Mitigating voltage lead errors of an AC Josephson voltage standard by impedance matching
Zhao, Dongsheng; van den Brom, Helko E.; Houtzager, Ernest
2017-09-01
A pulse-driven AC Josephson voltage standard (ACJVS) generates calculable AC voltage signals at low temperatures, whereas measurements are performed with a device under test (DUT) at room temperature. The voltage leads cause the output voltage to show deviations that scale with the frequency squared. Error correction mechanisms investigated so far allow the ACJVS to be operational for frequencies up to 100 kHz. In this paper, calculations are presented to deal with these errors in terms of reflected waves. Impedance matching at the source side of the system, which is loaded with a high-impedance DUT, is proposed as an accurate method to mitigate these errors for frequencies up to 1 MHz. Simulations show that the influence of non-ideal component characteristics, such as the tolerance of the matching resistor, the capacitance of the load input impedance, losses in the voltage leads, non-homogeneity in the voltage leads, a non-ideal on-chip connection and inductors between the Josephson junction array and the voltage leads, can be corrected for using the proposed procedures. The results show that an expanded uncertainty of 12 parts in 106 (k = 2) at 1 MHz and 0.5 part in 106 (k = 2) at 100 kHz is within reach.
Franek, James; Brandt, Steven; Berger, Birk; Liese, Martin; Barthel, Matthias; Schüngel, Edmund; Schulze, Julian
2015-05-01
We present a novel radio-frequency (RF) power supply and impedance matching to drive technological plasmas with customized voltage waveforms. It is based on a system of phase-locked RF generators that output single frequency voltage waveforms corresponding to multiple consecutive harmonics of a fundamental frequency. These signals are matched individually and combined to drive a RF plasma. Electrical filters are used to prevent parasitic interactions between the matching branches. By adjusting the harmonics' phases and voltage amplitudes individually, any voltage waveform can be approximated as a customized finite Fourier series. This RF supply system is easily adaptable to any technological plasma for industrial applications and allows the commercial utilization of process optimization based on voltage waveform tailoring for the first time. Here, this system is tested on a capacitive discharge based on three consecutive harmonics of 13.56 MHz. According to the Electrical Asymmetry Effect, tuning the phases between the applied harmonics results in an electrical control of the DC self-bias and the mean ion energy at almost constant ion flux. A comparison with the reference case of an electrically asymmetric dual-frequency discharge reveals that the control range of the mean ion energy can be significantly enlarged by using more than two consecutive harmonics.
Wang, Yonggang; Tong, Liqing; Liu, Kefu
2017-06-01
The purpose of impedance matching for a Marx generator and DBD lamp is to limit the output current of the Marx generator, provide a large discharge current at ignition, and obtain fast voltage rising/falling edges and large overshoot. In this paper, different impedance matching circuits (series inductor, parallel capacitor, and series inductor combined with parallel capacitor) are analyzed. It demonstrates that a series inductor could limit the Marx current. However, the discharge current is also limited. A parallel capacitor could provide a large discharge current, but the Marx current is also enlarged. A series inductor combined with a parallel capacitor takes full advantage of the inductor and capacitor, and avoids their shortcomings. Therefore, it is a good solution. Experimental results match the theoretical analysis well and show that both the series inductor and parallel capacitor improve the performance of the system. However, the series inductor combined with the parallel capacitor has the best performance. Compared with driving the DBD lamp with a Marx generator directly, an increase of 97.3% in radiant power and an increase of 59.3% in system efficiency are achieved using this matching circuit.
Impedance-matched Marx generators
Directory of Open Access Journals (Sweden)
W. A. Stygar
2017-04-01
Full Text Available We have conceived a new class of prime-power sources for pulsed-power accelerators: impedance-matched Marx generators (IMGs. The fundamental building block of an IMG is a brick, which consists of two capacitors connected electrically in series with a single switch. An IMG comprises a single stage or several stages distributed axially and connected in series. Each stage is powered by a single brick or several bricks distributed azimuthally within the stage and connected in parallel. The stages of a multistage IMG drive an impedance-matched coaxial transmission line with a conical center conductor. When the stages are triggered sequentially to launch a coherent traveling wave along the coaxial line, the IMG achieves electromagnetic-power amplification by triggered emission of radiation. Hence a multistage IMG is a pulsed-power analogue of a laser. To illustrate the IMG approach to prime power, we have developed conceptual designs of two ten-stage IMGs with LC time constants on the order of 100 ns. One design includes 20 bricks per stage, and delivers a peak electrical power of 1.05 TW to a matched-impedance 1.22-Ω load. The design generates 113 kV per stage and has a maximum energy efficiency of 89%. The other design includes a single brick per stage, delivers 68 GW to a matched-impedance 19-Ω load, generates 113 kV per stage, and has a maximum energy efficiency of 90%. For a given electrical-power-output time history, an IMG is less expensive and slightly more efficient than a linear transformer driver, since an IMG does not use ferromagnetic cores.
Impedance-matched Marx generators
Stygar, W. A.; LeChien, K. R.; Mazarakis, M. G.; Savage, M. E.; Stoltzfus, B. S.; Austin, K. N.; Breden, E. W.; Cuneo, M. E.; Hutsel, B. T.; Lewis, S. A.; McKee, G. R.; Moore, J. K.; Mulville, T. D.; Muron, D. J.; Reisman, D. B.; Sceiford, M. E.; Wisher, M. L.
2017-04-01
We have conceived a new class of prime-power sources for pulsed-power accelerators: impedance-matched Marx generators (IMGs). The fundamental building block of an IMG is a brick, which consists of two capacitors connected electrically in series with a single switch. An IMG comprises a single stage or several stages distributed axially and connected in series. Each stage is powered by a single brick or several bricks distributed azimuthally within the stage and connected in parallel. The stages of a multistage IMG drive an impedance-matched coaxial transmission line with a conical center conductor. When the stages are triggered sequentially to launch a coherent traveling wave along the coaxial line, the IMG achieves electromagnetic-power amplification by triggered emission of radiation. Hence a multistage IMG is a pulsed-power analogue of a laser. To illustrate the IMG approach to prime power, we have developed conceptual designs of two ten-stage IMGs with L C time constants on the order of 100 ns. One design includes 20 bricks per stage, and delivers a peak electrical power of 1.05 TW to a matched-impedance 1.22 -Ω load. The design generates 113 kV per stage and has a maximum energy efficiency of 89%. The other design includes a single brick per stage, delivers 68 GW to a matched-impedance 19 -Ω load, generates 113 kV per stage, and has a maximum energy efficiency of 90%. For a given electrical-power-output time history, an IMG is less expensive and slightly more efficient than a linear transformer driver, since an IMG does not use ferromagnetic cores.
Fractional Order Element Based Impedance Matching
Radwan, Ahmed Gomaa
2014-06-24
Disclosed are various embodiments of methods and systems related to fractional order element based impedance matching. In one embodiment, a method includes aligning a traditional Smith chart (|.alpha.|=1) with a fractional order Smith chart (|.alpha.|.noteq.1). A load impedance is located on the traditional Smith chart and projected onto the fractional order Smith chart. A fractional order matching element is determined by transitioning along a matching circle of the fractional order Smith chart based at least in part upon characteristic line impedance. In another embodiment, a system includes a fractional order impedance matching application executed in a computing device. The fractional order impedance matching application includes logic that obtains a first set of Smith chart coordinates at a first order, determines a second set of Smith chart coordinates at a second order, and determines a fractional order matching element from the second set of Smith chart coordinates.
Fractional Order Element Based Impedance Matching
Radwan, Ahmed Gomaa; Salama, Khaled N.; Shamim, Atif
2014-01-01
Disclosed are various embodiments of methods and systems related to fractional order element based impedance matching. In one embodiment, a method includes aligning a traditional Smith chart (|.alpha.|=1) with a fractional order Smith chart (|.alpha
Broadband electrical impedance matching for piezoelectric ultrasound transducers.
Huang, Haiying; Paramo, Daniel
2011-12-01
This paper presents a systematic method for designing broadband electrical impedance matching networks for piezoelectric ultrasound transducers. The design process involves three steps: 1) determine the equivalent circuit of the unmatched piezoelectric transducer based on its measured admittance; 2) design a set of impedance matching networks using a computerized Smith chart; and 3) establish the simulation model of the matched transducer to evaluate the gain and bandwidth of the impedance matching networks. The effectiveness of the presented approach is demonstrated through the design, implementation, and characterization of impedance matching networks for a broadband acoustic emission sensor. The impedance matching network improved the power of the acquired signal by 9 times.
Impedance matched thin metamaterials make metals absorbing.
Mattiucci, N; Bloemer, M J; Aközbek, N; D'Aguanno, G
2013-11-13
Metals are generally considered good reflectors over the entire electromagnetic spectrum up to their plasma frequency. Here we demonstrate an approach to tailor their absorbing characteristics based on the effective metamaterial properties of thin, periodic metallo-dielectric multilayers by exploiting a broadband, inherently non-resonant, surface impedance matching mechanism. Based on this mechanism, we design, fabricate and test omnidirectional, thin ( 99%) over a frequency range spanning from the UV to the IR. Our approach opens new venues to design cost effective materials for many applications such as thermo-photovoltaic energy conversion devices, light harvesting for solar cells, flat panel display, infrared detectors, stray light reduction, stealth and others.
Study on electrical impedance matching for broadband ultrasonic transducer
Energy Technology Data Exchange (ETDEWEB)
Kim, Geon Woo [University of Science and Technology, Daejeon (Korea, Republic of); Kim, Ki Bok [Korea Research Institute of Standards and Science, Center for Safety Measurement, Daejeon (Korea, Republic of); Baek, Kwang Sae [Elache Co., Busan (Korea, Republic of)
2017-02-15
Ultrasonic transducers with high resolution and resonant frequency are required to detect small defects (less than hundreds of μm) by ultrasonic testing. The resonance frequency and resolution of an ultrasonic transducer are closely related to the thickness of piezo-electric materials, backing materials, and the electric impedance matching technique. Among these factors, electrical impedance matching plays an important role because it can reduce the loss and reflection of ultrasonic energy differences in electrical impedance between an ultrasonic transducer and an ultrasonic defects detecting system. An LC matching circuit is the most frequently used electric matching method. It is necessary for the electrical impedance of an ultrasonic transducer to correspond to approximately 50 Ω to compensate the difference in electrical impedance between both connections. In this study, a 15 MHz immersion ultrasonic transducer was fabricated and an LC electrical impedance circuit was applied to that for having broad-band frequency characteristic.
Theory of fractional order elements based impedance matching networks
Radwan, Ahmed G.
2011-03-01
Fractional order circuit elements (inductors and capacitors) based impedance matching networks are introduced for the first time. In comparison to the conventional integer based L-type matching networks, fractional matching networks are much simpler and versatile. Any complex load can be matched utilizing a single series fractional element, which generally requires two elements for matching in the conventional approach. It is shown that all the Smith chart circles (resistance and reactance) are actually pairs of completely identical circles. They appear to be single for the conventional integer order case, where the identical circles completely overlap each other. The concept is supported by design equations and impedance matching examples. © 2010 IEEE.
Theory of fractional order elements based impedance matching networks
Radwan, Ahmed G.; Shamim, Atif; Salama, Khaled N.
2011-01-01
pairs of completely identical circles. They appear to be single for the conventional integer order case, where the identical circles completely overlap each other. The concept is supported by design equations and impedance matching examples. © 2010 IEEE.
An impact analysis of the fault impedance on voltage sags
Energy Technology Data Exchange (ETDEWEB)
Ramos, Alessandro Candido Lopes [CELG - Companhia Energetica de Goias, Goiania, GO (Brazil). Generation and Transmission. System' s Operation Center], E-mail: alessandro.clr@celg.com.br; Batista, Adalberto Jose [Federal University of Goias (UFG), Goiania, GO (Brazil)], E-mail: batista@eee.ufg.br; Leborgne, Roberto Chouhy [Federal University of Rio Grande do Sul (UFRGS), Porto Alegre, RS (Brazil)], E-mail: rcl@ece.ufrgs.br; Emiliano, Pedro Henrique Mota, E-mail: ph@phph.com.br
2009-07-01
This paper presents an impact analysis of the fault impedance, in terms of its module and angle, on voltage sags caused by faults. Symmetrical and asymmetrical faults are simulated, at transmission and distribution lines, by using a frequency-domain fault simulation software called ANAFAS. Voltage sags are monitored at buses where sensitive end-users are connected. In order to overcome some intrinsic limitations of this software concerning its automatic execution for several cases, a computational tool was developed in Java programming language. This solution allows the automatic simulation of cases including the effect of the fault position, the fault type, and the proper fault impedance. The main conclusion is that the module and angle of the fault impedance can have a significant influence on voltage sag depending on the fault characteristics. (author)
Impedance matching through a single passive fractional element
Radwan, Ahmed Gomaa
2012-07-01
For the first time, a generalized admittance Smith chart theory is introduced to represent fractional order circuit elements. The principles of fractional order matching circuits are described. We show that for fractional order α < 1, a single parallel fractional element can match a wider range of load impedances as compared to its series counterpart. Several matching examples demonstrate the versatility of fractional order series and parallel element matching as compared to the conventional approach. © 2012 IEEE.
Development of impedance matching technologies for ICRF antenna arrays
International Nuclear Information System (INIS)
Pinsker, R.I.
1998-01-01
All high-power ion cyclotron range of frequency (ICRF) heating systems include devices for matching the input impedance of the antenna array to the generator output impedance. For most types of antennas used, the input impedance is strongly time dependent on timescales as rapid as 10 -1 s, while the radio frequency (RF) generators used are capable of producing full power only into a stationary load impedance. Hence, the dynamic response of the matching method is of great practical importance. In this paper, world-wide developments in this field over the past decade are reviewed. These techniques may be divided into several classes. The edge plasma parameters that determine the antenna array's input impedance may be controlled to maintain a fixed load impedance. The frequency of the RF source can be feedback controlled to compensate for changes in the edge plasma conditions, or fast variable tuning elements in the transmission line between the generator output and the antenna input connections can provide the necessary time-varying impedance transformation. In 'lossy passive schemes', reflected power due to the time-varying impedance of the antenna array is diverted to a dummy load. Each of these techniques can be applied to a pre-existing antenna system. If a new antenna is to be designed, recent advances allow the antenna array to have the intrinsic property of presenting a constant load to the feeding transmission lines despite the varying load seen by each antenna in the array. (author)
Development of impedance matching technologies for ICRF antenna arrays
International Nuclear Information System (INIS)
Pinsker, R.I.
1998-03-01
All high power ICRF heating systems include devices for matching the input impedance of the antenna array to the generator output impedance. For most types of antennas used, the input impedance is strongly time-dependent on timescales as rapid as 10-4 s, while the rf generators used are capable of producing full power only into a stationary load impedance. Hence, the dynamic response of the matching method is of great practical importance. In this paper, world-wide developments in this field over the past decade are reviewed. These techniques may be divided into several classes. The edge plasma parameters that determine the antenna array's input impedance may be controlled to maintain a fixed load impedance. The frequency of the rf source can be feedback controlled to compensate for changes in the edge plasma conditions, or fast variable tuning elements in the transmission line between the generator output and the antenna input connections can provide the necessary time-varying impedance transformation. In lossy passive schemes, reflected power due to the time-varying impedance of the antenna array is diverted to a dummy load. Each of these techniques can be applied to a pre-existing antenna system. If a new antenna is to be designed, recent advances allow the antenna array to have the intrinsic property of presenting a constant load to the feeding transmission lines despite the varying load seen by each antenna in the array
Impedance Conjugate Matching of Plasmonic Nanoantenna in Optical Nanocircuits
DEFF Research Database (Denmark)
Sachkou, Yauhen; Andryieuski, Andrei; Lavrinenko, Andrei
2011-01-01
-shaped optical nanoantenna on its geometrical parameters. We show that several parameters – an arm length of the T-shaped nanoantenna, nanoantenna width and a width of the front facet of the connector – can influence on the impedance conjugate matching which gives flexibility in impedance tuning for coupling......Optical antennas are a state-of-the-art concept in modern plasmonics. Nanoantennas can be advantageously used to localize, enhance and detect radiation in nanoscale volumes. These abilities unlock an enormous potential applications ranging from optoelectronics engineering and integrated optical...... nanocircuitry to nanoscale optical microscopy and ultra-sensing. The coupling efficiency between nanoantenna and other elements in integrated optical nanocircuit strongly depends on its properties where the impedance matching plays a crucial role. We investigate the dependence of impedance of the T...
The frequency characteristics of medium voltage distribution system impedances
Directory of Open Access Journals (Sweden)
Liviu Emil Petrean
2009-10-01
Full Text Available In this paper we present the frequency characteristics of impedances involved in the electrical equivalent circuit of a large medium voltage distribution system. These impedances influence harmonics distortions propagation occurring due to the nonsinusoidal loads. We analyse the case of a 10 kV large urban distribution system which supplies industrial, commercial and residential customers. The influence of various parameters of the distribution network on the frequency characteristics are presented, in order to assess the interaction of harmonic distortion and distribution system network.
Impedance-match experiments using high intensity lasers
International Nuclear Information System (INIS)
Holmes, N.C.; Trainor, R.J.; Anderson, R.A.; Veeser, L.R.; Reeves, G.A.
1981-01-01
The results of a series of impedance-match experiments using copper-aluminum targets irradiated using the Janus Laser Facility are discussed. The results are compared to extrapolations of data obtained at lower pressures using impact techniques. The sources of errors are described and evaluated. The potential of lasers for high accuracy equation of state investigations are discussed
Impedance Matching Antenna-Integrated High-Efficiency Energy Harvesting Circuit
Shinki, Yuharu; Shibata, Kyohei; Mansour, Mohamed
2017-01-01
This paper describes the design of a high-efficiency energy harvesting circuit with an integrated antenna. The circuit is composed of series resonance and boost rectifier circuits for converting radio frequency power into boosted direct current (DC) voltage. The measured output DC voltage is 5.67 V for an input of 100 mV at 900 MHz. Antenna input impedance matching is optimized for greater efficiency and miniaturization. The measured efficiency of this antenna-integrated energy harvester is 60% for −4.85 dBm input power and a load resistance equal to 20 kΩ at 905 MHz. PMID:28763043
Impedance Matching Antenna-Integrated High-Efficiency Energy Harvesting Circuit
Directory of Open Access Journals (Sweden)
Yuharu Shinki
2017-08-01
Full Text Available This paper describes the design of a high-efficiency energy harvesting circuit with an integrated antenna. The circuit is composed of series resonance and boost rectifier circuits for converting radio frequency power into boosted direct current (DC voltage. The measured output DC voltage is 5.67 V for an input of 100 mV at 900 MHz. Antenna input impedance matching is optimized for greater efficiency and miniaturization. The measured efficiency of this antenna-integrated energy harvester is 60% for −4.85 dBm input power and a load resistance equal to 20 kΩ at 905 MHz.
Impedance Matching Antenna-Integrated High-Efficiency Energy Harvesting Circuit.
Shinki, Yuharu; Shibata, Kyohei; Mansour, Mohamed; Kanaya, Haruichi
2017-08-01
This paper describes the design of a high-efficiency energy harvesting circuit with an integrated antenna. The circuit is composed of series resonance and boost rectifier circuits for converting radio frequency power into boosted direct current (DC) voltage. The measured output DC voltage is 5.67 V for an input of 100 mV at 900 MHz. Antenna input impedance matching is optimized for greater efficiency and miniaturization. The measured efficiency of this antenna-integrated energy harvester is 60% for -4.85 dBm input power and a load resistance equal to 20 kΩ at 905 MHz.
Impedance matching wireless power transmission system for biomedical devices.
Lum, Kin Yun; Lindén, Maria; Tan, Tian Swee
2015-01-01
For medical application, the efficiency and transmission distance of the wireless power transfer (WPT) are always the main concern. Research has been showing that the impedance matching is one of the critical factors for dealing with the problem. However, there is not much work performed taking both the source and load sides into consideration. Both sides matching is crucial in achieving an optimum overall performance, and the present work proposes a circuit model analysis for design and implementation. The proposed technique was validated against experiment and software simulation. Result was showing an improvement in transmission distance up to 6 times, and efficiency at this transmission distance had been improved up to 7 times as compared to the impedance mismatch system. The system had demonstrated a near-constant transfer efficiency for an operating range of 2cm-12cm.
Bonding and impedance matching of acoustic transducers using silver epoxy.
Son, Kyu Tak; Lee, Chin C
2012-04-01
Silver epoxy was selected to bond transducer plates on glass substrates. The properties and thickness of the bonding medium affect the electrical input impedance of the transducer. Thus, the thickness of the silver epoxy bonding layer was used as a design parameter to optimize the structure for the transducer input impedance to match the 50 Ω output impedance of most radio frequency (RF) generators. Simulation and experimental results show that nearly perfect matching is achieved without using any matching circuit. At the matching condition, the transducer operates at a frequency band a little bit below the half-wavelength resonant frequency of the piezoelectric plate. In experiments, lead titanate (PT) piezoelectric plates were employed. Both full-size, 11.5 mm × 2 mm × 0.4 mm, and half-size, 5.75 mm × 2 mm × 0.4 mm, can be well matched using optimal silver epoxy thickness. The transducer assemblies demonstrate high efficiency. The conversion loss from electrical power to acoustic power in soda-lime glass is 4.3 dB. This loss is low considering the fact that the transducers operate at off-resonance by 12%. With proper choice of silver epoxy thickness, the transducer can be matched at the fundamental, the 3rd and 5th harmonic frequencies. This leads to the possible realization of triple-band transducers. Reliability was assessed with thermal cycling test according to Telcordia GR-468-Core recommendation. Of the 30 transducer assemblies tested, none broke until 2900 cycles and 27 have sustained beyond 4050 cycles. Copyright © 2011 Elsevier B.V. All rights reserved.
Directory of Open Access Journals (Sweden)
J. W. Horng
2010-12-01
Full Text Available A voltage-mode high input impedance first-order highpass, lowpass and allpass filters using two differential voltage current conveyors (DVCCs, one grounded capacitor and one grounded resistor is presented. The highpass, lowpass and allpass signals can be obtained simultaneously from the circuit configuration. The suggested filter uses a canonical number of passive components without requiring any component matching condition. The simulation results confirm the theoretical analysis.
Influence of the Antenna Impedance Variation and Input Matching Network Q on LNA Key Figures
DEFF Research Database (Denmark)
Ruaro, Andrea; Kvist, Søren Helstrup; Gülstorff, Steen
2012-01-01
In this paper we present an analysis of the behaviour of a 2:4 GHz Low Noise Amplifier (LNA) for Wireless Body Area Network (WBAN) applications facing antennadetuning issue. An amplifier with ultra-low power, low voltage, and with reduced component count is prototyped to validate simulation results......, then the behaviour is analyzed through simulations based on data measured on real users. The analysis shows that the designed LNA is stable to the antenna impedance variation expectable in most cases, while highlights the possible risks associated to a high-Q input matching network when used in a context prone...
Measurement scheme of kicker impedances via beam-induced voltages of coaxial cables
Energy Technology Data Exchange (ETDEWEB)
Shobuda, Yoshihiro, E-mail: yoshihiro.shobuda@j-parc.jp [J-PARC Center, JAEA and KEK, 2-4 Shirakata Shirane, Tokaimura, Nakagun, Ibaraki 319-1195 (Japan); Irie, Yoshiro [KEK, High Energy Accelerator Research Organization, 1-1 Oho, Tsukuba, Ibaraki 305-0801 (Japan); Toyama, Takeshi; Kamiya, Junichiro [J-PARC Center, JAEA and KEK, 2-4 Shirakata Shirane, Tokaimura, Nakagun, Ibaraki 319-1195 (Japan); Watanabe, Masao [Ministry of Education, Culture, Sports, Science and Technology, 3-2-2 Kasumigaseki, Chiyoda, Tokyo 100-8959 (Japan)
2013-06-11
A new theory, which satisfies the causality condition, is developed to describe impedances of kicker magnets with coaxial cables. The theoretical results well describe measurement results, which are obtained by standard wire methods. On the other hand, when beams pass through the kicker, voltages are induced at the terminals of coaxial cables. In other words, by analyzing the voltages, the kicker impedance for the beams can be obtained. The observed impedances are consistent with the theoretical results. The theory describes the impedance for non-relativistic beams, as well. The theoretical, simulation and measurement results indicate that the horizontal kicker impedance is drastically reduced by the non-relativistic effect. -- Highlights: ► We develop an innovative method to measure kicker impedance including power cable. ► By analyzing voltages at the ends of coaxial cables, the impedance is derived. ► The horizontal impedance is reduced as the beam becomes non-relativistic.
Phonon impedance matching: minimizing interfacial thermal resistance of thin films
Polanco, Carlos; Zhang, Jingjie; Ghosh, Avik
2014-03-01
The challenge to minimize interfacial thermal resistance is to allow a broad band spectrum of phonons, with non-linear dispersion and well defined translational and rotational symmetries, to cross the interface. We explain how to minimize this resistance using a frequency dependent broadening matrix that generalizes the notion of acoustic impedance to the whole phonon spectrum including symmetries. We show how to ``match'' two given materials by joining them with a single atomic layer, with a multilayer material and with a graded superlattice. Atomic layer ``matching'' requires a layer with a mass close to the arithmetic mean (or spring constant close to the harmonic mean) to favor high frequency phonon transmission. For multilayer ``matching,'' we want a material with a broadening close to the geometric mean to maximize transmission peaks. For graded superlattices, a continuous sequence of geometric means translates to an exponentially varying broadening that generates a wide-band antireflection coating for both the coherent and incoherent limits. Our results are supported by ``first principles'' calculations of thermal conductance for GaAs / Gax Al1 - x As / AlAs thin films using the Non-Equilibrium Greens Function formalism coupled with Density Functional Perturbation Theory. NSF-CAREER (QMHP 1028883), NSF-IDR (CBET 1134311), XSEDE.
DEFF Research Database (Denmark)
Savaghebi, Mehdi; Shafiee, Qobad; Vasquez, Juan Carlos
2015-01-01
and current inner control loops in order to fix the filter capacitor voltage and a virtual impedance loop mainly for voltage harmonics and unbalance compensation. The virtual impedance is set by the central secondary controller to mitigate the voltage distortion at sensitive load bus (SLB). Secondary...... controller is connected to a measurement unit to obtain the data of voltage harmonics and unbalance at microgrid SLB and broadcasts the commands for adjusting the virtual impedance of each unit. A general case with a combined voltage harmonic and unbalance distortion is considered. In such a case, voltage...... distortion is mitigated by selective insertion of capacitive virtual impedances for negative sequence of fundamental component as well as positive and negative sequences of main harmonics. The values of virtual capacitances are determined based on the required voltage quality at the load bus; thus...
Application of an impedance matching transformer to a plasma focus.
Bures, B L; James, C; Krishnan, M; Adler, R
2011-10-01
A plasma focus was constructed using an impedance matching transformer to improve power transfer between the pulse power and the dynamic plasma load. The system relied on two switches and twelve transformer cores to produce a 100 kA pulse in short circuit on the secondary at 27 kV on the primary with 110 J stored. With the two transformer systems in parallel, the Thevenin equivalent circuit parameters on the secondary side of the driver are: C = 10.9 μF, V(0) = 4.5 kV, L = 17 nH, and R = 5 mΩ. An equivalent direct drive circuit would require a large number of switches in parallel, to achieve the same Thevenin equivalent. The benefits of this approach are replacement of consumable switches with non-consumable transformer cores, reduction of the driver inductance and resistance as viewed by the dynamic load, and reduction of the stored energy to produce a given peak current. The system is designed to operate at 100 Hz, so minimizing the stored energy results in less load on the thermal management system. When operated at 1 Hz, the neutron yield from the transformer matched plasma focus was similar to the neutron yield from a conventional (directly driven) plasma focus at the same peak current.
Selective Harmonic Virtual Impedance for Voltage Source Inverters with LCL filter in Microgrids
DEFF Research Database (Denmark)
Savaghebi, Mehdi; Vasquez, Juan Carlos; Jalilian, Alireza Jalilian
2012-01-01
This paper presents a new control approach for voltage source inverters ended with LCL filters for microgrid applications. The control approach consists of voltage and current inner control loops in order to fix the filter capacitor voltage and a virtual impedance loop. The virtual impedance...... is added in order to mitigate the voltage distortion after the output inductor and improve the load sharing among parallel inverters. A general case with a combined voltage harmonic and unbalance distortion is considered. In such a case, voltage distortion is mitigated by inserting capacitive virtual...... impedance for negative sequence of fundamental component as well as positive and negative sequences of main harmonic components. Furthermore, resistive virtual impedances are added at these components in order to provide a proper load sharing and make the overall system more damped. Simulation results...
Improved Decoupling for 13C coil Arrays Using Non-Conventional Matching and Preamplifier Impedance
DEFF Research Database (Denmark)
Sanchez, Juan Diego; Johansen, Daniel Højrup; Hansen, Rie Beck
In this study, we describe a method to obtain improved preamplifier decoupling for receive-only coils. The method relies on the better decoupling obtained when coils are matched to an impedance higher than 50 . Preamplifiers with inductive imaginary impedance and low real impedance, increase...
Unified Impedance Model of Grid-Connected Voltage-Source Converters
DEFF Research Database (Denmark)
Wang, Xiongfei; Harnefors, Lennart; Blaabjerg, Frede
2018-01-01
This paper proposes a unified impedance model of grid-connected voltage-source converters for analyzing dynamic influences of the Phase-Locked Loop (PLL) and current control. The mathematical relations between the impedance models in the different domains are first explicitly revealed by means...... of complex transfer functions and complex space vectors. A stationary (αβ-) frame impedance model is then proposed, which not only predicts the stability impact of the PLL, but reveals also its frequency coupling effect explicitly. Furthermore, the impedance shaping effect of the PLL on the current control...... results and theoretical analysis confirm the effectiveness of the stationary-frame impedance model....
Energy Technology Data Exchange (ETDEWEB)
Bauer, Stephan; Palafox, Luis [Physikalisch-Technische Bundesanstalt (PTB), Braunschweig (Germany). Arbeitsgruppe 2.63 ' ' Josephson-Effekt, Spannung' '
2016-09-15
This article first explains the fundamental principle of an impedance measurement bridge on the base of Josephson voltage normals, before both types of measurement bridges realized in the PTB are more precisely discussed.
A Frequency-Tracking and Impedance-Matching Combined System for Robust Wireless Power Transfer
Directory of Open Access Journals (Sweden)
Yanting Luo
2017-01-01
Full Text Available One of the greatest challenges to power embedded devices using magnetically coupled resonant wireless power transfer (WPT system is that the amount of power delivered to the load is very sensitive to load impedance variations. Previous adaptive impedance-matching (IM technologies have drawbacks because adding IM networks, relay coils, or other compensating components in the receiver-side will significantly increase the receiver size. In this paper, a novel frequency-tracking and impedance-matching combined system is proposed to improve the robustness of wireless power transfer for embedded devices. The characteristics of the improved WPT system are investigated theoretically based on the two-port network model. Simulation and experimental studies are carried out to validate the proposed system. The results suggest that the frequency-tracking and impedance-matching combined WPT system can quickly find the best matching points and maintain high power transmission efficiency and output power when the load impedance changes.
Impedance-based Analysis of DC Link Control in Voltage Source Rectifiers
DEFF Research Database (Denmark)
Lu, Dapeng; Wang, Xiongfei; Blaabjerg, Frede
2018-01-01
This paper analyzes the dynamics influences of the outer dc link control in the voltage source rectifiers based on the impedance model. The ac-dc interactions are firstly presented by means of full order small signal model in dq frame, which shows the input voltage and load condition are the two...
Energy Technology Data Exchange (ETDEWEB)
Ao, Hiroyuki, E-mail: hiroyuki.ao@j-parc.jp [Japan Atomic Energy Agency (JAEA), J-PARC Center, Accelerator Division, 2-4, Shirakara Shirane, Tokai, Naka, Ibaraki 319-1195 (Japan); Asano, Hiroyuki [Japan Atomic Energy Agency (JAEA), J-PARC Center, Accelerator Division, 2-4, Shirakara Shirane, Tokai, Naka, Ibaraki 319-1195 (Japan); Naito, Fujio [High Energy Accelerator Research Organization (KEK), 1-1, Oho, Tsukuba, Ibaraki 305-0801 (Japan); Ouchi, Nobuo; Tamura, Jun [Japan Atomic Energy Agency (JAEA), J-PARC Center, Accelerator Division, 2-4, Shirakara Shirane, Tokai, Naka, Ibaraki 319-1195 (Japan); Takata, Koji [High Energy Accelerator Research Organization (KEK), 1-1, Oho, Tsukuba, Ibaraki 305-0801 (Japan)
2014-02-11
Impedance matching of RF windows that minimizes the RF reflection is necessary to prevent localized standing waves between an RF window and a cavity, which may cause thermal and/or multipactoring issues. It has been observed that the impedance matching condition of the pillbox-type RF window, checked by voltage standing wave ratio (VSWR) measurement, depends on the manufacturing lot of the window ceramic disk made of 95% purity Al{sub 2}O{sub 3}. The present report proposes new procedures for impedance matching as follows: (i) The relative dielectric constant of the ceramic window is directly measured using the resonant frequency of a cavity made by temporarily combining the pillbox part of the RF window and two short-circuiting plates. (ii) The dimensions of the pillbox section including the ceramic disk are fixed on the basis of the measured relative dielectric constant. To confirm this procedure, three RF windows were fabricated using the same type of ceramic material, and successful impedance matching of these windows was performed (VSWR<1.05). The measured results also suggest that the relative dielectric constant increases linearly with increasing density and that the impedance matching condition is mainly affected by variations of the relative dielectric constant due to shrinkage of the alumina during sintering. -- Highlights: • We measured the relative dielectric constant of an RF window ceramic directly. • We used the circular TE011-mode frequency of the pillbox part of an RF window itself. • The dimensions of the pillbox part were fixed on the basis of the measurement result. • Three RF windows were fabricated, and VSWR <1.05 for these windows was performed. • The relative dielectric constant increases linearly with increasing ceramic density.
International Nuclear Information System (INIS)
Ao, Hiroyuki; Asano, Hiroyuki; Naito, Fujio; Ouchi, Nobuo; Tamura, Jun; Takata, Koji
2014-01-01
Impedance matching of RF windows that minimizes the RF reflection is necessary to prevent localized standing waves between an RF window and a cavity, which may cause thermal and/or multipactoring issues. It has been observed that the impedance matching condition of the pillbox-type RF window, checked by voltage standing wave ratio (VSWR) measurement, depends on the manufacturing lot of the window ceramic disk made of 95% purity Al 2 O 3 . The present report proposes new procedures for impedance matching as follows: (i) The relative dielectric constant of the ceramic window is directly measured using the resonant frequency of a cavity made by temporarily combining the pillbox part of the RF window and two short-circuiting plates. (ii) The dimensions of the pillbox section including the ceramic disk are fixed on the basis of the measured relative dielectric constant. To confirm this procedure, three RF windows were fabricated using the same type of ceramic material, and successful impedance matching of these windows was performed (VSWR<1.05). The measured results also suggest that the relative dielectric constant increases linearly with increasing density and that the impedance matching condition is mainly affected by variations of the relative dielectric constant due to shrinkage of the alumina during sintering. -- Highlights: • We measured the relative dielectric constant of an RF window ceramic directly. • We used the circular TE011-mode frequency of the pillbox part of an RF window itself. • The dimensions of the pillbox part were fixed on the basis of the measurement result. • Three RF windows were fabricated, and VSWR <1.05 for these windows was performed. • The relative dielectric constant increases linearly with increasing ceramic density
Functionally graded materials for impedance matching in elastic media
International Nuclear Information System (INIS)
Chen, Shi; Zhang, Yinhong; Hao, Changchun; Lin, Shuyu; Fu, Zhiqiang
2014-01-01
When functionally graded material layers are inserted between two impedance mismatching media, passbands with extremely large bandwidths can appear in these layered systems. An accurate and effective iterative method is developed to deal with these layered systems with extremely large layer number.
Functionally graded materials for impedance matching in elastic media
Energy Technology Data Exchange (ETDEWEB)
Chen, Shi; Zhang, Yinhong, E-mail: zyh5337@163.com; Hao, Changchun; Lin, Shuyu; Fu, Zhiqiang
2014-01-03
When functionally graded material layers are inserted between two impedance mismatching media, passbands with extremely large bandwidths can appear in these layered systems. An accurate and effective iterative method is developed to deal with these layered systems with extremely large layer number.
Design and test of voltage and current probes for EAST ICRF antenna impedance measurement
Jianhua, WANG; Gen, CHEN; Yanping, ZHAO; Yuzhou, MAO; Shuai, YUAN; Xinjun, ZHANG; Hua, YANG; Chengming, QIN; Yan, CHENG; Yuqing, YANG; Guillaume, URBANCZYK; Lunan, LIU; Jian, CHENG
2018-04-01
On the experimental advanced superconducting tokamak (EAST), a pair of voltage and current probes (V/I probes) is installed on the ion cyclotron radio frequency transmission lines to measure the antenna input impedance, and supplement the conventional measurement technique based on voltage probe arrays. The coupling coefficients of V/I probes are sensitive to their sizes and installing locations, thus they should be determined properly to match the measurement range of data acquisition card. The V/I probes are tested in a testing platform at low power with various artificial loads. The testing results show that the deviation of coupling resistance is small for loads R L > 2.5 Ω, while the resistance deviations appear large for loads R L phase measurement error is the more significant factor leads to deleterious results rather than the amplitude measurement error. To exclude the possible ingredients that may lead to phase measurement error, the phase detector can be calibrated in steady L-mode scenario and then use the calibrated data for calculation under H-mode cases in EAST experiments.
A Unified Impedance Model of Voltage-Source Converters with Phase-Locked Loop Effect
DEFF Research Database (Denmark)
Wang, Xiongfei; Harnefors, Lennart; Blaabjerg, Frede
2016-01-01
This paper proposes a unified impedance model for analyzing the effect of Phase-Locked Loop (PLL) on the stability of grid-connected voltage-source converters. In the approach, the dq-frame impedance model is transformed into the stationary αβ-frame by means of complex transfer functions...... and complex space vectors, which not only predicts the stability impact of the PLL, but reveals also its frequency coupling effect in the phase domain. Thus, the impedance models previously developed in the different domains can be unified. Moreover, the impedance shaping effects of PLL are structurally...... characterized for the current control in the rotating dq-frame and the stationary αβ-frame. Case studies based on the unified impedance model are presented, which are then verified in the time-domain simulations and experiments. The results closely correlate with the impedance-based analysis....
Monitoring voltage-sensitive membrane impedance change using radio frequency interrogation.
Dharia, Sameera; Rabbitt, Richard D
2010-01-01
Here we present a new technique to monitor dynamic conformational changes in voltage-sensitive membrane-bound proteins using radio frequency (RF) impedance measurements. Xenopus oocytes were transfected to express ShakerB-IR K(+) ion channels, and step changes in membrane potential were applied using two-electrode voltage clamp (TEVC). Simultaneously, bipolar extracellular electrodes were used to measure the RF electrical impedance across the cell (300 kHz - 1 MHz). RF current will either pass through the media, around the cell, or displace charge across the cell membrane. The change in displacement current in the cell membrane during voltage clamp resulted in measurable RF impedance change. RF impedance change during DC membrane depolarization was significantly greater in ShakerB-IR expressing oocytes than in endogenous controls at 300 kHz, 500 kHz and, to a lesser extent, 1 MHz. Since the RF were too high to modulate ShakerB-IR protein conformational state (e.g. open channel probability), impedance changes are interpreted as reflections of voltage-dependent protein conformation and associated biophysics such as ion-channel dipole interactions, fluctuations in bound water, or charged lipid head-group rotations.
Hybrid silicon mode-locked laser with improved RF power by impedance matching
Tossoun, Bassem; Derickson, Dennis; Srinivasan, Sudharsanan; Bowers, John
2015-02-01
We design and discuss an impedance matching solution for a hybrid silicon mode-locked laser diode (MLLD) to improve peak optical power coming from the device. In order to develop an impedance matching solution, a thorough measurement and analysis of the MLLD as a function of bias on each of the laser segments was carried out. A passive component impedance matching network was designed at the operating frequency of 20 GHz to optimize RF power delivery to the laser. The hybrid silicon laser was packaged together in a module including the impedance matching circuit. The impedance matching design resulted in a 6 dB (electrical) improvement in the detected modulation spectrum power, as well as approximately a 10 dB phase noise improvement, from the MLLD. Also, looking ahead to possible future work, we discuss a Step Recovery Diode (SRD) driven impulse generator, which wave-shapes the RF drive to achieve efficient injection. This novel technique addresses the time varying impedance of the absorber as the optical pulse passes through it, to provide optimum optical pulse shaping.
International Nuclear Information System (INIS)
Mohamadou, Youssoufa; Oh, Tong In; Wi, Hun; Sohal, Harsh; Farooq, Adnan; Woo, Eung Je; McEwan, Alistair Lee
2012-01-01
Current sources are widely used in bio-impedance spectroscopy (BIS) measurement systems to maximize current injection for increased signal to noise while keeping within medical safety specifications. High-performance current sources based on the Howland current pump with optimized impedance converters are able to minimize stray capacitance of the cables and setup. This approach is limited at high frequencies primarily due to the deteriorated output impedance of the constant current source when situated in a real measurement system. For this reason, voltage sources have been suggested, but they require a current sensing resistor, and the SNR reduces at low impedance loads due to the lower current required to maintain constant voltage. In this paper, we compare the performance of a current source-based BIS and a voltage source-based BIS, which use common components. The current source BIS is based on a Howland current pump and generalized impedance converters to maintain a high output impedance of more than 1 MΩ at 2 MHz. The voltage source BIS is based on voltage division between an internal current sensing resistor (R s ) and an external sample. To maintain high SNR, R s is varied so that the source voltage is divided more or less equally. In order to calibrate the systems, we measured the transfer function of the BIS systems with several known resistor and capacitor loads. From this we may estimate the resistance and capacitance of biological tissues using the least-squares method to minimize error between the measured transimpedance excluding the system transfer function and that from an impedance model. When tested on realistic loads including discrete resistors and capacitors, and saline and agar phantoms, the voltage source-based BIS system had a wider bandwidth of 10 Hz to 2.2 MHz with less than 1% deviation from the expected spectra compared to more than 10% with the current source. The voltage source also showed an SNR of at least 60 dB up to 2.2 MHz
Chen, Mingji; Wang, Changxian; Cheng, Xiaodong; Gong, Congcheng; Song, Weili; Yuan, Xujin; Fang, Daining
2018-04-01
The realization of an ideal invisible cloak implementing transformation optics is still missing. An impedance matching concept is implanted into transformation optics cloak to generate an impedance matching cloak (IMC) shell. In this work, it is proved that impedance matching structure reduces the cloaking structure’s disturbance to a propagating electromagnetic field and improves its invisibility measured by scattering field intensity. Such a cylindrical IMC shell is designed, fabricated with proposed rounded rectangular split-ring-resonators (RR-SRRs), and experimental measurements show the total scattering field of a perfect electric conductor (PEC) cylinder surrounded by an IMC shell is improved greatly compared to the PEC cylinder showing electromagnetic wave front ripple suppression and a considerable scattering shrinking effect. IMC shell backward scattering field is suppressed down to 7.29%, compared to the previous value of 86.7% due to its impedance matching character, and overall scattering field intensity shrinking is down to 19.3% compared to the previously realized value of 56.4%. Sideward scattering field recorded in the experiment also has a remarkable improvement compared to the PEC cylinder. The impedance matching concept might enlighten the realization of an ideal cloak and other novel electromagnetic cloaking and shielding structures.
High transmission acoustic focusing by impedance-matched acoustic meta-surfaces
Al Jahdali, Rasha
2016-01-19
Impedance is an important issue in the design of acoustic lenses because mismatched impedance is detrimental to real focusing applications. Here, we report two designs of acoustic lenses that focus acoustic waves in water and air, respectively. They are tailored by acoustic meta-surfaces, which are rigid thin plates decorated with periodically distributed sub-wavelength slits. Their respective building blocks are constructed from the coiling-up spaces in water and the layered structures in air. Analytic analysis based on coupled-mode theory and transfer matrix reveals that the impedances of the lenses are matched to those of the background media. With these impedance-matched acoustic lenses, we demonstrate the acoustic focusing effect by finite-element simulations.
High transmission acoustic focusing by impedance-matched acoustic meta-surfaces
Al Jahdali, Rasha; Wu, Ying
2016-01-01
Impedance is an important issue in the design of acoustic lenses because mismatched impedance is detrimental to real focusing applications. Here, we report two designs of acoustic lenses that focus acoustic waves in water and air, respectively. They are tailored by acoustic meta-surfaces, which are rigid thin plates decorated with periodically distributed sub-wavelength slits. Their respective building blocks are constructed from the coiling-up spaces in water and the layered structures in air. Analytic analysis based on coupled-mode theory and transfer matrix reveals that the impedances of the lenses are matched to those of the background media. With these impedance-matched acoustic lenses, we demonstrate the acoustic focusing effect by finite-element simulations.
DEFF Research Database (Denmark)
Micallef, Alexander; Apap, Maurice; Spiteri-Staines, Cyril
2013-01-01
Parallel inverters having LCL output filters cause voltage distortions at the point of common coupling (PCC) in islanded microgrids when non-linear loads are present. A capacitive virtual impedance loop could be used to provide selective harmonic compensation in islanded microgrids, instead of in...... resistance for selective harmonic compensation in islanded microgrids. Simulation results were given to show the suitability of the proposed algorithms in reducing the voltage harmonics at the PCC.......Parallel inverters having LCL output filters cause voltage distortions at the point of common coupling (PCC) in islanded microgrids when non-linear loads are present. A capacitive virtual impedance loop could be used to provide selective harmonic compensation in islanded microgrids, instead...... of introducing additional active or passive filters into the system that could compromise the stability of the microgrid. However, the performance of these compensation loops becomes degraded when a virtual resistance is introduced with the aim to improve the overall stability of the parallel inverters...
Dey, Arka; Dhar, Joydeep; Sil, Sayantan; Jana, Rajkumar; Ray, Partha Pratim
2018-04-01
In this report, bias voltage-dependent dielectric and electron transport properties of ZnS nanoparticles were discussed. ZnS nanoparticles were synthesized by introducing a modified hydrothermal process. The powder XRD pattern indicates the phase purity, and field emission scanning electron microscope image demonstrates the morphology of the synthesized sample. The optical band gap energy (E g = 4.2 eV) from UV measurement explores semiconductor behavior of the synthesized material. The electrical properties were performed at room temperature using complex impedance spectroscopy (CIS) technique as a function of frequency (40 Hz-10 MHz) under different forward dc bias voltages (0-1 V). The CIS analysis demonstrates the contribution of bulk resistance in conduction mechanism and its dependency on forward dc bias voltages. The imaginary part of the impedance versus frequency curve exhibits the existence of relaxation peak which shifts with increasing dc forward bias voltages. The dc bias voltage-dependent ac and dc conductivity of the synthesized ZnS was studied on thin film structure. A possible hopping mechanism for electrical transport processes in the system was investigated. Finally, it is worth to mention that this analysis of bias voltage-dependent dielectric and transport properties of as-synthesized ZnS showed excellent properties for emerging energy applications.
