidation processes, unless forced exposure of the mussel lipid extracts to atmospheric oxygen was performed. CPE(d19:3/16:0) and its hydroxylated form, CPE(d19:3/2-OH-16:0), were found to be the prevailing species among CPE, whereas CAEP(d18:2/16:0), CAEP(d19:3/16:0) and CAEP(d19:3/2-OH-16:0) were the most abundant CAEP. Finally, ceramide lipids showed a remarkably higher stability, compared with glycerophospholipids, in mussels subjected to different thermal treatments. This finding opens interesting perspectives on the role of ceramide-based lipids in the adaptation of aquatic organisms to thermal stresses. Copyright © 2016 John Wiley & Sons, Ltd.
Shift sampling theory of Fourier transform computation
Institute of Scientific and Technical Information of China (English)
柴玉璞
1997-01-01
The DFT transform us extended to DFTξη transform and the relationship between FT and DFTξη is given by the Fourier transform discretization theorem. Based on the theorem, the DFTξη algorithm-error equation (DFTξη A-E equation) is established, and the minimization property of discrete effect and the oscillation property of truncation effect are demonstrated. All these construct the shift sampling theory——a new theory about Fourier transform computation.
On the $q$-Bessel Fourier transform
Dhaouadi, Lazhar
2013-01-01
In this work, we are interested by the $q$-Bessel Fourier transform with a new approach. Many important results of this $q$-integral transform are proved with a new constructive demonstrations and we establish in particular the associated $q$-Fourier-Neumen expansion which involves the $q$-little Jacobi polynomials.
Product Theorem for Quaternion Fourier Transform
Bahri, Mawardi
2014-01-01
In this paper we present the generalized convolution and correlation for the two-dim ensional discrete quaternion Fourier transform (DQFT). We provide several new properties of the generalizations. There results can be considered as the extension of correlation and convolution properties of real and complex Fourier transform to the DQFT domain.
The Geostationary Fourier Transform Spectrometer
Key, Richard; Sander, Stanley; Eldering, Annmarie; Blavier, Jean-Francois; Bekker, Dmitriy; Manatt, Ken; Rider, David; Wu, Yen-Hung
2012-01-01
The Geostationary Fourier Transform Spectrometer (GeoFTS) is an imaging spectrometer designed for a geostationary orbit (GEO) earth science mission to measure key atmospheric trace gases and process tracers related to climate change and human activity. GEO allows GeoFTS to continuously stare at a region of the earth for frequent sampling to capture the variability of biogenic fluxes and anthropogenic emissions from city to continental spatial scales and temporal scales from diurnal, synoptic, seasonal to interannual. The measurement strategy provides a process based understanding of the carbon cycle from contiguous maps of carbon dioxide (CO2), methane (CH4), carbon monoxide (CO), and chlorophyll fluorescence (CF) collected many times per day at high spatial resolution (2.7kmx2.7km at nadir). The CO2/CH4/CO/CF measurement suite in the near infrared spectral region provides the information needed to disentangle natural and anthropogenic contributions to atmospheric carbon concentrations and to minimize uncertainties in the flow of carbon between the atmosphere and surface. The half meter cube size GeoFTS instrument is based on a Michelson interferometer design that uses all high TRL components in a modular configuration to reduce complexity and cost. It is self-contained and as independent of the spacecraft as possible with simple spacecraft interfaces, making it ideal to be a "hosted" payload on a commercial communications satellite mission. The hosted payload approach for measuring the major carbon-containing gases in the atmosphere from the geostationary vantage point will affordably advance the scientific understating of carbon cycle processes and climate change.
Fourier transforms in radar and signal processing
Brandwood, David
2011-01-01
Fourier transforms are used widely, and are of particular value in the analysis of single functions and combinations of functions found in radar and signal processing. Still, many problems that could have been tackled by using Fourier transforms may have gone unsolved because they require integration that is difficult and tedious. This newly revised and expanded edition of a classic Artech House book provides you with an up-to-date, coordinated system for performing Fourier transforms on a wide variety of functions. Along numerous updates throughout the book, the Second Edition includes a crit
Fourier transform infrared spectra applications to chemical systems
Ferraro, John R
1978-01-01
Fourier Transform Infrared Spectroscopy: Applications to Chemical Systems presents the chemical applications of the Fourier transform interferometry (FT-IR).The book contains discussions on the applications of FT-IR in the fields of chromatography FT-IR, polymers and biological macromolecules, emission spectroscopy, matrix isolation, high-pressure interferometry, and far infrared interferometry. The final chapter is devoted to the presentation of the use of FT-IR in solving national technical problems such as air pollution, space exploration, and energy related subjects.Researc
Fourier transform infrared spectra applications to chemical systems
Ferraro, John R
1985-01-01
The final and largest volume to complete this four-volume treatise is published in response to the intense commercial and research interest in Fourier Transform Interferometry.Presenting current information from leading experts in the field, Volume 4 introduces new information on, for example, applications of Diffuse Reflectance Spectroscopy in the Far-Infrared Region. The editors place emphasis on surface studies and address advances in Capillary Gas Chromatography - Fourier Transform Interferometry.Volume 4 especially benefits spectroscopists and physicists, as well as researchers
Electronically-Scanned Fourier-Transform Spectrometer
Breckinridge, J. B.; Ocallaghan, F. G.
1984-01-01
Instrument efficient, lightweight, and stable. Fourier-transform spectrometer configuration uses electronic, instead of mechanical, scanning. Configuration insensitive to vibration-induced sampling errors introduced into mechanically scanned systems.
Fractional Fourier transform of Lorentz beams
Institute of Scientific and Technical Information of China (English)
Zhou Guo-Quan
2009-01-01
This paper introduces Lorentz beams to describe certain laser sources that produce highly divergent fields. The fractional Fourier transform (FRFT) is applied to treat the propagation of Lorentz beams. Based on the definition of convolution and the convolution theorem of the Fourier transform, an analytical expression for a Lorentz beam passing through a FRFT system has been derived. By using the derived formula, the properties of a Lorentz beam in the FRFT plane are illustrated numerically.
Glasser, L.
1987-01-01
This paper explores how Fourier Transform (FT) mimics spectral transformation, how this property can be exploited to advantage in spectroscopy, and how the FT can be used in data treatment. A table displays a number of important FT serial/spectral pairs related by Fourier Transformations. A bibliography and listing of computer software related to…
From Complex Fractional Fourier Transform to Complex Fractional Radon Transform
Institute of Scientific and Technical Information of China (English)
FANHong-Yi; JIANGNian-Quan
2004-01-01
We show that for n-dimensional complex fractional Fourier transform the corresponding complex fractional Radon transform can also be derived, however, it is different from the direct product of two n-dimensional real fractional Radon transforms. The complex fractional Radon transform of two-mode Wigner operator is calculated.
Mawardi Bahri
2013-01-01
In this paper we introduce convolution theorem for the Fourier transform (FT) of two complex functions. We show that the correlation theorem for the FT can be derived using properties of convolution. We develop this idea to derive the correlation theorem for the quaternion Fourier transform (QFT) of the two quaternion functions.
High order generalized permutational fractional Fourier transforms
Institute of Scientific and Technical Information of China (English)
Ran Qi-Wen; Yuan Lin; Tan Li-Ying; Ma Jing; Wang Qi
2004-01-01
We generalize the definition of the fractional Fourier transform (FRFT) by extending the new definition proposed by Shih. The generalized FRFT, called the high order generalized permutational fractional Fourier transform (HGPFRFT),is a generalized permutational transform. It is shown to have arbitrary natural number M periodic eigenvalues not only with respect to the order of Hermite-Gaussian functions but also to the order of the transform. This HGPFRFT will be reduced to the original FRFT proposed by Namias and Liu's generalized FRFT and Shih's FRFT at the three limits with M = +∞,M = 4k (k is a natural number), and M = 4, respectively. Therefore the HGPFRFT introduces a comprehensive approach to Shih's FRFT and the original definition. Some important properties of HGPFRFT are discussed. Lastly the results of computer simulations and symbolic representations of the transform are provided.
High order generalized permutational fractional Fourier transforms
Ran, Qi-Wen; Yuan, Lin; Tan, Li-Ying; Ma, Jing; Wang, Qi
2004-02-01
We generalize the definition of the fractional Fourier transform (FRFT) by extending the new definition proposed by Shih. The generalized FRFT, called the high order generalized permutational fractional Fourier transform (HGPFRFT), is a generalized permutational transform. It is shown to have arbitrary natural number M periodic eigenvalues not only with respect to the order of Hermite-Gaussian functions but also to the order of the transform. This HGPFRFT will be reduced to the original FRFT proposed by Namias and Liu's generalized FRFT and Shih's FRFT at the three limits with M = +infty, M = 4k (k is a natural number) and M = 4, respectively. Therefore the HGPFRFT introduces a comprehensive approach to Shih's FRFT and the original definition. Some important properties of HGPFRFT are discussed. Lastly the results of computer simulations and symbolic representations of the transform are provided.
Fourier transforms and convolutions for the experimentalist
Jennison, RC
1961-01-01
Fourier Transforms and Convolutions for the Experimentalist provides the experimentalist with a guide to the principles and practical uses of the Fourier transformation. It aims to bridge the gap between the more abstract account of a purely mathematical approach and the rule of thumb calculation and intuition of the practical worker. The monograph springs from a lecture course which the author has given in recent years and for which he has drawn upon a number of sources, including a set of notes compiled by the late Dr. I. C. Browne from a series of lectures given by Mr. J . A. Ratcliffe of t
Fractional Fourier Transform of Cantor Sets
Institute of Scientific and Technical Information of China (English)
LIAO Tian-ne; GAO Qiong
2005-01-01
@@ A new kind of multifractal is constructed by fractional Fourier transform of Cantor sets. The wavelet transform modulus maxima method is applied to calculate the singularity spectrum under an operational definition of multifractal. In particular, an analysing procedure to determine the spectrum is suggested for practice. Nonanalyticities of singularity spectra or phase transitions are discovered, which are interpreted as some indications on the range of Boltzmann temperature q, on which the scaling relation of partition function holds.
Fourier transforms on an amalgam type space
Liflyand, E
2012-01-01
We introduce an amalgam type space, a subspace of $L^1(\\mathbb R_+).$ Integrability results for the Fourier transform of a function with the derivative from such an amalgam space are proved. As an application we obtain estimates for the integrability of trigonometric series.
Ultrafast Fourier-transform parallel processor
Energy Technology Data Exchange (ETDEWEB)
Greenberg, W.L.
1980-04-01
A new, flexible, parallel-processing architecture is developed for a high-speed, high-precision Fourier transform processor. The processor is intended for use in 2-D signal processing including spatial filtering, matched filtering and image reconstruction from projections.
Directional Uncertainty Principle for Quaternion Fourier Transform
Hitzer, Eckhard
2013-01-01
This paper derives a new directional uncertainty principle for quaternion valued functions subject to the quaternion Fourier transformation. This can be generalized to establish directional uncertainty principles in Clifford geometric algebras with quaternion subalgebras. We demonstrate this with the example of a directional spacetime algebra function uncertainty principle related to multivector wave packets.
Energy Technology Data Exchange (ETDEWEB)
Tang, Yinjie; Pingitore, Francesco; Mukhopadhyay, Aindrila; Phan,Richard; Hazen, Terry C.; Keasling, Jay D.
2007-03-15
Flux distribution in central metabolic pathways ofDesulfovibrio vulgaris Hildenborough was examined using 13C tracerexperiments. Consistent with the current genome annotation andindependent evidence from enzyme activity assays, the isotopomer resultsfrom both GC-MS and Fourier Transform-Ion Cyclotron Resonance massspectrometry (FT-ICR MS) indicate the lack of oxidatively functional TCAcycle and an incomplete pentose phosphate pathway. Results from thisstudy suggest that fluxes through both pathways are limited tobiosynthesis. The data also indicate that>80 percent of the lactatewas converted to acetate and the reactions involved are the primary routeof energy production (NAD(P)H and ATP production). Independent of the TCAcycle, direct cleavage of acetyl-CoA to CO and 5,10-methyl-THF also leadsto production of NADH and ATP. Although the genome annotation implicatesa ferredoxin-dependentoxoglutarate synthase, isotopic evidence does notsupport flux through this reaction in either the oxidative or reductivemode; therefore, the TCA cycle is incomplete. FT-ICR MS was used tolocate the labeled carbon distribution in aspartate and glutamate andconfirmed the presence of an atypical enzyme for citrate formationsuggested in previous reports (the citrate synthesized by this enzyme isthe isotopic antipode of the citrate synthesized by the (S)-citratesynthase). These findings enable a better understanding of the relationbetween genome annotation and actual metabolic pathways in D. vulgaris,and also demonstrate FT-ICR MS as a powerful tool for isotopomeranalysis, overcoming problems in both GC-MS and NMRspectroscopy.
Kok, S.J.; Hankemeier, T.; Schoenmakers, P.J.
2005-01-01
The on-line coupling of comprehensive two-dimensional liquid chromatography (liquid chromatography × size-exclusion chromatography, LC × SEC) and infrared (IR) spectroscopy has been realized by means of an IR flow cell. The system has been assessed by the functional-group analysis of a series of sty
Debnath, Lokenath
2012-07-01
The profound study of nature is the most fertile source of mathematical discoveries. Not only does this study, by offering a definite goal to research, have the advantage of excluding vague questions and futile calculations, but it is also a sure means of moulding analysis itself, and discerning those elements in it which it is still essential to know and which science ought to conserve. These fundamental elements are those which recur in all natural phenomena. Joseph Fourier pure mathematics enables us to discover the concepts and laws connecting them, which gives us the key to the understanding of the phenomena of nature. Albert Einstein This article deals with a brief biographical sketch of Joseph Fourier, his first celebrated work on analytical theory of heat, his first great discovery of Fourier series and Fourier transforms. Included is a historical development of Fourier series and Fourier transforms with their properties, importance and applications. Special emphasis is made to his splendid research contributions to mathematical physics, pure and applied mathematics and his unprecedented public service accomplishments in the history of France. This is followed by historical comments about the significant and major impact of Fourier analysis on mathematical physics, probability and mathematical statistics, mathematical economics and many areas of pure and applied mathematics including geometry, harmonic analysis, signal analysis, wave propagation and wavelet analysis. Special attention is also given to the Fourier integral formula, Brownian motion and stochastic processes and many examples of applications including isoparametric inequality, everywhere continuous but nowhere differentiable functions, Heisenberg uncertainty principle, Dirichlets' theorem on primes in arithmetic progression, the Poisson summation formula and solutions of wave and diffusion equations. It is also shown that Fourier coefficients c n (t) in the Fourier expansion of a scalar field
Electro-optic imaging Fourier transform spectrometer
Chao, Tien-Hsin (Inventor); Znod, Hanying (Inventor)
2009-01-01
An Electro-Optic Imaging Fourier Transform Spectrometer (EOIFTS) for Hyperspectral Imaging is described. The EOIFTS includes an input polarizer, an output polarizer, and a plurality of birefringent phase elements. The relative orientations of the polarizers and birefringent phase elements can be changed mechanically or via a controller, using ferroelectric liquid crystals, to substantially measure the spectral Fourier components of light propagating through the EIOFTS. When achromatic switches are used as an integral part of the birefringent phase elements, the EIOFTS becomes suitable for broadband applications, with over 1 micron infrared bandwidth.
International Nuclear Information System (INIS)
Flux distribution in central metabolic pathways of Desulfovibrio vulgaris Hildenborough was examined using 13C tracer experiments. Consistent with the current genome annotation and independent evidence from enzyme activity assays, the isotopomer results from both GC-MS and Fourier Transform-Ion Cyclotron Resonance mass spectrometry (FT-ICR MS) indicate the lack of oxidatively functional TCA cycle and an incomplete pentose phosphate pathway. Results from this study suggest that fluxes through both pathways are limited to biosynthesis. The data also indicate that >80 percent of the lactate was converted to acetate and the reactions involved are the primary route of energy production (NAD(P)H and ATP production). Independent of the TCA cycle, direct cleavage of acetyl-CoA to CO and 5,10-methyl-THF also leads to production of NADH and ATP. Although the genome annotation implicates a ferredoxin-dependent oxoglutarate synthase, isotopic evidence does not support flux through this reaction in either the oxidative or reductive mode; therefore, the TCA cycle is incomplete. FT-ICR MS was used to locate the labeled carbon distribution in aspartate and glutamate and confirmed the presence of an atypical enzyme for citrate formation suggested in previous reports (the citrate synthesized by this enzyme is the isotopic antipode of the citrate synthesized by the (S)-citrate synthase). These findings enable a better understanding of the relation between genome annotation and actual metabolic pathways in D. vulgaris, and also demonstrate FT-ICR MS as a powerful tool for isotopomer analysis, overcoming problems in both GC-MS and NMR spectroscopy
Herniman, Julie M; Bristow, Tony W T; O'Connor, Gavin; Jarvis, Jackie; Langley, G John
2004-01-01
The use of a second electrospray nebuliser has proved to be highly successful for exact mass measurement during high-performance liquid chromatography/Fourier transform ion cyclotron resonance mass spectrometry (HPLC/FTICRMS). Much improved accuracy and precision of mass measurement were afforded by the introduction of the internal calibration solution, thus overcoming space charge issues due to the lack of control over relative ion abundances of the species eluting from the HPLC column. Further, issues of suppression of ionisation, observed when using a T-piece method, are addressed and this simple system has significant benefits over other more elaborate approaches providing data that compares very favourably with these other approaches. The technique is robust, flexible and transferable and can be used in conjunction with HPLC, infusion or flow injection analysis (FIA) to provide constant internal calibration signals to allow routine, accurate and precise mass measurements to be recorded.
Miniaturization of holographic Fourier-transform spectrometers.
Agladze, Nikolay I; Sievers, Albert J
2004-12-20
Wave propagation equations in the stationary-phase approximation have been used to identify the theoretical bounds of a miniature holographic Fourier-transform spectrometer (HFTS). It is demonstrated that the HFTS throughput can be larger than for a scanning Fourier-transform spectrometer. Given room- or a higher-temperature constraint, a small HFTS has the potential to outperform a small multichannel dispersive spectrograph with the same resolving power because of the size dependence of the signal-to-noise ratio. These predictions are used to analyze the performance of a miniature HFTS made from simple optical components covering a broad spectral range from the UV to the near IR. The importance of specific primary aberrations in limiting the HFTS performance has been both identified and verified. PMID:15646777
Fourier Transform Infrared Spectroscopic Studies in Flotation
Institute of Scientific and Technical Information of China (English)
无
2001-01-01
Fourier transform infrared (FTIR) spectroscopy has been extensively employed in flotation research.The work done by the author and co-workers has been reported.A comparison has been made among the different FTIR spectroscopic techniques,e.g.,transmission FTIR spectroscopy,diffuse reflectance infrared Fourier transform (DRIFT) spectroscopy,and attenuated total reflectance (ATR) FTIR spectroscopy.FTIR spectroscopy has been used to study the mechanism of interaction between the collector and the surfaces of different minerals,the mechanism of action of the depressant in improving the selectivity of flotation,and the mechanism of adsorption of the polymeric modifying reagent on mineral surfaces.The interaction between particles in mineral suspension has also been studied by FTIR spectroscopy.
Fourier-transform spectroscopy instrumentation engineering
Saptari, Vidi
2003-01-01
Many applications today require the Fourier-transform (FT) spectrometer to perform close to its limitations, such as taking many quantitative measurements in the visible and in the near infrared wavelength regions. In such cases, the instrument should not be considered as a perfect ""black box."" Knowing where the limitations of performance arise and which components must be improved are crucial to obtaining repeatable and accurate results. One of the objectives of this book is to help the user identify the instrument's bottleneck.
An Imaging Fourier Transform Spectrometer for NGST
Graham, J R
1999-01-01
Due to its simultaneous deep imaging and integral field spectroscopic capability, an Imaging Fourier Transform Spectrograph (IFTS) is ideally suited to the Next Generation Space Telescope (NGST) mission, and offers opportunities for tremendous scientific return in many fields of astrophysical inquiry. We describe the operation and quantify the advantages of an IFTS for space applications. The conceptual design of the Integral Field Infrared Spectrograph (IFIRS) is a wide field (5'.3 x 5'.3) four-port imaging Michelson interferometer.
Nepomuceno, Angelito I; Muddiman, David C; Bergen, H Robert; Craighead, James R; Burke, Michael J; Caskey, Patrick E; Allan, Jonathan A
2003-07-15
Fourier transform ion cyclotron resonance mass spectrometry (FT-ICR-MS) has rapidly established a prominent role in proteomics because of its unparalleled resolving power, sensitivity and ability to achieve high mass measurement accuracy (MMA) simultaneously. However, space-charge effects must be quantitatively, routinely, and confidently corrected because they are known to profoundly influence MMA. We argue that the most effective way to account for space-charge effects is to introduce an internal mass calibrant (IMC) using a dual electrospray ionization (ESI) source where the IMC is added from a separate ESI emitter. The major disadvantage of our initial dual ESI source to achieve high MMA, and arguably the only one, was the time required to switch between the analyte emitter and IMC emitter (i.e., >300 ms). While this "switching time" was acceptable for direct infusion experiments, it did not lend itself to high-throughput applications or when conducting on-line liquid separations. In this report, we completely redesigned the dual ESI source and demonstrate several key attributes. First, the new design allows for facile alignment of ESI emitters, undetectable vibration, and the ability to extend to multiple emitters. Second, the switching time was reduced to IMC to be accumulated "simultaneously" in the external ion reservoir and injected as a single ion packet into the ion cyclotron resonance cell, eliminating the need for a separate accumulation and ion injection event for the IMC. Third, by using a high concentration of the IMC, the residence time on this emitter could be reduced to approximately 80 ms, allowing for more time spent accumulating analyte ions of significantly lower concentration. Fourth, multiplexed on-line separations can be carried out providing increased throughput. Specifically, the new dual ESI source has demonstrated its ability to produce a stable ion current over a 45-min time period at 7 T resulting in mass accuracies of 1.08 ppm +/- 0
Fourier transform resampling: Theory and application
International Nuclear Information System (INIS)
One of the most challenging problems in medical imaging is the development of reconstruction algorithms for nonstandard geometries. This work focuses on the application of Fourier analysis to the problem of resampling or rebinning. Conventional resampling methods utilizing some form of interpolation almost always result in a loss of resolution in the tomographic image. Fourier Transform Resampling (FTRS) offers potential improvement because the Modulation Transfer Function (MTF) of the process behaves like an ideal low pass filter. The MTF, however, is nonstationary if the coordinate transformation is nonlinear. FTRS may be viewed as a generalization of the linear coordinate transformations of standard Fourier analysis. Simulated MTF's were obtained by projecting point sources at different transverse positions in the flat fan beam detector geometry. These MTF's were compared to the closed form expression for FIRS. Excellent agreement was obtained for frequencies at or below the estimated cutoff frequency. The resulting FTRS algorithm is applied to simulations with symmetric fan beam geometry, an elliptical orbit and uniform attenuation, with a normalized root mean square error (NRME) of 0.036. Also, a Tc-99m point source study (1 cm dia., placed in air 10 cm from the COR) for a circular fan beam acquisition was reconstructed with a hybrid resampling method. The FWHM of the hybrid resampling method was 11.28 mm and compares favorably with a direct reconstruction (FWHM: 11.03 mm)
Programs for high-speed Fourier, Mellin and Fourier-Bessel transforms
Ikhabisimov, D. K.; Debabov, A. S.; Kolosov, B. I.; Usikov, D. A.
1979-01-01
Several FORTRAN program modules for performing one-dimensional and two-dimensional discrete Fourier transforms, Mellin, and Fourier-Bessel transforms are described along with programs that realize the algebra of high speed Fourier transforms on a computer. The programs can perform numerical harmonic analysis of functions, synthesize complex optical filters on a computer, and model holographic image processing methods.
He, Xiao-Song; Xi, Bei-Dou; Li, Wen-Tao; Gao, Ru-Tai; Zhang, Hui; Tan, Wen-Bing; Huang, Cai-Hong
2015-11-13
Size exclusion chromatography and reversed-phase high-performance liquid chromatography (RP-HPLC) were combined with Fourier transform infrared spectra (FTIR) and nuclear magnetic resonance (NMR) based on two dimensional (2D) hetero-spectral correlation spectra techniques to fractionate compost-derived dissolved organic matter (DOM) and determine how size- and hydrophobicity-distinguished fractions differ in the composition and evolution. The results showed that the compost-derived DOM was comprised of protein- and humic-like species. The low apparent molecule weight (AMW) protein-like components were enriched in C-C=H3 and N-C=O, and showed more bioreactivity compared with the high AMW counterpart. The hydrophobic and hydrophilic protein-like components both consisted of CCH3 and N-C=O. However, the relatively hydrophilic protein-like components were more easily consumed. As to the humic-like species, the relatively hydrophilic components were slightly larger than the relatively hydrophobic ones. The high AMW and relatively hydrophilic humic-like components were high in C-H, OCH3, N-C=O, N-H, COO, O-H and aromatic C. The low AMW and relatively hydrophobic humic-like components were enriched in CCH3 and N-C=O, and were easily biodegraded during composting. 2D hetero-spectral correlation spectra techniques enhance the characterization of DOM and provide a promising way to elucidate the environmental behaviors of DOM.
Kok, S J; Wold, C A; Hankemeier, Th; Schoenmakers, P J
2003-10-31
Two commercial liquid chromatography-Fourier-transform infrared spectroscopy interfaces (LC-FTIR), viz. a flow cell and a solvent-elimination interface have been assessed for use in size-exclusion chromatography (SEC) with respect to their chromatographic integrity (i.e. peak asymmetry, chromatographic resolution), quantitative and qualitative aspects. A polycarbonate/aliphatic polyester (PC/APE) blend and a polycarbonate-co-polydimethylsiloxane (PC-co-PDMS) copolymer were selected for the assessment. Both samples were successfully and selectively analyzed. The relatively large volume of the flow cell and the inherent deposition characteristics of the solvent-elimination interface led to a comparable decrease in the chromatographic resolution. The separation of oligomers was diminished in comparison with SEC-ultra-violet (UV). However, the peak asymmetry was not significantly affected by either interface. For both interfaces, a linear relationship was obtained for the FTIR response versus the injected concentration. The sensitivity was found to be higher for the solvent-elimination interface. For the current model compounds, the flow-cell interface detection limits are worse. However, the repeatability of flow-cell SEC-FTIR, evaluated by means of four SEC-FTIR analyses of polycarbonate, was considerably better than for solvent-elimination SEC-FTIR. This is probably due to the well-defined optical path length of the sample in the flow cell. By spectral subtraction, it was very well possible to obtain qualitative (functional group) information for compound identification also with flow-cell SEC-FTIR.
Optical image encryption based on multifractional Fourier transforms.
Zhu, B; Liu, S; Ran, Q
2000-08-15
We propose a new image encryption algorithm based on a generalized fractional Fourier transform, to which we refer as a multifractional Fourier transform. We encrypt the input image simply by performing the multifractional Fourier transform with two keys. Numerical simulation results are given to verify the algorithm, and an optical implementation setup is also suggested. PMID:18066153
Quantum Optical Squeezing Transform for Generalizing Fractional Fourier Transform'
Institute of Scientific and Technical Information of China (English)
HU Li-Yun; FAN Hong-Yi
2008-01-01
By establishing the relation between the optical scaled fractional Fourier transform (FFT) and quantum mechanical squeezing-rotating operator transform, we employ the bipartite entangled state representation of two-mode squeezing operator to extend the scaled FFT to more general cases, such as scaled complex FFT and entangled scaled FFT. The additivity and eigenmodes are presented in quantum version. The relation between the scaled FFT and squeezing-rotating Wigner operator is studied.
The cyclic heptapeptide microcystin toxins produced by a strain of Microcystis aeruginosa that has not been investigated previously were separated by liquid chromatography and identified by high-accuracy m/z measurements of their [M + H]+ ions and the fragment i...
Fourier transform infrared spectroscopy of deuterated proteins
Marcano O., A.; Markushin, Y.; Melikechi, N.; Connolly, D.
2008-08-01
We report on Fourier transform spectra of deuterated proteins: Bovine Serum Albumin, Leptin, Insulin-like Growth Factor II, monoclonal antibody to ovarian cancer antigen CA125 and Osteopontin. The spectra exhibit changes in the relative amplitude and spectral width of certain peaks. New peaks not present in the non-deuterated sample are also observed. Ways for improving the deuteration of proteins by varying the temperature and dilution time are discussed. We propose the use of deuterated proteins to increase the sensitivity of immunoassays aimed for early diagnostic of diseases most notably cancer.
Two-dimensional fourier transform spectrometer
Energy Technology Data Exchange (ETDEWEB)
DeFlores, Lauren; Tokmakoff, Andrei
2016-10-25
The present invention relates to a system and methods for acquiring two-dimensional Fourier transform (2D FT) spectra. Overlap of a collinear pulse pair and probe induce a molecular response which is collected by spectral dispersion of the signal modulated probe beam. Simultaneous collection of the molecular response, pulse timing and characteristics permit real time phasing and rapid acquisition of spectra. Full spectra are acquired as a function of pulse pair timings and numerically transformed to achieve the full frequency-frequency spectrum. This method demonstrates the ability to acquire information on molecular dynamics, couplings and structure in a simple apparatus. Multi-dimensional methods can be used for diagnostic and analytical measurements in the biological, biomedical, and chemical fields.
Two-dimensional fourier transform spectrometer
DeFlores, Lauren; Tokmakoff, Andrei
2013-09-03
The present invention relates to a system and methods for acquiring two-dimensional Fourier transform (2D FT) spectra. Overlap of a collinear pulse pair and probe induce a molecular response which is collected by spectral dispersion of the signal modulated probe beam. Simultaneous collection of the molecular response, pulse timing and characteristics permit real time phasing and rapid acquisition of spectra. Full spectra are acquired as a function of pulse pair timings and numerically transformed to achieve the full frequency-frequency spectrum. This method demonstrates the ability to acquire information on molecular dynamics, couplings and structure in a simple apparatus. Multi-dimensional methods can be used for diagnostic and analytical measurements in the biological, biomedical, and chemical fields.
Research progress on discretization of fractional Fourier transform
Institute of Scientific and Technical Information of China (English)
TAO Ran; ZHANG Feng; WANG Yue
2008-01-01
As the fractional Fourier transform has attracted a considerable amount of atten-tion in the area of optics and signal processing,the discretization of the fractional Fourier transform becomes vital for the application of the fractional Fourier trans-form.Since the discretization of the fractional Fourier transform cannot be obtained by directly sampling in time domain and the fractional Fourier domain,the discre-tization of the fractional Fourier transform has been investigated recently.A sum-mary of discretizations of the fractional Fourier transform developed in the last nearly two decades is presented in this paper.The discretizations include sampling in the fractional Fourier domain,discrete-time fractional Fourier transform,frac-tional Fourier series,discrete fractional Fourier transform (including 3 main types:linear combination-type;sampling-type;and eigen decomposition-type),and other discrete fractional signal transform.It is hoped to offer a doorstep for the readers who are interested in the fractional Fourier transform.
The Fourier transform of tubular densities
Prior, C B
2012-05-18
We consider the Fourier transform of tubular volume densities, with arbitrary axial geometry and (possibly) twisted internal structure. This density can be used to represent, among others, magnetic flux or the electron density of biopolymer molecules. We consider tubes of both finite radii and unrestricted radius. When there is overlap of the tube structure the net density is calculated using the super-position principle. The Fourier transform of this density is composed of two expressions, one for which the radius of the tube is less than the curvature of the axis and one for which the radius is greater (which must have density overlap). This expression can accommodate an asymmetric density distribution and a tube structure which has non-uniform twisting. In addition we give several simpler expressions for isotropic densities, densities of finite radius, densities which decay at a rate sufficient to minimize local overlap and finally individual surfaces of the tube manifold. These simplified cases can often be expressed as arclength integrals and can be evaluated using a system of first-order ODEs. © 2012 IOP Publishing Ltd.
Compact Fourier transform spectrometer without moving parts
Huang, Chu-Yu; Estroff, B.; Wang, Wei-Chih
2012-04-01
Fourier transform spectroscopy (FTS) is a potent analytical tool for chemical and biological analysis, but is limited by system size, expense, and robustness. To make FTS technology more accessible, we present a compact, inexpensive FTS system based on a novel liquid crystal (LC) interferometer. This design is unique because the optical path difference (OPD) is controlled by voltage applied to the LC cell. The OPD is further improved by reflecting the polarized incident light through the LC several times before reaching the second polarizer and measurement. This paper presents the theoretical model and numerical simulations for the liquid crystal Fourier transform spectrometer (LCFTS), and experimental results from the prototype. Based on the experimental results, the LCFTS performs in accordance with the theoretical predictions, achieving a maximum OPD of 210μm and a resolution of 1nm at a wavelength of 630nm. The instrumental response refresh rate is just under 1 second. Absorbance measurements were conducted for single and mixed solutions of deionized water and isopropyl alcohol, demonstrating agreement with a commercial system and literature values. We also present the LCFTS transmission spectra for varying concentrations of potassium permanganate to show system sensitivity.
Fourier Transforms Simplified: Computing an Infrared Spectrum from an Interferogram
Hanley, Quentin S.
2012-01-01
Fourier transforms are used widely in chemistry and allied sciences. Examples include infrared, nuclear magnetic resonance, and mass spectroscopies. A thorough understanding of Fourier methods assists the understanding of microscopy, X-ray diffraction, and diffraction gratings. The theory of Fourier transforms has been presented in this "Journal",…
A high-resolution Fourier Transform Spectrometer for planetary spectroscopy
Cruikshank, D. P.; Sinton, W. M.
1973-01-01
The employment of a high-resolution Fourier Transform Spectrometer (FTS) is described for planetary and other astronomical spectroscopy in conjunction with the 88-inch telescope at Mauna Kea Observatory. The FTS system is designed for a broad range of uses, including double-beam laboratory spectroscopy, infrared gas chromatography, and nuclear magnetic resonance spectroscopy. The data system is well-suited to astronomical applications because of its great speed in acquiring and transforming data, and because of the enormous storage capability of the magnetic tape unit supplied with the system. The basic instrument is outlined 2nd some of the initial results from the first attempted use on the Mauna Kea 88-inch telescope are reported.
From fractional Fourier transformation to quantum mechanical fractional squeezing transformation
Institute of Scientific and Technical Information of China (English)
吕翠红; 范洪义; 李东韡
2015-01-01
By converting the triangular functions in the integration kernel of the fractional Fourier transformation to the hy-perbolic function, i.e., tanα→tanhα, sinα→sinhα, we find quantum mechanical fractional squeezing transformation (FrST) which satisfies additivity. By virtue of the integration technique within ordered product of operators (IWOP) wederive the unitary operator responsible for the FrST, which is composite and is made of eiπa†a/2 and exp[ iα2 (a2+a†2)]. The FrST may be implemented in combinations of quadratic nonlinear crystals with different phase mismatches.
Wang, Yilin; Ding, Chunguang; Du, Kehe; Xiao, Yao; Wu, Caisheng; Zhang, Jinlan; Qin, Hailin; Du, Guanhua
2009-09-01
Xiao-xu-ming decoction (XXMD) prescription is a traditional Chinese prescription that has been widely used to treat theoplegia and the sequela of theoplegia. Modern pharmacological research has also indicated that the active fraction from XXMD is able to treat cardiovascular diseases and Alzheimer's disease. In the study reported here, high-performance liquid chromatography coupled with Fourier transform ion cyclotron resonance mass spectrometry (HPLC/FTICR-MS) was developed to identify active compounds and their metabolites after oral administration of active fraction from Xiao-xu-ming decoction to rats, using parent mass list triggered data-dependent multiple-stage mass analysis at a resolving power of 100,000 in the external calibration mode. The mass accuracies obtained for full-scan MS were within 2 ppm in most cases. Fifteen constituents were identified in the active fraction from XXMD and the biological samples of rats. The fragmentation behaviors of these constituents were summarized which would be helpful for structural characterization. The profiles of the constituents in the active fraction and biological samples of rats were obtained which provided us with much information for a better understanding of the chemical basis of the pharmacologic actions of XXMD.
Fourier transform infrared spectroscopy for Mars science
Anderson, Mark S.; Andringa, Jason M.; Carlson, Robert W.; Conrad, Pamela; Hartford, Wayne; Shafer, Michael; Soto, Alejandro; Tsapin, Alexandre I.; Dybwad, Jens Peter; Wadsworth, Winthrop; Hand, Kevin
2005-03-01
Presented here is a Fourier transform infrared spectrometer (FTIR) for field studies that serves as a prototype for future Mars science applications. Infrared spectroscopy provides chemical information that is relevant to a number of Mars science questions. This includes mineralogical analysis, nitrogen compound recognition, truth testing of remote sensing measurements, and the ability to detect organic compounds. The challenges and scientific opportunities are given for the in situ FTIR analysis of Mars soil and rock samples. Various FTIR sampling techniques are assessed and compared to other analytical instrumentation. The prototype instrument presented is capable of providing field analysis in a Mars analog Antarctic environment. FTIR analysis of endolithic microbial communities in Antarctic rocks and a Mars meteor are given as analytical examples.
Uncertainty relation for the discrete Fourier transform.
Massar, Serge; Spindel, Philippe
2008-05-16
We derive an uncertainty relation for two unitary operators which obey a commutation relation of the form UV=e(i phi) VU. Its most important application is to constrain how much a quantum state can be localized simultaneously in two mutually unbiased bases related by a discrete fourier transform. It provides an uncertainty relation which smoothly interpolates between the well-known cases of the Pauli operators in two dimensions and the continuous variables position and momentum. This work also provides an uncertainty relation for modular variables, and could find applications in signal processing. In the finite dimensional case the minimum uncertainty states, discrete analogues of coherent and squeezed states, are minimum energy solutions of Harper's equation, a discrete version of the harmonic oscillator equation. PMID:18518426
Applying Quaternion Fourier Transforms for Enhancing Color Images
Directory of Open Access Journals (Sweden)
M.I. Khalil
2012-03-01
Full Text Available The Fourier transforms play a critical role in a broad range of image processing applications, including enhancement, analysis, restoration, and compression. Until recently, it was common to use the conventional methods to deal with colored images. These methods are based on RGB decomposition of the colored image by separating it into three separate scalar images and computing the Fourier transforms of these images separately. The computing of the Hypercomplex 2D Fourier transform of a color image as a whole unit has only recently been realized. This paper is concerned with frequency domain noise reduction of color images using quaternion Fourier transforms. The approach is based on obtaining quaternion Fourier transform of the color image and applying the Gaussian filter to it in the frequency domain. The filtered image is then obtained by calculating the inverse quaternion Fourier transforms.
Fourier-Laguerre transform, convolution and wavelets on the ball
McEwen, J D
2013-01-01
We review the Fourier-Laguerre transform, an alternative harmonic analysis on the three-dimensional ball to the usual Fourier-Bessel transform. The Fourier-Laguerre transform exhibits an exact quadrature rule and thus leads to a sampling theorem on the ball. We study the definition of convolution on the ball in this context, showing explicitly how translation on the radial line may be viewed as convolution with a shifted Dirac delta function. We review the exact Fourier-Laguerre wavelet transform on the ball, coined flaglets, and show that flaglets constitute a tight frame.
Weighted inequalities for Hilbert transforms and multiplicators of Fourier transforms
Directory of Open Access Journals (Sweden)
Kokilashvili V
1997-01-01
Full Text Available As is well known, invariant operators with a shift can be bounded from into only if . We show that the case might also hold for weighted spaces. We derive the sufficient conditions for the validity of strong (weak type inequalities for the Hilbert transform when . The examples of couple of weights which guarantee the fulfillness of two-weighted strong (weak type inequalities for singular integrals are presented. The method of proof of the main results allows us to generalize the results of this paper to the singular integrals which are defined on homogeneous groups. The Fourier multiplier theorem is also proved.
Fourier transform spectroscopy for future planetary missions
Brasunas, John C.; Hewagama, Tilak; Kolasinski, John R.; Kostiuk, Theodor
2015-11-01
Thermal-emission infrared spectroscopy is a powerful tool for exploring the composition, temperature structure, and dynamics of planetary atmospheres; and the temperature of solid surfaces. A host of Fourier transform spectrometers (FTS) such as Mariner IRIS, Voyager IRIS, and Cassini CIRS from NASA Goddard have made and continue to make important new discoveries throughout the solar system.Future FTS instruments will have to be more sensitive (when we concentrate on the colder, outer reaches of the solar system), and less massive and less power-hungry as we cope with decreasing resource allotments for future planetary science instruments. With this in mind, NASA Goddard was funded via the Planetary Instrument Definition and Development Progrem (PIDDP) to develop CIRS-lite, a smaller version of the CIRS FTS for future planetary missions. Following the initial validation of CIRS-lite operation in the laboratory, we have been acquiring atmospheric data in the 8-12 micron window at the 1.2 m telescope at the Goddard Geophysical and Astronomical Observatory (GGAO) in Greenbelt, MD. Targets so far have included Earth's atmosphere (in emission, and in absorption against the moon), and Venus.We will present the roadmap for making CIRS-lite a viable candidate for future planetary missions.
AN ANOMALY INTRUSION DETECTION METHOD USING FOURIER TRANSFORM
Institute of Scientific and Technical Information of China (English)
Yue Bing; Zhao Yuexia; Xu Zhoujun; Fu Hongjuan; Ma Fengning
2004-01-01
A set of discrete points obtained from audit records on a behavior session is processed with Fourier transform. The criterion of selecting Fourier transform coefficients is introduced, and is used to find a unified value from the set of coefficients. This unified value is compared with a threshold to determine whether the session is abnormal. Finally simple test results are reported.
Fiber Optic Fourier Transform White-Light Interferometry
Institute of Scientific and Technical Information of China (English)
Yi Jiang; Cai-Jie Tang
2008-01-01
Fiber optic Fourier transform white-light inter-fereometry is presented to interrogate the absolute optical path difference of an Mach-Zehnder inter-ferometer. The phase change of the interferometer caused by scanning wavelength can be calculated by a Fourier transform-based phase demodulation technique. A linear output is achieved.
Realization of quantum discrete Fourier transform with NMR
Institute of Scientific and Technical Information of China (English)
无
2000-01-01
The pulse sequences of the logic operations used in quantum discrete Fourier transform are designed for the experiment of nuclear magnetic resonance(NMR), and 2-qubit discrete Fourier transforms are implemented experimentally with NMR. The experimental errors are examined and methods for reducing the errors are proposed.
CORRELATION THEOREMS FOR TYPE II QUATERNION FOURIER TRANSFORM
Bahri, Mawardi; Ashino, Ryuichi
2013-01-01
We present the correlation within the framework of the quaternion algebra. We establish the correlation theorem for type II quaternion Fourier transform (QFT) and obtain some important properties of the relationship between the quaternion correlation and the type II QFT. Keywords: quaternion correlation; quaternion Fourier transform
Generalized Fourier-grid R-matrix theory: a discrete Fourier-Riccati-Bessel transform approach
Energy Technology Data Exchange (ETDEWEB)
Layton, E.G. (Joint Inst. for Lab. Astrophysics, Boulder, CO (United States)); Stade, E. (Colorado Univ., Boulder, CO (United States). Dept. of Mathematics)
1993-08-28
We present the latest developments in the Fourier-grid R-matrix theory of scattering. These developments are based on the generalized Fourier-grid formalism and use a new type of extended discrete Fourier transform: the discrete Fourier-Riccati-Bessel transform. We apply this new R-matrix approach to problems of potential scattering, to demonstrate how this method reduces computational effort by incorporating centrifugal effects into the representation. As this technique is quite new, we have hopes to broaden the formalism to many types of problems. (author).
Generalized Fourier-grid R-matrix theory: a discrete Fourier-Riccati-Bessel transform approach
International Nuclear Information System (INIS)
We present the latest developments in the Fourier-grid R-matrix theory of scattering. These developments are based on the generalized Fourier-grid formalism and use a new type of extended discrete Fourier transform: the discrete Fourier-Riccati-Bessel transform. We apply this new R-matrix approach to problems of potential scattering, to demonstrate how this method reduces computational effort by incorporating centrifugal effects into the representation. As this technique is quite new, we have hopes to broaden the formalism to many types of problems. (author)
Cryogenic Scan Mechanism for Fourier Transform Spectrometer
Brasunas, John C.; Francis, John L.
2011-01-01
A compact and lightweight mechanism has been developed to accurately move a Fourier transform spectrometer (FTS) scan mirror (a cube corner) in a near-linear fashion with near constant speed at cryogenic temperatures. This innovation includes a slide mechanism to restrict motion to one dimension, an actuator to drive the motion, and a linear velocity transducer (LVT) to measure the speed. The cube corner mirror is double-passed in one arm of the FTS; double-passing is required to compensate for optical beam shear resulting from tilting of the moving cube corner. The slide, actuator, and LVT are off-the-shelf components that are capable of cryogenic vacuum operation. The actuator drives the slide for the required travel of 2.5 cm. The LVT measures translation speed. A proportional feedback loop compares the LVT voltage with the set voltage (speed) to derive an error signal to drive the actuator and achieve near constant speed. When the end of the scan is reached, a personal computer reverses the set voltage. The actuator and LVT have no moving parts in contact, and have magnetic properties consistent with cryogenic operation. The unlubricated slide restricts motion to linear travel, using crossed roller bearings consistent with 100-million- stroke operation. The mechanism tilts several arc seconds during transport of the FTS mirror, which would compromise optical fringe efficiency when using a flat mirror. Consequently, a cube corner mirror is used, which converts a tilt into a shear. The sheared beam strikes (at normal incidence) a flat mirror at the end of the FTS arm with the moving mechanism, thereby returning upon itself and compensating for the shear
Color image registration based on quaternion Fourier transformation
Wang, Qiang; Wang, Zhengzhi
2012-05-01
The traditional Fourier Mellin transform is applied to quaternion algebra in order to investigate quaternion Fourier transformation properties useful for color image registration in frequency domain. Combining with the quaternion phase correlation, we propose a method for color image registration based on the quaternion Fourier transform. The registration method, which processes color image in a holistic manner, is convenient to realign color images differing in translation, rotation, and scaling. Experimental results on different types of color images indicate that the proposed method not only obtains high accuracy in similarity transform in the image plane but also is computationally efficient.
A discrete Fourier transform for virtual memory machines
Galant, David C.
1992-01-01
An algebraic theory of the Discrete Fourier Transform is developed in great detail. Examination of the details of the theory leads to a computationally efficient fast Fourier transform for the use on computers with virtual memory. Such an algorithm is of great use on modern desktop machines. A FORTRAN coded version of the algorithm is given for the case when the sequence of numbers to be transformed is a power of two.
Research progress of the fractional Fourier transform in signal processing
Institute of Scientific and Technical Information of China (English)
无
2006-01-01
The fractional Fourier transform is a generalization of the classical Fourier transform, which is introduced from the mathematic aspect by Namias at first and has many applications in optics quickly. Whereas its potential appears to have remained largely unknown to the signal processing community until 1990s. The fractional Fourier transform can be viewed as the chirp-basis expansion directly from its definition, but essentially it can be interpreted as a rotation in the time-frequency plane, i.e. the unified time-frequency transform. With the order from 0 increasing to 1, the fractional Fourier transform can show the characteristics of the signal changing from the time domain to the frequency domain. In this research paper, the fractional Fourier transform has been comprehensively and systematically treated from the signal processing point of view. Our aim is to provide a course from the definition to the applications of the fractional Fourier transform, especially as a reference and an introduction for researchers and interested readers.
Embolic Doppler ultrasound signal detection via fractional Fourier transform.
Gençer, Merve; Bilgin, Gökhan; Aydın, Nizamettin
2013-01-01
Computerized analysis of Doppler ultrasound signals can aid early detection of asymptomatic circulating emboli. For analysis, physicians use informative features extracted from Doppler ultrasound signals. Time -frequency analysis methods are useful tools to exploit the transient like signals such as Embolic signals. Detection of discriminative features would be the first step toward automated analysis of embolic Doppler ultrasound signals. The most problematic part of setting up emboli detection system is to differentiate embolic signals from confusing similar wave-like patterns such as Doppler speckle and artifacts caused by tissue movement, probe tapping, speaking etc. In this study, discrete version of fractional Fourier transform is presented as a solution in the detection of emboli in digitized Doppler ultrasound signals. An accurate set of parameters are extracted using short time Fourier transform and fractional Fourier transform and the results are compared to reveal detection quality. Experimental results prove the efficiency of fractional Fourier transform in which discriminative features becomes more evident.
Electro-Optic Imaging Fourier Transform Spectral Polarimeter Project
National Aeronautics and Space Administration — Boulder Nonlinear Systems, Inc. (BNS) proposes to develop an Electro-Optic Imaging Fourier Transform Spectral Polarimeter (E-O IFTSP). The polarimetric system is...
Signature Recognition using Multi Scale Fourier Descriptor And Wavelet Transform
Ismail, Ismail A; danaf, Talaat S El; Samak, Ahmed H
2010-01-01
This paper present a novel off-line signature recognition method based on multi scale Fourier Descriptor and wavelet transform . The main steps of constructing a signature recognition system are discussed and experiments on real data sets show that the average error rate can reach 1%. Finally we compare 8 distance measures between feature vectors with respect to the recognition performance. Key words: signature recognition; Fourier Descriptor; Wavelet transform; personal verification
Simple optical setup implementation for digital Fourier transform holography
Energy Technology Data Exchange (ETDEWEB)
De Oliveira, G N [Pos-graduacao em Engenharia Mecanica, TEM/PGMEC, Universidade Federal Fluminense, Rua Passo da Patria, 156, Niteroi, R.J., Cep.: 24.210-240 (Brazil); Rodrigues, D M C; Dos Santos, P A M, E-mail: pams@if.uff.br [Instituto de Fisica, Laboratorio de Optica Nao-linear e Aplicada, Universidade Federal Fluminense, Av. Gal. Nilton Tavares de Souza, s/n, Gragoata, Niteroi, R.J., Cep.:24.210-346 (Brazil)
2011-01-01
In the present work a simple implementation of Digital Fourier Transform Holography (DFTH) setup is discussed. This is obtained making a very simple modification in the classical setup arquiteture of the Fourier Transform holography. It is also demonstrated the easy and practical viability of the setup in an interferometric application for mechanical parameters determination. The work is also proposed as an interesting advanced introductory training for graduated students in digital holography.
Convolution Theorems for Quaternion Fourier Transform: Properties and Applications
Mawardi Bahri
2013-01-01
General convolution theorems for two-dimensional quaternion Fourier transforms(QFTs) are presented. It is shown that these theorems are not only valid for real-valued functions, but also for quaternion-valued functions. We describe some useful properties of generalized convolutions and compare with the convolution theorems of the classical Fourier transform. We finally apply the obtained results to study hypoellipticity and to solve the heat equation in quaternion algebra fr...
Convolution Theorems for Quaternion Fourier Transform: Properties and Applications
Directory of Open Access Journals (Sweden)
Mawardi Bahri
2013-01-01
Full Text Available General convolution theorems for two-dimensional quaternion Fourier transforms (QFTs are presented. It is shown that these theorems are valid not only for real-valued functions but also for quaternion-valued functions. We describe some useful properties of generalized convolutions and compare them with the convolution theorems of the classical Fourier transform. We finally apply the obtained results to study hypoellipticity and to solve the heat equation in quaternion algebra framework.
Convolution Theorems for Quaternion Fourier Transform: Properties and Applications
Mawardi Bahri; Ryuichi Ashino; Rémi Vaillancourt
2013-01-01
General convolution theorems for two-dimensional quaternion Fourier transforms (QFTs) are presented. It is shown that these theorems are valid not only for real-valued functions but also for quaternion-valued functions. We describe some useful properties of generalized convolutions and compare them with the convolution theorems of the classical Fourier transform. We finally apply the obtained results to study hypoellipticity and to solve the heat equation in quaternion algebra framework.
Quantum Fourier Transform and Phase Estimation in Qudit System
Institute of Scientific and Technical Information of China (English)
CAO Ye; PENG Shi-Guo; ZHENG Chao; LONG Gui-Lu
2011-01-01
The quantum Fourier transform and quantum phase estimation are the key components for many quantum algorithms, such as order-finding, factoring, and etc.In this article, the general procedure of quantum Fourier transform and phase estimation are investigated for high dimensional case.They can be seen as subroutines in a main program run in a qudit quantum computer, and the quantum circuits are given.
Rotation-invariant texture analysis using Radon and Fourier transforms
Institute of Scientific and Technical Information of China (English)
Songshan Xiao; Yongxing Wu
2007-01-01
@@ Texture analysis is a basic issue in image processing and computer vision, and how to attain the rotationinvariant texture characterization is a key problem. This paper proposes a rotation-invariant texture analysis technique using Radon and Fourier transforms. This method uses Radon transform to convert rotation to translation, then utilizes Fourier transform and takes the moduli of the Fourier transform of these functions to make the translation invariant. A k-nearest-neighbor rule is employed to classify texture images. The proposed method is robust to additive white noise as a result of summing pixel values to generate projections in the Radon transform step. Experiment results show the feasibility of the proposed method and its robustness to additive white noise.
Dynamic measurement of deformation using Fourier transform digital holographic interferometry
Gao, Xinya; Wu, Sijin; Yang, Lianxiang
2013-10-01
Digital holographic interferometry (DHI) is a well-established optical technique for measurement of nano-scale deformations. It has become more and more important due to the rapid development of applications in aerospace engineering and biomedicine. Traditionally, phase shift technique is used to quantitatively measure the deformations in DHI. However, it cannot be applied in dynamic measurement. Fourier transform phase extraction method, which can determine the phase distribution from only a single hologram, becomes a promising method to extract transient phases in DHI. This paper introduces a digital holographic interferometric system based on 2D Fourier transform phase extraction method, with which deformations of objects can be measured quickly. In the optical setup, the object beam strikes a CCD via a lens and aperture, and the reference beam is projected on the CCD through a single-mode fiber. A small inclination angle between the diverging reference beam and optical axial is introduced in order to physically separate the Fourier components in frequency domain. Phase maps are then obtained by the utilization of Fourier transform and windowed inverse Fourier transform. The capability of the Fourier transform DHI is discussed by theoretical discussion as well as experiments.
SAW chirp Fourier transform for MB-OFDM UWB receiver
Institute of Scientific and Technical Information of China (English)
HE Peng-fei; L(U) Ying-hua; ZHANG Hong-xin; WANG Ye-qiu; XU Yong
2006-01-01
In the conventional multiband orthogonal frequency division multiplexing ultra wideband (MB-OFDM UWB )receiver, the fast Fourier transform (FFT) algorithm is realized by the expensive and power-consuming digital signal processor (DSP) chips. In this article, the lower power, lower cost, and lower complexity real-time analog surface acoustic wave (SAW)chirp Fourier transform devices were used to replace the DSP part. A MB-OFDM UWB receiver based on the M-C-M SAW chirp Fourier transform was presented, and the step of signal transformation from input signals was also depicted. The simulation results show that the proposed receiver provides similar bit error performance compared to the fully digital receiver when used in the channel environments proposed by the IEEE 802.15SG3a.
Electro-Optical Imaging Fourier-Transform Spectrometer
Chao, Tien-Hsin; Zhou, Hanying
2006-01-01
An electro-optical (E-O) imaging Fourier-transform spectrometer (IFTS), now under development, is a prototype of improved imaging spectrometers to be used for hyperspectral imaging, especially in the infrared spectral region. Unlike both imaging and non-imaging traditional Fourier-transform spectrometers, the E-O IFTS does not contain any moving parts. Elimination of the moving parts and the associated actuator mechanisms and supporting structures would increase reliability while enabling reductions in size and mass, relative to traditional Fourier-transform spectrometers that offer equivalent capabilities. Elimination of moving parts would also eliminate the vibrations caused by the motions of those parts. Figure 1 schematically depicts a traditional Fourier-transform spectrometer, wherein a critical time delay is varied by translating one the mirrors of a Michelson interferometer. The time-dependent optical output is a periodic representation of the input spectrum. Data characterizing the input spectrum are generated through fast-Fourier-transform (FFT) post-processing of the output in conjunction with the varying time delay.
Methods of theme presentation "The Fourier transform of impulse functions"
Directory of Open Access Journals (Sweden)
Faniya Ahmetova
2016-09-01
Full Text Available The paper considers the Fourier transform of impulse functions, which is the mathematical basis of the tasks associated with the theory of reception and signal conversion in optoelectronic system. The method of calculating of two functions convolution, its Fourier image and the image of the Fourier-Bessel axisymmetric functions are demonstrated in details. A table, which summarizes the analytical expression for the shifted impulse functions and records their Fourier transforms, is provided. A wide range of examples of solving tasks, containing the graphic illustration, is analyzed. A structured approach to the presentation of the material, which combines basic theoretical information and analysis of typical tasks, will help second-year students of optoelectronic specialty in their independent work and homework.
Estimates for Fourier transform of measures supported on singular hypersurfaces
International Nuclear Information System (INIS)
We consider hypersurfaces S is contained in R3 with zero Gaussian curvature at every ordinary point with surface measure dS and we define the surface measure dμ ψ(x)dS(x) for smooth function ψ with compact support. We obtain uniform estimates of Fourier transform of measures concentrated on such hypersurfaces. We show that due to the damping effect of the surface measure the Fourier transform decays faster than O(vertical bar ξ vertical bar-1/h), where h is the height of the phase function. In particular, Fourier transform of measures supported on the exceptional surfaces decays as O(vertical bar ξ vertical bar-1/2). (author)
Energy Technology Data Exchange (ETDEWEB)
Rodriguez G, A.; Bowtell, R.; Mansfield, P. [Area de Procesamiento Digital de Senales e Imagenes Biomedicas. Universidad Autonoma Metropolitana Iztapalapa. Mexico D.F. 09340 Mexico (Mexico)
1998-12-31
Velocity maps were studied combining Doyle and Mansfield method (1986) with each of the following transforms: Fourier, window Fourier and wavelet (Mexican hat). Continuous wavelet transform was compared against the two Fourier transform to determine which technique is best suited to study blood maps generated by Half Fourier Echo-Planar Imaging. Coefficient images were calculated and plots of the pixel intensity variation are presented. Finally, contour maps are shown to visualize the behavior of the blood flow in the cardiac chambers for the wavelet technique. (Author)
Recording Fractional Fourier Transform Hologram Using Holographic Zone Plate
Institute of Scientific and Technical Information of China (English)
高峰; 曾阳素; 张怡霄; 杨静; 高福华; 郭永康
2002-01-01
FRTH(fractional Fourier transform hologram) is a new kind of hologram that differs from common Fresnel holograms and Fourier transform holograms. Due to the flexibility of zone plate. A method that uses the -1 order diffraction wave of zone plate as the object wave and the 0 order diffraction wave as the reference wave to record FRTH is presented. It provides a new simple way to record FRTH. In this paper, the theory of achieving FRT and recording FRTH using holographic zone plate is presented and experimental results are given.
Multifractional Fourier Transform Method and Its Applications to Image Encryption
Institute of Scientific and Technical Information of China (English)
RANQiwen; WANGQi; MAJing; TANLiying
2003-01-01
The multiplicity of the fractional Fourier transform(FRFT),which is intrinsic in any fractional operator,has been claimed by several authors,but never across-the-board developed.Particularly,the weight-type FRFT(WFRFT) has not been investigated.Starting with defining the multifractional Fourier transform (MFRFT),we gained the generalization permutation matrix group (GPMG)representation and multiplicity of the MFRFT,and the relationships among the MFRFT the standard WFRFT and the standard CFRFT.Finally,as a application,a novel image encryption method hased on the MFRFT is propounded.Similation results show that this method is safe,practicable and impactful.
Extending Fourier transformations to Hamilton's quaternions and Clifford's geometric algebras
Hitzer, Eckhard
2013-10-01
We show how Fourier transformations can be extended to Hamilton's algebra of quaternions. This was initially motivated by applications in nuclear magnetic resonance and electric engineering. Followed by an ever wider range of applications in color image and signal processing. Hamilton's algebra of quaternions is only one example of the larger class of Clifford's geometric algebras, complete algebras encoding a vector space and all its subspace elements. We introduce how Fourier transformations are extended to Clifford algebras and applied in electromagnetism, and in the processing of images, color images, vector field and climate data.
Fourier and Hadamard transform spectrometers - A limited comparison
Tai, M. H.; Harwit, M.
1976-01-01
An encoding figure of merit is established for a detector-noise limited Fourier transform spectrometer (FTS) and compared to the comparable figure for a Hadamard transform spectrometer (HTS). The limitation of the Fourier system is partly that it does not truly Fourier analyze the radiation. Instead a cosine squared modulation is imposed on the different spectral frequencies. An additional difficulty is that neither the cosine nor the cosine squared functions form an orthonormal set. This makes the Fellgett's advantage (root-mean-squared figure of merit) for a single detector Michelson interferometer a factor of the square root of (N/8) greater than for a conventional grating instrument - rather than the square root of (N/2). The theoretical limit would be the square root of N.
Fourier transformation methods in the field of gamma spectrometry
Indian Academy of Sciences (India)
A Abdel-Hafiez
2006-09-01
The basic principles of a new version of Fourier transformation is presented. This new version was applied to solve some main problems such as smoothing, and denoising in gamma spectroscopy. The mathematical procedures were first tested by simulated data and then by actual experimental data.
Quantum Discrete Fourier Transform in an Ion Trap System
Institute of Scientific and Technical Information of China (English)
ZHENG Shi-Biao
2007-01-01
We propose two schemes for the implementation of quantum discrete Fourier transform in the ion trap system. In each scheme we design a tunable two-qubit phase gate as the main ingredient. The experimental implementation of the schemes would be an important step toward complex quantum computation in the ion trap system.
Quaternion Fourier transforms for signal and image processing
Ell, Todd A; Sangwine, Stephen J
2014-01-01
Based on updates to signal and image processing technology made in the last two decades, this text examines the most recent research results pertaining to Quaternion Fourier Transforms. QFT is a central component of processing color images and complex valued signals. The book's attention to mathematical concepts, imaging applications, and Matlab compatibility render it an irreplaceable resource for students, scientists, researchers, and engineers.
SPICA/SAFARI fourier transform spectrometer mechanism evolutionary design
Dool, T.C. van den; Kruizinga, B.; Braam, B.C.; Hamelinck, R.F.M.M.; Loix, N.; Loon, D. van; Dams, J.
2012-01-01
TNO, together with its partners, have designed a cryogenic scanning mechanism for use in the SAFARI Fourier Transform Spectrometer (FTS) on board of the SPICA mission. SPICA is one of the M-class missions competing to be launched in ESA's Cosmic Vision Programme in 2022. JAXA leads the development o
Quasi- Chun- Ching Shih's Fractional Fourier Transform with Periodicity of 2,3 and M
Institute of Scientific and Technical Information of China (English)
FAN Xi-zhi
2004-01-01
Based on Chun-Ching Shih's idea, the basic transform was substituted and the quasi-ChunChing Shih's fractional Fourier transform with periodicity of 2, 3 and M was deduced. The two former transforms and the Chun-Ching Shih's fractional Fourier transform were only the particular cases of quasiChun-Ching Shih's fractional Fourier transform with periodicity of M.
On the finite Fourier transforms of functions with infinite discontinuities
Directory of Open Access Journals (Sweden)
Branko Saric
2002-01-01
Full Text Available The introductory part of the paper is provided to give a brief review of the stability theory of a matrix pencil for discrete linear time-invariant singular control systems, based on the causal relationship between Jordan's theorem from the theory of Fourier series and Laurent's theorem from the calculus of residues. The main part is concerned with the theory of the integral transforms, which has proved to be a powerful tool in the control systems theory. On the basis of a newly defined notion of the total value of improper integrals, throughout the main part of the paper, an attempt has been made to present the global theory of the integral transforms, which are slightly more general with respect to the Laplace and Fourier transforms. The paper ends with examples by which the results of the theory are verified.
An introduction to Laplace transforms and Fourier series
Dyke, Phil
2014-01-01
Laplace transforms continue to be a very important tool for the engineer, physicist and applied mathematician. They are also now useful to financial, economic and biological modellers as these disciplines become more quantitative. Any problem that has underlying linearity and with solution based on initial values can be expressed as an appropriate differential equation and hence be solved using Laplace transforms. In this book, there is a strong emphasis on application with the necessary mathematical grounding. There are plenty of worked examples with all solutions provided. This enlarged new edition includes generalised Fourier series and a completely new chapter on wavelets. Only knowledge of elementary trigonometry and calculus are required as prerequisites. An Introduction to Laplace Transforms and Fourier Series will be useful for second and third year undergraduate students in engineering, physics or mathematics, as well as for graduates in any discipline such as financial mathematics, econometrics and ...
A Student's Guide to Fourier Transforms - 2nd Edition
James, J. F.
2002-09-01
Fourier transform theory is of central importance in a vast range of applications in physical science, engineering, and applied mathematics. This new edition of a successful student text provides a concise introduction to the theory and practice of Fourier transforms, using qualitative arguments wherever possible and avoiding unnecessary mathematics. After a brief description of the basic ideas and theorems, the power of the technique is then illustrated by referring to particular applications in optics, spectroscopy, electronics and telecommunications. The rarely discussed but important field of multi-dimensional Fourier theory is covered, including a description of computer-aided tomography (CAT-scanning). The final chapter discusses digital methods, with particular attention to the fast Fourier transform. Throughout, discussion of these applications is reinforced by the inclusion of worked examples. The book assumes no previous knowledge of the subject, and will be invaluable to students of physics, electrical and electronic engineering, and computer science. Expanded to include more emphasis on applications An established successful textbook for undergraduate and graduate students Includes worked examples and copious diagrams throughout
The Fast Fourier Transform and its application to tidal oscillations
Directory of Open Access Journals (Sweden)
A. dos Santos Franco
1971-06-01
Full Text Available This paper proposes a new way of tidal spectral analysis based on the Cooley-Tukey algorithm, known as the Fast Fourier Transform. The Fast Fourier Transform analysis is used to compute both the harmonic constants of the tide and the power spectrum.The latter is obtained by means of a weighted sum. A new way is also derived to obtain the formula giving the number of the degrees of freedom,on which is based the confi dence interval corresponding to the noise spectrum.Este trabalho propõe um novo caminho para a analise espectral da maré baseada no algoritmo de Cooley-Tukey. A análise através da "Transformação Rápida de Fourier" (Fast Fourier Transform - FFT é empregada tanto para calcular as constantes harmônicas da maré quanto para a obtenção do espectro de energia. Este é calculado por meio de uma soma ponderada. Também é dada uma nova dedução da fórmula que exprime o número de graus de liberdade em que se baseia o intervalo de confiança correspondente ao espectro do ruído. O trabalho foi redigido em inglês a fim de facilitar o intercâmbio de informações.
Directory of Open Access Journals (Sweden)
E. Zumelzu
2015-01-01
Full Text Available The persistent adhesion of salmon muscle to food container walls after treatment with urea solution was observed. This work evaluated the diffusion of antibiotics from the salmon muscle to the polyethylene terephthalate (PET coating protecting the electrolytic chromium coated steel (ECCS plates. New aquaculture production systems employ antibiotics such as florfenicol, florfenicol amine, oxytetracycline, and erythromycin to control diseases. The introduction of antibiotics is a matter of concern regarding the effects on human health and biodiversity. It is important to determine their impact on the adhesion of postmortem salmon muscle to can walls and the surface and structural changes affecting the functionality of multilayers. This work characterized the changes occurring in the multilayer PET polymer and steel of containers by electron microscopy, 3D atomic force microscopy (3D-AFM, X-ray photoelectron spectroscopy (XPS, and Fourier transform infrared spectroscopy (FT-IR analyses. A robust mass spectrometry methodology was employed to determine the presence of antibiotic residues. No evidence of antibiotics was observed on the protective coating in the range between 0.001 and 2.0 ng/mL; however, the presence of proteins, cholesterol, and alpha-carotene was detected. This in-depth profiling of the matrix-level elements is relevant for the use of adequate materials in the canning export industry.
Fractional Fourier Transform for Ultrasonic Chirplet Signal Decomposition
Directory of Open Access Journals (Sweden)
Yufeng Lu
2012-01-01
Full Text Available A fractional fourier transform (FrFT based chirplet signal decomposition (FrFT-CSD algorithm is proposed to analyze ultrasonic signals for NDE applications. Particularly, this method is utilized to isolate dominant chirplet echoes for successive steps in signal decomposition and parameter estimation. FrFT rotates the signal with an optimal transform order. The search of optimal transform order is conducted by determining the highest kurtosis value of the signal in the transformed domain. A simulation study reveals the relationship among the kurtosis, the transform order of FrFT, and the chirp rate parameter in the simulated ultrasonic echoes. Benchmark and ultrasonic experimental data are used to evaluate the FrFT-CSD algorithm. Signal processing results show that FrFT-CSD not only reconstructs signal successfully, but also characterizes echoes and estimates echo parameters accurately. This study has a broad range of applications of importance in signal detection, estimation, and pattern recognition.
Fourier transform approach in modulation technique of experimental measurements.
Khazimullin, M V; Lebedev, Yu A
2010-04-01
An application of Fourier transform approach in modulation technique of experimental studies is considered. This method has obvious advantages compared with traditional lock-in amplifiers technique--simple experimental setup, a quickly available information on all the required harmonics, high speed of data processing using fast Fourier transform algorithm. A computationally simple, fast and accurate Fourier coefficients interpolation (FCI) method has been implemented to obtain a useful information from harmonics of a multimode signal. Our analysis shows that in this case FCI method has a systematical error (bias) of a signal parameters estimation, which became essential for the short data sets. Hence, a new differential Fourier coefficients interpolation (DFCI) method has been suggested, which is less sensitive to a presence of several modes in a signal. The analysis has been confirmed by simulations and measurements of a quartz wedge birefringence by means of the photoelastic modulator. The obtained bias, noise level, and measuring speed are comparable and even better than in lock-in amplifier technique. Moreover, presented DFCI method is expected to be promised candidate for using in actively developing imaging systems based on the modulation technique requiring fast digital signal processing of large data sets.
A simple scanning mirror mechanism for a Fourier transform spectrometer
Brasunas, John C.; Cushman, G. Mark
1998-04-01
We present a simple scanning mirror mechanism for a Fourier transform spectrometer, consisting of a voice coil actuator, a ball-bearing slide, and a single cube corner mirror or retroreflector. The use of a single cube corner reduces significantly the sensitivity of the Fourier transform spectrometer to the tilt of its moving mirror, enabling the simple mechanism. With simple velocity feedback using a linear velocity transducer, we achieve a scan speed stability of 2%-5% (root-mean-square) for scan speeds from 40 to 440 μm/s for a travel of 2.2 cm, corresponding to a spectral resolution of 0.06 cm-1. Fringe amplitude stability is about 5%. This scan mechanism was operated at standard temperature and pressure, but the relatively minute amount of lubricant in the ball-bearing slide may indicate possible operation under vacuum and/or cryogenic conditions.
Fourier transform spectrometer based on Fabry-Perot interferometer.
Al-Saeed, Tarek A; Khalil, Diaa A
2016-07-10
We analyze the Fourier transform spectrometer based on a symmetric/asymmetric Fabry-Perot interferometer. In this spectrometer, the interferogram is obtained by recording the intensity as a function of the interferometer length. Then, we recover the spectrum by applying the discrete Fourier transform (DFT) directly on the interferogram. This technique results in spectral harmonic overlap and fictitious wavenumber components outside the original spectral range. For this purpose, in this work, we propose a second method to recover the spectrum. This method is based on expanding the DFT of the interferogram and the spectrum by a Haar or box function. By this second method, we recovered the spectrum and got rid of the fictitious spectral components and spectral harmonic overlap.
On-chip photonic Fourier transform with surface plasmon polaritons
Institute of Scientific and Technical Information of China (English)
Shan Shan Kou; Guanghui Yuan; Qian wang; Luping Du; Eugeniu Balaur; Daohua Zhang; Dingyuan Tang
2016-01-01
The Fourier transform (FT),a cornerstone of optical processing,enables rapid evaluation of fundamental mathematical operations,such as derivatives and integrals.Conventionally,a converging lens performs an optical FT in free space when light passes through it.The speed of the transformation is limited by the thickness and the focal length of the lens.By usingthe wave nature of surface plasmon polaritons (SPPs),here we demonstrate that the FT can be implemented in a planar configuration with a minimal propagation distance of around 10 μm,resulting in an increase of speed by four to five orders of magnitude.The photonic FT was tested by synthesizing intricate SPP waves with their Fourier components.The reduced dimensionality in the minuscule device allows the future development of an ultrafast on-chip photonic information processing platform for large-scale optical computing.
On the Fourier transform of the greatest common divisor
van der Kamp, Peter H
2012-01-01
The discrete Fourier transform of the greatest common divisor is a multiplicative function that generalises both the gcd-sum function and Euler's totient function. On the one hand it is the Dirichlet convolution of the identity with Ramanujan's sum, and on the other hand it can be written as a generalised convolution product of the identity with the totient function. We show that this arithmetic function of two integers (a,m) counts the number of elements in the set of ordered pairs (i,j) such that i*j is equivalent to a modulo m. Furthermore we generalise a dozen known identities for the totient function, to identities which involve the discrete Fourier transform of the greatest common divisor, including its partial sums, and its Lambert series.
Optimal color image restoration: Wiener filter and quaternion Fourier transform
Grigoryan, Artyom M.; Agaian, Sos S.
2015-03-01
In this paper, we consider the model of quaternion signal degradation when the signal is convoluted and an additive noise is added. The classical model of such a model leads to the solution of the optimal Wiener filter, where the optimality with respect to the mean square error. The characteristic of this filter can be found in the frequency domain by using the Fourier transform. For quaternion signals, the inverse problem is complicated by the fact that the quaternion arithmetic is not commutative. The quaternion Fourier transform does not map the convolution to the operation of multiplication. In this paper, we analyze the linear model of the signal and image degradation with an additive independent noise and the optimal filtration of the signal and images in the frequency domain and in the quaternion space.
Multiparty Quantum Secret Sharing Using Quantum Fourier Transform
Institute of Scientific and Technical Information of China (English)
HUANG Da-Zu; CHEN Zhi-Gang; GUO Ying
2009-01-01
A (n, n )-threshold scheme of multiparty quantum secret sharing of classical or quantum message is proposed based on the discrete quantum Fourier transform.In our proposed scheme, the secret message, which is encoded by using the forward quantum Fourier transform and decoded by using the reverse, is split and shared in such a way that it can be reconstructed among them only if all the participants work in concert.Furthermore, we also discuss how this protocol must be carefully designed for correcting errors and checking eavesdropping or a dishonest participant.Security analysis shows that our scheme is secure.Also, this scheme has an advantage that it is completely compatible with quantum computation and easier to realize in the distributed quantum secure computation.
Time-frequency representation measurement based on temporal Fourier transformation
Suen, Yifan; Xiao, Shaoqiu; Hao, Sumin; Zhao, Xiaoxiang; Xiong, Yigao; Liu, Shenye
2016-10-01
We propose a new scheme to physically realize the short-time Fourier transform (STFT) of chirped optical pulse using time-lens array that enables us to get time-frequency representation without using FFT algorithm. The time-lens based upon the four-wave mixing is used to perform the process of temporal Fourier transformation. Pump pulse is used for both providing the quadratic phase and being the window function of STFT. The idea of STFT is physically realized in our scheme. Simulations have been done to investigate performance of the time-frequency representation scheme (TFRS) in comparison with STFT using FFT algorithm. Optimal measurement of resolution in time and frequency has been discussed.
Scaled Fractional Fourier Transform for Partially Coherent Beams
Institute of Scientific and Technical Information of China (English)
蔡阳健; 林强
2003-01-01
The definition of scaled fractional Fourier transform (SFRT) is extended to partially coherent beams based directly on the cross-spectral density. Based on this formula, an equivalent tensor ABCD law for the SFRT of partially coherent twisted anisotropic Gaussian-Schell model beams is derived. The derived formulae provide a powerful tool for analysing and calculating the SFRT of partially coherent beams. An application example is provided.
Measured Quantum Fourier Transform of 1024 Qubits on Fiber Optics
Tomita, Akihisa; Nakamura, Kazuo
2004-01-01
Quantum Fourier transform (QFT) is a key function to realize quantum computers. A QFT followed by measurement was demonstrated on a simple circuit based on fiber-optics. The QFT was shown to be robust against imperfections in the rotation gate. Error probability was estimated to be 0.01 per qubit, which corresponded to error-free operation on 100 qubits. The error probability can be further reduced by taking the majority of the accumulated results. The reduction of error probability resulted ...
Homology of balanced complexes via the Fourier transform
Meshulam, Roy
2011-01-01
Let G_0,...,G_k be finite abelian groups and let G_0*...*G_k be the join of the 0-dimensional complexes G_i. We give a characterization of the integral k-coboundaries of subcomplexes of G_0*...*G_k in terms of the Fourier transform on the group G_0 \\times ... \\times G_k. This leads to an extension of a recent result of Musiker and Reiner on a topological interpretation of the cyclotomic polynomial.
Quantum operation, quantum Fourier transform and semi-definite programming
Duan, Runyao; Ji, Zhengfeng; Feng, Yuan; Ying, Mingsheng
2003-01-01
We analyze a class of quantum operations based on a geometrical representation of $d-$level quantum system (or qudit for short). A sufficient and necessary condition of complete positivity, expressed in terms of the quantum Fourier transform, is found for this class of operations. A more general class of operations on qudits is also considered and its completely positive condition is reduced to the well-known semi-definite programming problem.
Structural Characterization of Carbohydrates by Fourier Transform Tandem Mass Spectrometry
Zhou, Wen; Håkansson, Kristina
2011-01-01
Fourier transform tandem mass spectrometry (MS/MS) provides high mass accuracy, high sensitivity, and analytical versatility and has therefore emerged as an indispensable tool for structural elucidation of biomolecules. Glycosylation is one of the most common posttranslational modifications, occurring in ~50% of proteins. However, due to the structural diversity of carbohydrates, arising from non-template driven biosynthesis, achievement of detailed structural insight is highly challenging. T...
Entangled Fractional Fourier Transform for the Multipartite Entangled State Representation
Institute of Scientific and Technical Information of China (English)
QIAN Xiao-Qing; SONG Tong-Qiang
2006-01-01
We deduce entangled fractional Fourier transformation (EFFT) for the multipartite entangled state representation, which was newly constructed with two mutually conjugate n-mode entangled states of continuum variables in n-mode Fock space. We establish a formalism of EFFT for quantum mechanical wave functions, which provides us a convenient way to derive some wave functions. We find that the eigenmode of EFFT is different from the usual Hermite Polynomials. We also derive the EFFT of the n-mode squeezed state.
Apparatus and methods for continuous beam fourier transform mass spectrometry
McLuckey, Scott A.; Goeringer, Douglas E.
2002-01-01
A continuous beam Fourier transform mass spectrometer in which a sample of ions to be analyzed is trapped in a trapping field, and the ions in the range of the mass-to-charge ratios to be analyzed are excited at their characteristic frequencies of motion by a continuous excitation signal. The excited ions in resonant motions generate real or image currents continuously which can be detected and processed to provide a mass spectrum.
10th International Conference on Progress in Fourier Transform Spectroscopy
Keresztury, Gábor; Kellner, Robert
1997-01-01
19 plenary lectures and 203 poster papers presented at the 10th International Conference of Fourier Transform Spectroscopy in Budapest 1995 give an overview on the state-of-the art of this technology and its wide range of applications. The reader will get information on any aspects of FTS including the latest instrumental developments, e.g. in diode array detection, time resolution FTS, microscopy and spectral mapping, double modulation and two-dimensional FTS.
Fourier-transform Ghost Imaging with Hard X-rays
Yu, Hong; Han, Shensheng; Xie, Honglan; Du, Guohao; Xiao, Tiqiao; Zhu, Daming
2016-01-01
Knowledge gained through X-ray crystallography fostered structural determination of materials and greatly facilitated the development of modern science and technology in the past century. Atomic details of sample structures is achievable by X-ray crystallography, however, it is only applied to crystalline structures. Imaging techniques based on X-ray coherent diffraction or zone plates are capable of resolving the internal structure of non-crystalline materials at nanoscales, but it is still a challenge to achieve atomic resolution. Here we demonstrate a novel lensless Fourier-transform ghost imaging method with pseudo-thermal hard X-rays by measuring the second-order intensity correlation function of the light. We show that high resolution Fourier-transform diffraction pattern of a complex structure can be achieved at Fresnel region, and the amplitude and phase distributions of a sample in spatial domain can be retrieved successfully. The method of lensless X-ray Fourier-transform ghost imaging extends X-ray...
Design of high-resolution Fourier transform lens
Zhang, Lei; Zhong, Xing; Jin, Guang
2007-12-01
With the development of optical information processing, high-resolution Fourier transform lens has often been used in holographic data storage system, spatial filtering and observation of particles. This paper studies the optical design method of high-resolution Fourier transform optical lenses system, which could be used in particles observation and holographic data storage system. According to Fourier transform relation between object and its frequency plane and the theory of geometrical optics, the system with working wavelength 532nm and resolution 3μm was designed based on ZEMAX. A multi-configuration method was adopted to optimize the system's lenses. In the optical system, a diaphragm was placed at the system's spectrum plane and the system demanded a low vacuum to cut down the influences of atmosphere and other particles. The result of finite element analysis indicated that the influences of vacuum pumping to optics spacing and mirror surface shape very minor, and the imaging quality not being affected. This system has many advantages, such as simple structure, good image quality and a high resolution of 3μm. So it has a wide application prospect and can be used both in holographic data storage system and particles observation.
Mass spectral peak distortion due to Fourier transform signal processing.
Rockwood, Alan L; Erve, John C L
2014-12-01
Distortions of peaks can occur when one uses the standard method of signal processing of data from the Orbitrap and other FT-based methods of mass spectrometry. These distortions arise because the standard method of signal processing is not a linear process. If one adds two or more functions, such as time-dependent signals from a Fourier transform mass spectrometer and performs a linear operation on the sum, the result is the same as if the operation was performed on separate functions and the results added. If this relationship is not valid, the operation is non-linear and can produce unexpected and/or distorted results. Although the Fourier transform itself is a linear operator, the standard algorithm for processing spectra in Fourier transform-based methods include non-linear mathematical operators such that spectra processed by the standard algorithm may become distorted. The most serious consequence is that apparent abundances of the peaks in the spectrum may be incorrect. In light of these considerations, we performed theoretical modeling studies to illustrate several distortion effects that can be observed, including abundance distortions. In addition, we discuss experimental systems where these effects may manifest, including suggested systems for study that should demonstrate these peak distortions. Finally, we point to several examples in the literature where peak distortions may be rationalized by the phenomena presented here.
Improved Spectral Representation for Birdcall Based on Fractional Fourier Transform
Institute of Scientific and Technical Information of China (English)
无
2008-01-01
A novel spectral representation based on fractional Fourier transform (FrFT) is proposed and applied to birdcall analysis. The FrFT-based spectrogram of a signal is derived and compared with its FT-based counterpart, and the spectrum gathering method is used to show the energy distribution related to the pitch frequency. The fixed transform order and adaptive orders for FrFT are tested. The fixed order can be obtained empirically or calculated according to the known chirp rate. The adaptive optimal orders are determined by using ambiguity function. Experimental results with birdcalls show that the FrFT-based spectrogram with an optimal transform order has higher resolution than its STFT-based counterpart, and the better performance can be achieved if adaptive orders are used.
Twin image elimination in digital holography by combination of Fourier transformations
Choudhury, Debesh
2013-01-01
We present a new technique for removing twin image in in-line digital Fourier holography using a combination of Fourier transformations. Instead of recording only a Fourier transform hologram of the object, we propose to record a combined Fourier transform hologram by simultaneously recording the hologram of the Fourier transform and the inverse Fourier transform of the object with suitable weighting coefficients. Twin image is eliminated by appropriate inverse combined Fourier transformation and proper choice of the weighting coefficients. An optical configuration is presented for recording combined Fourier transform holograms. Simulations demonstrate the feasibility of twin image elimination. The hologram reconstruction is sensitive to phase aberrations of the object, thereby opening a way for holographic phase sensing.
Denoise in the pseudopolar grid Fourier space using exact inverse pseudopolar Fourier transform
Wei, Fan Jun
2015-01-01
In this paper I show a matrix method to calculate the exact inverse pseudopolar grid Fourier transform, and use this transform to do noise removals in the k space of pseudopolar grids. I apply the Gaussian filter to this pseudopolar grid and find the advantages of the noise removals are very excellent by using pseudopolar grid, and finally I show the Cartesian grid denoise for comparisons. The results present the signal to noise ratio and the variance are much better when doing noise removals in the pseudopolar grid than the Cartesian grid. The noise removals of pseudopolar grid or Cartesian grid are both in the k space, and all these noises are added in the real space.
Fourier transform infrared studies in solid egg white lysozyme
International Nuclear Information System (INIS)
Fourier Transform Infrared (FTIR) Spectroscopy is the most recent addition to the arsenal of bioanalytical techniques capable of providing information about the secondary structure of proteins in a variety of environments. FTIR spectra have been obtained in solid egg white lysozyme. The spectra display the usual amide I, II and III bands. Secondary structural information obtained from the spectra after applying resolution enhancement techniques to the amide I band has been found consistent with the x-ray crystallographic data of the protein and also to the spectroscopic data of the protein in aqueous solution. (author). 17 refs, 6 figs, 2 tabs
Ash melting behavior by Fourier transform infrared spectroscopy
Institute of Scientific and Technical Information of China (English)
LI Han-xu; QIU Xiao-sheng; TANG Yong-xin
2008-01-01
A Fourier Transform Infrared Spectroscopic (FTIR) method involving a Fe2O3 flux was used to learn how China's coal ash melts. The relationship between ash fusion temperature and chemical composition, as well as the effects of Fe2O3 flux on the ash fusion temperature were studied. The relationship between ash fusion temperature and chemical composition, mineralogical phases and functional groups was analyzed with the FTIR method. The results show that the ash fusion temperature is related to the location and transmittance of certain absorption peaks, which is of great significance for the study of ash behavior.
Fourier Transform Infrared Spectroscopic Analysis of Protein Secondary Structures
Institute of Scientific and Technical Information of China (English)
Jilie KONG; Shaoning YU
2007-01-01
Infrared spectroscopy is one of the oldest and well established experimental techniques for the analysis of secondary structure of polypeptides and proteins. It is convenient, non-destructive, requires less sample preparation, and can be used under a wide variety of conditions. This review introduces the recent developments in Fourier transform infrared (FTIR) spectroscopy technique and its applications to protein structural studies. The experimental skills, data analysis, and correlations between the FTIR spectroscopic bands and protein secondary structure components are discussed. The applications of FTIR to the secondary structure analysis, conformational changes, structural dynamics and stability studies of proteins are also discussed.
Birefringent Fourier transform imaging spectrometer with a rotating retroreflector.
Bai, Caixun; Li, Jianxin; Shen, Yan; Zhou, Jianqiang
2016-08-01
A birefringent Fourier transform imaging spectrometer with a new lateral shearing interferometer is presented. The interferometer includes a Wollaston prism and a retroreflector. It splits an incident light beam into two shearing parallel parts to obtain interference fringe patterns of an imaging target, which is well established as an aid in reducing problems associated with optical alignment and manufacturing precision. Continuously rotating the retroreflector enables the spectrometer to acquire two-dimensional spectral images without spatial scanning. This technology, with a high work efficiency and low complexity, is inherently compact and robust. The effectiveness of the proposed method is demonstrated by the experimental results. PMID:27472640
Fast Computation of Voigt Functions via Fourier Transforms
Mendenhall, M H
2006-01-01
This work presents a method of computing Voigt functions and their derivatives, to high accuracy, on a uniform grid. It is based on an adaptation of Fourier-transform based convolution. The relative error of the result decreases as the fourth power of the computational effort. Because of its use of highly vectorizable operations for its core, it can be implemented very efficiently in scripting language environments which provide fast vector libraries. The availability of the derivatives makes it suitable as a function generator for non-linear fitting procedures.
Multi-band Image Registration Method Based on Fourier Transform
Institute of Scientific and Technical Information of China (English)
庹红娅; 刘允才
2004-01-01
This paper presented a registration method based on Fourier transform for multi-band images which is involved in translation and small rotation. Although different band images differ a lot in the intensity and features,they contain certain common information which we can exploit. A model was given that the multi-band images have linear correlations under the least-square sense. It is proved that the coefficients have no effect on the registration progress if two images have linear correlations. Finally, the steps of the registration method were proposed. The experiments show that the model is reasonable and the results are satisfying.
High-resolution wide-band Fast Fourier Transform spectrometers
Klein, Bernd; Hochgürtel, Stefan; Krämer, Ingo; Bell, Andreas; Meyer, Klaus; Güsten, Rolf
2012-01-01
We describe the performance of our latest generations of sensitive wide-band high-resolution digital Fast Fourier Transform Spectrometer (FFTS). Their design, optimized for a wide range of radio astronomical applications, is presented. Developed for operation with the GREAT far infrared heterodyne spectrometer on-board SOFIA, the eXtended bandwidth FFTS (XFFTS) offers a high instantaneous bandwidth of 2.5 GHz with 88.5 kHz spectral resolution and has been in routine operation during SOFIA's B...
Beam profile for the Herschel-SPIRE Fourier transform spectrometer.
Makiwa, Gibion; Naylor, David A; Ferlet, Marc; Salji, Carl; Swinyard, Bruce; Polehampton, Edward; van der Wiel, Matthijs H D
2013-06-01
One of the instruments on board the Herschel Space Observatory is the Spectral and Photometric Imaging Receiver (SPIRE). SPIRE employs a Fourier transform spectrometer with feed-horn-coupled bolometers to provide imaging spectroscopy. To interpret the resultant spectral images requires knowledge of the wavelength-dependent beam, which in the case of SPIRE is complicated by the use of multimoded feed horns. In this paper we describe a series of observations and the analysis conducted to determine the wavelength dependence of the SPIRE spectrometer beam profile.
A high-resolution Fourier-transform infrared spectrometer.
Johnson, H. L.; Forbes, F. F.; Thompson, R. I.; Steinmetz , D. L.; Harris, O.
1973-01-01
We have developed a Fourier-transform infrared spectrometer having a resolution of 0.5/cm over the range of wavelength from 1 to 5.5 microns. It has been used to observe the sun over this wavelength range from a Lear Jet flying at an altitude of 14 km, and to observe a number of stars from the ground, using the 229-cm telescope of the Steward Observatory and the 152-cm aluminum-mirror telescope at the Observatorio Astronomico Nacional in the Sierra de San Pedro Martir, Baja California, Mexico. The solar spectrum is given here, while the ground-based spectra are being published separately.
Double passing the Kitt Peak 1-m Fourier transform spectrometer
Jennings, D. E.; Hubbard, R.; Brault, J. W.
1985-01-01
Attention is given to a simple technique for performing the conversion of the Kitt Peak 1-m Fourier transform spectrometer's dual input/output optical configuration to a double pass configuration that improves spectral resolution by a factor of 2. The modification is made by placing a flat mirror in the output beam from each cat's eye, retroreflecting the beams back through the cat's eyes to the first beam splitter. A single detector is placed at the second input port, which then becomes the instrument's output.
Dispersive Fourier transform spectroscopy with gases in the visible region
Kerl, K.; Häusler, H.
1984-05-01
The method of dispersive Fourier transform spectroscopy (DFTS) with gases in the visible wavenumber range is described in detail and compared with the method of scanning-wavelength interferometry (SWI). Measurements of the dispersion of the complex refractive index of gases can be performed successively in several minutes using the same apparatus and gas sample conditions for both methods. In the reported experiments with CH 4 a very simple mirror drive was used. Nevertheless, reasonable results are obtained for the dispersion of the real refractive index of CH 4 in the wavenumber range 16,000 ⩽ σ ⩽ 23,000 cm -1.
The Non-uniform Fast Fourier Transform in Computed Tomography
Tang, Junqi
2016-01-01
This project is aimed at designing the fast forward projection algorithm and also the backprojection algorithm for cone beam CT imaging systems with circular X-ray source trajectory. The principle of the designs is based on utilizing the potential computational efficiency which the Fourier Slice Theorem and the Non-uniform Fast Fourier Transform (NUFFT) will bring forth. In this Masters report, the detailed design of the NUFFT based forward projector including a novel 3D (derivative of) Radon space resampling method will be given. Meanwhile the complexity of the NUFFT based forward projector is analysed and compared with the non-Fourier based CT projector, and the advantage of the NUFFT based forward projection in terms of the computational efficiency is demonstrated in this report. Base on the design of the forward algorithm, the NUFFT based 3D direct reconstruction algorithm will be derived. The experiments will be taken to test the performance of the forward algorithm and the backprojection algorithm to sh...
Fourier transform infrared spectroscopy techniques for the analysis of drugs of abuse
Kalasinsky, Kathryn S.; Levine, Barry K.; Smith, Michael L.; Magluilo, Joseph J.; Schaefer, Teresa
1994-01-01
Cryogenic deposition techniques for Gas Chromatography/Fourier Transform Infrared (GC/FT-IR) can be successfully employed in urinalysis for drugs of abuse with detection limits comparable to those of the established Gas Chromatography/Mass Spectrometry (GC/MS) technique. The additional confidence of the data that infrared analysis can offer has been helpful in identifying ambiguous results, particularly, in the case of amphetamines where drugs of abuse can be confused with over-the-counter medications or naturally occurring amines. Hair analysis has been important in drug testing when adulteration of urine samples has been a question. Functional group mapping can further assist the analysis and track drug use versus time.