Diamagneto-Dielectric Anisotropic Wide Angle Impedance Matching Layers for Active Phased Arrays
Silvestri, F.; Cifola, L.; Gerini, G.
2016-01-01
In this paper, we present the full process of designing anisotropic metamaterial (MM) wide angle impedance matching (WAIM) layers. These layers are used to reduce the scan losses that occur in active phased arrays for large scanning angles. Numerical results are provided to show the improvement in
Diamagneto-dielectric anisotropic wide angle impedance matching layers for active phased arrays
Silvestri, F.; Cifola, L.; Gerini, G.
2016-01-01
In this paper we present the full process of designing anisotropic metamaterial (MM) wide angle impedance matching (WAIM) layers. These layers are used to reduce the scan losses that occur in active phased arrays for large scanning angles. Numerical results are provided to show the improvement in
Influence of errors in the dimensions of a switched parasitic array on gain and impedance match
CSIR Research Space (South Africa)
Mofolo, MRO
2012-09-01
Full Text Available of these variations on the antenna performance attributes (e.g. gain and impedance match) is estimated using a Monte Carlo simulation. The simulation results demonstrate that the combined effect of all variations in the structural parameters quantify the impact...
Supplementary High-Input Impedance Voltage-Mode Universal Biquadratic Filter Using DVCCs
Directory of Open Access Journals (Sweden)
Jitendra Mohan
2012-01-01
Full Text Available To further extend the existing knowledge on voltage-mode universal biquadratic filter, in this paper, a new biquadratic filter circuit with single input and multiple outputs is proposed, employing three differential voltage current conveyors (DVCCs, three resistors, and two grounded capacitors. The proposed circuit realizes all the standard filter functions, that is, high-pass, band-pass, low-pass, notch, and all-pass filters simultaneously. The circuit enjoys the feature of high-input impedance, orthogonal control of resonance angular frequency (o, and quality factor (Q via grounded resistor and the use of grounded capacitors which is ideal for IC implementation.
Voltage Based Detection Method for High Impedance Fault in a Distribution System
Thomas, Mini Shaji; Bhaskar, Namrata; Prakash, Anupama
2016-09-01
High-impedance faults (HIFs) on distribution feeders cannot be detected by conventional protection schemes, as HIFs are characterized by their low fault current level and waveform distortion due to the nonlinearity of the ground return path. This paper proposes a method to identify the HIFs in distribution system and isolate the faulty section, to reduce downtime. This method is based on voltage measurements along the distribution feeder and utilizes the sequence components of the voltages. Three models of high impedance faults have been considered and source side and load side breaking of the conductor have been studied in this work to capture a wide range of scenarios. The effect of neutral grounding of the source side transformer is also accounted in this study. The results show that the algorithm detects the HIFs accurately and rapidly. Thus, the faulty section can be isolated and service can be restored to the rest of the consumers.
Theory of the low-voltage impedance of superconductor-- p insulator--normal metal tunnel junctions
International Nuclear Information System (INIS)
Lemberger, T.R.
1984-01-01
A theory for the low-voltage impedance of a superconductor-- p insulator--normal metal tunnel junction is developed that includes the effects of charge imbalance and of quasiparticle fluctuations. A novel, inelastic, charge-imbalance relaxation process is identified that is associated with the junction itself. This new process leads to the surprising result that the charge-imbalance component of the dc resistance of a junction becomes independent of the electron-phonon scattering rate as the insulator resistance decreases
Enhancing phonon flow through one-dimensional interfaces by impedance matching
Polanco, Carlos A.; Ghosh, Avik W.
2014-08-01
We extend concepts from microwave engineering to thermal interfaces and explore the principles of impedance matching in 1D. The extension is based on the generalization of acoustic impedance to nonlinear dispersions using the contact broadening matrix Γ(ω), extracted from the phonon self energy. For a single junction, we find that for coherent and incoherent phonons, the optimal thermal conductance occurs when the matching Γ(ω) equals the Geometric Mean of the contact broadenings. This criterion favors the transmission of both low and high frequency phonons by requiring that (1) the low frequency acoustic impedance of the junction matches that of the two contacts by minimizing the sum of interfacial resistances and (2) the cut-off frequency is near the minimum of the two contacts, thereby reducing the spillage of the states into the tunneling regime. For an ultimately scaled single atom/spring junction, the matching criterion transforms to the arithmetic mean for mass and the harmonic mean for spring constant. The matching can be further improved using a composite graded junction with an exponential varying broadening that functions like a broadband antireflection coating. There is, however, a trade off as the increased length of the interface brings in additional intrinsic sources of scattering.
Lucas, Greg M.; Love, Jeffrey J.; Kelbert, Anna
2018-02-01
Commonly, one-dimensional (1-D) Earth impedances have been used to calculate the voltages induced across electric power transmission lines during geomagnetic storms under the assumption that much of the three-dimensional structure of the Earth gets smoothed when integrating along power transmission lines. We calculate the voltage across power transmission lines in the mid-Atlantic region with both regional 1-D impedances and 64 empirical 3-D impedances obtained from a magnetotelluric survey. The use of 3-D impedances produces substantially more spatial variance in the calculated voltages, with the voltages being more than an order of magnitude different, both higher and lower, than the voltages calculated utilizing regional 1-D impedances. During the March 1989 geomagnetic storm 62 transmission lines exceed 100 V when utilizing empirical 3-D impedances, whereas 16 transmission lines exceed 100 V when utilizing regional 1-D impedances. This demonstrates the importance of using realistic impedances to understand and quantify the impact that a geomagnetic storm has on power grids.
Lucas, Greg M.; Love, Jeffrey J.; Kelbert, Anna
2018-01-01
Commonly, one-dimensional (1-D) Earth impedances have been used to calculate the voltages induced across electric power transmission lines during geomagnetic storms under the assumption that much of the three-dimensional structure of the Earth gets smoothed when integrating along power transmission lines. We calculate the voltage across power transmission lines in the mid-Atlantic region with both regional 1-D impedances and 64 empirical 3-D impedances obtained from a magnetotelluric survey. The use of 3-D impedances produces substantially more spatial variance in the calculated voltages, with the voltages being more than an order of magnitude different, both higher and lower, than the voltages calculated utilizing regional 1-D impedances. During the March 1989 geomagnetic storm 62 transmission lines exceed 100 V when utilizing empirical 3-D impedances, whereas 16 transmission lines exceed 100 V when utilizing regional 1-D impedances. This demonstrates the importance of using realistic impedances to understand and quantify the impact that a geomagnetic storm has on power grids.
Directory of Open Access Journals (Sweden)
Hua-Pin Chen
2010-01-01
Full Text Available This paper introduces a novel voltage-mode multifunction biquadratic filter with single input and four outputs using two plus-type differential difference current conveyors (DDCCs and four grounded passive components. The filter can realize inverting highpass, inverting bandpass, noninverting lowpass, and noninverting bandpass filter responses, simultaneously. It still maintains the following advantages: (i using grounded capacitors attractive for integration and absorbing shunt parasitic capacitance, (ii using grounded resistors at all X terminals of DDCCs suitable for the variations of filter parameters and absorbing series parasitic resistances at all X terminals of DDCCs, (iii high-input impedance good for cascadability, (iv no need to change the filter topology, (v no need to component-matching conditions, (vi low active and passive sensitivity performances, and (vii simpler configuration due to the use of plus-type DDCCs only. HSPICE and MATLAB simulations results are provided to demonstrate the theoretical analysis.
Linear variable voltage diode capacitor and adaptive matching networks
Larson, L.E.; De Vreede, L.C.N.
2006-01-01
An integrated variable voltage diode capacitor topology applied to a circuit providing a variable voltage load for controlling variable capacitance. The topology includes a first pair of anti-series varactor diodes, wherein the diode power-law exponent n for the first pair of anti-series varactor
New Power Sharing Control for Inverter-Dominated Microgrid Based on Impedance Match Concept
Gu, Herong; Wang, Deyu; Shen, Hong; Zhao, Wei; Guo, Xiaoqiang
2013-01-01
Power flow control is one of the most important issues for operating the inverter-dominated autonomous microgrid. A technical challenge is how to achieve the accurate active/reactive power sharing of inverters. P-F and Q-V droop control schemes have been widely used for power sharing in the past decades. But they suffer from the poor power sharing in the presence of unequal line impedance. In order to solve the problem, a comprehensive analysis of the power droop control is presented, and a new droop control based on the impedance match concept is proposed in this paper. In addition, the design guidelines of control coefficients and virtual impedance are provided. Finally, the performance evaluation is carried out, and the evaluation results verify the effectiveness of the proposed method. PMID:24453910
New power sharing control for inverter-dominated microgrid based on impedance match concept.
Gu, Herong; Wang, Deyu; Shen, Hong; Zhao, Wei; Guo, Xiaoqiang
2013-01-01
Power flow control is one of the most important issues for operating the inverter-dominated autonomous microgrid. A technical challenge is how to achieve the accurate active/reactive power sharing of inverters. P-F and Q-V droop control schemes have been widely used for power sharing in the past decades. But they suffer from the poor power sharing in the presence of unequal line impedance. In order to solve the problem, a comprehensive analysis of the power droop control is presented, and a new droop control based on the impedance match concept is proposed in this paper. In addition, the design guidelines of control coefficients and virtual impedance are provided. Finally, the performance evaluation is carried out, and the evaluation results verify the effectiveness of the proposed method.
Directory of Open Access Journals (Sweden)
Thuc Phi Duong
2015-07-01
Full Text Available To enable the geometrical freedom envisioned for wireless power transfer (WPT, fast dynamic adaptation to unpredictable changes in receiver position is needed. In this paper, we propose an adaptive impedance-searching system that achieves good impedance matching quickly. For fast and robust operation, the proposed method consists of three steps: system calibration, coarse search, and fine search. The proposed WPT system is characterized using distance variation and lateral and angular misalignment between coils. The measured results indicate that the proposed method significantly reduces searching time from a few minutes to approximately one second. Furthermore, the proposed system achieves impedance matching with good accuracy. The robust impedance-searching capability of the proposed system significantly improves power transfer efficiency. At 6.78 MHz, we achieve a maximum efficiency of 89.7% and a high efficiency of >80% up to a distance of 50 cm. When the center-to-center misalignment is 35 cm, the efficiency is improved from 48.4% to 74.1% with the proposed method. At a distance of 40 cm, the efficiency is higher than 74% for up to 60° of angular rotation. These results agree well with the simulated results obtained using a lumped-element circuit model.
Optimal workloop energetics of muscle-actuated systems: an impedance matching view.
Directory of Open Access Journals (Sweden)
Waleed A Farahat
2010-06-01
Full Text Available Integrative approaches to studying the coupled dynamics of skeletal muscles with their loads while under neural control have focused largely on questions pertaining to the postural and dynamical stability of animals and humans. Prior studies have focused on how the central nervous system actively modulates muscle mechanical impedance to generate and stabilize motion and posture. However, the question of whether muscle impedance properties can be neurally modulated to create favorable mechanical energetics, particularly in the context of periodic tasks, remains open. Through muscle stiffness tuning, we hypothesize that a pair of antagonist muscles acting against a common load may produce significantly more power synergistically than individually when impedance matching conditions are met between muscle and load. Since neurally modulated muscle stiffness contributes to the coupled muscle-load stiffness, we further anticipate that power-optimal oscillation frequencies will occur at frequencies greater than the natural frequency of the load. These hypotheses were evaluated computationally by applying optimal control methods to a bilinear muscle model, and also evaluated through in vitro measurements on frog Plantaris longus muscles acting individually and in pairs upon a mass-spring-damper load. We find a 7-fold increase in mechanical power when antagonist muscles act synergistically compared to individually at a frequency higher than the load natural frequency. These observed behaviors are interpreted in the context of resonance tuning and the engineering notion of impedance matching. These findings suggest that the central nervous system can adopt strategies to harness inherent muscle impedance in relation to external loads to attain favorable mechanical energetics.
Design and development of PLC based offline impedance matching system for ICRH experiment
International Nuclear Information System (INIS)
Joshi, Ramesh; Jadav, H.M.; Mali, Aniruddh; Kulkarni, S.V.
2015-01-01
Ion Cyclotron Resonance Heating (ICRH) transmission line has two impedance matching networks, one for offline matching which has been employed before experimental shot. Another is online impedance matching which has been employed during experimental shot. Offline matching network consists of two static stubs, coarse tuner and coarse phase shifter identical in both transmission lines. There are motorized arrangement installed in each stubs and phase shifters. Both stubs are being used to vary transmission line length. Phase shifter is used to match the frequency of generated RF power. Programmable Logic Controller (PLC) based automation and control technique has been designed and developed for the system. Offline matching should be operated below 1 kHz frequency in order to move stepper motors. Program generates required square pulses which employed to motor controller to move either in upward or downward direction. In existing system this operation has been carried out using VME. To reduce the load on VME, PLC based system has been designed and integrated with main DAC system. WinCC software has been used (as SCADA/HMI) to develop front end GUI which communicates with OPC server. Further, OPC communicates with PLC for control of motorized arrangement. This paper describes technical details,design and development of PLC based offline matching system using WinCC as user interface. The communication between WinCC application and hardware devices was realized by OPC technique. The developed system has friendly graphical user interface, high-level automation and comprehensive function such as experimental process control. The system was proved to be reliable and accurate in practical application. (author)
On the specification of bus voltages and radial link transfer impedance modes
Energy Technology Data Exchange (ETDEWEB)
El-Sadik, F.M. [Khartoum Univ., Aljamaa, Khartoum (Sudan)
2010-07-01
No algebraic equation has been derived for the steady-state stability limit (SSSL) of radial power system components in terms of their resistive sectional impedance elements and associated scalar node voltage constraints. While many criterion have been developed in the literature for the steady state angle and voltage stability limits of systems with losses, speculation exists about certain advocated metrics for these systems in terms of stability reserve margins as a measure of the risk of blackout and the representation of reactance modes associated with Putman's model for synchronous machines. This paper presented the results of a generalized algebraic statement for the reactance modes under stability conditions in voltage-specified power system components cited in the single machine-infinite bus (SMIB) and the radial power link (RPL) systems. The direct analytical solution to the problem enabled the identification of 2 different constraint relations for the sending (E) and receiving-end voltage regulator (VR) voltages. The paper discussed the general SSSL function plan, conditions for SMIB system reactances, and index results for the voltage stability of radial lines. The results for the R2 influence and influence on radial compensation levels were also presented. Index results for non-operational reactance zones were provided. It was concluded that the algebraic solution in a full representation of system losses would enable identification of additional function discontinuities that might not reveal in a step-by-step numerical algorithm and that may account for the many unresolved transmission system phenomenon associated with SSSL predictions and capacitance compensation schemes. 11 refs., 1 tab., 5 figs., 3 appendices.
International Nuclear Information System (INIS)
Kumazawa, R.; Saito, K.; Kasahara, H.; Seki, T.; Mutoh, T.; Shimpo, F.; Nomura, G.; Kato, A.; Okada, H.; Zhao, Y.; Kwak, J.G.; Yoon, J.S.
2008-01-01
Ion cyclotron range of frequency (ICRF) heating has been a reliable tool for steady-state plasma heating with high RF power of several tens of megawatts. However, a sudden increase in the reflected RF power during ICRF heating experiments with ELMy H-mode plasmas is an issue which must be solved for future fusion experimental devices or fusion reactors. This paper describes an innovative ICRF heating system using a frequency feedback control to reduce the reflected power in response to the rapid change in the plasma impedance in the ELMy H-mode plasma. A twin stub tuner has been newly invented for this purpose. The feasibility of keeping the reflected RF power fraction at a low level, e.g. 1%, is demonstrated even with a large change in plasma resistance, e.g. 2 ∼ 8Ω. Calculated and experimental results are presented for the conventional double stub tuner impedance matching system equipped with the twin stub tuner.
Antenna–load interactions at optical frequencies: impedance matching to quantum systems
International Nuclear Information System (INIS)
Olmon, R L; Raschke, M B
2012-01-01
The goal of antenna design at optical frequencies is to deliver optical electromagnetic energy to loads in the form of, e.g., atoms, molecules or nanostructures, or to enhance the radiative emission from such structures, or both. A true optical antenna would, on a qualitatively new level, control the light–matter interaction on the nanoscale for controlled optical signal transduction, radiative decay engineering, quantum coherent control, and super-resolution microscopy, and provide unprecedented sensitivity in spectroscopy. Resonant metallic structures have successfully been designed to approach these goals. They are called optical antennas in analogy to radiofrequency (RF) antennas due to their capability to collect and control electromagnetic fields at optical frequencies. However, in contrast to the RF, where exact design rules for antennas, waveguides, and antenna–load matching in terms of their impedances are well established, substantial physical differences limit the simple extension of the RF concepts into the optical regime. Key distinctions include, for one, intrinsic material resonances including quantum state excitations (metals, metal oxides, semiconductor homo- and heterostructures) and extrinsic resonances (surface plasmon/phonon polaritons) at optical frequencies. Second, in the absence of discrete inductors, capacitors, and resistors, new design strategies must be developed to impedance match the antenna to the load, ultimately in the form of a vibrational, electronic, or spin excitation on the quantum level. Third, there is as yet a lack of standard performance metrics for characterizing, comparing and quantifying optical antenna performance. Therefore, optical antenna development is currently challenged at all the levels of design, fabrication, and characterization. Here we generalize the ideal antenna–load interaction at optical frequencies, characterized by three main steps: (i) far-field reception of a propagating mode exciting an
Antenna-load interactions at optical frequencies: impedance matching to quantum systems.
Olmon, R L; Raschke, M B
2012-11-09
The goal of antenna design at optical frequencies is to deliver optical electromagnetic energy to loads in the form of, e.g., atoms, molecules or nanostructures, or to enhance the radiative emission from such structures, or both. A true optical antenna would, on a qualitatively new level, control the light-matter interaction on the nanoscale for controlled optical signal transduction, radiative decay engineering, quantum coherent control, and super-resolution microscopy, and provide unprecedented sensitivity in spectroscopy. Resonant metallic structures have successfully been designed to approach these goals. They are called optical antennas in analogy to radiofrequency (RF) antennas due to their capability to collect and control electromagnetic fields at optical frequencies. However, in contrast to the RF, where exact design rules for antennas, waveguides, and antenna-load matching in terms of their impedances are well established, substantial physical differences limit the simple extension of the RF concepts into the optical regime. Key distinctions include, for one, intrinsic material resonances including quantum state excitations (metals, metal oxides, semiconductor homo- and heterostructures) and extrinsic resonances (surface plasmon/phonon polaritons) at optical frequencies. Second, in the absence of discrete inductors, capacitors, and resistors, new design strategies must be developed to impedance match the antenna to the load, ultimately in the form of a vibrational, electronic, or spin excitation on the quantum level. Third, there is as yet a lack of standard performance metrics for characterizing, comparing and quantifying optical antenna performance. Therefore, optical antenna development is currently challenged at all the levels of design, fabrication, and characterization. Here we generalize the ideal antenna-load interaction at optical frequencies, characterized by three main steps: (i) far-field reception of a propagating mode exciting an antenna
High-efficiency resonant coupled wireless power transfer via tunable impedance matching
Anowar, Tanbir Ibne; Barman, Surajit Das; Wasif Reza, Ahmed; Kumar, Narendra
2017-10-01
For magnetic resonant coupled wireless power transfer (WPT), the axial movement of near-field coupled coils adversely degrades the power transfer efficiency (PTE) of the system and often creates sub-resonance. This paper presents a tunable impedance matching technique based on optimum coupling tuning to enhance the efficiency of resonant coupled WPT system. The optimum power transfer model is analysed from equivalent circuit model via reflected load principle, and the adequate matching are achieved through the optimum tuning of coupling coefficients at both the transmitting and receiving end of the system. Both simulations and experiments are performed to evaluate the theoretical model of the proposed matching technique, and results in a PTE over 80% at close coil proximity without shifting the original resonant frequency. Compared to the fixed coupled WPT, the extracted efficiency shows 15.1% and 19.9% improvements at the centre-to-centre misalignment of 10 and 70 cm, respectively. Applying this technique, the extracted S21 parameter shows more than 10 dB improvements at both strong and weak couplings. Through the developed model, the optimum coupling tuning also significantly improves the performance over matching techniques using frequency tracking and tunable matching circuits.
Derivation of elastic stiffness from site-matched mineral density and acoustic impedance maps
International Nuclear Information System (INIS)
Raum, Kay; Cleveland, Robin O; Peyrin, Francoise; Laugier, Pascal
2006-01-01
200 MHz acoustic impedance maps and site-matched synchrotron radiation micro computed tomography (SR-μCT) maps of tissue degree of mineralization of bone (DMB) were used to derive the elastic coefficient c 33 in cross sections of human cortical bone. To accomplish this goal, a model was developed to relate the DMB accessible with SR-μCT to mass density. The formulation incorporates the volume fractions and densities of the major bone tissue components (collagen, mineral and water), and accounts for tissue porosity. We found that the mass density can be well modelled by a second-order polynomial fit to DMB (R 2 = 0.999) and appears to be consistent with measurements of many different types of mineralized tissues. The derived elastic coefficient c 33 correlated more strongly with the acoustic impedance (R 2 = 0.996) than with mass density (R 2 = 0.310). This finding suggests that estimates of c 33 made from measurements of the acoustic impedance are more reliable than those made from density measurements. Mass density and elastic coefficient were in the range between 1.66 and 2.00 g cm -3 and 14.8 and 75.4 GPa, respectively. Although SAM inspection is limited to the evaluation of carefully prepared sample surfaces, it provides a two-dimensional quantitative estimate of elastic tissue properties at the tissue level
Detection of high-impedance fault in low-voltage DC distribution system via mathematical morphology
Directory of Open Access Journals (Sweden)
Yun-Sik Oh
2016-01-01
Full Text Available This study presents a method for high-impedance fault (HIF detection in a low-voltage DC (LVDC distribution system via mathematical morphology (MM, which is composed of two elementary transformations, namely, dilation and erosion. Various MM-based filters are used to detect abnormal signals of current waveform. The LVDC distribution system, including power conversion devices, such as AC/DC and DC/DC converters, is modelled with electromagnetic transient program (EMTP software to verify the proposed method. The HIF arc model in the DC system is also implemented with EMTP/MODELS, which is a symbolic language interpreter for EMTP. Simulation results show that the proposed method can be applied to detect HIF effectively in the LVDC distribution system.
Renk, Timothy; Simpson, Sean; Webb, Timothy; Mazarakis, Michael; Kiefer, Mark
2016-10-01
The SMP diode, fielded on the RITS-6 (3.5-8.5 MV) IVA accelerator at Sandia National Laboratories, produces a focused electron beam (transmission line (MITL) center conductors, of 40 and 80 ohms flow impedance. We have operated in-situ heating and discharge-cleaning hardware in the load region, in order to address the tendency of some shots to undergo premature impedance (Z) collapse, defined as a fall in impedance beyond that due to normal movement of electrode plasmas that reduces the effective A-K gap. The goal of heating/cleaning was to reduce the volume of evolving gases near the A-K gap. Despite clear evidence that the cleaning techniques removed the proton portion of beam current, we observed no consistent increase in diode impedance (ZDIODE). This forced an examination of the role that the IVA flow impedance has on ZDIODE. A preliminary conclusion is that ZDIODE should be at least 1.5 times the flow impedance before ZDIODE is a parameter independent of flow impedance. This has implications for SMP as a load for a IVA, since ZDIODE >100 ohms has not been consistently demonstrated. Data analysis is ongoing, and latest results will be reported. Sandia National Laboratories is a multi-program laboratory managed and operated by Sandia Corporation, a wholly owned subsidiary of Lockheed Martin Corporation, for the U.S. Department of Energy's National Nuclear Security Administration under contract DE-AC04-94AL85000.
Controllable transmission and total reflection through an impedance-matched acoustic metasurface
Mei, Jun
2014-12-02
A general design paradigm for a novel type of acoustic metasurface is proposed by introducing periodically repeated supercells on a rigid thin plate, where each supercell contains multiple cut-through slits that are filled with materials possessing different refractive indices but the same impedance as that of the host medium. When the wavelength of the incident wave is smaller than the periodicity, the direction of the transmitted wave with nearly unity transmittance can be chosen by engineering the phase discontinuities along the transverse direction. When the wavelength is larger than the periodicity, even though the metasurface is impedance matched to the host medium, most of the incident energy is reflected back and the remaining portion is converted into a surface-bound mode. We show that both the transmitted wave control and the high reflection with the surface mode excitation can be interpreted by a unified analytic model based on mode-coupling theory. Our general design principle not only supplies the functionalities of reflection-type acoustic metasurfaces, but also exhibits unprecedented flexibility and efficiency in various domains of wave manipulation for possible applications in fields like refracting, collimating, focusing or absorbing wave energy.
The origins of the cochlea and impedance matching hearing in synapsids
Directory of Open Access Journals (Sweden)
Michael Laass
2016-06-01
Full Text Available The origin of tympanic hearing in early synapsids is still controversial, because little is known about their inner ear and the function of their sound conducting apparatus. Here I describe the earliest known tympanic ear in the synapsid lineage, the ear of Pristerodon (Therapsida, Anomodontia from the Late Permian of South Africa, which was virtually reconstructed from neutron tomographic data. Although Pristerodon is not a direct ancestor of mammals, its inner ear with distinctive cochlear cavity represents a connecting link between the primitive therapsid inner ear and the mammalian inner ear. The anatomy of the sound conducting apparatus of Pristerodon and the increased sound pressure transformer ratio points to a sensitivity to airborne sound. Furthermore, the origins of the cochlea and impedance matching hearing in synapsids coincided with the loss of contact between head and substrate, which already took place at least in Late Permian therapsids even before the postdentary bones became detached from the mandible.
DEFF Research Database (Denmark)
Guo, Xiaoqiang; Zhang, Xue; Wang, Baocheng
2014-01-01
This letter presents a new control strategy of threephase grid-connected inverter for the positive sequence voltage recovery and negative sequence voltage reduction under asymmetrical grid faults. Unlike the conventional control strategy based on an assumption that the network impedance is mainly...... of the proposed solution for the flexible voltage support in a low-voltage grid, where thenetwork impedance is mainly resistive.......This letter presents a new control strategy of threephase grid-connected inverter for the positive sequence voltage recovery and negative sequence voltage reduction under asymmetrical grid faults. Unlike the conventional control strategy based on an assumption that the network impedance is mainly...... inductive, the proposed control strategy is more flexible and effective by considering the network impedance impact, which is of great importance for the high penetration of grid-connected renewable energy systems into low-voltage grids. The experimental tests are carried out to validate the effectiveness...
DEFF Research Database (Denmark)
Wang, Xiongfei; Blaabjerg, Frede; Chen, Zhe
2012-01-01
-sequence virtual resistance even in the case of feeding a balanced three-phase load. Furthermore, to adapt to the variety of unbalanced loads, a dynamically-tuned negative-sequence resistance loop is designed, such that a good compromise between the quality of inverter output voltage and the performance of load......The virtual output impedance loop is known as an effective way to enhance the load sharing stability and quality of droop-controlled parallel inverters. This paper proposes an improved design of virtual output impedance loop for parallel three-phase voltage source inverters. In the approach...... sharing can be obtained. Finally, laboratory test results of two parallel three-phase voltage source inverters are shown to confirm the validity of the proposed method....
Bunch motion in the presence of the self-induced voltage due to a reactive impedance with RF off
Energy Technology Data Exchange (ETDEWEB)
Shaposhnikova, E [European Organization for Nuclear Research, Geneva (Switzerland)
1996-08-01
Analytic self-consistent solutions have been found for the nonlinear Vlasov equation describing different types of behaviour with time of an intense bunch under the influence of voltage induced due to a reactive part of broad band impedance. The problem is solved for the particular type of the initial distribution function in longitudinal phase space which is elliptic and corresponds to parabolic line density. This paper is devoted to the consideration of the effects in the machine with RF off. In this case the induced voltage is changing with time and can significantly affect bunch motion. The same method applied in the case with RF on allows the time dependent effects of potential well distortion to be analysed. Numerical estimations for the CERN SPS show that effect of induced voltage is important for beam manipulations with RF off. Measurements of the change in the rate of debunching with intensity can be used to estimate the value of the reactive impedance. (author)
DEFF Research Database (Denmark)
Li, Chendan; Chaudhary, Sanjay Kumar; Savaghebi, Mehdi
2017-01-01
In the low-voltage (LV) ac microgrids (MGs), with a relatively high R/X ratio, virtual impedance is usually adopted to improve the performance of droop control applied to distributed generators (DGs). At the same time, LV dc MG using virtual impedance as droop control is emerging without adequate...... power flow studies. In this paper, power flow analyses for both ac and dc MGs are formulated and implemented. The mathematical models for both types of MGs considering the concept of virtual impedance are used to be in conformity with the practical control of the DGs. As a result, calculation accuracy...... is improved for both ac and dc MG power flow analyses, comparing with previous methods without considering virtual impedance. Case studies are conducted to verify the proposed power flow analyses in terms of convergence and accuracy. Investigation of the impact to the system of internal control parameters...
Jing, Mingyong; Yu, Bo; Hu, Jianyong; Hou, Huifang; Zhang, Guofeng; Xiao, Liantuan; Jia, Suotang
2017-05-15
In this paper, we present a novel ultra-narrow linewidth fiber resonator formed by a tunable polarization maintaining (PM) π-phase-shifted fiber Bragg grating and a PM uniform fiber Bragg grating with a certain length of PM single mode fiber patch cable between them. Theoretical prediction shows that this resonator has ultra-narrow linewidth resonant peaks and is easy to realize impedance matching. We experimentally obtain 3 MHz narrow linewidth impedance matched resonant peak in a 7.3 m ultra-long passive fiber cavity. The impedance self-matching characteristic of this resonator also makes itself particularly suitable for use in ultra-sensitive sensors, ultra-narrow band rejection optical filters and fiber lasers applications.
International Nuclear Information System (INIS)
Sudhir, Dass; Bandyopadhyay, M.; Chakraborty, A.; Kraus, W.; Gahlaut, A.; Bansal, G.
2014-01-01
Impedance matching circuit between radio frequency (RF) generator and the plasma load, placed between them, determines the RF power transfer from RF generator to the plasma load. The impedance of plasma load depends on the plasma parameters through skin depth and plasma conductivity or resistivity. Therefore, for long pulse operation of inductively coupled plasmas, particularly for high power (∼100 kW or more) where plasma load condition may vary due to different reasons (e.g., pressure, power, and thermal), online tuning of impedance matching circuit is necessary through feedback. In fusion grade ion source operation, such online methodology through feedback is not present but offline remote tuning by adjusting the matching circuit capacitors and tuning the driving frequency of the RF generator between the ion source operation pulses is envisaged. The present model is an approach for remote impedance tuning methodology for long pulse operation and corresponding online impedance matching algorithm based on RF coil antenna current measurement or coil antenna calorimetric measurement may be useful in this regard
Miao, Zhidong; Liu, Dake; Gong, Chen
2017-08-01
For an inductive wireless power transfer (IWPT) system, maintaining a reasonable power transfer efficiency and a stable output power are two most challenging design issues, especially when coil distance varies. To solve these issues, this paper presents a novel adaptive impedance matching network (IMN) for IWPT system. In our adaptive IMN IWPT system, the IMN is automatically reconfigured to keep matching with the coils and to adjust the output power adapting to coil distance variation. A closed loop control algorithm is used to change the capacitors continually, which can compensate mismatches and adjust output power simultaneously. The proposed adaptive IMN IWPT system is working at 125 kHz for 2 W power delivered to load. Comparing with the series resonant IWPT system and fixed IMN IWPT system, the power transfer efficiency of our system increases up to 31.79% and 60% when the coupling coefficient varies in a large range from 0.05 to 0.8 for 2 W output power.
Miao, Zhidong; Liu, Dake
2017-01-01
For an inductive wireless power transfer (IWPT) system, maintaining a reasonable power transfer efficiency and a stable output power are two most challenging design issues, especially when coil distance varies. To solve these issues, this paper presents a novel adaptive impedance matching network (IMN) for IWPT system. In our adaptive IMN IWPT system, the IMN is automatically reconfigured to keep matching with the coils and to adjust the output power adapting to coil distance variation. A closed loop control algorithm is used to change the capacitors continually, which can compensate mismatches and adjust output power simultaneously. The proposed adaptive IMN IWPT system is working at 125 kHz for 2 W power delivered to load. Comparing with the series resonant IWPT system and fixed IMN IWPT system, the power transfer efficiency of our system increases up to 31.79% and 60% when the coupling coefficient varies in a large range from 0.05 to 0.8 for 2 W output power. PMID:28763011
Song, Ai-Ling; Chen, Tian-Ning; Wang, Xiao-Peng; Wan, Le-Le
2016-08-01
The waveform distortion happens in most of the unidirectional acoustic transmission (UAT) devices proposed before. In this paper, a novel type of waveform-preserved UAT device composed of an impedance-matched acoustic metasurface (AMS) and a phononic crystal (PC) structure is proposed and numerically investigated. The acoustic pressure field distributions and transmittance are calculated by using the finite element method. The subwavelength AMS that can modulate the wavefront of the transmitted wave at will is designed and the band structure of the PC structure is calculated and analyzed. The sound pressure field distributions demonstrate that the unidirectional acoustic transmission can be realized by the proposed UAT device without changing the waveforms of the output waves, which is the distinctive feature compared with the previous UAT devices. The physical mechanism of the unidirectional acoustic transmission is discussed by analyzing the refraction angle changes and partial band gap map. The calculated transmission spectra show that the UAT device is valid within a relatively broad frequency range. The simulation results agree well with the theoretical predictions. The proposed UAT device provides a good reference for designing waveform-preserved UAT devices and has potential applications in many fields, such as medical ultrasound, acoustic rectifiers, and noise insulation.
Miao, Zhidong; Liu, Dake; Gong, Chen
2017-10-01
Inductive wireless power transfer (IWPT) is a promising power technology for implantable biomedical devices, where the power consumption is low and the efficiency is the most important consideration. In this paper, we propose an optimization method of impedance matching networks (IMN) to maximize the IWPT efficiency. The IMN at the load side is designed to achieve the optimal load, and the IMN at the source side is designed to deliver the required amount of power (no-more-no-less) from the power source to the load. The theoretical analyses and design procedure are given. An IWPT system for an implantable glaucoma therapeutic prototype is designed as an example. Compared with the efficiency of the resonant IWPT system, the efficiency of our optimized system increases with a factor of 1.73. Besides, the efficiency of our optimized IWPT system is 1.97 times higher than that of the IWPT system optimized by the traditional maximum power transfer method. All the discussions indicate that the optimization method proposed in this paper could achieve a high efficiency and long working time when the system is powered by a battery.
Directory of Open Access Journals (Sweden)
Yuan Yang
2017-12-01
Full Text Available For achieving the power maximum transmission, the electrical impedance matching (EIM for piezoelectric ultrasonic transducers is highly required. In this paper, the effect of EIM networks on the electromechanical characteristics of sandwiched piezoelectric ultrasonic transducers is investigated in time and frequency domains, based on the PSpice model of single sandwiched piezoelectric ultrasonic transducer. The above-mentioned EIM networks include, series capacitance and parallel inductance (I type and series inductance and parallel capacitance (II type. It is shown that when I and II type EIM networks are used, the resonance and anti-resonance frequencies and the received signal tailing are decreased; II type makes the electro-acoustic power ratio and the signal tailing smaller whereas it makes the electro-acoustic gain ratio larger at resonance frequency. In addition, I type makes the effective electromechanical coupling coefficient increase and II type makes it decrease; II type make the power spectral density at resonance frequency more dramatically increased. Specially, the electro-acoustic power ratio has maximum value near anti-resonance frequency, while the electro-acoustic gain ratio has maximum value near resonance frequency. It can be found that the theoretically analyzed results have good consistency with the measured ones.
A Low-Profile WLAN Antenna with Inductor and Tuning Stub for Broadband Impedance Matching
Directory of Open Access Journals (Sweden)
Woo-Su Kim
2014-01-01
Full Text Available This paper presents a low-profile multiband antenna suitable for wireless local area networks (WLANs, using a chip inductor and tuning stub for broadband impedance matching. The proposed antenna is compact 12×10×1 mm3 and covers three bands: 2.4-GHz (2.400–2.484 GHz, 5.2-GHz (5.150–5.350 GHz, and 5.8-GHz (5.725–5.825 GHz. The measured 10-dB bandwidths are 12.0% (2.28–2.57 GHz in the lower band for 2.4-GHz WLANs and 39.1% (4.81–7.15 GHz in the upper band for 5 GHz-WLANs. The measured peak gain of the antenna is between 2.7 and 4.39 dBi and the radiation patterns are omnidirectional.
Degtiarenko, Pavel V [Williamsburg, VA; Popov, Vladimir E [Newport News, VA
2011-03-22
A first stage electronic system for receiving charge or current from voltage-controlled sensors or detectors that includes a low input impedance current receiver/converter device (for example, a transimpedance amplifier), which is directly coupled to the sensor output, a source of bias voltage, and the device's power supply (or supplies), which use the biased voltage point as a baseline.
Sudhir, Dass; Bandyopadhyay, M; Chakraborty, A
2016-02-01
Plasma characterization and impedance matching are an integral part of any radio frequency (RF) based plasma source. In long pulse operation, particularly in high power operation where plasma load may vary due to different reasons (e.g. pressure and power), online tuning of impedance matching circuit and remote plasma density estimation are very useful. In some cases, due to remote interfaces, radio activation and, due to maintenance issues, power probes are not allowed to be incorporated in the ion source design for plasma characterization. Therefore, for characterization and impedance matching, more remote schemes are envisaged. Two such schemes by the same authors are suggested in these regards, which are based on air core transformer model of inductive coupled plasma (ICP) [M. Bandyopadhyay et al., Nucl. Fusion 55, 033017 (2015); D. Sudhir et al., Rev. Sci. Instrum. 85, 013510 (2014)]. However, the influence of the RF field interaction with the plasma to determine its impedance, a physics code HELIC [D. Arnush, Phys. Plasmas 7, 3042 (2000)] is coupled with the transformer model. This model can be useful for both types of RF sources, i.e., ICP and helicon sources.
Electron heating of voltage-driven and matched dual frequency discharges
International Nuclear Information System (INIS)
Lieberman, M A; Lichtenberg, A J
2010-01-01
In a dual frequency capacitive sheath, a high frequency uniform sheath motion is coupled with a low frequency Child law sheath motion. For current-driven high and low frequency sheaths, the high frequency sheath motion generates most of the ohmic and stochastic heating of the discharge electrons. The low frequency motion, in addition to its primary purpose of establishing the ion bombarding energy, also increases the heating by widening the sheath width and transporting the oscillating electrons to regions of lower plasma density, and hence higher sheath velocity. In this work, we show that for voltage-driven high and low frequency sheaths, increasing the low frequency voltage reduces the heating, due to the reduced high frequency current that flows through the sheath under voltage-driven conditions. We determine the dependence of the heating on various parameters and compare the results with the current-driven case. Particle-in-cell simulations are used to confirm this result. Discharges generally employ a matching network to maximize the power transmitted to the plasma. We obtain analytic expressions for the effect of the low frequency source under matched conditions and, again, find that the low frequency source reduces the heating.
DEFF Research Database (Denmark)
Yoon, Ick-Jae; Christensen, S.; Zhurbenko, Vitaliy
2016-01-01
The impedance bandwidth (BW) improvement property of a self-resonant folded spherical helix electric dipole and a spherical split ring (SSR) magnetic dipole is compared when a negative reactance element is loaded on the parasitic resonator of the antennas. They have the same electrical size of ka...
Farmann, Alexander; Waag, Wladislaw; Sauer, Dirk Uwe
2015-12-01
Robust algorithms using reduced order equivalent circuit model (ECM) for an accurate and reliable estimation of battery states in various applications become more popular. In this study, a novel adaptive, self-learning heuristic algorithm for on-board impedance parameters and voltage estimation of lithium-ion batteries (LIBs) in electric vehicles is introduced. The presented approach is verified using LIBs with different composition of chemistries (NMC/C, NMC/LTO, LFP/C) at different aging states. An impedance-based reduced order ECM incorporating ohmic resistance and a combination of a constant phase element and a resistance (so-called ZARC-element) is employed. Existing algorithms in vehicles are much more limited in the complexity of the ECMs. The algorithm is validated using seven day real vehicle data with high temperature variation including very low temperatures (from -20 °C to +30 °C) at different Depth-of-Discharges (DoDs). Two possibilities to approximate both ZARC-elements with finite number of RC-elements on-board are shown and the results of the voltage estimation are compared. Moreover, the current dependence of the charge-transfer resistance is considered by employing Butler-Volmer equation. Achieved results indicate that both models yield almost the same grade of accuracy.
Bunch motion in the presence of the self-induced voltage due to a reactive impedance; 1, RF off
Shaposhnikova, Elena
1995-01-01
Analytic self-consistent solutions have been found for the nonlinear Vlasov equation describing different types of behaviour with time of an intense bunch under the influence of voltage induced due to a reactive part of broad band impedance. The problem is solved for the particular type of the initial distribution function in longitudinal phase space which is elliptic and corresponds to parabolic line density. The first part of the paper is devoted to the consideration of the effects in the machine with RF off. In this case induced voltage is changing with time and, as in the case with RF on, can have a significant effect on bunch motion. Numerical estimations for the SPS show that this effect can be important for manipulations with beam at 26GeV. Measurements of the change in the rate of debunching with intensity can also be used to estimate the value of the impedance. The same method is applied in the second part of the paper to analyse time dependent effects of potential well distortion when RF is on.
Directory of Open Access Journals (Sweden)
Johannes Müller
Full Text Available BACKGROUND: Insights into the onset of evolutionary novelties are key to the understanding of amniote origins and diversification. The possession of an impedance-matching tympanic middle ear is characteristic of all terrestrial vertebrates with a sophisticated hearing sense and an adaptively important feature of many modern terrestrial vertebrates. Whereas tympanic ears seem to have evolved multiple times within tetrapods, especially among crown-group members such as frogs, mammals, squamates, turtles, crocodiles, and birds, the presence of true tympanic ears has never been recorded in a Paleozoic amniote, suggesting they evolved fairly recently in amniote history. METHODOLOGY/PRINCIPAL FINDINGS: In the present study, we performed a morphological examination and a phylogenetic analysis of poorly known parareptiles from the Middle Permian of the Mezen River Basin in Russia. We recovered a well-supported clade that is characterized by a unique cheek morphology indicative of a tympanum stretching across large parts of the temporal region to an extent not seen in other amniotes, fossil or extant, and a braincase specialized in showing modifications clearly related to an increase in auditory function, unlike the braincase of any other Paleozoic tetrapod. In addition, we estimated the ratio of the tympanum area relative to the stapedial footplate for the basalmost taxon of the clade, which, at 23:1, is in close correspondence to that of modern amniotes capable of efficient impedance-matching hearing. CONCLUSIONS/SIGNIFICANCE: Using modern amniotes as analogues, the possession of an impedance-matching middle ear in these parareptiles suggests unique ecological adaptations potentially related to living in dim-light environments. More importantly, our results demonstrate that already at an early stage of amniote diversification, and prior to the Permo-Triassic extinction event, the complexity of terrestrial vertebrate ecosystems had reached a level that
High aspect ratio MEMS capacitor for high frequency impedance matching applications
DEFF Research Database (Denmark)
Yalcinkaya, Arda Deniz; Jensen, Søren; Hansen, Ole
2003-01-01
We present a microelectromechanical tunable capacitor with a low control voltage, a wide tuning range and adequate electrical quality factor. The device is fabricated in a single-crystalline silicon layer using deep reactive ion etching (DRIE) for obtaining high-aspect ratio (> 20) parallel comb...
International Nuclear Information System (INIS)
Lee, Seung Hyun; Song, Ho Seung; Kim, Jeong Hwan; Cong, Truong Van; Kim, Hui Su; Yeon, Yeong Heum; Lee, Yong seok; Chai, Jong Seo
2015-01-01
The SKKUCY-9 is a compact cyclotron for radioactive isotopes (RI) production of positron emission tomography (PET). Charged particles such as H-ions are accelerated azimuthally within a high intensity electric field (E-field) generated from a radio frequency (RF) system in cyclotron. A high power RF signal is transmitted from an RF amplifier to an RF resonating cavity. The RF system of the SKKUCY-9 operates in continuous wave (CW) mode. If an ion beam were accelerated in the cyclotron, the vacuum level and permittivity would be changed because of beam loading. It causes an impedance shift of the RF resonating cavity. This impedance mismatch generates reflected power that decreases the RF transmitting power. To prevent this situation, an impedance matching system is necessary. This paper describes the impedance matching system of a 20 kW RF amplifier in an SKKUCY-9 compact cyclotron. The impedance matching circuit was designed using both an input stage and output stage, which are divided between the cathode and anode in a vacuum tube that is used as an amplifying device. The equivalent circuit of the matching system is made of passive elements. The characteristic results of designed circuit were calculated using a Smith chart. In assembling, the inductors were replaced by movable stub-line structures. The dimensions of the stub-line structures were optimized with equations and the measurement results. The experiment was performed to find the result values of matching circuit impedance and RF power amplitude
International Nuclear Information System (INIS)
Douvropoulos, Theodosios G
2012-01-01
An approximate formula for the period of pendulum motion beyond the small amplitude regime is obtained based on physical arguments. Two different schemes of different accuracy are developed: in the first less accurate scheme, emphasis is given on the non-quadratic form of the potential in connection to isochronism, and a specific form of a generic formula that is met in many previous works is produced, while the second and main result contains the Struve function which is further approximated by a simple sinusoidal expression based on its maximum value. The accuracy of the final formula gives a relative error of less than 0.2% for angles up to 140°. In addition, a simple relation between the Struve function and the complete elliptic integral of the first kind is produced, since they both constitute solutions of the pendulum period. This relation makes it possible for someone to connect different areas in physics and solve a difficult task by comparison with another much more simple one. As an example, a connection between the pendulum period and the acoustical radiation impedance is proposed through impedance matching and some interesting relations are produced. This paper is intended for undergraduate students to be useful for analysing pendulum experiments in introductory physics labs and it is also believed to offer valuable insight into some properties of the simple pendulum in undergraduate courses on general physics. (paper)
Impedance matching of a coaxial antenna for microwave in-situ processing of polluted soils.
Pauli, Mario; Kayser, Thorsten; Wiesbeck, Werner; Komarov, Vyacheslav
2011-01-01
The present paper is focused on the minimization of return loss of a slotted coaxial radiator proposed for a decontamination system for soils contaminated by volatile or semi-volatile organic compounds such as oils or fuels. The antenna upgrade is achieved by coating it with a 5 mm thick Teflon layer. The electromagnetic characteristics reflection coefficient and power density distribution around the antenna surrounded by soils with different moisture levels are analyzed numerically. Simplified analytical approaches are employed to accelerate the optimization of the given antenna for microwave heating systems. The improved antenna design shows a good matching of the antenna to the surrounding soil with varying moisture levels. This ensures a high efficiency of the proposed in-situ soil decontamination system.
Interdependence, Reflexivity, Fidelity, Impedance Matching, and the Evolution of Genetic Coding
Carter, Charles W; Wills, Peter R
2018-01-01
Abstract Genetic coding is generally thought to have required ribozymes whose functions were taken over by polypeptide aminoacyl-tRNA synthetases (aaRS). Two discoveries about aaRS and their interactions with tRNA substrates now furnish a unifying rationale for the opposite conclusion: that the key processes of the Central Dogma of molecular biology emerged simultaneously and naturally from simple origins in a peptide•RNA partnership, eliminating the epistemological utility of a prior RNA world. First, the two aaRS classes likely arose from opposite strands of the same ancestral gene, implying a simple genetic alphabet. The resulting inversion symmetries in aaRS structural biology would have stabilized the initial and subsequent differentiation of coding specificities, rapidly promoting diversity in the proteome. Second, amino acid physical chemistry maps onto tRNA identity elements, establishing reflexive, nanoenvironmental sensing in protein aaRS. Bootstrapping of increasingly detailed coding is thus intrinsic to polypeptide aaRS, but impossible in an RNA world. These notions underline the following concepts that contradict gradual replacement of ribozymal aaRS by polypeptide aaRS: 1) aaRS enzymes must be interdependent; 2) reflexivity intrinsic to polypeptide aaRS production dynamics promotes bootstrapping; 3) takeover of RNA-catalyzed aminoacylation by enzymes will necessarily degrade specificity; and 4) the Central Dogma’s emergence is most probable when replication and translation error rates remain comparable. These characteristics are necessary and sufficient for the essentially de novo emergence of a coupled gene–replicase–translatase system of genetic coding that would have continuously preserved the functional meaning of genetically encoded protein genes whose phylogenetic relationships match those observed today. PMID:29077934
Directory of Open Access Journals (Sweden)
A. ZAZOUA
2014-05-01
Full Text Available Impedance spectroscopy and capacitance-voltage (C-V methods are a rapidly developing electrochemical technique for the characterization of biomaterial–functionalized electrodes and biocatalytic transformations on the electrodes surface, and specifically for the transduction of biosensing events at electrodes. Such techniques have been used in our work as a tool for the characterization of a new nitrite biosensor for environmental applications based on the immobilization of catalase on insulator-semiconductor (IS systems (p-Si/SiO2/Si3N4. The principle of the developed biosensor includes the following: Catalase catalyzed the breakdown of H2O2 into H2O and O2. Nitrite was selected as an inhibitor of catalase. Under optimal conditions, i.e. buffer capacity corresponding to 3 mM phosphate buffer, the catalase enzyme insulator semiconductor sensors shows a high sensitivity to nitrite detection. In both cases, the responses of these biosensors based on nitrite additions are good with the detection limit around 10-11 M. It is expected that such an original and promising concept of inhibitor-based biosensors based on reactivation by inhibitive effects, will be useful for the development of environmental smart biosensors based on the integration of ENFET with the corresponding instrumentation in the same silicon chip.
Saffar, Saber; Abdullah, Amir
2012-01-01
The effective ultrasonic energy radiation into the air of piezoelectric transducers requires using multilayer matching systems with accurately selected acoustic impedances and the thickness of particular layers. One major problem of ultrasonic transducers, radiating acoustic energy into air, is to find the proper acoustic impedances of one or more matching layers. This work aims at developing an original solution to the acoustic impedance mismatch between transducer and air. If the acoustic impedance defences between transducer and air be more, then finding best matching layer(s) is harder. Therefore we consider PZT (lead zirconate titanate piezo electric) transducer and air that has huge acoustic impedance deference. The vibration source energy (PZT), which is used to generate the incident wave, consumes a part of the mechanical energy and converts it to an electrical one in theoretical calculation. After calculating matching layers, we consider the energy source as layer to design a transducer. However, this part of the mechanical energy will be neglected during the mathematical work. This approximation is correct only if the transducer is open-circuit. Since the possibilities of choosing material with required acoustic impedance are limited (the counted values cannot always be realized and applied in practice) it is necessary to correct the differences between theoretical values and the possibilities of practical application of given acoustic impedances. Such a correction can be done by manipulating other parameters of matching layers (e.g. by changing their thickness). The efficiency of the energy transmission from the piezoceramic transducer through different layers with different thickness and different attenuation enabling a compensation of non-ideal real values by changing their thickness was computer analyzed (base on genetic algorithm). Firstly, three theoretical solutions were investigated. Namely, Chebyshev, Desilets and Souquet theories. However, the
Rahman Khan, Motiur; Anjaneyulu, P.; Koteswara Rao, K. S. R.; Menon, R.
2017-03-01
We report on the analysis of temperature-dependent current-voltage characteristics and impedance measurements of electrochemically doped poly(3-methylthiophene) devices at different doping levels. The extent of doping is carefully tailored such that only the bulk-limited transport mechanism prevails. A transition from exponentially distributed trap-limited transport to trap-free space-charge-limited current is observed in current-voltage conduction upon increasing the doping. The obtained trap densities (3.2 × 1016 cm-3 and 8.6 × 1015 cm-3) and trap energies (31.7 meV and 16.6 meV) for different devices signify the variation in disorder with doping, which is later supported by impedance measurements. Impedance-frequency data for various devices can not be explained using the parallel resistance-capacitance (RC) model in the equivalent circuit. However, this was established by incorporating a constant phase element Q (CPE) instead of the capacitance parameter. It should be emphasized that low doping devices in particular are best simulated with two CPE elements, while the data related to other devices are fitted well with a single CPE element. It is also observed from evaluated circuit parameters that the spatial inhomogeneity and disorder are the cause of variability in different samples, which has an excellent correlation with the temperature-dependent current-voltage characteristics.
Time isolation high-voltage impulse generator
International Nuclear Information System (INIS)
Chodorow, A.M.
1975-01-01
Lewis' high-voltage impulse generator is analyzed in greater detail, demonstrating that voltage between adjacent nodes can be equalized by proper selection of parasitic impedances. This permits improved TEM mode propagation to a matched load, with more faithful source waveform preservation
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Loh, Poh Chiang; Blaabjerg, Frede
2014-01-01
This letter introduces a new versatile Y-shaped impedance network for realizing converters that demand a very high-voltage gain, while using a small duty ratio. To achieve that, the proposed network uses a tightly coupled transformer with three windings, whose obtained gain is presently not matched...... by existing networks operated at the same duty ratio. The proposed impedance network also has more degrees of freedom for varying its gain, and hence, more design freedom for meeting requirements demanded from it. This capability has been demonstrated by mathematical derivation, and proven in experiment...
Matching of Tore Supra ICRH antennas
International Nuclear Information System (INIS)
Ladurelle, L.; Beaumont, B.; Kuus, H.; Lombard, G.
1994-01-01
An automatic matching method is described for Tore Supra ICRH antennas based on impedance variations seen at their feed points. Error signals derived from directional voltage and phase measurements in the feeder allow to control the matching capacitors values for optimal power transmission. (author) 5 refs.; 9 figs
International Nuclear Information System (INIS)
Khan, Motiur Rahman; Koteswara Rao, K S R; Menon, R; Anjaneyulu, P
2017-01-01
We report on the analysis of temperature-dependent current–voltage characteristics and impedance measurements of electrochemically doped poly(3-methylthiophene) devices at different doping levels. The extent of doping is carefully tailored such that only the bulk-limited transport mechanism prevails. A transition from exponentially distributed trap-limited transport to trap-free space-charge-limited current is observed in current–voltage conduction upon increasing the doping. The obtained trap densities (3.2 × 10 16 cm −3 and 8.6 × 10 15 cm −3 ) and trap energies (31.7 meV and 16.6 meV) for different devices signify the variation in disorder with doping, which is later supported by impedance measurements. Impedance–frequency data for various devices can not be explained using the parallel resistance–capacitance ( RC ) model in the equivalent circuit. However, this was established by incorporating a constant phase element Q (CPE) instead of the capacitance parameter. It should be emphasized that low doping devices in particular are best simulated with two CPE elements, while the data related to other devices are fitted well with a single CPE element. It is also observed from evaluated circuit parameters that the spatial inhomogeneity and disorder are the cause of variability in different samples, which has an excellent correlation with the temperature-dependent current–voltage characteristics. (paper)
Feng, Guo-Hua; Liu, Wei-Fan
2013-10-09
This paper presents the microfabrication of an acoustic impedance gradient matching layer on a spherically-shaped piezoelectric ultrasonic transducer. The acoustic matching layer can be designed to achieve higher acoustic energy transmission and operating bandwidth. Also included in this paper are a theoretical analysis of the device design and a micromachining technique to produce the novel transducer. Based on a design of a lead titanium zirconium (PZT) micropillar array, the constructed gradient acoustic matching layer has much better acoustic transmission efficiency within a 20-50 MHz operation range compared to a matching layer with a conventional quarter-wavelength thickness Parylene deposition. To construct the transducer, periodic microcavities are built on a flexible copper sheet, and then the sheet forms a designed curvature with a ball shaping. After PZT slurry deposition, the constructed PZT micropillar array is released onto a curved thin PZT layer. Following Parylene conformal coating on the processed PZT micropillars, the PZT micropillars and the surrounding Parylene comprise a matching layer with gradient acoustic impedance. By using the proposed technique, the fabricated transducer achieves a center frequency of 26 MHz and a -6 dB bandwidth of approximately 65%.
A 350 KV nanosecond pulse voltage generator with adjustable pulsed-width
International Nuclear Information System (INIS)
Wang, X.; Wang, M.; Chen, Y.Q.; Zeng, L.G.; Han, M.
2002-01-01
This paper presents a 350 kV nanosecond pulse voltage generator (NPVG). The voltage pulsed-width can be adjusted from 30 to 160 ns. The generator consists of: Marx generator, pulsed forming line (PFL), main switch and matched impedance. The output voltage of Marx generator is over than nU c (n- the stage number of Marx generator, U c -the charging voltage of capacitor). When the pulse forming line is terminated with an impedance that is over than the characteristic impedance of PFL, the higher voltage pulse was provided for the load
DC-based smart PV-powered home energy management system based on voltage matching and RF module
Hasan, W. Z. W.
2017-01-01
The main tool for measuring system efficiency in homes and offices is the energy monitoring of the household appliances’ consumption. With the help of GUI through a PC or smart phone, there are various applications that can be developed for energy saving. This work describes the design and prototype implementation of a wireless PV-powered home energy management system under a DC-distribution environment, which allows remote monitoring of appliances’ energy consumptions and power rate quality. The system can be managed by a central computer, which obtains the energy data based on XBee RF modules that access the sensor measurements of system components. The proposed integrated prototype framework is characterized by low power consumption due to the lack of components and consists of three layers: XBee-based circuit for processing and communication architecture, solar charge controller, and solar-battery-load matching layers. Six precise analogue channels for data monitoring are considered to cover the energy measurements. Voltage, current and temperature analogue signals were accessed directly from the remote XBee node to be sent in real time with a sampling frequency of 11–123 Hz to capture the possible surge power. The performance shows that the developed prototype proves the DC voltage matching concept and is able to provide accurate and precise results. PMID:28934271
DC-based smart PV-powered home energy management system based on voltage matching and RF module.
Directory of Open Access Journals (Sweden)
Ahmad H Sabry
Full Text Available The main tool for measuring system efficiency in homes and offices is the energy monitoring of the household appliances' consumption. With the help of GUI through a PC or smart phone, there are various applications that can be developed for energy saving. This work describes the design and prototype implementation of a wireless PV-powered home energy management system under a DC-distribution environment, which allows remote monitoring of appliances' energy consumptions and power rate quality. The system can be managed by a central computer, which obtains the energy data based on XBee RF modules that access the sensor measurements of system components. The proposed integrated prototype framework is characterized by low power consumption due to the lack of components and consists of three layers: XBee-based circuit for processing and communication architecture, solar charge controller, and solar-battery-load matching layers. Six precise analogue channels for data monitoring are considered to cover the energy measurements. Voltage, current and temperature analogue signals were accessed directly from the remote XBee node to be sent in real time with a sampling frequency of 11-123 Hz to capture the possible surge power. The performance shows that the developed prototype proves the DC voltage matching concept and is able to provide accurate and precise results.
DC-based smart PV-powered home energy management system based on voltage matching and RF module.
Sabry, Ahmad H; Hasan, W Z W; Ab Kadir, Mza; Radzi, M A M; Shafie, S
2017-01-01
The main tool for measuring system efficiency in homes and offices is the energy monitoring of the household appliances' consumption. With the help of GUI through a PC or smart phone, there are various applications that can be developed for energy saving. This work describes the design and prototype implementation of a wireless PV-powered home energy management system under a DC-distribution environment, which allows remote monitoring of appliances' energy consumptions and power rate quality. The system can be managed by a central computer, which obtains the energy data based on XBee RF modules that access the sensor measurements of system components. The proposed integrated prototype framework is characterized by low power consumption due to the lack of components and consists of three layers: XBee-based circuit for processing and communication architecture, solar charge controller, and solar-battery-load matching layers. Six precise analogue channels for data monitoring are considered to cover the energy measurements. Voltage, current and temperature analogue signals were accessed directly from the remote XBee node to be sent in real time with a sampling frequency of 11-123 Hz to capture the possible surge power. The performance shows that the developed prototype proves the DC voltage matching concept and is able to provide accurate and precise results.
Yan Hong; Yong Wang; Wang Ling Goh; Yuan Gao; Lei Yao
2015-08-01
This paper presents a mathematic method and a cost-efficient circuit to measure the value of each component of the bio-impedance model at electrode-electrolyte interface. The proposed current excited triple-time-voltage oversampling (TTVO) method deduces the component values by solving triple simultaneous electric equation (TSEE) at different time nodes during a current excitation, which are the voltage functions of time. The proposed triple simultaneous electric equations (TSEEs) allows random selections of the time nodes, hence numerous solutions can be obtained during a single current excitation. Following that, the oversampling approach is engaged by averaging all solutions of multiple TSEEs acquired after a single current excitation, which increases the practical measurement accuracy through the improvement of the signal-to-noise ratio (SNR). In addition, a print circuit board (PCB) that consists a switched current exciter and an analog-to-digital converter (ADC) is designed for signal acquisition. This presents a great cost reduction when compared against other instrument-based measurement data reported [1]. Through testing, the measured values of this work is proven to be in superb agreements on the true component values of the electrode-electrolyte interface model. This work is most suited and also useful for biological and biomedical applications, to perform tasks such as stimulations, recordings, impedance characterizations, etc.
Energy Technology Data Exchange (ETDEWEB)
Pal, S.K. [National Metallurgical Laboratory, Jamshedpur 831007 (India)], E-mail: skp@nmlindia.org; Panda, A.K.; Mitra, A. [National Metallurgical Laboratory, Jamshedpur 831007 (India)
2008-02-15
Second harmonic amplitude (V{sub 2f}) of Giant Magneto-Impedance (GMI) voltage of (Fe{sub 6}Co{sub 94}){sub 72.5}Si{sub 12.5}B{sub 15} amorphous wires has been studied systematically for as-cast and Joule-heated states. Joule heating was carried out at current densities (J) of 20-40 A/mm{sup 2} for a period of 1-300 min. The amplitude of the second harmonic voltage (V{sub 2f}) initially increased with J, attained maximum and then decreased. The maximum amplitude of V{sub 2f} was observed for the sample annealed at J=25 A/mm{sup 2} for 5 min. The second harmonic peak shifted towards the higher field side with the increase of annealing current density and annealing period (t{sub p}). Few-fold increase in the amplitude of the second harmonic voltage (V{sub 2f}) was observed due to the presence of a small dc bias current (I{sub b}) of amplitude 1 mA. The asymmetry in V{sub 2f} was also observed in a Joule-heated sample. The observed asymmetry has been correlated with the bias field generated between the surface crystallized layer and the amorphous core. A small bias current of amplitude of 1 mA was applied along the direction of the applied field to minimize the asymmetry.
Active Match Load Circuit Intended for Testing Piezoelectric Transformers
DEFF Research Database (Denmark)
Andersen, Thomas; Rødgaard, Martin Schøler; Andersen, Michael A. E.
2012-01-01
An adjustable high voltage active load circuit for voltage amplitudes above 100 volts, especially intended for resistive matching the output impedance of a piezoelectric transformer (PT) is proposed in this paper. PTs have been around for over 50 years, were C. A. Rosen is common known for his...
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Blaabjerg, Frede; Galigekere, Veda Prakash
2016-01-01
A novel A-source impedance network is proposed in this letter. The A-source impedance network uses an autotransformer for realizing converters for any application that demand a very high dc voltage gain. The network utilizes a minimal turns ratio compared to other Magnetically Coupled Impedance S...
Y-source impedance-network-based isolated boost DC/DC converter
DEFF Research Database (Denmark)
Siwakoti, Yam P.; Town, Graham; Loh, Poh Chiang
2014-01-01
A dc-dc converter with very high voltage gain is proposed in this paper for any medium-power application requiring a high voltage boost with galvanic isolation. The proposed converter topology can be realized using only two switches. With this topology a very high voltage boost can be achieved even...... with a relatively low duty cycle of the switches, and the gain obtainable is presently not matched by any existing impedance network based converter operated at the same duty ratio. The proposed converter has a Y-source impedance network to boost the voltage at the intermediate dc-link side and a push......-pull transformer for square-wave AC inversion and isolation. The voltage-doubler rectifier provides a constant dc voltage at the output stage. A theoretical analysis of the converter is presented, supported by simulation and experimental results. A 250 W down-scaled prototype was implemented in the laboratory...
Czech Academy of Sciences Publication Activity Database
Toušek, J.; Toušková, J.; Chomutová, R.; Křivka, I.; Hajná, Milena; Stejskal, Jaroslav
2017-01-01
Roč. 234, December (2017), s. 161-165 ISSN 0379-6779 R&D Projects: GA ČR(CZ) GA16-02787S Institutional support: RVO:61389013 Keywords : polyaniline * impedance spectroscopy * CELIV Subject RIV: CD - Macromolecular Chemistry OBOR OECD: Polymer science Impact factor: 2.435, year: 2016
Institute of Scientific and Technical Information of China (English)
Zehua Liu; Xiaocong Tian; Xu Xu; Liang He; Mengyu Yan; Chunhua Han; Yan Li; Wei Yang; Liqiang Mai
2017-01-01
Planar micro-supercapacitors show great potential as the energy storage unit in miniaturized electronic devices.Asymmetric structures have been widely investigated in micro-supercapacitors,and carbon-based materials are commonly applied in the electrodes.To integrate different metal oxides in both electrodes in micro-supercapacitors,the critical challenge is the pairing of different faradic metal oxides.Herein,we propose a strategy of matching the voltage and capacitance of two faradic materials that are fully integrated into one high-performance asymmetric micro-supercapadtor by a fadle and controllable fabrication process.The fabricated micro-supercapacitors employ MnO2 as the positive active material and Fe2O3 as the negative active material,respectively.The planar asymmetric micro-supercapacitors possess a high capacitance of 60 F·cm-3,a high energy density of 12 mW·h·cm-3,and a broad operation voltage range up to 1.2 V.
Energy Technology Data Exchange (ETDEWEB)
Nagarajan, Adarsh [National Renewable Energy Lab. (NREL), Golden, CO (United States); Coddington, Michael H. [National Renewable Energy Lab. (NREL), Golden, CO (United States); Brown, David [Sacramento Municipal Utility District, CA (United States); Hassan, Sheikh [Sacramento Municipal Utility District, CA (United States); Franciosa, Leonardo [Sacramento Municipal Utility District, CA (United States); Sison-Lebrilla, Elaine [Sacramento Municipal Utility District, CA (United States)
2018-01-09
Voltage regulators perform as desired when regulating from the source to the load and when regulating from a strong source (utility) to a weak source (distributed generation). (See the glossary for definitions of a strong source and weak source.) Even when the control is provisioned for reverse operation, it has been observed that tap-changing voltage regulators do not perform as desired in reverse when attempting regulation from the weak source to the strong source. The region of performance that is not as well understood is the regulation between sources that are approaching equal strength. As part of this study, we explored all three scenarios: regulator control from a strong source to a weak source (classic case), control from a weak source to a strong source (during reverse power flow), and control between equivalent sources.
Short-circuit impedance measurement
DEFF Research Database (Denmark)
Pedersen, Knud Ole Helgesen; Nielsen, Arne Hejde; Poulsen, Niels Kjølstad
2003-01-01
Methods for estimating the short-circuit impedance in the power grid are investigated for various voltage levels and situations. The short-circuit impedance is measured, preferably from naturally occurring load changes in the grid, and it is shown that such a measurement system faces different...
Intraesophageal impedance monitoring: clinical studies
Conchillo Armendáriz, J.M.
2007-01-01
Electrical impedance (Z) between two electrodes is the ratio between applied voltage (U) and resulting current (I). In electrical impedance monitoring the resistance to electrical flow in an alternating current circuit is measured. Multichannel esophageal monitoring can be measured by using an
Ecker, Bernhard; Egelhaaf, Hans-Joachim; Steim, Roland; Parisi, Juergen; von Hauff', Elizabeth
2012-01-01
In this study we propose an equivalent circuit model to describe S-shaped current–voltage (I–V) characteristics in inverted solar cells with a TiOx interlayer between the cathode and the poly(3-hexylthiophene):[6,6]-phenyl C61 butyric acid methyl ester active layer. Initially the solar cells
International Nuclear Information System (INIS)
Baroch, P; Potocky, S; Saito, N
2011-01-01
Presented work focuses on the investigation and characterization of plasma discharges generated in water by newly developed bipolar pulse power supply. The main aim of our work was to solve and overcome problems with intensive arc discharge transition when the discharge is ignited and maintained by a low output impedance pulse power supply. For this purpose a novel type of bipolar pulse power supply was developed and tested. It was found that two distinguished stable modes of discharges generated in the water can be realized. Effects of water conductivity, pulse frequency and initial water temperature on the discharge properties were investigated. Optical emission spectroscopy was employed to study plasma parameters of the discharge and the correlation between the data obtained from the optical emission spectroscopy and the chemical species measured in the water was carried out.
Low Voltage, High-Q SOI MEMS Varactors for RF Applications
DEFF Research Database (Denmark)
Yalcinkaya, Arda Deniz; Jensen, Søren; Hansen, Ole
2003-01-01
A micro electromechanical tunable capacitor with a low control voltage, a wide tuning range and high electrical quality factor is presented with detailed characterizations. A 50μm thick single-crystalline silicon layer was etched using deep reactive ion etching (DRIE) for obtaining high-aspect ra...... is a suitable passive component to be used in band-pass filtering, voltage controlled oscillator or impedance matching applications on the very high frequency(VHF) and ultra high frequency (UHF) bands....
Impedance Scaling and Impedance Control
International Nuclear Information System (INIS)
Chou, W.; Griffin, J.
1997-06-01
When a machine becomes really large, such as the Very Large Hadron Collider (VLHC), of which the circumference could reach the order of megameters, beam instability could be an essential bottleneck. This paper studies the scaling of the instability threshold vs. machine size when the coupling impedance scales in a ''normal'' way. It is shown that the beam would be intrinsically unstable for the VLHC. As a possible solution to this problem, it is proposed to introduce local impedance inserts for controlling the machine impedance. In the longitudinal plane, this could be done by using a heavily detuned rf cavity (e.g., a biconical structure), which could provide large imaginary impedance with the right sign (i.e., inductive or capacitive) while keeping the real part small. In the transverse direction, a carefully designed variation of the cross section of a beam pipe could generate negative impedance that would partially compensate the transverse impedance in one plane
Superconducting active impedance converter
International Nuclear Information System (INIS)
Ginley, D.S.; Hietala, V.M.; Martens, J.S.
1993-01-01
A transimpedance amplifier for use with high temperature superconducting, other superconducting, and conventional semiconductors allows for appropriate signal amplification and impedance matching to processing electronics. The amplifier incorporates the superconducting flux flow transistor into a differential amplifier configuration which allows for operation over a wide temperature range, and is characterized by high gain, relatively low noise, and response times less than 200 picoseconds over at least a 10-80 K. temperature range. The invention is particularly useful when a signal derived from either far-IR focal plane detectors or from Josephson junctions is to be processed by higher signal/higher impedance electronics, such as conventional semiconductor technology. 12 figures
Bootstrapped Low-Voltage Analog Switches
DEFF Research Database (Denmark)
Steensgaard-Madsen, Jesper
1999-01-01
Novel low-voltage constant-impedance analog switch circuits are proposed. The switch element is a single MOSFET, and constant-impedance operation is obtained using simple circuits to adjust the gate and bulk voltages relative to the switched signal. Low-voltage (1-volt) operation is made feasible...
Energy Technology Data Exchange (ETDEWEB)
Das, Tarun K.; Mandal, Sushil K. [CSIR - National Metallurgical Laboratory, NDE and Magnetic Materials Group, MST Division, Jamshedpur (India); Banerji, Pallab [Indian Institute of Technology, Kharagpur, Materials Science Centre, Kharagpur (India)
2016-11-15
We report a systematic study of the influence of wire length, L, dependence of giant magneto-impedance (GMI) sensitivity of Co{sub 66}Fe{sub 2}Cr{sub 4}Si{sub 13}B{sub 15} soft magnetic amorphous wire of diameter ∝ 100 μm developed by in-water quenching technique. The magnetization behaviour (hysteresis loops) of the wire with different length (L = 1, 2, 3, 5, 8 and 10 cm) has been evaluated by fuxmetric induction method. It was observed that the behaviour of the hysteresis loops change drastically with the wire length, being attributed to the existence of a critical length, L{sub C}, found to be around 3 cm. GMI measurements have been taken using automated GMI measurement system and the GMI sensitivities in terms of intrinsic impedance sensitivity (S{sub Ω/Am}{sup -1}) and voltage sensitivity (S{sub V/Am}{sup -1}) of the wire have been evaluated under optimal bias field and excitation current. It was found that the maximum (S{sub Ω/Am}{sup -1}){sub max} ∼ 0.63 Ω/kAm{sup -1}/cm and (S{sub V/Am}{sup -1}){sub max} ∼ 3.10 V/kAm{sup -1}/cm were achieved at a critical length L{sub C} ∝ 3 cm of the wire for an AC current of 5 mA and a frequency of 5 MHz. These findings provide crucial insights for optimization of the geometrical dimensions of magnetic sensing elements and important practical guidance for designing high sensitive GMI sensors. The relevant combinations of magnetic material parameters and operating conditions that optimize the sensitivity are highlighted. (orig.)
Hybrid-Source Impedance Networks
DEFF Research Database (Denmark)
Li, Ding; Gao, Feng; Loh, Poh Chiang
2010-01-01
Hybrid-source impedance networks have attracted attention among researchers because of their flexibility in performing buck-boost energy conversion. To date, three distinct types of impedance networks can be summarized for implementing voltage-type inverters with another three types summarized...... for current-type inverters. These impedance networks can in principle be combined into two generic network entities, before multiple of them can further be connected together by applying any of the two proposed generalized cascading concepts. The resulting two-level and three-level inverters implemented using...
Pastor-Fernández, Carlos; Uddin, Kotub; Chouchelamane, Gael H.; Widanage, W. Dhammika; Marco, James
2017-08-01
Degradation of Lithium-ion batteries is a complex process that is caused by a variety of mechanisms. For simplicity, ageing mechanisms are often grouped into three degradation modes (DMs): conductivity loss (CL), loss of active material (LAM) and loss of lithium inventory (LLI). State of Health (SoH) is typically the parameter used by the Battery Management System (BMS) to quantify battery degradation based on the decrease in capacity and the increase in resistance. However, the definition of SoH within a BMS does not currently include an indication of the underlying DMs causing the degradation. Previous studies have analysed the effects of the DMs using incremental capacity and differential voltage (IC-DV) and electrochemical impedance spectroscopy (EIS). The aim of this study is to compare IC-DV and EIS on the same data set to evaluate if both techniques provide similar insights into the causes of battery degradation. For an experimental case of parallelized cells aged differently, the effects due to LAM and LLI were found to be the most pertinent, outlining that both techniques are correlated. This approach can be further implemented within a BMS to quantify the causes of battery ageing which would support battery lifetime control strategies and future battery designs.
New Magnetically Coupled Impedance (Z-) Source Networks
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Blaabjerg, Frede; Loh, Poh Chiang
2016-01-01
Various Magnetically Coupled Impedance Source (MCIS) networks have been proposed in the literature for increasing voltage gain and modulation index simultaneously, while reducing the number of passive components used in the converter. However, applications of such networks have been limited...
LIA longitudinal coupling impedance
International Nuclear Information System (INIS)
Faltens, A.
1980-01-01
The beam generated fields enter into the problems of waveform generation and longitudinal stability. In the former, provision must be made for the longitudinally defocusing forces due to the space charge and the beam loading effects on the accelerating voltage due to the current of a presumably known bunch. In the latter, the concern is for the growth of unintentional perturbations to unacceptably large values through the interaction of the charge and current fluctuations with the rest of the beam and the surrounding structures. These beam generated electric fields may be related to the beam current through a coupling impedance
Quadratic Boost A-Source Impedance Network
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Blaabjerg, Frede; Chub, Andrii
2016-01-01
A novel quadratic boost A-source impedance network is proposed to realize converters that demand very high voltage gain. To satisfy the requirement, the network uses an autotransformer where the obtained gain is quadratically dependent on the duty ratio and is unmatched by any existing impedance...
Matching Impedances and Modes in Acoustic Levitation
Barmatz, M. B.
1985-01-01
Temperature differences accommodated with tunable coupler. Report discusses schemes for coupling sound efficiently from cool outside atmosphere into hot acoustic-levitation chamber. Theoretical studies have practical implications for material-processing systems that employ acoustic levitation.
Current Matching in Multifold DBP/C70 Organic Solar Cells With Open-Circuit Voltages of up to 6.44 V
DEFF Research Database (Denmark)
Ahmadpour, Mehrad; Liu, Yiming; Rubahn, Horst-Günter
2017-01-01
In this paper, we demonstrate a novel method for achieving high open-circuit voltages (Voc) in organic solar cells based on tetraphenyldibenzoperiflanthen (DBP) as donor and fullerene (C70) as acceptor molecules, by fabrication of multifold bilayer single cells stacked on top of each other...
Modified impedance source inverter for power conditioning system
Indian Academy of Sciences (India)
DC link voltage boost, reduced total harmonic distortion of output current and voltage, better voltage gain and wide range of output voltage controlcan be achieved easily with improved power quality. Experimental set-up of the modified impedance source inverter with Field Programmable Gate Array (FPGA) controller has ...
International Nuclear Information System (INIS)
Detch, J.L. Jr.; Noel, B.W.
1981-01-01
Electrically gated proximity-focused channel intensifier tubes are often used as optical shutters. Optimum nanosecond shuttering requires both understanding the electrical pulse propagation across the device structure and proper impedance matching. A distributed-transmission-line model is developed that describes analytically the voltage- and current-wave propagation characteristics as functions of time for any point on the surface. The optical gain's spatial uniformity and shutter-open times are shown to depend on the electrical pulse width and amplitude, and on the applied bias. The driving-point impedance is derived from the model and is expressed as a function of an infinite sum of terms in the complex frequency. The synthesis in terms of lumped-constant network elements is realized in first- and second-Foster equivalent circuits. Experimental impedance data are compared with the model's predictions and deviations from the ideal model are discussed
Suppressing voltage transients in high voltage power supplies
International Nuclear Information System (INIS)
Lickel, K.F.; Stonebank, R.
1979-01-01
A high voltage power supply for an X-ray tubes includes voltage adjusting means, a high voltage transformer, switch means connected to make and interrupt the primary current of the transformer, and over-voltage suppression means to suppress the voltage transient produced when the current is switched on. In order to reduce the power losses in the suppression means, an impedance is connected in the transformer primary circuit on operation of the switch means and is subsequently short-circuited by a switch controlled by a timer after a period which is automatically adjusted to the duration of the transient overvoltage. (U.K.)
MVAC Submarine cable, impedance measurements and analysis
DEFF Research Database (Denmark)
Arentsen, Martin Trolle; Expethit, Adrian; Pedersen, Morten Virklund
2017-01-01
influence the losses. Secondly, cable parameters such as component impedances, positive-, zero-sequence impedances and losses are measured for varying currents and frequencies. Zero sequence is measured for two setups, one with armour grounded, and with armour open to match CIGREs formula assumption......-sequence impedance is found between the measurements and the CIGRÉ formulas. It is concluded that the formulas are not accurate for the specific cable under test....
Estimating the short-circuit impedance
DEFF Research Database (Denmark)
Nielsen, Arne Hejde; Pedersen, Knud Ole Helgesen; Poulsen, Niels Kjølstad
1997-01-01
A method for establishing a complex value of the short-circuit impedance from naturally occurring variations in voltage and current is discussed. It is the symmetrical three phase impedance at the fundamental grid frequency there is looked for. The positive sequence components in voltage...... and current are derived each period, and the short-circuit impedance is estimated from variations in these components created by load changes in the grid. Due to the noisy and dynamic grid with high harmonic distortion it is necessary to threat the calculated values statistical. This is done recursively...... through a RLS-algorithm. The algorithms have been tested and implemented on a PC at a 132 kV substation supplying a rolling mill. Knowing the short-circuit impedance gives the rolling mill an opportunity to adjust the arc furnace operation to keep flicker below a certain level. Therefore, the PC performs...
DEFF Research Database (Denmark)
Kwon, Jun Bum; Wang, Xiongfei; Blaabjerg, Frede
2014-01-01
This paper investigates the effect of grid impedance variation on harmonic resonant current controllers for gridconnected voltage source converters by means of impedance-based analysis. It reveals that the negative harmonic resistances tend to be derived from harmonic resonant controllers...... in the closed-loop output admittance of converter. Such negative resistances may interact with the grid impedance resulting in steady state error or unstable harmonic compensation. To deal with this problem, a design guideline for harmonic resonant controllers under a wide range of grid impedance is proposed...
International Nuclear Information System (INIS)
Stupakov, G.
2009-01-01
We review recent progress in the following areas of the impedance theory: calculation of impedance of tapers and small angle collimators; optical approximation and parabolic equation for the high-frequency impedance; impedance due to resistive inserts in a perfectly conducting pipe.
Microgrid Reactive and Harmonic Power Sharing Using Enhanced Virtual Impedance
DEFF Research Database (Denmark)
He, Jinwei; Wei Li, Yun; Guerrero, Josep M.
2013-01-01
To address the load sharing problem in islanding microgrids, this paper proposes an improved approach which regulates the distributed generation (DG) unit interfacing virtual impedance at fundamental and selected harmonic frequencies. In contrast to the conventional virtual impedance control where...... only a line current feed-forward term is added to the DG voltage reference, the proposed virtual impedances at fundamental and harmonic frequencies are realized using DG line current and point of common coupling (PCC) voltage feed-forward terms, respectively. With this modification, the mismatched DG...... feeder impedances can be properly compensated, resulting in accurate reactive and harmonic power sharing at the same time. In addition, this paper shows that the microgrid PCC harmonic voltages can be mitigated by reducing the magnitude of DG unit equivalent harmonic impedance. Finally, an improved...
Improved impedance transformation between microwave oscillator and Josephson junction series array
International Nuclear Information System (INIS)
Gutmann, P.; Vollmer, E.; Niemeyer, J.
1993-01-01
Superconducting microwave monolithic integrated circuits (S-MMIC), based on Josephson tunnel junctions, are a well-established tool to reproduce the volt at the highest level of accuracy. An external oscillator of a fixed frequency f supplies microwave energy through a waveguide to the S-MMIC. The wave changes its mode at a waveguide-antipodal finline-stripline taper before entering a series array stripline of up to 30 000 Josephson tunnel junctions and is dissipated as heat in a lossy stripline. Both striplines have a characteristic impedance Z of 2 to 5 Ω. An equivalent circuit is shown in figure 1. The oscillator is matched to the waveguide with a source resistance R G Z(waveguide) ∼ 550 Ω. The most critical part is the taper, which should work as a lossless impedance matching network at the frequency of the oscillator. Microwave energy is fed into the tunnel junctions by the surface current I HF of the travelling wave in the series array stripline producing an rf voltage amplitude U JHF across the capacitance C of each junction. The Josephson tunnel junctions work as self-oscillating parametric mixers producing steps of constant voltage V in the current-voltage characteristic whenever (nf - 2eV/h) = 0, with n denoting an integer and e and h denoting the elementary charge and Planck's constant, respectively. The equivalent circuit of a Josephson tunnel element used in a voltage standard for 1 V working at a frequency of f = 70 GHz is given by a lumped parallel resonant circuit with a nonlinear inductance on the order of L = φ 0 /2πI 0 ∼ 1 pH, flux quantum φ 0 = h/2e and a linear capacitance of C ∼ 40 pF. These tunnel junctions have a maximum zero voltage current of approximately I 0 ∼ 350 μA. (orig.)
Magnetically Coupled Impedance-Source Inverters
DEFF Research Database (Denmark)
Loh, Poh Chiang; Blaabjerg, Frede
2013-01-01
input-to-output gain and the presence of an impedance network. The former means a high dc-link voltage, which can stress the semiconductor switches unnecessarily. The latter leads to increases in cost and size, which similarly are undesirable. To lessen these concerns, an interesting approach is to use...
On Impedance Spectroscopy of Supercapacitors
Uchaikin, V. V.; Sibatov, R. T.; Ambrozevich, A. S.
2016-10-01
Supercapacitors are often characterized by responses measured by methods of impedance spectroscopy. In the frequency domain these responses have the form of power-law functions or their linear combinations. The inverse Fourier transform leads to relaxation equations with integro-differential operators of fractional order under assumption that the frequency response is independent of the working voltage. To compare long-term relaxation kinetics predicted by these equations with the observed one, charging-discharging of supercapacitors (with nominal capacitances of 0.22, 0.47, and 1.0 F) have been studied by means of registration of the current response to a step voltage signal. It is established that the reaction of devices under study to variations of the charging regime disagrees with the model of a homogeneous linear response. It is demonstrated that relaxation is well described by a fractional stretched exponent.
Electrical Impedance Measurements of PZT Nanofiber Sensors
Directory of Open Access Journals (Sweden)
Richard Galos
2017-01-01
Full Text Available Electrical impedance measurements of PZT nanofiber sensors were performed using a variety of methods over a frequency spectrum ranging from DC to 1.8 GHz. The nanofibers formed by electrospinning with diameters ranging from 10 to 150 nm were collected and integrated into sensors using microfabrication techniques. Special matching circuits with ultrahigh input impedance were fabricated to produce low noise, measurable sensor outputs. Material properties including resistivity and dielectric constant are derived from the impedance measurements. The resulting material properties are also compared with those of individual nanofibers being tested using conductive AFM and Scanning Conductive Microscopy.
Plummer, MD
1986-01-01
This study of matching theory deals with bipartite matching, network flows, and presents fundamental results for the non-bipartite case. It goes on to study elementary bipartite graphs and elementary graphs in general. Further discussed are 2-matchings, general matching problems as linear programs, the Edmonds Matching Algorithm (and other algorithmic approaches), f-factors and vertex packing.
Fan, Xu; Wang, Yunguang; Cheng, Haiping; Chong, Xiaochen
2016-02-01
The present circuit was designed to apply to human tissue impedance tuning and matching device in ultra-short wave treatment equipment. In order to judge if the optimum status of circuit parameter between energy emitter circuit and accepter circuit is in well syntony, we designed a high frequency envelope detect circuit to coordinate with automatic adjust device of accepter circuit, which would achieve the function of human tissue impedance matching and tuning. Using the sampling coil to receive the signal of amplitude-modulated wave, we compared the voltage signal of envelope detect circuit with electric current of energy emitter circuit. The result of experimental study was that the signal, which was transformed by the envelope detect circuit, was stable and could be recognized by low speed Analog to Digital Converter (ADC) and was proportional to the electric current signal of energy emitter circuit. It could be concluded that the voltage, transformed by envelope detect circuit can mirror the real circuit state of syntony and realize the function of human tissue impedance collecting.
Hybrid-source impedance network and its generalized cascading concepts
DEFF Research Database (Denmark)
Li, Ding; Gao, Feng; Loh, Poh Chiang
2009-01-01
Hybrid-source impedance networks have attracted attention among researchers because of their flexibility in performing buck-boost energy conversion. To date, three distinct types of impedance networks can be summarized for implementing voltage-type inverters, with another three types summarized...... for current-type inverters. These impedance networks can in principle be combined into a single generic network entity, before generalized cascading concepts are proposed for connecting multiple of them together to form energy converters with a higher output voltage gain and other unique advantages...
Wideband impedance measurements of DC motors under dynamic load conditions
Diouf, F.; Buesink, Frederik Johannes Karel; Leferink, Frank Bernardus Johannes; Duval, Fabrice; Bensetti, Mohamed
2013-01-01
One of the principal conducted EMI(electromagnetic interferences) sources of low voltage DC (direct current) motors is the commutation occurring during rotation. In this paper the small-signal impedance of low voltage DC motors under different functioning modes, including the dynamic one is studied
Nonsynchronous Noncommensurate Impedance Transformers
DEFF Research Database (Denmark)
Zhurbenko, Vitaliy; Kim, K
2012-01-01
Nonsynchronous noncommensurate impedance transformers consist of a combination of two types of transmission lines: transmission lines with a characteristic impedance equal to the impedance of the source, and transmission lines with a characteristic impedance equal to the load. The practical...... advantage of such transformers is that they can be constructed using sections of transmission lines with a limited variety of characteristic impedances. These transformers also provide comparatively compact size in applications where a wide transformation ratio is required. This paper presents the data...... matrix approach and experimentally verified by synthesizing a 12-section nonsynchronous noncommensurate impedance transformer. The measured characteristics of the transformer are compared to the characteristics of a conventional tapered line transformer....
Method for conducting nonlinear electrochemical impedance spectroscopy
Adler, Stuart B.; Wilson, Jamie R.; Huff, Shawn L.; Schwartz, Daniel T.
2015-06-02
A method for conducting nonlinear electrochemical impedance spectroscopy. The method includes quantifying the nonlinear response of an electrochemical system by measuring higher-order current or voltage harmonics generated by moderate-amplitude sinusoidal current or voltage perturbations. The method involves acquisition of the response signal followed by time apodization and fast Fourier transformation of the data into the frequency domain, where the magnitude and phase of each harmonic signal can be readily quantified. The method can be implemented on a computer as a software program.
Microfluidic device for cell capture and impedance measurement.
Jang, Ling-Sheng; Wang, Min-How
2007-10-01
This work presents a microfluidic device to capture physically single cells within microstructures inside a channel and to measure the impedance of a single HeLa cell (human cervical epithelioid carcinoma) using impedance spectroscopy. The device includes a glass substrate with electrodes and a PDMS channel with micro pillars. The commercial software CFD-ACE+ is used to study the flow of the microstructures in the channel. According to simulation results, the probability of cell capture by three micro pillars is about 10%. An equivalent circuit model of the device is established and fits closely to the experimental results. The circuit can be modeled electrically as cell impedance in parallel with dielectric capacitance and in series with a pair of electrode resistors. The system is operated at low frequency between 1 and 100 kHz. In this study, experiments show that the HeLa cell is successfully captured by the micro pillars and its impedance is measured by impedance spectroscopy. The magnitude of the HeLa cell impedance declines at all operation voltages with frequency because the HeLa cell is capacitive. Additionally, increasing the operation voltage reduces the magnitude of the HeLa cell because a strong electric field may promote the exchange of ions between the cytoplasm and the isotonic solution. Below an operating voltage of 0.9 V, the system impedance response is characteristic of a parallel circuit at under 30 kHz and of a series circuit at between 30 and 100 kHz. The phase of the HeLa cell impedance is characteristic of a series circuit when the operation voltage exceeds 0.8 V because the cell impedance becomes significant.
International Nuclear Information System (INIS)
Greffet, Jean-Jacques; Laroche, Marine; Marquier, Francois
2009-01-01
We introduce a generalized definition of the impedance of a nanoantenna that can be applied to any system. We also introduce a definition of the impedance of a two level system. Using this framework, we establish a link between the electrical engineering and the quantum optics picture of light emission.
RHIC injection kicker impedance
International Nuclear Information System (INIS)
Mane, V.; Peggs, S.; Trbojevic, D.; Zhang, W.
1995-01-01
The longitudinal impedance of the RHIC injection kicker is measured using the wire method up to a frequency of 3 GHz. The mismatch between the 50 ohm cable and the wire and pipe system is calibrated using the TRL calibration algorithm. Various methods of reducing the impedance, such as coated ceramic pipe and copper strips are investigated
Spheromak Impedance and Current Amplification
International Nuclear Information System (INIS)
Fowler, T K; Hua, D D; Stallard, B W
2002-01-01
It is shown that high current amplification can be achieved only by injecting helicity on the timescale for reconnection, τ REC , which determines the effective impedance of the spheromak. An approximate equation for current amplification is: dI TOR 2 /dt ∼ I 2 /τ REC - I TOR 2 /τ closed where I is the gun current, I TOR is the spheromak toroidal current and τ CLOSED is the ohmic decay time of the spheromak. Achieving high current amplification, I TOR >> I, requires τ REC CLOSED . For resistive reconnection, this requires reconnection in a cold zone feeding helicity into a hot zone. Here we propose an impedance model based on these ideas in a form that can be implemented in the Corsica-based helicity transport code. The most important feature of the model is the possibility that τ REC actually increases as the spheromak temperature increases, perhaps accounting for the ''voltage sag'' observed in some experiments, and a tendency toward a constant ratio of field to current, B ∝ I, or I TOR ∼ I. Program implications are discussed
Harmonic current prediction by impedance modeling of grid-tied inverters
DEFF Research Database (Denmark)
Pereira, Heverton A.; Freijedo, Francisco D.; Silva, M. M.