A Matrix Formulation of Discrete Chirp Fourier Transform Algorithms
Institute of Scientific and Technical Information of China (English)
Juan Pablo Soto Quiros; Domingo Rodriguez
2014-01-01
This work presents a computational matrix framework in terms of tensor signal algebra for the formulation of discrete chirp Fourier transform algorithms. These algorithms are used in this work to estimate the point target functions (impulse response functions) of multiple-input multiple-output (MIMO) synthetic aperture radar (SAR) systems. This estimation technique is being studied as an alternative to the estimation of point target functions using the discrete cross-ambiguity function for certain types of environmental surveillance applications. The tensor signal algebra is presented as a mathematics environment composed of signal spaces, finite dimensional linear operators, and special matrices where algebraic methods are used to generate these signal transforms as computational estimators. Also, the tensor signal algebra contributes to analysis, design, and implementation of parallel algorithms. An instantiation of the framework was performed by using the MATLAB Parallel Computing Toolbox, where all the algorithms presented in this paper were implemented.
Fourier transform for fermionic systems and the spectral tensor network.
Ferris, Andrew J
2014-07-01
Leveraging the decomposability of the fast Fourier transform, I propose a new class of tensor network that is efficiently contractible and able to represent many-body systems with local entanglement that is greater than the area law. Translationally invariant systems of free fermions in arbitrary dimensions as well as 1D systems solved by the Jordan-Wigner transformation are shown to be exactly represented in this class. Further, it is proposed that these tensor networks be used as generic structures to variationally describe more complicated systems, such as interacting fermions. This class shares some similarities with the Evenbly-Vidal branching multiscale entanglement renormalization ansatz, but with some important differences and greatly reduced computational demands.
Fast Fourier Transform Co-Processor (FFTC)- Towards Embedded GFLOPs
Kuehl, Christopher; Liebstueckel, Uwe; Tejerina, Isaac; Uemminghaus, Michael; Wite, Felix; Kolb, Michael; Suess, Martin; Weigand, Roland
2012-08-01
Many signal processing applications and algorithms perform their operations on the data in the transform domain to gain efficiency. The Fourier Transform Co- Processor has been developed with the aim to offload General Purpose Processors from performing these transformations and therefore to boast the overall performance of a processing module. The IP of the commercial PowerFFT processor has been selected and adapted to meet the constraints of the space environment.In frame of the ESA activity “Fast Fourier Transform DSP Co-processor (FFTC)” (ESTEC/Contract No. 15314/07/NL/LvH/ma) the objectives were the following:Production of prototypes of a space qualified version of the commercial PowerFFT chip called FFTC based on the PowerFFT IP.The development of a stand-alone FFTC Accelerator Board (FTAB) based on the FFTC including the Controller FPGA and SpaceWire Interfaces to verify the FFTC function and performance.The FFTC chip performs its calculations with floating point precision. Stand alone it is capable computing FFTs of up to 1K complex samples in length in only 10μsec. This corresponds to an equivalent processing performance of 4.7 GFlops. In this mode the maximum sustained data throughput reaches 6.4Gbit/s. When connected to up to 4 EDAC protected SDRAM memory banks the FFTC can perform long FFTs with up to 1M complex samples in length or multidimensional FFT- based processing tasks.A Controller FPGA on the FTAB takes care of the SDRAM addressing. The instructions commanded via the Controller FPGA are used to set up the data flow and generate the memory addresses.The presentation will give and overview on the project, including the results of the validation of the FFTC ASIC prototypes.
Fast Fourier Transform Co-processor (FFTC), towards embedded GFLOPs
Kuehl, Christopher; Liebstueckel, Uwe; Tejerina, Isaac; Uemminghaus, Michael; Witte, Felix; Kolb, Michael; Suess, Martin; Weigand, Roland; Kopp, Nicholas
2012-10-01
Many signal processing applications and algorithms perform their operations on the data in the transform domain to gain efficiency. The Fourier Transform Co-Processor has been developed with the aim to offload General Purpose Processors from performing these transformations and therefore to boast the overall performance of a processing module. The IP of the commercial PowerFFT processor has been selected and adapted to meet the constraints of the space environment. In frame of the ESA activity "Fast Fourier Transform DSP Co-processor (FFTC)" (ESTEC/Contract No. 15314/07/NL/LvH/ma) the objectives were the following: • Production of prototypes of a space qualified version of the commercial PowerFFT chip called FFTC based on the PowerFFT IP. • The development of a stand-alone FFTC Accelerator Board (FTAB) based on the FFTC including the Controller FPGA and SpaceWire Interfaces to verify the FFTC function and performance. The FFTC chip performs its calculations with floating point precision. Stand alone it is capable computing FFTs of up to 1K complex samples in length in only 10μsec. This corresponds to an equivalent processing performance of 4.7 GFlops. In this mode the maximum sustained data throughput reaches 6.4Gbit/s. When connected to up to 4 EDAC protected SDRAM memory banks the FFTC can perform long FFTs with up to 1M complex samples in length or multidimensional FFT-based processing tasks. A Controller FPGA on the FTAB takes care of the SDRAM addressing. The instructions commanded via the Controller FPGA are used to set up the data flow and generate the memory addresses. The paper will give an overview on the project, including the results of the validation of the FFTC ASIC prototypes.
Fourier-Transform Ghost Imaging with Hard X Rays
Yu, Hong; Lu, Ronghua; Han, Shensheng; Xie, Honglan; Du, Guohao; Xiao, Tiqiao; Zhu, Daming
2016-09-01
Knowledge gained through x-ray crystallography fostered structural determination of materials and greatly facilitated the development of modern science and technology in the past century. However, it is only applied to crystalline structures and cannot resolve noncrystalline materials. Here we demonstrate a novel lensless Fourier-transform ghost imaging method with pseudothermal hard x rays that extends x-ray crystallography to noncrystalline samples. By measuring the second-order intensity correlation function of the light, Fourier-transform diffraction pattern of a complex amplitude sample is achieved at the Fresnel region in our experiment and the amplitude and phase distributions of the sample in the spatial domain are retrieved successfully. For the first time, ghost imaging is experimentally realized with x rays. Since a highly coherent x-ray source is not required, the method can be implemented with laboratory x-ray sources and it also provides a potential solution for lensless diffraction imaging with fermions, such as neutrons and electrons where intensive coherent sources usually are not available.
Chiral Analysis of Isopulegol by Fourier Transform Molecular Rotational Spectroscopy
Evangelisti, Luca; Seifert, Nathan A.; Spada, Lorenzo; Pate, Brooks
2016-06-01
Chiral analysis on molecules with multiple chiral centers can be performed using pulsed-jet Fourier transform rotational spectroscopy. This analysis includes quantitative measurement of diastereomer products and, with the three wave mixing methods developed by Patterson, Schnell, and Doyle (Nature 497, 475-477 (2013)), quantitative determination of the enantiomeric excess of each diastereomer. The high resolution features enable to perform the analysis directly on complex samples without the need for chromatographic separation. Isopulegol has been chosen to show the capabilities of Fourier transform rotational spectroscopy for chiral analysis. Broadband rotational spectroscopy produces spectra with signal-to-noise ratio exceeding 1000:1. The ability to identify low-abundance (0.1-1%) diastereomers in the sample will be described. Methods to rapidly identify rotational spectra from isotopologues at natural abundance will be shown and the molecular structures obtained from this analysis will be compared to theory. The role that quantum chemistry calculations play in identifying structural minima and estimating their spectroscopic properties to aid spectral analysis will be described. Finally, the implementation of three wave mixing techniques to measure the enantiomeric excess of each diastereomer and determine the absolute configuration of the enantiomer in excess will be described.
Integrated optics in an electrically scanned imaging Fourier transform spectrometer
Breckinridge, James B. (Inventor); Ocallaghan, Fred G. (Inventor)
1982-01-01
An efficient, lightweight and stable, Fourier transform spectrometer was developed. The mechanical slide mechanism needed to create a path difference was eliminated by the use of retro-reflecting mirrors in a monolithic interferometer assembly in which the mirrors are not at 90 degrees to the propagation vector of the radiation, but rather at a small angle. The resulting plane wave fronts create a double-sided inteferogram of the source irradiance distribution which is detected by a charge-coupled device image sensor array. The position of each CCD pixel in the array is an indication of the path difference between the two retro-reflecting mirrors in the monolithic optical structure. The Fourier transform of the signals generated by the image sensor provide the spectral irradiance distribution of the source. For imaging, the interferometer assembly scans the source of irradiation by moving the entire instrument, such as would occur if it was fixedly mounted to a moving platform, i.e., a spacecraft. During scanning, the entrace slot to the monolithic optical structure sends different pixels to corresponding interferograms detected by adjacent columns of pixels of the image sensor.
[Fourier Transform Spectrometer Based on Rotating Parallel-Mirror-Pair].
Zhao, Bao-wei; Xiangli, Bin; Cai, Qi-sheng; Lü, Qun-bo; Zhou, Jin-song
2015-11-01
In the temporally-modulated Fourier transform spectroscopy, the translational moving mirror is difficult to drive accurately, causing tilt and shear problems. While, a rotational moving mirror can solve these problems. A rotary Fourier transform spectrometer is recommanded in this paper. Its principle is analyzed and the optical path difference is deduced. Also, the constrains for engineering realization are presented. This spectrometer consists of one beamsplitter, two fixed mirrors, one rotating parallel mirror pair, a collimating lens, a collecting lens, and one detector. From it's principle, this spectrometer show a simple structure, and it is assembled and adjustmented easily because the two split light are interfered with each other after reflected through the same plane mirror; By calculating the expression of it's optical path difference, the spectrometer is easy to realize large optical path difference, meaning high spectral resolution; Through analyzing it's engineering design constraints and computer simulation, it is known that the spectrometer should get the high resolution sample by high-speed spinning motor, so it is easy to achieve precise motion control, good stability, fast measurement speed.
Energy Technology Data Exchange (ETDEWEB)
Bartosch, T. [Erlanger-Nuernberg Univ., Erlanger (Germany). Lehrstul fuer Nachrichtentechnik I; Seidl, D. [Seismologisches Zentralobservatorium Graefenberg, Erlanegen (Greece). Bundesanstalt fuer Geiwissenschaften und Rohstoffe
1999-06-01
Among a variety of spectrogram methods short-time Fourier transform (STFT) and continuous wavelet transform (CWT) were selected to analyse transients in non-stationary signals. Depending on the properties of the tremor signals from the volcanos Mt. Stromboli, Mt. Semeru and Mt. Pinatubo were analyzed using both methods. The CWT can also be used to extend the definition of coherency into a time-varying coherency spectrogram. An example is given using array data from the volcano Mt. Stromboli (Italy).
Institute of Scientific and Technical Information of China (English)
FAN Hong-Yi
2004-01-01
We find quantum mechanical Fourier-Hankel representation transform for an electron moving in a uniform magnetic field. The physical meaning of Fourier decomposition states of electron's coordinate eigenstate and the momentum eigenstate are revealed.
Micro wishbone interferometer for Fourier transform infrared spectrometry
International Nuclear Information System (INIS)
A miniature wishbone-type Si interferometer with electrically actuated rotary comb drive actuators is designed and fabricated to apply a Fourier transform infrared (FTIR) spectrometer. Corner cube mirrors are mounted on the end of the Si interferometer that is formed on a glass substrate. The total size of the interferometer is approximately 8 mm × 8 mm. The corner cube mirrors with sharp edges with a size of approximately 1 × 1 × 0.5 mm3 are fabricated using an indentation technique. The rotation angle of rotary comb drive actuators is approximately 11° with an applied voltage of 180 V. Hereby, the maximum optical path difference of approximately 2640 µm is achieved, which corresponds to the highest resolution of ∼4 cm−1 as a spectrometer
Indirect Fourier transform in the context of statistical inference.
Muthig, Michael; Prévost, Sylvain; Orglmeister, Reinhold; Gradzielski, Michael
2016-09-01
Inferring structural information from the intensity of a small-angle scattering (SAS) experiment is an ill-posed inverse problem. Thus, the determination of a solution is in general non-trivial. In this work, the indirect Fourier transform (IFT), which determines the pair distance distribution function from the intensity and hence yields structural information, is discussed within two different statistical inference approaches, namely a frequentist one and a Bayesian one, in order to determine a solution objectively From the frequentist approach the cross-validation method is obtained as a good practical objective function for selecting an IFT solution. Moreover, modern machine learning methods are employed to suppress oscillatory behaviour of the solution, hence extracting only meaningful features of the solution. By comparing the results yielded by the different methods presented here, the reliability of the outcome can be improved and thus the approach should enable more reliable information to be deduced from SAS experiments. PMID:27580204
A rheumatoid arthritis study by Fourier transform infrared spectroscopy
Carvalho, Carolina S.; Silva, Ana Carla A.; Santos, Tatiano J. P. S.; Martin, Airton A.; dos Santos Fernandes, Ana Célia; Andrade, Luís E.; Raniero, Leandro
2012-01-01
Rheumatoid arthritis is a systemic inflammatory disease of unknown causes and a new methods to identify it in early stages are needed. The main purpose of this work is the biochemical differentiation of sera between normal and RA patients, through the establishment of a statistical method that can be appropriately used for serological analysis. The human sera from 39 healthy donors and 39 rheumatics donors were collected and analyzed by Fourier Transform Infrared Spectroscopy. The results show significant spectral variations with p<0.05 in regions corresponding to protein, lipids and immunoglobulins. The technique of latex particles, coated with human IgG and monoclonal anti-CRP by indirect agglutination known as FR and CRP, was performed to confirm possible false-negative results within the groups, facilitating the statistical interpretation and validation of the technique.
Spatially Resolved Fourier Transform Spectroscopy in the Extreme Ultraviolet
Jansen, G S M; Freisem, L; Eikema, K S E; Witte, S
2016-01-01
Coherent extreme ultraviolet (XUV) radiation produced by table-top high-harmonic generation (HHG) sources provides a wealth of possibilities in research areas ranging from attosecond physics to high resolution coherent imaging. However, it remains challenging to fully exploit the coherence of such sources for interferometry and Fourier transform spectroscopy (FTS). This is due to the need for a measurement system that is stable at the level of a wavelength fraction, yet allowing a controlled scanning of time delays. Here we demonstrate XUV interferometry and FTS in the 17-55 nm wavelength range using an ultrastable common-path interferometer suitable for high-intensity laser pulses that drive the HHG process. This approach enables the generation of fully coherent XUV pulse pairs with sub-attosecond timing variation, tunable time delay and a clean Gaussian spatial mode profile. We demonstrate the capabilities of our XUV interferometer by performing spatially resolved FTS on a thin film composed of titanium and...
Persian Sign Language Recognition Using Radial Distance and Fourier Transform
Directory of Open Access Journals (Sweden)
Bahare Jalilian
2013-11-01
Full Text Available This paper provides a novel hand gesture recognition method to recognize 32 static signs of the Persian Sign Language (PSL alphabets. Accurate hand segmentation is the first and important step in sign language recognition systems. Here, we propose a method for hand segmentation that helps to build a better vision based sign language recognition system. The proposed method is based on YCbCr color space, single Gaussian model and Bayes rule. It detects region of hand in complex background and non-uniform illumination. Hand gesture features are extracted by radial distance and Fourier transform. Finally, the Euclidean distanceis used to compute the similarity between the input signs and all training feature vectors in the database. The system is tested on 480 posture images of the PSL, 15 images for each 32 signs. Experimental results show that our approach is capable to recognize all 32 PSL alphabets with 95.62% recognition rate.
Control Of Cryogenic Fourier Transform Spectrometer Scanning Mirrors
Tripathi, S. S.; Gowrinathan, S.
1981-12-01
The Perkin-Elmer Corporation has designed and built a cryogenically cooled Fourier transform spectrometer for spaceborne applications. In operation, the spectrometer requires mirrors moving at constant velocity in both forward and reverse directions. To maintain efficiency and accuracy, the time taken to reverse direction and the vibration induced due to this reversal must be kept within low limits. This paper deals with the control system design for maintaining a constant velocity during forward and reverse scans and for smooth direction reversals. The systems aspects of the problem are described, and time-domain techniques of modern control theory are applied for optimization of turn-around profile. The analysis leads to a suboptimal design easily implemented by using analog-type components. Test results of satisfactory performance are also included.
Highly sensitive Fourier transform spectroscopy with LED sources
Serdyukov, V. I.; Sinitsa, L. N.; Vasil'chenko, S. S.
2013-08-01
It is shown that the use of high luminance LED emitters as a light source for Fourier transform spectrometers permits to enhance their threshold sensitivity in the visible range by orders of magnitude. Using a 2.5 W Edixeon EDEI-1LS3 emitter in the range of 11,350-11,700 cm-1 as a light source for the spectrometer with a 60-cm multipass cell during a 24-h measurement time, we have achieved a signal-to-noise ratio of 4.5 × 104 which corresponds to the minimal detectable absorption coefficient of 1.2 × 10-8 cm-1. Such enhanced sensitivity spectrometer has been used to measure the transition frequencies of CO2 vibrational bands 00051-00001 and 01151-01101 in the range of 11,400-11,500 cm-1.
Mid-Infrared Frequency Comb Fourier Transform Spectrometer
Adler, Florian; Foltynowicz, Aleksandra; Cossel, Kevin C; Briles, Travis C; Hartl, Ingmar; Ye, Jun
2010-01-01
Optical frequency-comb-based-high-resolution spectrometers offer enormous potential for spectroscopic applications. Although various implementations have been demonstrated, the lack of suitable mid-infrared comb sources has impeded explorations of molecular fingerprinting. Here we present for the first time a frequency-comb Fourier transform spectrometer operating in the 2100-to-3700-cm-1 spectral region that allows fast and simultaneous acquisitions of broadband absorption spectra with up to 0.0056 cm-1 resolution. We demonstrate part-per-billion detection limits in 30 seconds of integration time for various important molecules including methane, ethane, isoprene, and nitrous oxide. Our system enables precise concentration measurements even in gas mixtures that exhibit continuous absorption bands, and it allows detection of molecules at levels below the noise floor via simultaneous analysis of multiple spectral features. This system represents a near real-time, high-resolution, high-bandwidth mid-infrared sp...
Extreme-ultraviolet lensless Fourier-transform holography.
Lee, S H; Naulleau, P; Goldberg, K A; Cho, C H; Jeong, S; Bokor, J
2001-06-01
We demonstrate 100-nm-resolution holographic aerial image monitoring based on lensless Fourier-transform holography at extreme-UV (EUV) wavelengths, using synchrotron-based illumination. This method can be used to monitor the coherent imaging performance of EUV lithographic optical systems. The system has been implemented in the EUV phase-shifting point-diffraction interferometer recently developed at Lawrence Berkeley National Laboratory. Here we introduce the idea of the holographic aerial image-recording technique and present imaging performance characterization results for a 10x Schwarzschild objective, a prototype EUV lithographic optic. The results are compared with simulations, and good agreement is obtained. Various object patterns, including phase-shift-enhanced patterns, have been studied. Finally, the application of the holographic aerial image-recording technique to EUV multilayer mask-blank defect characterization is discussed. PMID:18357280
Multifunctional metasurface lens for imaging and Fourier transform
Wen, Dandan; Yue, Fuyong; Ardron, Marcus; Chen, Xianzhong
2016-06-01
A metasurface can manipulate light in a desirable manner by imparting local and space-variant abrupt phase change. Benefiting from such an unprecedented capability, the conventional concept of what constitutes an optical lens continues to evolve. Ultrathin optical metasurface lenses have been demonstrated based on various nanoantennas such as V-shape structures, nanorods and nanoslits. A single device that can integrate two different types of lenses and polarities is desirable for system integration and device miniaturization. We experimentally demonstrate such an ultrathin metasurface lens that can function either as a spherical lens or a cylindrical lens, depending on the helicity of the incident light. Helicity-controllable focal line and focal point in the real focal plane, as well as imaging and 1D/2D Fourier transforms, are observed on the same lens. Our work provides a unique tool for polarization imaging, image processing and particle trapping.
Instrument concept of the imaging Fourier transform spectrometer GLORIA
Directory of Open Access Journals (Sweden)
F. Friedl-Vallon
2014-10-01
Full Text Available The Gimballed Limb Observer for Radiance Imaging of the Atmosphere (GLORIA is an imaging limb emission sounder operating in the thermal infrared region. It is designed to provide measurements of the upper troposphere/lower stratosphere with high spatial and high spectral resolution. The instrument consists of an imaging Fourier transform spectrometer integrated into a gimbal. The assembly can be mounted in the belly pod of the German High Altitude and Long Range research aircraft (HALO and in instrument bays of the Russian M55 Geophysica. Measurements are made in two distinct modes: the chemistry mode emphasises chemical analysis with high spectral resolution, and the dynamics mode focuses on dynamical processes of the atmosphere with very high spatial resolution. In addition, the instrument allows tomographic analyses of air volumes. The first measurement campaigns have shown compliance with key performance and operational requirements.
Optical design of the ATMOS Fourier transform spectrometer
Abel, I. R.; Reynolds, B. R.; Breckinridge, J. B.; Pritchard, J.
1979-01-01
The optical system design of the ATMOS Fourier transform spectrometer to be operated from Spacelab for the measurement of stratospheric trace molecules is described. The design contains features which can achieve the required fringe contrast of 80% and spectral resolution of 0.02/cm over a spectral range of 2-16 microns. In particular, the design is based on the following features which alleviate the usual requirements for alignment precision: (1) 'cat's eye' mirror configuration in the two arms of the interferometer for retroreflection stability, (2) tilt-compensated system of beamsplitter, compensator, and fold mirrors for wavefront directional stability, (3) paraboloidal 'cat's eye' primary mirror for wavefront stability against shear, (4) rotatable compensator for matching chromatic dispersion, and (5) wedged refractive components to avoid channel spectra due to the Fabry-Perot effect.
Calibration of the Geostationary Imaging Fourier Transform Spectrometer (GIFTS)
Best, F. A.; Revercomb, H. E.; Bingham, G. E.; Knuteson, R. O.; Tobin, D. C.; LaPorte, D. D.; Smith, W. L.
2001-01-01
The NASA New Millennium Program's Geostationary Imaging Fourier Transform Spectrometer (GIFTS) requires highly accurate radiometric and spectral calibration in order to carry out its mission to provide water vapor, wind, temperature, and trace gas profiling from geostationary orbit. A calibration concept has been developed for the GIFTS Phase A instrument design. The in-flight calibration is performed using views of two on-board blackbody sources along with cold space. A radiometric calibration uncertainty analysis has been developed and used to show that the expected performance for GIFTS exceeds its top level requirement to measure brightness temperature to better than 1 K. For the Phase A GIFTS design, the spectral calibration is established by the highly stable diode laser used as the reference for interferogram sampling, and verified with comparisons to atmospheric calculations.
A direct digital synthesis chirped pulse Fourier transform microwave spectrometer.
Finneran, Ian A; Holland, Daniel B; Carroll, P Brandon; Blake, Geoffrey A
2013-08-01
Chirped pulse Fourier transform microwave (CP-FTMW) spectrometers have become the instrument of choice for acquiring rotational spectra, due to their high sensitivity, fast acquisition rate, and large bandwidth. Here we present the design and capabilities of a recently constructed CP-FTMW spectrometer using direct digital synthesis (DDS) as a new method for chirped pulse generation, through both a suite of extensive microwave characterizations and deep averaging of the 10-14 GHz spectrum of jet-cooled acetone. The use of DDS is more suited for in situ applications of CP-FTMW spectroscopy, as it reduces the size, weight, and power consumption of the chirp generation segment of the spectrometer all by more than an order of magnitude, while matching the performance of traditional designs. The performance of the instrument was further improved by the use of a high speed digitizer with dedicated signal averaging electronics, which facilitates a data acquisition rate of 2.1 kHz.
Observing Extended Sources with the \\Herschel SPIRE Fourier Transform Spectrometer
Wu, Ronin; Etxaluze, Mireya; Makiwa, Gibion; Naylor, David A; Salji, Carl; Swinyard, Bruce M; Ferlet, Marc; van der Wiel, Matthijs H D; Smith, Anthony J; Fulton, Trevor; Griffin, Matt J; Baluteau, Jean-Paul; Benielli, Dominique; Glenn, Jason; Hopwood, Rosalind; Imhof, Peter; Lim, Tanya; Lu, Nanyao; Panuzzo, Pasquale; Pearson, Chris; Sidher, Sunil; Valtchanov, Ivan
2013-01-01
The Spectral and Photometric Imaging Receiver (SPIRE) on the European Space Agency's Herschel Space Observatory utilizes a pioneering design for its imaging spectrometer in the form of a Fourier Transform Spectrometer (FTS). The standard FTS data reduction and calibration schemes are aimed at objects with either a spatial extent much larger than the beam size or a source that can be approximated as a point source within the beam. However, when sources are of intermediate spatial extent, neither of these calibrations schemes is appropriate and both the spatial response of the instrument and the source's light profile must be taken into account and the coupling between them explicitly derived. To that end, we derive the necessary corrections using an observed spectrum of a fully extended source with the beam profile and the source's light profile taken into account. We apply the derived correction to several observations of planets and compare the corrected spectra with their spectral models to study the beam c...
A Fourier transform Raman spectrometer with visible laser excitation
Dzsaber, S; Bernáth, B; Gyüre, B; Fehér, T; Kramberger, C; Pichler, T; Simon, F
2014-01-01
We present the development and performance of a Fourier transformation (FT) based Raman spectrometer working with visible laser (532 nm) excitation. It is generally thought that FT-Raman spectrometers are not viable in the visible range where shot-noise limits the detector performance and therein they are outperformed by grating based, dispersive ones. We show that contrary to this common belief, the recent advances of high-performance interference filters makes the FT-Raman design a valid alternative to dispersive Raman spectrometers for samples which do not luminesce. We critically compare the performance of our spectrometer to two dispersive ones: a home-built single channel and a state-of-the-art CCD based instruments. We demonstrate a similar or even better sensitivity than the CCD based dispersive spectrometer particularly when the laser power density is considered. The instrument possesses all the known advantages of the FT principle of spectral accuracy, high throughput, and economic design. We also d...
[Influence of collimation system on static Fourier transform spectrometer].
Jiang, Cheng-Zhi; Liang, Jing-Qiu; Liang, Zhong-Zhu; Sun, Qiang; Wang, Wei-Biao
2014-01-01
Collimation system provides collimated light for the static Fourier-transform spectroscopy (SFTS). Its quality is crucial to the signal to noise ratio (SNR) of SFTS. In the present paper, the physical model of SFTS was established based on the Fresnel diffraction theory by means of numerical software. The influence of collimation system on the SFTS was discussed in detail focusing on the aberrations of collimation lens and the quality of extended source. The results of simulation show that the influences of different kinds of aberrations on SNR take on obvious regularity, and in particular, the influences of off-axis aberrations on SNR are closely related to the location of off-axis point source. Finally the extended source's maximum radius allowed was obtained by simulation, which equals to 0.65 mm. The discussion results will be used for the design of collimation system.
How to tickle spins with a fourier transform NMR spectrometer.
Segawa, Takuya F; Carnevale, Diego; Bodenhausen, Geoffrey
2013-02-01
In the long bygone days of continuous-wave nuclear magnetic resonance (NMR) spectroscopy, a selected transition within a multiplet of a high-resolution spectrum could be irradiated by a highly selective continuous-wave (CW) radio-frequency (rf) field with a very weak amplitude ω(2)/(2π)≤J. This causes splittings of connected transitions, allowing one to map the connectivities of all transitions within the energy-level diagram of the spin system. Such "tickling" experiments stimulated the invention of two-dimensional spectroscopy, but seem to have been forgotten for nearly 50 years. We show that tickling can readily be achieved in homonuclear systems with Fourier transform spectrometers by applying short pulses in the intervals between the sampling points. Extensions to heteronuclear systems are even more straightforward since they can be carried out using very weak CW rf fields.
Tow-dimensional Strain Analysis by Fourier Transform Moire Interferometry
International Nuclear Information System (INIS)
Moire interferometry using a diffraction grating and a laser is a powerful technique for analyzing small deformation of a specimen. In the method, the x and y-directional fringe patterns are obtained by using the x and y-directional sets of two beams. If the both sets of two beams are simultaneously incident to the specimen, the x and y-directional fringe patterns are super imposed. In this case, it is difficult to separate each directional fringe pattern. Therefore each fringe pattern has been separately recorded by selecting each set of two beams. In order to analyze a two-dimensional strain changing with time, Moire interferometry using the two-dimensional fourier transform method is proposed and the x and y-directional fringes are separated. By this method, the thermal deformation of a glass plate is analyzed
3-D Printed Slit Nozzles for Fourier Transform Microwave Spectroscopy
Dewberry, Chris; Mackenzie, Becca; Green, Susan; Leopold, Ken
2015-06-01
3-D printing is a new technology whose applications are only beginning to be explored. In this report, we describe the application of 3-D printing to the facile design and construction of supersonic nozzles. The efficacy of a variety of designs is assessed by examining rotational spectra OCS and Ar-OCS using a Fourier transform microwave spectrometer with tandem cavity and chirped-pulse capabilities. This work focuses primarily on the use of slit nozzles but other designs have been tested as well. New nozzles can be created for 0.50 or less each, and the ease and low cost should facilitate the optimization of nozzle performance (e.g., jet temperature or cluster size distribution) for the needs of any particular experiment.
Radix-3 Algorithm for Realization of Discrete Fourier Transform
Directory of Open Access Journals (Sweden)
M.Narayan Murty
2016-07-01
Full Text Available In this paper, a new radix-3 algorithm for realization of discrete Fourier transform (DFT of length N = 3m (m = 1, 2, 3,... is presented. The DFT of length N can be realized from three DFT sequences, each of length N/3. If the input signal has length N, direct calculation of DFT requires O (N 2 complex multiplications (4N 2 real multiplications and some additions. This radix-3 algorithm reduces the number of multiplications required for realizing DFT. For example, the number of complex multiplications required for realizing 9-point DFT using the proposed radix-3 algorithm is 60. Thus, saving in time can be achieved in the realization of proposed algorithm.
Dispersive Fourier Transformation for Versatile Microwave Photonics Applications
Directory of Open Access Journals (Sweden)
Chao Wang
2014-12-01
Full Text Available Dispersive Fourier transformation (DFT maps the broadband spectrum of an ultrashort optical pulse into a time stretched waveform with its intensity profile mirroring the spectrum using chromatic dispersion. Owing to its capability of continuous pulse-by-pulse spectroscopic measurement and manipulation, DFT has become an emerging technique for ultrafast signal generation and processing, and high-throughput real-time measurements, where the speed of traditional optical instruments falls short. In this paper, the principle and implementation methods of DFT are first introduced and the recent development in employing DFT technique for widespread microwave photonics applications are presented, with emphasis on real-time spectroscopy, microwave arbitrary waveform generation, and microwave spectrum sensing. Finally, possible future research directions for DFT-based microwave photonics techniques are discussed as well.
Motion saliency detection using a temporal fourier transform
Chen, Zhe; Wang, Xin; Sun, Zhen; Wang, Zhijian
2016-06-01
Motion saliency detection aims at detecting the dynamic semantic regions in a video sequence. It is very important for many vision tasks. This paper proposes a new type of motion saliency detection method, Temporal Fourier Transform, for fast motion saliency detection. Different from conventional motion saliency detection methods that use complex mathematical models or features, variations in the phase spectrum of consecutive frames are identified and extracted as the key to obtaining the location of salient motion. As all the calculation is made on the temporal frequency spectrum, our model is independent of features, background models, or other forms of prior knowledge about scenes. The benefits of the proposed approach are evaluated for various videos where the number of moving objects, illumination, and background are all different. Compared with some the state of the art methods, our method achieves both good accuracy and fast computation.
Adaptive Controller for Compact Fourier Transform Spectrometer with Space Applications
Keymeulen, D.; Yiu, P.; Berisford, D. F.; Hand, K. P.; Carlson, R. W.; Conroy, M.
2014-12-01
Here we present noise mitigation techniques developed as part of an adaptive controller for a very compact Compositional InfraRed Interferometric Spectrometer (CIRIS) implemented on a stand-alone field programmable gate array (FPGA) architecture with emphasis on space applications in high radiation environments such as Europa. CIRIS is a novel take on traditional Fourier Transform Spectrometers (FTS) and replaces linearly moving mirrors (characteristic of Michelson interferometers) with a constant-velocity rotating refractor to variably phase shift and alter the path length of incoming light. The design eschews a monochromatic reference laser typically used for sampling clock generation and instead utilizes constant time-sampling via internally generated clocks. This allows for a compact and robust device, making it ideal for spaceborne measurements in the near-IR to thermal-IR band (2-12 µm) on planetary exploration missions. The instrument's embedded microcontroller is implemented on a VIRTEX-5 FPGA and a PowerPC with the aim of sampling the instrument's detector and optical rotary encoder in order to construct interferograms. Subsequent onboard signal processing provides spectral immunity from the noise effects introduced by the compact design's removal of a reference laser and by the radiation encountered during space flight to destinations such as Europa. A variety of signal processing techniques including resampling, radiation peak removal, Fast Fourier Transform (FFT), spectral feature alignment, dispersion correction and calibration processes are applied to compose the sample spectrum in real-time with signal-to-noise-ratio (SNR) performance comparable to laser-based FTS designs in radiation-free environments. The instrument's FPGA controller is demonstrated with the FTS to characterize its noise mitigation techniques and highlight its suitability for implementation in space systems.
Institute of Scientific and Technical Information of China (English)
Chujun Zheng; Peng Han; Hongsen Chang
2006-01-01
@@ A new one-step four-quadrant spatial phase-shifting Fourier transform digital holography is presented for recording of cosine transform coefficients, because cosine transform is a real-even symmetric Fourier transform. This approach implements four quadrant spatial phase shifting at a time using a special phase mask, which is located in the reference arm, and the phase distributions of its four-quadrants are 0, π/2, π,and 3π/2 respectively. The theoretical analysis and computer simulation results show that cosine transform coefficients of real-valued image can be calculated by capturing single four-quadrant spatial phase-shifting Fourier transform digital hologram.
Identification of formation interfaces by using wavelet and Fourier transforms
Mukherjee, Bappa; Srivardhan, V.; Roy, P. N. S.
2016-05-01
The identification of formation interfaces is of prime importance from well log data. The interfaces are not clearly discernible due to the presence of high and low frequency noise in the log response. Accurate bed boundary information is very crucial in hydrocarbon exploration and the problem has received considerable attention and many techniques have been proposed. Frequency spectrum based filtering techniques aids us in interpretation, but usually leads to inaccurate amplification of unwanted components of the log response. Wavelet transform is very effective in denoising the log response and can be carried out to filter low and high frequency components of signal. The use of Fourier and Wavelet transform in denoising the log data for obtaining formation interfaces is demonstrated in this work. The feasibility of the proposed technique is tested so that it can be used in the industry to decipher formation interfaces. The work flow is demonstrated by testing on wells belonging to the Upper Assam Basin, which are self-potential, gamma ray, and resistivity log responses.
The use of Fourier reverse transforms in crystallographic phase refinement
International Nuclear Information System (INIS)
Often a crystallographer obtains an electron density map which shows only part of the structure. In such cases, the phasing of the trial model is poor enough that the electron density map may show peaks in some of the atomic positions, but other atomic positions are not visible. There may also be extraneous peaks present which are not due to atomic positions. A method for determination of crystal structures that have resisted solution through normal crystallographic methods has been developed. PHASER is a series of FORTRAN programs which aids in the structure solution of poorly phased electron density maps by refining the crystallographic phases. It facilitates the refinement of such poorly phased electron density maps for difficult structures which might otherwise not be solvable. The trial model, which serves as the starting point for the phase refinement, may be acquired by several routes such as direct methods or Patterson methods. Modifications are made to the reverse transform process based on several assumptions. First, the starting electron density map is modified based on the fact that physically the electron density map must be non-negative at all points. In practice a small positive cutoff is used. A reverse Fourier transform is computed based on the modified electron density map. Secondly, the authors assume that a better electron density map will result by using the observed magnitudes of the structure factors combined with the phases calculated in the reverse transform. After convergence has been reached, more atomic positions and less extraneous peaks are observed in the refined electron density map. The starting model need not be very large to achieve success with PHASER; successful phase refinement has been achieved with a starting model that consists of only 5% of the total scattering power of the full molecule. The second part of the thesis discusses three crystal structure determinations
The use of Fourier reverse transforms in crystallographic phase refinement
Energy Technology Data Exchange (ETDEWEB)
Ringrose, S.
1997-10-08
Often a crystallographer obtains an electron density map which shows only part of the structure. In such cases, the phasing of the trial model is poor enough that the electron density map may show peaks in some of the atomic positions, but other atomic positions are not visible. There may also be extraneous peaks present which are not due to atomic positions. A method for determination of crystal structures that have resisted solution through normal crystallographic methods has been developed. PHASER is a series of FORTRAN programs which aids in the structure solution of poorly phased electron density maps by refining the crystallographic phases. It facilitates the refinement of such poorly phased electron density maps for difficult structures which might otherwise not be solvable. The trial model, which serves as the starting point for the phase refinement, may be acquired by several routes such as direct methods or Patterson methods. Modifications are made to the reverse transform process based on several assumptions. First, the starting electron density map is modified based on the fact that physically the electron density map must be non-negative at all points. In practice a small positive cutoff is used. A reverse Fourier transform is computed based on the modified electron density map. Secondly, the authors assume that a better electron density map will result by using the observed magnitudes of the structure factors combined with the phases calculated in the reverse transform. After convergence has been reached, more atomic positions and less extraneous peaks are observed in the refined electron density map. The starting model need not be very large to achieve success with PHASER; successful phase refinement has been achieved with a starting model that consists of only 5% of the total scattering power of the full molecule. The second part of the thesis discusses three crystal structure determinations.
[Biological Process Oriented Online Fourier Transform Infrared Spectrometer].
Xie, Fei; Wu, Qiong-shui; Zeng, Li-bo
2015-08-01
An online Fourier Transform Infrared Spectrometer and an ATR (Attenuated Total Reflection) probe, specifically at the application of real time measurement of the reaction substrate concentration in biological processes, were designed. (1) The spectrometer combined the theories of double cube-corner reflectors and flat mirror, which created a kind of high performance interferometer system. The light path folding way was utilized to makes the interferometer compact structure. Adopting double cube-corner reflectors, greatly reduces the influence of factors in the process of moving mirror movement such as rotation, tilt, etc. The parallelogram oscillation flexible support device was utilized to support the moving mirror moves. It cancelled the friction and vibration during mirror moving, and ensures the smooth operation. The ZnSe splitter significantly improved the hardware reliability in high moisture environment. The method of 60° entrance to light splitter improves the luminous flux. (2) An ATR in situ measuring probe with simple structure, large-flux, economical and practical character was designed in this article. The transmission of incident light and the light output utilized the infrared pipe with large diameter and innerplanted-high plating membrane, which conducted for the infrared transmission media of ATR probe. It greatly reduced the energy loss of infrared light after multiple reflection on the inner wall of the light pipe. Therefore, the ATR probe obtained high flux, improved the signal strength, which make the signal detected easily. Finally, the high sensitivity of MCT (Mercury Cadmium Telluride) detector was utilized to realize infrared interference signal collection, and improved the data quality of detection. The test results showed that the system yields the advantages of perfect moisture-proof performance, luminous flux, online measurement, etc. The designed online Fourier infrared spectrometer can real-time measured common reactant substrates
[Biological Process Oriented Online Fourier Transform Infrared Spectrometer].
Xie, Fei; Wu, Qiong-shui; Zeng, Li-bo
2015-08-01
An online Fourier Transform Infrared Spectrometer and an ATR (Attenuated Total Reflection) probe, specifically at the application of real time measurement of the reaction substrate concentration in biological processes, were designed. (1) The spectrometer combined the theories of double cube-corner reflectors and flat mirror, which created a kind of high performance interferometer system. The light path folding way was utilized to makes the interferometer compact structure. Adopting double cube-corner reflectors, greatly reduces the influence of factors in the process of moving mirror movement such as rotation, tilt, etc. The parallelogram oscillation flexible support device was utilized to support the moving mirror moves. It cancelled the friction and vibration during mirror moving, and ensures the smooth operation. The ZnSe splitter significantly improved the hardware reliability in high moisture environment. The method of 60° entrance to light splitter improves the luminous flux. (2) An ATR in situ measuring probe with simple structure, large-flux, economical and practical character was designed in this article. The transmission of incident light and the light output utilized the infrared pipe with large diameter and innerplanted-high plating membrane, which conducted for the infrared transmission media of ATR probe. It greatly reduced the energy loss of infrared light after multiple reflection on the inner wall of the light pipe. Therefore, the ATR probe obtained high flux, improved the signal strength, which make the signal detected easily. Finally, the high sensitivity of MCT (Mercury Cadmium Telluride) detector was utilized to realize infrared interference signal collection, and improved the data quality of detection. The test results showed that the system yields the advantages of perfect moisture-proof performance, luminous flux, online measurement, etc. The designed online Fourier infrared spectrometer can real-time measured common reactant substrates
[Using Fourier transform to calculate gas concentration in DOAS].
Liu, Qian-lin; Wang, Li-shi; Huang, Xin-jian; Wu, Yan-dan; Xiao, Ming-wei
2008-12-01
Being an analysis tool of high sensitivity, high resolution, multicomponents, real-time and fast monitoring, the differential optical absorption spectrometry (DOAS) is becoming a new method in atmosphere pollution monitoring. In the DOAS technique, many gases spectra have periodicity evidently, such as those from SO2, NO, NH3 and NO2. Aiming at three kinds of main air-polluted gases, i.e., SO2, NO and NO2 in atmosphere, the DOAS technique is used to monitor them, and Fourier transform is used to analyse the above-mentioned absorption spectra. Under the condition of Hanning Windows, Fourier transforma is used to process various gases spectra which have periodicity. In the process, the value of the characteristic frequency has a linearity relation to the gas concentration. So a new analysis method of DOAS is proposed, which is utilizing the relation between the value of the characteristic frequency and the gas concentration to deduce a linearity formula to calculate the gas concentration. So the value of the characteristic frequency can be used to get the gas concentration. For the gases with evident spectrum periodicity, such as SO2 and NO, this method is good. But for some gases with periodicity not evident, the error in the calculated concentration is beyond the allowable value. So in this method, the important process is frequency separation. It is also the main part in the future study. In a word, this method frees itself from the basic theory in the DOAS technique, cuts down on the process of the concentration calculation and the spectral analysis, and deserves further study. PMID:19248493
Institute of Scientific and Technical Information of China (English)
Wang Chuandan; Zhang Zhongpei; Li Shaoqian
2007-01-01
The method of FRactional Fourier Transform (FRFT) is introduced to Transform Domain Communication System (TDCS) for signal transforming in the paper after theoretical analysis. The method yields optimal Basis Function (BF) by FRFT with optimal transform angle. The TDCS using the proposed method has wider usable spectrum, stronger robustness and better ability of anti non-stationary jamming than using usual methods, such as Fourier Transform (FT), Auto Regressive (AR), Wavelet Transform (WT), etc. The main simulation results are as follows. First, the Bit Error Rate (BER) Pb is close to theoretical bound of no jamming no matter in single tone or in linear chirp interference. Second, the interference-to-signal ratio J/E is at least 12dB more than that of Direct Spread Spectrum System (DSSS) under the same BER if the spectrum hopping-to-signal ratio is 1:20 in chirp plus hopping interfering. Third, the Eb/No (when estimation difference is 90% between transmitter and receiver) is about 3.5dB or about 0.5dB (when estimation difference is 10% between transmitter and receiver) more than that of theoretical result when no estimation difference under Pb = 10-2.
Gas Analysis by Fourier Transform Mm-Wave Spectroscopy
Harris, Brent J.; Steber, Amanda L.; Lehmann, Kevin K.; Pate, Brooks H.
2013-06-01
Molecular rotational spectroscopy of low pressure, room temperature gases offers high chemical selectivity and sensitivity with the potential for a wide range of applications in gas analysis. A strength of the technique is the potential to identify molecules that have not been previously studied by rotational spectroscopy by comparing experimental results to predictions of the spectroscopic parameters from quantum chemistry -6 so called library-free detection. The development of Fourier transform mm-wave spectrometers using high peak power (30 mW) active multiplier chain mm-wave sources brings new measurement capabilities to the analysis of complex gas mixtures. Strategies for gas analysis based on high-throughput mm-wave spectroscopy and arbitrary waveform generator driven mm-wave sources are described. Several new measurement capabilities come from the intrinsic time-domain measurement technique. High-sensitivity double-resonance measurements can be performed to speed the analysis of a complex gas sample containing several species. This technique uses a "pi-pulse" to selectively invert the population of two selected rotational energy levels and the effect of this excitation pulse on all other transitions in the spectrometer operating range is monitored using segmented chirped-pulse Fourier transform spectroscopy. This method can lead to automated determination of the molecular rotational constants. Rapid pulse duration scan experiments can be used to estimate the magnitude and direction of the dipole moment of the molecule from an unknown spectrum. Coherent pulse echo experiments, using the traditional Hahn sequence or two-color population recovery methods, can be used to determine the collisional relaxation rate of the unknown molecule. This rate determination improves the ability to estimate the mass of the unknown molecule from the determination of the Doppler dephasing rate. By performing a suite of automated, high-throughput measurements, there is the
Hodges, Colin M.; Akhavan, Jacqueline
For routine identification of forensic samples many techniques are employed. These include ultraviolet spectrophotometry, combined gas chromatography—mass spectroscopy together with high performance liquid chromatography, infrared spectroscopy and X-ray powder diffraction. Conventional Raman spectroscopy is not routinely used by forensic laboratories for the identification of drugs and explosives because of high background scatter and time consuming sample alignment. One way of overcoming these problems is to use the newly developed technique of Fourier Transform Raman spectroscopy. Here negligible sample alignment is required, and there is reduced sample fluorescence. FTR spectra were recorded of pure and contaminated illicit drug samples, together with some explosive materials. Identification of an unknown explosive (Semtex) was also conducted. FTR provides a simple and satisfactory method of identifying certain drugs and explosives. The technique is non-destructive, utilizing small samples with no sample preparation being required.
Institute of Scientific and Technical Information of China (English)
张润生; 王跨陡; 龚飞君; 叶海英; 张玉荣; 严松茂; 杜一平; 张维冰
2012-01-01
采用气相色谱一傅立叶变换红外光谱联用技术,建立了9种苯丙胺类毒品及其衍生物的分析鉴别方法.采用HP-1(30 m×0.32 mm,0.25 μm)毛细管柱,MCT红外检测器与氢火焰检测器同时检测,在程序升温条件下,以十七烷为内标物,研究已知对照品的色谱保留行为及其气态红外光谱图特征,建立相应的特征吸收峰数据库,作为鉴别分析的依据.色谱保留时间与红外特征吸收峰联合鉴别法极大地提高了鉴别的准确性.将发展的方法应用于可疑毒品物证中苯丙胺类及其衍生物毒品样品的鉴别,获得了理想的结果.本方法可用于涉毒案件毒品物证的检验,特别适用于混合毒品成分的检验.%Gas chromatography-infrared spectroscopy technique has been used for the quantitative analysis of amphetamine-type stimulants. Capillary column of HP-1 (30 m×0. 32 mm. 0. 25 μm) was coupled with MCT infrared spectroscopy and flame ionization detecters using temperature programming for identification of nine ampletamine-type stimulants. Chromatographic retention behaviors were studied using heptadecane as internal standard for separation and identification of the stimulants. On another hand, infrared spectra of the gaseous amphetamine-type stimulants were measured using chemical reference substances of the stimulants and the characteristic peaks from the collected spectra were supplied as an assistant identification. The cooperation of information from retention time and infrared spectrum should improve the reliability of the method. The analysis of real samples showed satisfactory results. The method could be applied to the stimulant identification, especially to that of complicated sample.