2017-01-01
and harmonic voltage profiles. Results reinforce that impedance models can represent with relatively accuracy the harmonic current emitted by the PV plants at the point of common coupling (PCC). Lastly, a stress test is performed to show how a variation in the harmonic voltage phase angle impacts the PV plant...... impedance models when used in harmonic integration studies. It is aimed to estimate the harmonic current contribution as a function of the background harmonic voltages components. Time domain simulations based on detailed and average models are compared with the impedance model developed in frequency domain....... In grids with harmonic voltages, impedance models can predict the current distortion for all active power injection scenarios. Furthermore, measurements in a 1.4 MW PV plant connected in a distributed grid are used to validate the simulation based on impedance models during different power injections...
Impedance and Collective Effects
Metral, E; Rumolo, R; Herr, W
2013-01-01
This document is part of Subvolume C 'Accelerators and Colliders' of Volume 21 'Elementary Particles' of Landolt-Börnstein - Group I 'Elementary Particles, Nuclei and Atoms'. It contains the Chapter '4 Impedance and Collective Effects' with the content: 4 Impedance and Collective Effects Introduction 4.1 Space Charge 4.2 Wake Fields and Impedances 4.3 Coherent Instabilities 4.4 Landau Damping 4.5 Two-Stream Effects (Electron Cloud and Ions) 4.6 Beam-Beam Effects 4.7 Numerical Modelling
Broadband impedance of the NESTOR storage ring
International Nuclear Information System (INIS)
Androsov, V.P.; Gladkikh, P.I.; Gvozd, A.M.; Karnaukhov, I.M.; Telegin, Yu.N.
2011-01-01
The contributions from lossy and inductive vacuum chamber components to the broadband impedance of the NESTOR storage ring are obtained by using both low-frequency analytical approaches and computer simulations. As was expected considering the small ring circumference (15.44m), the main contributions both to the longitudinal impedance Z || /n and the loss factor k loss come from the RF-cavity. Cavity impedance was also estimated with CST Microwave Studio (CST Studio Suite TM 2006) by simulating coaxial wire method commonly used for impedance measurements. Both estimates agree well. Finally, we performed the simulations of a number of inductive elements with CST Particle Studio 2010 by using wake field solver. We have also evaluated the bunch length in NESTOR taking the conservative estimate of 3 Ohm for the ring broadband impedance and have found that the bunch length s z = 0.5 cm could be obtained in steady state operation mode for the designed bunch current of 10 mA and RF-voltage of 250 kV.
Characterization of lithium-thionyl chloride cells by impedance techniques
Walsh, F.; Pozin, M.; Cherniy, A.; Tikhonov, K.
The main contributor to voltage drop observed on initial discharge of lithium-thionyl chloride cells is the resistance of the passive layer on the lithium anode, as can be determined from the Nyquist plot of a lithium-thionyl chloride cell. At extremely low discharge currents, initial voltage drop corresponds to the value found from the impedance measurements; at higher current, an empirical correction based on the experimental results is required. The dispersion in the values of the impedance parameters and thus in initial voltage drop of individual cells was analyzed. The condition of the lithium surface after assembly was shown not to be the only reason for high dispersion in impedance parameter values.
Bias-dependent model of the electrical impedance of ionic polymer-metal composites.
Cha, Youngsu; Porfiri, Maurizio
2013-02-01
In this paper, we analyze the charge dynamics of ionic polymer-metal composites (IPMCs) in response to voltage inputs composed of a large dc bias and a small superimposed time-varying voltage. The IPMC chemoelectrical behavior is described through the modified Poisson-Nernst-Planck framework, in which steric effects are taken into consideration. The physics of charge build-up and mass transfer in the proximity of the high surface electrodes is modeled by schematizing the IPMC as the stacked sequence of five layers, in which the ionomeric membrane is separated from the metal electrodes by two composite layers. The method of matched asymptotic expansions is used to derive a semianalytical solution for the concentration of mobile counterions and the electric potential in the IPMC, which is, in turn, used to establish an equivalent circuit model for the IPMC electrical response. The circuit model consists of the series connection of a resistor and two complex elements, each constituted by the parallel connection of a capacitor and a Warburg impedance. The resistor is associated with ion transport in the ionomeric membrane and is independent of the dc bias. The capacitors and the Warburg impedance idealize charge build-up and mass transfer in the vicinity of the electrodes and their value is controlled by the dc bias. The proposed approach is validated against experimental results on in-house fabricated IPMCs and the accuracy of the equivalent circuit is assessed through comparison with finite element results.
Directory of Open Access Journals (Sweden)
Supachai Klungtong
2017-01-01
Full Text Available This paper presents a second-order voltage-mode filter with three inputs and single-output voltage using single commercially available IC, one resistor, and two capacitors. The used commercially available IC, called LT1228, is manufactured by Linear Technology Corporation. The proposed filter is based on parallel RLC circuit. The filter provides five output filter responses, namely, band-pass (BP, band-reject (BR, low-pass (LP, high-pass (HP, and all-pass (AP functions. The selection of each filter response can be done without the requirement of active and passive component matching condition. Furthermore, the natural frequency and quality factor are electronically controlled. Besides, the nonideal case is also investigated. The output voltage node exhibits low impedance. The experimental results can validate the theoretical analyses.
Twenty-channel high-voltage pulse generators
International Nuclear Information System (INIS)
Anan'in, P.S.; Kashirin, A.P.
1980-01-01
A 20-channel high-voltage pulse generator operating with a mismatched load is described. The generator contains shaping lines 20 m long made of coaxial cable, a trigatron-type discharged, and isolating plates. The channel characteristic impedance is 50 Ohm. The maximum pulse amplitude is up to 15 kV on a high-resistance load and 7.5 kV on a matched one. The pulse duration is 100 ns at a pulse rise time of 12 ns, the delay introduced by the generator is 200 +-2.5 ns. Provision is made in the control circuit for compensation of the shaped pulse and separation of a pulse reflected from the load. The reflected pulse shape and amplitude characterize load parameters. Generator tests proved its high operational reliability (after 10 5 operations no significant changes in generator performances have been observed). The generator is intended for filmless data output from spark chambers
Electrical Impedance Tomography Technology
National Aeronautics and Space Administration — The goal for the Electrical Impedance Tomography Technology (EITT) project is to develop a reliable portable, lightweight device providing two-dimensional...
Impedance and component heating
Métral, E; Mounet, N; Pieloni, T; Salvant, B
2015-01-01
The impedance is a complex function of frequency, which represents, for the plane under consideration (longitudinal, horizontal or vertical), the force integrated over the length of an element, from a “source” to a “test” wave, normalized by their charges. In general, the impedance in a given plane is a nonlinear function of the test and source transverse coordinates, but it is most of the time sufficient to consider only the first few linear terms. Impedances can influence the motion of trailing particles, in the longitudinal and in one or both transverse directions, leading to energy loss, beam instabilities, or producing undesirable secondary effects such as excessive heating of sensitive components at or near the chamber wall, called beam-induced RF heating. The LHC performance limitations linked to impedances encountered during the 2010-2012 run are reviewed and the currently expected situation during the HL-LHC era is discussed.
Energy Technology Data Exchange (ETDEWEB)
Zotter, B [European Organization for Nuclear Research, Geneva (Switzerland)
1996-08-01
This report describes a number of measurements and computations of the impedance of the Large Electron Positron collider LEP at CERN. The work has been performed over several years, together with D. Brandt, K. Cornelis, A. Hofmann, G. Sabbi and many others. The agreement between measurements of single bunch instabilities on the machine and computer simulations is in general excellent and gives confidence in the impedance model used. (author)
Subnanosecond-rise-time, low-impedance pulse generator
International Nuclear Information System (INIS)
Druce, R.; Vogtlin, G.
1983-01-01
This paper describes a fast rise, low-impedance pulse generator that has been developed at the Lawrence Livermore National Laboratory. The design specifications of this generator are: 50-kV operating voltage, 1-ohm output impedance, subnanosecond rise time, and a 2 to 10 nanosecond pulse length. High repetition rate is not required. The design chosen is a parallel-plate, folded Blumlein generator. A tack switch is utilized for its simple construction and high performance. The primary diagnostic is a capacitive voltage divider with a B probe used to measure the current waveform
Subnanosecond-rise-time, low-impedance pulse generator
Energy Technology Data Exchange (ETDEWEB)
Druce, R.; Vogtlin, G.
1983-06-03
This paper describes a fast rise, low-impedance pulse generator that has been developed at the Lawrence Livermore National Laboratory. The design specifications of this generator are: 50-kV operating voltage, 1-ohm output impedance, subnanosecond rise time, and a 2 to 10 nanosecond pulse length. High repetition rate is not required. The design chosen is a parallel-plate, folded Blumlein generator. A tack switch is utilized for its simple construction and high performance. The primary diagnostic is a capacitive voltage divider with a B probe used to measure the current waveform.
Subnanosecond-rise-time, low-impedance pulse generator
Druce, R.; Vigtlin, G.
1983-06-01
A fast rise, low impedance pulse generator developed at the Lawrence Livermore National Laboratory is described. The design specifications of this generator are: 50-kV operating voltage, 1-ohm output impedance, subnanosecond rise time, and a 2 to 10 nanosecond pulse length. High repetition rate is not required. The design chosen is a parallel plate, folded Blumlein generator. A tack switch is utilized for its simple construction and high performance. The primary diagnostic is a capacitive voltage divider with a B probe used to measure the current waveform.
International Nuclear Information System (INIS)
Bogónez-Franco, P; Nescolarde, L; Bragós, R; Rosell-Ferrer, J; Yandiola, I
2009-01-01
The purpose of this study is to compare measurement errors in two commercially available multi-frequency bioimpedance analyzers, a Xitron 4000B and an ImpediMed SFB7, including electrode impedance mismatch. The comparison was made using resistive electrical models and in ten human volunteers. We used three different electrical models simulating three different body segments: the right-side, leg and thorax. In the electrical models, we tested the effect of the capacitive coupling of the patient to ground and the skin–electrode impedance mismatch. Results showed that both sets of equipment are optimized for right-side measurements and for moderate skin–electrode impedance mismatch. In right-side measurements with mismatch electrode, 4000B is more accurate than SFB7. When an electrode impedance mismatch was simulated, errors increased in both bioimpedance analyzers and the effect of the mismatch in the voltage detection leads was greater than that in current injection leads. For segments with lower impedance as the leg and thorax, SFB7 is more accurate than 4000B and also shows less dependence on electrode mismatch. In both devices, impedance measurements were not significantly affected (p > 0.05) by the capacitive coupling to ground
Effect of single aerosol droplets on plasma impedance in the inductively coupled plasma
Energy Technology Data Exchange (ETDEWEB)
Chan, George C.-Y., E-mail: gcchan@indiana.edu; Zhu, Zhenli; Hieftje, Gary M.
2012-10-15
The impedance of an inductively coupled plasma was indirectly monitored by two different means-through a RF-probe coil placed inside the torch housing and from tapping the phase-detector signal of the impedance-matching network. During single-droplet introduction, temporal spikes in both the RF-probe coil and the phase-detector signals were readily observed, indicating a momentary change in plasma impedance. The changes in plasma impedance were found to be due solely to plasma perturbation by droplet introduction, and not to an artifact caused by imperfect automatic impedance matching. The temporal changes in plasma impedance were found to be directly proportional to the temporally integrated atomic emission of hydrogen, which is assumed in turn to be directly proportional to the volume of the introduced droplet. A small satellite droplet, with an estimated diameter of 27 {mu}m (i.e., {approx} 10 pL in volume), caused a readily measurable change in plasma impedance. By assuming that the change in RF-probe voltage is directly proportional to the variation in RF power delivered by the load coil, the instantaneous power change coupled to the plasma during single-droplet introduction was estimated. Typical increases in peak RF power and total energy coupled to the plasma, for a single 50-{mu}m droplet introduction, were thereby estimated to be around 8 to 11 W and 0.03 to 0.04 J, respectively. This impedance change was also exploited as a trigger to signal the droplet-introduction event into the plasma. This trigger signal was obtained through a combination of the RF-probe and the phase-detector signals and offered typical jitter from 1 to 2 ms. With the proper choice of a trigger threshold, no trigger misfire resulted and the achievable efficiencies of the trigger signal were 99.95, 97.18 and 74.33% for plasma forward power levels of 900, 1200, and 1500 W, respectively. The baseline noise on the RF-probe coil and the phase-detector signals, which increase with plasma
Impedance-Source Networks for Electric Power Conversion Part II
DEFF Research Database (Denmark)
Siwakoti, Yam P.; Peng, Fang Zheng; Blaabjerg, Frede
2015-01-01
Impedance-source networks cover the entire spectrum of electric power conversion applications (dc-dc, dc-ac, ac-dc, ac-ac) controlled and modulated by different modulation strategies to generate the desired dc or ac voltage and current at the output. A comprehensive review of various impedance......-source-network-based power converters has been covered in a previous paper and main topologies were discussed from an application point of view. Now Part II provides a comprehensive review of the most popular control and modulation strategies for impedance-source network-based power converters/inverters. These methods...
Electrochemical Characterization of a PEMEC Using Impedance Spectroscopy
DEFF Research Database (Denmark)
Elsøe, Katrine; Grahl-Madsen, L.; Hjelm, Johan
2017-01-01
In this study, electrochemical impedance spectroscopy (EIS) is applied in combination with cyclic voltammetry (CV) and current density – cell voltage curves (iV-curves) to investigate the processes contributing to the total impedance of a polymer electrolyte membrane electrolysis cell (PEMEC). i......V-curves were linear above 0.35 A cm−2 implying ohmic processes to be performance limiting, however the impedance spectra showed three arcs indicating three electrochemical reactions at these conditions not to be purely ohmic, but also to have capacitive properties. A hypothesis that the composite Ir...
A Versatile and Reproducible Multi-Frequency Electrical Impedance Tomography System
Directory of Open Access Journals (Sweden)
James Avery
2017-01-01
Full Text Available A highly versatile Electrical Impedance Tomography (EIT system, nicknamed the ScouseTom, has been developed. The system allows control over current amplitude, frequency, number of electrodes, injection protocol and data processing. Current is injected using a Keithley 6221 current source, and voltages are recorded with a 24-bit EEG system with minimum bandwidth of 3.2 kHz. Custom PCBs interface with a PC to control the measurement process, electrode addressing and triggering of external stimuli. The performance of the system was characterised using resistor phantoms to represent human scalp recordings, with an SNR of 77.5 dB, stable across a four hour recording and 20 Hz to 20 kHz. In studies of both haeomorrhage using scalp electrodes, and evoked activity using epicortical electrode mats in rats, it was possible to reconstruct images matching established literature at known areas of onset. Data collected using scalp electrode in humans matched known tissue impedance spectra and was stable over frequency. The experimental procedure is software controlled and is readily adaptable to new paradigms. Where possible, commercial or open-source components were used, to minimise the complexity in reproduction. The hardware designs and software for the system have been released under an open source licence, encouraging contributions and allowing for rapid replication.
Directory of Open Access Journals (Sweden)
Heng-Yi Su
2016-11-01
Full Text Available This paper proposes an efficient approach for the computation of voltage stability margin (VSM in a large-scale power grid. The objective is to accurately and rapidly determine the load power margin which corresponds to voltage collapse phenomena. The proposed approach is based on the impedance match-based technique and the model-based technique. It combines the Thevenin equivalent (TE network method with cubic spline extrapolation technique and the continuation technique to achieve fast and accurate VSM computation for a bulk power grid. Moreover, the generator Q limits are taken into account for practical applications. Extensive case studies carried out on Institute of Electrical and Electronics Engineers (IEEE benchmark systems and the Taiwan Power Company (Taipower, Taipei, Taiwan system are used to demonstrate the effectiveness of the proposed approach.
Energy Technology Data Exchange (ETDEWEB)
Kopp, Joachim; Liu, Jia [PRISMA Cluster of Excellence & Mainz Institute for Theoretical Physics,Johannes Gutenberg University,Staudingerweg 7, 55099 Mainz (Germany); Slatyer, Tracy R. [Center for Theoretical Physics, Massachusetts Institute of Technology,Cambridge, MA 02139 (United States); Wang, Xiao-Ping [PRISMA Cluster of Excellence & Mainz Institute for Theoretical Physics,Johannes Gutenberg University,Staudingerweg 7, 55099 Mainz (Germany); Xue, Wei [Center for Theoretical Physics, Massachusetts Institute of Technology,Cambridge, MA 02139 (United States)
2016-12-12
We consider dark matter models in which the mass splitting between the dark matter particles and their annihilation products is tiny. Compared to the previously proposed Forbidden Dark Matter scenario, the mass splittings we consider are much smaller, and are allowed to be either positive or negative. To emphasize this modification, we dub our scenario “Impeded Dark Matter”. We demonstrate that Impeded Dark Matter can be easily realized without requiring tuning of model parameters. For negative mass splitting, we demonstrate that the annihilation cross-section for Impeded Dark Matter depends linearly on the dark matter velocity or may even be kinematically forbidden, making this scenario almost insensitive to constraints from the cosmic microwave background and from observations of dwarf galaxies. Accordingly, it may be possible for Impeded Dark Matter to yield observable signals in clusters or the Galactic center, with no corresponding signal in dwarfs. For positive mass splitting, we show that the annihilation cross-section is suppressed by the small mass splitting, which helps light dark matter to survive increasingly stringent constraints from indirect searches. As specific realizations for Impeded Dark Matter, we introduce a model of vector dark matter from a hidden SU(2) sector, and a composite dark matter scenario based on a QCD-like dark sector.
International Nuclear Information System (INIS)
Kopp, Joachim; Liu, Jia; Slatyer, Tracy R.; Wang, Xiao-Ping; Xue, Wei
2016-01-01
We consider dark matter models in which the mass splitting between the dark matter particles and their annihilation products is tiny. Compared to the previously proposed Forbidden Dark Matter scenario, the mass splittings we consider are much smaller, and are allowed to be either positive or negative. To emphasize this modification, we dub our scenario “Impeded Dark Matter”. We demonstrate that Impeded Dark Matter can be easily realized without requiring tuning of model parameters. For negative mass splitting, we demonstrate that the annihilation cross-section for Impeded Dark Matter depends linearly on the dark matter velocity or may even be kinematically forbidden, making this scenario almost insensitive to constraints from the cosmic microwave background and from observations of dwarf galaxies. Accordingly, it may be possible for Impeded Dark Matter to yield observable signals in clusters or the Galactic center, with no corresponding signal in dwarfs. For positive mass splitting, we show that the annihilation cross-section is suppressed by the small mass splitting, which helps light dark matter to survive increasingly stringent constraints from indirect searches. As specific realizations for Impeded Dark Matter, we introduce a model of vector dark matter from a hidden SU(2) sector, and a composite dark matter scenario based on a QCD-like dark sector.
International Nuclear Information System (INIS)
Borghi, Carlo A; Cristofolini, Andrea; Grandi, Gabriele; Neretti, Gabriele; Seri, Paolo
2015-01-01
In this work a high voltage—high frequency generator for the power supply of a dielectric barrier discharge (DBD) plasma actuator for the aerodynamic control obtained by the electro-hydro-dynamic (EHD) interaction is described and tested. The generator can produce different voltage waveforms. The operating frequency is independent of the load characteristics and does not require impedance matching. The peak-to-peak voltage is 30 kV at a frequency up to 20 kHz and time variation rates up to 60 kV μs −1 . The performance of the actuator when supplied by several voltage waveforms is investigated. The tests have been performed in still air at atmospheric pressure. Voltage and current time behaviors have been measured. The evaluation of the energy delivered to the actuator allowed the estimation of the periods in which the plasma was ignited. Vibrational and rotational temperatures of the plasma have been estimated through spectroscopic acquisitions. The flow field induced in the region above the surface of the DBD actuator has been studied and the EHD conversion efficiency has been evaluated for the voltage waveforms investigated. The nearly sinusoidal multilevel voltage of the proposed generator and the sinusoidal voltage waveform of a conventional ac generator obtain comparable plasma features, EHD effects, and efficiencies. Inverse saw tooth waveform presents the highest effects and efficiency. The rectangular waveform generates suitable EHD effects but with the lowest efficiency. The voltage waveforms that induce plasmas with higher rotational temperatures are less efficient for the conversion of the electric into kinetic energy. (paper)
Coupled Transmission Lines as Impedance Transformer
DEFF Research Database (Denmark)
Jensen, Thomas; Zhurbenko, Vitaliy; Krozer, Viktor
2007-01-01
A theoretical investigation of the use of a coupled line section as an impedance transformer is presented. We show how to properly select the terminations of the coupled line structures for effective matching of real and complex loads in both narrow and wide frequency ranges. The corresponding...... circuit configurations and the design procedures are proposed. Synthesis relations are derived and provided for efficient matching circuit construction. Design examples are given to demonstrate the flexibility and limitations of the design methods and to show their validity for practical applications...
High-Voltage, High-Impedance Ion Beam Production
2009-06-01
the anode tube with a loosely-crumpled, thin aluminized- mylar foil. This spoils the virtual cathode and greatly reduces the neutron signal, as seen...ions follow ballistic (straight-line) trajectories in the drift tube (see Sec. VIII), then (except for the small displacement associated with bending...mTorr) ambient in the drift tube . Based on our previous experience, we would expect charge, but not necessarily current, neutralization of the beam
Impedance of accelerator components
International Nuclear Information System (INIS)
Corlett, J.N.
1996-05-01
As demands for high luminosity and low emittance particle beams increase, an understanding of the electromagnetic interaction of these beams with their vacuum chamber environment becomes more important in order to maintain the quality of the beam. This interaction is described in terms of the wake field in time domain, and the beam impedance in frequency domain. These concepts are introduced, and related quantities such as the loss factor are presented. The broadband Q = 1 resonator impedance model is discussed. Perturbation and coaxial wire methods of measurement of real components are reviewed
An Islanding Microgrid Power Sharing Approach Using Enhanced Virtual Impedance Control Scheme
DEFF Research Database (Denmark)
He, Jinwei; Li, Yun Wei; Guerrero, Josep M.
2013-01-01
In order to address the load sharing problem in islanding microgrids, this paper proposes an enhanced distributed generation (DG) unit virtual impedance control approach. The proposed method can realize accurate regulation of DG unit equivalent impedance at both fundamental and selected harmonic...... and PCC harmonic voltage compensation are achieved without using any fundamental and harmonic components extractions. Experimental results from a scaled single-phase microgrid prototype are provided to validate the feasibility of the proposed virtual impedance control approach....... frequencies. In contrast to conventional virtual impedance control methods, where only a line current feed-forward term is added to the DG voltage reference, the proposed virtual impedance at fundamental and harmonic frequencies is regulated using DG line current and point of common coupling (PCC) voltage...
Impedance Characteristics Modeling of a Two-Terminal Active Capacitor
DEFF Research Database (Denmark)
Wang, Haoran; Wang, Huai; Lu, Minghui
2017-01-01
to overcome the above issues. In this paper, the modeling of the active capacitor is investigated and a voltage feed-forward compensation scheme is proposed for overshoot reduction. Therefore, the impedance, equivalent capacitance, ESR, and ESL, of the active capacitor can be specified. A case study based...
Detailed characterization of anode-supported SOFCs by impedance spectroscopy
DEFF Research Database (Denmark)
Barfod, R.; Mogensen, Mogens Bjerg; Klemensø, Trine
2007-01-01
Anode-supported thin electrolyte cells are studied by electrochemical impedance spectroscopy (EIS). The aim is to describe how the losses of this type of cells are distributed at low current density (around open-circuit voltage) as a function of temperature. An equivalent circuit consisting...
Measurement of Phase Dependent Impedance for 3-phase Diode Rectifier
DEFF Research Database (Denmark)
Kwon, Jun Bum; Wang, Xiongfei; Bak, Claus Leth
2016-01-01
This paper presents a new method to measure the phase dependent impedance from an experimental set up. Though most of power electronics based system is gradually migrating to IGBT based voltage source converter due to their controllability, the rectifier composed of diode or thyristor components...
Sheath impedance effects in very high frequency plasma experiments
International Nuclear Information System (INIS)
Schwarzenbach, W.; Howling, A.A.; Fivaz, M.; Brunner, S.; Hollenstein, C.
1995-05-01
The frequency dependence (13.56 MHz to 70 MHz) of the ion energy distribution at the ground electrode was measured by mass spectrometry in a symmetrical capacitive argon discharge. Reduced sheath impedance at Very High Frequency allows high levels of plasma power and substrate ion flux whilst maintaining low levels of ion energy and electrode voltage. The lower limit of ion bombardment energy is fixed by the sheath floating potential at high frequency, in contrast to low frequencies where only the rf voltage amplitude is determinant. The capacitive sheaths are thinner at high frequencies which accentuates the high frequency reduction in sheath impedance. It is argued that the frequency dependence of sheath impedance is responsible for the principal characteristics of Very High Frequency plasmas. The measurements are summarised by simple physical descriptions and compared with a Particle-In-Cell simulation. (author) figs., tabs., refs
... to know FAQ Living donation What is living donation? Organs Types Being a living donor First steps Being ... brochures What Every Patient Needs to Know Living Donation Multiple Listing Visit UNOS Store Learn more How organs are matched How to become a living donor ...
Experimental evaluation of voltage unbalance compensation in an islanded microgrid
DEFF Research Database (Denmark)
Savaghebi, Mehdi; Guerrero, Josep M.; Jalilian, Alireza
2011-01-01
In this paper, a method for voltage unbalance compensation in an islanded microgrid based on the proper control of distributed generators (DGs) interface converter is proposed. In this method, active and reactive power control loops are considered to control the power sharing among the DGs. Also......, a virtual impedance loop and voltage and current proportional-resonant controllers are included. Experimental results show the effectiveness of the proposed method for compensating voltage unbalance to an acceptable level....
Tapping mode microwave impedance microscopy
Lai, K.; Kundhikanjana, W.; Peng, H.; Cui, Y.; Kelly, M. A.; Shen, Z. X.
2009-01-01
We report tapping mode microwave impedance imaging based on atomic force microscope platforms. The shielded cantilever probe is critical to localize the tip-sample interaction near the tip apex. The modulated tip-sample impedance can be accurately
International Nuclear Information System (INIS)
Colton, E.P.; Wang, T.F.
1985-01-01
The longitudinal and transverse complex impedances Z/sub l//n and Z/sub t/, respectively, have been calculated for both the SSC injection and abort kickers. The calculations assumed that no attempt was made to shield the beam from the kickers. We took the injection and abort kickers to be as specified. The injection kickers were ferrite with a single-turn design, and the abort kickers were of a ''window-frame design'' with tape wound cores
Impedance analysis of subwoofer systems
Berkhoff, Arthur P.
The electrical impedance of four low-frequency loudspeaker systems is analyzed. The expression for this impedance is obtained directly from the acoustical analogous circuit. Formulas are derived for calculating the small-signal parameters from the frequencies of impedance minima and maxima of two
Acoustic impedances of ear canals measured by impedance tube
DEFF Research Database (Denmark)
Ciric, Dejan; Hammershøi, Dorte
2007-01-01
During hearing sensitivity tests, the sound field is commonly generated by an earphone placed on a subject ear. One of the factors that can affect the sound transmission in the ear is the acoustic impedance of the ear canal. Its importance is related to the contribution of other elements involved...... in the transmission such as the earphone impedance. In order to determine the acoustic impedances of human ear canals, the standardized method for measurement of complex impedances used for the measurement of the audiometric earphone impedances is applied. It is based on the transfer function between two microphone...... locations in an impedance tube. The end of the tube representing the measurement plane is placed at the ear canal entrance. Thus, the impedance seen from the entrance inward is measured on 25 subjects. Most subjects participated in the previous measurement of the ratio between the pressures at the open...
Something different - caching applied to calculation of impedance matrix elements
CSIR Research Space (South Africa)
Lysko, AA
2012-09-01
Full Text Available of the multipliers, the approximating functions are used any required parameters, such as input impedance or gain pattern etc. The method is relatively straightforward but, especially for small to medium matrices, requires spending time on filling... of the computing the impedance matrix for the method of moments, or a similar method, such as boundary element method (BEM) [22], with the help of the flowchart shown in Figure 1. Input Parameters (a) Search the cached data for a match (b) A match found...
Beam measurements of the SPS longitudinal impedance
Lasheen, A
2017-01-01
Longitudinal instabilities are one of the main limitationsin the CERN SPS to reach the beam parameters requiredfor the High Luminosity LHC project. In preparation tothe SPS upgrade, possible remedies are studied by perform-ing macroparticle simulations using the machine impedancemodel obtained from electromagnetic simulations and mea-surements. To benchmark the impedance model, the resultsof simulations are compared with various beam measure-ments. In this study, the reactive part of the impedance wasprobed by measuring the quadrupole frequency shift withintensity, obtained from bunch length oscillations at mis-matched injection into the SPS. This method was appliedover many last years to follow up the evolution of the SPSimpedance, injecting bunches with the same bunch length.A novel approach, giving significantly more information,consists in varying the injected bunch length. The compari-son of these measurements with macroparticle simulationsallowed to test the existing model and identify some missingSPS i...
Observations involving broadband impedance modelling
Energy Technology Data Exchange (ETDEWEB)
Berg, J S [Stanford Linear Accelerator Center, Menlo Park, CA (United States)
1996-08-01
Results for single- and multi-bunch instabilities can be significantly affected by the precise model that is used for the broadband impedance. This paper discusses three aspects of broadband impedance modelling. The first is an observation of the effect that a seemingly minor change in an impedance model has on the single-bunch mode coupling threshold. The second is a successful attempt to construct a model for the high-frequency tails of an r.f. cavity. The last is a discussion of requirements for the mathematical form of an impedance which follow from the general properties of impedances. (author)
Application of impedance spectroscopy to SOFC research
Energy Technology Data Exchange (ETDEWEB)
Hsieh, G.; Mason, T.O. [Northwestern Univ., Evanston, IL (United States); Pederson, L.R. [Pacific Northwest National Lab., Richland, WA (United States)
1996-12-31
With the resurgence of interest in solid oxide fuel cells and other solid state electrochemical devices, techniques originally developed for characterizing aqueous systems are being adapted and applied to solid state systems. One of these techniques, three-electrode impedance spectroscopy, is particularly powerful as it allows characterization of subcomponent and interfacial properties. Obtaining accurate impedance spectra, however, is difficult as reference electrode impedance is usually non-negligible and solid electrolytes typically have much lower conductance than aqueous solutions. Faidi et al and Chechirlian et al have both identified problems associated with low conductivity media. Other sources of error are still being uncovered. Ford et al identified resistive contacts with large time constants as a possibility, while Me et al showed that the small contact capacitance of the reference electrode was at fault. Still others show that instrument limitations play a role. Using the voltage divider concept, a simplified model that demonstrates the interplay of these various factors, predicts the form of possible distortions, and offers means to minimize errors is presented.
International Nuclear Information System (INIS)
Hu Jing; Rovey, Joshua L.
2011-01-01
A movable Faraday cup design with simple structure and adjustable impedance is described in this work. This Faraday cup has external adjustable shunt resistance for self-biased measurement setup and 50 Ω characteristic impedance to match with 50 Ω standard BNC coaxial cable and vacuum feedthroughs for nanosecond-level pulse signal measurements. Adjustable shunt resistance allows self-biased measurements to be quickly acquired to determine the electron energy distribution function. The performance of the Faraday cup is validated by tests of response time and amplitude of output signal. When compared with a reference source, the percent difference of the Faraday cup signal fall time is less than 10% for fall times greater than 10 ns. The percent difference of the Faraday cup signal pulse width is below 6.7% for pulse widths greater than 10 ns. A pseudospark-generated electron beam is used to compare the amplitude of the Faraday cup signal with a calibrated F-70 commercial current transformer. The error of the Faraday cup output amplitude is below 10% for the 4-14 kV tested pseudospark voltages. The main benefit of this Faraday cup is demonstrated by adjusting the external shunt resistance and performing the self-biased method for obtaining the electron energy distribution function. Results from a 4 kV pseudospark discharge indicate a ''double-humped'' energy distribution.
Hu, Jing; Rovey, Joshua L
2011-07-01
A movable Faraday cup design with simple structure and adjustable impedance is described in this work. This Faraday cup has external adjustable shunt resistance for self-biased measurement setup and 50 Ω characteristic impedance to match with 50 Ω standard BNC coaxial cable and vacuum feedthroughs for nanosecond-level pulse signal measurements. Adjustable shunt resistance allows self-biased measurements to be quickly acquired to determine the electron energy distribution function. The performance of the Faraday cup is validated by tests of response time and amplitude of output signal. When compared with a reference source, the percent difference of the Faraday cup signal fall time is less than 10% for fall times greater than 10 ns. The percent difference of the Faraday cup signal pulse width is below 6.7% for pulse widths greater than 10 ns. A pseudospark-generated electron beam is used to compare the amplitude of the Faraday cup signal with a calibrated F-70 commercial current transformer. The error of the Faraday cup output amplitude is below 10% for the 4-14 kV tested pseudospark voltages. The main benefit of this Faraday cup is demonstrated by adjusting the external shunt resistance and performing the self-biased method for obtaining the electron energy distribution function. Results from a 4 kV pseudospark discharge indicate a "double-humped" energy distribution.
RF impedance measurement calibration
International Nuclear Information System (INIS)
Matthews, P.J.; Song, J.J.
1993-01-01
The intent of this note is not to explain all of the available calibration methods in detail. Instead, we will focus on the calibration methods of interest for RF impedance coupling measurements and attempt to explain: (1). The standards and measurements necessary for the various calibration techniques. (2). The advantages and disadvantages of each technique. (3). The mathematical manipulations that need to be applied to the measured standards and devices. (4). An outline of the steps needed for writing a calibration routine that operated from a remote computer. For further details of the various techniques presented in this note, the reader should consult the references
Design of optimized impedance transformer for ICRF antenna in LHD
Energy Technology Data Exchange (ETDEWEB)
Saito, K., E-mail: saito@nifs.ac.jp [National Institute for Fusion Science, Toki, Gifu, 509-5292 (Japan); Seki, T.; Kasahara, H.; Seki, R.; Kumazawa, R.; Nomura, G.; Shimpo, F.; Mutoh, T. [National Institute for Fusion Science, Toki, Gifu, 509-5292 (Japan)
2013-10-15
Highlights: ► We developed optimization method of impedance transformer for ICRF antenna. ► Power loss will be one-third with the optimized impedance transformer. ► Possibility of damage on the transmission line will be drastically reduced. ► High performance will be kept in the wide antenna impedance region. -- Abstract: A pair of ion cyclotron range of frequencies (ICRF) antennas in the large helical device (LHD), HAS antennas showed high efficiency in minority ion heating. However the low loading resistance must be increased to prevent breakdown in transmission line. Moreover, the voltage and the current around the feed-through must be reduced to protect it. For these purpose, we developed a design procedure of the impedance transformer for HAS antennas. To optimize the transformer, the inner conductors were divided into several segments and the radii of them were given discretely and independently. The maximum effective loading resistance was obtained by using all combinations of radii within the limitations of the voltage and current at the feed-through and the electric field on the transformer. To get a precise solution, this procedure was repeated several times by narrowing the range of radii of inner conductors. Then the optimized impedance transformer was designed by smoothing the radii of inner conductors. For the typical discharge, the voltage and current at the feed-through were reduced to the half of the original values with the same power. The effective loading resistance was increased to more than four times. High performance is expected in wide impedance region.
Distributed stability control using intelligent voltage-margin relay
Energy Technology Data Exchange (ETDEWEB)
Wiszniewski, A.; Rebizant, W. [Wroclaw Univ. of Technology (Poland); Klimek, A. [Powertech Labs Inc., Surrey, BC (Canada)
2010-07-01
This paper presented an intelligent relay that operates if the load to source impedance ratio decreases to a level that is dangerously close to the stability limit, which leads to power system blackouts. The intelligent voltage-margin/difference relay installed at receiving substations automatically initiates action if the voltage stability margin drops to a dangerously low level. The relay decides if the tap changing devices are to be blocked and if under-voltage load shedding should be initiated, thereby mitigating an evolving instability. The intelligent relay has two levels of operation. At the first stage, which corresponds to the higher load to source impedance ratio, the relay initiates blocking of the tap changer. At the second stage, corresponding to the lower source to load impedance ratio, load shedding is initiated. The relay operates when the load to source impedance ratio reaches a certain predetermined level, but it does not depend either on the level of the source voltage or on the difference of source and load impedance phase angles. The algorithm for the relay is relatively simple and uses only locally available signals. Consequently, the transformer is well controlled to eliminate the cases of voltage instability. 6 refs., 7 figs.
Droop Control with an Adjustable Complex Virtual Impedance Loop based on Cloud Model Theory
DEFF Research Database (Denmark)
Li, Yan; Shuai, Zhikang; Xu, Qinming
2016-01-01
Droop control framework with an adjustable virtual impedance loop is proposed in this paper, which is based on the cloud model theory. The proposed virtual impedance loop includes two terms: a negative virtual resistor and an adjustable virtual inductance. The negative virtual resistor term...... sometimes. The cloud model theory is applied to get online the changing line impedance value, which relies on the relevance of the reactive power responding the changing line impedance. The verification of the proposed control strategy is done according to the simulation in a low voltage microgrid in Matlab....
2005-01-01
A voltage regulating circuit comprising a rectifier (2) for receiving an AC voltage (Vmains) and for generating a rectified AC voltage (vrec), and a capacitor (3) connected in parallel with said rectified AC voltage for providing a DC voltage (VDC) over a load (5), characterized by a unidirectional
Electrical impedance tomography
Lobo, Beatriz; Hermosa, Cecilia; Abella, Ana
2018-01-01
Continuous assessment of respiratory status is one of the cornerstones of modern intensive care unit (ICU) monitoring systems. Electrical impedance tomography (EIT), although with some constraints, may play the lead as a new diagnostic and guiding tool for an adequate optimization of mechanical ventilation in critically ill patients. EIT may assist in defining mechanical ventilation settings, assess distribution of tidal volume and of end-expiratory lung volume (EELV) and contribute to titrate positive end-expiratory pressure (PEEP)/tidal volume combinations. It may also quantify gains (recruitment) and losses (overdistention or derecruitment), granting a more realistic evaluation of different ventilator modes or recruitment maneuvers, and helping in the identification of responders and non-responders to such maneuvers. Moreover, EIT also contributes to the management of life-threatening lung diseases such as pneumothorax, and aids in guiding fluid management in the critical care setting. Lastly, assessment of cardiac function and lung perfusion through electrical impedance is on the way. PMID:29430443
Low profile conformal antenna arrays on high impedance substrate
Singh, Hema; Jha, Rakesh Mohan
2016-01-01
This book presents electromagnetic (EM) design and analysis of dipole antenna array over high impedance substrate (HIS). HIS is a preferred substrate for low-profile antenna design, owing to its unique boundary conditions. Such substrates permit radiating elements to be printed on them without any disturbance in the radiation characteristics. Moreover HIS provides improved impedance matching, enhanced bandwidth, and increased broadside directivity owing to total reflection from the reactive surface and high input impedance. This book considers different configurations of HIS for array design on planar and non-planar high-impedance surfaces. Results are presented for cylindrical dipole, printed dipole, and folded dipole over single- and double-layered square-patch-based HIS and dogbone-based HIS. The performance of antenna arrays is analyzed in terms of performance parameters such as return loss and radiation pattern. The design presented shows acceptable return loss and mainlobe gain of radiation pattern. Thi...
Microwave Impedance Measurement for Nanoelectronics
Directory of Open Access Journals (Sweden)
M. Randus
2011-04-01
Full Text Available The rapid progress in nanoelectronics showed an urgent need for microwave measurement of impedances extremely different from the 50Ω reference impedance of measurement instruments. In commonly used methods input impedance or admittance of a device under test (DUT is derived from measured value of its reflection coefficient causing serious accuracy problems for very high and very low impedances due to insufficient sensitivity of the reflection coefficient to impedance of the DUT. This paper brings theoretical description and experimental verification of a method developed especially for measurement of extreme impedances. The method can significantly improve measurement sensitivity and reduce errors caused by the VNA. It is based on subtraction (or addition of a reference reflection coefficient and the reflection coefficient of the DUT by a passive network, amplifying the resulting signal by an amplifier and measuring the amplified signal as a transmission coefficient by a common vector network analyzer (VNA. A suitable calibration technique is also presented.
Directory of Open Access Journals (Sweden)
Xin-Bing Cheng
2010-07-01
Full Text Available The Blumlein pulse forming line (BPFL consisting of an inner coaxial pulse forming line (PFL and an outer coaxial PFL is widely used in the field of pulsed power, especially for intense electron-beam accelerators (IEBA. The output voltage waveform determines the quality and characteristics of the output beam current of the IEBA. Comparing with the conventional BPFL, an IEBA based on a helical type BPFL can increase the duration of the output voltage in the same geometrical volume. However, for the helical type BPFL, the voltage waveform on a matched load may be distorted which influences the electron-beam quality. In this paper, an IEBA based on helical type BPFL is studied theoretically. Based on telegrapher equations of the BPFL, a formula for the output voltage of IEBA is obtained when the transition section is taken into account, where the transition section is between the middle cylinder of BPFL and the load. From the theoretical analysis, it is found that the wave impedance and transit time of the transition section influence considerably the main pulse voltage waveform at the load, a step is formed in front of the main pulse, and a sharp spike is also formed at the end of the main pulse. In order to get a well-shaped square waveform at the load and to improve the electron-beam quality of such an accelerator, the wave impedance of the transition section should be equal to that of the inner PFL of helical type BPFL and the transit time of the transition section should be designed as short as possible. Experiments performed on an IEBA with the helical type BPFL show reasonable agreement with theoretical analysis.
NOTE: Impedance magnetocardiogram
Kandori, Akihiko; Miyashita, Tsuyoshi; Suzuki, Daisuke; Yokosawa, Koichi; Tsukada, Keiji
2001-02-01
We have developed an impedance magnetocardiogram (IMCG) system to detect the change of magnetic field corresponding to changes in blood volume in the heart. A low magnetic field from the electrical activity of the human heart - the so-called magnetocardiogram (MCG) - can be simultaneously detected by using this system. Because the mechanical and electrical functions in the heart can be monitored by non-invasive and non-contact measurements, it is easy to observe the cardiovascular functions from an accurate sensor position. This system uses a technique to demodulate induced current in a subject. A flux-locked circuit of a superconducting quantum interference device has a wide frequency range (above 1 MHz) because a constant current (40 kHz) is fed through the subject. It is shown for the first time that the system could measure IMCG signals at the same time as MCG signals.
Monotonicity-based electrical impedance tomography for lung imaging
Zhou, Liangdong; Harrach, Bastian; Seo, Jin Keun
2018-04-01
This paper presents a monotonicity-based spatiotemporal conductivity imaging method for continuous regional lung monitoring using electrical impedance tomography (EIT). The EIT data (i.e. the boundary current-voltage data) can be decomposed into pulmonary, cardiac and other parts using their different periodic natures. The time-differential current-voltage operator corresponding to the lung ventilation can be viewed as either semi-positive or semi-negative definite owing to monotonic conductivity changes within the lung regions. We used these monotonicity constraints to improve the quality of lung EIT imaging. We tested the proposed methods in numerical simulations, phantom experiments and human experiments.
Frequency-Tunable antenna by input-impedance-tunable CMOS RF-Frontend
Haider, Nadia; Oude Alink, M.S.; Caratelli, Diego; Klumperink, Eric A.M.; Yarovoy, Alexander G.
2013-01-01
Variable-impedance matching between the antenna and the RF-frontend provides several potential advantages, including changing operational frequency, compensating for unintentional mismatch, improving scanning capability, and reducing noise and interference signal levels. In this article a concept of
Plasma focus matching conditions
International Nuclear Information System (INIS)
Soliman, H.M.; Masoud, M.M.; Elkhalafawy, T.A.
1988-01-01
A snow-plough and slug models have been used to obtain the optimum matching conditions of the plasma in the focus. The dimensions of the plasma focus device are, inner electrode radius = 2 cm, outer electrode radius = 5.5 cm, and its length = 8 cm. It was found that the maximum magnetic energy of 12.26 kJ has to be delivered to plasma focus whose density is 10 19 /cm 3 at focusing time of 2.55 μs and with total external inductance of 24.2 n H. The same method is used to evaluate the optimum matching conditions for the previous coaxial discharge system which had inner electrode radius = 1.6 cm, outer electrode radius = 3.3 cm and its length = 31.5 cm. These conditions are charging voltage = 12 kV, capacity of the condenser bank = 430 μf, plasma focus density = 10 19 /cm 3 focusing time = 8 μs and total external inductance = 60.32 n H.3 fig., 2 tab
Electrical impedance tomography of the 1995 OGI gasoline release
International Nuclear Information System (INIS)
Daily, W.; Ramirez, A.
1996-01-01
Electrical impedance tomography (EIT) was used to image the plume resulting from a release of 378 liters (100 gallons) of gasoline into a sandy acquifer. Images were made in 5 planes before and 5 times during the release, to generate a detailed picture of the spatial as well as the temporal development of the plume as it spread at the water table. Information of the electrical impedance (both in phase and out of phase voltages) was used or several different frequencies to produce images. We observed little dispersion in the images either before or after the gasoline entered the acquifer. Likewise, despite some laboratory measurements of impedances, there was no evidence of a change in the reactance in the soil because of the gasoline
Impedance models in time domain
Rienstra, S.W.
2005-01-01
Necessary conditions for an impedance function are derived. Methods available in the literature are discussed. A format with recipe is proposed for an exact impedance condition in time domain on a time grid, based on the Helmholtz resonator model. An explicit solution is given of a pulse reflecting
Impedance Source Power Electronic Converters
DEFF Research Database (Denmark)
Liu, Yushan; Abu-Rub, Haitham; Ge, Baoming
Impedance Source Power Electronic Converters brings together state of the art knowledge and cutting edge techniques in various stages of research related to the ever more popular impedance source converters/inverters. Significant research efforts are underway to develop commercially viable...... and technically feasible, efficient and reliable power converters for renewable energy, electric transportation and for various industrial applications. This book provides a detailed understanding of the concepts, designs, controls, and application demonstrations of the impedance source converters/inverters. Key...... features: Comprehensive analysis of the impedance source converter/inverter topologies, including typical topologies and derived topologies. Fully explains the design and control techniques of impedance source converters/inverters, including hardware design and control parameter design for corresponding...
Impedance source power electronic converters
Liu, Yushan; Ge, Baoming; Blaabjerg, Frede; Ellabban, Omar; Loh, Poh Chiang
2016-01-01
Impedance Source Power Electronic Converters brings together state of the art knowledge and cutting edge techniques in various stages of research related to the ever more popular impedance source converters/inverters. Significant research efforts are underway to develop commercially viable and technically feasible, efficient and reliable power converters for renewable energy, electric transportation and for various industrial applications. This book provides a detailed understanding of the concepts, designs, controls, and application demonstrations of the impedance source converters/inverters. Key features: Comprehensive analysis of the impedance source converter/inverter topologies, including typical topologies and derived topologies. Fully explains the design and control techniques of impedance source converters/inverters, including hardware design and control parameter design for corresponding control methods. Presents the latest power conversion solutions that aim to advance the role of pow...
International Nuclear Information System (INIS)
Liu Guozhi
2003-01-01
The energy spectrum of the electron beam generated by low-impedance diode and the influence of external magnetic field on the impedance of diode are studied numerically in this paper. The results show that the beam generated by the diode has an energy spread, even with constant applied voltage. Additionally, external magnetic field has great but reverse influence on the impedance of low-impedance diode, which is, according to the author's analysis, the result of the change of the electron's track due to external magnetic field. If the beam current is less than the critical one for self-pinch, the impedance will be constant with the variation of external magnetic field
Impedance Changes Indicate Proximal Ventriculoperitoneal Shunt Obstruction In Vitro.
Basati, Sukhraaj; Tangen, Kevin; Hsu, Ying; Lin, Hanna; Frim, David; Linninger, Andreas
2015-12-01
Extracranial cerebrospinal fluid (CSF) shunt obstruction is one of the most important problems in hydrocephalus patient management. Despite ongoing research into better shunt design, robust and reliable detection of shunt malfunction remains elusive. The authors present a novel method of correlating degree of tissue ingrowth into ventricular CSF drainage catheters with internal electrical impedance. The impedance based sensor is able to continuously monitor shunt patency using intraluminal electrodes. Prototype obstruction sensors were fabricated for in-vitro analysis of cellular ingrowth into a shunt under static and dynamic flow conditions. Primary astrocyte cell lines and C6 glioma cells were allowed to proliferate up to 7 days within a shunt catheter and the impedance waveform was observed. During cell ingrowth a significant change in the peak-to-peak voltage signal as well as the root-mean-square voltage level was observed, allowing the impedance sensor to potentially anticipate shunt malfunction long before it affects fluid drainage. Finite element modeling was employed to demonstrate that the electrical signal used to monitor tissue ingrowth is contained inside the catheter lumen and does not endanger tissue surrounding the shunt. These results may herald the development of "next generation" shunt technology that allows prediction of malfunction before it affects patient outcome.
Magnetically coupled impedance-source inverters
DEFF Research Database (Denmark)
Loh, Poh Chiang; Blaabjerg, Frede
2012-01-01
input-to-output gain, and the presence of an impedance network. The former means a high dc-link voltage, which can stress the semiconductor switches unnecessarily. The latter leads to increases in cost and size, which similarly are undesirable. To lessen these concerns, an interesting approach is to use...... magnetically coupled transformers or inductors to raise the gain and modulation ratio simultaneously, while reducing the number of passive components needed. A study of the approach is now presented to show how various existing magnetically coupled inverters can be derived by applying a generic methodology....... The same methodology is then applied to develop more magnetically coupled Z-source inverters with advantages that have not been identified in the literature. These findings have already been proven in experiments....
Virtual Impedance Based Stability Improvement for DC Microgrids with Constant Power Loads
DEFF Research Database (Denmark)
Lu, Xiaonan; Sun, Kai; Huang, Lipei
2014-01-01
DC microgrid provides an efficient way to integrate different kinds of renewable energy sources with DC couplings. In this paper, in order to improve the stability of DC microgrids with constant power loads (CPLs), a virtual impedance based method is proposed. The CPLs have inherent instability....... To validate the stability with the above stabilizers in a DC microgrid with parallel interfacing converters and CPL, the impedance matching approach is employed. The output impedance of the source converter and input impedance of the load are calculated respectively, and the influence of droop control...
Moderately nonlinear diffuse-charge dynamics under an ac voltage.
Stout, Robert F; Khair, Aditya S
2015-09-01
The response of a symmetric binary electrolyte between two parallel, blocking electrodes to a moderate amplitude ac voltage is quantified. The diffuse charge dynamics are modeled via the Poisson-Nernst-Planck equations for a dilute solution of point-like ions. The solution to these equations is expressed as a Fourier series with a voltage perturbation expansion for arbitrary Debye layer thickness and ac frequency. Here, the perturbation expansion in voltage proceeds in powers of V_{o}/(k_{B}T/e), where V_{o} is the amplitude of the driving voltage and k_{B}T/e is the thermal voltage with k_{B} as Boltzmann's constant, T as the temperature, and e as the fundamental charge. We show that the response of the electrolyte remains essentially linear in voltage amplitude at frequencies greater than the RC frequency of Debye layer charging, D/λ_{D}L, where D is the ion diffusivity, λ_{D} is the Debye layer thickness, and L is half the cell width. In contrast, nonlinear response is predicted at frequencies below the RC frequency. We find that the ion densities exhibit symmetric deviations from the (uniform) equilibrium density at even orders of the voltage amplitude. This leads to the voltage dependence of the current in the external circuit arising from the odd orders of voltage. For instance, the first nonlinear contribution to the current is O(V_{o}^{3}) which contains the expected third harmonic but also a component oscillating at the applied frequency. We use this to compute a generalized impedance for moderate voltages, the first nonlinear contribution to which is quadratic in V_{o}. This contribution predicts a decrease in the imaginary part of the impedance at low frequency, which is due to the increase in Debye layer capacitance with increasing V_{o}. In contrast, the real part of the impedance increases at low frequency, due to adsorption of neutral salt from the bulk to the Debye layer.
Moderately nonlinear diffuse-charge dynamics under an ac voltage
Stout, Robert F.; Khair, Aditya S.
2015-09-01
The response of a symmetric binary electrolyte between two parallel, blocking electrodes to a moderate amplitude ac voltage is quantified. The diffuse charge dynamics are modeled via the Poisson-Nernst-Planck equations for a dilute solution of point-like ions. The solution to these equations is expressed as a Fourier series with a voltage perturbation expansion for arbitrary Debye layer thickness and ac frequency. Here, the perturbation expansion in voltage proceeds in powers of Vo/(kBT /e ) , where Vo is the amplitude of the driving voltage and kBT /e is the thermal voltage with kB as Boltzmann's constant, T as the temperature, and e as the fundamental charge. We show that the response of the electrolyte remains essentially linear in voltage amplitude at frequencies greater than the RC frequency of Debye layer charging, D /λDL , where D is the ion diffusivity, λD is the Debye layer thickness, and L is half the cell width. In contrast, nonlinear response is predicted at frequencies below the RC frequency. We find that the ion densities exhibit symmetric deviations from the (uniform) equilibrium density at even orders of the voltage amplitude. This leads to the voltage dependence of the current in the external circuit arising from the odd orders of voltage. For instance, the first nonlinear contribution to the current is O (Vo3) which contains the expected third harmonic but also a component oscillating at the applied frequency. We use this to compute a generalized impedance for moderate voltages, the first nonlinear contribution to which is quadratic in Vo. This contribution predicts a decrease in the imaginary part of the impedance at low frequency, which is due to the increase in Debye layer capacitance with increasing Vo. In contrast, the real part of the impedance increases at low frequency, due to adsorption of neutral salt from the bulk to the Debye layer.
A model for voltage collapse study considering load characteristics
Energy Technology Data Exchange (ETDEWEB)
Aguiar, L B [Companhia de Energia Eletrica da Bahia (COELBA), Salvador, BA (Brazil)
1994-12-31
This paper presents a model for analysis of voltage collapse and instability problem considering the load characteristics. The model considers fundamentally the transmission lines represented by exact from through the generalized constants A, B, C, D and the loads as function of the voltage, emphasizing the cases of constant power, constant current and constant impedance. the study treats of the system behavior on steady state and presents illustrative graphics about the problem. (author) 12 refs., 4 figs.
Directory of Open Access Journals (Sweden)
Chaichana Amornchai
2017-01-01
Full Text Available In this paper, a voltage mode multifunction filter based on single voltage differencing differential difference amplifier (VDDDA is presented. The proposed filter with three input voltages and single output voltage is constructed with single VDDDA, two capacitors and two resistors. Its quality factor can be adjusted without affecting natural frequency. Also, the natural frequency can be electronically tuned via adjusting of bias current. The filter can offer five output responses, high-pas (HP, band-pass (BP, band-reject (BR, low-pass (LP and all-ass (AP functions in the same circuit topology. The output response can be selected by choosing the suitable input voltage without the component matching condition and the requirement of additional double gain voltage amplifier. PSpice simulation results to confirm an operation of the proposed filter correspond to the theory.
Microwave integrated circuit for Josephson voltage standards
Holdeman, L. B.; Toots, J.; Chang, C. C. (Inventor)
1980-01-01
A microwave integrated circuit comprised of one or more Josephson junctions and short sections of microstrip or stripline transmission line is fabricated from thin layers of superconducting metal on a dielectric substrate. The short sections of transmission are combined to form the elements of the circuit and particularly, two microwave resonators. The Josephson junctions are located between the resonators and the impedance of the Josephson junctions forms part of the circuitry that couples the two resonators. The microwave integrated circuit has an application in Josephson voltage standards. In this application, the device is asymmetrically driven at a selected frequency (approximately equal to the resonance frequency of the resonators), and a d.c. bias is applied to the junction. By observing the current voltage characteristic of the junction, a precise voltage, proportional to the frequency of the microwave drive signal, is obtained.
Coplanar strips for Josephson voltage standard circuits
International Nuclear Information System (INIS)
Schubert, M.; May, T.; Wende, G.; Fritzsch, L.; Meyer, H.-G.
2001-01-01
We present a microwave circuit for Josephson voltage standards. Here, the Josephson junctions are integrated in a microwave transmission line designed as coplanar strips (CPS). The new layout offers the possibility of achieving a higher scale of integration and to considerably simplify the fabrication technology. The characteristic impedance of the CPS is about 50 Ω, and this should be of interest for programmable Josephson voltage standard circuits with SNS or SINIS junctions. To demonstrate the function of the microwave circuit design, conventional 10 V Josephson voltage standard circuits with 17000 Nb/AlO x /Nb junctions were prepared and tested. Stable Shapiro steps at the 10 V level were generated. Furthermore, arrays of 1400 SINIS junctions in this microwave layout exhibited first-order Shapiro steps. Copyright 2001 American Institute of Physics
Nonlinear Impedance of Whole Cells Near an Electrode as a Probe of Mitochondrial Activity
Directory of Open Access Journals (Sweden)
John H. Miller Jr.
2011-04-01
Full Text Available By simultaneously measuring the bulk media and electrode interface voltages of a yeast (Saccharomyces cerevisiae suspension subjected to an AC voltage, a yeast-dependent nonlinear response was found only near the current injection electrodes. Computer simulation of yeast near a current injection electrode found an enhanced voltage drop across the yeast near the electrode due to slowed charging of the electrode interfacial capacitance. This voltage drop is sufficient to induce conformation change in membrane proteins. Disruption of the mitochondrial electron transport chain is found to significantly change the measured nonlinear current response, suggesting nonlinear impedance can be used as a non-invasive probe of cellular metabolic activity.
Small Signal Loudspeaker Impedance Emulator
DEFF Research Database (Denmark)
Iversen, Niels Elkjær; Knott, Arnold
2014-01-01
Specifying the performance of audio amplifiers is typically done by playing sine waves into a pure ohmic load. However real loudspeaker impedances are not purely ohmic but characterized by the mechanical resonance between the mass of the diaphragm and the compliance of its suspension which vary...... from driver to driver. Therefore, a loudspeaker emulator capable of adjusting its impedance to that of a given driver is desired for measurement purposes. This paper proposes a loudspeaker emulator circuit for small signals. Simulations and experimental results are compared and show that it is possible...... to emulate the loudspeaker impedance with an electric circuit and that its resonance frequency can be changed by tuning two resistors....
Modified coaxial wire method for measurement of transfer impedance of beam position monitors
Kumar, Mukesh; Babbar, L. K.; Deo, R. K.; Puntambekar, T. A.; Senecha, V. K.
2018-05-01
The transfer impedance is a very important parameter of a beam position monitor (BPM) which relates its output signal with the beam current. The coaxial wire method is a standard technique to measure transfer impedance of the BPM. The conventional coaxial wire method requires impedance matching between coaxial wire and external circuits (vector network analyzer and associated cables). This paper presents a modified coaxial wire method for bench measurement of the transfer impedance of capacitive pickups like button electrodes and shoe box BPMs. Unlike the conventional coaxial wire method, in the modified coaxial wire method no impedance matching elements have been used between the device under test and the external circuit. The effect of impedance mismatch has been solved mathematically and a new expression of transfer impedance has been derived. The proposed method is verified through simulation of a button electrode BPM using cst studio suite. The new method is also applied to measure transfer impedance of a button electrode BPM developed for insertion devices of Indus-2 and the results are also compared with its simulations. Close agreement between measured and simulation results suggests that the modified coaxial wire setup can be exploited for the measurement of transfer impedance of capacitive BPMs like button electrodes and shoe box BPM.
Modified coaxial wire method for measurement of transfer impedance of beam position monitors
Directory of Open Access Journals (Sweden)
Mukesh Kumar
2018-05-01
Full Text Available The transfer impedance is a very important parameter of a beam position monitor (BPM which relates its output signal with the beam current. The coaxial wire method is a standard technique to measure transfer impedance of the BPM. The conventional coaxial wire method requires impedance matching between coaxial wire and external circuits (vector network analyzer and associated cables. This paper presents a modified coaxial wire method for bench measurement of the transfer impedance of capacitive pickups like button electrodes and shoe box BPMs. Unlike the conventional coaxial wire method, in the modified coaxial wire method no impedance matching elements have been used between the device under test and the external circuit. The effect of impedance mismatch has been solved mathematically and a new expression of transfer impedance has been derived. The proposed method is verified through simulation of a button electrode BPM using cst studio suite. The new method is also applied to measure transfer impedance of a button electrode BPM developed for insertion devices of Indus-2 and the results are also compared with its simulations. Close agreement between measured and simulation results suggests that the modified coaxial wire setup can be exploited for the measurement of transfer impedance of capacitive BPMs like button electrodes and shoe box BPM.
International Nuclear Information System (INIS)
Martin, M.
1991-01-01
Industrial processes usually require electrical power. This power is used to drive motors, to heat materials, or in electrochemical processes. Often the power requirements of a plant require the electric power to be delivered at high voltage. In this paper high voltage is considered any voltage over 600 V. This voltage could be as high as 138,000 V for some very large facilities. The characteristics of this voltage and the enormous amounts of power being transmitted necessitate special safety considerations. Safety must be considered during the four activities associated with a high voltage electrical system. These activities are: Design; Installation; Operation; and Maintenance
Reid, Margaret A.
1989-01-01
Impedances of fifteen electrodes form each of the four U.S. manufactures were measured at 0.200 V vs. the Hg/HgO reference electrode. This corresponds to a voltage of 1.145 for a Ni/H2 cell. Measurements were also made of a representative sample of these at 0.44 V. At the higher voltage, the impedances were small and very similar, but at the lower voltage there were major differences between manufacturers. Electrodes from the same manufacturers showed only small differences. The impedances of electrodes from two manufacturers were considerably different in 26 percent KOH from those in 31 percent KOH. These preliminary results seen to correlate with the limited data from earlier life testing of cells from these manufacturers. The impedances of cells being tested for Space Station Freedom are being followed, and more impendance measurements of electrodes are being performed as functions of manufacturer, voltage, electrolyte concentration, and cycle history in hopes of finding better correlations of impedance with life.
Electrical impedance along connective tissue planes associated with acupuncture meridians
Directory of Open Access Journals (Sweden)
Hammerschlag Richard
2005-05-01
Full Text Available Abstract Background Acupuncture points and meridians are commonly believed to possess unique electrical properties. The experimental support for this claim is limited given the technical and methodological shortcomings of prior studies. Recent studies indicate a correspondence between acupuncture meridians and connective tissue planes. We hypothesized that segments of acupuncture meridians that are associated with loose connective tissue planes (between muscles or between muscle and bone visible by ultrasound have greater electrical conductance (less electrical impedance than non-meridian, parallel control segments. Methods We used a four-electrode method to measure the electrical impedance along segments of the Pericardium and Spleen meridians and corresponding parallel control segments in 23 human subjects. Meridian segments were determined by palpation and proportional measurements. Connective tissue planes underlying those segments were imaged with an ultrasound scanner. Along each meridian segment, four gold-plated needles were inserted along a straight line and used as electrodes. A parallel series of four control needles were placed 0.8 cm medial to the meridian needles. For each set of four needles, a 3.3 kHz alternating (AC constant amplitude current was introduced at three different amplitudes (20, 40, and 80 μAmps to the outer two needles, while the voltage was measured between the inner two needles. Tissue impedance between the two inner needles was calculated based on Ohm's law (ratio of voltage to current intensity. Results At the Pericardium location, mean tissue impedance was significantly lower at meridian segments (70.4 ± 5.7 Ω compared with control segments (75.0 ± 5.9 Ω (p = 0.0003. At the Spleen location, mean impedance for meridian (67.8 ± 6.8 Ω and control segments (68.5 ± 7.5 Ω were not significantly different (p = 0.70. Conclusion Tissue impedance was on average lower along the Pericardium meridian, but not
DEFF Research Database (Denmark)
Wang, Xiongfei; Blaabjerg, Frede; Chen, Zhe
2012-01-01
This paper proposes a harmonic impedance synthesis technique for voltage-controlled distributed generation inverter in order to damp harmonic voltage distortion on a distribution network. The approach employs a multiloop control scheme, where a selective load harmonic current feedforward loop bas...
Probabilistic seismic history matching using binary images
Davolio, Alessandra; Schiozer, Denis Jose
2018-02-01
Currently, the goal of history-matching procedures is not only to provide a model matching any observed data but also to generate multiple matched models to properly handle uncertainties. One such approach is a probabilistic history-matching methodology based on the discrete Latin Hypercube sampling algorithm, proposed in previous works, which was particularly efficient for matching well data (production rates and pressure). 4D seismic (4DS) data have been increasingly included into history-matching procedures. A key issue in seismic history matching (SHM) is to transfer data into a common domain: impedance, amplitude or pressure, and saturation. In any case, seismic inversions and/or modeling are required, which can be time consuming. An alternative to avoid these procedures is using binary images in SHM as they allow the shape, rather than the physical values, of observed anomalies to be matched. This work presents the incorporation of binary images in SHM within the aforementioned probabilistic history matching. The application was performed with real data from a segment of the Norne benchmark case that presents strong 4D anomalies, including softening signals due to pressure build up. The binary images are used to match the pressurized zones observed in time-lapse data. Three history matchings were conducted using: only well data, well and 4DS data, and only 4DS. The methodology is very flexible and successfully utilized the addition of binary images for seismic objective functions. Results proved the good convergence of the method in few iterations for all three cases. The matched models of the first two cases provided the best results, with similar well matching quality. The second case provided models presenting pore pressure changes according to the expected dynamic behavior (pressurized zones) observed on 4DS data. The use of binary images in SHM is relatively new with few examples in the literature. This work enriches this discussion by presenting a new
Three-Level Z-Source Inverters Using a Single LC Impedance Network
DEFF Research Database (Denmark)
Loh, Poh Chiang; Lim, Sok Wei; Gao, Feng
2007-01-01
two LC impedance networks and two isolated dc sources, which can significantly increase the overall system cost and require a more complex modulator for balancing the network inductive voltage boosting. Offering a number of less costly alternatives, this letter presents the design and control of two...... three-level Z-source inverters, whose output voltage can be stepped down or up using only a single LC impedance network connected between the dc input source and either a neutral-point-clamped (NPC) or dc-link cascaded inverter circuitry. Through careful design of their modulation scheme, both inverters...
Impedance characteristics of the Bz diode on the LION accelerator
International Nuclear Information System (INIS)
Meyerhofer, D.D.; Horioka, K.; Kusse, B.; Rondeau, G.; Struckman, C.
1987-01-01
The LION accelerator at Cornell University is being used to study the characteristics of the applied B/sub z/, or 'barrel' diode. This 0.8 TW, 4 ohm, ion accelerator has the ability to take several shots per day, and hence alloys systematic scans to be performed. An important result of a recent series of experiments is that the diode impedance remains relatively constant, decaying only slowly, during the 50 nsec pulse. When the diode is operated with a 4.5 mm gap and a 21 kG insulating magnetic field, the typical diode parameters, are a voltage of 1 MV and a total current of 250 kA, leading to a diode impedance of 4 ohms and power of 0.25 TW. The diode impedance decays with a 100 nsec time constant. The ion beams have peak currents of roughly 125 kA and typical impedances of Bohms, which decays with a time constant of 25 nsec. The Child-Langmuir gap was approximately 2 mm and closed with a velocity of roughly 2X10/sup 6/ cm/sec. Current experimental work is aimed at characterizing the impedance of the B/sub z/ diode as a function of the applied magnetic field, the A-K gap, the anode curvature, and the anode groove parameters. In addition, the effect of changing the voltage rise with a plasma opening switch and of adding an electron limiter is examined. The ion beam quality is examined at the focus of the barrel diode with a swept Thomson parabola and various Rutherford scattering diagnostics
Voltage regulator for generator
Energy Technology Data Exchange (ETDEWEB)
Naoi, K
1989-01-17
It is an object of this invention to provide a voltage regulator for a generator charging a battery, wherein even if the ambient temperature at the voltage regulator rises abnormally high, possible thermal breakage of the semiconductor elements constituting the voltage regulator can be avoided. A feature of this invention is that the semiconductor elements can be protected from thermal breakage, even at an abnormal ambient temperature rise at the voltage regulator for the battery charging generator, by controlling a maximum conduction ratio of a power transistor in the voltage regulator in accordance with the temperature at the voltage regulator. This is achieved through a switching device connected in series to the field coil of the generator and adapted to be controlled in accordance with an output voltage of the generator and the ambient temperature at the voltage regulator. 6 figs.
Automatic voltage imbalance detector
Bobbett, Ronald E.; McCormick, J. Byron; Kerwin, William J.
1984-01-01
A device for indicating and preventing damage to voltage cells such as galvanic cells and fuel cells connected in series by detecting sequential voltages and comparing these voltages to adjacent voltage cells. The device is implemented by using operational amplifiers and switching circuitry is provided by transistors. The device can be utilized in battery powered electric vehicles to prevent galvanic cell damage and also in series connected fuel cells to prevent fuel cell damage.
Online Detection and Estimation of Grid Impedance Variation for Distributed Power Generation
DEFF Research Database (Denmark)
Jebali-Ben Ghorbal, Manel; Ghzaiel, Walid; Slama-Belkhodja, Ilhem
2012-01-01
A better knowledge of the grid impedance is essential in order to improve power quality and control of the Distributed Power Generation Systems (DPGS) and also for a safe connection or reconnection to the utility grid. An LCL-filter associated to a Voltage Source Inverter (VSI) is usually used...... to disconnect the DPG systems of the network. This work presents a rapid and simple technique to detect the grid impedance variation and to determine the grid impedance before and after grid faults accurs. Implementation on FPGA control board, simulations and experimental results are presented to validate...
Directory of Open Access Journals (Sweden)
Yufei Teng
2017-03-01
Full Text Available In order to improve the fault monitoring performance of grounding electrode lines in ultra-high voltage DC (UHVDC transmission systems, a novel fault monitoring approach based on the high-frequency voltage standing-wave ratio (VSWR is proposed in this paper. The VSWR is defined considering a lossless transmission line, and the characteristics of the VSWR under different conditions are analyzed. It is shown that the VSWR equals 1 when the terminal resistance completely matches the characteristic impedance of the line, and when a short circuit fault occurs on the grounding electrode line, the VSWR will be greater than 1. The VSWR will approach positive infinity under metallic earth fault conditions, whereas the VSWR in non-metallic earth faults will be smaller. Based on these analytical results, a fault supervision criterion is formulated. The effectiveness of the proposed VSWR-based fault supervision technique is verified with a typical UHVDC project established in Power Systems Computer Aided Design/Electromagnetic Transients including DC(PSCAD/EMTDC. Simulation results indicate that the proposed strategy can reliably identify the grounding electrode line fault and has strong anti-fault resistance capability.
Secondary control for voltage unbalance compensation in an islanded microgrid
DEFF Research Database (Denmark)
Savaghebi, Mehdi; Guerrero, Josep M.; Jalilian, Alireza
2011-01-01
In this paper, the concept of secondary control is applied for voltage unbalance compensation in an islanded microgrid. The aim of the proposed control approach is to enhance the voltage quality at the point of common coupling (PCC). Unbalance compensation is achieved by proper control...... of distributed generators (DGs). The DGs control structure mainly consists of active and reactive power controllers, virtual impedance loop and voltage and current proportional-resonant controllers. Simulation results are presented for different cases. The results show the effectiveness of the proposed approach...
Stupin, Daniil D.; Koniakhin, Sergei V.; Verlov, Nikolay A.; Dubina, Michael V.
2017-05-01
The time-domain technique for impedance spectroscopy consists of computing the excitation voltage and current response Fourier images by fast or discrete Fourier transformation and calculating their relation. Here we propose an alternative method for excitation voltage and current response processing for deriving a system impedance spectrum based on a fast and flexible adaptive filtering method. We show the equivalence between the problem of adaptive filter learning and deriving the system impedance spectrum. To be specific, we express the impedance via the adaptive filter weight coefficients. The noise-canceling property of adaptive filtering is also justified. Using the RLC circuit as a model system, we experimentally show that adaptive filtering yields correct admittance spectra and elements ratings in the high-noise conditions when the Fourier-transform technique fails. Providing the additional sensitivity of impedance spectroscopy, adaptive filtering can be applied to otherwise impossible-to-interpret time-domain impedance data. The advantages of adaptive filtering are justified with practical living-cell impedance measurements.
A 0.18 μm CMOS LDO Regulator for an On-Chip Sensor Array Impedance Measurement System.
Pérez-Bailón, Jorge; Márquez, Alejandro; Calvo, Belén; Medrano, Nicolás
2018-05-02
This paper presents a fully integrated 0.18 μm CMOS Low-Dropout (LDO) Voltage Regulator specifically designed to meet the stringent requirements of a battery-operated impedance spectrometry multichannel CMOS micro-instrument. The proposed LDO provides a regulated 1.8 V voltage from a 3.6 V to 1.94 V battery voltage over a −40 °C to 100 °C temperature range, with a compact topology (sensors.
Efficiency Investigation of Switch Mode Power Amplifier Drving Low Impedance Transducers
DEFF Research Database (Denmark)
Iversen, Niels Elkjær; Schneider, Henrik; Knott, Arnold
2015-01-01
the amplifier rail voltage requirement as a function of the voice coil nominal resistance is presented. The method is based on a crest factor analysis of music signals and estimation of the electrical power requirement from a specific target of the sound pressure level. Experimental measurements confirms a huge...... performance leap in terms of efficiency compared to a conventional battery driven sound system. Future optimization of low voltage, high current amplifiers for low impedance loudspeaker drivers are discussed....
Portable audio electronics for impedance-based measurements in microfluidics
International Nuclear Information System (INIS)
Wood, Paul; Sinton, David
2010-01-01
We demonstrate the use of audio electronics-based signals to perform on-chip electrochemical measurements. Cell phones and portable music players are examples of consumer electronics that are easily operated and are ubiquitous worldwide. Audio output (play) and input (record) signals are voltage based and contain frequency and amplitude information. A cell phone, laptop soundcard and two compact audio players are compared with respect to frequency response; the laptop soundcard provides the most uniform frequency response, while the cell phone performance is found to be insufficient. The audio signals in the common portable music players and laptop soundcard operate in the range of 20 Hz to 20 kHz and are found to be applicable, as voltage input and output signals, to impedance-based electrochemical measurements in microfluidic systems. Validated impedance-based measurements of concentration (0.1–50 mM), flow rate (2–120 µL min −1 ) and particle detection (32 µm diameter) are demonstrated. The prevailing, lossless, wave audio file format is found to be suitable for data transmission to and from external sources, such as a centralized lab, and the cost of all hardware (in addition to audio devices) is ∼10 USD. The utility demonstrated here, in combination with the ubiquitous nature of portable audio electronics, presents new opportunities for impedance-based measurements in portable microfluidic systems. (technical note)
Signal Processing for the Impedance Measurement on an Electrochemical Generator
Directory of Open Access Journals (Sweden)
El-Hassane AGLZIM
2008-04-01
Full Text Available Improving the life time of batteries or fuel cells requires the optimization of components such as membranes and electrodes and enhancement of the flow of gases [1], [2]. These goals could be reached by using a real time measurement on loaded generator. The impedance spectroscopy is a new way that was recently investigated. In this paper, we present an electronic measurement instrumentation developed in our laboratory to measure and plot the impedance of a loaded electrochemical generator like batteries and fuel cells. Impedance measures were done according to variations of the frequency in a larger band than what is usually used. The electronic instrumentation is controlled by Hpvee® software which allows us to plot the Nyquist graph of the electrochemical generator impedance. The theoretical results obtained in simulation under Pspice® confirm the choice of the method and its advantage. For safety reasons, the experimental preliminary tests were done on a 12 V vehicle battery, having an input current of 330 A and a capacity of 40 Ah and are now extended to a fuel cell. The results were plotted at various nominal voltages of the battery (12.7 V, 10 V, 8 V and 5 V and with two imposed currents (0.6 A and 4 A. The Nyquist diagram resulting from the experimental data enable us to show an influence of the load of the battery on its internal impedance. The similitude in the graph form and in order of magnitude of the values obtained (both theoretical and practical enables us to validate our electronic measurement instrumentation. Different sensors (temperature, pressure were placed around the device under test (DUT. These influence parameters were permanently recorded. Results presented here concern a classic loaded 12 V vehicle battery. The Nyquist diagram resulting from the experimental data confirms the influence of the load of the DUT on its internal impedance.
Water Electrolysis at Different Current - Voltage Regimes
International Nuclear Information System (INIS)
Kleperis, J.; Blums, J.; Vanags, M.
2007-01-01
Full text: Electrochemical impedance and volt-amperic methods were used to compare an efficiency of water electrolysis for different materials and different electrode configurations. Two and three electrode measurements were made, using standard calomel reference electrode. Non-standard capacitative electrolysis was analyzed in special cell made from cylindrical steel electrodes. Volt-amperic measurements from - 15V to +15V DC didn't indicated the presence of oxidation - reduction reactions when distilled water was used as electrolyte. Impedance measurements showed unusual frequency behavior when the AC voltage increased till 0.5V. Different nickel and carbon electrodes (plate, porous and textile - type) were used to learn classical Faraday electrolysis in strong alkali solutions. Flying increase of current was indicator of the presence of electrolysis, and characteristic potential was used differ between materials accordingly they effectiveness for usage in an electrolyser device. (Aithors)
Development of a fast voltage control method for electrostatic accelerators
International Nuclear Information System (INIS)
Lobanov, Nikolai R.; Linardakis, Peter; Tsifakis, Dimitrios
2014-01-01
The concept of a novel fast voltage control loop for tandem electrostatic accelerators is described. This control loop utilises high-frequency components of the ion beam current intercepted by the image slits to generate a correction voltage that is applied to the first few gaps of the low- and high-energy acceleration tubes adjoining the high voltage terminal. New techniques for the direct measurement of the transfer function of an ultra-high impedance structure, such as an electrostatic accelerator, have been developed. For the first time, the transfer function for the fast feedback loop has been measured directly. Slow voltage variations are stabilised with common corona control loop and the relationship between transfer functions for the slow and new fast control loops required for optimum operation is discussed. The main source of terminal voltage instabilities, which are due to variation of the charging current caused by mechanical oscillations of charging chains, has been analysed
Impedances of nickel electrodes cycled in various KOH concentrations
Reid, Margaret A.; Loyselle, Patricia L.
1991-01-01
Recent tests at Hughes have shown that Ni/H2 cells cycled in 26 percent KOH have much longer lives than those cycled in other concentrations. As part of an ongoing program to try to correlate the impedances of nickel electrodes with their life and performance, impedances were measured of a number of electrodes from these tests that had been cycled in concentrations from 21 to 36 percent KOH. These had ranged from about 1000 to 40,000 cycles. After cycling ten times to reduce possible changes due to storage, impedances were measured at five voltages corresponding to low states of charge. The results were analyzed using a standard circuit model including Warburg impedance term. Lower kinetic resistances and Warburg slopes were found for several electrodes which had been cycled in 26 percent KOH even though they had been cycled for a much longer time than the others. Interpretation of the data is complicated by the fact that the cycle lives, storage times, and failure mechanisms varied. Several other circuit models have also been examined, but the best correlations with life were found with parameters obtained from the simple model.
Multi-Electrode Impedance Method for Detection of Regional Ventilation
International Nuclear Information System (INIS)
Furuya, Norio; Sakamoto, Katsuyuki
2013-01-01
By means of computer simulation and experiment, we investigated the feasibility of simultaneously measuring the transfer impedance changes in the right apex, left apex, right base and left base of the lungs using the multi-electrode impedance method. To obtain the transfer impedance in each region, while suppressing the effects of other regions, changing the amplitude and polarity of the applied current must localize the high sensitivity areas in the interest region. Twelve current and eight voltage electrodes were equidistantly arranged on the anterior and posterior chest walls. The amplitudes and polarities of the currents that were simultaneously applied to the current electrodes, and which provided the appropriate sensitivity distribution, were theoretically obtained. The effects of the localized sensitivity distribution were verified by comparing the simulation results of the investigated method with the results of the conventional four-electrode method. From the results of the computer simulation, we developed a multi-electrode impedance pneumography and applied it to healthy adult volunteers who were both in sitting position and in left decubitus. We found that the measurement results were physiologically reasonable.
Jairo Blanco; Ruben Darío Leal; Jonathan Jacome; Johann F. Petit; Gabriel Ordoñez; Víctor Barrera
2011-01-01
This article presents an analysis of voltage sag propagation. The ATPDraw tool was selected for simulating the IEEE 34 node test feeder. It takes into account both voltage sags caused by electrical fault network, as well as voltage sag propagation characteristics caused by induction motor starting and transformer energising. The analysis was aimed at assessing the influence of transformer winding connections, the impedance of these transformers, lines and cables, summarising the...
A Reduced Switch Voltage Stress Class E Power Amplifier Using Harmonic Control Network
Ali Reza Zirak; Sobhan Roshani
2016-01-01
In this paper, a harmonic control network (HCN) is presented to reduce the voltage stress (maximum MOSFET voltage) of the class E power amplifier (PA). Effects of the HCN on the amplifier specifications are investigated. The results show that the proposed HCN affects several specifications of the amplifier, such as drain voltage, switch current, output power capability (Cp factor), and drain impedance. The output power capability of the presented amplifier is also improved, compared with the ...
Fast FPGA Implementation of an Original Impedance Analyser
Directory of Open Access Journals (Sweden)
Abdulrahman HAMED
2011-02-01
Full Text Available This article describes in detail the design and rapid prototyping of an embedded impedance analyzer. The measurement principle is based on the feedback control of the excitation voltage VD during a fast frequency sweeping. This function is carried out by a high precision synthesizer whose output resistance RG is digitally adjustable. Real and imaginary parts of the dipole impedance are determined from RG and the phase of VD. The digital architecture design uses the hardware-in-the-loop simulation in which the dipole is modeled using an RLC parallel circuit and a Butterworth Van Dyke structure. All digital functions are implemented on a Stratix II FPGA board with a 100 MHz frequency clock. The parameters taken into account are the frequency range (0 to 5 MHz, speed and resolution of the analysis and the quality factor of the resonant dipole. To reduce the analysis duration, the frequency sweeping rate is adjusted in real time.
Distance protection of multiple-circuit shared tower transmission lines with different voltages
DEFF Research Database (Denmark)
Silva, Filipe Miguel Faria da; Bak, Claus Leth
2017-01-01
combined faults, being advised to increase the resistive limit of the protection zone, if the network has lower short-circuit power. It is recommended to assure that the fault can only happen for cases where the faulted phase from the higher voltage level leads the faulted phase from the lower voltage......Multiple-circuit transmission lines combining different voltage levels in one tower present extra challenges when setting a protection philosophy, as faults between voltage levels are possible. In this study, the fault loop impedance of combined faults is compared with the fault loop impedance......-phase-to-ground faults. It is also demonstrated that the fault loop impedance of combined faults is more resistive, when compared with equivalent single-phase-to-ground faults. It is concluded that the settings used to protect a line against single-phase-to-ground faults are capable of protecting the line against...
Energy Technology Data Exchange (ETDEWEB)
Kimura, Tomoharu; Yamada, Hirofumi, E-mail: h-yamada@kuee.kyoto-u.ac.jp [Department of Electronic Science and Engineering, Kyoto University, Kyoto 615-8510 (Japan); Kobayashi, Kei [Department of Electronic Science and Engineering, Kyoto University, Kyoto 615-8510 (Japan); The Hakubi Center for Advanced Research, Kyoto University, Kyoto 615-8520 (Japan)
2015-08-07
The device performances of organic thin film transistors are often limited by the metal–organic interface because of the disordered molecular layers at the interface and the energy barriers against the carrier injection. It is important to study the local impedance at the interface without being affected by the interface morphology. We combined frequency modulation atomic force microscopy with scanning impedance microscopy (SIM) to sensitively measure the ac responses of the interface to an ac voltage applied across the interface and the dc potential drop at the interface. By using the frequency-modulation SIM (FM-SIM) technique, we characterized the interface impedance of a Pt electrode and a single pentacene grain as a parallel circuit of a contact resistance and a capacitance. We found that the reduction of the contact resistance was caused by the reduction of the energy level mismatch at the interface by the FM-SIM measurements, demonstrating the usefulness of the FM-SIM technique for investigation of the local interface impedance without being affected by its morphology.
Technological Aspects: High Voltage
International Nuclear Information System (INIS)
Faircloth, D C
2013-01-01
This paper covers the theory and technological aspects of high-voltage design for ion sources. Electric field strengths are critical to understanding high-voltage breakdown. The equations governing electric fields and the techniques to solve them are discussed. The fundamental physics of high-voltage breakdown and electrical discharges are outlined. Different types of electrical discharges are catalogued and their behaviour in environments ranging from air to vacuum are detailed. The importance of surfaces is discussed. The principles of designing electrodes and insulators are introduced. The use of high-voltage platforms and their relation to system design are discussed. The use of commercially available high-voltage technology such as connectors, feedthroughs and cables are considered. Different power supply technologies and their procurement are briefly outlined. High-voltage safety, electric shocks and system design rules are covered. (author)
Technological Aspects: High Voltage
Faircloth, D.C.
2013-12-16
This paper covers the theory and technological aspects of high-voltage design for ion sources. Electric field strengths are critical to understanding high-voltage breakdown. The equations governing electric fields and the techniques to solve them are discussed. The fundamental physics of high-voltage breakdown and electrical discharges are outlined. Different types of electrical discharges are catalogued and their behaviour in environments ranging from air to vacuum are detailed. The importance of surfaces is discussed. The principles of designing electrodes and insulators are introduced. The use of high-voltage platforms and their relation to system design are discussed. The use of commercially available high-voltage technology such as connectors, feedthroughs and cables are considered. Different power supply technologies and their procurement are briefly outlined. High-voltage safety, electric shocks and system design rules are covered.
Rababah, A S; Walsh, S J; Manoharan, G; Walsh, P R; Escalona, O J
2016-07-01
Intracardiac impedance (ICI) is a major determinant of success during internal cardioversion of atrial fibrillation (AF). However, there have been few studies that have examined the dynamic behaviour of atrial impedance during internal cardioversion in relation to clinical outcome. In this study, voltage and current waveforms captured during internal cardioversion of acute AF in ovine models using novel radiofrequency (RF) generated low-tilt rectilinear and conventional capacitor-discharge based shock waveforms were retrospectively analysed using a digital signal processing algorithm to investigate the dynamic behaviour of atrial impedance during cardioversion. The algorithm was specifically designed to facilitate the simultaneous analysis of multiple impedance parameters, including: mean intracardiac impedance (Z M), intracardiac impedance variance (ICIV) and impedance amplitude spectrum area (IAMSA) for each cardioversion event. A significant reduction in ICI was observed when comparing two successive shocks of increasing energy where cardioversion outcome was successful. In addition, ICIV and IAMSA variables were found to inversely correlate to the magnitude of energy delivered; with a stronger correlation found to the former parameter. In conclusion, ICIV and IAMSA have been evidenced as two key dynamic intracardiac impedance variables that may prove useful in better understanding of the cardioversion process and that could potentially act as prognostic markers with respect to clinical outcome.
Directory of Open Access Journals (Sweden)
A. Lasheen
2018-03-01
Full Text Available Microwave instability in the Super Proton Synchrotron (SPS at CERN is one of the main limitations to reach the requirements for the High Luminosity-LHC project (increased beam intensity by a factor 2. To identify the impedance source responsible of the instability, beam measurements were carried out to probe the SPS impedance. The method presented in this paper relies on measurements of the unstable spectra of single bunches, injected in the SPS with the rf voltage switched off. The modulation of the bunch profile gives information about the main impedance sources driving microwave instability, and is compared to particle simulations using the SPS impedance model to identify the most important contributions. This allowed us to identify the vacuum flanges as the main impedance source for microwave instability in the SPS, and to evaluate possible missing impedance sources.
Lasheen, A.; Argyropoulos, T.; Bohl, T.; Esteban Müller, J. F.; Timko, H.; Shaposhnikova, E.
2018-03-01
Microwave instability in the Super Proton Synchrotron (SPS) at CERN is one of the main limitations to reach the requirements for the High Luminosity-LHC project (increased beam intensity by a factor 2). To identify the impedance source responsible of the instability, beam measurements were carried out to probe the SPS impedance. The method presented in this paper relies on measurements of the unstable spectra of single bunches, injected in the SPS with the rf voltage switched off. The modulation of the bunch profile gives information about the main impedance sources driving microwave instability, and is compared to particle simulations using the SPS impedance model to identify the most important contributions. This allowed us to identify the vacuum flanges as the main impedance source for microwave instability in the SPS, and to evaluate possible missing impedance sources.
Homma, Akira
2011-07-01
A novel annular parallel-strip transmission line was devised to construct high-voltage high-speed pulse isolation transformers. The transmission lines can easily realize stable high-voltage operation and good impedance matching between primary and secondary circuits. The time constant for the step response of the transformer was calculated by introducing a simple low-frequency equivalent circuit model. Results show that the relation between the time constant and low-cut-off frequency of the transformer conforms to the theory of the general first-order linear time-invariant system. Results also show that the test transformer composed of the new transmission lines can transmit about 600 ps rise time pulses across the dc potential difference of more than 150 kV with insertion loss of -2.5 dB. The measured effective time constant of 12 ns agreed exactly with the theoretically predicted value. For practical applications involving the delivery of synchronized trigger signals to a dc high-voltage electron gun station, the transformer described in this paper exhibited advantages over methods using fiber optic cables for the signal transfer system. This transformer has no jitter or breakdown problems that invariably occur in active circuit components.
Transvers Impedance Measurements of the Modified DARHT-2 Accelerator Cell Design
International Nuclear Information System (INIS)
Briggs, Dick; Waldron, Will
2005-01-01
The DARHT-2 accelerator cells have been redesigned to make their high voltage performance more robust. At the outset of the DARHT-2 development program about 8 years ago, an extensive campaign was mounted to minimize the transverse impedance of the original cell design. Since the initial spec on the machine was a beam current of 4 kA, the control of beam-breakup (BBU) amplification with a 2 microsecond pulse length was considered to be one the most critical issues in the design. Even after advances in detector technology allowed the beam current requirement to be lowered to 2 kA, the goal for the standard cell impedance was kept at ∼300 ohms/meter to allow for the possibility of future beam current upgrades to 4 kA without any modifications in the cells. The results of this campaign to minimize the transverse impedance are described in detail in Reference 1. After several iterations in the design of ferrite dampers and the anode finger stock shape, the measured (peak) impedance of the original standard cell was determined to be about 280 ohms/meter. (As a reference point, the measured impedance of the DARHT-1 cell is about 880 ohms/meter). This impedance provided such a wide safety margin against BBU amplification at 2 kA that it was felt that the cell redesign could focus on voltage holding without any detailed considerations of impacts on the transverse impedance. Now that a baseline design for the DARHT-2 cell has been established and tested, however, it was felt that a measurement of its impedance would be prudent. The results of these impedance measurements are presented in this note. The objective was mainly to do a ''quick check'' to ensure that there were no surprises, and to provide an estimate of the BBU frequencies and growth rates to the experimental test program
Current phase control test based on real-time measurement of impedance matrix of ICRF antennas
Energy Technology Data Exchange (ETDEWEB)
Saito, K., E-mail: saito@nifs.ac.jp [National Institute for Fusion Science, Toki, Gifu 509-5292 (Japan); Kumazawa, R.; Seki, T.; Kasahara, H.; Yokota, M.; Nomura, G.; Shimpo, F.; Mutoh, T. [National Institute for Fusion Science, Toki, Gifu 509-5292 (Japan)
2011-10-15
New ion cyclotron range of frequencies (ICRF) antennas have just been installed in the large helical device (LHD). These side-by-side ICRF antennas are symmetrical and designed to launch fast waves with various wave numbers parallel to the magnetic field line. The wave number can be controlled by changing the current phase on the straps; however, the mutual coupling between antennas changes antenna impedances, even if the plasma parameters are constant, leading to an increase in the reflected power. In addition to the current phase control, impedance matching devices must be tuned for the protection of tetrode tubes and efficient power injection. For this purpose, the impedance matrix of ICRF antennas must be determined, and it can be deduced from the forward and reflected waves at the outlet of the power amplifier by assuming geometric symmetry and reciprocity of the antennas. Using half-scale antennas, we successfully demonstrated simultaneous impedance matching and current phase control.
DEFF Research Database (Denmark)
Hyvönen, N.; Majander, H.; Staboulis, Stratos
2017-01-01
Electrical impedance tomography aims at reconstructing the conductivity inside a physical body from boundary measurements of current and voltage at a finite number of contact electrodes. In many practical applications, the shape of the imaged object is subject to considerable uncertainties...
Unbalanced Impedance of the Aperture Coils of Some LHC Main Dipole Magnets
Ravaioli, Emanuele; Verweij, A.P.; ten Kate, Herman H.J.
2013-01-01
During the Large Hadron Collider operation in 2010 and 2011, it was observed that about 50% of the main dipole magnets exhibiting two apertures showed unbalanced dynamic-impedance behavior not well understood. When a main dipole circuit is switched off, voltage waves with a frequency of 28 Hz travel
Energy Technology Data Exchange (ETDEWEB)
Jamali, B.; Piercy, R.; Dick, P. [Kinetrics Inc., Toronto, ON (Canada). Transmission and Distribution Technologies
2008-04-09
This report discussed issues related to farm stray voltage and evaluated mitigation strategies and costs for limiting voltage to farms. A 3-phase, 3-wire system with no neutral ground was used throughout North America before the 1930s. Transformers were connected phase to phase without any electrical connection between the primary and secondary sides of the transformers. Distribution voltage levels were then increased and multi-grounded neutral wires were added. The earth now forms a parallel return path for the neutral current that allows part of the neutral current to flow continuously through the earth. The arrangement is responsible for causing stray voltage. Stray voltage causes uneven milk production, increased incidences of mastitis, and can create a reluctance to drink water amongst cows when stray voltages are present. Off-farm sources of stray voltage include phase unbalances, undersized neutral wire, and high resistance splices on the neutral wire. Mitigation strategies for reducing stray voltage include phase balancing; conversion from single to 3-phase; increasing distribution voltage levels, and changing pole configurations. 22 refs., 5 tabs., 13 figs.