Directory of Open Access Journals (Sweden)
D. Seidl
1999-06-01
Full Text Available Among a variety of spectrogram methods Short-Time Fourier Transform (STFT and Continuous Wavelet Transform (CWT were selected to analyse transients in non-stationary tremor signals. Depending on the properties of the tremor signal a more suitable representation of the signal is gained by CWT. Three selected broadband tremor signals from the volcanos Mt. Stromboli, Mt. Semeru and Mt. Pinatubo were analyzed using both methods. The CWT can also be used to extend the definition of coherency into a time-varying coherency spectrogram. An example is given using array data from the volcano Mt. Stromboli.
Single beam Fourier transform digital holographic quantitative phase microscopy
Energy Technology Data Exchange (ETDEWEB)
Anand, A., E-mail: arun-nair-in@yahoo.com; Chhaniwal, V. K.; Mahajan, S.; Trivedi, V. [Optics Laboratory, Applied Physics Department, Faculty of Technology and Engineering, M.S. University of Baroda, Vadodara 390001 (India); Faridian, A.; Pedrini, G.; Osten, W. [Institut für Technische Optik, Universität Stuttgart, Pfaffenwaldring 9, 70569 Stuttgart (Germany); Dubey, S. K. [Siemens Technology and Services Pvt. Ltd, Corporate Technology—Research and Technology Centre, Bangalore 560100 (India); Javidi, B. [Department of Electrical and Computer Engineering, U-4157, University of Connecticut, Storrs, Connecticut 06269-2157 (United States)
2014-03-10
Quantitative phase contrast microscopy reveals thickness or height information of a biological or technical micro-object under investigation. The information obtained from this process provides a means to study their dynamics. Digital holographic (DH) microscopy is one of the most used, state of the art single-shot quantitative techniques for three dimensional imaging of living cells. Conventional off axis DH microscopy directly provides phase contrast images of the objects. However, this process requires two separate beams and their ratio adjustment for high contrast interference fringes. Also the use of two separate beams may make the system more vulnerable to vibrations. Single beam techniques can overcome these hurdles while remaining compact as well. Here, we describe the development of a single beam DH microscope providing whole field imaging of micro-objects. A hologram of the magnified object projected on to a diffuser co-located with a pinhole is recorded with the use of a commercially available diode laser and an arrayed sensor. A Fourier transform of the recorded hologram directly yields the complex amplitude at the image plane. The method proposed was investigated using various phase objects. It was also used to image the dynamics of human red blood cells in which sub-micrometer level thickness variation were measurable.
Spectroscopic Stokes polarimetry based on Fourier transform spectrometer
Liu, Yeng-Cheng; Lo, Yu-Lung; Li, Chang-Ye; Liao, Chia-Chi
2015-02-01
Two methods are proposed for measuring the spectroscopic Stokes parameters using a Fourier transform spectrometer. In the first method, it is designed for single point measurement. The parameters are extracted using an optical setup comprising a white light source, a polarizer set to 0°, a quarter-wave plate and a scanning Michelson interferometer. In the proposed approach, the parameters are extracted from the intensity distributions of the interferograms produced with the quarter-wave plate rotated to 0°, 22.5°, 45° and -45°, respectively. For the second approach, the full-field and dynamic measurement can be designed based upon the first method with special angle design in a polarizer and a quarter-wave plate. Hence, the interferograms of two-dimensional detection also can be simultaneously extracted via a pixelated phase-retarder and polarizer array on a high-speed CCD camera and a parallel read-out circuit with a multi-channel analog to digital converter. Thus, a full-field and dynamic spectroscopic Stokes polarimetry without any rotating components could be developed. The validity of the proposed methods is demonstrated both numerically and experimentally. To the authors' knowledge, this could be the simplest optical arrangement in extracting the spectral Stokes parameters. Importantly, the latter one method avoids the need for rotating components within the optical system and therefore provides an experimentally straightforward means of extracting the dynamic spectral Stokes parameters.
Online Botnet Detection Based on Incremental Discrete Fourier Transform
Directory of Open Access Journals (Sweden)
Xiaocong Yu
2010-05-01
Full Text Available Botnet detection has attracted lots of attention since botnet attack is becoming one of the most serious threats on the Internet. But little work has considered the online detection. In this paper, we propose a novel approach that can monitor the botnet activities in an online way. We define the concept of “feature streams” to describe raw network traffic. If some feature streams show high similarities, the corresponding hosts will be regarded as suspected bots which will be added into the suspected bot hosts set. After activity analysis, bot hosts will be confirmed as soon as possible. We present a simple method by computing the average Euclidean distance for similarity measurement. To avoid huge calculation among feature streams, classical Discrete Fourier Transform (DFT technique is adopted. Then an incremental calculation of DFT coefficients is introduced to obtain the optimal execution time. The experimental evaluations show that our approach can detect both centralized and distributed botnet activities successfully with high efficiency and low false positive rate.
Analysis of ovarian tumor pathology by Fourier Transform Infrared Spectroscopy
Directory of Open Access Journals (Sweden)
Mehrotra Ranjana
2010-12-01
Full Text Available Abstract Background Ovarian cancer is the second most common cancer among women and the leading cause of death among gynecologic malignancies. In recent years, infrared (IR spectroscopy has gained attention as a simple and inexpensive method for the biomedical study of several diseases. In the present study infrared spectra of normal and malignant ovarian tissues were recorded in the 650 cm-1 to 4000 cm-1 region. Methods Post surgical tissue samples were taken from the normal and tumor sections of the tissue. Fourier Transform Infrared (FTIR data on twelve cases of ovarian cancer with different grades of malignancy from patients of different age groups were analyzed. Results Significant spectral differences between the normal and the ovarian cancerous tissues were observed. In particular changes in frequency and intensity in the spectral region of protein, nucleic acid and lipid vibrational modes were observed. It was evident that the sample-to-sample or patient-to-patient variations were small and the spectral differences between normal and diseased tissues were reproducible. Conclusion The measured spectroscopic features, which are the spectroscopic fingerprints of the tissues, provided the important differentiating information about the malignant and normal tissues. The findings of this study demonstrate the possible use of infrared spectroscopy in differentiating normal and malignant ovarian tissues.
Quantitative analysis of iron oxides using Fourier transform infrared spectrophotometry
International Nuclear Information System (INIS)
In this study, a systematic approach based on the application of Fourier transform infrared spectrophotometry (FTIR) was taken, in order to quantitatively analyze the corrosion products formed in the secondary cycle of pressurized water reactors (PWR). Binary mixtures of iron oxides were prepared with known compositions containing pure commercial magnetite (Fe3O4), maghemite (γ-Fe2O3), and hematite (α-Fe2O3) for calibration purposes. Calcium oxide (lime) was added to all samples as a standard reference in obtaining the calibration curves. Using regression analysis, relationships were developed for intensity versus concentration for absorption bands corresponding to each of the phases in their corresponding FTIR spectrum. Correlation coefficients, R2, of 0.82, 0.87, and 0.86 were obtained for maghemite-magnetite, hematite-magnetite, and hematite-maghemite systems, respectively. The calibration curves generated were used to quantify phases in multi-component unknown field samples that were obtained from different components (moisture separators, condensers, and high- and low- pressure heaters) of the two units (units 1 and 2) of the secondary cycle of the Comanche Peak PWR
High-resolution wide-band Fast Fourier Transform spectrometers
Klein, Bernd; Krämer, Ingo; Bell, Andreas; Meyer, Klaus; Güsten, Rolf
2012-01-01
We describe the performance of our latest generations of sensitive wide-band high-resolution digital Fast Fourier Transform Spectrometer (FFTS). Their design, optimized for a wide range of radio astronomical applications, is presented. Developed for operation with the GREAT far infrared heterodyne spectrometer on-board SOFIA, the eXtended bandwidth FFTS (XFFTS) offers a high instantaneous bandwidth of 2.5 GHz with 88.5 kHz spectral resolution and has been in routine operation during SOFIA's Basic Science since July 2011. We discuss the advanced field programmable gate array (FPGA) signal processing pipeline, with an optimized multi-tap polyphase filter bank algorithm that provides a nearly loss-less time-to-frequency data conversion with significantly reduced frequency scallop and fast sidelobe fall-off. Our digital spectrometers have been proven to be extremely reliable and robust, even under the harsh environmental conditions of an airborne observatory, with Allan-variance stability times of several 1000 se...
Realization of a scalable coherent quantum Fourier transform
Debnath, Shantanu; Linke, Norbert; Figgatt, Caroline; Landsman, Kevin; Wright, Ken; Monroe, Chris
2016-05-01
The exponential speed-up in some quantum algorithms is a direct result of parallel function-evaluation paths that interfere through a quantum Fourier transform (QFT). We report the implementation of a fully coherent QFT on five trapped Yb+ atomic qubits using sequences of fundamental quantum logic gates. These modular gates can be used to program arbitrary sequences nearly independent of system size and distance between qubits. We use this capability to first perform a Deutsch-Jozsa algorithm where several instances of three-qubit balanced and constant functions are implemented and then examined using single qubit QFTs. Secondly, we apply a fully coherent five-qubit QFT as a part of a quantum phase estimation protocol. Here, the QFT operates on a five-qubit superposition state with a particular phase modulation of its coefficients and directly produces the corresponding phase to five-bit precision. Finally, we examine the performance of the QFT in the period finding problem in the context of Shor's factorization algorithm. This work is supported by the ARO with funding from the IARPA MQCO program and the AFOSR MURI on Quantum Measurement and Verification.
Fourier transform ion cyclotron resonance mass spectrometry: a primer.
Marshall, A G; Hendrickson, C L; Jackson, G S
1998-01-01
This review offers an introduction to the principles and generic applications of FT-ICR mass spectrometry, directed to readers with no prior experience with the technique. We are able to explain the fundamental FT-ICR phenomena from a simplified theoretical treatment of ion behavior in idealized magnetic and electric fields. The effects of trapping voltage, trap size and shape, and other nonidealities are manifested mainly as perturbations that preserve the idealized ion behavior modified by appropriate numerical correction factors. Topics include: effect of ion mass, charge, magnetic field, and trapping voltage on ion cyclotron frequency; excitation and detection of ICR signals; mass calibration; mass resolving power and mass accuracy; upper mass limit(s); dynamic range; detection limit, strategies for mass and energy selection for MSn; ion axialization, cooling, and remeasurement; and means for guiding externally formed ions into the ion trap. The relation of FT-ICR MS to other types of Fourier transform spectroscopy and to the Paul (quadrupole) ion trap is described. The article concludes with selected applications, an appendix listing accurate fundamental constants needed for ultrahigh-precision analysis, and an annotated list of selected reviews and primary source publications that describe in further detail various FT-ICR MS techniques and applications.
Optical Fourier transform scatterometry for LER and LWR metrology
Boher, P.; Petit, J.; Leroux, T.; Foucher, J.; Desieres, Y.; Hazart, J.; Chaton, P.
2005-05-01
We present an innovating instrument based on optical Fourier transform (OFT) capable to measure simultaneously the specular and non specular diffraction pattern of sub-micronic periodic structures. The sample is illuminated at fixed wavelength (green laser) versus a large angular aperture both in incidence (0 to 80°) and azimuth (0 to 180°). In the present paper we focus on the possibility to measure line edge roughness (LER) and line width roughness (LWR) using this new technique. To understand the problem, different gratings with artificial periodic LER and LWR roughness have been fabricated and characterized precisely by atomic force microscopy (AFM). Different light scattering measurements have been performed using the OFT instrument with different illuminations in order to understand precisely the optical behavior of these systems. We show that we can distinguish LER and LWR by measuring simultaneously the diffracted contributions coming from the grating and from the periodic roughness. In phase LER with small LWR does not give first order diffraction contribution for the periodic roughness. In contrast, LER in opposite phase with large LWR gives a strong contribution for the first order of diffraction of the periodic roughness. In any case, the sensitivity to LER and LWR is better than 5nm for 500nm period gratings measured at 532nm. This result can be extended to samples with real LER and LWR. It shows without ambiguity that simultaneous measurement of the specular and diffracted light diffraction patterns is necessary to extract separately the two parameters.
SAR-based vibrometry using the fractional Fourier transform
Campbell, Justin B.; Wang, Qi; Ade-Bello, Jelili; Caudana, Humberto; Trujillo, Nicole B.; Bhatta, Ishwor; Dunkel, Ralf; Atwood, Thomas; Doerry, Armin; Gerstle, Walter H.; Santhanam, Balu; Hayat, Majeed M.
2015-05-01
A fundamental assumption when applying Synthetic Aperture Radar (SAR) to a ground scene is that all targets are motionless. If a target is not stationary, but instead vibrating in the scene, it will introduce a non-stationary phase modulation, termed the micro-Doppler effect, into the returned SAR signals. Previously, the authors proposed a pseudosubspace method, a modification to the Discrete Fractional Fourier Transform (DFRFT), which demonstrated success for estimating the instantaneous accelerations of vibrating objects. However, this method may not yield reliable results when clutter in the SAR image is strong. Simulations and experimental results have shown that the DFRFT method can yield reliable results when the signal-to-clutter ratio (SCR) > 8 dB. Here, we provide the capability to determine a target's frequency and amplitude in a low SCR environment by presenting two methods that can perform vibration estimations when SCR < 3 dB. The first method is a variation and continuation of the subspace approach proposed previously in conjunction with the DFRFT. In the second method, we employ the dual-beam SAR collection architecture combined with the extended Kalman filter (EKF) to extract information from the returned SAR signals about the vibrating target. We also show the potential for extending this SAR-based capability to remotely detect and classify objects housed inside buildings or other cover based on knowing the location of vibrations as well as the vibration histories of the vibrating structures that house the vibrating objects.
Fourier Transform Infrared Spectroscopy and Photoacoustic Spectroscopy for Saliva Analysis.
Mikkonen, Jopi J W; Raittila, Jussi; Rieppo, Lassi; Lappalainen, Reijo; Kullaa, Arja M; Myllymaa, Sami
2016-09-01
Saliva provides a valuable tool for assessing oral and systemic diseases, but concentrations of salivary components are very small, calling the need for precise analysis methods. In this work, Fourier transform infrared (FT-IR) spectroscopy using transmission and photoacoustic (PA) modes were compared for quantitative analysis of saliva. The performance of these techniques was compared with a calibration series. The linearity of spectrum output was verified by using albumin-thiocyanate (SCN(-)) solution at different SCN(-) concentrations. Saliva samples used as a comparison were obtained from healthy subjects. Saliva droplets of 15 µL were applied on the silicon sample substrate, 6 drops for each specimen, and dried at 37 ℃ overnight. The measurements were carried out using an FT-IR spectrometer in conjunction with an accessory unit for PA measurements. The findings with both transmission and PA modes mirror each other. The major bands presented were 1500-1750 cm(-1) for proteins and 1050-1200 cm(-1) for carbohydrates. In addition, the distinct spectral band at 2050 cm(-1) derives from SCN(-) anions, which is converted by salivary peroxidases to hypothiocyanate (OSCN(-)). The correlation between the spectroscopic data with SCN(-) concentration (r > 0.990 for transmission and r = 0.967 for PA mode) was found to be significant (P < 0.01), thus promising to be utilized in future applications.
Continued Development of a Planetary Imaging Fourier Transform Spectrometer (PIFTS)
Sromovsky, L. A.
2002-01-01
This report describes continued efforts to evaluate a breadboard of a Planetary Imaging Fourier Transform Spectrometer (PIFTS). The PIFTS breadboard was developed under prior PIDDP funding. That effort is described in the final report for NASA Grant NAG5-6248 and in two conference papers (Sromovsky et al. 2000; Revercomb et al. 2000). The PIFTS breadboard was designed for near-IR (1-5.2 micrometer imaging of planetary targets with spectral resolving powers of several hundred to several thousand, using an InSb detector array providing at least 64x64 pixels imaging detail. The major focus of the development effort was to combine existing technologies to produce a small and low power design compatible with a very low mass flyable instrument. The objective of this grant (NAG5-10729) was further characterization of the breadboard performance, including intercomparisons with the highly accurate non-imaging Advanced Emitted Radiance Interferometer (AERI) (Revercomb et al. 1994; Best et al. 1997).
Geosynchronous Imaging Fourier Transform Spectrometer (GIFTS): Imaging and Tracking Capability
Zhou, D. K.; Larar, A. M.; Liu, Xu; Reisse, R. A.; Smith, W. L.; Revercomb, H. E.; Bingham, G. E.; Zollinger, L. J.; Tansock, J. J.; Huppi, Ronald J.
2007-01-01
The geosynchronous-imaging Fourier transform spectrometer (GIFTS) engineering demonstration unit (EDU) is an imaging infrared spectrometer designed for atmospheric soundings. It measures the infrared spectrum in two spectral bands (14.6 to 8.8 microns, 6.0 to 4.4 microns) using two 128 128 detector arrays with a spectral resolution of 0.57/cm with a scan duration of approx. 11 seconds. From a geosynchronous orbit, the instrument will have the capability of taking successive measurements of such data to scan desired regions of the globe, from which atmospheric status, cloud parameters, wind field profiles, and other derived products can be retrieved. The GIFTS EDU provides a flexible and accurate testbed for the new challenges of the emerging hyperspectral era. The EDU ground-based measurement experiment, held in Logan, Utah during September 2006, demonstrated its extensive capabilities and potential for geosynchronous and other applications (e.g., Earth observing environmental measurements). This paper addresses the experiment objectives and overall performance of the sensor system with a focus on the GIFTS EDU imaging capability and proof of the GIFTS measurement concept.
Calibration of the Herschel SPIRE Fourier Transform Spectrometer
Swinyard, B M; Hopwood, R; Valtchanov, I; Lu, N; Fulton, T; Benielli, D; Imhof, P; Marchili, N; Baluteau, J -P; Bendo, G J; Ferlet, M; Griffin, M J; Lim, T L; Makiwa, G; Naylor, D A; Orton, G S; Papageorgiou, A; Pearson, C P; Schulz, B; Sidher, S D; Spencer, L D; van der Wiel, M H D; Wu, R
2014-01-01
The Herschel SPIRE instrument consists of an imaging photometric camera and an imaging Fourier Transform Spectrometer (FTS), both operating over a frequency range of 450-1550 GHz. In this paper, we briefly review the FTS design, operation, and data reduction, and describe in detail the approach taken to relative calibration (removal of instrument signatures) and absolute calibration against standard astronomical sources. The calibration scheme assumes a spatially extended source and uses the Herschel telescope as primary calibrator. Conversion from extended to point-source calibration is carried out using observations of the planet Uranus. The model of the telescope emission is shown to be accurate to within 6% and repeatable to better than 0.06% and, by comparison with models of Mars and Neptune, the Uranus model is shown to be accurate to within 3%. Multiple observations of a number of point-like sources show that the repeatability of the calibration is better than 1%, if the effects of the satellite absolu...
Fourier transform infrared spectroscopy (FTIR) of laser-irradiated cementum
Rechmann, Peter; White, Joel M.; Cecchini, Silvia C. M.; Hennig, Thomas
2003-06-01
Utilizing Fourier Transform Infrared Spectroscopy (FTIR) in specular reflectance mode chemical changes of root cement surfaces due to laser radiation were investigated. A total of 18 samples of root cement were analyzed, six served as controls. In this study laser energies were set to those known for removal of calculus or for disinfection of periodontal pockets. Major changes in organic as well as inorganic components of the cementum were observed following Nd:YAG laser irradiation (wavelength 1064 nm, pulse duration 250 μs, free running, pulse repetition rate 20 Hz, fiber diameter 320 μm, contact mode; Iskra Twinlight, Fontona, Slovenia). Er:YAG laser irradiation (wavelength 2.94 μm, pulse duration 250 μs, free running, pulse repetition rate 6 Hz, focus diameter 620 μm, air water cooling 30 ml/min; Iskra Twinlight, Fontona, Slovenia) significantly reduced the Amid bands due to changes in the organic components. After irradiation with a frequency doubled Alexandrite laser (wavelength 377 nm, pulse duration 200 ns, q-switched, pulse repetition rate 20 Hz, beam diameter 800 μm, contact mode, water cooling 30 ml/min; laboratory prototype) only minimal reductions in the peak intensity of the Amide-II band were detected.
Multi-channel sampling theorems for band-limited signals with fractional Fourier transform
Institute of Scientific and Technical Information of China (English)
2008-01-01
Multi-channel sampling for band-limited signals is fundamental in the theory of multi-channel parallel A/D environment and multiplexing wireless communication environment. As the fractional Fourier transform has been found wide applications in signal processing fields, it is necessary to consider the multi-channel sampling theorem based on the fractional Fourier transform. In this paper, the multi-channel sampling theorem for the fractional band-limited signal is firstly proposed, which is the generalization of the well-known sampling theorem for the fractional Fourier transform. Since the periodic nonuniformly sampled signal in the fractional Fourier domain has valuable applications, the reconstruction expression for the periodic nonuniformly sampled signal has been then obtained by using the derived multi-channel sampling theorem and the specific space-shifting and phase-shifting properties of the fractional Fourier transform. Moreover, by designing different fractional Fourier filters, we can obtain reconstruction methods for other sampling strategies.
Summation of the Fourier Transform of Measures and Four Denominator Estimates
International Nuclear Information System (INIS)
In this paper we consider convergence exponent for the Fourier transform of surface-carried measures. We apply the obtained bound for the Fourier transform of measures to so-called four denominator estimate related to the Schroedinger operator on a lattice. (author)
An Introduction to Fast Fourier Transforms through the Study of Oscillating Reactions.
Eastman, M. P.; And Others
1986-01-01
Discusses an experiment designed to introduce students to the basic principles of the fast Fourier transform and Fourier smoothing through transformation of time-dependent optical absorption data from an oscillating reaction. Uses the Belousov-Zhabotinskii reaction. Describes the experimental setup and data analysis techniques.
Precise and fast spatial-frequency analysis using the iterative local Fourier transform.
Lee, Sukmock; Choi, Heejoo; Kim, Dae Wook
2016-09-19
The use of the discrete Fourier transform has decreased since the introduction of the fast Fourier transform (fFT), which is a numerically efficient computing process. This paper presents the iterative local Fourier transform (ilFT), a set of new processing algorithms that iteratively apply the discrete Fourier transform within a local and optimal frequency domain. The new technique achieves 210 times higher frequency resolution than the fFT within a comparable computation time. The method's superb computing efficiency, high resolution, spectrum zoom-in capability, and overall performance are evaluated and compared to other advanced high-resolution Fourier transform techniques, such as the fFT combined with several fitting methods. The effectiveness of the ilFT is demonstrated through the data analysis of a set of Talbot self-images (1280 × 1024 pixels) obtained with an experimental setup using grating in a diverging beam produced by a coherent point source.
Fourier Transform Spectrometer measurements of Atmospheric Carbon Dioxide and Methane
Kivi, Rigel; Heikkinen, Pauli; Chen, Huilin; Hatakka, Juha; Laurila, Tuomas
2016-04-01
Ground based remote sensing measurements of column CO2 and CH4 using Fourier Transform Spectrometers (FTS) within the Total Carbon Column Observing Network (TCCON) are known for high precision and accuracy. These measurements are performed at various locations globally and they have been widely used in carbon cycle studies and validation of space born measurements. The relevant satellite missions include the Orbiting Carbon Observatory-2 (OCO-2) by the National Aeronautics and Space Administration (NASA); the SCanning Imaging Absorption SpectroMeter for Atmospheric CHartographY (SCIAMACHY) by the European Space Agency (ESA); the Greenhouse gases Observing SATellite (GOSAT) by the Japan Aerospace Exploration Agency (JAXA) and the upcoming Sentinel-5 Precursor mission, which is an ESA mission and scheduled for launch in 2016. Results of the column CO2 and CH4 measurements at Sodankylä in northern Finland (at 67.4° N, 26.6° E) are reported in this study. The measurements have been performed on regular basis since the beginning of the program in early 2009. We also present evaluation of the data quality of the ground based measurements and comparisons with the available satellite based retrievals. In case of comparisons between the GOSAT and ground based retrievals of CO2 and CH4 no significant biases were found. Sodankylä is one of the northernmost stations in the TCCON network. However, the data coverage has been relatively good thanks to the progress towards automation of the FTS measurement system. At Sodankylä the retrievals have been also compared with the balloon borne AirCore measurements at the site. AirCore sampling system is directly related to the World Meteorological Organization in situ trace gas measurement scales. The balloon platform allows sampling in both stratosphere and troposphere, which is a benefit, compared to the aircraft in situ measurements.
Geostationary Imaging Fourier Transform Spectrometer (GIFTS): science applications
Smith, W. L.; Revercomb, H. E.; Zhou, D. K.; Bingham, G. E.; Feltz, W. F.; Huang, H. L.; Knuteson, R. O.; Larar, A. M.; Liu, X.; Reisse, R.; Tobin, D. C.
2006-12-01
A revolutionary satellite weather forecasting instrument, called the "GIFTS" which stands for the "Geostationary Imaging Fourier Transform Spectrometer", was recently completed and successfully tested in a space chamber at the Utah State University's Space Dynamics Laboratory. The GIFTS was originally proposed by the NASA Langley Research Center, the University of Wisconsin, and the Utah State University and selected for flight demonstration as NASA's New Millennium Program (NMP) Earth Observing-3 (EO-3) mission, which was unfortunately cancelled in 2004. GIFTS is like a digital 3-d movie camera that, when mounted on a geostationary satellite, would provide from space a revolutionary four-dimensional view of the Earth's atmosphere. GIFTS will measure the distribution, change, and movement of atmospheric moisture, temperature, and certain pollutant gases, such as carbon monoxide and ozone. The observation of the convergence of invisible water vapor, and the change of atmospheric temperature, provides meteorologists with the observations needed to predict where, and when, severe thunderstorms, and possibly tornados, would occur, before they are visible on radar or in satellite cloud imagery. The ability of GIFTS to observe the motion of moisture and clouds at different altitudes enables atmospheric winds to be observed over vast, and otherwise data sparse, oceanic regions of the globe. These wind observations would provide the means to greatly improve the forecast of where tropical storms and hurricanes will move and where and when they will come ashore (i.e., their landfall position and time). GIFTS, if flown into geostationary orbit, would provide about 80,000 vertical profiles per minute, each one like a low vertical resolution (1-2km) weather balloon sounding, but with a spacing of 4 km. GIFTS is a revolutionary atmospheric sensing tool. A glimpse of the science measurement capabilities of GIFTS is provided through airborne measurements with the NPOESS Airborne
Continuously tunable optical multidimensional Fourier-transform spectrometer.
Dey, P; Paul, J; Bylsma, J; Deminico, S; Karaiskaj, D
2013-02-01
A multidimensional optical nonlinear spectrometer (MONSTR) is a robust, ultrastable platform consisting of nested and folded Michelson interferometers that can be actively phase stabilized. The MONSTR provides output pulses for nonlinear excitation of materials and phase-stabilized reference pulses for heterodyne detection of the induced signal. This platform generates a square of identical laser pulses that can be adjusted to have arbitrary time delays between them while maintaining phase stability. This arrangement is ideal for performing coherent optical experiments, such as multidimensional Fourier-transform spectroscopy. The present work reports on overcoming some important limitations on the original design of the MONSTR apparatus. One important advantage of the MONSTR is the fact that it is a closed platform, which provides the high stability. Once the optical alignment is performed, it is desirable to maintain the alignment over long periods of time. The previous design of the MONSTR was limited to a narrow spectral range defined by the optical coating of the beam splitters. In order to achieve tunability over a broad spectral range the internal optics needed to be changed. By using broadband coated and wedged beam splitters and compensator plates, combined with modifications of the beam paths, continuous tunability can be achieved from 520 nm to 1100 nm without changing any optics or performing alignment of the internal components of the MONSTR. Furthermore, in order to achieve continuous tunability in the spectral region between 520 nm and 720 nm, crucially important for studies on numerous biological molecules, a single longitudinal mode laser at 488.5 nm was identified and used as a metrology laser. The shorter wavelength of the metrology laser as compared to the usual HeNe laser has also increased the phase stability of the system. Finally, in order to perform experiments in the reflection geometry, a simple method to achieve active phase stabilization
Bivariate raising and lowering differential operators for eigenfunctions of a 2D Fourier transform
International Nuclear Information System (INIS)
We define a two-dimensional (2D) Fourier transform that self-reproduces a one-parameter family of bivariate Hermite functions; these are eigenfunctions of a Hamiltonian differential operator of second order, whose exponential is that transform. We find explicit forms of the bivariate raising and lowering partial differential operators of first degree for the eigenfunctions of this 2D Fourier transform. (paper)
A Synthetic Quadrature Phase Detector/Demodulator for Fourier Transform Transform Spectrometers
Campbell, Joel
2008-01-01
A method is developed to demodulate (velocity correct) Fourier transform spectrometer (FTS) data that is taken with an analog to digital converter that digitizes equally spaced in time. This method makes it possible to use simple low cost, high resolution audio digitizers to record high quality data without the need for an event timer or quadrature laser hardware, and makes it possible to use a metrology laser of any wavelength. The reduced parts count and simplicity implementation makes it an attractive alternative in space based applications when compared to previous methods such as the Brault algorithm.
Thyroid lesions diagnosis by Fourier transformed infrared absorption spectroscopy (FTIR)
International Nuclear Information System (INIS)
Thyroid nodules are a common disorder, with 4-7% of incidence in the Brazilian population. Although the fine needle aspiration (FNA) is an accurate method for thyroid tumors diagnosis, the discrimination between benign and malignant neoplasm is currently not possible in some cases with high incidence of false negative diagnosis, leading to a surgical intervention due to the risk of carcinomas. The aim of this study was to verify if the Fourier Transform infrared spectroscopy (FTIR) can contribute to the diagnosis of thyroid carcinomas and goiters, using samples of tissue and aspirates. Samples of FNA, homogenates and tissues of thyroid nodules with histopathological diagnosis were obtained and prepared for FTIR spectroscopy analysis. The FNA and homogenates samples were measured by μ-FTIR (between 950 . 1750 cm-1), at a nominal resolution of 4 cm-1 and 120 scans). Tissue samples were analyzed directly by ATR-FTIR technique, at a resolution 2 cm-1, with 60 scans in the same region. All spectra were corrected by the baseline and normalized by amides area (1550-1640 cm-1) in order to minimize variations of sample homogeneity. Then, spectra were converted into second derivatives using the Savitzk-Golay algorithm with a 13 points window. The Ward's minimum variance algorithm and Euclidean distances among the points were used for cluster analysis. Some FNA samples showed complex spectral pattern. All samples showed some cell pellets and large amount of hormone, represented by the bands of 1545 and 1655 cm-1. Bands in 1409, 1412, 1414, 1578 and 1579 cm-1 were also found, indicating possible presence of sugar, DNA, citric acid or metabolic products. In this study, it was obtained an excellent separation between goiter and malign lesion for the samples of tissues, with 100% of specificity in specific cluster and 67% sensibility and 50 of specificity. In homogenate and FNA samples this sensibility and specificity were lower, because among these samples, it were included
Rapid Bacterial Identification Using Fourier Transform Infrared Spectroscopy
Energy Technology Data Exchange (ETDEWEB)
Valentine, Nancy B.; Johnson, Timothy J.; Su, Yin-Fong; Forrester, Joel B.
2007-02-01
Recent studies at Pacific Northwest National Laboratory (PNNL) using infrared spectroscopy combined with statistical analysis have shown the ability to identify and discriminate vegetative bacteria, bacterial spores and background interferents from one another. Since the anthrax releases in 2001, rapid identification of unknown powders has become a necessity. Bacterial endospores are formed by some Bacillus species as a result of the vegetative bacteria undergoing environmental stress, e.g. a lack of nutrients. Endospores are formed as a survival mechanism and are extremely resistant to heat, cold, sunlight and some chemicals. They become airborne easily and are thus readily dispersed which was demonstrated in the Hart building. Fourier Transform Infrared (FTIR) spectroscopy is one of several rapid analytical methods used for bacterial endospore identification. The most common means of bacterial identification is culturing, but this is a time-consuming process, taking hours to days. It is difficult to rapidly identify potentially harmful bacterial agents in a highly reproducible way. Various analytical methods, including FTIR, Raman, photoacoustic FTIR and Matrix Assisted Laser Desorption/Ionization (MALDI) have been used to identify vegetative bacteria and bacterial endospores. Each has shown certain areas of promise, but each has shortcomings in terms of sensitivity, measurement time or portability. IR spectroscopy has been successfully used to distinguish between the sporulated and vegetative state. [1,2] It has also shown its utility at distinguishing between the spores of different species. [2-4] There are several Bacillus species that occur commonly in nature, so it is important to be able to distinguish between the many different species versus those that present an imminent health threat. The spectra of the different sporulated species are all quite similar, though there are some subtle yet reproducible spectroscopic differences. Thus, a more robust and
On the restriction of the Fourier transform to polynomial curves
Dendrinos, S.
2007-01-01
We prove a Fourier restriction theorem on curves parametrised by the mapping P(t) = (P1(t),..., Pn(t)), where each of the P1,..., Pn is a real-valued polynomial and t belongs to an interval on which each of the P1,..., Pn "resembles" a monomial.
The Boundedness of Maximal Operators and Singular Integrals via Fourier Transform Estimates
Institute of Scientific and Technical Information of China (English)
Hong Hai LIU
2012-01-01
In this paper,the author studies the mapping properties for some general maximal operators and singular integrals on certain function spaces via Fourier transform estimates.Also,some concrete maximal operators and singular integrals are studied as applications.
D'Astous, Y.; Blanchard, M.
1982-05-01
In the past years, the Journal has published a number of articles1-5 devoted to the introduction of Fourier transform spectroscopy in the undergraduate labs. In most papers, the proposed experimental setup consists of a Michelson interferometer, a light source, a light detector, and a chart recorder. The student uses this setup to record an interferogram which is then Fourier transformed to obtain the spectrogram of the light source. Although attempts have been made to ease the task of performing the required Fourier transform,6 the use of computers and Cooley-Tukey's fast Fourier transform (FFT) algorithm7 is by far the simplest method to use. However, to be able to use FFT, one has to get a number of samples of the interferogram, a tedious job which should be kept to a minimum. (AIP)
The Fourier Transform in Chemistry. Part 1. Nuclear Magnetic Resonance: Introduction.
King, Roy W.; Williams, Kathryn R.
1989-01-01
Using fourier transformation methods in nuclear magnetic resonance has made possible increased sensitivity in chemical analysis. This article describes these methods as they relate to magnetization, the RF magnetic field, nuclear relaxation, the RF pulse, and free induction decay. (CW)
Cageao, R.; Sander, S.; Blavier, J.; Jiang, Y.; Nemtchinov, V.
2000-01-01
A compact, high resolution Fourier-transform spectrometer for atmospheric near ultraviolet spectroscopy has been installed at the Jet Propulsion Laboratory's Table Mountain Facility (34.4N, 117.7 W, elevation 2290m).
Implementation of Period-Finding Algorithm by Means of Simulating Quantum Fourier Transform
Directory of Open Access Journals (Sweden)
Zohreh Moghareh Abed
2010-01-01
Full Text Available In this paper, we introduce quantum fourier transform as a key ingredient for many useful algorithms. These algorithms make a solution for problems which is considered to be intractable problems on a classical computer. Quantum Fourier transform is propounded as a key for quantum phase estimation algorithm. In this paper our aim is the implementation of period-finding algorithm.Quantum computer solves this problem, exponentially faster than classical one. Quantum phase estimation algorithm is the key for the period-finding problem .Therefore, by means of simulating quantum Fourier transform, we are able to implement the period-finding algorithm. In this paper, the simulation of quantum Fourier transform is carried out by Matlab software.
Improved method of generating bit reversed numbers for calculating fast fourier transform
Digital Repository Service at National Institute of Oceanography (India)
Suresh, T.
Fast Fourier Transform (FFT) is an important tool required for signal processing in defence applications. This paper reports an improved method for generating bit reversed numbers needed in calculating FFT using radix-2. The refined algorithm takes...
Building a symbolic computer algebra toolbox to compute 2D Fourier transforms in polar coordinates.
Dovlo, Edem; Baddour, Natalie
2015-01-01
The development of a symbolic computer algebra toolbox for the computation of two dimensional (2D) Fourier transforms in polar coordinates is presented. Multidimensional Fourier transforms are widely used in image processing, tomographic reconstructions and in fact any application that requires a multidimensional convolution. By examining a function in the frequency domain, additional information and insights may be obtained. The advantages of our method include: •The implementation of the 2D Fourier transform in polar coordinates within the toolbox via the combination of two significantly simpler transforms.•The modular approach along with the idea of lookup tables implemented help avoid the issue of indeterminate results which may occur when attempting to directly evaluate the transform.•The concept also helps prevent unnecessary computation of already known transforms thereby saving memory and processing time.
Pulse shaping using the optical Fourier transform technique - for ultra-high-speed signal processing
DEFF Research Database (Denmark)
Palushani, Evarist; Oxenløwe, Leif Katsuo; Galili, Michael;
2009-01-01
This paper reports on the generation of a 1.6 ps FWHM flat-top pulse using the optical Fourier transform technique. The pulse is validated in a 320 Gbit/s demultiplexing experiment.......This paper reports on the generation of a 1.6 ps FWHM flat-top pulse using the optical Fourier transform technique. The pulse is validated in a 320 Gbit/s demultiplexing experiment....
Time-Domain Optical Fourier Transformation for OTDM-DWDM and DWDM-OTDM Conversion
DEFF Research Database (Denmark)
Mulvad, Hans Christian Hansen; Palushani, Evarist; Galili, Michael;
2011-01-01
Applications of time-domain optical Fourier transformation (OFT) in ultra-high-speed optical time-division multiplexed systems (OTDM) are reviewed, with emphasis on the recent demonstrations of OFT-based conversion between the OTDM and DWDM formats.......Applications of time-domain optical Fourier transformation (OFT) in ultra-high-speed optical time-division multiplexed systems (OTDM) are reviewed, with emphasis on the recent demonstrations of OFT-based conversion between the OTDM and DWDM formats....
Institute of Scientific and Technical Information of China (English)
CHEN Lin-Fei; ZHAO Dao-Mu
2006-01-01
@@ We propose a new method to add different images together by optical implementation that is realized by the multi-exposure based on fractional Fourier transform hologram. Partial image fusion is proposed and realized by this method. Multiple images encryption can also be implemented by the multi-exposure of the hologram based on fractional Fourier transform. Computer simulations prove that this method is valid.
The Design of Lens Imaging System by Means of Fractional Fourier Transform
Institute of Scientific and Technical Information of China (English)
CHEN Jiannong; XU Qiang; D.R.Selviah
2002-01-01
The relation between the 2nd fractional Fourier transform and the imaging process of an optical system is discussed. By changing the coordinate scales of the input plane in respect to the magnification of the optical imaging system, the fractional Fourier transform can be a powerful tool in designing specific imaging system. The Gaussian imaging formula of single lens is obtained by using the tool. Finally the procedures are generalized for designing a double-lens imaging system through an example.
Accelerating the Non-equispaced Fast Fourier Transform on Commodity Graphics Hardware
DEFF Research Database (Denmark)
Sørensen, Thomas Sangild; Schaeffter, Tobias; Noe, Karsten Østergaard;
2008-01-01
We present a fast parallel algorithm to compute the Non-equispaced fast Fourier transform on commodity graphics hardware (the GPU). We focus particularly on a novel implementation of the convolution step in the transform, which was previously its most time consuming part. We describe the performa......We present a fast parallel algorithm to compute the Non-equispaced fast Fourier transform on commodity graphics hardware (the GPU). We focus particularly on a novel implementation of the convolution step in the transform, which was previously its most time consuming part. We describe...
Energy Technology Data Exchange (ETDEWEB)
Zheng, Y. [Pennsylvania State Univ., University Park, PA (United States)]|[Lawrence Berkeley Lab., CA (United States); Shirley, D.A. [Pennsylvania State Univ., University Park, PA (United States)
1995-02-01
The authors show by Fourier analyses of experimental data, with no further treatment, that the positions of all the strong peaks in Fourier transforms of angle-resolved photoemission extended fine structure (ARPEFS) from adsorbed surfaces can be explicitly predicted from a trial structure with an accuracy of about {+-} 0.3 {angstrom} based on a single-scattering cluster model together with the concept of a strong backscattering cone, and without any additional analysis. This characteristic of ARPEFS Fourier transforms can be developed as a simple method for determining the structures of adsorbed surfaces to an accuracy of about {+-} 0.1 {angstrom}.
DEFF Research Database (Denmark)
Nielsen, Michael L; Savitski, Mikhail M; Kjeldsen, Frank;
2004-01-01
Sequence verification and mapping of posttranslational modifications require nearly 100% sequence coverage in the "bottom-up" protein analysis. Even in favorable cases, routine liquid chromatography-mass spectrometry detects from protein digests peptides covering 50-90% of the sequence. Here we...... investigated the reasons for limited peptide detection, considering various physicochemical aspects of peptide behavior in liquid chromatography-Fourier transform mass spectrometry (LC-FTMS). No overall correlation was found between the detection probability and peptide mass. In agreement with literature data......, the signal increased with peptide hydrophobicity. Surprisingly, the pI values exhibited an opposite trend, with more acidic tryptic peptides detected with higher probability. A mixture of synthesized peptides of similar masses confirmed the hydrophobicity dependence but showed strong positive correlation...
Fourier-space inversion of the star transform
Zhao, Fan; Markel, Vadim A
2014-01-01
We define the star transform as a generalization of the broken ray transform introduced by us previously. The advantages of using the star transform include the possibility to reconstruct the absorption and the scattering coefficients of the medium separately and simultaneously (from the same data) and the possibility to utilize scattered radiation which, in the case of the conventional X-ray tomography, is discarded. In this paper, we derive the star transform from physical principles, discuss its mathematical properties and analyze numerical stability of inversion. In particular, it is shown that stable inversion of the star transform can be obtained only for configurations involving odd number of rays. Several computationally-efficient inversion algorithms are derived and tested numerically.
Luo, David; Kudenov, Michael W
2016-05-16
Systematic phase errors in Fourier transform spectroscopy can severely degrade the calculated spectra. Compensation of these errors is typically accomplished using post-processing techniques, such as Fourier deconvolution, linear unmixing, or iterative solvers. This results in increased computational complexity when reconstructing and calibrating many parallel interference patterns. In this paper, we describe a new method of calibrating a Fourier transform spectrometer based on the use of artificial neural networks (ANNs). In this way, it is demonstrated that a simpler and more straightforward reconstruction process can be achieved at the cost of additional calibration equipment. To this end, we provide a theoretical model for general systematic phase errors in a polarization birefringent interferometer. This is followed by a discussion of our experimental setup and a demonstration of our technique, as applied to data with and without phase error. The technique's utility is then supported by comparison to alternative reconstruction techniques using fast Fourier transforms (FFTs) and linear unmixing.
Factoring and Fourier Transformation with a Mach-Zehnder Interferometer
Summhammer, J
1997-01-01
The scheme of Clauser and Dowling (Phys. Rev. A 53, 4587 (1996)) for factoring N by means of an N-slit interference experiment is translated into an experiment with a single Mach-Zehnder interferometer. With dispersive phase shifters the ratio of the coherence length to wavelength limits the numbers that can be factored. A conservative estimate permits $N \\approx 10^7$. It is furthermore shown, that sine and cosine Fourier coefficients of a real periodic function can be obtained with such an interferometer.
Magneto-sensor circuit efficiency incremented by Fourier-transformation
Talukdar, Abdul Hafiz Ibne
2011-10-01
In this paper detection by recognized intelligent algorithm for different magnetic films with the aid of a cost-effective and simple high efficient circuit are realized. Well-known, magnetic films generate oscillating frequencies when they stay a part of an LC- oscillatory circuit. These frequencies can be further analyzed to gather information about their magnetic properties. For the first time in this work we apply the signal analysis in frequency domain to create the Fourier frequency spectra which was used to detect the sample properties and their recognition. In this paper we have summarized both the simulation and experimental results. © 2011 Elsevier Ltd. All rights reserved.
Determination of amino acids in Chinese rice wine by fourier transform near-infrared spectroscopy.
Shen, Fei; Niu, Xiaoying; Yang, Danting; Ying, Yibin; Li, Bobin; Zhu, Geqing; Wu, Jian
2010-09-01
Chinese rice wine is abundant in amino acids. The possibility of quantitative detection of 16 free amino acids (aspartic acid, threonine, serine, glutamic acid, proline, glycine, alanine, valine, methionine, isoleucine, leucine, tyrosine, phenylalanine, lysine, histidine, and arginine) in Chinese rice wine by Fourier transform near-infrared (NIR) spectroscopy was investigated for the first time in this study. A total of 98 samples from vintage 2007 rice wines with different aging times were analyzed by NIR spectroscopy in transmission mode. Calibration models were developed using partial least-squares regression (PLSR) with high-performance liquid chromatography (HPLC) by postcolumn derivatization and diode array detection as a reference method. To validate the calibration models, full cross (leave-one-out) validation was employed. The results showed that the calibration statistics were good (rcal>0.94) for all amino acids except proline, histidine, and arginine. The correlation coefficient in cross validation (rcv) was >0.81 for 12 amino acids. The residual predictive deviation (RPD) value obtained was >1.5 in all amino acids except proline and arginine, and it was >2.0 in 6 amino acids. The results obtained in this study indicated that NIR spectroscopy could be used as an easy, rapid, and novel tool to quantitatively predict free amino acids in Chinese rice wine without sophisticated methods. PMID:20707307
Digital watermarking for still image based on discrete fractional fourier transform
Institute of Scientific and Technical Information of China (English)
无
2001-01-01
Presents a digital watermarking technique based on discrete fractional Fourier transform(DFRFT), discusses the transformation of the original image by DFRFT, and the modification of DFRFT coefficients of the original image by the information of watermark, and concludes from experimental results that the proposed tech nique is robust to lossy compression attack.
Fast Inverse Nonlinear Fourier Transforms for Fiber Bragg Grating Design and Related Problems
Wahls, Sander
2016-01-01
The problem of constructing a fiber Bragg grating profile numerically such that the reflection coefficient of the grating matches a given specification is considered. The well-known analytic solution to this problem is given by a suitable inverse nonlinear Fourier transform (also known as inverse scattering transform) of the specificed reflection coefficient. Many different algorithms have been proposed to compute this inverse nonlinear Fourier transform numerically. The most efficient ones require $\\mathcal{O}(D^{2})$ floating point operations (flops) to generate $D$ samples of the grating profile. In this paper, two new fast inverse nonlinear Fourier transform algorithms that require only $\\mathcal{O}(D\\log^{2}D)$ flops are proposed. The merits of our algorithms are demonstrated in numerical examples, in which they are compared to a conventional layer peeling method, the Toeplitz inner bordering method and integral layer peeling. One of our two algorithms also extends to the design problem for fiber-assiste...