Rizk, Farouk AM
2014-01-01
Inspired by a new revival of worldwide interest in extra-high-voltage (EHV) and ultra-high-voltage (UHV) transmission, High Voltage Engineering merges the latest research with the extensive experience of the best in the field to deliver a comprehensive treatment of electrical insulation systems for the next generation of utility engineers and electric power professionals. The book offers extensive coverage of the physical basis of high-voltage engineering, from insulation stress and strength to lightning attachment and protection and beyond. Presenting information critical to the design, selec
Kind, Dieter
2001-01-01
The second edition of High Voltage Test Techniques has been completely revised. The present revision takes into account the latest international developments in High Voltage and Measurement technology, making it an essential reference for engineers in the testing field.High Voltage Technology belongs to the traditional area of Electrical Engineering. However, this is not to say that the area has stood still. New insulating materials, computing methods and voltage levels repeatedly pose new problems or open up methods of solution; electromagnetic compatibility (EMC) or components and systems al
Sato-Akaba, Hideo
2014-01-01
A low impedance transceiver circuit consisting of a transmit-receive switch circuit, a class-D amplifier and a transimpedance amplifier (TIA) was newly designed and tested for a nitrogen-14 NQR. An NQR signal at 1.37MHz from imidazole was successfully observed with the dead time of ~85µs under the high Q transmission (Q~120) and reception (Q~140). The noise performance of the low impedance TIA with an NQR probe was comparable with a commercial low noise 50Ω amplifier (voltage input noise: 0.25 nV/Hz) which was also connected to the probe. The protection voltage for the pre-amplifier using the low impedance transceiver was ~10 times smaller than that for the pre-amplifier using a 50Ω conventional transceiver, which is suitable for NQR remote sensing applications. Copyright © 2014 Elsevier Inc. All rights reserved.
Rivera-Serrano, Nilka; Pagan, Miraida; Colón-Rodríguez, Joanisse; Fuster, Christian; Vélez, Román; Almodovar-Faria, Jose; Jiménez-Rivera, Carlos; Cunci, Lisandro
2018-02-06
In this study, electrochemical impedance spectroscopy was used for the first time to study the adsorption of dopamine in carbon fiber microelectrodes. In order to show a proof-of-concept, static and dynamic measurements were taken at potentials ranging from -0.4 to 0.8 V versus Ag|AgCl to demonstrate the versatility of this technique to study dopamine without the need of its oxidation. We used electrochemical impedance spectroscopy and single frequency electrochemical impedance to measure different concentrations of dopamine as low as 1 nM. Moreover, the capacitance of the microelectrodes surface was found to decrease due to dopamine adsorption, which is dependent on its concentration. The effect of dissolved oxygen and electrochemical oxidation of the surface in the detection of dopamine was also studied. Nonoxidized and oxidized carbon fiber microelectrodes were prepared and characterized by optical microscopy, scanning electron microscopy, cyclic voltammetry, and electrochemical impedance spectroscopy. Optimum working parameters of the electrodes, such as frequency and voltage, were obtained for better measurement. Electrochemical impedance of dopamine was determined at different concentration, voltages, and frequencies. Finally, dynamic experiments were conducted using a flow cell and single frequency impedance in order to study continuous and real-time measurements of dopamine.
Desai, Sharanya Arcot; Rolston, John D.; Guo, Liang; Potter, Steve M.
2010-01-01
Implantable microelectrode arrays (MEAs) have been a boon for neural stimulation and recording experiments. Commercially available MEAs have high impedances, due to their low surface area and small tip diameters, which are suitable for recording single unit activity. Lowering the electrode impedance, but preserving the small diameter, would provide a number of advantages, including reduced stimulation voltages, reduced stimulation artifacts and improved signal-to-noise ratio. Impedance reductions can be achieved by electroplating the MEAs with platinum (Pt) black, which increases the surface area but has little effect on the physical extent of the electrodes. However, because of the low durability of Pt black plating, this method has not been popular for chronic use. Sonicoplating (i.e. electroplating under ultrasonic agitation) has been shown to improve the durability of Pt black on the base metals of macro-electrodes used for cyclic voltammetry. This method has not previously been characterized for MEAs used in chronic neural implants. We show here that sonicoplating can lower the impedances of microwire multi-electrode arrays (MMEA) by an order of magnitude or more (depending on the time and voltage of electroplating), with better durability compared to pulsed plating or traditional DC methods. We also show the improved stimulation and recording performance that can be achieved in an in vivo implantation study with the sonicoplated low-impedance MMEAs, compared to high-impedance unplated electrodes. PMID:20485478
Cheng, Xin-bing; Liu, Jin-liang; Qian, Bao-liang; Zhang, Yu; Zhang, Hong-bo
2009-11-01
A high voltage pulse transformer (HVPT) is usually used as a charging device for the pulse forming line (PFL) of intense electron-beam accelerators (IEBAs). Insulation of the HVPT is one of the important factors that restrict the development of the HVPT. Until now, considerable effort has been focused on minimizing high field regions to avoid insulation breakdown between windings. Characteristics of the HVPT have been widely discussed to achieve these goals, but the effects of the PFL and load resistance on HVPT are usually neglected. In this paper, a HVPT is used as a charging device for the PFL of an IEBA and the effect of the change in the load resistance on the HVPT of the IEBA is presented. When the load resistance does not match the wave impedance of the PFL, a high-frequency bipolar oscillating voltage will occur, and the amplitude of the oscillating voltage will increase with the decrease in the load resistance. The load resistance approximates to zero and the amplitude of the oscillating voltage is much higher. This makes it easier for surface flashover along the insulation materials to form and decrease the lifetime of the HVPT.
Plasma diagnosis of RF discharge by using impedance measurement
International Nuclear Information System (INIS)
Huang Jianjun; Teuner, D.
2001-01-01
It is presented that the method known from network analysis with home-made probe and experimental setup to measure current, voltage and phase angle of RF discharge in He gas more accurately. The sheath thickness and the real and imaginary parts of the plasma impedance were obtained by using the equivalent circuit model and taking account stray capacitances of the set-up. In addition, making use of Godyak's RF discharge simple model, the electron density in the discharge was calculated at different pressure and current density
High-Voltage, Low-Power BNC Feedthrough Terminator
Bearden, Douglas
2012-01-01
This innovation is a high-voltage, lowpower BNC (Bayonet Neill-Concelman) feedthrough that enables the user to terminate an instrumentation cable properly while connected to a high voltage, without the use of a voltage divider. This feedthrough is low power, which will not load the source, and will properly terminate the instrumentation cable to the instrumentation, even if the cable impedance is not constant. The Space Shuttle Program had a requirement to measure voltage transients on the orbiter bus through the Ground Lightning Measurement System (GLMS). This measurement has a bandwidth requirement of 1 MHz. The GLMS voltage measurement is connected to the orbiter through a DC panel. The DC panel is connected to the bus through a nonuniform cable that is approximately 75 ft (approximately equal to 23 m) long. A 15-ft (approximately equal to 5-m), 50-ohm triaxial cable is connected between the DC panel and the digitizer. Based on calculations and simulations, cable resonances and reflections due to mismatched impedances of the cable connecting the orbiter bus and the digitizer causes the output not to reflect accurately what is on the bus. A voltage divider at the DC panel, and terminating the 50-ohm cable properly, would eliminate this issue. Due to implementation issues, an alternative design was needed to terminate the cable properly without the use of a voltage divider. Analysis shows how the cable resonances and reflections due to the mismatched impedances of the cable connecting the orbiter bus and the digitizer causes the output not to reflect accurately what is on the bus. After simulating a dampening circuit located at the digitizer, simulations were performed to show how the cable resonances were dampened and the accuracy was improved significantly. Test cables built to verify simulations were accurate. Since the dampening circuit is low power, it can be packaged in a BNC feedthrough.
A method of accurate determination of voltage stability margin
Energy Technology Data Exchange (ETDEWEB)
Wiszniewski, A.; Rebizant, W. [Wroclaw Univ. of Technology, Wroclaw (Poland); Klimek, A. [AREVA Transmission and Distribution, Stafford (United Kingdom)
2008-07-01
In the process of developing power system disturbance, voltage instability at the receiving substations often contributes to deteriorating system stability, which eventually may lead to severe blackouts. The voltage stability margin at receiving substations may be used to determine measures to prevent voltage collapse, primarily by operating or blocking the transformer tap changing device, or by load shedding. The best measure of the stability margin is the actual load to source impedance ratio and its critical value, which is unity. This paper presented an accurate method of calculating the load to source impedance ratio, derived from the Thevenin's equivalent circuit of the system, which led to calculation of the stability margin. The paper described the calculation of the load to source impedance ratio including the supporting equations. The calculation was based on the very definition of voltage stability, which says that system stability is maintained as long as the change of power, which follows the increase of admittance is positive. The testing of the stability margin assessment method was performed in a simulative way for a number of power network structures and simulation scenarios. Results of the simulations revealed that this method is accurate and stable for all possible events occurring downstream of the device location. 3 refs., 8 figs.
Impedance Analysis of Silicon Nanowire Lithium Ion Battery Anodes
Ruffo, Riccardo
2009-07-02
The impedance behavior of silicon nanowire electrodes has been investigated to understand the electrochemical process kinetics that influences the performance when used as a high-capacity anode in a lithium ion battery. The ac response was measured by using impedance spectroscopy in equilibrium conditions at different lithium compositions and during several cycles of charge and discharge in a half cell vs. metallic lithium. The impedance analysis shows the contribution of both surface resistance and solid state diffusion through the bulk of the nanowires. The surface process is dominated by a solid electrolyte layer (SEI) consisting of an inner, inorganic insoluble part and several organic compounds at the outer interface, as seen by XPS analysis. The surface resistivity, which seems to be correlated with the Coulombic efficiency of the electrode, grows at very high lithium contents due to an increase in the inorganic SEI thickness. We estimate the diffusion coefficient of about 2 × 10 -10 cm 2/s for lithium diffusion in silicon. A large increase in the electrode impedance was observed at very low lithium compositions, probably due to a different mechanism for lithium diffusion inside the wires. Restricting the discharge voltage to 0.7 V prevents this large impedance and improves the electrode lifetime. Cells cycled between 0.07 and 0.70 V vs. metallic lithium at a current density of 0.84 A/g (C/5) showed good Coulombic efficiency (about 99%) and maintained a capacity of about 2000 mAh/g after 80 cycles. © 2009 American Chemical Society.
DEFF Research Database (Denmark)
Zhao, Xin; Firoozabadi, Mehdi Savaghebi; Quintero, Juan Carlos Vasquez
2015-01-01
. In this paper, a voltage support strategy based on negative sequence droop control, which regulate the positive/negative sequence active and reactive power flow by means of sending proper voltage reference to the inner control loop, is proposed for the grid connected MGs to ride through voltage sags under...... complex line impedance conditions. In this case, the MGs should inject a certain amount of positive and negative sequence power to the grid so that the voltage quality at load side can be maintained at a satisfied level. A two layer hierarchical control strategy is proposed in this paper. The primary...... control loop consists of voltage and current inner loops, conventional droop control and virtual impedance loop while the secondary control loop is based on positive/negative sequence droop control which can achieve power injection under voltage sags. Experimental results with asymmetrical voltage sags...
Ida, Midori; Hirata, Masakazu; Hosoda, Kiminori; Nakao, Kazuwa
2013-02-01
Two novel bioelectrical impedance analysis (BIA) methods have been developed recently for evaluation of intra-abdominal fat accumulation. Both methods use electrodes that are placed on abdominal wall and allow evaluation of intra-abdominal fat area (IAFA) easily without radiation exposure. Of these, "abdominal BIA" method measures impedance distribution along abdominal anterior-posterior axis, and IAFA by BIA method(BIA-IAFA) is calculated from waist circumference and the voltage occurring at the flank. Dual BIA method measures impedance of trunk and body surface at the abdominal level and calculates BIA-IAFA from transverse and antero-posterior diameters of the abdomen and the impedance of trunk and abdominal surface. BIA-IAFA by these two BIA methods correlated well with IAFA measured by abdominal CT (CT-IAFA) with correlatipn coefficient of 0.88 (n = 91, p abdominal adiposity in clinical study and routine clinical practice of metabolic syndrome and obesity.
Optimized transmission-line impedance transformers for petawatt-class pulsed-power accelerators
Directory of Open Access Journals (Sweden)
D. R. Welch
2008-03-01
Full Text Available We have developed 1D analytic and 2D fully electromagnetic models of radial transmission-line impedance transformers. The models have been used to quantify the power-transport efficiency and pulse sharpening of such transformers as a function of voltage pulse width and impedance profile. For the cases considered, we find that in the limit as Γ→0 (where Γ is the ratio of the pulse width to the one-way transit time of the transformer, the transport efficiency is maximized when the impedance profile is exponential. As Γ increases from zero, the optimum profile gradually deviates from an exponential. A numerical procedure is presented that determines the optimum profile for a given pulse shape and width. The procedure can be applied to optimize the design of impedance transformers used in petawatt-class pulsed-power accelerators.
Mathematical modeling of the electrochemical impedance spectroscopy in lithium ion battery cycling
International Nuclear Information System (INIS)
Xie, Yuanyuan; Li, Jianyang; Yuan, Chris
2014-01-01
Electrochemical impedance spectroscopy (EIS) has been widely utilized as an experimental method for understanding the internal mechanisms and aging effect of lithium ion battery. However, the impedance interpretation still has a lot of difficulties. In this study, a multi-physics based EIS simulation approach is developed to study the cycling effect on the battery impedance responses. The SEI film growth during cycling is coherently coupled with the complicated charge, mass and energy transport processes. The EIS simulation is carried out by applying a perturbation voltage on the electrode surface, and the numerical results on cycled cells are compared with the corresponding experimental data. The effect of electrical double layer, electrode open circuit potential as well as the diffusivity of binary electrolyte are simulated on battery impedance responses. The influence of different SEI growth rate, thermal conditions and charging-discharging rate during cycling are also studied. This developed method can be potentially utilized for interpretation and analysis of experimental EIS results
Transition metal oxide as anode interface buffer for impedance spectroscopy
Xu, Hui; Tang, Chao; Wang, Xu-Liang; Zhai, Wen-Juan; Liu, Rui-Lan; Rong, Zhou; Pang, Zong-Qiang; Jiang, Bing; Fan, Qu-Li; Huang, Wei
2015-12-01
Impedance spectroscopy is a strong method in electric measurement, which also shows powerful function in research of carrier dynamics in organic semiconductors when suitable mathematical physical models are used. Apart from this, another requirement is that the contact interface between the electrode and materials should at least be quasi-ohmic contact. So in this report, three different transitional metal oxides, V2O5, MoO3 and WO3 were used as hole injection buffer for interface of ITO/NPB. Through the impedance spectroscopy and PSO algorithm, the carrier mobilities and I-V characteristics of the NPB in different devices were measured. Then the data curves were compared with the single layer device without the interface layer in order to investigate the influence of transitional metal oxides on the carrier mobility. The careful research showed that when the work function (WF) of the buffer material was just between the work function of anode and the HOMO of the organic material, such interface material could work as a good bridge for carrier injection. Under such condition, the carrier mobility measured through impedance spectroscopy should be close to the intrinsic value. Considering that the HOMO (or LUMO) of most organic semiconductors did not match with the work function of the electrode, this report also provides a method for wide application of impedance spectroscopy to the research of carrier dynamics.
Tapping mode microwave impedance microscopy
Lai, K.
2009-01-01
We report tapping mode microwave impedance imaging based on atomic force microscope platforms. The shielded cantilever probe is critical to localize the tip-sample interaction near the tip apex. The modulated tip-sample impedance can be accurately simulated by the finite-element analysis and the result agrees quantitatively to the experimental data on a series of thin-film dielectric samples. The tapping mode microwave imaging is also superior to the contact mode in that the thermal drift in a long time scale is totally eliminated and an absolute measurement on the dielectric properties is possible. We demonstrated tapping images on working nanodevices, and the data are consistent with the transport results. © 2009 American Institute of Physics.
Tuning and matching of the BPX ICH system
International Nuclear Information System (INIS)
Swain, D.W.
1991-01-01
Two methods of tuning and matching the ion cyclotron heating (ICH) antennas for the Burning Plasma Experiment (BPX) to the BPX plasma have been analyzed. Both appear to provide adequate tuning and matching capabilities. However, there are trade-offs between the frequency range that can be covered and the compactness of the high-voltage region of the transmission lines that makes up of the matching network. 4 refs., 5 figs
Electrical impedance tomography: topology optimization
International Nuclear Information System (INIS)
Miranda, Lenine Campos
2013-01-01
The Electrical Impedance Tomography (EIT) is a study of body parts who use electric current. Is studied through computers resistance or conductivity of these parts, producing an image used for medical diagnosis. A body is wrapped in a blanket placed with small electrodes and receivers of electric current, potential difference. Based on data obtained from a series of measurements at the electrodes, one by one, sending and receiving, you can perform a numerical phantom, where each 'voxel' of the image formed computationally represents the impedance of biological tissue. In Brazil, studies on electrical impedance tomography (EIT) has not yet started. Such equipment are measured tensions - potential difference - between each electrode / sensor one by one, as a way to Simple Combinatorial Analysis. The sequence and the way it is measured strains are in the final image quality. Finite Element Method Interactive, whose algorithm is based on Dialectical Method. We use an initial function with the objective of maximizing the data quantitatively, for better qualitative analysis. Topology Optimization methods are used to improve the image reconstruction. Currently the study is quite primitive related to the theory that shows how to power the new science studied. The high quality images requires a difficulty in obtaining. This work is not intended for detailed for analysis in any tissue or organ specific, but in general terms. And the formation of the 2D image. 3D need a reconstructor to part. (author)
Cascaded impedance networks for NPC inverter
DEFF Research Database (Denmark)
Li, Ding; Gao, Feng; Loh, Poh Chiang
2010-01-01
they are subject to the renewable sources. To date, three distinct types of impedance networks can be summarized for implementing a hybrid source impedance network, which can in principle be combined and cascaded before connected to a NPC inverter by proposed two ways. The resulting cascaded impedance network NPC...
Impedance analysis of acupuncture points and pathways
International Nuclear Information System (INIS)
Teplan, Michal; Kukucka, Marek; Ondrejkovicová, Alena
2011-01-01
Investigation of impedance characteristics of acupuncture points from acoustic to radio frequency range is addressed. Discernment and localization of acupuncture points in initial single subject study was unsuccessfully attempted by impedance map technique. Vector impedance analyses determined possible resonant zones in MHz region.
Journal bearing impedance descriptions for rotordynamic applications
Childs, D.W.; Moes, H.; Leeuwen, van H.J.
1977-01-01
Bearing impedance vectors are introduced for plain journal bearings which define the bearing reaction force components as a function of the bearing motion. Impedance descriptions are developed directly for the approximate Ocvirk (short) and Sommerfeld (long) bearing solutions. The impedance vector
... for Kids ▸ Stinging Insect Matching Game Share | Stinging Insect Matching Game Stinging insects can ruin summer fun for those who are ... the difference between the different kinds of stinging insects in order to keep your summer safe and ...
Control voltage and power fluctuations when connecting wind farms
Berinde, Ioan; Bǎlan, Horia; Oros Pop, Teodora Susana
2015-12-01
Voltage, frequency, active power and reactive power are very important parameters in terms of power quality. These parameters are followed when connecting any power plant, the more the connection of wind farms. Connecting wind farms to the electricity system must not cause interference outside the limits set by regulations. Modern solutions for fast and automatic voltage control and power fluctuations using electronic control systems of reactive power flows. FACTS (Flexible Alternating Current Transmision System) systems, established on the basis of power electronic circuits ensure control of electrical status quantities to achieve the necessary transfer of power to the power grid. FACTS devices can quickly control parameters and sizes of state power lines, such as impedance line voltages and phase angles of the voltages of the two ends of the line. Their use can lead to improvement in power system operation by increasing the transmission capacity of power lines, power flow control lines, improved static and transient stability reserve.
Control voltage and power fluctuations when connecting wind farms
International Nuclear Information System (INIS)
Berinde, Ioan; Bălan, Horia; Oros, Teodora Susana
2015-01-01
Voltage, frequency, active power and reactive power are very important parameters in terms of power quality. These parameters are followed when connecting any power plant, the more the connection of wind farms. Connecting wind farms to the electricity system must not cause interference outside the limits set by regulations. Modern solutions for fast and automatic voltage control and power fluctuations using electronic control systems of reactive power flows. FACTS (Flexible Alternating Current Transmision System) systems, established on the basis of power electronic circuits ensure control of electrical status quantities to achieve the necessary transfer of power to the power grid. FACTS devices can quickly control parameters and sizes of state power lines, such as impedance line voltages and phase angles of the voltages of the two ends of the line. Their use can lead to improvement in power system operation by increasing the transmission capacity of power lines, power flow control lines, improved static and transient stability reserve
Control voltage and power fluctuations when connecting wind farms
Energy Technology Data Exchange (ETDEWEB)
Berinde, Ioan, E-mail: ioan-berinde@yahoo.com; Bălan, Horia, E-mail: hbalan@mail.utcluj.ro; Oros, Teodora Susana, E-mail: teodoraoros-87@yahoo.com [Technical University of Cluj-Napoca, Romania, Faculty of Electrical Engineering, Department of Power Engineering and Management (Romania)
2015-12-23
Voltage, frequency, active power and reactive power are very important parameters in terms of power quality. These parameters are followed when connecting any power plant, the more the connection of wind farms. Connecting wind farms to the electricity system must not cause interference outside the limits set by regulations. Modern solutions for fast and automatic voltage control and power fluctuations using electronic control systems of reactive power flows. FACTS (Flexible Alternating Current Transmision System) systems, established on the basis of power electronic circuits ensure control of electrical status quantities to achieve the necessary transfer of power to the power grid. FACTS devices can quickly control parameters and sizes of state power lines, such as impedance line voltages and phase angles of the voltages of the two ends of the line. Their use can lead to improvement in power system operation by increasing the transmission capacity of power lines, power flow control lines, improved static and transient stability reserve.
Meandered-line antenna with integrated high-impedance surface.
Energy Technology Data Exchange (ETDEWEB)
Forman, Michael A.
2010-09-01
A reduced-volume antenna composed of a meandered-line dipole antenna over a finite-width, high-impedance surface is presented. The structure is novel in that the high-impedance surface is implemented with four Sievenpiper via-mushroom unit cells, whose area is optimized to match the meandered-line dipole antenna. The result is an antenna similar in performance to patch antenna but one fourth the area that can be deployed directly on the surface of a conductor. Simulations demonstrate a 3.5 cm ({lambda}/4) square antenna with a bandwidth of 4% and a gain of 4.8 dBi at 2.5 GHz.
Bench measurements of coupling impedance of AGS Booster components
International Nuclear Information System (INIS)
Ratti, A.; Shea, T.J.
1991-01-01
Quantifying instability thresholds for modern synchrotrons and storage rings requires some knowledge of the accelerator's coupling impedance. To this end, the wire technique has been implemented to measure the longitudinal coupling impedance of AGS Booster devices. The techniques are being refined to allow measurement of RHIC devices at higher frequencies. All the measurements are performed using an HP 8753 Network Analyzer controlled via GPIB by a Macintosh computer. The computer provides an environment for automated data acquisition, data analysis, and report generation. Resistive matches between the 50ω analyzer cables and the 300ω pipe-and-wire structure allow the use of a simple response calibration in the measurement of S21 to 400MHz. Results from ferrite loaded rf cavities, position monitors and kickers are presented. 4 refs., 4 figs
based dynamic voltage restorer
African Journals Online (AJOL)
HOD
operation due to presence of increased use of nonlinear loads (computers, microcontrollers ... simulations of a dynamic voltage restorer (DVR) was achieved using MATLAB/Simulink. ..... using Discrete PWM generator, then the IGBT inverter.
Best matching theory & applications
Moghaddam, Mohsen
2017-01-01
Mismatch or best match? This book demonstrates that best matching of individual entities to each other is essential to ensure smooth conduct and successful competitiveness in any distributed system, natural and artificial. Interactions must be optimized through best matching in planning and scheduling, enterprise network design, transportation and construction planning, recruitment, problem solving, selective assembly, team formation, sensor network design, and more. Fundamentals of best matching in distributed and collaborative systems are explained by providing: § Methodical analysis of various multidimensional best matching processes § Comprehensive taxonomy, comparing different best matching problems and processes § Systematic identification of systems’ hierarchy, nature of interactions, and distribution of decision-making and control functions § Practical formulation of solutions based on a library of best matching algorithms and protocols, ready for direct applications and apps development. Design...
Directory of Open Access Journals (Sweden)
Dejan Trifunovic
2017-08-01
Full Text Available In this paper, we present the problem of matching students to schools by using different matching mechanisms. This market is specific since public schools are free and the price mechanism cannot be used to determine the optimal allocation of children in schools. Therefore, it is necessary to use different matching algorithms that mimic the market mechanism and enable us to determine the core of the cooperative game. In this paper, we will determine that it is possible to apply cooperative game theory in matching problems. This review paper is based on illustrative examples aiming to compare matching algorithms in terms of the incentive compatibility, stability and efficiency of the matching. In this paper we will present some specific problems that may occur in matching, such as improving the quality of schools, favoring minority students, the limited length of the list of preferences and generating strict priorities from weak priorities.
Control and Protection Cooperation Strategy for Voltage Instability
DEFF Research Database (Denmark)
Liu, Zhou; Chen, Zhe; Sun, Haishun
2012-01-01
Most cascaded blackouts are caused by unexpected backup relay operations due to low voltage or overload state caused by post fault load restoration dynamics. If such state can be sensed and adjusted appropriately prior to those relay actions, system stability might be sustained. This paper proposed...... a control and protection cooperation strategy to prevent post fault voltage instability. The multi-agent technology is applied for the strategy implementation; the criteria based on wide area measured apparent impedances are defined to choose the control strategy, such as tap changer adjusting or load...
Number-to-voltage converter on commutated condensers
International Nuclear Information System (INIS)
Grekhov, Yu.N.
1975-01-01
A code-voltage converter using precision voltage dividers based on commutated capacitors [1] is described which is distinguished by the absence of precision elements. Each digit includes eight field-effect transistors in two 1KT682 microcircuit assemblies and three microcapacitors with a conventional unstable capacitance 6200 pF +- 50%. The converter has a speed of response that is not inferior to that of converters based on R-2R matrices, while in time stability of the characteristics, low interference level, and low output impedance it is superior to such converters
OFCC based voltage and transadmittance mode instrumentation amplifier
Nand, Deva; Pandey, Neeta; Pandey, Rajeshwari; Tripathi, Prateek; Gola, Prashant
2017-07-01
The operational floating current conveyor (OFCC) is a versatile active block due to the availability of both low and high input and output impedance terminals. This paper addresses the realization of OFCC based voltage and transadmittance mode instrumentation amplifiers (VMIA and TAM IA). It employs three OFCCs and seven resistors. The transadmittance mode operation can easily be obtained by simply connecting an OFCC based voltage to current converter at the output. The effect of non-idealities of OFCC, in particular finite transimpedance and tracking error, on system performance is also dealt with and corresponding mathematical expressions are derived. The functional verification is performed through SPICE simulation using CMOS based implementation of OFCC.
Investigation of body's impedance under different conditions for electro-acupuncture
International Nuclear Information System (INIS)
Ahmed, M.M.; Abrarov, S.; Khan, R.R.; Maqsood, R.S.; Qaiser, M.A.; Karimov, Kh. S.
2001-01-01
A computer controlled automated setup has been designed to investigate the body acupuncture points (bio-active points) by using a probes matrix which exerts a uniform pressure on the body. 16 probes matrix was placed in a 15 : 15 mm/sup 2/ dielectric substrate with 5 mm inter probe distance, compatible with the average diameter of the points. These probes have been designed to facilitate a semiconductor injection laser for probing of the points along with optical and/or electric signal. The bioactive points were identified by evaluating the impedance between each probe and a hand held electrode through a micro-controlled scan. This also allowed the selection of an appropriate signal - DC, AC or tidal waveform, for the electric treatment of bioactive points. It has been found that body impedance decreases with the increase of measuring voltage. Moreover, for current-voltage characteristics a nonlinearity coefficient in the range 2-3 was also observed. The study revealed that at low applied voltages 0.l V, the impedance depends on the polarity of the applied signal. Furthermore, body impedance decreases nonlinearly by increasing the probe's pressure on the skin, which may be attributed to piezo resistive effect. By using the AC and Dc measurements an appropriate body equivalent circuit is proposed in this investigation. (author)
High voltage engineering fundamentals
Kuffel, E; Hammond, P
1984-01-01
Provides a comprehensive treatment of high voltage engineering fundamentals at the introductory and intermediate levels. It covers: techniques used for generation and measurement of high direct, alternating and surge voltages for general application in industrial testing and selected special examples found in basic research; analytical and numerical calculation of electrostatic fields in simple practical insulation system; basic ionisation and decay processes in gases and breakdown mechanisms of gaseous, liquid and solid dielectrics; partial discharges and modern discharge detectors; and over
DEFF Research Database (Denmark)
Chen, Yandong; Guerrero, Josep M.; Shuai, Zhikang
2016-01-01
In this paper, an inverter using resistivecapacitive output impedance (RC-type inverter) is proposed not only to provide fast reactive power sharing to support microgrid voltage, and but also to reduce circulating currents and damp high-frequency resonances among inverters. Introducing the RC......-frequency resonances among parallel inverters are quantitatively analyzed. The control parameters are systematically selected, and effect of virtual complex impedance on the inverter output voltage is depicted. The RC-type inverter can reduce circulating currents and damp resonances due to different equivalent output...
International Nuclear Information System (INIS)
Glyavin, M. Yu.; Zavolskiy, N. A.; Sedov, A. S.; Nusinovich, G. S.
2013-01-01
For a long time, the gyrotrons were primarily developed for electron cyclotron heating and current drive of plasmas in controlled fusion reactors where a multi-megawatt, quasi-continuous millimeter-wave power is required. In addition to this important application, there are other applications (and their number increases with time) which do not require a very high power level, but such issues as the ability to operate at low voltages and have compact devices are very important. For example, gyrotrons are of interest for a dynamic nuclear polarization, which improves the sensitivity of the nuclear magnetic resonance spectroscopy. In this paper, some issues important for operation of gyrotrons driven by low-voltage electron beams are analyzed. An emphasis is made on the efficiency of low-voltage gyrotron operation at the fundamental and higher cyclotron harmonics. These efficiencies calculated with the account for ohmic losses were, first, determined in the framework of the generalized gyrotron theory based on the cold-cavity approximation. Then, more accurate, self-consistent calculations for the fundamental and second harmonic low-voltage sub-THz gyrotron designs were carried out. Results of these calculations are presented and discussed. It is shown that operation of the fundamental and second harmonic gyrotrons with noticeable efficiencies is possible even at voltages as low as 5–10 kV. Even the third harmonic gyrotrons can operate at voltages about 15 kV, albeit with rather low efficiency (1%–2% in the submillimeter wavelength region).
Surge Protection in Low-Voltage AC Power Circuits: An Anthology
Martzloff, F. D.
2002-10-01
The papers included in this part of the Anthology provide basic information on the propagation of surges in low-voltage AC power circuits. The subject was approached by a combination of experiments and theoretical considerations. One important distinction is made between voltage surges and current surges. Historically, voltage surges were the initial concern. After the introduction and widespread use of current-diverting surge-protective devices at the point-of-use, the propagation of current surges became a significant factor. The papers included in this part reflect this dual dichotomy of voltage versus current and impedance mismatch effects versus simple circuit theory.
Manohar, Aswin K; Bretschger, Orianna; Nealson, Kenneth H; Mansfeld, Florian
2008-04-01
Electrochemical impedance spectroscopy (EIS) has been used to determine several electrochemical properties of the anode and cathode of a mediator-less microbial fuel cell (MFC) under different operational conditions. These operational conditions included a system with and without the bacterial catalyst and EIS measurements at the open-circuit potential of the anode and the cathode or at an applied cell voltage. In all cases the impedance spectra followed a simple one-time-constant model (OTCM) in which the solution resistance is in series with a parallel combination of the polarization resistance and the electrode capacitance. Analysis of the impedance spectra showed that addition of Shewanella oneidensis MR-1 to a solution of buffer and lactate greatly increased the rate of the lactate oxidation at the anode under open-circuit conditions. The large decrease of open-circuit potential of the anode increased the cell voltage of the MFC and its power output. Measurements of impedance spectra for the MFC at different cell voltages resulted in determining the internal resistance (R(int)) of the MFC and it was found that R(int) is a function of cell voltage. Additionally, R(int) was equal to R(ext) at the cell voltage corresponding to maximum power, where R(ext) is the external resistance that must be applied across the circuit to obtain the maximum power output.
Observations involving broadband impedance modelling
International Nuclear Information System (INIS)
Berg, J.S.
1995-08-01
Results for single- and multi-bunch instabilities can be significantly affected by the precise model that is used for the broadband impendance. This paper discusses three aspects of broadband impendance modeling. The first is an observation of the effect that a seemingly minor change in an impedance model has on the single-bunch mode coupling threshold. The second is a successful attempt to construct a model for the high-frequency tails of an r.f cavity. The last is a discussion of requirements for the mathematical form of an impendance which follow from the general properties of impendances
Programmable differential capacitance-to-voltage converter for MEMS accelerometers
Royo, G.; Sánchez-Azqueta, C.; Gimeno, C.; Aldea, C.; Celma, S.
2017-05-01
Capacitive MEMS sensors exhibit an excellent noise performance, high sensitivity and low power consumption. They offer a huge range of applications, being the accelerometer one of its main uses. In this work, we present the design of a capacitance-to-voltage converter in CMOS technology to measure the acceleration from the capacitance variations. It is based on a low-power, fully-differential transimpedance amplifier with low input impedance and a very low input noise.
Impedance studies of 2D azimuthally symmetric devices of finite length
Biancacci, N; Métral, E; Salvant, B; Migliorati, M; Palumbo, L
2014-01-01
In particle accelerators, the beam quality can be strongly affected by the interaction with self-induced electromagnetic fields excited by the beam in the passage through the elements of the accelerator. The beam coupling impedance quantifies this interaction and allows predicting the stability of the dynamics of high intensity, high brilliance beams. The coupling impedance can be evaluated with finite element methods or using analytical approaches, such as field matching or mode matching. In this paper we present an application of the mode matching technique for an azimuthally uniform structure of finite length: a cylindrical cavity loaded with a toroidal slab of lossy dielectric, connected with cylindrical beam pipes. In order to take into account the finite length of the structure, with respect to the infinite length approximation, we decompose the fields in the cavity into a set of orthonormal modes. We obtain a complete set of equations using the magnetic field matching and the nonuniform convergence of ...
Device for monitoring cell voltage
Doepke, Matthias [Garbsen, DE; Eisermann, Henning [Edermissen, DE
2012-08-21
A device for monitoring a rechargeable battery having a number of electrically connected cells includes at least one current interruption switch for interrupting current flowing through at least one associated cell and a plurality of monitoring units for detecting cell voltage. Each monitoring unit is associated with a single cell and includes a reference voltage unit for producing a defined reference threshold voltage and a voltage comparison unit for comparing the reference threshold voltage with a partial cell voltage of the associated cell. The reference voltage unit is electrically supplied from the cell voltage of the associated cell. The voltage comparison unit is coupled to the at least one current interruption switch for interrupting the current of at least the current flowing through the associated cell, with a defined minimum difference between the reference threshold voltage and the partial cell voltage.
DEFF Research Database (Denmark)
Wei, Baoze; Guerrero, Josep M.; Quintero, Juan Carlos Vasquez
2016-01-01
This paper describes a theoretical with experiment study on a control strategy for the parallel operation of threephase voltage source inverters (VSI), to be applied to uninterruptible power systems (UPS). A circulating current suppression strategy for parallel VSIs is proposed in this paper based...... on circulating current control loops used to modify the reference currents by compensating the error currents among parallel inverters. Both of the cross and zero-sequence circulating currents are considered. The proposed method is coordinated together with droop and virtual impedance control. In this paper......, droop control is used to generate the reference voltage of each inverter, and the virtual impedance is used to fix the output impedance of the inverters. In addition, a secondary control is used in order to recover the voltage deviation caused by the virtual impedance. And the auxiliary current control...
International Nuclear Information System (INIS)
Bertram, F.; Evertsson, J.; Messing, M. E.; Mikkelsen, A.; Lundgren, E.; Zhang, F.; Pan, J.; Carlà, F.; Nilsson, J.-O.
2014-01-01
We present results from the anodization of an aluminum single crystal [Al(111)] and an aluminum alloy [Al 6060] studied by in situ x-ray reflectivity, in situ electrochemical impedance spectroscopy and ex situ scanning electron microscopy. For both samples, a linear increase of oxide film thickness with increasing anodization voltage was found. However, the slope is much higher in the single crystal case, and the break-up of the oxide film grown on the alloy occurs at a lower anodization potential than on the single crystal. The reasons for these observations are discussed as are the measured differences observed for x-ray reflectivity and electrochemical impedance spectroscopy.
Pumping slots: impedances and power losses
Energy Technology Data Exchange (ETDEWEB)
Kurennoy, S [Maryland Univ., College Park, MD (United States). Dept. of Physics
1996-08-01
Contributions of pumping slots to the beam coupling impedances and power losses in a B-factory ring are considered. While their leading contribution is to the inductive impedance, for high-intensity machines with short bunches like e{sup +}e{sup -} B-factories the real part of the impedance and related loss factors are also important. Using an analytical approach we calculate the coupling impedances and loss factors due to slots in a ring with an arbitrary cross section of the vacuum chamber. Effects of the slot tilt on the beam impedance are also considered, and restrictions on the tilt angle are derived from limitations on the impedance increase. The power leakage through the slots is discussed briefly. The results are applied to the KEK B-factory. (author)
Mehling, Joshua S.; Holley, James; O'Malley, Marcia K.
2015-01-01
The fidelity with which series elastic actuators (SEAs) render desired impedances is important. Numerous approaches to SEA impedance control have been developed under the premise that high-precision actuator torque control is a prerequisite. Indeed, the design of an inner torque compensator has a significant impact on actuator impedance rendering. The disturbance observer (DOB) based torque control implemented in NASA's Valkyrie robot is considered here and a mathematical model of this torque control, cascaded with an outer impedance compensator, is constructed. While previous work has examined the impact a disturbance observer has on torque control performance, little has been done regarding DOBs and impedance rendering accuracy. Both simulation and a series of experiments are used to demonstrate the significant improvements possible in an SEA's ability to render desired dynamic behaviors when utilizing a DOB. Actuator transparency at low impedances is improved, closed loop hysteresis is reduced, and the actuator's dynamic response to both commands and interaction torques more faithfully matches that of the desired model. All of this is achieved by leveraging DOB based control rather than increasing compensator gains, thus making improved SEA impedance control easier to achieve in practice.
High frequency breakdown voltage
International Nuclear Information System (INIS)
Chu, Thanh Duy.
1992-03-01
This report contains information about the effect of frequency on the breakdown voltage of an air gap at standard pressure and temperature, 76 mm Hg and O degrees C, respectively. The frequencies of interest are 47 MHz and 60 MHz. Additionally, the breakdown in vacuum is briefly considered. The breakdown mechanism is explained on the basis of collision and ionization. The presence of the positive ions produced by ionization enhances the field in the gap, and thus determines the breakdown. When a low-frequency voltage is applied across the gap, the breakdown mechanism is the same as that caused by the DC or static voltage. However, when the frequency exceeds the first critical value f c , the positive ions are trapped in the gap, increasing the field considerably. This makes the breakdown occur earlier; in other words, the breakdown voltage is lowered. As the frequency increases two decades or more, the second critical frequency, f ce , is reached. This time the electrons start being trapped in the gap. Those electrons that travel multiple times across the gap before reaching the positive electrode result in an enormous number of electrons and positive ions being present in the gap. The result is a further decrease of the breakdown voltage. However, increasing the frequency does not decrease the breakdown voltage correspondingly. In fact, the associated breakdown field intensity is almost constant (about 29 kV/cm).The reason is that the recombination rate increases and counterbalances the production rate, thus reducing the effect of the positive ions' concentration in the gap. The theory of collision and ionization does not apply to the breakdown in vacuum. It seems that the breakdown in vacuum is primarily determined by the irregularities on the surfaces of the electrodes. Therefore, the effect of frequency on the breakdown, if any, is of secondary importance
An integrated microsystem device with matched interdigitated microelectrode chip was fabricated for enrichment and detection of Escherichia coli O157:H7. The microsystem has integrated with positive dielectrophoresis (pDEP) enrichment and in situ impedance detection, whose total volume is only 3.0 ×...
Liu, Hai-Tao; Wen, Zhi-Yu; Xu, Yi; Shang, Zheng-Guo; Peng, Jin-Lan; Tian, Peng
2017-09-01
In this paper, an integrated microfluidic analysis microsystems with bacterial capture enrichment and in-situ impedance detection was purposed based on microfluidic chips dielectrophoresis technique and electrochemical impedance detection principle. The microsystems include microfluidic chip, main control module, and drive and control module, and signal detection and processing modulet and result display unit. The main control module produce the work sequence of impedance detection system parts and achieve data communication functions, the drive and control circuit generate AC signal which amplitude and frequency adjustable, and it was applied on the foodborne pathogens impedance analysis microsystems to realize the capture enrichment and impedance detection. The signal detection and processing circuit translate the current signal into impendence of bacteria, and transfer to computer, the last detection result is displayed on the computer. The experiment sample was prepared by adding Escherichia coli standard sample into chicken sample solution, and the samples were tested on the dielectrophoresis chip capture enrichment and in-situ impedance detection microsystems with micro-array electrode microfluidic chips. The experiments show that the Escherichia coli detection limit of microsystems is 5 × 104 CFU/mL and the detection time is within 6 min in the optimization of voltage detection 10 V and detection frequency 500 KHz operating conditions. The integrated microfluidic analysis microsystems laid the solid foundation for rapid real-time in-situ detection of bacteria.
Determination of salt content in various depth of pork chop by electrical impedance spectroscopy
International Nuclear Information System (INIS)
Kaltenecker, P; Szöllösi, D; Vozáry, E; Friedrich, L
2013-01-01
The salt concentration was determined inside of pork chop both by electrical impedance spectroscopy and by a conventional chemical method (according to Mohr). The pork chop in various depths (4 mm, 10 mm, 20 mm and 25 mm) was punctured with two stainless steel electrodes. The length of electrodes was 60 mm, and they were insulated along the length except 1 cm section on the end, so the measurement of impedance was realized in various depths. The magnitude and phase angle of impedance were measured with a HP 4284A and a HP 4285A LCR meters from 30 Hz up to 1 MHz and from 75 kHz up to 30 MHz frequency range, respectively at 1 V voltage. The distance between the electrodes was 1 cm. The impedance magnitude decreased as the salt concentration increased. The magnitude of open-short corrected impedance values at various frequencies (10 kHz, 100 kHz, 125 kHz, 1.1 MHz and 8 MHz) showed a good correlation with salt content determined by chemical procedure. The electrical impedance spectroscopy seems a prospective method for determination the salt concentration inside the meat in various depths during the curing procedure.
International Nuclear Information System (INIS)
Hu Yixiang; Qiu Aici; Sun Fengju; Huang Tao; Cong Peitian; Wang Liangping; Zeng Jiangtao; Li Yan; Zhang Xinjun; Lei Tianshi
2011-01-01
Based on the transmission line code TLCODE, a 1D circuit model for a transmission-line impedance transformer was developed and the simulation results were compared with those in the literature. The model was used to quantify the efficiencies of voltage-transport, energy-transport and power-transport for a transmission-line impedance transformer as functions of ψ (the ratio of the output impedance to the input impedance of the transformer) and Γ (the ratio of the pulse width to the one-way transit time of the transformer) under a large scale of m (the coefficient of the generalized exponential impedance profile). Simulation results suggest that with the increase in Γ, from 0 to ∞, the power transport efficiency first increases and then decreases. The maximum power transport efficiency can reach 90% or even higher for an exponential impedance profile (m = 1). With a consideration of dissipative loss in the dielectric and electrodes of the transformer, two representative designs of the water-insulated transformer are investigated for the next generation of petawatt-class z-pinch drivers. It is found that the dissipative losses in the electrodes are negligibly small, below 0.1%, but the dissipative loss in the water dielectric is about 1% to 4%. (fusion engineering)
Stress-Dependent Voltage Offsets From Polymer Insulators Used in Rock Mechanics and Material Testing
Carlson, G. G.; Dahlgren, Robert; Gray, Amber; Vanderbilt, V. C.; Freund, F.; Johnston, M. J.; Dunson, C.
2013-01-01
Dielectric insulators are used in a variety of laboratory settings when performing experiments in rock mechanics, petrology, and electromagnetic studies of rocks in the fields of geophysics,material science, and civil engineering. These components may be used to electrically isolate geological samples from the experimental equipment, to perform a mechanical compliance function between brittle samples and the loading equipment, to match ultrasonic transducers, or perform other functions. In manyexperimental configurations the insulators bear the full brunt of force applied to the sample but do not need to withstand high voltages, therefore the insulators are often thin sheets of mechanically tough polymers. From an instrument perspective, transduction from various types of mechanical perturbation has beenqualitatively compared for a number of polymers [1, 2] and these error sources are readily apparent duringhigh-impedance measurements if not mitigated. However even when following best practices, a force dependent voltage signal still remains and its behavior is explored in this presentation. In this experimenttwo thin sheets (0.25 mm) of high-density polyethylene (HDPE) were set up in a stack, held alternatelybetween three aluminum bars; this stack was placed on the platen of a 60T capacity hydraulic testingmachine. The surface area, A, over which the force is applied to the PE sheets in this sandwich is roughly 40 square cm, each sheet forming a parallel-plate capacitor having roughly 320 pF [3], assuming therelative dielectric permittivity of PE is approximately 2.3. The outer two aluminum bars were connected to the LO input ofthe electrometer and the central aluminum bar was connected to the HI input of a Keithley model 617 electrometer. Once the stack is mechanically well-seated with no air gaps, the voltage offset is observed tobe a linear function of the baseline voltage for a given change in applied force. For a periodically appliedforce of 66.7 kN the
Optimal Packed String Matching
DEFF Research Database (Denmark)
Ben-Kiki, Oren; Bille, Philip; Breslauer, Dany
2011-01-01
In the packed string matching problem, each machine word accommodates – characters, thus an n-character text occupies n/– memory words. We extend the Crochemore-Perrin constantspace O(n)-time string matching algorithm to run in optimal O(n/–) time and even in real-time, achieving a factor – speed...
Ontology Matching Across Domains
2010-05-01
matching include GMO [1], Anchor-Prompt [2], and Similarity Flooding [3]. GMO is an iterative structural matcher, which uses RDF bipartite graphs to...AFRL under contract# FA8750-09-C-0058. References [1] Hu, W., Jian, N., Qu, Y., Wang, Y., “ GMO : a graph matching for ontologies”, in: Proceedings of
Directory of Open Access Journals (Sweden)
Miska Prasad
2016-12-01
Full Text Available The voltage events namely voltage sags and voltage swells represent the most common, frequent and important power quality events in today’s power system. Dynamic voltage restorer (DVR is one of the key components used to mitigate the supply voltage quality disturbances in terms of voltage sags and swells in the distribution system. It consists of an energy storage unit, a voltage source inverter, a filter, a coupling transformer and the control system. This paper presents three different inverter configurations of dynamic voltage restorer (DVR for mitigation of voltage events such as voltage sags and swells with sudden addition or removal of the nonlinear load. These three configurations are voltage source inverter based DVR (VSI-DVR, current source inverter based DVR (CSI-DVR and impedance or Z-source inverter based DVR (ZSI-DVR. The d-q control technique is used to control the operation of the DVR. The response of ZSI-DVR for mitigation of voltage sags and swells are investigated and compared with VSI-DVR and CSI-DVR using MATLAB/SIMULINK environment.
A compact broadband nonsynchronous noncommensurate impedance transformer
DEFF Research Database (Denmark)
Zhurbenko, Vitaliy; Kim, Kseniya; Narenda, Kumar
2012-01-01
Nonsynchronous noncommensurate impedance transformers consist of a combination of high‐ and low‐impedance transmission lines. High‐impedance lines have narrow tracks in strip and microstrip technology, which allows for high flexibility and miniaturization of the layout in comparison...... to the traditional tapered line transformers. This flexibility of the broadband nonsynchronous noncommensurate impedance transformers is experimentally demonstrated in this article allowing the length reduction by almost three times. © 2012 Wiley Periodicals, Inc. Microwave Opt Technol Lett 54:1832–1835, 2012; View...
Electrical impedance tomography of electrolysis.
Directory of Open Access Journals (Sweden)
Arie Meir
Full Text Available The primary goal of this study is to explore the hypothesis that changes in pH during electrolysis can be detected with Electrical Impedance Tomography (EIT. The study has relevance to real time control of minimally invasive surgery with electrolytic ablation. To investigate the hypothesis, we compare EIT reconstructed images to optical images acquired using pH-sensitive dyes embedded in a physiological saline agar gel phantom treated with electrolysis. We further demonstrate the biological relevance of our work using a bacterial E.Coli model, grown on the phantom. The results demonstrate the ability of EIT to image pH changes in a physiological saline phantom and show that these changes correlate with cell death in the E.coli model. The results are promising, and invite further experimental explorations.
International Nuclear Information System (INIS)
Fatehmulla, Amanullah; Farooq, W. A.; Aslam, M.; Atif, M.; Ali, S.M.; Al-Dhafir, A. M.; Yakuphanoglu, F.; Yahia, I.S.
2014-01-01
Cadmium Sulphide (CdS) quantum dots (QDs) were deposited on nanostructured TiO 2 film using a modified Successive Ionic Layer Adsorption and Reaction (SILAR) method. Nanostructured TiO 2 on FTO glass and Platinum on FTO are used as photoelectrode and Counter electrode respectively. High resolution Transmission Electron Microscopy (HRT EM) image revealed CdS QDs adsorbed on nanostructured TiO 2 . The photovoltaic characteristics and impedance spectroscopy properties of CdS quantum dot sensitized solar cell (QDSSC) were analyzed under air mass 1.5 illuminations. At the SILAR adsorption time of 2 min (10 cycles), the QDSSC measured a short circuit current density of 2 mA/cm 2 and an open circuit voltage of 0.45 V under air mass 1.5. In a widespread frequency range, the capacitance – voltage, the conductance – voltage, the series resistance - voltage measurements were carried out for the QDSSC applications. A conduct of positive to negative capacitance was observed from the measured characteristics of capacitance - voltage which is attributed to the injection of electrons from FTO electrode into TiO 2 . Key words: Nanostructured TiO 2 , CdS QDSSC, SILAR method, photovoltaic measurements, impedance characteristic
International Nuclear Information System (INIS)
Zhou Zhicheng
1992-01-01
A digital voltage discriminator is described, which is synthesized by digital comparator and ADC. The threshold is program controllable with high stability. Digital region of confusion is approximately equal to 1.5 LSB. This discriminator has a single channel analyzer function model with channel width of 1.5 LSB
Energy Technology Data Exchange (ETDEWEB)
Bugl, Andrea; Ball, Markus; Boehmer, Michael; Doerheim, Sverre; Hoenle, Andreas; Konorov, Igor [Technische Universitaet Muenchen, Garching (Germany); Ketzer, Bernhard [Technische Universitaet Muenchen, Garching (Germany); Helmholtz-Institut fuer Strahlen- und Kernphysik, Bonn (Germany)
2014-07-01
Current measurements in the nano- and picoampere region on high voltage are an important tool to understand charge transfer processes in micropattern gas detectors like the Gas Electron Multiplier (GEM). They are currently used to e.g. optimize the field configuration in a multi-GEM stack to be used in the ALICE TPC after the upgrade of the experiment during the 2nd long shutdown of the LHC. Devices which allow measurements down to 1pA at high voltage up to 6 kV have been developed at TU Muenchen. They are based on analog current measurements via the voltage drop over a switchable shunt. A microcontroller collects 128 digital ADC values and calculates their mean and standard deviation. This information is sent with a wireless transmitting unit to a computer and stored in a root file. A nearly unlimited number of devices can be operated simultaneously and read out by a single receiver. The results can also be displayed on a LCD directly at the device. Battery operation and the wireless readout are important to protect the user from any contact to high voltage. The principle of the device is explained, and systematic studies of their properties are shown.
Geomagnetism and Induced Voltage
Abdul-Razzaq, W.; Biller, R. D.
2010-01-01
Introductory physics laboratories have seen an influx of "conceptual integrated science" over time in their classrooms with elements of other sciences such as chemistry, biology, Earth science, and astronomy. We describe a laboratory to introduce this development, as it attracts attention to the voltage induced in the human brain as it…
DEFF Research Database (Denmark)
Mustafa, Ghullam; Bak-Jensen, Birgitte; Mahat, Pukar
2013-01-01
Any problem with voltage in a power network is undesirable as it aggravates the quality of the power. Power electronic devices such as Voltage Source Converter (VSC) based Static Synchronous Compensator (STATCOM) etc. can be used to mitigate the voltage problems in the distribution system...... to unbalanced faults. The compensation of unbalanced voltage sags and voltage unbalance in the CIGRE distribution network is done by using the four STATCOM compensators already existing in the test grid. The simulations are carried out in DIgSILENT power factory software version 15.0........ The voltage problems dealt with in this paper are to show how to mitigate unbalanced voltage sags and voltage unbalance in the CIGRE Low Voltage (LV) test network and net-works like this. The voltage unbalances, for the tested cases in the CIGRE LV test network are mainly due to single phase loads and due...
Realization of Nth-Order Voltage Transfer Function using Current Conveyors CCII
Directory of Open Access Journals (Sweden)
K. Vrba
1997-06-01
Full Text Available A universal method for the realization of arbitrary voltage transfer function in canonic form is presented. A voltage-controlled current-source using a plus-type second-generation current conveyor is here applied as the basic building element. Filters designed according to this method have a high input impedance and low sensitivity to variations of circuit parameters. All passive elements are grounded.
DEFF Research Database (Denmark)
Forouzesh, Mojtaba; Abdelhakim, Ahmed; Siwakoti, Yam
2018-01-01
Magnetically coupled impedance source (MCIS) converters are prone to high voltage spikes across the inverter bridge (or dc-link) due to the presence of leakage and stray inductances in the high frequency loop. The problem manifolds because of a shoot-through state in impedance source converters......, but the solutions are not generic (i.e. structure-oriented) and they are quite lossy with intuitive modification in the circuit itself, resulting in significant changes in the performance of the power converter (e.g. increase in components stresses). To address this concern, a general passive regenerative inductor......-capacitor-diode (L-C-D) snubber is presented in this paper for all MCIS converters without any modification in the original circuit. The proposed circuit rechannel the leakage energy of the coupled magnetics and feedback it to input or network itself, which does not only avoid extreme voltage spikes across...
Valente, Virgilio; Dai Jiang; Demosthenous, Andreas
2015-08-01
This paper presents the preliminary design and simulation of a flexible and programmable analog front-end (AFE) circuit with current and voltage readout capabilities for electric impedance spectroscopy (EIS). The AFE is part of a fully integrated multifrequency EIS platform. The current readout comprises of a transimpedance stage and an automatic gain control (AGC) unit designed to accommodate impedance changes larger than 3 order of magnitude. The AGC is based on a dynamic peak detector that tracks changes in the input current over time and regulates the gain of a programmable gain amplifier in order to optimise the signal-to-noise ratio. The system works up to 1 MHz. The voltage readout consists of a 2 stages of fully differential current-feedback instrumentation amplifier which provide 100 dB of CMRR and a programmable gain up to 20 V/V per stage with a bandwidth in excess of 10MHz.
Ansory, Achmad; Prajitno, Prawito; Wijaya, Sastra Kusuma
2018-02-01
Electrical Impedance Tomography (EIT) is an imaging method that is able to estimate electrical impedance distribution inside an object. This EIT system is developed by using 32 electrodes and microcontroller based module. From a pair of electrodes, sinusoidal current of 3 mA is injected and the voltage differences between other pairs of electrodes are measured. Voltage measurement data are then sent to MATLAB and EIDORS software; the data are used to reconstruct two dimensions image. The system can detect and determine the position of a phantom in the tank. The object's position is accurately reconstructed and determined with the average shifting of 0.69 cm but object's area cannot be accurately reconstructed. The object's image is more accurately reconstructed when the object is located near to electrodes, has a larger size, and when the current injected to the system has a frequency of 100 kHz or 200kHz.
Directory of Open Access Journals (Sweden)
Schullcke Benjamin
2016-09-01
Full Text Available Electrical impedance tomography (EIT is used to monitor the regional distribution of ventilation in a transversal plane of the thorax. In this manuscript we evaluate the impact of different quantities of electrodes used for current injection and voltage measurement on the reconstructed shape of the lungs. Results indicate that the shape of reconstructed impedance changes in the body depends on the number of electrodes. In this manuscript, we demonstrate that a higher number of electrodes do not necessarily increase the image quality. For the used stimulation pattern, utilizing neighboring electrodes for current injection and voltage measurement, we conclude that the shape of the lungs is best reconstructed if 16 electrodes are used.
CMOS Silicon-on-Sapphire RF Tunable Matching Networks
Directory of Open Access Journals (Sweden)
Chamseddine Ahmad
2006-01-01
Full Text Available This paper describes the design and optimization of an RF tunable network capable of matching highly mismatched loads to 50 at 1.9 GHz. Tuning was achieved using switched capacitors with low-loss, single-transistor switches. Simulations show that the performance of the matching network depends strongly on the switch performances and on the inductor losses. A 0.5 m silicon-on-sapphire (SOS CMOS technology was chosen for network implementation because of the relatively high-quality monolithic inductors achievable in the process. The matching network provides very good matching for inductive loads, and acceptable matching for highly capacitive loads. A 1 dB compression point greater than dBm was obtained for a wide range of load impedances.
Method, system and computer-readable media for measuring impedance of an energy storage device
Morrison, John L.; Morrison, William H.; Christophersen, Jon P.; Motloch, Chester G.
2016-01-26
Real-time battery impedance spectrum is acquired using a one-time record. Fast Summation Transformation (FST) is a parallel method of acquiring a real-time battery impedance spectrum using a one-time record that enables battery diagnostics. An excitation current to a battery is a sum of equal amplitude sine waves of frequencies that are octave harmonics spread over a range of interest. A sample frequency is also octave and harmonically related to all frequencies in the sum. A time profile of this sampled signal has a duration that is a few periods of the lowest frequency. A voltage response of the battery, average deleted, is an impedance of the battery in a time domain. Since the excitation frequencies are known and octave and harmonically related, a simple algorithm, FST, processes the time profile by rectifying relative to sine and cosine of each frequency. Another algorithm yields real and imaginary components for each frequency.
International Nuclear Information System (INIS)
Leinonen, Matti; Hakula, Harri; Hyvönen, Nuutti
2014-01-01
The aim of electrical impedance tomography is to determine the internal conductivity distribution of some physical body from boundary measurements of current and voltage. The most accurate forward model for impedance tomography is the complete electrode model, which consists of the conductivity equation coupled with boundary conditions that take into account the electrode shapes and the contact resistances at the corresponding interfaces. If the reconstruction task of impedance tomography is recast as a Bayesian inference problem, it is essential to be able to solve the complete electrode model forward problem with the conductivity and the contact resistances treated as a random field and random variables, respectively. In this work, we apply a stochastic Galerkin finite element method to the ensuing elliptic stochastic boundary value problem and compare the results with Monte Carlo simulations
Mitigation of Voltage Sags in CIGRE Low Voltage Distribution Network
DEFF Research Database (Denmark)
Mustafa, Ghullam; Bak-Jensen, Birgitte; Mahat, Pukar
2013-01-01
Any problem in voltage in a power network is undesirable as it aggravates the quality of the power. Power electronic devices such as Voltage Source Converter (VSC) based Static Synchronous Compensator (STATCOM), Dynamic Voltage Restorer (DVR) etc. are commonly used for the mitigation of voltage p....... The compensation of voltage sags in the different parts of CIGRE distribution network is done by using the four STATCOM compensators already existing in the test grid. The simulations are carried out in DIgSILENT power factory software version 15.0.......Any problem in voltage in a power network is undesirable as it aggravates the quality of the power. Power electronic devices such as Voltage Source Converter (VSC) based Static Synchronous Compensator (STATCOM), Dynamic Voltage Restorer (DVR) etc. are commonly used for the mitigation of voltage...... problems in the distribution system. The voltage problems dealt with in this paper are to show how to mitigate voltage sags in the CIGRE Low Voltage (LV) test network and networks like this. The voltage sags, for the tested cases in the CIGRE LV test network are mainly due to three phase faults...
rf impedance of the accelerating beam gap and its significance to the TRIUMF rf system
International Nuclear Information System (INIS)
Poirier, R.
1979-03-01
The rf system at TRIUMF is now operating with the highest Q, the lowest rf leakage into the beam gap, the best voltage stability, and the lowest resonator strongback temperatures ever measured since it was first put into operation. This paper describes the calculation of the rf impedance of the beam gap and its correlation to the rf problems encountered, which eventually led to modifications to the flux guides and resonator tips to accomplish the improved operation of the rf system
Frequency-Division Multiplexing for Electrical Impedance Tomography in Biomedical Applications
Directory of Open Access Journals (Sweden)
Yair Granot
2007-01-01
Full Text Available Electrical impedance tomography (EIT produces an image of the electrical impedance distribution of tissues in the body, using electrodes that are placed on the periphery of the imaged area. These electrodes inject currents and measure voltages and from these data, the impedance can be computed. Traditional EIT systems usually inject current patterns in a serial manner which means that the impedance is computed from data collected at slightly different times. It is usually also a time-consuming process. In this paper, we propose a method for collecting data concurrently from all of the current patterns in biomedical applications of EIT. This is achieved by injecting current through all of the current injecting electrodes simultaneously, and measuring all of the resulting voltages at once. The signals from various current injecting electrodes are separated by injecting different frequencies through each electrode. This is called frequency-division multiplexing (FDM. At the voltage measurement electrodes, the voltage related to each current injecting electrode is isolated by using Fourier decomposition. In biomedical applications, using different frequencies has important implications due to dispersions as the tissue's electrical properties change with frequency. Another significant issue arises when we are recording data in a dynamic environment where the properties change very fast. This method allows simultaneous measurements of all the current patterns, which may be important in applications where the tissue changes occur in the same time scale as the measurement. We discuss the FDM EIT method from the biomedical point of view and show results obtained with a simple experimental system.
DEFF Research Database (Denmark)
Wang, Xiongfei; Blaabjerg, Frede; Chen, Zhe
2012-01-01
This paper proposes a harmonic impedance synthesis technique for voltage-controlled distributed generation inverters in order to damp harmonic voltage distortion on a distribution network. The approach employs a multiloop control scheme, where a selective harmonic load current feedforward loop...... at the dominant harmonic frequencies. Thus, the harmonic voltage drop on the grid-side inductance and the harmonic resonances throughout a distribution feeder with multiple shunt-connected capacitors can be effectively attenuated. Simulation and laboratory test results validate the performance of the proposed...
Quantification of coating aging using impedance measurements
Westing, E.P.M. van; Weijde, D.H. van der; Vreijling, M.P.W.; Ferrari, G.M.; Wit, J.H.W. de
1998-01-01
This chapter shows the application results of a novel approach to quantify the ageing of organic coatings using impedance measurements. The ageing quantification is based on the typical impedance behaviour of barrier coatings in immersion. This immersion behaviour is used to determine the limiting
High temperature impedance spectroscopy of barium stannate
Indian Academy of Sciences (India)
... differential thermal analysis, thermogravimetric analysis and Fourier transform infrared techniques. Electrical properties were studied using a.c. impedance spectroscopy technique in the temperature range of 50–650 °C and frequency range of 10 Hz–13 MHz. The complex impedance plots at temperature ≥ 300 °C show ...
Impedance of a slotted-pipe kicker
Energy Technology Data Exchange (ETDEWEB)
Zhou, Feng [Academia Sinica, Beijing, BJ (China). Inst. of High Energy Physics
1996-08-01
This paper introduces the principle of a new slotted kicker simply, which is made by using vacuum pipe itself with proper slits as current conductors, and then, presents a rough estimation of its longitudinal and transverse impedance, respectively. Calculation shows that its impedance is reduced significantly compared to our present air-coil kicker. (author)
impedance calculations of induction machine rotor conductors.
African Journals Online (AJOL)
Dr Obe
computed. The parallel R-L network shown in figure 3 is used in the modeling of the rotor bars. The network total impedance is given by,. (19). Where,. 5. simulation Results. MATLAB m-file for the calculation of the total impedance of the rectangular and trapezoidal rotor bars is developed [10]. The parameters of the bars.
Impedance of Surface Footings on Layered Ground
DEFF Research Database (Denmark)
Andersen, Lars; Clausen, Johan Christian
2005-01-01
is discussed. Based on the Green's function for a stratified half-space, the impedance of a surface footing with arbitrary shape is computed. A wind turbine foundation is analysed in the frequency range 0 to 3 Hz. Analyses show that soil stratification may lead to a significant changes in the impedance related...
Impedance of Surface Footings on Layered Ground
DEFF Research Database (Denmark)
Andersen, Lars; Clausen, Johan
2008-01-01
is discussed. Based on the Green's function for a stratified half-space, the impedance of a surface footing with arbitrary shape is computed. A wind turbine foundation is analysed in the frequency range 0-3 Hz. Analyses show that soil stratification may lead to significant changes in the impedance related...
Impedance interactions in bidirectional cascaded converter
DEFF Research Database (Denmark)
Tian, Yanjun; Loh, Poh Chiang; Chen, Zhe
2016-01-01
A cascaded converter is built by connecting one elementary converter to another. Output impedance of one converter will therefore interact with input impedance of the other converter. This interaction will change when power flow reverses. To compare this difference, an investigation is performed...
FDTD modeling of thin impedance sheets
Luebbers, Raymond J.; Kunz, Karl S.
1991-01-01
Thin sheets of resistive or dielectric material are commonly encountered in radar cross section calculations. Analysis of such sheets is simplified by using sheet impedances. In this paper it is shown that sheet impedances can be modeled easily and accurately using Finite Difference Time Domain (FDTD) methods.
Possibilities of electrical impedance tomography in gynecology
International Nuclear Information System (INIS)
Trokhanova O V; Chijova Y A; Okhapkin M B; Korjenevsky A V; Tuykin T S
2013-01-01
The paper describes results of comprehensive EIT diagnostics of mammary glands and cervix. The data were obtained from examinations of 170 patients by EIT system MEM (multi-frequency electrical impedance mammograph) and EIT system GIT (gynecological impedance tomograph). Mutual dependence is discussed.
Infographic explaining NCI-COG Pediatric MATCH, a cancer treatment clinical trial for children and adolescents, from 1 to 21 years of age, that is testing the use of precision medicine for pediatric cancers.
Data Matching Imputation System
National Oceanic and Atmospheric Administration, Department of Commerce — The DMIS dataset is a flat file record of the matching of several data set collections. Primarily it consists of VTRs, dealer records, Observer data in conjunction...
Impedance of finite length resistive cylinder
Directory of Open Access Journals (Sweden)
S. Krinsky
2004-11-01
Full Text Available We determine the impedance of a cylindrical metal tube (resistor of radius a, length g, and conductivity σ attached at each end to perfect conductors of semi-infinite length. Our main interest is in the asymptotic behavior of the impedance at high frequency (k≫1/a. In the equilibrium regime, ka^{2}≪g, the impedance per unit length is accurately described by the well-known result for an infinite length tube with conductivity σ. In the transient regime, ka^{2}≫g, where the contribution of transition radiation arising from the discontinuity in conductivity is important, we derive an analytic expression for the impedance and compute the short-range wakefield. The analytic results are shown to agree with numerical evaluation of the impedance.
Design and characterization of a high-power ultrasound driver with ultralow-output impedance
Lewis, George K.; Olbricht, William L.
2009-11-01
We describe a pocket-sized ultrasound driver with an ultralow-output impedance amplifier circuit (less than 0.05 Ω) that can transfer more than 99% of the voltage from a power supply to the ultrasound transducer with minimal reflections. The device produces high-power acoustical energy waves while operating at lower voltages than conventional ultrasound driving systems because energy losses owing to mismatched impedance are minimized. The peak performance of the driver is measured experimentally with a PZT-4, 1.54 MHz, piezoelectric ceramic, and modeled using an adjusted Mason model over a range of transducer resonant frequencies. The ultrasound driver can deliver a 100 Vpp (peak to peak) square-wave signal across 0-8 MHz ultrasound transducers in 5 ms bursts through continuous wave operation, producing acoustic powers exceeding 130 W. Effects of frequency, output impedance of the driver, and input impedance of the transducer on the maximum acoustic output power of piezoelectric transducers are examined. The small size, high power, and efficiency of the ultrasound driver make this technology useful for research, medical, and industrial ultrasonic applications.
Dynamic impedance model of the skin-electrode interface for transcutaneous electrical stimulation.
Directory of Open Access Journals (Sweden)
José Luis Vargas Luna
Full Text Available Transcutaneous electrical stimulation can depolarize nerve or muscle cells applying impulses through electrodes attached on the skin. For these applications, the electrode-skin impedance is an important factor which influences effectiveness. Various models describe the interface using constant or current-depending resistive-capacitive equivalent circuit. Here, we develop a dynamic impedance model valid for a wide range stimulation intensities. The model considers electroporation and charge-dependent effects to describe the impedance variation, which allows to describe high-charge pulses. The parameters were adjusted based on rectangular, biphasic stimulation pulses generated by a stimulator, providing optionally current or voltage-controlled impulses, and applied through electrodes of different sizes. Both control methods deliver a different electrical field to the tissue, which is constant throughout the impulse duration for current-controlled mode or have a very current peak for voltage-controlled. The results show a predominant dependence in the current intensity in the case of both stimulation techniques that allows to keep a simple model. A verification simulation using the proposed dynamic model shows coefficient of determination of around 0.99 in both stimulation types. The presented method for fitting electrode-skin impedance can be simple extended to other stimulation waveforms and electrode configuration. Therefore, it can be embedded in optimization algorithms for designing electrical stimulation applications even for pulses with high charges and high current spikes.
Lindén, J.; Lindberg, M.; Greggas, A.; Jylhävuori, N.; Norrgrann, H.; Lill, J. O.
2017-07-01
In addition to the main ingredients; sulfur, potassium chlorate and carbon, ordinary safety matches contain various dyes, glues etc, giving the head of the match an even texture and appealing color. Among the common reddish-brown matches there are several types, which after ignition can be attracted by a strong magnet. Before ignition the match head is generally not attracted by the magnet. An elemental analysis based on proton-induced x-ray emission was performed to single out iron as the element responsible for the observed magnetism. 57Fe Mössbauer spectroscopy was used for identifying the various types of iron-compounds, present before and after ignition, responsible for the macroscopic magnetism: Fe2O3 before and Fe3O4 after. The reaction was verified by mixing the main chemicals in the match-head with Fe2O3 in glue and mounting the mixture on a match stick. The ash residue after igniting the mixture was magnetic.
Applications of Electrical Impedance Tomography (EIT): A Short Review
Kanti Bera, Tushar
2018-03-01
Electrical Impedance Tomography (EIT) is a tomographic imaging method which solves an ill posed inverse problem using the boundary voltage-current data collected from the surface of the object under test. Though the spatial resolution is comparatively low compared to conventional tomographic imaging modalities, due to several advantages EIT has been studied for a number of applications such as medical imaging, material engineering, civil engineering, biotechnology, chemical engineering, MEMS and other fields of engineering and applied sciences. In this paper, the applications of EIT have been reviewed and presented as a short summary. The working principal, instrumentation and advantages are briefly discussed followed by a detail discussion on the applications of EIT technology in different areas of engineering, technology and applied sciences.
Assessment of breast tumor size in electrical impedance scanning
International Nuclear Information System (INIS)
Kim, Sungwhan
2012-01-01
Electrical impedance scanning (EIS) is a newly introduced imaging technique for early breast cancer detection. In EIS, we apply a sinusoidal voltage between a hand-held electrode and a scanning probe placed on the breast skin to make current travel through the breast. We measure induced currents (Neumann data) through the scanning probe. In this paper, we investigate the frequency-dependent behavior of the induced complex potential and show how the frequency differential of the current measurement on the scanning probe reflects the contrast in complex conductivity values between surrounding and cancerous tissues. Furthermore, we develop the formula for breast tumor size using the frequency differential of the current measurement and provide its feasibility. (paper)
Emira, Ahmed A.; Abdelghany, Mohamed A.; Elsayed, Mohannad Yomn; Elshurafa, Amro M; Salama, Khaled N.
2014-01-01
Various embodiments of a high voltage charge pump are described. One embodiment is a charge pump circuit that comprises a plurality of switching stages each including a clock input, a clock input inverse, a clock output, and a clock output inverse. The circuit further comprises a plurality of pumping capacitors, wherein one or more pumping capacitors are coupled to a corresponding switching stage. The circuit also comprises a maximum selection circuit coupled to a last switching stage among the plurality of switching stages, the maximum selection circuit configured to filter noise on the output clock and the output clock inverse of the last switching stage, the maximum selection circuit further configured to generate a DC output voltage based on the output clock and the output clock inverse of the last switching stage.
Emira, Ahmed A.
2014-10-09
Various embodiments of a high voltage charge pump are described. One embodiment is a charge pump circuit that comprises a plurality of switching stages each including a clock input, a clock input inverse, a clock output, and a clock output inverse. The circuit further comprises a plurality of pumping capacitors, wherein one or more pumping capacitors are coupled to a corresponding switching stage. The circuit also comprises a maximum selection circuit coupled to a last switching stage among the plurality of switching stages, the maximum selection circuit configured to filter noise on the output clock and the output clock inverse of the last switching stage, the maximum selection circuit further configured to generate a DC output voltage based on the output clock and the output clock inverse of the last switching stage.
Progress in Noise Thermometry at 505 K and 693 K Using Quantized Voltage Noise Ratio Spectra
Tew, W. L.; Benz, S. P.; Dresselhaus, P. D.; Coakley, K. J.; Rogalla, H.; White, D. R.; Labenski, J. R.
2010-09-01
Technical advances and new results in noise thermometry at temperatures near the tin freezing point and the zinc freezing point using a quantized voltage noise source (QVNS) are reported. The temperatures are derived by comparing the power spectral density of QVNS synthesized noise with that of Johnson noise from a known resistance at both 505 K and 693 K. Reference noise is digitally synthesized so that the average power spectra of the QVNS match those of the thermal noise, resulting in ratios of power spectra close to unity in the low-frequency limit. Three-parameter models are used to account for differences in impedance-related time constants in the spectra. Direct comparison of noise temperatures to the International Temperature Scale of 1990 (ITS-90) is achieved in a comparison furnace with standard platinum resistance thermometers. The observed noise temperatures determined by operating the noise thermometer in both absolute and relative modes, and related statistics together with estimated uncertainties are reported. The relative noise thermometry results are combined with results from other thermodynamic determinations at temperatures near the tin freezing point to calculate a value of T - T 90 = +4(18) mK for temperatures near the zinc freezing point. These latest results achieve a lower uncertainty than that of our earlier efforts. The present value of T - T 90 is compared to other published determinations from noise thermometry and other methods.
High Voltage Seismic Generator
Bogacz, Adrian; Pala, Damian; Knafel, Marcin
2015-04-01
This contribution describes the preliminary result of annual cooperation of three student research groups from AGH UST in Krakow, Poland. The aim of this cooperation was to develop and construct a high voltage seismic wave generator. Constructed device uses a high-energy electrical discharge to generate seismic wave in ground. This type of device can be applied in several different methods of seismic measurement, but because of its limited power it is mainly dedicated for engineering geophysics. The source operates on a basic physical principles. The energy is stored in capacitor bank, which is charged by two stage low to high voltage converter. Stored energy is then released in very short time through high voltage thyristor in spark gap. The whole appliance is powered from li-ion battery and controlled by ATmega microcontroller. It is possible to construct larger and more powerful device. In this contribution the structure of device with technical specifications is resented. As a part of the investigation the prototype was built and series of experiments conducted. System parameter was measured, on this basis specification of elements for the final device were chosen. First stage of the project was successful. It was possible to efficiently generate seismic waves with constructed device. Then the field test was conducted. Spark gap wasplaced in shallowborehole(0.5 m) filled with salt water. Geophones were placed on the ground in straight line. The comparison of signal registered with hammer source and sparker source was made. The results of the test measurements are presented and discussed. Analysis of the collected data shows that characteristic of generated seismic signal is very promising, thus confirms possibility of practical application of the new high voltage generator. The biggest advantage of presented device after signal characteristics is its size which is 0.5 x 0.25 x 0.2 m and weight approximately 7 kg. This features with small li-ion battery makes
Increased voltage photovoltaic cell
Ross, B.; Bickler, D. B.; Gallagher, B. D. (Inventor)
1985-01-01
A photovoltaic cell, such as a solar cell, is provided which has a higher output voltage than prior cells. The improved cell includes a substrate of doped silicon, a first layer of silicon disposed on the substrate and having opposite doping, and a second layer of silicon carbide disposed on the first layer. The silicon carbide preferably has the same type of doping as the first layer.
Methods for calculating the electrode position Jacobian for impedance imaging.
Boyle, A; Crabb, M G; Jehl, M; Lionheart, W R B; Adler, A
2017-03-01
Electrical impedance tomography (EIT) or electrical resistivity tomography (ERT) current and measure voltages at the boundary of a domain through electrodes. The movement or incorrect placement of electrodes may lead to modelling errors that result in significant reconstructed image artifacts. These errors may be accounted for by allowing for electrode position estimates in the model. Movement may be reconstructed through a first-order approximation, the electrode position Jacobian. A reconstruction that incorporates electrode position estimates and conductivity can significantly reduce image artifacts. Conversely, if electrode position is ignored it can be difficult to distinguish true conductivity changes from reconstruction artifacts which may increase the risk of a flawed interpretation. In this work, we aim to determine the fastest, most accurate approach for estimating the electrode position Jacobian. Four methods of calculating the electrode position Jacobian were evaluated on a homogeneous halfspace. Results show that Fréchet derivative and rank-one update methods are competitive in computational efficiency but achieve different solutions for certain values of contact impedance and mesh density.
A differential low-voltage high gain current-mode integrated RF receiver front-end
Energy Technology Data Exchange (ETDEWEB)
Wang Chunhua; Ma Minglin; Sun Jingru; Du Sichun; Guo Xiaorong; He Haizhen, E-mail: wch1227164@sina.com [School of Information Science and Technology, Hunan University, Changsha 410082 (China)
2011-02-15
A differential low-voltage high gain current-mode integrated RF front end for an 802.11b WLAN is proposed. It contains a differential transconductance low noise amplifier (G{sub m}-LNA) and a differential current-mode down converted mixer. The single terminal of the G{sub m}-LNA contains just one MOS transistor, two capacitors and two inductors. The gate-source shunt capacitors, C{sub x1} and C{sub x2}, can not only reduce the effects of gate-source C{sub gs} on resonance frequency and input-matching impedance, but they also enable the gate inductance L{sub g1,2} to be selected at a very small value. The current-mode mixer is composed of four switched current mirrors. Adjusting the ratio of the drain channel sizes of the switched current mirrors can increase the gain of the mixer and accordingly increase the gain of RF receiver front-end. The RF front-end operates under 1 V supply voltage. The receiver RFIC was fabricated using a chartered 0.18 {mu}m CMOS process. The integrated RF receiver front-end has a measured power conversion gain of 17.48 dB and an input referred third-order intercept point (IIP3) of -7.02 dBm. The total noise figure is 4.5 dB and the power is only 14 mW by post-simulations. (semiconductor integrated circuits)
Jain, Anil K; Feng, Jianjiang
2009-06-01
The evidential value of palmprints in forensic applications is clear as about 30 percent of the latents recovered from crime scenes are from palms. While biometric systems for palmprint-based personal authentication in access control type of applications have been developed, they mostly deal with low-resolution (about 100 ppi) palmprints and only perform full-to-full palmprint matching. We propose a latent-to-full palmprint matching system that is needed in forensic applications. Our system deals with palmprints captured at 500 ppi (the current standard in forensic applications) or higher resolution and uses minutiae as features to be compatible with the methodology used by latent experts. Latent palmprint matching is a challenging problem because latent prints lifted at crime scenes are of poor image quality, cover only a small area of the palm, and have a complex background. Other difficulties include a large number of minutiae in full prints (about 10 times as many as fingerprints), and the presence of many creases in latents and full prints. A robust algorithm to reliably estimate the local ridge direction and frequency in palmprints is developed. This facilitates the extraction of ridge and minutiae features even in poor quality palmprints. A fixed-length minutia descriptor, MinutiaCode, is utilized to capture distinctive information around each minutia and an alignment-based minutiae matching algorithm is used to match two palmprints. Two sets of partial palmprints (150 live-scan partial palmprints and 100 latent palmprints) are matched to a background database of 10,200 full palmprints to test the proposed system. Despite the inherent difficulty of latent-to-full palmprint matching, rank-1 recognition rates of 78.7 and 69 percent, respectively, were achieved in searching live-scan partial palmprints and latent palmprints against the background database.
Modeling charge polarization voltage for large lithium-ion batteries in electric vehicles
Directory of Open Access Journals (Sweden)
Yan Jiang
2013-06-01
Full Text Available Purpose: Polarization voltage of the lithium-ion battery is an important parameter that has direct influence on battery performance. The paper aims to analyze the impedance characteristics of the lithium-ion battery based on EIS data. Design/methodology/approach: The effects of currents, initial SOC of the battery on charge polarization voltage are investigated, which is approximately linear function of charge current. The change of charge polarization voltage is also analyzed with the gradient analytical method in the SOC domain. The charge polarization model with two RC networks is presented, and parts of model parameters like Ohmic resistance and charge transfer impedance are estimated by both EIS method and battery constant current testing method. Findings: This paper reveals that the Ohmic resistance accounts for much contribution to battery total polarization compared to charge transfer impedance. Practical implications: Experimental results demonstrate the efficacy of the model with the proposed identification method, which provides the foundation for battery charging optimization. Originality/value: The paper analyzed the impedance characteristics of the lithium-ion battery based on EIS data, presented a charge polarization model with two RC networks, and estimated parameters like Ohmic resistance and charge transfer impedance.
A Study on the Optimal Receiver Impedance for SNR Maximization in Broadband PLC
Directory of Open Access Journals (Sweden)
Massimo Antoniali
2013-01-01
Full Text Available We consider the design of the front-end receiver for broadband power line communications. We focus on the design of the input impedance that maximizes the signal-to-noise ratio (SNR at the receiver. We show that the amplitude, rather than the power, of the received signal is important for communication purposes. Furthermore, we show that the receiver impedance impacts the amplitude of the noise term. We focus on the background noise, and we propose a novel description of the noise experienced at the receiver port of a PLC network. We model the noise as the sum of four uncorrelated contributions, that is, the active, resistive, receiver, and coupled noise components. We study the optimal impedance design problem for real in-home grids that we assessed with experimental measurements. We describe the results of the measurement campaign, and we report the statistics of the optimal impedance. Hence, we study the best attainable performance when the optimal receiver impedance is deployed. We focus on the SNR and the maximum achievable rate, and we show that power matching is suboptimal with respect to the proposed impedance design approach.
Energy Technology Data Exchange (ETDEWEB)
Niedermayer, U., E-mail: u.niedermayer@gsi.de [Technische Universitaet Darmstadt, Institut fuer Theorie Elektromagnetischer Felder, Schlossgartenstrasse 8, 64289 Darmstadt (Germany); Boine-Frankenheim, O. [Technische Universitaet Darmstadt, Institut fuer Theorie Elektromagnetischer Felder, Schlossgartenstrasse 8, 64289 Darmstadt (Germany)
2012-09-21
The resistive wall impedance is one of the main sources for beam instabilities in synchrotrons and storage rings. The fast ramped SIS18 synchrotron at GSI and the projected SIS100 synchrotron for FAIR both employ thin (0.3 mm) stainless steel beam pipes in order to reduce eddy current effects. The lowest betatron sidebands are at about 100 kHz, which demands accurate impedance predictions in the low frequency (LF) range where the beam pipe and possibly also the structures behind the pipe are the dominating impedance sources. The longitudinal and transverse resistive wall impedances of a circular multi-layer pipe are calculated analytically using the field matching technique. We compare the impedances obtained from a radial wave model, which corresponds to the setup used in bench measurements, with the axial wave model, which corresponds to an actual beam moving with relativistic velocity. For thin beam pipes the induced wall current and the corresponding shielding properties of the pipe are important. In both models the wall current is obtained analytically. The characteristic frequencies for the onset of the wall current are calculated from equivalent lumped element circuits corresponding to the radial model. For more complex structures, like the SIS100 beam pipe, we use a numerical method, in which the impedance is obtained from the total power loss. The method is validated by the analytic expressions for circular beam pipes.
Impedance adaptation methods of the piezoelectric energy harvesting
Kim, Hyeoungwoo
In this study, the important issues of energy recovery were addressed and a comprehensive investigation was performed on harvesting electrical power from an ambient mechanical vibration source. Also discussed are the impedance matching methods used to increase the efficiency of energy transfer from the environment to the application. Initially, the mechanical impedance matching method was investigated to increase mechanical energy transferred to the transducer from the environment. This was done by reducing the mechanical impedance such as damping factor and energy reflection ratio. The vibration source and the transducer were modeled by a two-degree-of-freedom dynamic system with mass, spring constant, and damper. The transmissibility employed to show how much mechanical energy that was transferred in this system was affected by the damping ratio and the stiffness of elastic materials. The mechanical impedance of the system was described by electrical system using analogy between the two systems in order to simply the total mechanical impedance. Secondly, the transduction rate of mechanical energy to electrical energy was improved by using a PZT material which has a high figure of merit and a high electromechanical coupling factor for electrical power generation, and a piezoelectric transducer which has a high transduction rate was designed and fabricated. The high g material (g33 = 40 [10-3Vm/N]) was developed to improve the figure of merit of the PZT ceramics. The cymbal composite transducer has been found as a promising structure for piezoelectric energy harvesting under high force at cyclic conditions (10--200 Hz), because it has almost 40 times higher effective strain coefficient than PZT ceramics. The endcap of cymbal also enhances the endurance of the ceramic to sustain ac load along with stress amplification. In addition, a macro fiber composite (MFC) was employed as a strain component because of its flexibility and the high electromechanical coupling
Directory of Open Access Journals (Sweden)
Xiaoming Zha
2016-11-01
Full Text Available Power hardware-in-the-loop (PHIL systems are advanced, real-time platforms for combined software and hardware testing. Two paramount issues in PHIL simulations are the closed-loop stability and simulation accuracy. This paper presents a virtual impedance (VI method for PHIL simulations that improves the simulation’s stability and accuracy. Through the establishment of an impedance model for a PHIL simulation circuit, which is composed of a voltage-source converter and a simple network, the stability and accuracy of the PHIL system are analyzed. Then, the proposed VI method is implemented in a digital real-time simulator and used to correct the combined impedance in the impedance model, achieving higher stability and accuracy of the results. The validity of the VI method is verified through the PHIL simulation of two typical PHIL examples.
A centralized control architecture for harmonic voltage suppression in islanded microgrids
DEFF Research Database (Denmark)
Wang, Xiongfei; Blaabjerg, Frede; Chen, Zhe
2011-01-01
Resource (DER) units. Thus the harmonic voltage distortion caused by the mismatch between the harmonic conductance and characteristic impedance of the distribution feeder can be reduced. Moreover, to overcome the constraint on transmitting harmonic signals by a low-bandwidth communication line, a Park...
Tracking of electrochemical impedance of batteries
Piret, H.; Granjon, P.; Guillet, N.; Cattin, V.
2016-04-01
This paper presents an evolutionary battery impedance estimation method, which can be easily embedded in vehicles or nomad devices. The proposed method not only allows an accurate frequency impedance estimation, but also a tracking of its temporal evolution contrary to classical electrochemical impedance spectroscopy methods. Taking into account constraints of cost and complexity, we propose to use the existing electronics of current control to perform a frequency evolutionary estimation of the electrochemical impedance. The developed method uses a simple wideband input signal, and relies on a recursive local average of Fourier transforms. The averaging is controlled by a single parameter, managing a trade-off between tracking and estimation performance. This normalized parameter allows to correctly adapt the behavior of the proposed estimator to the variations of the impedance. The advantage of the proposed method is twofold: the method is easy to embed into a simple electronic circuit, and the battery impedance estimator is evolutionary. The ability of the method to monitor the impedance over time is demonstrated on a simulator, and on a real Lithium ion battery, on which a repeatability study is carried out. The experiments reveal good tracking results, and estimation performance as accurate as the usual laboratory approaches.
Approaches for Stereo Matching
Directory of Open Access Journals (Sweden)
Takouhi Ozanian
1995-04-01
Full Text Available This review focuses on the last decade's development of the computational stereopsis for recovering three-dimensional information. The main components of the stereo analysis are exposed: image acquisition and camera modeling, feature selection, feature matching and disparity interpretation. A brief survey is given of the well known feature selection approaches and the estimation parameters for this selection are mentioned. The difficulties in identifying correspondent locations in the two images are explained. Methods as to how effectively to constrain the search for correct solution of the correspondence problem are discussed, as are strategies for the whole matching process. Reasons for the occurrence of matching errors are considered. Some recently proposed approaches, employing new ideas in the modeling of stereo matching in terms of energy minimization, are described. Acknowledging the importance of computation time for real-time applications, special attention is paid to parallelism as a way to achieve the required level of performance. The development of trinocular stereo analysis as an alternative to the conventional binocular one, is described. Finally a classification based on the test images for verification of the stereo matching algorithms, is supplied.
Benchmarking of Voltage Sag Generators
DEFF Research Database (Denmark)
Yang, Yongheng; Blaabjerg, Frede; Zou, Zhixiang
2012-01-01
The increased penetration of renewable energy systems, like photovoltaic and wind power systems, rises the concern about the power quality and stability of the utility grid. Some regulations for Low Voltage Ride-Through (LVRT) for medium voltage or high voltage applications, are coming into force...
Brainard, John P [Albuquerque, NM; Christenson, Todd R [Albuquerque, NM
2009-11-03
A charge-pump voltage converter for converting a low voltage provided by a low-voltage source to a higher voltage. Charge is inductively generated on a transfer rotor electrode during its transit past an inductor stator electrode and subsequently transferred by the rotating rotor to a collector stator electrode for storage or use. Repetition of the charge transfer process leads to a build-up of voltage on a charge-receiving device. Connection of multiple charge-pump voltage converters in series can generate higher voltages, and connection of multiple charge-pump voltage converters in parallel can generate higher currents. Microelectromechanical (MEMS) embodiments of this invention provide a small and compact high-voltage (several hundred V) voltage source starting with a few-V initial voltage source. The microscale size of many embodiments of this invention make it ideally suited for MEMS- and other micro-applications where integration of the voltage or charge source in a small package is highly desirable.
Liang, Zhen; Li, Bin; Huang, Mo; Zheng, Yanqi; Ye, Hui; Xu, Ken; Deng, Fangming
2017-04-19
In this work, a low cost Bluetooth Low Energy (BLE) transceiver for wireless sensor network (WSN) applications, with a receiver (RX)-matching network-reusing power amplifier (PA) load inductor, is presented. In order to decrease the die area, only two inductors were used in this work. Besides the one used in the voltage control oscillator (VCO), the PA load inductor was reused as the RX impedance matching component in the front-end. Proper controls have been applied to achieve high transmitter (TX) input impedance when the transceiver is in the receiving mode, and vice versa. This allows the TRX-switch/matching network integration without significant performance degradation. The RX adopted a low-IF structure and integrated a single-ended low noise amplifier (LNA), a current bleeding mixer, a 4th complex filter and a delta-sigma continuous time (CT) analog-to-digital converter (ADC). The TX employed a two-point PLL-based architecture with a non-linear PA. The RX achieved a sensitivity of -93 dBm and consumes 9.7 mW, while the TX achieved a 2.97% error vector magnitude (EVM) with 9.4 mW at 0 dBm output power. This design was fabricated in a 0.11 μm complementary metal oxide semiconductor (CMOS) technology and the front-end circuit only occupies 0.24 mm². The measurement results verify the effectiveness and applicability of the proposed BLE transceiver for WSN applications.