Rejection of Linear FM Interference in DSSS System Based on Fractional Fourier Transform
Institute of Scientific and Technical Information of China (English)
QI Lin; TAO Ran; ZHOU Si-yong
2005-01-01
A new method for the rejection of linear frequency modulation (LFM) interference in direct sequence spread spectrum (DSSS) system based on the fractional Fourier transform is proposed, and the configuration of the receiver with an interference exciser is also presented. Based on the property that the fractional Fourier transform of a signal is equivalent to rotating the signal in the time-frequency plane, the received signal is transform into a certain fractional Fourier domain, this transform will result in the least spectrum overlap between the signal and interference. Then, a narrowband filter is exploited to extract most of the interference energy. The performance analyses show that remarkable improvements in signal-to-noise ratio (SNR) and biterror-ratio (BER) are obtained.
Safari, M J; Afarideh, H; Jamili, S; Bayat, E
2016-01-01
A Discrete Fourier Transform Method (DFTM) for discrimination between the signal of neutrons and gamma rays in organic scintillation detectors is presented. The method is based on the transformation of signals into the frequency domain using the sine and cosine Fourier transforms in combination with the discrete Fourier transform. The method is largely benefited from considerable differences that usually is available between the zero-frequency components of sine and cosine and the norm of the amplitude of the DFT for neutrons and gamma-ray signals. Moreover, working in frequency domain naturally results in considerable suppression of the unwanted effects of various noise sources that is expected to be effective in time domain methods. The proposed method could also be assumed as a generalized nonlinear weighting method that could result in a new class of pulse shape discrimination methods, beyond definition of the DFT. A comparison to the traditional Charge Integration Method (CIM), as well as the Frequency G...
Institute of Scientific and Technical Information of China (English)
无
2006-01-01
A new method of resolving overlapped peak, Fourier self-deconvolution (FSD) using approximation CN obtained from frequency domain wavelet transform of F(ω) yielded by Fourier transform of overlapped peak signals f(t) as the linear function, was presented in this paper.Compared with classical FSD, the new method exhibits excellent resolution for different overlapped peak signals such as HPLC signals, and have some characteristics such as an extensive applicability for any overlapped peak shape signals and a simple operation because of no selection procedure of the linear function. Its excellent resolution for those different overlapped peak signals is mainly because F(ω) obtained from Fourier transform of f(t) and CN obtained from wavelet transform of F(ω) have the similar linearity and peak width. The effect of those fake peaks can be eliminated by the algorithm proposed by authors. This method has good potential in the process of different overlapped peak signals.
Does the Entorhinal Cortex use the Fourier Transform?
Directory of Open Access Journals (Sweden)
Jeff eOrchard
2013-12-01
Full Text Available Some neurons in the entorhinal cortex (EC fire bursts when the animal occupies locations organized in a hexagonal grid pattern in their spatial environment. Place cells have also been observed, firing bursts only when the animal occupies a particular region of the environment. Both of these types of cells exhibit theta-cycle modulation, firing bursts in the 4-12Hz range. Grid cells fire bursts of action potentials that precess with respect to the theta cycle, a phenomenon dubbed "theta precession". Various models have been proposed to explain these phenomena, and how they relate to navigation. Among the most promising are the oscillator interference models. The bank-of-oscillators model proposed by Welday et al. (2011 exhibits all these features. However, their simulations are based on theoretical oscillators, and not implemented entirely with spiking neurons. We extend their work in a number of ways. First, we place the oscillators in a frequency domain and reformulate the model in terms of Fourier theory. Second, this perspective suggests a division of labour for implementing spatial maps: position, versus map layout. The animal's position is encoded in the phases of the oscillators, while the spatial map shape is encoded implicitly in the weights of the connections between the oscillators and the read-out nodes. Third, it reveals that the oscillator phases all need to conform to a linear relationship across the frequency domain. Fourth, we implement a partial model of the EC using spiking leaky integrate-and-fire (LIF neurons. Fifth, we devise new coupling mechanisms, enlightened by the global phase constraint, and show they are capable of keeping spiking neural oscillators in consistent formation. Our model demonstrates place cells, grid cells, and phase precession. The Fourier model also gives direction for future investigations, such as integrating sensory feedback to combat drift, or explaining why grid cells exist at all.
Fourier and Hadamard transform spectrometers - A limited comparison. II
Harwit, M.; Tai, M. H.
1977-01-01
A mathematical approach was used to compare interferometric spectrometers and Hadamard transform spectrometers. The principle results are reported, noting that the simple Hadamard spectrometer encodes more efficiently than a Michelson interferometer which, in turn, encodes less efficiently than is usually acknowledged. Hirschfeld's (1977) major objections to these findings are discussed, although it is noted that none of his objections is supported by evidence.
Method of local pointed function reduction of original shape in Fourier transformation
Dosch, H
2002-01-01
The method for analytical reduction of the original shape in the one-dimensional Fourier transformation by the fourier image modulus is proposed. The basic concept of the method consists in the presentation of the model shape in the form of the local peak functions sum. The eigenfunctions, generated by the linear differential equations with the polynomial coefficients, are selected as the latter ones. This provides for the possibility of managing the Fourier transformation without numerical integration. This reduces the reverse task to the nonlinear regression with a small number of the evaluated parameters and to the numerical or asymptotic study on the model peak functions - the eigenfunctions of the differential tasks and their fourier images
Locke, Jonathan; White, Paul R
2011-10-01
The analysis of cetacean vocalizations is considered using Fourier-based techniques that employ chirp functions in their decomposition. In particular, the paper considers a short-time methods based on the fractional Fourier transform for detecting frequency modulated narrow-band signals, such as dolphin whistles, and compares this to the classical short-time Fourier methods. The fractional Fourier technique explored computes transforms associated with a range of chirp rates and automatically selects the rate for the final analysis. This avoids the need for prior knowledge of signal's chirp rate. An analysis is presented that details the performance of both methods as signal detectors and allows one to determine their detection thresholds. These thresholds are then used to measure the detectability of synthetic signals. This principle is then extended to measure performance on a set of recordings of narrow-band vocalizations from a range of cetacean species.
Fourier-transform Ghost Imaging for pure phase object based on Compressive Sampling algorithm
Wang, Hui; Han, Shensheng
2009-01-01
A special algorithm for the Fourier-transform Ghost Imaging (GI) scheme is discussed based on the Compressive Sampling (CS) theory. Though developed mostly in real space, CS algorithm could also be used for the Fourier spectrum reconstruction of pure phase object by setting a proper sensing matrix. This could find its application in diffraction imaging of X-ray, neutron and electron with higher efficiency and resolution. Simulation and experiment results are also presented to prove the feasib...
Fast Fourier and Wavelet Transforms for Wavefront Reconstruction in Adaptive Optics
Energy Technology Data Exchange (ETDEWEB)
Dowla, F U; Brase, J M; Olivier, S S
2000-07-28
Wavefront reconstruction techniques using the least-squares estimators are computationally quite expensive. We compare wavelet and Fourier transforms techniques in addressing the computation issues of wavefront reconstruction in adaptive optics. It is shown that because the Fourier approach is not simply a numerical approximation technique unlike the wavelet method, the Fourier approach might have advantages in terms of numerical accuracy. However, strictly from a numerical computations viewpoint, the wavelet approximation method might have advantage in terms of speed. To optimize the wavelet method, a statistical study might be necessary to use the best basis functions or ''approximation tree.''
Adaptive motion mapping in pancreatic SBRT patients using Fourier transforms
Jones, Bernard L; Miften, Moyed
2015-01-01
Recent studies suggest that 4DCT is unable to accurately measure respiratory-induced pancreatic tumor motion. In this work, we assessed the daily motion of pancreatic tumors treated with SBRT, and developed adaptive strategies to predict and account for this motion. The daily motion trajectory of pancreatic tumors during CBCT acquisition was calculated using a model which reconstructs the instantaneous 3D position in each 2D CBCT projection image. We developed a metric (termed "Spectral Coherence," SC) based on the Fourier frequency spectrum of motion in the SI direction, and analyzed the ability of SC to predict motion-based errors and classify patients according to motion characteristics. The amplitude of daily motion exceeded the predictions of pre-treatment 4DCT imaging by an average of 3.0 mm, 2.3 mm, and 3.5 mm in the AP, LR, and SI directions. SC was correlated with daily motion differences and tumor dose coverage. In a simulated adaptive protocol, target margins were adjusted based on SC, resulting in...
Grid-Based Fourier Transform Phase Contrast Imaging
Tahir, Sajjad
Low contrast in x-ray attenuation imaging between different materials of low electron density is a limitation of traditional x-ray radiography. Phase contrast imaging offers the potential to improve the contrast between such materials, but due to the requirements on the spatial coherence of the x-ray beam, practical implementation of such systems with tabletop (i.e. non-synchrotron) sources has been limited. One recently developed phase imaging technique employs multiple fine-pitched gratings. However, the strict manufacturing tolerances and precise alignment requirements have limited the widespread adoption of grating-based techniques. In this work, we have investigated a technique recently demonstrated by Bennett et al. that utilizes a single grid of much coarser pitch. Our system consisted of a low power 100 microm spot Mo source, a CCD with 22 microm pixel pitch, and either a focused mammography linear grid or a stainless steel woven mesh. Phase is extracted from a single image by windowing and comparing data localized about harmonics of the grid in the Fourier domain. A Matlab code was written to perform the image processing. For the first time, the effects on the diffraction phase contrast and scattering amplitude images of varying grid types and periods, and of varying the window function type used to separate the harmonics, and the window widths, were investigated. Using the wire mesh, derivatives of the phase along two orthogonal directions were obtained and new methods investigated to form improved phase contrast images.
bahram noshad; morteza razaz; seyed ghodratollah seifossadat
2012-01-01
One of mal-operations of the transformer differential protection during the unload transformer energizing with additional line/load from the supplying side is ultra-saturation phenomenon. In this paper, first a new model according to Discrete Fourier Transform (DFT) algorithm for investigating the ultra-saturation phenomenon during the unload transformer energizing with additional line/load from the supplying side is presented and its effect on the differential protection of the transformer i...
Prime Factor Cyclotomic Fourier Transforms with Reduced Complexity over Finite Fields
Wu, Xuebin; Chen, Ning; Wagh, Meghanad
2010-01-01
Discrete Fourier transforms~(DFTs) over finite fields have widespread applications in error correction coding. Hence, reducing the computational complexities of DFTs is of great significance, especially for long DFTs as increasingly longer error control codes are chosen for digital communication and storage systems. Since DFTs involve both multiplications and additions over finite fields and multiplications are much more complex than additions, recently proposed cyclotomic fast Fourier transforms (CFFTs) are promising due to their low multiplicative complexity. Unfortunately, they have very high additive complexity. Techniques such as common subexpression elimination (CSE) can be used to reduce the additive complexity of CFFTs, but their effectiveness for long DFTs is limited by their complexity. In this paper, we propose prime factor cyclotomic Fourier transforms (PFCFTs), which use CFFTs as sub-DFTs via the prime factor algorithm. When the length of DFTs is prime, our PFCFTs reduce to CFFTs. When the length...
Image encryption based on the reality-preserving multiple-parameter fractional Fourier transform
Lang, Jun
2012-05-01
In recent years, a number of methods have been proposed in the literature for the encryption of two-dimensional information by using the fractional Fourier transform, but most of their encryptions are complex values and need digital hologram technique to record information, which is inconvenient for digital transmission. In this paper, we propose a new approach for image encryption based on the real-valuedness and decorrelation property of the reality-preserving multiple-parameter fractional Fourier transform in order to meet the requirements of the secure image transmission. In the proposed scheme, the original and encrypted images are respectively in the spatial domain and the reality-preserving multiple-parameter fractional Fourier transformed domain determined by the encryption keys. Numerical simulations are performed to demonstrate that the proposed method is reliable and more robust to blind decryption than several existing methods.
Fourier-transform spectroscopy: new methods and applications: introduction by the feature editors.
Traub, W A; Winkel, R J; Goldman, A
1996-06-01
We are pleased to introduce this special issue of papers on Fourier-transform spectroscopy, which grew out of a recent topical meeting sponsored by the Optical Society of America. The topical meeting welcomed all researchers who practice the art of Fourier-transform spectroscopy in the laboratory, in the atmosphere, and in space. The power and the wide applicability of Fourier-transform spectroscopy unite these fields with a common mathematical and instrumental bond. The meeting probed each of these areas in depth, bringing out new ideas for instrumentation, analysis, and applications. There was a strong sentiment at the meeting that the quality of papers and posters was exceptionally high and that it would be important for future progress in the field to have the results of this meeting captured in print. This special issue is the fruit of that effort.
Inversion of Fourier Transforms by Means of Scale-Frequency Series
Directory of Open Access Journals (Sweden)
Nassar H. S. Haidar
2014-01-01
Full Text Available We report on inversion of the Fourier transform when the frequency variable can be scaled in a variety of different ways that improve the resolution of certain parts of the frequency domain. The corresponding inverse Fourier transform is shown to exist in the form of two dual scale-frequency series. Upon discretization of the continuous scale factor, this Fourier transform series inverse becomes a certain nonharmonic double series, a discretized scale-frequency (DSF series. The DSF series is also demonstrated, theoretically and practically, to be rate-optimizable with respect to its two free parameters, when it satisfies, as an entropy maximizer, a pertaining recursive nonlinear programming problem incorporating the entropy-based uncertainty principle.
Goodman, Roe W
2016-01-01
This textbook for undergraduate mathematics, science, and engineering students introduces the theory and applications of discrete Fourier and wavelet transforms using elementary linear algebra, without assuming prior knowledge of signal processing or advanced analysis.It explains how to use the Fourier matrix to extract frequency information from a digital signal and how to use circulant matrices to emphasize selected frequency ranges. It introduces discrete wavelet transforms for digital signals through the lifting method and illustrates through examples and computer explorations how these transforms are used in signal and image processing. Then the general theory of discrete wavelet transforms is developed via the matrix algebra of two-channel filter banks. Finally, wavelet transforms for analog signals are constructed based on filter bank results already presented, and the mathematical framework of multiresolution analysis is examined.
q-Extension of Mehta's eigenvectors of the finite Fourier transform for q, a root of unity
Energy Technology Data Exchange (ETDEWEB)
Atakishiyeva, Mesuma K [Facultad de Ciencias, Universidad Autonoma del Estado de Morelos, CP 62250 Cuernavaca, Morelos (Mexico); Atakishiyev, Natig M [Instituto de Matematicas, Unidad Cuernavaca Universidad Nacional Autonoma de Mexico, AP 273-3 Admon 3, Cuernavaca, Morelos 62251 (Mexico); Koornwinder, Tom H [Korteweg-de Vries Institute, University of Amsterdam, PO Box 94248, 1090 GE Amsterdam (Netherlands)], E-mail: mesuma@servm.fc.uaem.mx, E-mail: natig@matcuer.unam.mx, E-mail: T.H.Koornwinder@uva.nl
2009-10-30
It is shown that the continuous q-Hermite polynomials for q, a root of unity, have simple transformation properties with respect to the classical Fourier transform. This result is then used to construct q-extended eigenvectors of the finite Fourier transform in terms of these polynomials.
DEFF Research Database (Denmark)
Guan, Pengyu; Røge, Kasper Meldgaard; Kjøller, Niels-Kristian;
2015-01-01
We propose a novel all-optical WDM regeneration scheme for DPSK signals based on optical Fourier transformation and phase sensitive amplification. Phase regeneration of a WDM signal consisting of 4x10-Gbit/s phase noise degraded DPSK channels is demonstrated for the first time.......We propose a novel all-optical WDM regeneration scheme for DPSK signals based on optical Fourier transformation and phase sensitive amplification. Phase regeneration of a WDM signal consisting of 4x10-Gbit/s phase noise degraded DPSK channels is demonstrated for the first time....
OTDM-WDM Conversion Based on Time-Domain Optical Fourier Transformation with Spectral Compression
DEFF Research Database (Denmark)
Mulvad, Hans Christian Hansen; Palushani, Evarist; Galili, Michael;
2011-01-01
We propose a scheme enabling direct serial-to-parallel conversion of OTDM data tributaries onto a WDM grid, based on optical Fourier transformation with spectral compression. Demonstrations on 320 Gbit/s and 640 Gbit/s OTDM data are shown.......We propose a scheme enabling direct serial-to-parallel conversion of OTDM data tributaries onto a WDM grid, based on optical Fourier transformation with spectral compression. Demonstrations on 320 Gbit/s and 640 Gbit/s OTDM data are shown....
DWDM-TO-OTDM Conversion by Time-Domain Optical Fourier Transformation
DEFF Research Database (Denmark)
Mulvad, Hans Christian Hansen; Hu, Hao; Galili, Michael;
2011-01-01
We propose DWDM-OTDM conversion by time-domain optical Fourier transformation. Error-free conversion of a 16×10 Gbit/s 50 GHz-spacing DWDM data signal to a 160 Gbit/s OTDM signal with a 2.1 dB average penalty is demonstrated.......We propose DWDM-OTDM conversion by time-domain optical Fourier transformation. Error-free conversion of a 16×10 Gbit/s 50 GHz-spacing DWDM data signal to a 160 Gbit/s OTDM signal with a 2.1 dB average penalty is demonstrated....
The Pegg-Barnett phase operator and the discrete Fourier transform
Perez-Leija, Armando; Andrade-Morales, Luis A.; Soto-Eguibar, Francisco; Szameit, Alexander; Moya-Cessa, Héctor M.
2016-04-01
In quantum mechanics the position and momentum operators are related to each other via the Fourier transform. In the same way, here we show that the so-called Pegg-Barnett phase operator can be obtained by the application of the discrete Fourier transform to the number operators defined in a finite-dimensional Hilbert space. Furthermore, we show that the structure of the London-Susskind-Glogower phase operator, whose natural logarithm gives rise to the Pegg-Barnett phase operator, is contained in the Hamiltonian of circular waveguide arrays. Our results may find applications in the development of new finite-dimensional photonic systems with interesting phase-dependent properties.
Estimation of structural modal parameters by fourier transform with an optimal window
Institute of Scientific and Technical Information of China (English)
ZHU Hong-ping; WAN Xin-hua
2005-01-01
An adaptive Fourier Transform (FT) with an optimal window has been proposed for the time-frequency analysis of nonstationary time series. The method allows for a good estimation of both frequency and amplitude of the spectrum and can be easily applied to the general case of time-varying signals. The evaluation of the proposed approach has been performed on measured time-varying signals from a suspension bridge model and a steel frame model whose data have the typical non-stationary characteristics. The numerical results show that the proposed approach can overcome some of the difficulties encountered in the classic Fourier transform technique and can achieve higher computation accuracy.
DEFF Research Database (Denmark)
Hu, Hao; Kong, Deming; Palushani, Evarist;
2013-01-01
We demonstrate transmission of a 1.28-Tbaud Nyquist-OTDM signal over a record distance of 100 km with detection by time-domain optical Fourier transformation followed by FEC decoding, resulting in error-free performance for all tributaries.......We demonstrate transmission of a 1.28-Tbaud Nyquist-OTDM signal over a record distance of 100 km with detection by time-domain optical Fourier transformation followed by FEC decoding, resulting in error-free performance for all tributaries....
Interferometric time delay correction for Fourier transform spectroscopy in the extreme ultraviolet
Meng, Yijian; Zhang, Chunmei; Marceau, Claude; Naumov, A. Yu.; Corkum, P. B.; Villeneuve, D. M.
2016-09-01
We demonstrate a Fourier transform spectrometer in the extreme ultraviolet (XUV) spectrum using a high-harmonic source, with wavelengths as short as 32 nm. The femtosecond infrared laser source is divided into two separate foci in the same gas jet to create two synchronized XUV sources. An interferometric method to determine the relative delay between the two sources is shown to improve the accuracy of the delay time, with corrections of up to 200 asec required. By correcting the time base before the Fourier transform, the frequency resolution is improved by up to an order of magnitude.
Fast 3D shape measurement using Fourier transform profilometry without phase unwrapping
Song, Kechen; Hu, Shaopeng; Wen, Xin; Yan, Yunhui
2016-09-01
This paper presents a novel, simple, yet fast 3D shape measurement method using Fourier transform profilometry. Different from the conventional Fourier transform profilometry, this proposed method introduces the binocular stereo vision and employs two image pairs (i.e., original image pairs and fringe image pairs) to restructure 3D shape. In this proposed method, instead of phase unwrapping algorithm, a coarse disparity map is adopted as a constraint condition to realize phase matching using wrapped phase. Since the local phase matching and sub-pixel disparity refinement are proposed to obtain high measuring accuracy, high-quality phase is not required. The validity of the proposed method is verified by experiments.
Goda, Keisuke; Jalali, Bahram
2008-01-01
Dispersive Fourier transformation is a powerful technique in which the spectrum of an optical pulse is mapped into a time-domain waveform using chromatic dispersion. It replaces a diffraction grating and detector array with a dispersive fiber and single photodetector. This simplifies the system and, more importantly, enables fast real-time measurements. Here we describe a novel ultrafast barcode reader and displacement sensor that employs internally-amplified dispersive Fourier transformation. This technique amplifies and simultaneously maps the spectrally encoded barcode into a temporal waveform. It achieves a record acquisition speed of 25 MHz -- four orders of magnitude faster than the current state-of-the-art.
Goda, Keisuke; Tsia, Kevin K.; Jalali, Bahram
2008-09-01
Dispersive Fourier transformation is a powerful technique in which the spectrum of an optical pulse is mapped into a time-domain waveform using chromatic dispersion. It replaces a diffraction grating and detector array with a dispersive fiber and single photodetector. This simplifies the system and, more importantly, enables fast real-time measurements. Here we describe a novel ultrafast barcode reader and displacement sensor that employs internally amplified dispersive Fourier transformation. This technique amplifies and simultaneously maps the spectrally encoded barcode into a temporal waveform. It achieves a record acquisition speed of 25MHz—four orders of magnitude faster than the current state of the art.
Security of image encryption scheme based on multi-parameter fractional Fourier transform
Zhao, Tieyu; Ran, Qiwen; Yuan, Lin; Chi, Yingying; Ma, Jing
2016-10-01
Recently, multi-parameter fractional Fourier transform (MPFRFT) has been widely applied in the optics cryptosystem, which has attracted more and more researchers' attention. However, in further study we find a serious security problem on the MPFRFT which is the multi-choice of decryption key corresponding to an encryption key. The existence of multi-decryption-key hinders the application of this algorithm. We present a new generalized fractional Fourier transform, which can overcome the problem and enlarge the key space. The simulation results show that the proposed algorithm has higher security and key sensitivity.
Change of Scale Formulas for Wiener Integrals Related to Fourier-Feynman Transform and Convolution
Directory of Open Access Journals (Sweden)
Bong Jin Kim
2014-01-01
Full Text Available Cameron and Storvick discovered change of scale formulas for Wiener integrals of functionals in Banach algebra S on classical Wiener space. Yoo and Skoug extended these results for functionals in the Fresnel class F(B and in a generalized Fresnel class FA1,A2 on abstract Wiener space. We express Fourier-Feynman transform and convolution product of functionals in S as limits of Wiener integrals. Moreover we obtain change of scale formulas for Wiener integrals related to Fourier-Feynman transform and convolution product of these functionals.
Abnormality Detection in ECG Signal Using Wavelets and Fourier Transform
Directory of Open Access Journals (Sweden)
Sayali Kambire
2016-06-01
Full Text Available Electrocardiogram (ECG is used to record the electrical activity in the heart. It is the most important physiological parameter that gives the correct assessment regarding the functioning of the heart. The paper proposes a method based on signal processing correlation technique to find out whether the ECG is normal or abnormal. Many of the abnormal ECGs are called Arrhythmias. Aim of this study to analyze the ECG signal using MATLAB and to find whether the signal is normal or abnormal. According to different arrhythmia it helps to analyze the electrocardiogram (ECG signal, extract the features, for the classification of heart beats. ECG feature extraction system has been developed and evaluated based on the multi-resolution wavelet transform.
Fourier Transform Near Infrared Spectrometry: Using Interferograms To Determine Chemical Composition
Hoy, R. M.; McClure, W. Fred
1989-12-01
Previous research conducted in this laboratory has demonstrated several advantages accrued by transforming near infrared spectra from the wavelength domain to the Fourier domain. Those advantages include: [1] smoothing wavelength domain data without loss of end points, [2] correcting for particle size phenomena encountered in solid sample analyses by simply omitting the mean term Fourier coefficient from the "retransformation process", [3] minimizing the multicollinearity problem prevalent in wavelength space, [4] generating wavelength-space derivatives from Fourier space without loss of end points, [5] performing band enhancements via Fourier self-deconvolution, [6] identifying sample type using Fourier vectors, [7] estimating chemical composition using only the first few Fourier coefficients, [8] cutting of computer storage requirements by more than 96%, [9] cutting of calibration time by more than 96%, hence [10] reducing the drudgery of maintaining calibrations. That the first 12 Fourier coefficients contain sufficient information to determine chemical constituents in many products has turned out to be a major advantage leading us to understand that the chemical absorption information in the wavelength spectrum of a sample obtained with an interferometer was also present in the interferogram.
Novel Polynomial Basis with Fast Fourier Transform and Its Application to Reed-Solomon Erasure Codes
Lin, Sian-Jheng
2016-09-13
In this paper, we present a fast Fourier transform (FFT) algorithm over extension binary fields, where the polynomial is represented in a non-standard basis. The proposed Fourier-like transform requires O(h lg(h)) field operations, where h is the number of evaluation points. Based on the proposed Fourier-like algorithm, we then develop the encoding/ decoding algorithms for (n = 2m; k) Reed-Solomon erasure codes. The proposed encoding/erasure decoding algorithm requires O(n lg(n)), in both additive and multiplicative complexities. As the complexity leading factor is small, the proposed algorithms are advantageous in practical applications. Finally, the approaches to convert the basis between the monomial basis and the new basis are proposed.
Multi-dimensional option pricing using radial basis functions and the generalized Fourier transform
Larsson, Elisabeth; Ahlander, Krister; Hall, Andreas
2008-12-01
We show that the generalized Fourier transform can be used for reducing the computational cost and memory requirements of radial basis function methods for multi-dimensional option pricing. We derive a general algorithm, including a transformation of the Black-Scholes equation into the heat equation, that can be used in any number of dimensions. Numerical experiments in two and three dimensions show that the gain is substantial even for small problem sizes. Furthermore, the gain increases with the number of dimensions.
The quantum state vector in phase space and Gabor's windowed Fourier transform
Bracken, A. J.; Watson, P.
2010-01-01
Representations of quantum state vectors by complex phase space amplitudes, complementing the description of the density operator by the Wigner function, have been defined by applying the Weyl-Wigner transform to dyadic operators, linear in the state vector and anti-linear in a fixed `window state vector'. Here aspects of this construction are explored, with emphasis on the connection with Gabor's `windowed Fourier transform'. The amplitudes that arise for simple quantum states from various c...
Fourier transforms on Cantor sets: A study in non-Diophantine arithmetic and calculus
Aerts, Diederik; Czachor, Marek; Kuna, Maciej
2016-10-01
Fractals equipped with intrinsic arithmetic lead to a natural definition of differentiation, integration and complex numbers. Applying the formalism to the problem of a Fourier transform on fractals we show that the resulting transform has all the expected basic properties. As an example we discuss a sawtooth signal on the ternary middle-third Cantor set. The formalism works also for fractals that are not self-similar.
Directory of Open Access Journals (Sweden)
Percival Almoro
1998-12-01
Full Text Available Microscopic deformations on the surface of a circular diaphragm were measured using double exposure holographic interferometry and Fourier transform method (FTM. The three-dimensional surface deformations were successfully visualized by applying FTM to holographic interferogram analysis. The minimum surface displacement measured was 0.317 µm. This was calibrated via the Michelson interferometry technique.
Soil science research is increasingly applying Fourier transform infrared (FTIR) spectroscopy for analysis of soil organic matter (SOM). However, the compositional complexity of soils and the dominance of the mineral component can limit spectroscopic resolution of SOM and other minor components. The...
Lidar, Daniel A.
2003-01-01
A Comment on the paper "Quantum waveguide array generator for performing Fourier transforms: Alternate route to quantum computing" by R. Akis and D.K. Ferry, Appl. Phys. Lett. 79, 2823 (2001). The authors reply in Appl. Phys. Lett. 80, 2420 (2002).
Subpixel translation of MEMS measured by discrete fourier transform analysis of CCD images
Yamahata, C.; Sarajlic, E.; Stranczl, M.; Krijnen, G.J.M.; Gijs, M.A.M.
2011-01-01
We present a straightforward method for measuring in-plane linear displacements of microelectromechanical systems (MEMS) with subnanometer resolution. The technique is based on Fourier transform analysis of a video recorded with a Charge-Coupled Device (CCD) camera attached to an optical microscope
Hermann, Peter; Hoehl, Arne; Patoka, Piotr; Huth, Florian; Rühl, Eckart; Ulm, Gerhard
2013-02-11
We demonstrate scanning near-field optical microscopy with a spatial resolution below 100 nm by using low intensity broadband synchrotron radiation in the IR regime. The use of such a broadband radiation source opens up the possibility to perform nano-Fourier-transform infrared spectroscopy over a wide spectral range. PMID:23481749
Amir, W.; Planchon, T. A.; Durfee, C. G.; Squier, J. A.
2007-04-01
Spatiotemporal pulse shaping is characterized with two-dimensional Fourier transform spectral interferometry. A deformable-mirror-based bidimensional pulse shaper is used to create simple spatiotemporal structures on a femtosecond pulse, structures that are directly calculated from the measured spatiospectral phases and intensities.
Directory of Open Access Journals (Sweden)
M. Aslam Chaudhry
2004-01-01
Full Text Available We present a Fourier transform representation of the gamma functions, which leads naturally to a distributional representation for them. Both of these representations lead to new identities for the integrals of gamma functions multiplied by other functions, which are also presented here.
Wiedermann, Guenter; Jennings, D. E.; Hanel, R. H.; Kunde, V. G.; Moseley, S. H.
1989-01-01
A postdispersion system for astronomical observations with Fourier transform spectrometers in the thermal infrared has been developed which improves the sensitivity of radiation noise limited observations by reducing the spectral range incident on the detector. Special attention is given to the first-generation blocked impurity band detector. Planetary, solar, and stellar observations are reported.
DEFF Research Database (Denmark)
Clausen, Sønnik; Morgenstjerne, Axel; Rathmann, Ole
1996-01-01
Surface temperatures are estimated with high precision based on a multitemperature method for Fourier-transform spectrometers. The method is based on Planck's radiation law and a nonlinear least-squares fitting algorithm applied to two or more spectra at different sample temperatures and a single...
Barbu, I.M.
2008-01-01
This thesis describes, the use of a Fourier Transform Ion Cyclotron (FTICR) mass spectrometer in the study of biological samples with, imaging mass spectrometry (MS). To achieve this goal experiments were performed on an in-house modified FTICR-MS instrument (for which special acquisition software w
VUV Fourier-transform absorption study of the Lyman and Werner bands in D-2
Lange, de A.; Dickenson, G.D.; Salumbides, E.J.; Ubachs, W.M.G.; Oliveira, de N.; Joyeux, D.; Nahon, L.
2012-01-01
An extensive survey of the D-2 absorption spectrum has been performed with the high-resolution VUV Fourier-transform spectrometer employing synchrotron radiation. The frequency range of 90 000-119 000 cm (1) covers the full depth of the potential wells of the B (1)Sigma(+)(u), B' 1 Sigma(+)(u), and
Dool, T.C. van den; Hamelinck, R.F.M.M.; Kruizinga, B.; Gielesen, W.L.M.; Braam, B.C.; Nijenhuis, J.R.; Loix, N.; Luyckx, S.; Loon, D. van; Kooijman, P.P.; Swinyard, B.M.
2010-01-01
TNO, together with its partners Micromega and SRON, have designed a cryogenic scanning mechanism for use in the SAFARI Fourier Transform Spectrometer (FTS) on board of the SPICA mission. The optics of the FTS scanning mechanism (FTSM) consists of two back-to-back cat's-eyes. The optics are mounted o
Alpha-rooting method of color image enhancement by discrete quaternion Fourier transform
Grigoryan, Artyom M.; Agaian, Sos S.
2014-02-01
This paper presents a novel method for color image enhancement based on the discrete quaternion Fourier transform. We choose the quaternion Fourier transform, because it well-suited for color image processing applications, it processes all 3 color components (R,G,B) simultaneously, it capture the inherent correlation between the components, it does not generate color artifacts or blending , finally it does not need an additional color restoration process. Also we introduce a new CEME measure to evaluate the quality of the enhanced color images. Preliminary results show that the α-rooting based on the quaternion Fourier transform enhancement method out-performs other enhancement methods such as the Fourier transform based α-rooting algorithm and the Multi scale Retinex. On top, the new method not only provides true color fidelity for poor quality images but also averages the color components to gray value for balancing colors. It can be used to enhance edge information and sharp features in images, as well as for enhancing even low contrast images. The proposed algorithms are simple to apply and design, which makes them very practical in image enhancement.
DEFF Research Database (Denmark)
Uceda Otero, E. P.; Eliel, G. S. N.; Fonseca, E. J. S.;
2013-01-01
In this work we have used Fourier transform infrared (FTIR) / vibrational absorption (VA) spectroscopy to study two cancer cell lines: the Henrietta Lacks (HeLa) human cervix carcinoma and 5637 human bladder carcinoma cell lines. Our goal is to experimentally investigate biochemical changes and d...
Shifting and Variational Properties for Fourier-Feynman Transform and Convolution
Directory of Open Access Journals (Sweden)
Byoung Soo Kim
2015-01-01
Full Text Available Shifting, scaling, modulation, and variational properties for Fourier-Feynman transform of functionals in a Banach algebra S are given. Cameron and Storvick's translation theorem can be obtained as a corollary of our result. We also study shifting, scaling, and modulation properties for the convolution product of functionals in S.
Wuttke, Joachim
2009-01-01
An algorithm is described for computing the Laplace transform (one-sided Fourier sine and cosine transform) of the stretched (or compressed) exponential function exp(-t^beta) (also known as Kohlrausch-Williams-Watts function, as characteristic function of a Levy stable distribution, or as complementary cumulative Weibull distribution) for exponents beta between 0.1 and 2. For low and high frequencies, the well-known series expansions are used; for intermediate frequencies, the explicit integration is strongly accelerated by the Ooura-Mori double exponential transformation. The algorithm is implemented in C as library libkww. The source code is available at http://www.messen-und-deuten.de/kww
Gundreddy, Rohith Reddy; Tan, Maxine; Qui, Yuchen; Zheng, Bin
2015-03-01
The purpose of this study is to develop and test a new content-based image retrieval (CBIR) scheme that enables to achieve higher reproducibility when it is implemented in an interactive computer-aided diagnosis (CAD) system without significantly reducing lesion classification performance. This is a new Fourier transform based CBIR algorithm that determines image similarity of two regions of interest (ROI) based on the difference of average regional image pixel value distribution in two Fourier transform mapped images under comparison. A reference image database involving 227 ROIs depicting the verified soft-tissue breast lesions was used. For each testing ROI, the queried lesion center was systematically shifted from 10 to 50 pixels to simulate inter-user variation of querying suspicious lesion center when using an interactive CAD system. The lesion classification performance and reproducibility as the queried lesion center shift were assessed and compared among the three CBIR schemes based on Fourier transform, mutual information and Pearson correlation. Each CBIR scheme retrieved 10 most similar reference ROIs and computed a likelihood score of the queried ROI depicting a malignant lesion. The experimental results shown that three CBIR schemes yielded very comparable lesion classification performance as measured by the areas under ROC curves with the p-value greater than 0.498. However, the CBIR scheme using Fourier transform yielded the highest invariance to both queried lesion center shift and lesion size change. This study demonstrated the feasibility of improving robustness of the interactive CAD systems by adding a new Fourier transform based image feature to CBIR schemes.
Fractional Fourier transform for partially coherent beam in spatial-frequency domain
Institute of Scientific and Technical Information of China (English)
Cai Yang-Jian; Lin Qiang
2004-01-01
By using Fourier transform and the tensor analysis method, the fractional Fourier transform (FRT) in the spatialfrequency domain for partially coherent beams is derived. Based on the FRT in the spatial-frequency domain, an analytical transform formula is derived for a partially coherent twisted anisotropic Gaussian-Schell model (GSM) beam passing through the FRT system. The connections between the FRT formula and the generalized diffraction integral formulae for partially coherent beams through an aligned optical system and a misaligned optical system in the spatialfrequency domain are discussed, separately. By using the derived formula, the intensity distribution of partially coherent twisted anisotropic GSM beams in the FRT plane are studied in detail. The formula derived provide a convenient tool for analysing and calculating the FRTs of the partially coherent beams in spatial-frequency domain.
Park, G Barratt; Field, Robert W
2016-05-28
Since its invention in 2006, the broadband chirped pulse Fourier transform spectrometer has transformed the field of microwave spectroscopy. The technique enables the collection of a ≥10 GHz bandwidth spectrum in a single shot of the spectrometer, which allows broadband, high-resolution microwave spectra to be acquired several orders of magnitude faster than what was previously possible. We discuss the advantages and challenges associated with the technique and look back on the first ten years of chirped pulse Fourier transform spectroscopy. In addition to enabling faster-than-ever structure determination of increasingly complex species, the technique has given rise to an assortment of entirely new classes of experiments, ranging from chiral sensing by three-wave mixing to microwave detection of multichannel reaction kinetics. However, this is only the beginning. Future generations of microwave experiments will make increasingly creative use of frequency-agile pulse sequences for the coherent manipulation and interrogation of molecular dynamics.
Efficient method for localized functions using domain transformation and Fourier sine series
Jovanovic, Raka; Alharbi, Fahhad H
2014-01-01
An efficient approach to handle localized states by using spectral methods (SM) in one and three dimensions is presented. The method consists of transformation of the infinite domain to the bounded domain in $(0, \\pi)$ and using the Fourier sine series as a set of basis functions for the SM. It is shown that with an appropriate choice of transformation functions, this method manages to preserve the good properties of original SMs; more precisely, superb computational efficiency when high level of accuracy is necessary. This is made possible by analytically exploiting the properties of the transformation function and the Fourier sine series. An especially important property of this approach is the possibility of calculating the Hartree energy very efficiently. This is done by exploiting the positive properties of the sine series as a basis set and conducting an extinctive part of the calculations analytically. We illustrate the efficiency of this method and implement it to solve the Poisson's and Helmholtz equ...
Perspective: The first ten years of broadband chirped pulse Fourier transform microwave spectroscopy
Park, G. Barratt; Field, Robert W.
2016-05-01
Since its invention in 2006, the broadband chirped pulse Fourier transform spectrometer has transformed the field of microwave spectroscopy. The technique enables the collection of a ≥10 GHz bandwidth spectrum in a single shot of the spectrometer, which allows broadband, high-resolution microwave spectra to be acquired several orders of magnitude faster than what was previously possible. We discuss the advantages and challenges associated with the technique and look back on the first ten years of chirped pulse Fourier transform spectroscopy. In addition to enabling faster-than-ever structure determination of increasingly complex species, the technique has given rise to an assortment of entirely new classes of experiments, ranging from chiral sensing by three-wave mixing to microwave detection of multichannel reaction kinetics. However, this is only the beginning. Future generations of microwave experiments will make increasingly creative use of frequency-agile pulse sequences for the coherent manipulation and interrogation of molecular dynamics.
DEFF Research Database (Denmark)
Clausen, Anders; Guan, Pengyu; Mulvad, Hans Christian Hansen;
2014-01-01
All-optical time-domain Optical Fourier Transformation utilised for signal processing of ultra-high-speed OTDM signals and OFDM signals will be presented.......All-optical time-domain Optical Fourier Transformation utilised for signal processing of ultra-high-speed OTDM signals and OFDM signals will be presented....
Directory of Open Access Journals (Sweden)
Vahid Bagheri
2014-11-01
Full Text Available This paper introduce a new data SONAR classification method based on Short-Time Fractional Fourier Transform (STFrFT analysis. The passive SONAR system receives the acoustic signals radiated by vessels and attempts to categorize them as a function of the similarities between vessels of the same class.Here, a time-frequency processing and feature extraction method is developed in order to improve the performance of a feedforwardneural network, which is used to classify five classes of vessels.Processing of time-varying signals in fractional fourier domain allows us to estimate the signal with higher concentration than conventional fourier domain, making the technique robust against additive noise, maintaining same computational complexity. With the purpose of dimension reduction and classification improvement, we use Linear Discriminant Analysis (LDA technique. The feasibility of the proposed technique (STFrFTLDA has been tested experimentally using a real database. The experimental results show the superiority of the proposed method
Institute of Scientific and Technical Information of China (English)
QI Lin; TAO Ran; ZHOU Siyong; WANG Yue
2004-01-01
This paper presents a new method for the detection and parameter estimation of multicomponent LFM signals based on the fractional Fourier transform. For the optimization in the fractional Fourier domain, an algorithm based on Quasi-Newton method is proposed which consists of two steps of searching, leading to a reduction in computation without loss of accuracy. And for multicomponent signals, we further propose a signal separation technique in the fractional Fourier domain which can effectively suppress the interferences on the detection of the weak components brought by the stronger components. The statistical analysis of the estimate errors is also performed which perfects the method theoretically, and finally, simulation results are provided to show the validity of our method.
Lu, Wenlong; Xie, Junwei; Wang, Heming; Sheng, Chuan
2016-01-01
Inspired by track-before-detection technology in radar, a novel time-frequency transform, namely polynomial chirping Fourier transform (PCFT), is exploited to extract components from noisy multicomponent signal. The PCFT combines advantages of Fourier transform and polynomial chirplet transform to accumulate component energy along a polynomial chirping curve in the time-frequency plane. The particle swarm optimization algorithm is employed to search optimal polynomial parameters with which the PCFT will achieve a most concentrated energy ridge in the time-frequency plane for the target component. The component can be well separated in the polynomial chirping Fourier domain with a narrow-band filter and then reconstructed by inverse PCFT. Furthermore, an iterative procedure, involving parameter estimation, PCFT, filtering and recovery, is introduced to extract components from a noisy multicomponent signal successively. The Simulations and experiments show that the proposed method has better performance in component extraction from noisy multicomponent signal as well as provides more time-frequency details about the analyzed signal than conventional methods. PMID:27512636
Maccone, Claudio
2007-04-01
The present article describes that the range of any radiotelescope (and radar in general) may be increased by virtue of software, if one replaces the fast Fourier transform by the Karhunen Loève transform. The range increases with the inverse of the fourth root of the signal-to-noise ratio when this ratio decreases. Thus, the range on any radiotelescope (and radar) may be increased without changing the hardware at all, but by changing the software only. This improvement in the range of the radiotelescope is currently implemented at the 32-m antenna located at Medicina, near Bologna, in Italy, for both SETI and general radioastronomy.
Fast large scale structure perturbation theory using one-dimensional fast Fourier transforms
Schmittfull, Marcel; Vlah, Zvonimir; McDonald, Patrick
2016-05-01
The usual fluid equations describing the large-scale evolution of mass density in the universe can be written as local in the density, velocity divergence, and velocity potential fields. As a result, the perturbative expansion in small density fluctuations, usually written in terms of convolutions in Fourier space, can be written as a series of products of these fields evaluated at the same location in configuration space. Based on this, we establish a new method to numerically evaluate the 1-loop power spectrum (i.e., Fourier transform of the 2-point correlation function) with one-dimensional fast Fourier transforms. This is exact and a few orders of magnitude faster than previously used numerical approaches. Numerical results of the new method are in excellent agreement with the standard quadrature integration method. This fast model evaluation can in principle be extended to higher loop order where existing codes become painfully slow. Our approach follows by writing higher order corrections to the 2-point correlation function as, e.g., the correlation between two second-order fields or the correlation between a linear and a third-order field. These are then decomposed into products of correlations of linear fields and derivatives of linear fields. The method can also be viewed as evaluating three-dimensional Fourier space convolutions using products in configuration space, which may also be useful in other contexts where similar integrals appear.
Ion collision cross section measurements in Fourier transform-based mass analyzers.
Li, Dayu; Tang, Yang; Xu, Wei
2016-06-01
With the increasing demands of molecular structure analysis, several methods have been developed to measure ion collision cross sections within Fourier transform (FT) based mass analyzers. Particularly in the recent three years since 2012, the method of obtaining biomolecule collision cross sections was achieved in Fourier transform ion cyclotron resonance (FT-ICR) cells. Furthermore, similar methods have been realized or proposed for orbitraps and quadrupole ion traps. This technique adds a new ion structure analysis capability to FT-based mass analyzers. By providing complementary ion structure information, it could be used together with tandem mass spectrometry and ion mobility spectroscopy techniques. Although many questions and challenges remain, this technique potentially would greatly enhance the ion structure analysis capability of a mass spectrometer, and provide a new tool for chemists and biochemists. PMID:26788551
Kaneko, Tak
2008-01-01
Context: Fourier transform (or lag) correlators in radio interferometers can serve as an efficient means of synthesising spectral channels. However aliasing corrupts the edge channels so they usually have to be excluded from the data set. In systems with around 10 channels, the loss in sensitivity can be significant. In addition, the low level of residual aliasing in the remaining channels may cause systematic errors. Moreover, delay errors have been widely reported in implementations of broadband analogue correlators and simulations have shown that delay errors exasperate the effects of aliasing. Aims: We describe a software-based approach that suppresses aliasing by oversampling the cross-correlation function. This method can be applied to interferometers with individually-tracking antennas equipped with a discrete path compensator system. It is based on the well-known property of interferometers where the drift scan response is the Fourier transform of the source's band-limited spectrum. Methods: In this p...
Sun, Zhiwei; Zhi, Ya'nan; Liu, Liren; Sun, Jianfeng; Zhou, Yu; Hou, Peipei
2013-09-01
The synthetic aperture imaging ladar (SAIL) systems typically generate large amounts of data difficult to compress with digital method. This paper presents an optical SAIL processor based on compensation of quadratic phase of echo in azimuth direction and two dimensional Fourier transform. The optical processor mainly consists of one phase-only liquid crystal spatial modulator(LCSLM) to load the phase data of target echo and one cylindrical lens to compensate the quadratic phase and one spherical lens to fulfill the task of two dimensional Fourier transform. We show the imaging processing result of practical target echo obtained by a synthetic aperture imaging ladar demonstrator. The optical processor is compact and lightweight and could provide inherent parallel and the speed-of-light computing capability, it has a promising application future especially in onboard and satellite borne SAIL systems.
DEFF Research Database (Denmark)
Palushani, Evarist; Oxenløwe, Leif Katsuo; Galili, Michael;
2009-01-01
This paper reports on the generation of 1.6-ps fullwidth at half-maximum flat-top pulses by the optical Fourier transform technique, and the utilization of these pulses in a 320-Gb/s demultiplexing experiment. It is demonstrated how a narrow pulse having a 15-nm wide third-order super-Gaussian sp......This paper reports on the generation of 1.6-ps fullwidth at half-maximum flat-top pulses by the optical Fourier transform technique, and the utilization of these pulses in a 320-Gb/s demultiplexing experiment. It is demonstrated how a narrow pulse having a 15-nm wide third-order super...
Shi, Xiaoyan; Zhao, Daomu
2011-02-10
Based on joint transform correlator (JTC) architecture and holographic techniques, a new method for image hiding is presented. A hidden image encrypted by JTC architecture is embedded in the Fourier hologram of the host image. Inverse Fourier transform can be used to obtain the watermarked image, and JTC architecture is used to decode the hidden image from the watermarked hologram. Unlike other watermarking techniques, by prechoosing information, the noise added to the recovered hidden image by the host can be reduced. Unlike other watermarking systems based on double random-phase encoding, no conjugate key is used to recover the hidden image. Theoretical analyses have shown the system's feasibility. Computer simulations are presented to verify the system's validity and efficiency. Numerical simulations also show that the proposed system is robust enough to resist attacks, such as occlusion, noise, and filtering.
Sharpening the response of an FIR filter using Fractional Fourier Transform
Directory of Open Access Journals (Sweden)
Somesh Chaturvedi
2012-03-01
Full Text Available In this paper we have implemented FIR filter with the help of Kaiser Window and Fractional Fourier Transform (FRFT. The window shape parameter is tuned for the transition band by considering linear phase FRFT Finite Impulse Response (FIR filter. Here FRFT of Kaiser Window is taken and convolved with the response function for tuning purposes of the transition band which makes effective transition band. This proposed method includes the change of parameters of Kaiser window by which other windows like Rectangle, Bartlett, Hamming Blackman and Hanning windows are generated by using FRFT. The efficiencies of this method in terms of main lobe and side ripples are better than the above mentioned windows under Fourier transform.
Information Transmission using the Nonlinear Fourier Transform, Part I: Mathematical Tools
Yousefi, Mansoor I
2012-01-01
The nonlinear Fourier transform (NFT), a powerful tool in soliton theory and exactly solvable models, is a method for solving integrable partial differential equations governing wave propagation in certain nonlinear media. The NFT decorrelates signal degrees-of-freedom in such models, in much the same way that the Fourier transform does for linear systems. In this paper, this observation is exploited for data transmission over integrable channels such as optical fibers, where pulse propagation is governed by the nonlinear Schr\\"odinger equation. In this transmission scheme, which can be viewed as a nonlinear analogue of orthogonal frequency division multiplexing commonly used in linear channels, information is encoded in the spectral amplitudes associated with nonlinear frequencies. Unlike most other fiber-optic transmission schemes, this technique deals with both dispersion and nonlinearity directly and unconditionally without the need for dispersion or nonlinearity compensation methods. This paper explains ...
Nagler, Peter C; Kogut, Alan; Tucker, Gregory S
2015-01-01
The detection of the primordial B-mode polarization signal of the cosmic microwave background (CMB) would provide evidence for inflation. Yet as has become increasingly clear, the detection of a such a faint signal requires an instrument with both wide frequency coverage to reject foregrounds and excellent control over instrumental systematic effects. Using a polarizing Fourier transform spectrometer (FTS) for CMB observations meets both these requirements. In this work, we present an analysis of instrumental systematic effects in polarizing Fourier transform spectrometers, using the Primordial Inflation Explorer (PIXIE) as a worked example. We analytically solve for the most important systematic effects inherent to the FTS - emissive optical components, misaligned optical components, sampling and phase errors, and spin synchronous effects - and demonstrate that residual systematic error terms after corrections will all be at the sub-nK level, well below the predicted 100 nK B-mode signal.
Institute of Scientific and Technical Information of China (English)
无
2007-01-01
The mismatch between echo and replica caused by underwater moving target(UMT)'s radial velocity degrades the detection performance of the matched filter(MF) for the linear frequency modulation(LFM) signal. By using the focusing property of fractional Fourier transform(FRFT) to that signal, a detection algorithm for UMT's LFM echo based on the discrete fractional Fourier transform(DFRFT) is proposed. This algorithm is less affected by the target's radial velocity compared with the other MF detection algorithm utilizing zero radial velocity replica(ZRVR), and the mathematical relation between the output peak positions of these two algorithms exists in the case of existence of target echo. The algorithm can also estimate the target distance by using this relation. The simulation and experiment show that this algorithm's detection performance is better than or equivalent to that of the other MF algorithm utilizing ZRVR for the LFM echo of UMT with unknown radial velocity under reverberation noise background.
Gaseous effluent monitoring and identification using an imaging Fourier transform spectrometer
Energy Technology Data Exchange (ETDEWEB)
Carter, M.R.; Bennett, C.L.; Fields, D.J.; Hernandez, J.
1993-10-01
We are developing an imaging Fourier transform spectrometer for chemical effluent monitoring. The system consists of a 2-D infrared imaging array in the focal plane of a Michelson interferometer. Individual images are coordinated with the positioning of a moving mirror in the Michelson interferometer. A three dimensional data cube with two spatial dimensions and one interferogram dimension is then Fourier transformed to produce a hyperspectral data cube with one spectral dimension and two spatial dimensions. The spectral range of the instrument is determined by the choice of optical components and the spectral range of the focal plane array. Measurements in the near UV, visible, near IR, and mid-IR ranges are possible with the existing instrument. Gaseous effluent monitoring and identification measurements will be primarily in the ``fingerprint`` region of the spectrum, ({lambda} = 8 to 12 {mu}m). Initial measurements of effluent using this imaging interferometer in the mid-IR will be presented.
Ion collision cross section measurements in Fourier transform-based mass analyzers.
Li, Dayu; Tang, Yang; Xu, Wei
2016-06-01
With the increasing demands of molecular structure analysis, several methods have been developed to measure ion collision cross sections within Fourier transform (FT) based mass analyzers. Particularly in the recent three years since 2012, the method of obtaining biomolecule collision cross sections was achieved in Fourier transform ion cyclotron resonance (FT-ICR) cells. Furthermore, similar methods have been realized or proposed for orbitraps and quadrupole ion traps. This technique adds a new ion structure analysis capability to FT-based mass analyzers. By providing complementary ion structure information, it could be used together with tandem mass spectrometry and ion mobility spectroscopy techniques. Although many questions and challenges remain, this technique potentially would greatly enhance the ion structure analysis capability of a mass spectrometer, and provide a new tool for chemists and biochemists.
The data processing pipeline for the Herschel SPIRE Fourier Transform Spectrometer
Fulton, T; Polehampton, E T; Valtchanov, I; Hopwood, R; Lu, N; Baluteau, J -P; Mainetti, G; Pearson, C; Papageorgiou, A; Guest, S; Zhang, L; Imhof, P; Swinyard, B M; Griffin, M J; Lim, T L
2016-01-01
We present the data processing pipeline to generate calibrated data products from the Spectral and Photometric Imaging Receiver (SPIRE) imaging Fourier Transform Spectrometer on the Herschel Space Observatory. The pipeline processes telemetry from SPIRE observations and produces calibrated spectra for all resolution modes. The spectrometer pipeline shares some elements with the SPIRE photometer pipeline, including the conversion of telemetry packets into data timelines and calculation of bolometer voltages. We present the following fundamental processing steps unique to the spectrometer: temporal and spatial interpolation of the scan mechanism and detector data to create interferograms; Fourier transformation; apodization; and creation of a data cube. We also describe the corrections for various instrumental effects including first- and second-level glitch identification and removal, correction of the effects due to emission from the Herschel telescope and from within the spectrometer instrument, interferogra...
Surpassing the Path-Limited Resolution of a Fourier Transform Spectrometer with Frequency Combs
Maslowski, Piotr; Johansson, Alexandra C; Khodabakhsh, Amir; Kowzan, Grzegorz; Rutkowski, Lucile; Mills, Andrew A; Mohr, Christian; Jiang, Jie; Fermann, Martin E; Foltynowicz, Aleksandra
2015-01-01
Fourier transform spectroscopy based on incoherent light sources is a well-established tool in research fields from molecular spectroscopy and atmospheric monitoring to material science and biophysics. It provides broadband molecular spectra and information about the molecular structure and composition of absorptive media. However, the spectral resolution is fundamentally limited by the maximum delay range ({\\Delta}$_{max}$) of the interferometer, so acquisition of high-resolution spectra implies long measurement times and large instrument size. We overcome this limit by combining the Fourier transform spectrometer with an optical frequency comb and measuring the intensities of individual comb lines by precisely matching the {\\Delta}$_{max}$ to the comb line spacing. This allows measurements of absorption lines narrower than the nominal (optical path-limited) resolution without ringing effects from the instrumental lineshape and reduces the acquisition time and interferometer length by orders of magnitude.
Suppression law of quantum states in a 3D photonic fast Fourier transform chip
Crespi, Andrea; Osellame, Roberto; Ramponi, Roberta; Bentivegna, Marco; Flamini, Fulvio; Spagnolo, Nicolò; Viggianiello, Niko; Innocenti, Luca; Mataloni, Paolo; Sciarrino, Fabio
2016-01-01
The identification of phenomena able to pinpoint quantum interference is attracting large interest. Indeed, a generalization of the Hong–Ou–Mandel effect valid for any number of photons and optical modes would represent an important leap ahead both from a fundamental perspective and for practical applications, such as certification of photonic quantum devices, whose computational speedup is expected to depend critically on multi-particle interference. Quantum distinctive features have been predicted for many particles injected into multimode interferometers implementing the Fourier transform over the optical modes. Here we develop a scalable approach for the implementation of the fast Fourier transform algorithm using three-dimensional photonic integrated interferometers, fabricated via femtosecond laser writing technique. We observe the suppression law for a large number of output states with four- and eight-mode optical circuits: the experimental results demonstrate genuine quantum interference between the injected photons, thus offering a powerful tool for diagnostic of photonic platforms. PMID:26843135
Directory of Open Access Journals (Sweden)
Eduardo O. Cerqueira
2000-10-01
Full Text Available Instrumental data always present some noise. The analytical data information and instrumental noise generally has different frequencies. Thus is possible to remove the noise using a digital filter based on Fourier transform and inverse Fourier transform. This procedure enhance the signal/noise ratio and consecutively increase the detection limits on instrumental analysis. The basic principle of Fourier transform filter with modifications implemented to improve its performance is presented. A numerical example, as well as a real voltammetric example are showed to demonstrate the Fourier transform filter implementation. The programs to perform the Fourier transform filter, in Matlab and Visual Basic languages, are included as appendices
Fourier Transform Infrared (FTIR) Spektroskopinin Gıda Analizlerine Uygulanması (İngilizce)
Erkahveci, Ayşe; Karaali, Artemis
1996-01-01
The rapid analysis of food products is crucial in food industry for adjustment of manufacturing processes while production is underway. Recent research has led to development of rapid quality-control methodologies for the food industry through the application of Fourier Transform Infrared (FTIR) spectroscopy and FTIR technology has offered substantial potential as a quantitative quality control tool for the food industry. FTIR analysis methods are convenient, rapid and automatable, and in con...
The Fourier Transform and FTIR System as a Technique in Semiconductor Research
International Nuclear Information System (INIS)
In this paper a simple treatment of Fourier transform in FTIR with an introductory picture of its use in Semiconductor Research is described. A brief account of research foucssing on optical characterization of silicon (Si) wafers is outlined. The smeasurement of residual oxygen (O2) concentration is an important indicator in determining the overall quality of the finished Si wafers. The O2 concentration is determined directly from an infrared absorption band occurring at 1106 cm-1 in the Si Lattice
Directory of Open Access Journals (Sweden)
Mark J. Tobin
2013-08-01
Full Text Available Fourier transform infrared (FTIR microspectroscopy shows potential as a benign, objective and rapid tool to screen pluripotent and multipotent stem cells for clinical use. It offers a new experimental approach that provides a holistic measurement of macromolecular composition such that a signature representing the internal cellular phenotype is obtained. The use of this technique therefore contributes information that is complementary to that acquired by conventional genetic and immunohistochemical methods.
Approaches towards Implementation of Multi-bit Digital Receiver using Fast Fourier Transform
Abhijit S. Kulkarni; Vijesh P.; Hemant V. Paranjape; K. Maheswara Reddy
2013-01-01
This paper compares different digital receiver signal processing schemes as applied to current ESM/RWR systems. The schemes include fast fourier transform (FFT)-based, FIR filter-based and mixed architectures. Use of polyphase FFT and IIR filters is also discussed. The specifications and signal processing requirements of a modern digital electronic warfare (EW) receiver are discussed. The design procedures and architectures for all the schemes are brought out. The tradeoffs involved in select...
Institute of Scientific and Technical Information of China (English)
FAN Hong-Yi; FAN Yue
2003-01-01
By introducing a convenient complex form of the α-th 2-dimensional fractional Fourier transform (CFFT) operation we find that it possesses new eigenmodes which are two-mode Hermite polynomials. We prove the eigenvalues of propagation in quadratic graded-index medium over a definite distance are the same as the eigenvalues of the α-th CFFT, which means that our definition of the α-th CFFT is physically meaningful.
Mossoulina, O. A.; Kirilenko, M. S.; Khonina, S. N.
2016-08-01
We use radial Fractional Fourier transform to model vortex laser beams propagation in optical waveguides with parabolic dependence of the refractive index. To overcome calculation difficulties at distances proportional to a quarter of the period we use varied calculation step. Numerical results for vortex modes superposition propagation in a parabolic optical fiber show that the transverse beam structure can be changed significantly during the propagation. To provide stable transverse distribution input scale modes should be in accordance with fiber parameters.
Computing Greeks for L\\'evy Models: The Fourier Transform Approach
Federico De Olivera; Ernesto Mordecki
2014-01-01
The computation of Greeks for exponential L\\'evy models are usually approached by Malliavin Calculus and other methods, as the Likelihood Ratio and the finite difference method. In this paper we obtain exact formulas for Greeks of European options based on the Lewis formula for the option value. Therefore, it is possible to obtain accurate approximations using Fast Fourier Transform. We will present an exhaustive development of Greeks for Call options. The error is shown for all Greeks in the...
Büchl, Nicole Ramona
2009-01-01
Im Rahmen der vorliegenden Arbeit wurden drei Systeme zur Identifizierung von Hefen basierend auf der Anwendung von Fourier-transform Infrarotspektroskopie und künstlichen neuronalen Netzen etabliert. Die zur Entwicklung verwendeten Referenzstämme wurden mittels verschiedener molekularbiologischer Methoden zur Identifizierung auf Art- und Stammebene überprüft. So konnten zuverlässige Methoden zur eindeutigen Identifizierung der im Lebens-, Futtermittel- und klinischen Bereich vorkommenden Hef...
Song, Xinbing; Sun, Yifan; Li, Pengyun; Qin, Hongwei; Zhang, Xiangdong
2015-09-15
We perform Bell's measurement for the non-separable correlation between polarization and orbital angular momentum from the same classical vortex beam. The violation of Bell's inequality for such a non-separable classical correlation has been demonstrated experimentally. Based on the classical vortex beam and non-quantum entanglement between the polarization and the orbital angular momentum, the Hadamard gates and conditional phase gates have been designed. Furthermore, a quantum Fourier transform has been implemented experimentally.
Analysis of Index Gases of Coal Spontaneous Combustion Using Fourier Transform Infrared Spectrometer
Xiaojun Tang; Yuntao Liang; Haozhe Dong; Yong Sun; Haizhu Luo
2014-01-01
Analysis of the index gases of coal for the prevention of spontaneous combustion is of great importance for the enhancement of coal mine safety. In this work, Fourier Transform Infrared Spectrometer (FTIRS) is presented to be used to analyze the index gases of coal in real time to monitor spontaneous combustion conditions. Both the instrument parameters and the analysis method are introduced at first by combining characteristics of the absorption spectra of the target analyte with the analysi...
Using single buffers and data reorganization to implement a multi-megasample fast Fourier transform
Brown, R. D.
1992-01-01
Data ordering in large fast Fourier transforms (FFT's) is both conceptually and implementationally difficult. Discribed here is a method of visualizing data orderings as vectors of address bits, which enables the engineer to use more efficient data orderings and reduce double-buffer memory designs. Also detailed are the difficulties and algorithmic solutions involved in FFT lengths up to 4 megasamples (Msamples) and sample rates up to 80 MHz.
Exhibition of the periodicity of Quantum Fourier Transformation in Nuclear Magnetic Resonance
Peng, Xinhua; Zhu, Xiwen; Fang, Ximing; Feng, Mang; Yang, Xiaodong; Liu, Maili; Gao, Kelin
2002-01-01
The remarkable capability of quantum Fourier transformation (QFT) to extract the periodicity of a given periodic function has been exhibited by using nuclear magnetic resonance (NMR) techniques. Two separate sets of experiments were performed. In a full QFT, the periodicity were validated with state tomography and fidelity measurements. For a simplified QFT, the three-qubit pseudo-pure state was created by introducting an additional observer spin, and the spectra recorded on the observer spin...
On the Entangled Fractional Fourier Transform in Tripartite Entangled State Representation
Institute of Scientific and Technical Information of China (English)
FAN Hong-Yi; JIANG Nian-Quan
2003-01-01
Based on the newly constructed two mutually conjugate 3-mode entangled states of continuum variablesin three-mode Fock space we introduce entangled fractional Fourier transform (EFFT) for the tripartite entangled staterepresentations, which are not a direct product of three 1-dimensional FFTs. The eigenmodes of EFFT are obtained,which is different from the usual Hermite polynomials. The EFFT of the three-mode squeezed state is derived.
Raman spectroscopy of gases with a Fourier transform spectrometer - The spectrum of D2
Jennings, D. E.; Weber, A.; Brault, J. W.
1986-01-01
Fourier transform spectrometry (FTS) is presently used to record the spontaneous incoherent laser Raman spectra of gases. The high resolution, sensitivity, calibration accuracy and spectral coverage achieved demonstrate the viability of FTS for Raman spectroscopy. Attention is given to the coefficients obtained by fitting measurements obtained from the spectrum of D2, containing both v = 0-0 and 1-0 transitions, to the Dunham (1932) expansion of the vibration-rotation energy levels.
Indian Academy of Sciences (India)
Amit K Sharma; D P Chhachhia; D Mohan; A K Aggarwal
2008-01-01
This paper describes a simple method for making dual beam encoded extended fractional Fourier transform (EFRT) security holograms. The hologram possesses different stages of encoding so that security features are concealed and remain invisible to the counterfeiter. These concealed and encoded anticounterfeit security features in the security hologram can only be read through a key hologram. Key hologram also facilitates in-built repositioning of security hologram. The method of fabrication, the principle of reconstruction and the experimental results are presented.
Automotive FMCW Radar-enhanced Range Estimation via a Local Resampling Fourier Transform
Cailing Wang; Huajun Liu; Guang Han; Xiaoyuan Jing
2016-01-01
In complex traffic scenarios, more accurate measurement and discrimination for an automotive frequency-modulated continuous-wave (FMCW) radar is required for intelligent robots, driverless cars and driver-assistant systems. A more accurate range estimation method based on a local resampling Fourier transform (LRFT) for a FMCW radar is developed in this paper. Radar signal correlation in the phase space sees a higher signal-noise-ratio (SNR) to achieve more accurate ranging, and the LRFT - whi...
An Imaging Fourier Transform Spectrometer for the Next Generation Space Telescope
Graham, J R
1999-01-01
Due to its simultaneous deep imaging and integral field spectroscopic capability, an Imaging Fourier Transform Spectrograph (IFTS) is ideally suited to the Next Generation Space Telescope (NGST) mission, and offers opportunities for tremendous scientific return in many fields of astrophysical inquiry. We describe the operation and quantify the advantages of an IFTS for space applications. The conceptual design of the Integral Field Infrared Spectrograph (IFIRS) is a wide field (5'.3 x 5'.3) four-port imaging Michelson interferometer.
Kumar, Manoj; Shakher, Chandra
2016-04-01
In this paper, moisture induced deformation and shrinkage behaviour of deodar wood during convective drying is experimentally investigated by using digital holographic interferometry. There induces dimensional changes in wood due to the moisture absorption and desorption. Lensless Fourier transform digital holographic interferometry (LLFTDH) is used to study the moisture induced deformation and strain distribution in deodar wood. The proposed technique having high sensitivity and enables the observation of deformation and strain distribution during the variations of moisture content in the deodar wood.
Kamnev, A. A.; Ristić, M.; Antonyuk, L. P.; Chernyshev, A. V.; Ignatov, V. V.
1997-06-01
The data of Fourier transform infrared (FTIR) spectroscopic measurements performed on intact cells of the soil nitrogen-fixing bacterium Azospirillum brasilense grown in a standard medium and under the conditions of an increased metal uptake are compared and discussed. The structural FTIR information obtained is considered together with atomic absorption spectrometry (AAS) data on the content of metal cations in the bacterial cells. Some methodological aspects concerning preparation of bacterial cell samples for FTIR measurements are also discussed.
International Nuclear Information System (INIS)
The subject of this thesis is gas phase ion/molecule reactions as studied by Fourier Transform Ion Cyclotron Resonance (FT-ICR) mass spectrometry (chapter 2 contains a short description of this method). Three chapters are mainly concerned with mechanistic aspects of gas phase ion/molecule reactions. An equally important aspect of the thesis is the stability and reactivity of α-thio carbanions, dipole stabilized carbanions and homoenolate anions, dealt with in the other four chapters. (Auth.)
Fourier Transform Analysis of STM Images of Multilayer Graphene Moir\\'e Patterns
Joucken, Frédéric; Frising, Fernande; Sporken, Robert
2014-01-01
With the help of a simple model, we analyze Scanning Tunneling Microscopy images of simple and double moir\\'e patterns resulting from misoriented bi- and tri-layers graphene stacks. It is found that the model reproduces surprisingly well non-trivial features observed in the Fast Fourier Transform of the images. We point out difficulties due to those features in interpreting the patterns seen on the FFT.
Fuzzy Logic Classification of Imaging Laser Desorption Fourier Transform Mass Spectrometry Data
McJunkin, Timothy R.; Scott, Jill R.
2006-01-01
A fuzzy logic based classification engine has been developed for classifying mass spectra obtained with an imaging internal source Fourier transform mass spectrometer (I^2LD-FTMS). Traditionally, an operator uses the relative abundance of ions with specific mass-to-charge (m/z) ratios to categorize spectra. An operator does this by comparing the spectrum of m/z versus abundance of an unknown sample against a library of spectra from known samples. Automated positioning and acquisition allow I^...
A compact Fourier transform imaging spectrometer employing a variable gap Fabry-Perot interferometer
Lucey, Paul G.; Akagi, Jason; Bingham, Adam L.; Hinrichs, John L.; Knobbe, Edward T.
2014-05-01
Fourier transform spectroscopy is a widely employed method for obtaining visible and infrared spectral imagery, with applications ranging from the desktop to remote sensing. Most fielded Fourier transform spectrometers (FTS) employ the Michelson interferometer and measure the spectrum encoded in a time-varying signal imposed by the source spectrum interaction with the interferometer. A second, less widely used form of FTS is the spatial FTS, where the spectrum is encoded in a pattern sampled by a detector array. Recently we described using a Fabry-Perot interferometer, with a deliberately wedged gap geometry and engineered surface reflectivities, to produce an imaging spatial FTS. The Fabry-Perot interferometer can be much lighter and more compact than a conventional interferometer configuration, thereby making them suitable for portable and handheld applications. This approach is suitable for use over many spectral regimes of interest, including visible and infrared regions. Primary efforts to date have focused on development and demonstration of long wave infrared (LWIR) spectral imagers. The LWIR version of the miniaturized Fabry-Perot has been shown to be effective for various applications including spectral imaging-based chemical detection. The compact LWIR spectral imager employs uncooled optics and a microbolometer camera; a handheld version is envisioned for future development. Recent advancements associated with the spatial Fourier Transform imaging spectrometer system are described.
[A novel spatial modulation Fourier transform spectrometer with adjustable spectral resolution].
Lian, Yu-Sheng; Liao, Ning-Fang; Lü, Hang; Wu, Wen-Min; Dong, Zhi-Gang
2014-11-01
In the premise of fulfilling the application requirement, the adjustment of spectral resolution can improve efficiency of data acquisition, data processing and data saving. So, by adjusting the spectral resolution, the performance of spectrometer can be improved, and its application range can be extended. To avoid the problems of the fixed spectral resolution of classical Fourier transform spectrometer, a novel type of spatial modulation Fourier transform spectrometer with adjustable spectral resolution is proposed in this paper. The principle of the novel spectrometer and its interferometer is described. The general expressions of the optical path difference and the lateral shear are induced by a ray tracing procedure. The equivalent model of the novel interferometer is analyzed. Meanwhile, the principle of the adjustment of spectral resolution is analyzed. The result shows that the novel spectrometer has the merits of adjustable spectral resolution, high stability, easy assemblage and adjustment etc. This theoretical study will provide the theoretical basis for the design of the spectrometer with adjustable spectral resolution and expand the application range of Fourier transform spectrometer.
Infrared small target enhancement via phase spectrum of Quaternion Fourier Transform
Qi, Shengxiang; Ma, Jie; Li, Hang; Zhang, Shuiping; Tian, Jinwen
2014-01-01
Small target enhancement is one of the crucial stages in infrared small target detection. In this paper, we propose a new method using phase spectrum of Quaternion Fourier Transform to enhance small targets while suppressing backgrounds for infrared images. This is inspired by the property that regularly Gaussian-like shape small targets could be considered as attractively salient signal in infrared images and the location information of such signal is implicitly contained in the phase spectrum from frequency domain. Formally, in the proposed method, we adopt the phase spectrum of Quaternion Fourier Transform instead of using traditional Fourier Transform to enhance the targets since the quaternion provides at most four data channels than only one for the latter, which could be helpful to broad types of background clutters by adding more information. For the construction of the quaternion, we present a second-order directional derivative filter via facet model to compute four second order directional derivative maps from four directions respectively as the four data channels. This filter is used to suppress noises and distinguish the targets and backgrounds into separably different textures so that it would boost the robustness of small target enhancement. In experiments, some typical infrared images with various scenes are tested to validate the effectiveness of the proposed method. The results demonstrate that our method actually has good performance and outperforms several state-of-the-art methods, which can be further used for infrared small target detection and tracking.
Color pattern recognition based on the joint fractional Fourier transform correlator
Institute of Scientific and Technical Information of China (English)
Weimin Jin; Yupei Zhang
2007-01-01
A new system of multi-channel single-output joint fractional Fourier transform correlator (JFRTC) for color pattern recognition is proposed based on the conventional system of multi-channel single-output joint transform correlator (JTC). The theoretical analysis and optical experiments are performed. With this method, one can obtain three correlation peaks at the output plane which show a pair of desired cross-correlation peaks and one auto-correlation peak. In comparison, the conventional system leads to more correlation peaks playing a noise role in color pattern recognition.
Fourier and Wavelet Transform Analysis of Pressure Signals during Explosive Boiling
Institute of Scientific and Technical Information of China (English)
YIN Tie-Nan; HUAI Xiu-Lan
2008-01-01
@@ The transient pressure in a liquid-pool during explosive boiling of acetone is measured by a micro-pressure-measuring system.The Fast Fourier transform and continuous wavelet transform methods are applied to investigate the frequency characteristics.The results show that the dominant frequency of the explosive boiling is 0-2MHz,and the bubble cluster formed by numerous tiny bubbles departs twice.Analysis and discussions are also conducted to explain the bubble evolution during the explosive boiling.
[The principle and analysis of micro-Fourier transform spectrometer based on MEMS micro-mirror].
Chen, Jian-Jun; Zhu, Yong; Liu, Bo; Wei, Wei; Zhang, Jie; Wang, Ning
2012-11-01
The present paper puts forward a novel micro-Fourier transform infrared spectrometer based on programmable MEMS micro-mirror. This design uses a MEMS micro-mirror and a slantwise reflector to replace the moving mirror system on traditional spectrometer. This paper analyzes the operating principle of this spectrometer and gives the simulation result to prove the feasibility of this method. The results show that the spectral resolution is less than 5 nm in near-infrared wave band, the wave-length accuracy is approximately 1 nm and the sampling period of this spectrometer is approximately 50 ms. This method can use Hadamard transform to improve the SNR.
Drifting Sub-Pulse Analysis Using the Two-Dimensional Fourier Transform
Edwards, R,A; Stappers, B. W.
2002-01-01
The basic form of drifting sub-pulses is that of a periodicity whose phase depends (approximately linearly) on both pulse longitude and pulse number. As such, we argue that the two-dimensional Fourier transform of the longitude-time data (called the Two-Dimensional Fluctuation Spectrum; 2DFS) presents an ideal basis for studies of this phenomenon. We examine the 2DFS of a pulsar signal synthesized using the parameters of an empirical model for sub-pulse behaviour. We show that the transform c...
Directory of Open Access Journals (Sweden)
Siesler H. W.
2006-11-01
Full Text Available The recent extension of the Fourier-Transform (FT technique to the Raman effect has launched Raman spectroscopy into a new era of polymer chemical and physical applications. Thus, the increase in signal-to-noise ratio and the improvement in time resolution have largely enhanced the potential of FT-Raman spectroscopy for analytical applications, the characterization of time-dependent phenomena and the on-line combination with other techniques. Primarily the suppression of fluorescence by shifting the excitation line to the near-infrared (NIR region has contributed to the fast acceptance as an industrial routine tool. Furthermore, the application of fiber optics has opened up the areas of process-control and remote sensing. Les applications de la spectroscopie Raman dans le domaine des polymères sont entrées dans une ère nouvelle, grâce aux récents développements de la technique à transformée de Fourier avec excitation dans le proche infrarouge. L'augmentation du rapport signal sur bruit et l'amélioration de la résolution temporelle ont fortement renforcé les potentialités de la technique en ce qui concerne les applications analytiques, la caractérisation de phénomènes qui dépendent du temps et le couplage en ligne avec d'autres techniques. La suppression du phénomène de fluorescence par déplacement de la longueur d'onde de l'excitatrice dans le proche infrarouge a contribué à l'intégration rapide de l'outil en site industriel. L'emploi de fibres optiques a permis l'accroissement des applications dans le domaine du contrôle des procédés et d'analyser à distance.
Use of Fourier Transformation for Detection of Faults in Underground Power Cables
Institute of Scientific and Technical Information of China (English)
Abhishek Pandey; Nicolas H. Younan
2011-01-01
An analysis of underground power cables is performed using Fourier analysis with the objective of detecting fault and average life of the cables. Three types of cables are used in this experiment： a normal cable, a shorted cable, and a cable with holes. The impedance in each case is computed and Fourier transformation is applied so that the re- suiting impedance magnitude and impedance phase can be examined in the frequency domain. Various windowing tech- niques are applied to the experimental data to eliminate any interference. Fourier analysis is then applied to the imped- ance data calculated from both the sending end voltage and differential voltage. This analysis reveals differences in the frequency response of the three different types of a cable and can eventually be used as a measure for fault detection. Preliminary results reveal the differences in the frequency response. Accordingly, Fourier type methods can be effectively used as low cost and viable solutions to identify and detect faults in underground cables.
Directory of Open Access Journals (Sweden)
Chiu Shen
2005-01-01
Full Text Available A relatively unknown yet powerful technique, the so-called fractional Fourier transform (FrFT, is applied to SAR along-track interferometry (SAR-ATI in order to estimate moving target parameters. By mapping a target's signal onto a fractional Fourier axis, the FrFT permits a constant-velocity target to be focused in the fractional Fourier domain thereby affording orders of magnitude improvement in SCR. Moving target velocity and position parameters are derived and expressed in terms of an optimum fractional angle and a measured fractional Fourier position , allowing a target to be accurately repositioned and its velocity components computed without actually forming an SAR image. The new estimation algorithm is compared with the matched filter bank approach, showing some of the advantages of the FrFT method. The proposed technique is applied to the data acquired by the two-aperture CV580 airborne radar system configured in its along-track mode. Results show that the method is effective in estimating target velocity and position parameters.
MATLAB. Application on Fourier Transform%MATLAB在傅里叶变换中的应用
Institute of Scientific and Technical Information of China (English)
周玲
2011-01-01
Fourier transform knowledge is always difficult to be understood because of high abstract characteristics in the signals and systems. It is realized for the visualization of various operations by the powerfully signal processing functions of MATLAB based on the content of Fourier transform, Fourier series and convolution theorem. The concept of time and frequency domains and the analysis method are deeply understood by MATLAB programming.%针对信号与系统中傅里叶变换具有抽象度高、难于理解的特点，利用MATLAB强大的信号处理功能，结合傅里叶变换、傅里叶级数、卷积定理的内容，通过MATLAB程序设计实现傅里叶变换中各种运算的直观演示。通过MATLAB编程加深对信号时域和频域概念、分析方法的理解。
Sangwine, Stephen J
2010-01-01
We show that the discrete complex, and numerous hypercomplex, Fourier transforms defined and used so far by a number of different researchers can be unified into a single theoretical framework based on a matrix exponential version of Euler's formula $e^{j\\theta}=\\cos\\theta+j\\sin\\theta$, and a matrix root of -1 in place of the imaginary root $j$. The transforms thus defined can be computed numerically using standard matrix multiplications and additions with no hypercomplex code, the complex or hypercomplex algebra being represented by the form of the matrix root of -1, so that the matrix multiplications are equivalent to multiplications in the appropriate algebra. We present examples from the complex, quaternion, and biquaternion algebras, and believe that Clifford algebras would fit the same framework. The significance of this result is both in the theoretical unification achieved, and also in the scope it affords for insight into the structure of the various transforms, since the formulation is such a simple...
Digital watermarking algorithm research of color images based on quaternion Fourier transform
An, Mali; Wang, Weijiang; Zhao, Zhen
2013-10-01
A watermarking algorithm of color images based on the quaternion Fourier Transform (QFFT) and improved quantization index algorithm (QIM) is proposed in this paper. The original image is transformed by QFFT, the watermark image is processed by compression and quantization coding, and then the processed watermark image is embedded into the components of the transformed original image. It achieves embedding and blind extraction of the watermark image. The experimental results show that the watermarking algorithm based on the improved QIM algorithm with distortion compensation achieves a good tradeoff between invisibility and robustness, and better robustness for the attacks of Gaussian noises, salt and pepper noises, JPEG compression, cropping, filtering and image enhancement than the traditional QIM algorithm.
Asymmetric multiple-image encryption based on the cascaded fractional Fourier transform
Li, Yanbin; Zhang, Feng; Li, Yuanchao; Tao, Ran
2015-09-01
A multiple-image cryptosystem is proposed based on the cascaded fractional Fourier transform. During an encryption procedure, each of the original images is directly separated into two phase masks. A portion of the masks is subsequently modulated into an interim mask, which is encrypted into the ciphertext image; the others are used as the encryption keys. Using phase truncation in the fractional Fourier domain, one can use an asymmetric cryptosystem to produce a real-valued noise-like ciphertext, while a legal user can reconstruct all of the original images using a different group of phase masks. The encryption key is an indivisible part of the corresponding original image and is still useful during decryption. The proposed system has high resistance to various potential attacks, including the chosen-plaintext attack. Numerical simulations also demonstrate the security and feasibility of the proposed scheme.
A portable Fourier transform infrared gas analyzer with a photoacoustic detector performed reliably during pollution prevention research at two industrial facilities. It exhibited good agreement (within approximately 6%) with other analytical instruments (dispersive infrared and ...
Institute of Scientific and Technical Information of China (English)
FENG; Shaotong; HAN; Dianrong; DING; Heping
2004-01-01
By means of experimental technique of optical fractional Fourier transform,we have determined the Hurst exponent of a regular self-affine fractal pattern to demonstrate the feasibility of this approach. Then we extend this method to determine the Hurst exponents of some irregular self-affine fractal patterns. Experimental results show that optical fractional Fourier transform is a practical method for analyzing the self-affine fractal patterns.
Fedorenko, Sergei V
2011-01-01
A novel method for computation of the discrete Fourier transform over a finite field with reduced multiplicative complexity is described. If the number of multiplications is to be minimized, then the novel method for the finite field of even extension degree is the best known method of the discrete Fourier transform computation. A constructive method of constructing for a cyclic convolution over a finite field is introduced.
DEFF Research Database (Denmark)
Guan, Pengyu; Røge, Kasper Meldgaard; Morioka, Toshio;
2016-01-01
We review recent progress in the use of time lens based optical Fourier transformation for advanced optical signal processing, with focus on all-optical generation, detection and format conversion of spectrally-efficient OFDM and N-WDM signals.......We review recent progress in the use of time lens based optical Fourier transformation for advanced optical signal processing, with focus on all-optical generation, detection and format conversion of spectrally-efficient OFDM and N-WDM signals....
Kamalian, Morteza; Prilepsky, Jaroslaw E; Le, Son Thai; Turitsyn, Sergei K
2016-08-01
In this work, we introduce the periodic nonlinear Fourier transform (PNFT) method as an alternative and efficacious tool for compensation of the nonlinear transmission effects in optical fiber links. In the Part I, we introduce the algorithmic platform of the technique, describing in details the direct and inverse PNFT operations, also known as the inverse scattering transform for periodic (in time variable) nonlinear Schrödinger equation (NLSE). We pay a special attention to explaining the potential advantages of the PNFT-based processing over the previously studied nonlinear Fourier transform (NFT) based methods. Further, we elucidate the issue of the numerical PNFT computation: we compare the performance of four known numerical methods applicable for the calculation of nonlinear spectral data (the direct PNFT), in particular, taking the main spectrum (utilized further in Part II for the modulation and transmission) associated with some simple example waveforms as the quality indicator for each method. We show that the Ablowitz-Ladik discretization approach for the direct PNFT provides the best performance in terms of the accuracy and computational time consumption. PMID:27505799
Zhang, Leihong; Liang, Dong; Li, Bei; Kang, Yi; Pan, Zilan; Zhang, Dawei; Gao, Xiumin; Ma, Xiuhua
2016-07-01
On the basis of analyzing the cosine light field with determined analytic expression and the pseudo-inverse method, the object is illuminated by a presetting light field with a determined discrete Fourier transform measurement matrix, and the object image is reconstructed by the pseudo-inverse method. The analytic expression of the algorithm of computational ghost imaging based on discrete Fourier transform measurement matrix is deduced theoretically, and compared with the algorithm of compressive computational ghost imaging based on random measurement matrix. The reconstruction process and the reconstruction error are analyzed. On this basis, the simulation is done to verify the theoretical analysis. When the sampling measurement number is similar to the number of object pixel, the rank of discrete Fourier transform matrix is the same as the one of the random measurement matrix, the PSNR of the reconstruction image of FGI algorithm and PGI algorithm are similar, the reconstruction error of the traditional CGI algorithm is lower than that of reconstruction image based on FGI algorithm and PGI algorithm. As the decreasing of the number of sampling measurement, the PSNR of reconstruction image based on FGI algorithm decreases slowly, and the PSNR of reconstruction image based on PGI algorithm and CGI algorithm decreases sharply. The reconstruction time of FGI algorithm is lower than that of other algorithms and is not affected by the number of sampling measurement. The FGI algorithm can effectively filter out the random white noise through a low-pass filter and realize the reconstruction denoising which has a higher denoising capability than that of the CGI algorithm. The FGI algorithm can improve the reconstruction accuracy and the reconstruction speed of computational ghost imaging.
A hybrid algorithm for the rapid Fourier transform of extensive series of data
Directory of Open Access Journals (Sweden)
A. S Franco
1971-12-01
Full Text Available A technique is described for the rapid Fourier transform of large series of numbers. The technique takes advantage of the fact that most digital series are highly factorizable by the number 2, which permits the use of the F.F.T. algorithm. Using two magnetic tape units, or alternatively magnetic disk facilities, very large series can be transformed efficiently with only modest computer facilities. For the transformation of odd-valued series the Thomas Prime-Factor and Gentleman and Sande algorithms are treated in detail.Apresenta-se neste trabalho uma técnica de transformação rápida de Fourier aplicada a uma longa série de valores numéricos. A técnica tira partido do fato de que a grande maioria das séries digitalizadas é, em geral, suscetível de fatoração onde aparece frequentemente o fator 2, o que permite o emprego do algorítmo da transformação rápida de Fourier (F.F.T.. Com o emprego de duas fitas magnéticas ou discos, pode ser efetuada eficientemente a transformação de longas séries em computadores de modesta memória. O algorítmo de fatores primos de Thomas e o de Gentleman e Sande são, respectivamente, tratados em detalhe, na transformação de séries com numero ímpar de valores.
Galizzi, Gustavo E.; Cuadrado-Laborde, Christian
2015-10-01
In this work we study the joint transform correlator setup, finding two analytical expressions for the extensions of the joint power spectrum and its inverse Fourier transform. We found that an optimum efficiency is reached, when the bandwidth of the key code is equal to the sum of the bandwidths of the image plus the random phase mask (RPM). The quality of the decryption is also affected by the ratio between the bandwidths of the RPM and the input image, being better as this ratio increases. In addition, the effect on the decrypted image when the detection area is lower than the encrypted signal extension was analyzed. We illustrate these results through several numerical examples.
Indian Academy of Sciences (India)
Shishir B Sahay; T Meghasyam; Rahul K Roy; Gaurav Pooniwala; Sasank Chilamkurthy; Vikram Gadre
2015-06-01
This paper is targeted towards a general readership in signal processing. It intends to provide a brief tutorial exposure to the Fractional Fourier Transform, followed by a report on experiments performed by the authors on a Generalized Time Frequency Transform (GTFT) proposed by them in an earlier paper. The paper also discusses the extension of the uncertainty principle to the GTFT. This paper discusses some analytical results of the GTFT. We identify the eigenfunctions and eigenvalues of the GTFT. The time shift property of the GTFT is discussed. The paper describes methods for estimation of parameters of individual chirp signals on receipt of a noisy mixture of chirps. A priori knowledge of the nature of chirp signals in the mixture – linear or quadratic is required, as the two proposed methods fall in the category of model-dependent methods for chirp parameter estimation.
Strehl ratio as the Fourier transform of a probability density of error differences.
Alonso, Miguel A; Forbes, G W
2016-08-15
To give useful insight into the impact of mid-spatial frequency structure on optical performance, the Strehl ratio is shown to correspond to the Fourier transform of a simple statistical characterization of the aberration in the exit pupil. This statistical description is found simply by autocorrelating a histogram of the aberration values. In practice, the histogram itself can often be approximated by a convolution of underlying histograms associated with fabrication steps and, together with the final autocorrelation, it follows from the central limit theorem that the Strehl ratio as a function of the scale of the phase error is generally approximated well by a Gaussian. PMID:27519076
Wave scattering theory a series approach based on the Fourier transformation
Eom, Hyo J
2001-01-01
The book provides a unified technique of Fourier transform to solve the wave scattering, diffraction, penetration, and radiation problems where the technique of separation of variables is applicable. The book discusses wave scattering from waveguide discontinuities, various apertures, and coupling structures, often encountered in electromagnetic, electrostatic, magnetostatic, and acoustic problems. A system of simultaneous equations for the modal coefficients is formulated and the rapidly-convergent series solutions amenable to numerical computation are presented. The series solutions find practical applications in the design of microwave/acoustic transmission lines, waveguide filters, antennas, and electromagnetic interference/compatibilty-related problems.
Osowiecki, Gaël. D.; Madi, Mohammad; Shorubalko, Ivan; Philipoussis, Irène; Alberti, Edoardo; Scharf, Toralf; Herzig, Hans P.
2015-09-01
We show the miniaturization and parallelization of a scanning standing wave spectrometer with a long term goal of creating a compact imaging spectrometer. In our standing wave integrated Fourier transform spectrometer, light is injected with micro-lenses into several optical polymer waveguides. A piezo actuated mirror located at the waveguide end-facet can shift the interferogram to increase its sampling frequency. The spatial distribution of the standing wave intensity inside the waveguide is partially scattered out of the plane by a periodic metallic grating and recorded by a CCD camera. We present spectra acquisition for six adjacent waveguides simultaneously at a wavelength of 632.8 nm.
An analog of the Fourier transform associated with a nonlinear one-dimensional Schroedinger equation
International Nuclear Information System (INIS)
We consider an eigenvalue problem which includes a nonlinear Schroedinger equation on the half-line [0, ∞) and certain boundary conditions. It is shown that the spectrum of this problem fills a half-line and that to each point of the spectrum there corresponds a unique eigenfunction. The main result of the paper is that an arbitrary infinitely differentiable function g(x) rapidly decaying as x → ∞ and satisfying suitable boundary conditions at the point x = 0 can be uniquely expanded into an integral over eigenfunctions similar to the representation of functions by the Fourier transform (the latter is obviously associated with a linear self-adjoint eigenvalue problem)
DEFF Research Database (Denmark)
Bruun, Susanne Wrang; Kohler, Achim; Adt, Isabelle;
2006-01-01
Fourier transform infrared (FT-IR) spectroscopy is a valuable technique for characterization of biological samples, providing a detailed fingerprint of the major chemical constituents. However, water vapor and CO(2) in the beam path often cause interferences in the spectra, which can hamper...... an absorption band from either water vapor or CO(2). From two calibration data sets, gas model spectra were estimated in each of the four spectral regions, and these model spectra were applied for correction of gas absorptions in two independent test sets (spectra of aqueous solutions and a yeast biofilm (C...
INVERSE CORRECTION OF FOURIER TRANSFORMS FOR ONE-DIMENSIONAL STRONGLY SCATTERING APERIODIC LATTICES
Directory of Open Access Journals (Sweden)
Y. F. Hsin
2016-05-01
Full Text Available The accuracy of the Fourier transform (FT, advantageous for the aperiodic lattice (AL design, is significantly improved for strongly scattering periodic lattices (PLs and ALs. The approach is to inversely obtain corrected parameters from an accurate transfer matrix method for the FT. We establish a corrected FT in order to improve the spectral inaccuracy of strongly scattering PLs by redefining wave numbers and reflective intensity. We further correct the FT for strongly scattering ALs by implementing improvements applied to strongly scattering PLs and then making detailed wave number adjustments in the main band spectral region. Silicon lattice simulations are presented.
Application of micro-Fourier transform infrared spectroscopy to the examination of paint samples
Zięba-Palus, J.
1999-11-01
The examination and identification of automobile paints is an important problem in road accidents investigations. Since the real sample available is very small, only sensitive microtechniques can be applied. The methods of optical microscopy and micro-Fourier transform infrared spectroscopy (MK-FTIR) supported by scanning electron microscopy together with X-ray microanalysis (SEM-EDX) allow one to carry out the examination of each paint layer without any separation procedure. In this paper an attempt is made to discriminate between different automobile paints of the same colour by the use of these methods for criminalistic investigations.
Simple atmospheric transmittance calculation based on a Fourier-transformed Voigt profile.
Kobayashi, Hirokazu
2002-11-20
A method of line-by-line transmission calculation for a homogeneous atmospheric layer that uses the Fourier-transformed Voigt profile is presented. The method is based on a pure Voigt function with no approximation and an interference term that takes into account the line-mixing effect. One can use the method to calculate transmittance, considering each line shape as it is affected by temperature and pressure, with a line database with an arbitrary wave-number range and resolution. To show that the method is feasible for practical model development, we compared the calculated transmittance with that obtained with a conventional model, and good consistency was observed. PMID:12463237
Townley-Smith, Keeley; Nave, Gillian; Pickering, Juliet C.; Blackwell-Whitehead, Richard J.
2016-09-01
We expand on the comprehensive study of hyperfine structure (HFS) in Mn II conducted by Holt et al. (1999) by verifying hyperfine magnetic dipole constants (A) for 20 levels previously measured by Holt et al. (1999) and deriving A constants for 47 previously unstudied levels. The HFS patterns were measured in archival spectra from Fourier transform (FT) spectrometers at Imperial College London and the National Institute of Standards and Technology. Analysis of the FT spectra was carried out in XGREMLIN. Our A constant for the ground level has a lower uncertainty by a factor of 6 than that of Blackwell-Whitehead et al.
Fourier transform acousto-optic imaging with a custom-designed CMOS smart-pixels array.
Barjean, Kinia; Contreras, Kevin; Laudereau, Jean-Baptiste; Tinet, Éric; Ettori, Dominique; Ramaz, François; Tualle, Jean-Michel
2015-03-01
We report acousto-optic imaging (AOI) into a scattering medium using a Fourier Transform (FT) analysis to achieve axial resolution. The measurement system was implemented using a CMOS smart-pixels sensor dedicated to the real-time analysis of speckle patterns. This first proof-of-principle of FT-AOI demonstrates some of its potential advantages, with a signal-to-noise ratio comparable to the one obtained without axial resolution, and with an acquisition rate compatible with a use on living biological tissue.
Hettich, Robert L.; Jin, Changming; Compton, Robert N.; Buseck, Peter R.; Tsipursky, Semeon J.
1993-10-01
Fourier transform mass spectrometry (FTMS) employing both laser desorption/ionization and thermal desorption/electron ionization is useful for the detection and structural characterization of fullerenes and chemically-modified fullerenes. Examination of a carbon-rich shungite rock sample from Russia by transmission electron microscopy and FTMS provided evidence of naturally-occurring fullerenes. Ion-molecule reactions can be studied with FTMS to investigate the electron affinities of modified fullerenes. By monitoring charge exchange reactions, the electron affinities of C60Fx (x=44,46) and C70Fy (y=52,54) were found to be substantially higher than the values for the parent fullerenes.
Parallel 3-dim fast Fourier transforms with load balancing of the plane waves
Gao, Xingyu; Fang, Jun; Wang, Han
2016-01-01
The plane wave method is most widely used for solving the Kohn-Sham equations in first-principles materials science computations. In this procedure, the three-dimensional (3-dim) trial wave functions' fast Fourier transform (FFT) is a regular operation and one of the most demanding algorithms in terms of the scalability on a parallel machine. We propose a new partitioning algorithm for the 3-dim FFT grid to accomplish the trade-off between the communication overhead and load balancing of the plane waves. It is shown by qualitative analysis and numerical results that our approach could scale the plane wave first-principles calculations up to more nodes.
Mastering the discrete Fourier transform in one, two or several dimensions pitfalls and artifacts
Amidror, Isaac
2013-01-01
The discrete Fourier transform (DFT) is an extremely useful tool that finds application in many different disciplines. However, its use requires caution. The aim of this book is to explain the DFT and its various artifacts and pitfalls and to show how to avoid these (whenever possible), or at least how to recognize them in order to avoid misinterpretations. This concentrated treatment of the DFT artifacts and pitfalls in a single volume is, indeed, new, and it makes this book a valuable source of information for the widest possible range of DFT users. Special attention is given to the one and
Radar cross-section reduction based on an iterative fast Fourier transform optimized metasurface
Song, Yi-Chuan; Ding, Jun; Guo, Chen-Jiang; Ren, Yu-Hui; Zhang, Jia-Kai
2016-07-01
A novel polarization insensitive metasurface with over 25 dB monostatic radar cross-section (RCS) reduction is introduced. The proposed metasurface is comprised of carefully arranged unit cells with spatially varied dimension, which enables approximate uniform diffusion of incoming electromagnetic (EM) energy and reduces the threat from bistatic radar system. An iterative fast Fourier transform (FFT) method for conventional antenna array pattern synthesis is innovatively applied to find the best unit cell geometry parameter arrangement. Finally, a metasurface sample is fabricated and tested to validate RCS reduction behavior predicted by full wave simulation software Ansys HFSSTM and marvelous agreement is observed.
Pengjuan Liang; Hao Wang; Chaoyin Chen; Feng Ge; Diqiu Liu; Shiqi Li; Benyong Han; Xianfeng Xiong; Shenglan Zhao
2013-01-01
Currently, the authentication of virgin walnut oil (VWO) has become very important due to the possible adulteration of VWO with cheaper plant oils such as soybean oil (SO), puer tea seed oil (PO), and sunflower oil (SFO). Methods involving Fourier transform infrared (FT-IR) spectroscopy combined with chemometric techniques (partial least square) were developed for quantification of SO, PO, and SFO in VWO. IR spectra of oil samples were recorded at frequency regions of 4000–650 cm−1 on horizon...
Beecken, Brian P.; Kleinman, Randall R.
2004-01-01
New developments in infrared sensor technology have potentially made possible a new space-based system which can measure far-infrared radiation at lower costs (mass, power and expense). The Stationary Imaging Fourier Transform Spectrometer (SIFTS) proposed by NASA Langley Research Center, makes use of new detector array technology. A mathematical model which simulates resolution and spectral range relationships has been developed for analyzing the utility of such a radically new approach to spectroscopy. Calculations with this forward model emulate the effects of a detector array on the ability to retrieve accurate spectral features. Initial computations indicate significant attenuation at high wavenumbers.
Parsons, C. L.; Gerlach, J. C.; Whitehurst, M.
1982-01-01
The development of a prototype, ground-based, Sun-pointed Michelson interferometric spectrometer is described. Its intended use is to measure the atmospheric amount of various gases which absorb in the near-infrared, visible, and near-ultraviolet portions of the electromagnetic spectrum. Preliminary spectra which contain the alpha, 0.8 micrometer, and rho sigma tau water vapor absorption bands in the near-infrared are presented to indicate the present capability of the system. Ultimately, the spectrometer can be used to explore the feasible applications of Fourier transform spectroscopy in the ultraviolet where grating spectrometers were used exclusively.
Analysis of the Interference Modulation Depth in the Fourier Transform Spectrometer
Directory of Open Access Journals (Sweden)
Rilong Liu
2015-01-01
Full Text Available Based on the principle of the Michelson interferometer, the paper briefly describes the theoretical significance and calculates and deduces three expressions of the interference modulation depth. The influence of the surface shape error of plane mirror on modulation depth is analyzed, and the tolerance of error is also pointed out. Moreover, the dependence of modulation depth on the reflectance change of beam splitter interface is also analyzed, and the curve is given. It is concluded that this paper is of general significance for the Fourier transform spectrometer based on the principle of the Michelson two-beam interference.
Zheng, Ying; Liang, Jingqiu; Liang, Zhongzhu
2013-01-14
A model of miniaturized space-modulated Fourier transform infrared spectrometer (FTIR) is given. The two step mirrors as the key components are designed and a lithography-electroplating technique used to fabricate the small step mirror is proposed. We analyze the effect of the experiment results resulted from fabricating technics on the recovery spectrum in theory, and demonstrate that the lithography-electroplating technique is an effective method to fabricate the step mirror, which make miniaturized FTIR realized. We believe that the performances of FTIR can be better realized by optimizing experimental conditions to make this fabricating method more attractive.
Chelliah, Pandian; Sahoo, Trilochan; Singh, Sheela; Sujatha, Annie
2015-10-20
A Fourier transform spectrometer (FTS) used for interrogating a fiber Bragg grating (FBG) consists of a scanning-type interferometer. The FTS has a broad wavelength range of operation and good multiplexing capability. However, it has poor wavelength resolution and interrogation speed. We propose a modification to the FTS using path delay multiplexing to improve the same. Using this method, spatial resolution and interrogation time can be improved by n times by using n path delays. In this paper, simulation results for n=2, 5 are shown.
Planetary infrared astronomy using a cryogenic postdisperser on Fourier transform spectrometers
Jennings, D. E.; Kunde, V. G.; Hanel, R. A.; Maguire, W. C.; Lamb, G. M.
1986-01-01
High resolution infrared spectra of planets from ground-based observatories were analyzed and instrumentation to improve sensitivity was developed. A cryogenic postdisperser (a narrow bandpass spectral filter) for use with Fourier transform spectrometers (FTS's) at facility observatories was constructed. This instrument has improved the sensitivity of FTS observations at 8 to 20 microns by about an order of magnitude. Spectra of Jupiter, Saturn and Comet Halley were obtained using the postdisperser with FTS facilities at the Kitt Peak 4-meter and McMath telescopes. Spectral resolution as high as 0.01/cm was achieved.
The optimum scheme of a static Fourier-transform spectrometer based on birefringent crystal
Institute of Scientific and Technical Information of China (English)
Dongqing Zhang(张冬青); Fuquan Wu(吴福全); Shuhai Fan(范树海)
2003-01-01
An optimum design of static Fourier-transform spectrometer based on Savert prisms is presented in this paper. A new method of increasing path difference and resolution of spectrometer is given. When the angle between the crystal optical axis of the first Savert prism and the incident interface is 58° and the angle between the crystal optical axis of the second Savert prism and the incident interface is 28°, the maximum path difference will be 0.63 mm, the maximum resolution will be 15.8 cm-1, and the whole field-of-view will reach 6°.
Côté, Alex; Levasseur, Simon; Boudreau, Sylvain; Genest, Jérôme
2014-07-01
This paper presents a concept using field cameras in combination with Fourier transform spectrometers. The device can produce five-dimensional (position-angle-spectra) data cubes. This can lead to accurate measurements in both spectrum and distance and allows a thorough characterization of the interferometer, as well as adds passive ranging information to hyperspectral images. Shear and tilt fringes are simultaneously observed in a fixed optical path difference interferometer, and a passive spectral ranging demonstration is done in both absorption and emission for the 500-900 nm spectral bands.
Design of spatio-temporally modulated static infrared imaging Fourier transform spectrometer.
Wang, WenCong; Liang, JingQiu; Liang, ZhongZhu; Lü, JinGuang; Qin, YuXin; Tian, Chao; Wang, WeiBiao
2014-08-15
A novel static medium wave infrared (MWIR) imaging Fourier transform spectrometer (IFTS) is conceptually proposed and experimentally demonstrated. In this system, the moving mirror in traditional temporally modulated IFTS is replaced by multi-step micro-mirrors to realize the static design. Compared with the traditional spatially modulated IFTS, they have no slit system and are superior with larger luminous flux and higher energy efficiency. The use of the multi-step micro-mirrors can also make the system compact and light.
Coherent coupling of magneto-excitons probed by two-dimensional Fourier transform spectroscopy
Paul, Jagannath; Liu, Cunming; McGill, Stephen; Hilton, David; Karaiskaj, Denis
We present the coherent two dimensional Fourier Transform (2DFT) spectra of magneto-excitons in undoped GaAs quantum wells at high magnetic field up to 10 Tesla. The 2DFT data reveal strong coherent coupling between resonances and line shapes which are strikingly different from the zero field spectra. 2DFT spectra measured using co-linear and co-circular polarizations at low temperatures will be discussed. The work at USF and UAB was supported by the National Science Foundation under Grant Number DMR-1409473. The work at NHMFL, Florida State University was supported by the National Science Foundation under Grant Numbers DMR-1157490 and DMR-1229217.
Directory of Open Access Journals (Sweden)
Venedikt A. Kapitanov
2010-01-01
Full Text Available Measurements of ethylene absorption spectra with Fourier Transform (FT and Photoacoustic (PA spectrometers within 6035–6210 cm−1 are described. The methodology used for building the frequency scale for both spectrometers is presented. The methane absorption spectrum, included into the HITRAN database, was used in both cases to calibrate the frequency scale. Ethylene absorption spectra were obtained with the two recording methods; a coincidence of the measured line center positions was obtained with an accuracy of 0.0005 cm−1.
A highly linear superconducting bolometer for quantitative THz Fourier transform spectroscopy.
Kehrt, Mathias; Monte, Christian; Beyer, Jörn; Hollandt, Jörg
2015-05-01
A superconducting transition edge sensor (TES) bolometer operating in the spectral range from 0.1 THz to 3 THz was designed. It is especially intended for Fourier transform spectroscopy and features a higher dynamic range and a highly linear response at a similar response compared to commercially available silicon composite bolometers. The design is based on a thin film metal mesh absorber, a superconducting thermistor and Si3N4 membrane technology. A prototype was set up, characterized and successfully used in first applications. PMID:25969213
Institute of Scientific and Technical Information of China (English)
无
2011-01-01
Real time digital recording and numerical reconstruction of a temporal digital hologram sequence have become feasible in recent years.They provide a new measurement method which enjoys the valuable advantages of being full-field,noncontact and high precision.In this paper,a combined method of temporal digital hologram sequence and windowed Fourier transform is proposed to measure the kinematic parameters of random vibration.A series of holograms are recorded by CCD camera and the original phase can be recon...
Applications of asynoptic space - Time Fourier transform methods to scanning satellite measurements
Lait, Leslie R.; Stanford, John L.
1988-01-01
A method proposed by Salby (1982) for computing the zonal space-time Fourier transform of asynoptically acquired satellite data is discussed. The method and its relationship to other techniques are briefly described, and possible problems in applying it to real data are outlined. Examples of results obtained using this technique are given which demonstrate its sensitivity to small-amplitude signals. A number of waves are found which have previously been observed as well as two not heretofore reported. A possible extension of the method which could increase temporal and longitudinal resolution is described.
Opto-digital image encryption by using Baker mapping and 1-D fractional Fourier transform
Liu, Zhengjun; Li, She; Liu, Wei; Liu, Shutian
2013-03-01
We present an optical encryption method based on the Baker mapping in one-dimensional fractional Fourier transform (1D FrFT) domains. A thin cylinder lens is controlled by computer for implementing 1D FrFT at horizontal direction or vertical direction. The Baker mapping is introduced to scramble the amplitude distribution of complex function. The amplitude and phase of the output of encryption system are regarded as encrypted image and key. Numerical simulation has been performed for testing the validity of this encryption scheme.
Reconstruction of piecewise homogeneous images from partial knowledge of their Fourier Transform
Féron, Olivier; Chama, Zouaoui; Mohammad-Djafari, Ali
2004-11-01
Fourier synthesis (FS) inverse problem consists in reconstructing a multi-variable function from the measured data which correspond to partial and uncertain knowledge of its Fourier Transform (FT). By partial knowledge we mean either partial support and/or the knowledge of only the module and by uncertain we mean both uncertainty of the model and noisy data. This inverse problem arises in many applications such as : optical imaging, radio astronomy, magnetic resonance imaging (MRI) and diffraction scattering (ultrasounds or microwave imaging). Most classical methods of inversion are based on interpolation of the data and fast inverse FT. But when the data do not fill uniformly the Fourier domain or when the phase of the signal is lacking as in optical interferometry, the results obtained by such methods are not satisfactory, because these inverse problems are ill-posed. The Bayesian estimation approach, via an appropriate modeling of the unknown functions gives the possibility of compensating the lack of information in the data, thus giving satisfactory results. In this paper we study the case where the observations are a part of the FT modulus of objects which are composed of a few number of homogeneous materials. To model such objects we use a Hierarchical Hidden Markov Modeling (HMM) and propose a Bayesian inversion method using appropriate Markov Chain Monte Carlo (MCMC) algorithms.
Institute of Scientific and Technical Information of China (English)
Liu Zhi-Ming; Liu Wen-qing; Gao Ming-Guang; Tong Jing-Jing; Zhang Wian-Shu; Xu Liang; Wei Xiuai
2008-01-01
Passive Fourier transform infrared (FTIR) remote sensing measurement of chemical gas cloud is a vital technology.It takes an important part in many fields for the detection of released gases.The principle of concentration measurement is based on the Beer-Lambert law.Unlike the active measurement,for the passive remote sensing,in most cases,the difference between the temperature of the gas cloud and the brightness temperature of the background is usually a few kelvins.The gas cloud emission is almost equal to the background emission,thereby the emission of the gas cloud cannot be ignored.The concentration retrieval algorithm is quite different from the active measurement.In this paper,the concentration retrieval algorithm for the passive FTIR remote measurement of gas cloud is presented in detail,which involves radiative transfer model,radiometric calibration,absorption coefficient calculation,et al.The background spectrum has a broad feature,which is a slowly varying function of frequency.In this paper,the background spectrum is fitted with a polynomial by using the Levenberg-Marquardt method which is a kind of nonlinear least squares fitting algorithm.No background spectra are required.Thus,this method allows mobile,real-time and fast measurements of gas clouds.
Intermittent Clutter Removal Approac for wind Profiler Data Based on Fractional Fourier Transform
Institute of Scientific and Technical Information of China (English)
YUAN Lei; AN Jian-ping; BU Xiang-yuan
2009-01-01
Through the analysis of the target characteristics and according to the intermittent clutter bursting and short duration characteristics, a new method for the clutter recognition based on the fractional Fourier transform (FRFT) is proposed. This method is predicated on the fact that the FRFT perfectly localizes a chirp signal as an impulse when the angle parameter of the transform matches the chirp rate of the chirp signal. The method involves detecting the presence of the intermittent clutter and correctly estimating its orientation in the time-frequency plane, removing the intermittent clutter in the fractional domain, and completing wind estimation by the power spectrum. By testing the artificial WPR-like signal and data measured from the field, we verify that the FRFT-based method is very effective.
High-resolution Fourier-transform spectrometer chip with microphotonic silicon spiral waveguides.
Velasco, Aitor V; Cheben, Pavel; Bock, Przemek J; Delâge, André; Schmid, Jens H; Lapointe, Jean; Janz, Siegfried; Calvo, María L; Xu, Dan-Xia; Florjańczyk, Mirosław; Vachon, Martin
2013-03-01
We report a stationary Fourier-transform spectrometer chip implemented in silicon microphotonic waveguides. The device comprises an array of 32 Mach-Zehnder interferometers (MZIs) with linearly increasing optical path delays between the MZI arms across the array. The optical delays are achieved by using Si-wire waveguides arranged in tightly coiled spirals with a compact device footprint of 12 mm2. Spectral retrieval is demonstrated in a single measurement of the stationary spatial interferogram formed at the output waveguides of the array, with a wavelength resolution of 40 pm within a free spectral range of 0.75 nm. The phase and amplitude errors arising from fabrication imperfections are compensated using a transformation matrix spectral retrieval algorithm.
The quantum state vector in phase space and Gabor's windowed Fourier transform
Bracken, A J
2010-01-01
Representations of quantum state vectors by complex phase space amplitudes, complementing the description of the density operator by the Wigner function, have been defined by applying the Weyl-Wigner transform to dyadic operators, linear in the state vector and anti-linear in a fixed `window state vector'. Here aspects of this construction are explored, with emphasis on the connection with Gabor's `windowed Fourier transform'. The amplitudes that arise for simple quantum states from various choices of window are presented as illustrations. Generalized Bargmann representations of the state vector appear as special cases, associated with Gaussian windows. For every choice of window, amplitudes lie in a corresponding linear subspace of square-integrable functions on phase space. A generalized Born interpretation of amplitudes is described, with both the Wigner function and a generalized Husimi function appearing as quantities linear in an amplitude and anti-linear in its complex conjugate. Schr\\"odinger's time-d...
Tensor representation of color images and fast 2D quaternion discrete Fourier transform
Grigoryan, Artyom M.; Agaian, Sos S.
2015-03-01
In this paper, a general, efficient, split algorithm to compute the two-dimensional quaternion discrete Fourier transform (2-D QDFT), by using the special partitioning in the frequency domain, is introduced. The partition determines an effective transformation, or color image representation in the form of 1-D quaternion signals which allow for splitting the N × M-point 2-D QDFT into a set of 1-D QDFTs. Comparative estimates revealing the efficiency of the proposed algorithms with respect to the known ones are given. In particular, a proposed method of calculating the 2r × 2r -point 2-D QDFT uses 18N2 less multiplications than the well-known column-row method and method of calculation based on the symplectic decomposition. The proposed algorithm is simple to apply and design, which makes it very practical in color image processing in the frequency domain.
Foliage Plant Retrieval Using Polar Fourier Transform, Color Moments and Vein Features
Directory of Open Access Journals (Sweden)
Abdul Kadir
2011-09-01
Full Text Available This paper proposed a method that combines Polar Fo urier Transform, color moments, and vein features to retrieve leaf images based on a leaf image. The method is very useful to help people in recognizing foliage plants. Foliage plants are plants that have various colors and unique patterns in the leaf. Th erefore, the colors and its patterns are information that sh ould be counted on in the processing of plant identification. To compare the performance of retri eving system to other result, the experiments used Flavia dataset, which is very popular in recognizin g plants. The result shows that the method gave bet ter performance than PNN, SVM, and Fourier Transform. T he method was also tested using foliage plants with various colors. The accuracy was 90.80% for 50 kinds of plants.
International Nuclear Information System (INIS)
The Integral Fourier Transform has a large range of applications in such areas as communication theory, circuit theory, physics, etc. In order to perform discrete Fourier Transform the Finite Fourier Transform is defined; it operates upon N samples of a uniformely sampled continuous function. All the properties known in the continuous case can be found in the discrete case also. The first part of the paper presents the relationship between the Finite Fourier Transform and the Integral one. The computing of a Finite Fourier Transform is a problem in itself since in order to transform a set of N data we have to perform N2 ''operations'' if the transformation relations are used directly. An algorithm known as the Fast Fourier Transform (FFT) reduces this figure from N2 to a more reasonable Nlog2N, when N is a power of two. The original Cooley and Tuckey algorithm for FFT can be further improved when higher basis are used. The price to be paid in this case is the increase in complexity of such algorithms. The recurrence relations and a comparation among such algorithms are presented. The key point in understanding the application of FFT resides in the convolution theorem which states that the convolution (an N2 type procedure) of the primitive functions is equivalent to the ordinar multiplication of their transforms. Since filtering is actually a convolution process we present several procedures to perform digital filtering by means of FFT. The best is the one using the segmentation of records and the transformation of pairs of records. In the digital processing of signals, besides digital filtering a special attention is paid to the estimation of various statistical characteristics of a signal as: autocorrelation and correlation functions, periodiograms, density power sepctrum, etc. We give several algorithms for the consistent and unbiased estimation of such functions, by means of FFT. (author)
The design of ROM-type holographic memory with iterative Fourier transform algorithm
Akamatsu, Hideki; Yamada, Kai; Unno, Noriyuki; Yoshida, Shuhei; Taniguchi, Jun; Yamamoto, Manabu
2013-03-01
The research and development of the holographic data storage (HDS) is advanced, as one of the high-speed, mass storage systems of the next generation. Recently, along the development of the write-once system that uses photopolymer media, large capacity ROM type HDS which can replace conventional optical discs becomes important. In this study, we develop the ROM type HDS using a diffractive optical element (DOE), and verify the effectiveness of our approach. In order to design DOE, iterative Fourier transform algorithm was adopted, and DOE is fabricated with electron beam (EB) cutting and nanoimprint lithography. We optimize the phase distribution of the hologram by iterative Fourier transform algorithm known as Gerchberg-Saxton (GS) algorithm with the angular spectrum method. In the fabrication process, the phase distribution of the hologram is implicated as the concavity and convexity structure by the EB cutting and transcribed with nanoimprint lithography. At this time, the mold is formed as multiple-stage concavity and convexity. The purpose of multiple-stage concavity and convexity is to obtain high diffraction efficiency and signal-to-noise ratio (SNR). Fabricated trial model DOE is evaluated by the experiment.
The data processing pipeline for the Herschel SPIRE Fourier Transform Spectrometer
Fulton, T.; Naylor, D. A.; Polehampton, E. T.; Valtchanov, I.; Hopwood, R.; Lu, N.; Baluteau, J.-P.; Mainetti, G.; Pearson, C.; Papageorgiou, A.; Guest, S.; Zhang, L.; Imhof, P.; Swinyard, B. M.; Griffin, M. J.; Lim, T. L.
2016-05-01
We present the data processing pipeline to generate calibrated data products from the Spectral and Photometric Imaging Receiver (SPIRE) imaging Fourier Transform Spectrometer on the Herschel Space Observatory. The pipeline processes telemetry from SPIRE observations and produces calibrated spectra for all resolution modes. The spectrometer pipeline shares some elements with the SPIRE photometer pipeline, including the conversion of telemetry packets into data timelines and calculation of bolometer voltages. We present the following fundamental processing steps unique to the spectrometer: temporal and spatial interpolation of the scan mechanism and detector data to create interferograms; Fourier transformation; apodization; and creation of a data cube. We also describe the corrections for various instrumental effects including first- and second-level glitch identification and removal, correction of the effects due to emission from the Herschel telescope and from within the spectrometer instrument, interferogram baseline correction, temporal and spatial phase correction, non-linear response of the bolometers, and variation of instrument performance across the focal plane arrays. Astronomical calibration is based on combinations of observations of standard astronomical sources and regions of space known to contain minimal emission.
Source brightness fluctuation correction of solar absorption fourier transform mid infrared spectra
Ridder, T.; Warneke, T.; Notholt, J.
2011-06-01
The precision and accuracy of trace gas observations using solar absorption Fourier Transform infrared spectrometry depend on the stability of the light source. Fluctuations in the source brightness, however, cannot always be avoided. Current correction schemes, which calculate a corrected interferogram as the ratio of the raw DC interferogram and a smoothed DC interferogram, are applicable only to near infrared measurements. Spectra in the mid infrared spectral region below 2000 cm-1 are generally considered uncorrectable, if they are measured with a MCT detector. Such measurements introduce an unknown offset to MCT interferograms, which prevents the established source brightness fluctuation correction. This problem can be overcome by a determination of the offset using the modulation efficiency of the instrument. With known modulation efficiency the offset can be calculated, and the source brightness correction can be performed on the basis of offset-corrected interferograms. We present a source brightness fluctuation correction method which performs the smoothing of the raw DC interferogram in the interferogram domain by an application of a running mean instead of high-pass filtering the corresponding spectrum after Fourier transformation of the raw DC interferogram. This smoothing can be performed with the onboard software of commercial instruments. The improvement of MCT spectra and subsequent ozone profile and total column retrievals is demonstrated. Application to InSb interferograms in the near infrared spectral region proves the equivalence with the established correction scheme.
Wang, Guangjun; Wang, Huaying; Wang, Dayong; Xie, Jianjun; Zhao, Jie
2007-12-01
A simple holographic high-resolution imaging system without pre-magnification, which is based on off-axis lensless Fourier transform configuration, has been developed. Experimental investigations are performed on USAF resolution test target. The method based on angular spectrum theory for reconstructing lensless Fourier hologram is given. The reconstructed results of the same hologram at different reconstructing distances are presented for what is to our knowledge the first time. Approximate diffraction limited lateral resolution is achieved. The results show that the angular spectrum method has several advantages over more commonly used Fresnel transform method. Lossless reconstruction can be achieved for any numerical aperture holograms as long as the wave field is calculated at a special reconstructing distance, which is determined by the light wavelength and the chip size and the pixel size of the CCD sensor. This is very important for reconstructing an extremely large numerical aperture hologram. Frequency-domain spectrum filtering can be applied conveniently to remove the disturbance of zero-order. The reconstructed image wave field is accurate so long as the sampling theorem is not violated. The experimental results also demonstrate that for a high quality hologram, special image processing is unnecessary to obtain a high quality image.
Institute of Scientific and Technical Information of China (English)
Muthuramalingam Uthaya Siva; Mohideen Abdul Badhul Haq; Deivasigamani Selvam; Ganesan Dinesh Babu; Rathinam Bakyaraj
2013-01-01
Objective: To investigate functional groups of toxic spines in stingray by Fourier transform infrared spectroscopic analysis.Methods:sephen were centrifuged at 6000 r/min for 10 min. The supernatant was collected and preserved separately in methanol, ethanol, chloroform, acetone (1:2) and then soaked in the mentioned solvents for 48 h. Then extracts were filtered and used for Fourier transform infrared spectroscopic analysis.Results:The venom extract of Himantura gerrardi, Himantura imbricata and Pastinachus and random coiled secondary structure. The presence of O-H stretch, C=O stretch, C-H stretch, N-H deformation, O-H deformation and C-O stretch in the sample aligned with standard bovine serum albumin. The influence of functional groups within the molecule was because of the impact of preferred spatial orientation, chemical and physical interaction on the molecule. In conclusion, compared to bovine serum albumin, Himantura imbricata consists of two C=O stretch, are involved in the hydrogen bonding that takes place between the different elements of secondary structure.Conclusions:The results identified that the presence of free amino acids and protein having β-sheet medicine not available for treatment against injuries causing stingray. Therefore, it's the baseline study, to motivate further process and produce effective antibiotics. The venom of poisonous animals has been extensively studied, since standard.
Directory of Open Access Journals (Sweden)
Muthuramalingam Uthaya Siva
2013-10-01
Full Text Available Objective: To investigate functional groups of toxic spines in stingray by Fourier transform infrared spectroscopic analysis. Methods: The venom extract of Himantura gerrardi, Himantura imbricata and Pastinachus sephen were centrifuged at 6 000 r/min for 10 min. The supernatant was collected and preserved separately in methanol, ethanol, chloroform, acetone (1:2 and then soaked in the mentioned solvents for 48 h. Then extracts were filtered and used for Fourier transform infrared spectroscopic analysis. Results: The results identified that the presence of free amino acids and protein having β-sheet and random coiled secondary structure. The presence of O-H stretch, C=O stretch, C-H stretch, N-H deformation, O-H deformation and C-O stretch in the sample aligned with standard bovine serum albumin. The influence of functional groups within the molecule was because of the impact of preferred spatial orientation, chemical and physical interaction on the molecule. In conclusion, compared to bovine serum albumin, Himantura imbricata consists of two C=O stretch, are involved in the hydrogen bonding that takes place between the different elements of secondary structure. Conclusions: The venom of poisonous animals has been extensively studied, since standard medicine not available for treatment against injuries causing stingray. Therefore, it's the baseline study, to motivate further process and produce effective antibiotics.
Institute of Scientific and Technical Information of China (English)
Ding Lu; Weimin Jin
2011-01-01
A novel fully phase color image encryption/decryption scheme based on joint fractional Fourier transform correlator (JFRTC) and phase retrieval algorithm (PRA) is proposed. The security of the system is enhanced by the fractional order as a new added key. This method takes full advantage of the parallel processing features of the optical system and could optically realize single-channel color image encryption.The system and operation procedures are simplified. The simulation results of a color image indicate that the new method provides efficient solutions with a strong sense of security.%@@ A novel fully phase color image encryption/decryption scheme based on joint fractional Fourier transform correlator (JFRTC) and phase retrieval algorithm (PRA) is proposed. The security of the system is enhanced by the fractional order as a new added key. This method takes full advantage of the parallel processing features of the optical system and could optically realize single-channel color image encryption. The system and operation procedures are simplified. The simulation results of a color image indicate that the new method provides efficient solutions with a strong sense of security.
Directory of Open Access Journals (Sweden)
Qiu Bo
2008-01-01
Full Text Available Binaural cue coding (BCC is an efficient technique for spatial audio rendering by using the side information such as interchannel level difference (ICLD, interchannel time difference (ICTD, and interchannel correlation (ICC. Of the side information, the ICTD plays an important role to the auditory spatial image. However, inaccurate estimation of the ICTD may lead to the audio quality degradation. In this paper, we develop a novel ICTD estimation algorithm based on the nonuniform discrete Fourier transform (NDFT and integrate it with the BCC approach to improve the decoded auditory image. Furthermore, a new subjective assessment method is proposed for the evaluation of auditory image widths of decoded signals. The test results demonstrate that the NDFT-based scheme can achieve much wider and more externalized auditory image than the existing BCC scheme based on the discrete Fourier transform (DFT. It is found that the present technique, regardless of the image width, does not deteriorate the sound quality at the decoder compared to the traditional scheme without ICTD estimation.
Directory of Open Access Journals (Sweden)
Bo Qiu
2008-05-01
Full Text Available Binaural cue coding (BCC is an efficient technique for spatial audio rendering by using the side information such as interchannel level difference (ICLD, interchannel time difference (ICTD, and interchannel correlation (ICC. Of the side information, the ICTD plays an important role to the auditory spatial image. However, inaccurate estimation of the ICTD may lead to the audio quality degradation. In this paper, we develop a novel ICTD estimation algorithm based on the nonuniform discrete Fourier transform (NDFT and integrate it with the BCC approach to improve the decoded auditory image. Furthermore, a new subjective assessment method is proposed for the evaluation of auditory image widths of decoded signals. The test results demonstrate that the NDFT-based scheme can achieve much wider and more externalized auditory image than the existing BCC scheme based on the discrete Fourier transform (DFT. It is found that the present technique, regardless of the image width, does not deteriorate the sound quality at the decoder compared to the traditional scheme without ICTD estimation.
Iterative Fourier transform algorithm: different approaches to diffractive optical element design
Skeren, Marek; Richter, Ivan; Fiala, Pavel
2002-10-01
This contribution focuses on the study and comparison of different design approaches for designing phase-only diffractive optical elements (PDOEs) for different possible applications in laser beam shaping. Especially, new results and approaches, concerning the iterative Fourier transform algorithm, are analyzed, implemented, and compared. Namely, various approaches within the iterative Fourier transform algorithm (IFTA) are analyzed for the case of phase-only diffractive optical elements with quantizied phase levels (either binary or multilevel structures). First, the general scheme of the IFTA iterative approach with partial quantization is briefly presented and discussed. Then, the special assortment of the general IFTA scheme is given with respect to quantization constraint strategies. Based on such a special classification, the three practically interesting approaches are chosen, further-analyzed, and compared to eachother. The performance of these algorithms is compared in detail in terms of the signal-to-noise ratio characteristic developments with respect to the numberof iterations, for various input diffusive-type objects chose. Also, the performance is documented on the complex spectra developments for typical computer reconstruction results. The advantages and drawbacks of all approaches are discussed, and a brief guide on the choice of a particular approach for typical design tasks is given. Finally, the two ways of amplitude elimination within the design procedure are considered, namely the direct elimination and partial elimination of the amplitude of the complex hologram function.
Binary encoding method to encrypt Fourier-transformed information of digital images
Lin, Kuang Tsan
2009-02-01
An encoding method is used to encrypt the Fourier-transformed information of a hidden (covert) digital image in an overt image, while the Fourier-transformed information must be encoded with binary codes. All of the pixels in an overt image are classified into five groups that are called identification, type, tracing, dimension, and information codes. Identification codes are used to judge if the overt image contains codes that belong to the proposed encoding method or not; type codes are used to judge the encoding type; tracing codes are used to judge the encoding trace; dimension codes are used to judge the size of the hidden information; and information codes are used to decode the hidden information. Applying the proposed encoding method is rather easy, and host images corresponding to overt images are not needed for decoding work. The experiment has demonstrated four types of encoding for the proposed encoding method to reconstruct covert images without any distortion or only with a little distortion.
Algorithm, applications and evaluation for protein comparison by Ramanujan Fourier transform.
Zhao, Jian; Wang, Jiasong; Hua, Wei; Ouyang, Pingkai
2015-12-01
The amino acid sequence of a protein determines its chemical properties, chain conformation and biological functions. Protein sequence comparison is of great importance to identify similarities of protein structures and infer their functions. Many properties of a protein correspond to the low-frequency signals within the sequence. Low frequency modes in protein sequences are linked to the secondary structures, membrane protein types, and sub-cellular localizations of the proteins. In this paper, we present Ramanujan Fourier transform (RFT) with a fast algorithm to analyze the low-frequency signals of protein sequences. The RFT method is applied to similarity analysis of protein sequences with the Resonant Recognition Model (RRM). The results show that the proposed fast RFT method on protein comparison is more efficient than commonly used discrete Fourier transform (DFT). RFT can detect common frequencies as significant feature for specific protein families, and the RFT spectrum heat-map of protein sequences demonstrates the information conservation in the sequence comparison. The proposed method offers a new tool for pattern recognition, feature extraction and structural analysis on protein sequences.
A fractional Fourier transform analysis of a bubble excited by an ultrasonic chirp.
Barlow, Euan; Mulholland, Anthony J
2011-11-01
The fractional Fourier transform is proposed here as a model based, signal processing technique for determining the size of a bubble in a fluid. The bubble is insonified with an ultrasonic chirp and the radiated pressure field is recorded. This experimental bubble response is then compared with a series of theoretical model responses to identify the most accurate match between experiment and theory which allows the correct bubble size to be identified. The fractional Fourier transform is used to produce a more detailed description of each response, and two-dimensional cross correlation is then employed to identify the similarities between the experimental response and each theoretical response. In this paper the experimental bubble response is simulated by adding various levels of noise to the theoretical model output. The method is compared to the standard technique of using time-domain cross correlation. The proposed method is shown to be far more robust at correctly sizing the bubble and can cope with much lower signal to noise ratios.
Otten, Leonard John, III; Butler, Eugene W.; Rafert, Bruce; Sellar, R. Glenn
1995-06-01
Kestrel Corporation and the Florida Institute of Technology have designed, and are now manufacturing, a Fourier transform visible hyperspectral imager system for use in a single engine light aircraft. The system is composed of a Sagnac-based interferometer optical subsystem, a data management system, and an aircraft attitude and current position sybsystem. The system is designed to have better than 5 nm spectral resolution at 450 nm, operates over the 440 nm to 1150 nm spectral band and has a 2D spatial resolution of 0.8 mrad. An internal calibration source is recorded with every frame of data to retain radiometric accuracy. The entire system fits into a Cessna 206 and uses a conventional downward looking view port located in the baggage compartment. During operation, data are collected at a rate of 15 Mbytes per second and stored direct to a disk array. Data storage has been sized to accommodate 56 minutes of observations. Designed for environmental mapping, this Fourier transform imager has uses in emergency response and military operations.
Fourier Transform Infrared Spectroscopic Study of Sodium Phosphate Solids and Solutions
Institute of Scientific and Technical Information of China (English)
龚文琪
2001-01-01
Solids and solutions of sodium phosphates with various chain lengths have been studied by using the techniques of diffuse reflectance infrared Fourier transform (DRIFT) spectroscopy and attenuated total reflection-Fourier transform infrared (ATR-FTIR) spectroscopy, respectively. A systematic study of the infrared spectra of the solid sodium phosphates has been conducted on the basis of the information available in the literatures to establish the assignments of the infrared vibrations of the different groups in the phosphate molecules. The infrared spectra of the solutions of sodium phosphates have been analyzed according to the infrared study on the relevant solids, in conjunction with the study of the phosphate species distribution in solution on the basis of the acid-base reaction equilibria. The results obtained have revealed the correlations between the infrared absorption spectra and the structure of the different P-O groups in different kinds of phosphates and are useful in the analysis of phosphate solids and solutions widely used in the various operations of mineral processing.
[A phase error correction method for the new Fourier transforms spectrometer].
Wang, Ning; Gong, Tian-Cheng; Chen, Jian-Jun; Li, Yang; Yang, Yi-Ning; Zhu, Yong; Zhang, Jie; Chen, Wei-Min
2014-11-01
To decrease the distortion of the recovered spectrum, improve the quantity of the recovered spectrum and decrease the influence of the phase error of the new spectrum detection system based on MEMS (micro-electro-mechanical systems) micro-mirrors, a new phase error correction method for this system is proposed in the present paper. The source of phase error of the spectrum detection system based on MEMS micro-mirrors is analyzed firstly. The analyzed result indicated that the phase error of the new spectral Fourier transform detection system is the zero drift of the optical path difference, and the phase error can be corrected by Zero-crossing sampling which is realized by improving the structure of the interferometer system and Mertz product The spectrum detection system is set up and the phase error correction method is verified by this system. The experiment result is show that the quantity of the recovered spectrum of the spectrum detection is improved obviously by using the improved interferometer system and Mertz product, and the recovered spectrum has no negative peaks and the side lobes is suppressed markedly. This correction method can reduce the influence caused by phase error to the system performance well and improve the spectral detection performance effectively. In this paper, the origin of the system phase error based on the new MEMS micromirror Fourier transform spectroscopy detection system is analyzed, and the phase error correction method is proposed. This method can improve the performance of the spectrum detection system.
A prototype stationary Fourier transform spectrometer for near-infrared absorption spectroscopy.
Li, Jinyang; Lu, Dan-feng; Qi, Zhi-mei
2015-09-01
A prototype stationary Fourier transform spectrometer (FTS) was constructed with a fiber-coupled lithium niobate (LiNbO3) waveguide Mach-Zehnder interferometer (MZI) for the purpose of rapid on-site spectroscopy of biological and chemical measurands. The MZI contains push-pull electrodes for electro-optic modulation, and its interferogram as a plot of intensity against voltage was obtained by scanning the modulating voltage from -60 to +60 V in 50 ms. The power spectrum of input signal was retrieved by Fourier transform processing of the interferogram combined with the wavelength dispersion of half-wave voltage determined for the MZI used. The prototype FTS operates in the single-mode wavelength range from 1200 to 1700 nm and allows for reproducible spectroscopy. A linear concentration dependence of the absorbance at λmax = 1451 nm for water in ethanolic solution was obtained using the prototype FTS. The near-infrared spectroscopy of solid samples was also implemented, and the different spectra obtained with different materials evidenced the chemical recognition capability of the prototype FTS. To make this prototype FTS practically applicable, work on improving its spectral resolution by increasing the maximum optical path length difference is in progress.
Hewagama, TIlak; Aslam, Shahid; Talabac, Stephen; Allen, John E., Jr.; Annen, John N.; Jennings, Donald E.
2011-01-01
Fourier transform spectrometers have a venerable heritage as flight instruments. However, obtaining an accurate spectrum exacts a penalty in instrument mass and power requirements. Recent advances in a broad class of non-scanning Fourier transform spectrometer (FTS) devices, generally called spatial heterodyne spectrometers, offer distinct advantages as flight optimized systems. We are developing a miniaturized system that employs photonics lightwave circuit principles and functions as an FTS operating in the 7-14 micrometer spectral region. The inteferogram is constructed from an ensemble of Mach-Zehnder interferometers with path length differences calibrated to mimic scan mirror sample positions of a classic Michelson type FTS. One potential long-term application of this technology in low cost planetary missions is the concept of a self-contained sensor system. We are developing a systems architecture concept for wide area in situ and remote monitoring of characteristic properties that are of scientific interest. The system will be based on wavelength- and resolution-independent spectroscopic sensors for studying atmospheric and surface chemistry, physics, and mineralogy. The self-contained sensor network is based on our concept of an Addressable Photonics Cube (APC) which has real-time flexibility and broad science applications. It is envisaged that a spatially distributed autonomous sensor web concept that integrates multiple APCs will be reactive and dynamically driven. The network is designed to respond in an event- or model-driven manner or reconfigured as needed.
Quantization maps, algebra representation, and non-commutative Fourier transform for Lie groups
Energy Technology Data Exchange (ETDEWEB)
Guedes, Carlos; Oriti, Daniele [Max Planck Institute for Gravitational Physics (Albert Einstein Institute), Am Mühlenberg 1, 14476 Potsdam (Germany); Raasakka, Matti [Max Planck Institute for Gravitational Physics (Albert Einstein Institute), Am Mühlenberg 1, 14476 Potsdam (Germany); LIPN, Institut Galilée, Université Paris-Nord, 99, av. Clement, 93430 Villetaneuse (France)
2013-08-15
The phase space given by the cotangent bundle of a Lie group appears in the context of several models for physical systems. A representation for the quantum system in terms of non-commutative functions on the (dual) Lie algebra, and a generalized notion of (non-commutative) Fourier transform, different from standard harmonic analysis, has been recently developed, and found several applications, especially in the quantum gravity literature. We show that this algebra representation can be defined on the sole basis of a quantization map of the classical Poisson algebra, and identify the conditions for its existence. In particular, the corresponding non-commutative star-product carried by this representation is obtained directly from the quantization map via deformation quantization. We then clarify under which conditions a unitary intertwiner between such algebra representation and the usual group representation can be constructed giving rise to the non-commutative plane waves and consequently, the non-commutative Fourier transform. The compact groups U(1) and SU(2) are considered for different choices of quantization maps, such as the symmetric and the Duflo map, and we exhibit the corresponding star-products, algebra representations, and non-commutative plane waves.
Monitoring wine aging with Fourier transform infrared spectroscopy (FT-IR
Directory of Open Access Journals (Sweden)
Basalekou Marianthi
2015-01-01
Full Text Available Oak wood has commonly been used in wine aging but recently other wood types such as Acacia and Chestnut, have attracted the interest of the researchers due to their possible positive contribution to wine quality. However, only the use of oak and chestnut woods is approved by the International Enological Codex of the International Organisation of Vine and Wine. In this study Fourier Transform (FT-mid-infrared spectroscopy combined with Discriminant Analysis was used to differentiate wines aged in barrels made from French oak, American oak, Acacia and Chestnut and in tanks with oak chips, over a period of 12 months. Two red (Mandilaria, Kotsifali and two white (Vilana, Dafni native Greek grape varieties where used to produce four wines. The Fourier Transform Infrared (FT-IR spectra of the samples were recorded on a Zinc Selenide (ZnSe window after incubation at 40 °C for 30 min. A complete differentiation of the samples according to both the type of wood used and the contact time was achieved based on their FT-IR spectra.
Dziuba, B
2013-01-01
Fourier transform infrared spectroscopy (FTIR) and artificial neural networks (ANN's) were used to identify species of Propionibacteria strains. The aim of the study was to improve the methodology to identify species of Propionibacteria strains, in which the differentiation index D, calculated based on Pearson's correlation and cluster analyses were used to describe the correlation between the Fourier transform infrared spectra and bacteria as molecular systems brought unsatisfactory results. More advanced statistical methods of identification of the FTIR spectra with application of artificial neural networks (ANN's) were used. In this experiment, the FTIR spectra of Propionibacteria strains stored in the library were used to develop artificial neural networks for their identification. Several multilayer perceptrons (MLP) and probabilistic neural networks (PNN) were tested. The practical value of selected artificial neural networks was assessed based on identification results of spectra of 9 reference strains and 28 isolates. To verify results of isolates identification, the PCR based method with the pairs of species-specific primers was used. The use of artificial neural networks in FTIR spectral analyses as the most advanced chemometric method supported correct identification of 93% bacteria of the genus Propionibacterium to the species level.
International Nuclear Information System (INIS)
We have developed a rapid method to screen large numbers of mutant plants for a broad range of cell wall phenotypes using Fourier transform infrared (FTIR) microspectroscopy of leaves. We established and validated a model that can discriminate between the leaves of wild-type and a previously defined set of cell-wall mutants of Arabidopsis. Exploratory principal component analysis indicated that mutants deficient in different cell-wall sugars can be distinguished from each other. Discrimination of cell-wall mutants from wild-type was independent of variability in starch content or additional unrelated mutations that might be present in a heavily mutagenised population. We then developed an analysis of FTIR spectra of leaves obtained from over 1000 mutagenised flax plants, and selected 59 plants whose spectral variation from wild-type was significantly out of the range of a wild-type population, determined by Mahalanobis distance. Cell wall sugars from the leaves of selected putative mutants were assayed by gas chromatography-mass spectrometry and 42 showed significant differences in neutral sugar composition. The FTIR spectra indicated that six of the remaining 17 plants have altered ester or protein content. We conclude that linear discriminant analysis of FTIR spectra is a robust method to identify a broad range of structural and architectural alterations in cell walls, appearing as a consequence of developmental regulation, environmental adaptation or genetic modification. (author)
Zhou, Xibin; Shi, Quan; Zhang, Yahe; Zhao, Suoqi; Zhang, Rui; Chung, Keng H; Xu, Chunming
2012-04-01
A novel technique was developed for characterization of saturated hydrocarbons. Linear alkanes were selectively oxidized to ketones by ruthenium ion catalyzed oxidation (RICO). Branched and cyclic alkanes were oxidized to alcohols and ketones. The ketones were then reduced to alcohols by lithium aluminum hydride (LiAlH(4)). The monohydric alcohols (O(1)) in the products obtained from the RICO and RICO-LiAlH(4) reduction reactions were characterized using negative-ion electrospray ionization (ESI) Fourier transform ion cyclotron resonance mass spectrometry (FTICR MS) for identification of iso-paraffins, acyclic paraffins and cyclic paraffins. Various model saturated compounds were used to determine the RICO reaction and ionization selectivity. The results from the FTICR MS analysis on the petroleum distillates derived saturated fraction were in agreement with those from field ionization gas chromatography time-of-flight mass spectrometry (FI GC-TOF MS) analysis. The technique was also used to characterize a petroleum vacuum residue (VR) derived saturates. The results showed that the saturated molecules in the VR contained up to 11 cyclic rings, and the maximum carbon number was up to 92. PMID:22424498
Karami, A; Balke, S T; Schunk, T C
2001-03-01
Quantitative evaluation of polymer composition across the SEC chromatogram can provide more accurate characterization of heterogeneous polymer samples for problem solving and for material specification. To this end Fourier-transform infrared spectroscopy (FTIR) with solvent-evaporation interfaces has become a very powerful detector for size-exclusion chromatography (SEC). The solvent-evaporation interface removes the mobile phase at the exit of the chromatograph and deposits the separated molecular sizes as polymer films on infrared transparent substrates. Quantitative interpretation of the FTIR spectra obtained from these films has recently been found to be best accomplished by using partial least squares. In this paper, polystyrene and poly(methylmethacrylate), alone, as blends, and a copolymer were analyzed in a SEC equipped with an evaporative interface. Molecular weight effects, wavelength selection, the effect of averaging spectra on results, and selection of the best data preprocessing method were investigated. General methods of evaluating these variables were developed to arrive at conditions for this particular "model" situation in order to provide a basis for the analysis of more complex polymers.
Markopoulou, C K; Koundourellis, J E; Orkoula, M G; Kontoyannis, C G
2008-02-01
Two different nondestructive spectroscopy methods based on near-infrared (NIR) and Fourier transform (FT) Raman spectroscopy were developed for the determination of ticlopidine-hydrochloride (TCL) in pharmaceutical formulations and the results were compared to those obtained by high-performance liquid chromatography (HPLC). An NIR assay was performed by reflectance over the 850-1700 nm region using a partial least squares (PLS) prediction model, while the absolute FT-Raman intensity of TCL's most intense vibration was used for constructing the calibration curve. For both methodologies the spectra were obtained from the as-received film-coated tablets of TCL. The two quantitative techniques were built using five "manual compressed" tablets containing different concentrations and validated by evaluating the calibration model as well as the accuracy and precision. The models were applied to commercial preparations (Ticlid). The results were compared to those obtained from the application of HPLC using the methodology described by "Sanofi Research Department" and were found to be in excellent agreement, proving that NIR, using fiber-optic probes, and FT-Raman spectroscopy can be used for the fast and reliable determination of the major component in pharmaceutical analysis. PMID:18284803
Hu, Yaxi; Pan, Zhi Jie; Liao, Wen; Li, Jiaqi; Gruget, Pierre; Kitts, David D; Lu, Xiaonan
2016-07-01
Antioxidant capacity and phenolic content of chocolate, containing different amounts of cacao (35-100%), were determined using attenuated total reflectance (ATR)-Fourier transformed-infrared (FT-IR) spectroscopy (4000-550cm(-1)). Antioxidant capacities were first characterized using DPPH (2,2-diphenyl-1-picrylhydrazyl) and ORAC (oxygen radical absorbance capacity) assays. Phenolic contents, including total phenol and procyanidins monomers, were quantified using the Folin-Ciocalteu assay and high performance liquid chromatography coupled with photodiode array detector (HPLC-DAD), respectively. Five partial least-squares regression (PLSR) models were constructed and cross-validated using FT-IR spectra from 18 types of chocolate and corresponding reference values determined using DPPH, ORAC, Folin-Ciocalteu, and HPLC assays. The models were validated using seven unknown samples of chocolate. PLSR models showed good prediction capability for DPPH [R(2)-P (prediction)=0.88, RMSEP (root mean squares error of prediction)=12.62μmol Trolox/g DFW], ORAC (R(2)-P=0.90, RMSEP=37.92), Folin-Ciocalteu (R(2)-P=0.88, RMSEP=5.08), and (+)-catechin (R(2)-P=0.86, RMSEP=0.10), but lacked accuracy in the prediction of (-)-epicatechin (R(2)-P=0.72, RMSEP=0.57). ATR-FT-IR spectroscopy can be used for rapid prediction of antioxidant capacity, total phenolic content, and (+)-catechin in chocolate. PMID:26920292
Chen, Li; Yap, Yee Leng
2008-01-01
This study describes a new algorithm for charge state determination of complex isotope-resolved mass spectra. This algorithm is based on peak-target Fourier transform (PTFT) of isotope packets. It is modified from the widely used Fourier transform method because Fourier transform may give ambiguous charge state assignment for low signal-to-noise ratio (S/N) or overlapping isotopic clusters. The PTFT algorithm applies a novel "folding" strategy to enhance peaks that are symmetrically spaced about the targeted peak before applying the FT. The "folding" strategy multiplies each point to the high-m/z side of the targeted peak by its counterpart on the low-m/z side. A Fourier transform of this "folded" spectrum is thus simplified, emphasizing the charge state of the "chosen" ion, whereas ions of other charge states contribute less to the transformed data. An intensity-dependent technique is also proposed for charge state determination from frequency signals. The performance of PTFT is demonstrated using experimental electrospray ionization Fourier transform ion cyclotron resonance mass spectra. The results show that PTFT is robust for charge state determination of low S/N and overlapping isotopic clusters, and also useful for manual verification of potential hidden isotopic clusters that may be missed by the current analysis algorithms, i.e., AID-MS or THRASH.
Celeghini, Enrico
2014-01-01
Quantum Mechanics and Signal Processing in the line R, are strictly related to Fourier Transform and Weyl-Heisenberg algebra. We discuss here the addition of a new discrete variable that measures the degree of the Hermite functions and allows to obtain the projective algebra io(2). A Rigged Hilbert space is found and a new discrete basis in R obtained. The operators {O[R]} defined on R are shown to belong to the Universal Enveloping Algebra UEA[io(2)] allowing, in this way, their algebraic discussion. Introducing in the half-line a Fourier-like Transform, the procedure is extended to R^+ and can be easily generalized to R^n and to spherical reference systems.
Directory of Open Access Journals (Sweden)
bahram noshad
2012-11-01
Full Text Available One of mal-operations of the transformer differential protection during the unload transformer energizing with additional line/load from the supplying side is ultra-saturation phenomenon. In this paper, first a new model according to Discrete Fourier Transform (DFT algorithm for investigating the ultra-saturation phenomenon during the unload transformer energizing with additional line/load from the supplying side is presented and its effect on the differential protection of the transformer is considered. In this model, the nonlinear characteristic of the transformer core and the effect of current transformer are taken into account. It is assumed that the additional line/load from the supplying side of the power transformer is a resistive-inductive load. Also, the effect of the residual flux, inception angle and additional line/load from the supplying side on the ultra-saturation phenomenon is investigated. Then, the mal-operation of differential protection by using DFT algorithm is described.
Real-time Fourier transformation of lightwave spectra and application in optical reflectometry.
Malacarne, Antonio; Park, Yongwoo; Li, Ming; LaRochelle, Sophie; Azaña, José
2015-12-14
We propose and experimentally demonstrate a fiber-optics scheme for real-time analog Fourier transform (FT) of a lightwave energy spectrum, such that the output signal maps the FT of the spectrum of interest along the time axis. This scheme avoids the need for analog-to-digital conversion and subsequent digital signal post-processing of the photo-detected spectrum, thus being capable of providing the desired FT processing directly in the optical domain at megahertz update rates. The proposed concept is particularly attractive for applications requiring FT analysis of optical spectra, such as in many optical Fourier-domain reflectrometry (OFDR), interferometry, spectroscopy and sensing systems. Examples are reported to illustrate the use of the method for real-time OFDR, where the target axial-line profile is directly observed in a single-shot oscilloscope trace, similarly to a time-of-flight measurement, but with a resolution and depth of range dictated by the underlying interferometry scheme. PMID:26699041
Real-time Fourier transformation of lightwave spectra and application in optical reflectometry.
Malacarne, Antonio; Park, Yongwoo; Li, Ming; LaRochelle, Sophie; Azaña, José
2015-12-14
We propose and experimentally demonstrate a fiber-optics scheme for real-time analog Fourier transform (FT) of a lightwave energy spectrum, such that the output signal maps the FT of the spectrum of interest along the time axis. This scheme avoids the need for analog-to-digital conversion and subsequent digital signal post-processing of the photo-detected spectrum, thus being capable of providing the desired FT processing directly in the optical domain at megahertz update rates. The proposed concept is particularly attractive for applications requiring FT analysis of optical spectra, such as in many optical Fourier-domain reflectrometry (OFDR), interferometry, spectroscopy and sensing systems. Examples are reported to illustrate the use of the method for real-time OFDR, where the target axial-line profile is directly observed in a single-shot oscilloscope trace, similarly to a time-of-flight measurement, but with a resolution and depth of range dictated by the underlying interferometry scheme.
Chirp-Rate Resolution of Fractional Fourier Transform in Multi-component LFM Signal
Institute of Scientific and Technical Information of China (English)
LIU Feng; HUANG Yu; TAO Ran; WANG Yue
2009-01-01
Distinguishing close chirp-rates of different linear frequency modulation (LFM) signals under concentrated and complicated signal environment was studied.Firstly,detection and parameter estimation of multi-component LFM signal were used by discrete fast fractional Fourier transform (FrFT).Then the expression of chirp-rate resolution in fractional Fourier domain (FrFD) was deduced from discrete normalize time-frequency distribution,when multi-component LFM signal had only one center frequency.Furthermore,the detail influence of the sampling time,sampling frequeney and chirp-rate upon the resolution was analyzed by partial differential equation.Simulation results and analysis indicate that increasing the sampling time can enhance the resolution,but the influence of the sampling frequency can be omitted.What's more,in multi-component LFM signal,the chirp-rate resolution of FrFT is no less than a minimal value,and it mainly dependent on the biggest value of chirp-rates,with which it has an approximately positive exponential relationship.
Double-resolution electron holography with simple Fourier transform of fringe-shifted holograms
Energy Technology Data Exchange (ETDEWEB)
Volkov, V.V., E-mail: volkov@bnl.gov; Han, M.G.; Zhu, Y.
2013-11-15
We propose a fringe-shifting holographic method with an appropriate image wave recovery algorithm leading to exact solution of holographic equations. With this new method the complex object image wave recovered from holograms appears to have much less traditional artifacts caused by the autocorrelation band present practically in all Fourier transformed holograms. The new analytical solutions make possible a double-resolution electron holography free from autocorrelation band artifacts and thus push the limits for phase resolution. The new image wave recovery algorithm uses a popular Fourier solution of the side band-pass filter technique, while the fringe-shifting holographic method is simple to implement in practice. - Highlights: • We propose a fringe-shifting holographic method simple enough for practical implementations. • Our new image-wave-recovery algorithm follows from exact solution of holographic equations. • With autocorrelation band removal from holograms it is possible to achieve double-resolution electron holography data free from several commonly known artifacts. • The new fringe-shifting method can reach an image wave resolution close to single fringe spacing.
Real-time discrimination of ventricular tachyarrhythmia with Fourier-transform neural network.
Minami, K; Nakajima, H; Toyoshima, T
1999-02-01
We have developed a method to discriminate life-threatening ventricular arrhythmias by observing the QRS complex of the electrocardiogram (ECG) in each heartbeat. Changes in QRS complexes due to rhythm origination and conduction path were observed with the Fourier transform, and three kinds of rhythms were discriminated by a neural network. In this paper, the potential of our method for clinical uses and real-time detection was examined using human surface ECG's and intracardiac electrograms (EGM's). The method achieved high sensitivity and specificity (> or = 0.98) in discrimination of supraventricular rhythms from ventricular ones. We also present a hardware implementation of the algorithm on a commercial single-chip CPU. PMID:9932339
Amirfattahi, Rassoul
2013-10-01
Owing to its simplicity radix-2 is a popular algorithm to implement fast fourier transform. Radix-2(p) algorithms have the same order of computational complexity as higher radices algorithms, but still retain the simplicity of radix-2. By defining a new concept, twiddle factor template, in this paper, we propose a method for exact calculation of multiplicative complexity for radix-2(p) algorithms. The methodology is described for radix-2, radix-2 (2) and radix-2 (3) algorithms. Results show that radix-2 (2) and radix-2 (3) have significantly less computational complexity compared with radix-2. Another interesting result is that while the number of complex multiplications in radix-2 (3) algorithm is slightly more than radix-2 (2), the number of real multiplications for radix-2 (3) is less than radix-2 (2). This is because of the twiddle factors in the form of which need less number of real multiplications and are more frequent in radix-2 (3) algorithm.
Suto, Hiroshi; Yoshida, Jun; Desbiens, Raphael; Kawashima, Takahiro; Kuze, Akihiko
2013-07-10
Microvibrations onboard greenhouse gases observing satellite (GOSAT) cause scan speed variations in the TANSO Fourier transform spectrometer. The associated periodic sampling errors generate ghost features in O2 A-band spectra, where surface pressure and aerosol properties are retrieved to determine the optical path through the atmosphere. A correction algorithm has been developed to re-compute the interferograms at equally spaced sampling intervals. The key is to determine iteratively the amplitude and phase of sinusoidal perturbations with predetermined frequencies to minimize the magnitude of the out-of-band ghosts artifacts after correction of the sampling grid. This correction algorithm drastically reduces errors in retrieved surface pressure and improves agreement with ground-based observations.
Directory of Open Access Journals (Sweden)
Jinsoo Park
2014-01-01
Full Text Available A new voice activity detector for noisy environments is proposed. In conventional algorithms, the endpoint of speech is found by applying an edge detection filter that finds the abrupt changing point in a feature domain. However, since the frame energy feature is unstable in noisy environments, it is difficult to accurately find the endpoint of speech. Therefore, a novel feature extraction algorithm based on the double-combined Fourier transform and envelope line fitting is proposed. It is combined with an edge detection filter for effective detection of endpoints. Effectiveness of the proposed algorithm is evaluated and compared to other VAD algorithms using two different databases, which are AURORA 2.0 database and SITEC database. Experimental results show that the proposed algorithm performs well under a variety of noisy conditions.
Wavelength-scale stationary-wave integrated Fourier-transform spectrometry
Coarer, Etienne Le; Benech, Pierre; Stefanon, Ilan; Morand, Alain; Lérondel, Gilles; Leblond, Grégory; Kern, Pierre; Fedeli, Jean Marc; Royer, Pascal
2007-01-01
Spectrometry is a general physical-analysis approach for investigating light-matter interactions. However, the complex designs of existing spectrometers render them resistant to simplification and miniaturization, both of which are vital for applications in micro- and nanotechnology and which are now undergoing intensive research. Stationary-wave integrated Fourier-transform spectrometry (SWIFTS)-an approach based on direct intensity detection of a standing wave resulting from either reflection (as in the principle of colour photography by Gabriel Lippmann) or counterpropagative interference phenomenon-is expected to be able to overcome this drawback. Here, we present a SWIFTS-based spectrometer relying on an original optical near-field detection method in which optical nanoprobes are used to sample directly the evanescent standing wave in the waveguide. Combined with integrated optics, we report a way of reducing the volume of the spectrometer to a few hundreds of cubic wavelengths. This is the first attempt...
Chen, Cheng; Liang, Jingqiu; Liang, Zhongzhu; Lü, Jinguang; Qin, Yuxin; Tian, Chao; Wang, Weibiao
2015-12-01
A method of "slope splicing" is proposed to build a tall-stepped mirror with high precision in a stepped-mirror-based static Fourier transform infrared spectrometer. The structural parameters were designed, and their errors were analyzed. We present the test results and an analysis of the combined effect of the errors on the recovered spectrum. The spectrum-constructing error of the constructed spectrum, 5.81%, meets the requirements for the system and suitable for realization of a miniaturized spectrometer. We performed experiments with the tall-stepped mirror to obtain the interferogram and spectrum of a silicon carbide light source. Further work is needed to optimize the capability of the system.
International Nuclear Information System (INIS)
Recent progress is described resulting from the past three years of DOE support for studies of combustion-related photofragmentation dynamics, energy transfer, and reaction processes using a time-resolved Fourier transform infrared (FTIR) emission technique. The FTIR is coupled to a high repetition rate excimer laser which produces radicals by photolysis to obtain novel, high resolution measurements on vibrational and rotational state dynamics. The results are important for the study of numerous radical species relevant to combustion processes. The method has been applied to the detailed study of photofragmentation dynamics in systems such as acetylene, which produces C2H; chlorofluoroethylene to study the HF product channel; vinyl chloride and dichloroethylene, which produce HCl; acetone, which produces CO and CH3; and ammonia, which produces NH2. In addition, we have recently demonstrated use of the FTIR technique for preliminary studies of energy transfer events under near single collision conditions, radical-radical reactions, and laser-initiated chain reaction processes
Supercontinuum Fourier transform spectrometry with balanced detection on a single photodiode.
Goncharov, Vasily V; Hall, Gregory E
2016-08-28
We demonstrate a method of combining a supercontinuum light source with a commercial Fourier transform spectrometer, using a novel approach to dual-beam balanced detection, implemented with phase-sensitive detection on a single light detector. A 40 dB reduction in the relative intensity noise is achieved for broadband light, analogous to conventional balanced detection methods using two matched photodetectors. Unlike conventional balanced detection, however, this method exploits the time structure of the broadband source to interleave signal and reference pulse trains in the time domain, recording the broadband differential signal at the fundamental pulse repetition frequency of the supercontinuum. The method is capable of real-time correction for instability in the supercontinuum spectral structure over a broad range of wavelengths and is compatible with commercially designed spectrometers. A proof-of-principle experimental setup is demonstrated for weak absorption in the 1500-1600 nm region. PMID:27586915
Davis, Benjamin L; Shields, Douglas W; Kennefick, Julia; Kennefick, Daniel; Seigar, Marc S; Lacy, Claud H S; Puerari, Ivânio
2012-01-01
A logarithmic spiral is a prominent feature appearing in a majority of observed galaxies. This feature has long been associated with the traditional Hubble classification scheme, but historical quotes of pitch angle of spiral galaxies have been almost exclusively qualitative. We have developed a methodology, utilizing two-dimensional fast Fourier transformations of images of spiral galaxies, in order to isolate and measure the pitch angles of their spiral arms. Our technique provides a quantitative way to measure this morphological feature. This will allow comparison of spiral galaxy pitch angle to other galactic parameters and test spiral arm genesis theories. In this work, we detail our image processing and analysis of spiral galaxy images and discuss the robustness of our analysis techniques.
Melin, A M; Perromat, A; Déléris, G
2000-01-01
Microsomal fractions from rat liver were examined by means of Fourier transform IR (FTIR) spectroscopy to study the in vivo toxic effect of carbon tetrachloride administered by intraperitoneal injection. Lipid content was significantly enhanced in the liver of treated rats compared with untreated ones. The level of saturated fatty acids largely increased while that of unsaturated acids slightly decreased as a consequence of lipid peroxidation induced by the xenobiotic compound. The conformational structure of membrane proteins was changed, which was shown by the large decrease in the alpha-helical configuration. In the polysaccharide region we observed an important loss in glucidic structures that could be related to the metabolic changes caused by carbon tetrachloride intoxication. Thus, FTIR spectroscopy appears to be a useful tool to rapidly investigate the chemical alterations induced by this drug in liver microsomes and to correlate them with biochemical and physiological data.
Fourier transform based iterative method for x-ray differential phase-contrast computed tomography
Cong, Wenxiang; Wang, Ge
2011-01-01
Biological soft tissues encountered in clinical and pre-clinical imaging mainly consist of light element atoms, and their composition is nearly uniform with little density variation. Thus, x-ray attenuation imaging suffers from low image contrast resolution. By contrast, x-ray phase shift of soft tissues is about a thousand times greater than x-ray absorption over the diagnostic energy range, thereby a significantly higher sensitivity can be achieved in terms of phase shift. In this paper, we propose a novel Fourier transform based iterative method to perform x-ray tomographic imaging of the refractive index directly from differential phase shift data. This approach offers distinct advantages in cases of incomplete and noisy data than analytic reconstruction, and especially suitable for phase-contrast interior tomography by incorporating prior knowledge in a region of interest (ROI). Biological experiments demonstrate the merits of the proposed approach.
McNaughton, Don; Godfrey, Peter D.; Jahn, Michaela K.; Dewald, David A.; Grabow, Jens-Uwe
2011-04-01
The pure rotational spectra of the bicyclic aromatic nitrogen heterocycle molecules, quinazoline, quinoxaline, and phthalazine, have been recorded and assigned in the region 13-87 GHz. An analysis, guided by ab initio molecular orbital predictions, of frequency-scanned Stark modulated, jet-cooled millimeter wave absorption spectra (48-87 GHz) yielded a preliminary set of rotational and centrifugal distortion constants. Subsequent spectral analysis at higher resolution was carried out with Fourier transform microwave (FT-MW) spectroscopy (13-18 GHz) of a supersonic rotationally cold molecular beam. The high spectral resolution of the FT-MW instrument provided an improved set of rotational and centrifugal distortion constants together with nitrogen quadrupole coupling constants for all three species. Density functional theory calculations at the B3LYP/6-311+G** level of theory closely predict rotational constants and are useful in predicting quadrupole coupling constants and dipole moments for such species.
Reference Ultraviolet Wavelengths of Cr III Measured by Fourier Transform Spectrometry
Smillie, D.G.; Pickering, J.C.; Smith, P.L.
2008-01-01
We report Cr III ultraviolet (UV) transition wavelengths measured using a high-resolution Fourier transform spectrometer (FTS), for the first time, available for use as wavelength standards. The doubly ionized iron group element spectra dominate the observed opacity of hot B stars in the UV, and improved, accurate, wavelengths are required for the analysis of astronomical spectra. The spectrum was excited using a chromium-neon Penning discharge lamp and measured with the Imperial College vacuum ultraviolet FTS. 140 classified 3d(exp 3)4s- 3d(exp 3)4p Cr III transition lines, in the spectral range 38,000 to 49,000 cm(exp -1) (2632 to 2041 A), the strongest having wavelength uncertainties less than one part in 10(exp 7), are presented.
An Analog of the Fourier Transform Associated with a Nonlinear One-Dimensional Schroedinger Equation
Zhidkov, E P
2001-01-01
We consider an eigenvalue problem which includes a nonlinear Schroedinger equation on the half-line [0,\\infty) and certain boundary conditions. It is shown that the spectrum of this problem fills a half-line and that to each point of the spectrum there corresponds a unique eigenfunction. The main result of the paper is that an arbitrary infinitely differentiable function g(x) rapidly decaying as x\\to\\infty and satisfying suitable boundary conditions at the point x=0 can be uniquely expanded into an integral over eigenfunctions similar to the representation of functions by the Fourier transform (the latter is obviously associated with a linear self-adjoint eigenvalue problem).
High-definition Fourier transform infrared spectroscopic imaging of prostate tissue
Wrobel, Tomasz P.; Kwak, Jin Tae; Kadjacsy-Balla, Andre; Bhargava, Rohit
2016-03-01
Histopathology forms the gold standard for cancer diagnosis and therapy, and generally relies on manual examination of microscopic structural morphology within tissue. Fourier-Transform Infrared (FT-IR) imaging is an emerging vibrational spectroscopic imaging technique, especially in a High-Definition (HD) format, that provides the spatial specificity of microscopy at magnifications used in diagnostic surgical pathology. While it has been shown for standard imaging that IR absorption by tissue creates a strong signal where the spectrum at each pixel is a quantitative "fingerprint" of the molecular composition of the sample, here we show that this fingerprint also enables direct digital pathology without the need for stains or dyes for HD imaging. An assessment of the potential of HD imaging to improve diagnostic pathology accuracy is presented.
Rohman, A; Man, Yb Che; Sismindari
2009-10-01
Today, virgin coconut oil (VCO) is becoming valuable oil and is receiving an attractive topic for researchers because of its several biological activities. In cosmetics industry, VCO is excellent material which functions as a skin moisturizer and softener. Therefore, it is important to develop a quantitative analytical method offering a fast and reliable technique. Fourier transform infrared (FTIR) spectroscopy with sample handling technique of attenuated total reflectance (ATR) can be successfully used to analyze VCO quantitatively in cream cosmetic preparations. A multivariate analysis using calibration of partial least square (PLS) model revealed the good relationship between actual value and FTIR-predicted value of VCO with coefficient of determination (R2) of 0.998.
Detection of Left-Sided and Right-Sided Hearing Loss via Fractional Fourier Transform
Directory of Open Access Journals (Sweden)
Shuihua Wang
2016-05-01
Full Text Available In order to detect hearing loss more efficiently and accurately, this study proposed a new method based on fractional Fourier transform (FRFT. Three-dimensional volumetric magnetic resonance images were obtained from 15 patients with left-sided hearing loss (LHL, 20 healthy controls (HC, and 14 patients with right-sided hearing loss (RHL. Twenty-five FRFT spectrums were reduced by principal component analysis with thresholds of 90%, 95%, and 98%, respectively. The classifier is the single-hidden-layer feed-forward neural network (SFN trained by the Levenberg–Marquardt algorithm. The results showed that the accuracies of all three classes are higher than 95%. In all, our method is promising and may raise interest from other researchers.
Speeding-up exchange-mediated saturation transfer experiments by Fourier transform
Energy Technology Data Exchange (ETDEWEB)
Carneiro, Marta G.; Reddy, Jithender G.; Griesinger, Christian; Lee, Donghan, E-mail: dole@nmr.mpibpc.mpg.de [Max-Planck Institute for Biophysical chemistry, Department of NMR-based Structural Biology (Germany)
2015-11-15
Protein motions over various time scales are crucial for protein function. NMR relaxation dispersion experiments play a key role in explaining these motions. However, the study of slow conformational changes with lowly populated states remained elusive. The recently developed exchange-mediated saturation transfer experiments allow the detection and characterization of such motions, but require extensive measurement time. Here we show that, by making use of Fourier transform, the total acquisition time required to measure an exchange-mediated saturation transfer profile can be reduced by twofold in case that one applies linear prediction. In addition, we demonstrate that the analytical solution for R{sub 1}ρ experiments can be used for fitting the exchange-mediated saturation transfer profile. Furthermore, we show that simultaneous analysis of exchange-mediated saturation transfer profiles with two different radio-frequency field strengths is required for accurate and precise characterization of the exchange process and the exchanging states.
Robust digital speckle photography based on radon and Fourier-Mellin transforms
Kamau, E. N.; Falldorf, C.; von Kopylow, C.; Bergmann, R. B.
2011-11-01
We present a new and robust method for determining in-plane displacements of an object from distorted Digital Speckle Photography (DSP) images. This new approach is designed particularly to facilitate accurate measurement of deformations of steel samples during a gas quenching heat treatment process, where rigid body motion and large deformations lead to unwanted image distortions. The new method allows the computation and correction of image rotation and magnification via Radon and Fourier-Mellin Transformations prior to calculations of in-plane displacements, thereby alleviating the inaccuracy that arises from the cross correlation of distorted images in conventional DSP. The method is validated through simulation and measurements with predefined deformations. Initial studies show that the new method is well suited for this application and that it enables measurement of displacements with high accuracy in the micrometer range.
Jusman, Yessi; Mat Isa, Nor Ashidi; Ng, Siew-Cheok; Hasikin, Khairunnisa; Abu Osman, Noor Azuan
2016-07-01
Fourier transform infrared (FTIR) spectroscopy technique can detect the abnormality of a cervical cell that occurs before the morphological change could be observed under the light microscope as employed in conventional techniques. This paper presents developed features extraction for an automated screening system for cervical precancerous cell based on the FTIR spectroscopy as a second opinion to pathologists. The automated system generally consists of the developed features extraction and classification stages. Signal processing techniques are used in the features extraction stage. Then, discriminant analysis and principal component analysis are employed to select dominant features for the classification process. The datasets of the cervical precancerous cells obtained from the feature selection process are classified using a hybrid multilayered perceptron network. The proposed system achieved 92% accuracy.
Zimmerman, G. A.; Gulkis, S.
1991-01-01
The sensitivity of a matched filter-detection system to a finite-duration continuous wave (CW) tone is compared with the sensitivities of a windowed discrete Fourier transform (DFT) system and an ideal bandpass filter-bank system. These comparisons are made in the context of the NASA Search for Extraterrestrial Intelligence (SETI) microwave observing project (MOP) sky survey. A review of the theory of polyphase-DFT filter banks and its relationship to the well-known windowed-DFT process is presented. The polyphase-DFT system approximates the ideal bandpass filter bank by using as few as eight filter taps per polyphase branch. An improvement in sensitivity of approx. 3 dB over a windowed-DFT system can be obtained by using the polyphase-DFT approach. Sidelobe rejection of the polyphase-DFT system is vastly superior to the windowed-DFT system, thereby improving its performance in the presence of radio frequency interference (RFI).
Studies of beam propagation characteristics on apertured fractional Fourier transforming systems
Institute of Scientific and Technical Information of China (English)
Hongjie Liu(刘红婕); Daomu Zhao(赵道木); Haidan Mao(毛海丹); Shaomin Wang(王绍民); Feng Jing(景峰); Qihua Zhu(朱启华); Xiaofeng Wei(魏晓峰); Xiaomin Zhang(张小民)
2004-01-01
Based on the principle that a rectangular function can be expanded into a sum of complex Gaussian functions with finite numbers, propagation characteristics of a Gaussian beam or a plane wave passing through apertured fractional Fourier transforming systems are analyzed and corresponding analytical formulae are obtained. Analytical formulae in different fractional orders are numerically simulated and compared with the diffraction integral formulae, the applicable range and exactness of analytical formulae are confirmed.It is shown that the calculating speed of using the obtained approximate analytical formulae, is several hundred times faster than that of using diffraction integral directly. Meanwhile, by using analytical formulae the effect of different aperture sizes on Gaussian beam propagation characteristics is numerically simulated, it is shown that the diffraction effect can be neglected when the aperture size is 5 times larger than the beam waist size.
Energy Technology Data Exchange (ETDEWEB)
Nagler, Peter C.; Tucker, Gregory S. [Department of Physics, Brown University, Providence, RI 02912 (United States); Fixsen, Dale J.; Kogut, Alan, E-mail: peter.c.nagler@nasa.gov [NASA/Goddard Space Flight Center, Code 553, Greenbelt, MD 20771 (United States)
2015-11-15
The detection of the primordial B-mode polarization signal of the cosmic microwave background (CMB) would provide evidence for inflation. Yet as has become increasingly clear, the detection of a such a faint signal requires an instrument with both wide frequency coverage to reject foregrounds and excellent control over instrumental systematic effects. Using a polarizing Fourier transform spectrometer (FTS) for CMB observations meets both of these requirements. In this work, we present an analysis of instrumental systematic effects in polarizing FTSs, using the Primordial Inflation Explorer (PIXIE) as a worked example. We analytically solve for the most important systematic effects inherent to the FTS—emissive optical components, misaligned optical components, sampling and phase errors, and spin synchronous effects—and demonstrate that residual systematic error terms after corrections will all be at the sub-nK level, well below the predicted 100 nK B-mode signal.
Irudayaraj, Joseph; Yang, Hong; Sakhamuri, Sivakesava
2002-03-01
Fourier transform infrared photoacoustic spectroscopy (FTIR-PAS) was used to differentiate and identify microorganisms on a food (apple) surface. Microorganisms considered include bacteria ( Lactobacillus casei, Bacillus cereus, and Escherichia coli), yeast ( Saccharomyces cerevisiae), and fungi ( Aspergillus niger and Fusarium verticilliodes). Discriminant analysis was used to differentiate apples contaminated with the different microorganisms from uncontaminated apple. Mahalanobis distances were calculated to quantify the differences. The higher the value of the Mahalanobis distance metric between different microorganisms, the greater is their difference. Additionally, pathogenic (O157:H7) E. coli was successfully differentiated from non-pathogenic strains. Results demonstrate that FTIR-PAS spectroscopy has the potential to become a non-destructive analysis tool in food safety related research.
Directory of Open Access Journals (Sweden)
Hsiu-An eChu
2013-05-01
Full Text Available The photosystem II reaction center mediates the light-induced transfer of electrons from water to plastoquinone, with concomitant production of O2. Water oxidation chemistry occurs in the oxygen-evolving complex (OEC, which consists of an inorganic Mn4CaO5 cluster and its surrounding protein matrix. Light-induced Fourier transform infrared (FTIR difference spectroscopy has been successfully used to study the molecular mechanism of photosynthetic water oxidation. This powerful technique has enabled the characterization of the dynamic structural changes in active water molecules, the Mn4CaO5 cluster, and its surrounding protein matrix during the catalytic cycle. This mini-review presents an overview of recent important progress in FTIR studies of the OEC and implications for revealing the molecular mechanism of photosynthetic water oxidation.
Energy Technology Data Exchange (ETDEWEB)
Hettich, R.L.; Jin, C.; Compton, R.N. (Oak Ridge National Laboratory, Oak Ridge, Tennessee 37831-6120 (United States)); Buseck, P.R.; Tsipursky, S.J. (Department of Geology, Arizona State University, Tempe, Arizona 85287 (United States))
1993-10-10
Fourier transform mass spectrometry (FTMS) employing both laser desorption/ionization and thermal desorption/electron ionization is useful for the detection and structural characterization of fullerenes and chemically-modified fullerenes. Examination of a carbon-rich shungite rock sample from Russia by transmission electron microscopy and FTMS provided evidence of naturally-occurring fullerenes. Ion-molecule reactions can be studied with FTMS to investigate the electron affinities of modified fullerenes. By monitoring charge exchange reactions, the electron affinities of C[sub 60]F[sub x] (x=44,46) and C[sub 70]F[sub y] (y=52,54) were found to be substantially higher than the values for the parent fullerenes.
Spherical T-Duality and the spherical Fourier-Mukai transform
Bouwknegt, Peter; Mathai, Varghese
2015-01-01
In [arxiv:1405.5844], we introduced spherical T-duality, which relates pairs of the form $(P,H)$ consisting of a principal $SU(2)$-bundle $P\\rightarrow M$ and a 7-cocycle $H$ on $P$. Intuitively, spherical T-duality exchanges $H$ with the second Chern class $c_2(P)$. Unless $\\mathrm{dim}(M)\\leq 4$, not all pairs admit spherical T-duals and the spherical T-duals are not always unique. In this paper, we define a canonical spherical Poincar\\'e vector bundle $\\mathcal P$ on $SU(2)\\times SU(2)$ and the spherical Fourier-Mukai transform, which implements a degree shifting isomorphism in K-theory on the trivial $SU(2)$-bundle with trivial 7-flux, and then (partially) generalise it to prove that all spherical T-dualities induce a natural degree-shifting isomorphism on the 7-twisted K-theories of the bundles when $\\mathrm{dim}(M)\\leq 4$.
A JOINT TIMING OFFSET AND CHANNEL ESTIMATION USING FRACTIONAL FOURIER TRANSFORM AND CAZAC SEQUENCES
Directory of Open Access Journals (Sweden)
Cherif Rezgui
2016-03-01
Full Text Available This paper deals with symbol timing offset and channel estimation in OFDM (orthogonal frequency division multiplexing system in fast varying channel. Symbol timing offset (STO estimation is a major task in OFDM. Most of existing methods for estimating STO used cyclic prefix or training sequences. In this paper, we consider a new system for STO estimation using constant amplitude zero autocorrelation (CAZAC sequences as pilot sequences in conjunction with fractional Fourier transform (FRFT. After STO estimation is done, timing compensation is made. Thereafter, channel is estimated to well recover the original transmitted signal. This method gives good results in terms of MSE in comparison with other known techniques, it estimated well the channel and it is important for fast varying channel. MATLAB Monte-Carlo simulations are used to evaluate the performance of the proposed estimator.
Relative-coordinate determination for visual double stars by applying Fourier transforms
Directory of Open Access Journals (Sweden)
Radović Viktor
2013-01-01
Full Text Available We discuss the software developed for the purpose of determining the relative coordinates (position angle θ and separation ρ for visual double or multiple stars. It is based on application of Fourier transforms in treating CCD frames of these systems. The objective was to determine the relative coordinates automatically to an extent as large as possible. In this way the time needed for the reduction of many CCD frames becomes shorter. The capabilities and limitations of the software are examined. Besides, the possibility of improving is also considered. The software has been tested and checked on a sample consisting of CCD frames of 165 double or multiple stars obtained with the 2m telescope at NAO Rozhen in Bulgaria in October 2011. The results have been compared with the corresponding results obtained by applying different software and the agreement is found to be very good.
Label-free identification of individual bacteria using Fourier transform light scattering
Jo, YoungJu; Kim, Min-hyeok; Park, HyunJoo; Kang, Suk-Jo; Park, YongKeun
2015-01-01
Rapid identification of bacterial species is crucial in medicine and food hygiene. In order to achieve rapid and label-free identification of bacterial species at the single bacterium level, we propose and experimentally demonstrate an optical method based on Fourier transform light scattering (FTLS) measurements and statistical classification. For individual rod-shaped bacteria belonging to four bacterial species (Listeria monocytogenes, Escherichia coli, Lactobacillus casei, and Bacillus subtilis), two-dimensional angle-resolved light scattering maps are precisely measured using FTLS technique. The scattering maps are then systematically analyzed, employing statistical classification in order to extract the unique fingerprint patterns for each species, so that a new unidentified bacterium can be identified by a single light scattering measurement. The single-bacterial and label-free nature of our method suggests wide applicability for rapid point-of-care bacterial diagnosis.
Singular Spectrum Analysis: A Note on Data Processing for Fourier Transform Hyperspectral Imagers.
Rafert, J Bruce; Zabalza, Jaime; Marshall, Stephen; Ren, Jinchang
2016-09-01
Hyperspectral remote sensing is experiencing a dazzling proliferation of new sensors, platforms, systems, and applications with the introduction of novel, low-cost, low-weight sensors. Curiously, relatively little development is now occurring in the use of Fourier transform (FT) systems, which have the potential to operate at extremely high throughput without use of a slit or reductions in both spatial and spectral resolution that thin film based mosaic sensors introduce. This study introduces a new physics-based analytical framework called singular spectrum analysis (SSA) to process raw hyperspectral imagery collected with FT imagers that addresses some of the data processing issues associated with the use of the inverse FT. Synthetic interferogram data are analyzed using SSA, which adaptively decomposes the original synthetic interferogram into several independent components associated with the signal, photon and system noise, and the field illumination pattern.
A device for fluorescence temperature measurement based on fast fourier transform
Institute of Scientific and Technical Information of China (English)
WANG Dong-sheng; WANG Gui-mei; PAN Wei-wei; WANG Yu-tian
2008-01-01
A sapphire fiber thermal probe with Cr3+ion-doped end was grown using the laser heated pedestal method.The fluores-ccnce themal probe offers advantages of compact structure,high performance and the ability to sustain high temperature from the room temperature to 450℃.Based on the fast fourier transform(FFT),the fluorescence lifetime is obtained from the tangent function of the phase angle of the first non-zeroth item of FFT result.Compared with other traditional fitting methods,our method has advantages such as fast speed,high accuracy and being free from the influence of the base signal.The standard deviation of FFT method is about half of that of the Prony method and close to the one of the Marquardt method.In addition.since the FFT method is immunity to the background noise of the signal,the background noise analysis can be skipped.
Johansson, Alexandra C; Khodabakhsh, Amir; Foltynowicz, Aleksandra
2016-01-01
We present a thorough analysis of the signal line shapes of Fourier transform-based noise-immune cavity-enhanced optical frequency comb spectroscopy (NICE-OFCS). We discuss the signal dependence on the ratio of the modulation frequency, f${_m}$, to the molecular line width, {\\Gamma}. We compare a full model of the signals and a simplified absorption-like analytical model that has high accuracy for low f${_m}$/{\\Gamma} ratios and is much faster to compute. We verify the theory experimentally by measuring and fitting NICE-OFCS spectra of CO${_2}$ at 1575 nm using a system based on an Er:fiber femtosecond laser and a cavity with a finesse of ~11000.
Institute of Scientific and Technical Information of China (English)
XIE Yi-bin; LIU Qian; HE Fei; GUO Chun-guang; WANG Cheng-feng; ZHAO Ping
2011-01-01
Background Fourier transform infrared spectroscopy (FT-IR) combined with chemometrics discriminant analysis technology could improve diagnosis. The present study aimed to evaluate the effects of FT-IR on malignant colon tissue samples in diagnosis of colon cancer.Methods Principal component analysis (PCA) and support vector machine classification were used to discriminate FT-IR spectra from malignant and normal tissue. Colon tissues samples from 85 patients were used to demonstrate the procedure.Results For this set of colon spectral data, the sensitivity and specificity of the support vector machine (SVM)classification were found both higher than 90%.Conclusions FT-IR provided important information about cancerous tissue, which could be used to discriminate malignant from normal tissues. The combination of PCA and SVM classification indicated that FT-IR has a potential clinical application in diagnosis of colon cancer.
International Nuclear Information System (INIS)
The electronic part of the Mesooptical Fourier-Transform Microscope (MFTM) and the software for the automatic recording of the straight line particle tracks of minimum ionization in nuclear emulsion are described. The features of the data at the output of the MFTM depending on the geometrical position of a straight line particle track and the algorithm of the read-out of this data are given. The units moving the carriages and rotating the platform of the optical interface of the MFTM are described. The programming supplements to the standard FOKAL'71 subroutines used in these experiments as well as the routine programme used for the evaluation of the mesooptical images of straight line particle tracks are presented
Exhibition of the periodicity of Quantum Fourier Transformation in Nuclear Magnetic Resonance
Peng, X; Fang, X; Feng, M; Yang, X; Liu, M; Gao, K; Peng, Xinhua; Zhu, Xiwen; Fang, Ximing; Feng, Mang; Yang, Xiaodong; Liu, Maili; Gao, Kelin
2002-01-01
The remarkable capability of quantum Fourier transformation (QFT) to extract the periodicity of a given periodic function has been exhibited by using nuclear magnetic resonance (NMR) techniques. Two separate sets of experiments were performed. In a full QFT, the periodicity were validated with state tomography and fidelity measurements. For a simplified QFT, the three-qubit pseudo-pure state was created by introducting an additional observer spin, and the spectra recorded on the observer spin showed intuitively the power of QFT\\ to find the periodicity. Experimentally realizing the QFT provides a critical step to implement the renowned Shor's quantum factoring algorithm and many other algorithms. Moveover, it can be applied to the study of quantum chaos and other quantum information processing.
Submillimeter fourier-transform spectrometer measurements of atmospheric opacity above mauna kea.
Serabyn, E; Weisstein, E W; Lis, D C; Pardo, J R
1998-04-20
We present accurately calibrated submillimeter atmospheric transmission spectra obtained with a Fourier-transform spectrometer at the Caltech Submillimeter Observatory on Mauna Kea, Hawaii. These measurements cover the 0.9-0.3-mm wavelength range and are the first in a series aimed at defining the terrestrial long-wave atmospheric transmission curve. The 4.1-km altitude of the Mauna Kea site provides access to extremely low zenith water-vapor columns, permitting atmospheric observations at frequencies well above those possible from sea level. We describe the calibration procedures, present our first well-calibrated transmission spectra, and compare our results with those of a single-layer atmospheric transmission model, AT. With an empirical best-fit continuum opacity term included, this simple single-layer model provides a remarkably good fit to the opacity data for H(2)O line profiles described by either van Vleck-Weisskopf or kinetic shapes.
Accelerating the 2-point and 3-point galaxy correlation functions using Fourier transforms
Slepian, Zachary
2015-01-01
Though Fourier Transforms (FTs) are a common technique for finding correlation functions, they are not typically used in computations of the anisotropy of the two-point correlation function (2PCF) about the line of sight in wide-angle surveys because the line-of-sight direction is not constant on the Cartesian grid. Here we show how FTs can be used to compute the multipole moments of the anisotropic 2PCF. We also show how FTs can be used to accelerate the 3PCF algorithm of Slepian & Eisenstein (2015). In both cases, these FT methods allow one to avoid the computational cost of pair counting, which scales as the square of the number density of objects in the survey. With the upcoming large datasets of DESI, Euclid, and LSST, FT techniques will therefore offer an important complement to simple pair or triplet counts.
An Archive of Spectra from the Mayall Fourier Transform Spectrometer at Kitt Peak
Pilachowski, C A; Young, M D; Dennis, H B; Gopu, A; Henschel, R; Hayashi, S
2016-01-01
We describe the SpArc science gateway for spectral data obtained during the period from 1975 through 1995 at the Kitt Peak National Observatory using the Fourier Transform Spectrometer (FTS) in operation at the Mayall 4-m telescope. SpArc is hosted by Indiana University Bloomington and is available for public access. The archive includes nearly 10,000 individual spectra of more than 800 different astronomical sources including stars, nebulae, galaxies, and Solar System objects. We briefly describe the FTS instrument itself, and summarize the conversion of the original interferograms into spectral data and the process for recovering the data into FITS files. The architecture of the archive is discussed, and the process for retrieving data from the archive is introduced. Sample use cases showing typical FTS spectra are presented.