Energy Technology Data Exchange (ETDEWEB)
Boix, Pablo P.; Guerrero, Antonio; Garcia-Belmonte, Germa; Bisquert, Juan [Photovoltaic and Optoelectronic Devices Group, Departament de Fisica, Universitat Jaume I, ES-12071 Castello (Spain); Marchesi, Luis F. [Laboratorio Interdisciplinar de, Eletroquimica e Ceramica (LIEC), Universidade Federal de Sao Carlos (Brazil); Photovoltaic and Optoelectronic Devices Group, Departament de Fisica, Universitat Jaume I, ES-12071 Castello (Spain)
2011-11-15
A connection is established between recombination and series resistances extracted from impedance spectroscopy and current-voltage curves of polythiophene:fullerene organic solar cells. Recombination is shown to depend exclusively on the (Fermi level) voltage, which allows construction of the current-voltage characteristics in any required conditions based on a restricted set of measurements. The analysis highlights carrier recombination current as the determining mechanism of organic solar cell performance. (Copyright copyright 2011 WILEY-VCH Verlag GmbH and Co. KGaA, Weinheim)
Novel T-Z source inverter with high voltage gain and reduced transformer turn ratio
DEFF Research Database (Denmark)
Mostaan, Ali; Sharifi Malfejani, Saeed; Soltani, Mohsen
2015-01-01
Novel voltage source inverter based on the Z source inverter structure is introduced in this paper. In this new inverter, two inductors of the impedance network in conventional Z source inverter are replaced with two transformers or coupled inductors, but unlike the T-Z source inverter that it......'s voltage gain is raised with increasing the transformers turn ratio, the voltage gain in the proposed inverter is raised with lowering the transformer turn ratio. Therefore, transformers with lower turn ratio are required in the proposed inverter in compare with T- Z source inverter that can lead to lower...
Universal Voltage Conveyor and Current Conveyor in Fast Full-Wave Rectifier
Directory of Open Access Journals (Sweden)
Josef Burian
2012-12-01
Full Text Available This paper deals about the design of a fast voltage-mode full-wave rectifier, where universal voltage conveyor and second-generation current conveyor are used as active elements. Thanks to the active elements, the input and output impedance of the non-linear circuit is infinitely high respectively zero in theory. For the rectification only two diodes and three resistors are required as passive elements. The performance of the circuit is shown on experimental measurement results showing the dynamic range, time response, frequency dependent DC transient value and RMS error for different values of input voltage amplitudes.
Experimental Observation of RF Radiation Generated by an Explosively Driven Voltage Generator
National Research Council Canada - National Science Library
Rader, Mark S; Sullivan, Carol; Andreadis, Tim D
2005-01-01
.... located in Rolla, Missouri. A pulse-forming network and transmitting antenna were constructed and attached to the voltage-pulse generator, and a matched receiving antenna was located approximately 3 m from the transmitting antenna...
Transient voltage sharing in series-coupled high voltage switches
Directory of Open Access Journals (Sweden)
Editorial Office
1992-07-01
Full Text Available For switching voltages in excess of the maximum blocking voltage of a switching element (for example, thyristor, MOSFET or bipolar transistor such elements are often coupled in series - and additional circuitry has to be provided to ensure equal voltage sharing. Between each such series element and system ground there is a certain parasitic capacitance that may draw a significant current during high-speed voltage transients. The "open" switch is modelled as a ladder network. Analysis reveals an exponential progression in the distribution of the applied voltage across the elements. Overstressing thus occurs in some of the elements at levels of the total voltage that are significantly below the design value. This difficulty is overcome by grading the voltage sharing circuitry, coupled in parallel with each element, in a prescribed manner, as set out here.
Wearable impedance monitoring system for dialysis patients.
Bonnet, S; Bourgerette, A; Gharbi, S; Rubeck, C; Arkouche, W; Massot, B; McAdams, E; Montalibet, A; Jallon, P
2016-08-01
This paper describes the development and the validation of a prototype wearable miniaturized impedance monitoring system for remote monitoring in home-based dialysis patients. This device is intended to assess the hydration status of dialysis patients using calf impedance measurements. The system is based on the low-power AD8302 component. The impedance calibration procedure is described together with the Cole parameter estimation and the hydric volume estimation. Results are given on a test cell to validate the design and on preliminary calf measurements showing Cole parameter variations during hemodialysis.
Identification of irradiated potatoes by impedance
International Nuclear Information System (INIS)
Singh, Rita; Singh, Antaryami; Wadhawan, A.K.
1997-01-01
The electrical impedance of potatoes irradiated at 60, 90, 150 and 1000 Gy was measured using various frequencies of alternating current. The impedance of the irradiated potatoes was higher than the unirradiated potatoes particularly in the frequency range of 100 Hz to 10 kHz. The ratio of the impedance at 5 kHz to that at 50 Hz (Z5k/Z50) was found to be the best indicator for detection of radiation treatment. (author). 4 refs., 2 figs
Small-signal Loudspeaker Impedance Emulator
DEFF Research Database (Denmark)
Knott, Arnold; Iversen, Niels Elkjær
2014-01-01
Specifying the performance of audio ampliers is typically done by playing sine waves into a pure ohmic load. However real loudspeaker impedances are not purely ohmic but characterised by the mechanical resonance between the mass of the diaphragm and the compliance of its' suspension which vary from...... driver to driver. Therefore a loudspeaker emulator capable of adjusting its' impedance to a given driver is in need for measurement purposes. This paper proposes a loudspeaker emulator circuit for small signals. Simulations and experimental results are compared and show that it is possible to emulate...... the loudspeaker impedance with an electric circuit and that its' resonance frequency can be changed by tuning two resistors....
RHIC ABORT KICKER WITH REDUCED COUPLING IMPEDANCE
International Nuclear Information System (INIS)
HAHN, H.; DAVINO, D.
2002-01-01
Kicker magnets typically represent the most important contributors to the transverse impedance budget of accelerators and storage rings. Methods of reducing the impedance value of the SNS extraction kicker presently under construction and, in view of a future performance upgrade, that of the RHIC abort kicker have been thoroughly studied at this laboratory. In this paper, the investigation of a potential improvement from using ferrite different from the BNL standard CMD5005 is reported. Permeability measurements of several ferrite types have been performed. Measurements on two kicker magnets using CMD5005 and C2050 suggest that the impedance of a magnet without external resistive damping, such as the RHIC abort kicker, would benefit
SPATIAL VARIABILITY OF PEDOZEMS MECHANICAL IMPEDANCE
Directory of Open Access Journals (Sweden)
Zhukov A.V.
2013-04-01
Full Text Available We studied the spatial variability of pedozem mechanical impedance in ResearchRemediation Center of the Dnipropetrovsk State Agrarian University in Ordzhonikidze. Thestatistical distribution of the soil mechanical impedance within the studied area is characterized by deviation from the normal law in 0–10 and 30–50 cm layers from the surface. 2D and 3D modeling shows the structural design of the soil as locations of high mechanical impedance which found in the soils with less hardness.
Sound Propagation An impedance Based Approach
Kim, Yang-Hann
2010-01-01
In Sound Propagation: An Impedance Based Approach , Professor Yang-Hann Kim introduces acoustics and sound fields by using the concept of impedance. Kim starts with vibrations and waves, demonstrating how vibration can be envisaged as a kind of wave, mathematically and physically. One-dimensional waves are used to convey the fundamental concepts. Readers can then understand wave propagation in terms of characteristic and driving point impedance. The essential measures for acoustic waves, such as dB scale, octave scale, acoustic pressure, energy, and intensity, are explained. These measures are
Relationship between moisture content and electrical impedance of carrot slices during drying
Kertész, Ákos; Hlaváčová, Zuzana; Vozáry, Eszter; Staroňová, Lenka
2015-01-01
Electrical properties of food materials can give information about the inner structure and physiological state of biological tissues. Generally, the process of drying of fruits and vegetables is followed by weight loss. The aim of this study was to measure the impedance spectra of carrot slices during drying and to correlate impedance parameters to moisture content in different drying periods. Cylindrical slices were cut out from the carrot root along the axis. The slices were dried in a Venticell 111 air oven at 50°C. The weight of the slices was measured with a Denver SI-603 electronic analytical and precision balance. The weighing of the samples was performed every 30 min at the beginning of drying and every 60 min after the process. The moisture content of the samples was calculated on wet basis. The magnitude and phase angle of electrical impedance of the slices were measured with HP 4284A and 4285A precision LCR meters in the frequency range from 30 Hz to 1 MHz and from 75 kHz to 30 MHz, respectively, at voltage 1 V. The impedance measurement was performed after weighting. The change in the magnitude of impedance during drying showed a good correlation with the change in the moisture content.
DEFF Research Database (Denmark)
Nainar, Karthikeyan; Pokhrel, Basanta Raj; Pillai, Jayakrishnan Radhakrishna
2017-01-01
This paper reviews and analyzes the existing voltage control methods of distributed solar PV inverters to improve the voltage regulation and thereby the hosting capacity of a low-voltage distribution network. A novel coordinated voltage control method is proposed based on voltage sensitivity...... optimization. The proposed method is used to calculate the voltage bands and droop settings of PV inverters at each node by the supervisory controller. The local controller of each PV inverter implements the volt/var control and if necessary, the active power curtailment as per the received settings and based...... on measured local voltages. The advantage of the proposed method is that the calculated reactive power and active power droop settings enable fair contribution of the PV inverters at each node to the voltage regulation. Simulation studies are conducted using DigSilent Power factory software on a simplified...
Non-foster matching of an RFID antenna
Mohamed Hassan Salem, Nedime Pelin
2011-07-01
Novel designs of radio-frequency identification (RFID) tag antennas with better matching characteristics to achieve extended range for passive tags are investigated in ultra-high frequency (UHF) band. A microstrip dipole antenna with or without an integrated negative impedance converter designed to cancel out the antenna\\'s input capacitance at resonance frequency was designed, simulated, constructed and measured for implementation in RFID applications. © 2011 IEEE.
Design of Matched Absorbing Layers for Surface Plasmon-Polaritons
Directory of Open Access Journals (Sweden)
Sergio de la Cruz
2012-01-01
Full Text Available We describe a procedure for designing metal-metal boundaries for the strong attenuation of surface plasmon-polaritons without the introduction of reflections or scattering effects. Solutions associated with different sets of matching materials are found. To illustrate the results and the consequences of adopting different solutions, we present calculations based on an integral equation formulation for the scattering problem and the use of a nonlocal impedance boundary condition.
Sensing voltage across lipid membranes
Swartz, Kenton J.
2009-01-01
The detection of electrical potentials across lipid bilayers by specialized membrane proteins is required for many fundamental cellular processes such as the generation and propagation of nerve impulses. These membrane proteins possess modular voltage-sensing domains, a notable example being the S1-S4 domains of voltage-activated ion channels. Ground-breaking structural studies on these domains explain how voltage sensors are designed and reveal important interactions with the surrounding lipid membrane. Although further structures are needed to fully understand the conformational changes that occur during voltage sensing, the available data help to frame several key concepts that are fundamental to the mechanism of voltage sensing. PMID:19092925
Matching problems in pulse power radial transmission lines
International Nuclear Information System (INIS)
Mittag, K.; Brandelik, A.
1984-12-01
In this report we study the power transfer from a generator along a coaxial transmission line followed by a radial transmission line into a load, which in our application is a pseudo-spark plasma of about one millimeter diameter and about 15 cm in length. First the theoretical background based on transmission line theory is described. Then numerical results are presented. The main conclusion is that when matching the pulse power generator to the pseudo-spark plasma, the effect of the impedance transformation caused by the radial transmission line has to be taken into account. The conditions to obtain an optimal match are described. (orig.) [de
Autonomous Voltage Oscillations in a Direct Methanol Fuel Cell
International Nuclear Information System (INIS)
Nogueira, Jéssica A.; Peña Arias, Ivonne K.; Hanke-Rauschenbach, Richard; Vidakovic-Koch, Tanja; Varela, Hamilton; Sundmacher, Kai
2016-01-01
Proton exchange membrane fuel cells fed with H_2/CO mixtures at the anode have a considerably lower performance than fuel cells fed with pure hydrogen. However, when operated in an autonomous oscillatory regime, the overall voltage loss decreases due to a self-cleaning mechanism. Another molecule, also widely used as feed in the fuel cell and susceptible to kinetic instabilities, is methanol. To the best of our knowledge, there are no reports on autonomous voltage oscillations in the direct methanol fuel cell (DMFC). The purpose of this work was to explore if such instabilities also occur in the DMFC system. Initially, half-cell experiments with a gas diffusion electrode were performed. Then, a DMFC was operated under current control and studied by means of electrochemical impedance spectroscopy. The half-cell measurements revealed that the induction period for oscillations depends on the mass transfer conditions, where on stagnant electrode the induction time was shorter than in the case of forced convection. The DMFC showed also autonomous voltage oscillations above a certain threshold current. The results obtained by electrochemical impedance spectroscopy give evidence of a negative differential resistance in the fuel cell, hitherto not described in the literature, which can be related to the appearance of oscillations during galvanostatic methanol electro-oxidation. These results open the possibility to evaluate the performance of low-temperature fuel cells fed with carbon-containing fuels under oscillatory operating conditions.
Theory for electrochemical impedance spectroscopy of ...
Indian Academy of Sciences (India)
Shweta Dhillon
generalize phenomenological theory for the Randles-Ershler admittance at the electrode with double layer capacitance ..... impedance (equation 12) curves accounting for the elec- ..... R.K. thanks University of Delhi for the financial support.
Battery impedance spectroscopy using bidirectional grid connected
Indian Academy of Sciences (India)
Keywords. Impedance spectroscopy; grid connection; battery converter; state of charge; health monitoring ... The converter is grid connected and controlled to operate at unity power factor. Additional ... Sadhana. Current Issue : Vol. 43, Issue 6.
Impedance and collective effects in the LHC
Energy Technology Data Exchange (ETDEWEB)
Gareyte, J [European Organization for Nuclear Research, Geneva (Switzerland)
1996-08-01
After a review of the main LHC parameters, and a brief description of the RF and vacuum systems, the coupling impedances of the main machine elements are given, as well as the resulting thresholds for instabilities. (author)
Electrochemical Impedance Studies of SOFC Cathodes
DEFF Research Database (Denmark)
Hjelm, Johan; Søgaard, Martin; Wandel, Marie
2007-01-01
Mixed ion- and electron-conducting composite electrodes consisting of doped ceria and perovskite have been studied by electrochemical impedance spectroscopy (EIS) at different temperatures and oxygen partial pressures. This paper aims to describe the different contributions to the polarisation...
Loudspeaker impedance emulator for multi resonant systems
DEFF Research Database (Denmark)
Iversen, Niels Elkjær; Knott, Arnold
2015-01-01
Specifying the performance of audio amplifiers is typically done by playing sine waves into a pure ohmic load. However real loudspeaker impedances are not purely ohmic but characterised by its electrical, mechanical and acoustical properties. Therefore a loudspeaker emulator capable of adjusting...... its impedance to that of a given loudspeaker is desired for measurement purposes. An adjustable RLC based emulator is implemented with switch controlled capacitors, air gap controlled inductors and potentiometers. Calculations and experimental results are compared and show that it is possible...... to emulate the loudspeaker impedance infinite baffle-, closed box- and the multi resonant vented box-loudspeaker by tuning the component values in the proposed circuit. Future work is outlined and encourage that the proposed impedance emulator is used as part of a control circuit in a switch-mode based...
Modeling degradation in SOEC impedance spectra
DEFF Research Database (Denmark)
Jensen, Søren Højgaard; Hauch, Anne; Knibbe, Ruth
2013-01-01
Solid oxide cell (SOC) performance is limited by various processes. One way to investigate these processes is by electrochemical impedance spectroscopy. In order to quantify and characterize the processes, an equivalent circuit can be used to model the SOC impedance spectra (IS). Unfortunately......, the optimal equivalent circuit is often unknown and to complicate matters further, several processes contribute to the SOC impedance - making detailed process characterization difficult. In this work we analyze and model a series of IS measured during steam electrolysis operation of an SOC. During testing......, degradation is only observed in the Ni/YSZ electrode and not in the electrolyte or the LSM/YSZ electrode. A batch fit of the differences between the IS shows that a modified Gerischer element provides a better fit to the Ni/YSZ electrode impedance than the frequently used RQ element - albeit neither...
Applications of Nonlinear Electrochemical Impedance Spectroscopy (NLEIS)
Adler, S. B.
2013-01-01
This paper reviews the use of nonlinear electrochemical impedance spectroscopy (NLEIS) in the analysis of SOFC electrode reactions. By combining EIS and NLEIS, as well as other independent information about an electrode material, it becomes possible
School Library Media Activities Monthly, 1996
1996-01-01
Giving students a chance to associate numbers with subjects can be useful in speeding their location of desired print or nonprint materials and helping students feel independent when browsing. A matching game for helping students learn the Dewey numbers is presented. Instructions for the library media specialist or teacher, instructions for…
Directory of Open Access Journals (Sweden)
Nima Tabatabaei
2018-04-01
Full Text Available Conventional infrared thermography techniques, including pulsed and lock-in thermography, have shown great potential for non-destructive evaluation of broad spectrum of materials, spanning from metals to polymers to biological tissues. However, performance of these techniques is often limited due to the diffuse nature of thermal wave fields, resulting in an inherent compromise between inspection depth and depth resolution. Recently, matched-filter thermography has been introduced as a means for overcoming this classic limitation to enable depth-resolved subsurface thermal imaging and improving axial/depth resolution. This paper reviews the basic principles and experimental results of matched-filter thermography: first, mathematical and signal processing concepts related to matched-fileting and pulse compression are discussed. Next, theoretical modeling of thermal-wave responses to matched-filter thermography using two categories of pulse compression techniques (linear frequency modulation and binary phase coding are reviewed. Key experimental results from literature demonstrating the maintenance of axial resolution while inspecting deep into opaque and turbid media are also presented and discussed. Finally, the concept of thermal coherence tomography for deconvolution of thermal responses of axially superposed sources and creation of depth-selective images in a diffusion-wave field is reviewed.
DEFF Research Database (Denmark)
Hartelius, Karsten; Carstensen, Jens Michael
2003-01-01
A method for locating distorted grid structures in images is presented. The method is based on the theories of template matching and Bayesian image restoration. The grid is modeled as a deformable template. Prior knowledge of the grid is described through a Markov random field (MRF) model which r...
Jeuring, J.T.
1995-01-01
The pattern matching problem can be informally specified as follows: given a pattern and a text, find all occurrences of the pattern in the text. The pattern and the text may both be lists, or they may both be trees, or they may both be multi-dimensional arrays, etc. This paper describes a general
Gallistel, C. R.; King, Adam Philip; Gottlieb, Daniel; Balci, Fuat; Papachristos, Efstathios B.; Szalecki, Matthew; Carbone, Kimberly S.
2007-01-01
Experimentally naive mice matched the proportions of their temporal investments (visit durations) in two feeding hoppers to the proportions of the food income (pellets per unit session time) derived from them in three experiments that varied the coupling between the behavioral investment and food income, from no coupling to strict coupling.…
Matching Supernovae to Galaxies
Kohler, Susanna
2016-12-01
developed a new automated algorithm for matching supernovae to their host galaxies. Their work builds on currently existing algorithms and makes use of information about the nearby galaxies, accounts for the uncertainty of the match, and even includes a machine learning component to improve the matching accuracy.Gupta and collaborators test their matching algorithm on catalogs of galaxies and simulated supernova events to quantify how well the algorithm is able to accurately recover the true hosts.Successful MatchingThe matching algorithms accuracy (purity) as a function of the true supernova-host separation, the supernova redshift, the true hosts brightness, and the true hosts size. [Gupta et al. 2016]The authors find that when the basic algorithm is run on catalog data, it matches supernovae to their hosts with 91% accuracy. Including the machine learning component, which is run after the initial matching algorithm, improves the accuracy of the matching to 97%.The encouraging results of this work which was intended as a proof of concept suggest that methods similar to this could prove very practical for tackling future survey data. And the method explored here has use beyond matching just supernovae to their host galaxies: it could also be applied to other extragalactic transients, such as gamma-ray bursts, tidal disruption events, or electromagnetic counterparts to gravitational-wave detections.CitationRavi R. Gupta et al 2016 AJ 152 154. doi:10.3847/0004-6256/152/6/154
CSR Impedance for Non-Ultrarelativistic Beams
International Nuclear Information System (INIS)
Li, Rui; Tsai, Cheng Y.
2015-09-01
For the analysis of the coherent synchrotron radiation (CSR)-induced microbunching gain in the low energy regime, such as when a high-brightness electron beam is transported through a low-energy merger in an energy-recovery linac (ERL) design, it is necessary to extend the CSR impedance expression in the ultrarelativistic limit to the non-ultrarelativistic regime. This paper presents our analysis of CSR impedance for general beam energies.
Ultrahigh impedance method to assess electrostatic accelerator performance
Directory of Open Access Journals (Sweden)
Nikolai R. Lobanov
2015-06-01
Full Text Available This paper describes an investigation of problem-solving procedures to troubleshoot electrostatic accelerators. A novel technique to diagnose issues with high-voltage components is described. The main application of this technique is noninvasive testing of electrostatic accelerator high-voltage grading systems, measuring insulation resistance, or determining the volume and surface resistivity of insulation materials used in column posts and acceleration tubes. In addition, this technique allows verification of the continuity of the resistive divider assembly as a complete circuit, revealing if an electrical path exists between equipotential rings, resistors, tube electrodes, and column post-to-tube conductors. It is capable of identifying and locating a “microbreak” in a resistor and the experimental validation of the transfer function of the high impedance energy control element. A simple and practical fault-finding procedure has been developed based on fundamental principles. The experimental distributions of relative resistance deviations (ΔR/R for both accelerating tubes and posts were collected during five scheduled accelerator maintenance tank openings during 2013 and 2014. Components with measured ΔR/R>±2.5% were considered faulty and put through a detailed examination, with faults categorized. In total, thirty four unique fault categories were identified and most would not be identifiable without the new technique described. The most common failure mode was permanent and irreversible insulator current leakage that developed after being exposed to the ambient environment. As a result of efficient in situ troubleshooting and fault-elimination techniques, the maximum values of |ΔR/R| are kept below 2.5% at the conclusion of maintenance procedures. The acceptance margin could be narrowed even further by a factor of 2.5 by increasing the test voltage from 40 V up to 100 V. Based on experience over the last two years, resistor and
Low Impedance Carbon Adhesive Electrodes with Long Shelf Life.
Posada-Quintero, Hugo F; Reyes, Bersaín A; Burnham, Ken; Pennace, John; Chon, Ki H
2015-10-01
A novel electrocardiogram (ECG) electrode film is developed by mixing carbon black powder and a quaternary salt with a visco-elastic polymeric adhesive. Unlike traditional wet gel-based electrodes, carbon/salt/adhesive (CSA) electrodes should theoretically have an infinite shelf life as they do not dehydrate even after a prolonged period of storage. The CSA electrodes are electrically activated for use through the process of electrophoresis. Specifically, the activation procedure involves sending a high voltage and current through the electrode, which results in significant reduction of impedance so that high fidelity ECG signals can be obtained. Using the activation procedure, the ideal concentration of carbon black powder in the mixture with the adhesive was examined. It was determined that the optimum concentration of carbon black which minimized post-activation impedance was 10%. Once the optimal carbon black powder concentration was determined, extensive signal analysis was performed to compare the performance of the CSA electrodes to the standard silver-silver chloride (Ag/AgCl) electrodes. As a part of data analysis, electrode-skin contact impedance of the CSA was measured and compared to the standard Ag/AgCl electrodes; we found consistently lower impedance for CSA electrodes. For quantitative data analysis, we simultaneously collected ECG data with CSA and Ag/AgCl electrodes from 17 healthy subjects. Heart rate variability (HRV) indices and ECG morphological waveforms were calculated to compare CSA and Ag/AgCl electrodes. Non-significant differences for most of the HRV indices between CSA and Ag/AgCl electrodes were found. Of the morphological waveform metrics consisting of R-wave peak amplitude, ST-segment elevation and QT interval, only the first index was found to be significantly different between the two media. The response of CSA electrodes to motion artifacts was also tested, and we found in general no difference in the quality of the ECG signal
DEFF Research Database (Denmark)
Nørgaard, Kren Rahbek; Fernandez Grande, Efren
2017-01-01
The ear-canal acoustic impedance and reflectance are useful for assessing conductive hearing disorders and calibrating stimulus levels in situ. However, such probe-based measurements are affected by errors due to the presence of evanescent modes and incorrect estimates or assumptions regarding...... characteristic impedance. This paper proposes a method to compensate for evanescent modes in measurements of acoustic impedance, reflectance, and sound pressure in waveguides, as well as estimating the characteristic impedance immediately in front of the probe. This is achieved by adjusting the characteristic...... impedance and subtracting an acoustic inertance from the measured impedance such that the non-causality in the reflectance is minimized in the frequency domain using the Hilbert transform. The method is thus capable of estimating plane-wave quantities of the sought-for parameters by supplying only...
DEFF Research Database (Denmark)
Savaghebi, Mehdi; Guerrero, Josep M.; Jalilian, Alireza
2011-01-01
In this paper, a microgrid hierarchical control scheme is proposed which includes primary and secondary control levels. The primary level comprises distributed generators (DGs) local controllers. The local controller mainly consists of active and reactive power controllers, voltage and current...... controllers, and virtual impedance loop. A novel virtual impedance structure is proposed to achieve proper sharing of non-fundamental power among the microgrid DGs. The secondary level is designed to manage compensation of voltage harmonics at the microgrid load bus (LB) to which the sensitive loads may...... be connected. Also, restoration of LB voltage amplitude and microgrid frequency to the rated values is directed by the secondary level. These functions are achieved by sending proper control signals to the local controllers. The simulation results show the effectiveness of the proposed control scheme....
[Propensity score matching in SPSS].
Huang, Fuqiang; DU, Chunlin; Sun, Menghui; Ning, Bing; Luo, Ying; An, Shengli
2015-11-01
To realize propensity score matching in PS Matching module of SPSS and interpret the analysis results. The R software and plug-in that could link with the corresponding versions of SPSS and propensity score matching package were installed. A PS matching module was added in the SPSS interface, and its use was demonstrated with test data. Score estimation and nearest neighbor matching was achieved with the PS matching module, and the results of qualitative and quantitative statistical description and evaluation were presented in the form of a graph matching. Propensity score matching can be accomplished conveniently using SPSS software.
Exponential current pulse generation for efficient very high-impedance multisite stimulation.
Ethier, S; Sawan, M
2011-02-01
We describe in this paper an intracortical current-pulse generator for high-impedance microstimulation. This dual-chip system features a stimuli generator and a high-voltage electrode driver. The stimuli generator produces flexible rising exponential pulses in addition to standard rectangular stimuli. This novel stimulation waveform is expected to provide superior energy efficiency for action potential triggering while releasing less toxic reduced ions in the cortical tissues. The proposed fully integrated electrode driver is used as the output stage where high-voltage supplies are generated on-chip to significantly increase the voltage compliance for stimulation through high-impedance electrode-tissue interfaces. The stimuli generator has been implemented in 0.18-μm CMOS technology while a 0.8-μm CMOS/DMOS process has been used to integrate the high-voltage output stage. Experimental results show that the rectangular pulses cover a range of 1.6 to 167.2 μA with a DNL and an INL of 0.098 and 0.163 least-significant bit, respectively. The maximal dynamic range of the generated exponential reaches 34.36 dB at full scale within an error of ± 0.5 dB while all of its parameters (amplitude, duration, and time constant) are independently programmable over wide ranges. This chip consumes a maximum of 88.3 μ W in the exponential mode. High-voltage supplies of 8.95 and -8.46 V are generated by the output stage, boosting the voltage swing up to 13.6 V for a load as high as 100 kΩ.
Heat-pump performance: voltage dip/sag, under-voltage and over-voltage
Directory of Open Access Journals (Sweden)
William J.B. Heffernan
2014-12-01
Full Text Available Reverse cycle air-source heat-pumps are an increasingly significant load in New Zealand and in many other countries. This has raised concern over the impact wide-spread use of heat-pumps may have on the grid. The characteristics of the loads connected to the power system are changing because of heat-pumps. Their performance during under-voltage events such as voltage dips has the potential to compound the event and possibly cause voltage collapse. In this study, results from testing six heat-pumps are presented to assess their performance at various voltages and hence their impact on voltage stability.
Directory of Open Access Journals (Sweden)
ABYANEH, H. A.
2010-11-01
Full Text Available Development of distribution network and power consumption growth, increase voltage drop on the line impedance and therefore voltage drop in system buses. In some cases consumption is so high that voltage in some buses exceed from standard. In this paper, effect of the fuel cell and photovoltaic hybrid system on distribution network for solving expressed problem is studied. For determining the capacity of each distributed generation source, voltage limitation on the bus voltages under different conditions is considered. Simulation is done by using DIgSILENT software on the part of the 20 kV real life Sirjan distribution system. In this article, optimum location with regard to system and environmental conditions are studied in two different viewpoints.
International Nuclear Information System (INIS)
Barnes, C.W.; Jarboe, T.R.; Marklin, G.J.; Knox, S.O.; Henins, I.
1989-01-01
Electrostatic (dc) helicity injection has previously been shown to successfully sustain the magnetic fields of spheromaks and tokamaks. The magnitude of the injected magnetic helicity balances (within experimental error) the flux lost be resistive decay of the toroidal equilibrium. The problem of optimizing this current drive scheme hence involves maximizing the injected helicity (the voltage-connecting-flux product) while minimizing the current (which multiplied by the voltage represents the energy input and also possible damage to the electrodes). The impedance (voltage-to-current ratio) and energy efficiency of a dc helicity injection experiment are studied on the CTX spheromak. Over several years changes were made in the physical geometry of the coaxial magnetized plasmas source as well as changes in the external electrical circuit. The source could be operated over a wide range of external charging voltage (and hence current), applied axial flux, and source gas flow rate. A database of resulting voltage, helicity injection, efficiency, electron density, and rotation has been created. These experimental results are compared to an ideal magnetohydrodynamic theory of magnetic flux flow. The theory is parameterized by the dimensionless Hall parameter, the ratio of electric to mass current. For a constant Hall parameter the theory explains why the voltage depends quadratically on the current at constant flux. The theory also explains the approximately linear dependence of the impedance-to-current ratio on the current-to-flux ratio of the source. 9 refs., 6 figs
EPG waveforms of blue green sharpshooter: impedance and voltage level effects on stylet probing
Blue-green sharpshooter (BGSS), Graphocephala atropunctata, is a native California vector of Xylella fastidiosa (Xf) a foregut-borne bacterium that is the causal agent of Pierce’s disease in grapevines. A 3rd-generation, AC-DC electropenetrograph (EPG) was used to record probing behaviors of adult B...
Design and Control of Parallel Three Phase Voltage Source Inverters in Low Voltage AC Microgrid
Directory of Open Access Journals (Sweden)
El Hassane Margoum
2017-01-01
Full Text Available Design and hierarchical control of three phase parallel Voltage Source Inverters are developed in this paper. The control scheme is based on synchronous reference frame and consists of primary and secondary control levels. The primary control consists of the droop control and the virtual output impedance loops. This control level is designed to share the active and reactive power correctly between the connected VSIs in order to avoid the undesired circulating current and overload of the connected VSIs. The secondary control is designed to clear the magnitude and the frequency deviations caused by the primary control. The control structure is validated through dynamics simulations.The obtained results demonstrate the effectiveness of the control structure.
Microwave noise detection of a quantum dot with stub impedance matching
Hasler, Thomas
2016-01-01
Noise is defined as random fluctuations of a signal in time. The fundamental requirement for noise is some sort of randomness. Noise is well-known and infamous to every experimentalist - whether he is working in the field of electronics, optics, acoustics or anywhere else - since such fluctuations are inherent and unavoidable in many systems. For most of us, the word noise has a negative connotation. It is considered to be an unwanted disturbance superposed on a useful signal, which tend...
Decrossas, Emmanuel (Inventor); Chattopadhyay, Goutam (Inventor); Chahat, Nacer (Inventor); Tang, Adrian J. (Inventor)
2016-01-01
A lens for interconnecting a metallic waveguide with a dielectric waveguide is provided. The lens may be coupled a metallic waveguide and a dielectric waveguide, and minimize a signal loss between the metallic waveguide and the dielectric waveguide.
Controllable transmission and total reflection through an impedance-matched acoustic metasurface
Mei, Jun; Wu, Ying
2014-01-01
supplies the functionalities of reflection-type acoustic metasurfaces, but also exhibits unprecedented flexibility and efficiency in various domains of wave manipulation for possible applications in fields like refracting, collimating, focusing or absorbing
Adaptive Discrete Hypergraph Matching.
Yan, Junchi; Li, Changsheng; Li, Yin; Cao, Guitao
2018-02-01
This paper addresses the problem of hypergraph matching using higher-order affinity information. We propose a solver that iteratively updates the solution in the discrete domain by linear assignment approximation. The proposed method is guaranteed to converge to a stationary discrete solution and avoids the annealing procedure and ad-hoc post binarization step that are required in several previous methods. Specifically, we start with a simple iterative discrete gradient assignment solver. This solver can be trapped in an -circle sequence under moderate conditions, where is the order of the graph matching problem. We then devise an adaptive relaxation mechanism to jump out this degenerating case and show that the resulting new path will converge to a fixed solution in the discrete domain. The proposed method is tested on both synthetic and real-world benchmarks. The experimental results corroborate the efficacy of our method.
Electromagnetic wave matching device
International Nuclear Information System (INIS)
Hirata, Yosuke; Mitsunaka, Yoshika; Hayashi, Ken-ichi; Ito, Yasuyuki.
1997-01-01
The present invention provides a matching device capable of increasing an efficiency of combining beams of electromagnetic waves outputted from an output window of a gyrotron which is expected for plasma heating of a thermonuclear reactor and an electromagnetic wave transmission system as high as possible. Namely, an electromagnetic wave matching device reflects beams of electromagnetic waves incident from an inlet by a plurality of phase correction mirrors and combines them to an external transmission system through an exit. In this case, the phase correction mirrors change the phase of the beams of electromagnetic waves incident to the phase correction mirrors by a predetermined amount corresponding to the position of the reflection mirrors. Then, the beams of electromagnetic waves outputted, for example, from a gyrotron can properly be shaped as desired for the intensity and the phase. As a result, combination efficiency with the transmission system can be increased. (I.S.)
High voltage isolation transformer
Clatterbuck, C. H.; Ruitberg, A. P. (Inventor)
1985-01-01
A high voltage isolation transformer is provided with primary and secondary coils separated by discrete electrostatic shields from the surfaces of insulating spools on which the coils are wound. The electrostatic shields are formed by coatings of a compound with a low electrical conductivity which completely encase the coils and adhere to the surfaces of the insulating spools adjacent to the coils. Coatings of the compound also line axial bores of the spools, thereby forming electrostatic shields separating the spools from legs of a ferromagnetic core extending through the bores. The transformer is able to isolate a high constant potential applied to one of its coils, without the occurrence of sparking or corona, by coupling the coatings, lining the axial bores to the ferromagnetic core and by coupling one terminal of each coil to the respective coating encasing the coil.
Directory of Open Access Journals (Sweden)
Mohamed ZELLAGUI
2012-08-01
Full Text Available This paper presents and compares the impact of SSSC on measured impedance for single phase to ground fault condition. The presence of Static Synchronous SSSC on a transmission line has a great influence on the ZRelay in distance protection. The protection of the high voltage 220 kV single circuit transmission line in eastern Algerian electrical transmission networks is affected in the case with resistance fault RF. The paper investigate the effect of Static Synchronous Series Compensator (SSSC on the measured impedance (Relay taking into account the distance fault point (n and fault resistance (RF. The resultants simulation is performed in MATLAB software environment.
International Nuclear Information System (INIS)
Valeev, R.I.; Nikiforov, M.G.; Kharchenko, A.F.
1988-01-01
The design is described and the test results of a four-channel pulse-voltage generator with maximum output voltage 200 kV are presented. The measurement results of generator triggering time depending on the value and polarity of the triggering voltage pulse for different triggering circuits are presented. The tests have shown stable triggering of all four channels of the generator in the range up to 40 % from selfbreakdown voltage. The generator triggering delay in the given range is <25 ns, asynchronism in channel triggering is <±1 ns
Voltage Dependence of Supercapacitor Capacitance
Directory of Open Access Journals (Sweden)
Szewczyk Arkadiusz
2016-09-01
Full Text Available Electronic Double-Layer Capacitors (EDLC, called Supercapacitors (SC, are electronic devices that are capable to store a relatively high amount of energy in a small volume comparing to other types of capacitors. They are composed of an activated carbon layer and electrolyte solution. The charge is stored on electrodes, forming the Helmholtz layer, and in electrolyte. The capacitance of supercapacitor is voltage- dependent. We propose an experimental method, based on monitoring of charging and discharging a supercapacitor, which enables to evaluate the charge in an SC structure as well as the Capacitance-Voltage (C-V dependence. The measurement setup, method and experimental results of charging/discharging commercially available supercapacitors in various voltage and current conditions are presented. The total charge stored in an SC structure is proportional to the square of voltage at SC electrodes while the charge on electrodes increases linearly with the voltage on SC electrodes. The Helmholtz capacitance increases linearly with the voltage bias while a sublinear increase of total capacitance was found. The voltage on SC increases after the discharge of electrodes due to diffusion of charges from the electrolyte to the electrodes. We have found that the recovery voltage value is linearly proportional to the initial bias voltage value.
State of charge classification for lithium-ion batteries using impedance based features
Directory of Open Access Journals (Sweden)
M. P. Felder
2017-09-01
Full Text Available Currently, the electrification of the drive train of passenger cars takes place, and the task of obtaining precise knowledge about the condition of the on board batteries gains importance. Due to a flat open circuit voltage (OCV to state of charge (SoC characteristic of lithium ion batteries, methods employed in applications with other cell chemistries cannot be adapted. Exploiting the higher significance of the impedance for state estimation for that chemistry, new impedance based features are proposed by this work. To evaluate the suitability of these features, simulations have been conducted using a simplified on-board power supply net as excitation source. The simulation outcome has been investigated regarding the cross correlation factor rxy and in a polynomial regression scenario. The results of the simulations show a best case error below 1 % SoC, which is 3 percentage points lower than using terminal voltage and impedance. When increasing the measurement uncertainty, the difference remains around 2 percent points.
Magnetic resonance electrical impedance tomography (MREIT): conductivity and current density imaging
International Nuclear Information System (INIS)
Seo, Jin Keun; Kwon, Ohin; Woo, Eung Je
2005-01-01
This paper reviews the latest impedance imaging technique called Magnetic Resonance Electrical Impedance Tomography (MREIT) providing information on electrical conductivity and current density distributions inside an electrically conducting domain such as the human body. The motivation for this research is explained by discussing conductivity changes related with physiological and pathological events, electromagnetic source imaging and electromagnetic stimulations. We briefly summarize the related technique of Electrical Impedance Tomography (EIT) that deals with cross-sectional image reconstructions of conductivity distributions from boundary measurements of current-voltage data. Noting that EIT suffers from the ill-posed nature of the corresponding inverse problem, we introduce MREIT as a new conductivity imaging modality providing images with better spatial resolution and accuracy. MREIT utilizes internal information on the induced magnetic field in addition to the boundary current-voltage measurements to produce three-dimensional images of conductivity and current density distributions. Mathematical theory, algorithms, and experimental methods of current MREIT research are described. With numerous potential applications in mind, future research directions in MREIT are proposed
TECHNICAL NOTE: Portable audio electronics for impedance-based measurements in microfluidics
Wood, Paul; Sinton, David
2010-08-01
We demonstrate the use of audio electronics-based signals to perform on-chip electrochemical measurements. Cell phones and portable music players are examples of consumer electronics that are easily operated and are ubiquitous worldwide. Audio output (play) and input (record) signals are voltage based and contain frequency and amplitude information. A cell phone, laptop soundcard and two compact audio players are compared with respect to frequency response; the laptop soundcard provides the most uniform frequency response, while the cell phone performance is found to be insufficient. The audio signals in the common portable music players and laptop soundcard operate in the range of 20 Hz to 20 kHz and are found to be applicable, as voltage input and output signals, to impedance-based electrochemical measurements in microfluidic systems. Validated impedance-based measurements of concentration (0.1-50 mM), flow rate (2-120 µL min-1) and particle detection (32 µm diameter) are demonstrated. The prevailing, lossless, wave audio file format is found to be suitable for data transmission to and from external sources, such as a centralized lab, and the cost of all hardware (in addition to audio devices) is ~10 USD. The utility demonstrated here, in combination with the ubiquitous nature of portable audio electronics, presents new opportunities for impedance-based measurements in portable microfluidic systems.
DEFF Research Database (Denmark)
Wei, Baoze; Guerrero, Josep M.; Quintero, Juan Carlos Vasquez
2017-01-01
This paper presents a theoretical study with experimental validation of a circulating-current suppression method for parallel operation of three-phase voltage source inverters (VSI), which may be suitable for modular parallel uninterruptible power supply systems or hybrid AC/DC microgrid applicat......This paper presents a theoretical study with experimental validation of a circulating-current suppression method for parallel operation of three-phase voltage source inverters (VSI), which may be suitable for modular parallel uninterruptible power supply systems or hybrid AC/DC microgrid......, and added into the conventional droop plus virtual impedance control. In the control architecture, the reference voltages of the inverters are generated by the primary control loop which consists of a droop control and a virtual impedance. The secondary control is used to compensate the voltage drop...
VISUALIZATION OF BIOLOGICAL TISSUE IMPEDANCE PARAMETERS
Directory of Open Access Journals (Sweden)
V. I. Bankov
2016-01-01
Full Text Available Objective. Investigation the opportunity for measurement of biological tissue impedance to visualize its parameters.Materials and methods. Studies were undertook on the experimental facility, consists of registrating measuring cell, constructed from flat inductors system, formed in oscillatory circuit, herewith investigated biological tissue is the part of this oscillatory circuit. An excitation of oscillatory circuit fulfilled by means of exciter inductor which forms impulse complex modulated electromagnetic field (ICM EMF. The measurement process and visualizations provided by set of certificated instruments: a digital oscillograph AKTAKOM ADS-2221MV, a digital generator АКТАКОМ AWG-4150 (both with software and a gauge RLC E7-22. Comparative dynamic studies of fixed volume and weight pig’s blood, adipose tissue, muscular tissue impedance were conducted by contact versus contactless methods. Contactless method in contrast to contact method gives opportunity to obtain the real morphological visualization of biological tissue irrespective of their nature.Results. Comparison of contact and contactless methods of impedance measurement shows that the inductance to capacitance ratio X(L / X(C was equal: 17 – for muscular tissue, 4 – for blood, 1 – for adipose tissue. It demonstrates the technical correspondence of both impedance registration methods. If propose the base relevance of X (L and X (C parameters for biological tissue impedance so contactless measurement method for sure shows insulating properties of adipose tissue and high conductivity for blood and muscular tissue in fixed volume-weight parameters. Registration of biological tissue impedance complex parameters by contactless method with the help of induced ICM EMF in fixed volume of biological tissue uncovers the most important informative volumes to characterize morphofunctional condition of biological tissue namely X (L / X (C.Conclusion. Contactless method of biological
High voltage wide range marx generator design and construction
International Nuclear Information System (INIS)
Thompson, J.E.
1976-01-01
A wide range, long pulse, Marx generator has been designed and constructed for the purpose of exciting a thermionic electron gun utilized for quasi-cw gas laser medium ionization. The Marx generator has been specifically designed to operate over a voltage range variable from 100 kV to 200 kV into a resistive load of between 83 kΩ and open circuit. This wide operating range, both in voltage and load impedance, was obtained using interstage coupling capacitors to assure overvoltage and subsequent breakdown of the three element spark gap switches used. This paper will discuss the motivation and specific application for the Marx generator and will present the relevant design procedure with particular emphasis on the interstage coupling and triggering techniques employed. Experimental data regarding the measured Marx generator performance will also be presented
Review of Stratum Corneum Impedance Measurement in Non-Invasive Penetration Application
Directory of Open Access Journals (Sweden)
Fei Lu
2018-03-01
Full Text Available Due to advances in telemedicine, mobile medical care, wearable health monitoring, and electronic skin, great efforts have been directed to non-invasive monitoring and treatment of disease. These processes generally involve disease detection from interstitial fluid (ISF instead of blood, and transdermal drug delivery. However, the quantitative extraction of ISF and the level of drug absorption are greatly affected by the individual’s skin permeability, which is closely related to the properties of the stratum corneum (SC. Therefore, measurement of SC impedance has been proposed as an appropriate way for assessing individual skin differences. In order to figure out the current status and research direction of human SC impedance detection, investigations regarding skin impedance measurement have been reviewed in this paper. Future directions are concluded after a review of impedance models, electrodes, measurement methods and systems, and their applications in treatment. It is believed that a well-matched skin impedance model and measurement method will be established for clinical and point-of care applications in the near future.
Laboratory scale tests of electrical impedence tomography
International Nuclear Information System (INIS)
Binley, A; Daily, W; LaBredcque, D; Ramirez, A.
1998-01-01
Electrical impedance tomographs (magnitude and phase) of known, laboratory-scale targets are reported. Three methods are used to invert electrical impedance data and their tomographs compared. The first method uses an electrical resistance tomography (ERT) algonthm (designed for DC resistivity inversion) to perform impedance magnitude inversion and a linearized perturbation approach (PA) to invert the imaginary part. The second approximate method compares ERT magnitude inversions at two frequencies and uses the frequency effect (FE) to compute phase tomographs. The third approach, electrrcal impedance tomography (EIT), employs fully complex algebra to account for the real and imaginary components of electrical impedance data. The EIT approach provided useful magnitude and phase images for the frequency range of 0.0625 to 64 Hz; images for higher frequencies were not reliable. Comparisons of the ERT and EIT magnitude images show that both methods provided equivalent results for the water blank, copper rod and PVC rod targets. The EIT magnitude images showed better spatial resolutron for a sand-lead mixture target. Phase images located anomalies of both high and low contrast IP and provided better spatial resolution than the magnitude images. When IP was absent from the data, the EIT algorithm reconstructed phase values consistent with the data noise levels
Impedance Spectroscopy of Dielectrics and Electronic Conductors
DEFF Research Database (Denmark)
Bonanos, Nikolaos; Pissis, Polycarpos; Macdonald, J. Ross
2013-01-01
Impedance spectroscopy is used for the characterization of materials, such as electroceramics, solid and liquid electrochemical cells, dielectrics and also fully integrated devices, such as fuel cells. It consists of measuring the electrical impedance - or a closely related property, such as admi......Impedance spectroscopy is used for the characterization of materials, such as electroceramics, solid and liquid electrochemical cells, dielectrics and also fully integrated devices, such as fuel cells. It consists of measuring the electrical impedance - or a closely related property......, such as admittance or dielectric constant - as a function of frequency and comparing the results with expectations based on physical, chemical, and microstructural models. This article reviews the principles and practical aspects of the technique, the representations of the results, the analysis of data......, and procedures for the correction of measurement errors. The applications of impedance spectroscopy are illustrated with examples from electroceramics and polymer-based dielectric systems. The way in which the technique is applied to the two classes of materials is compared with reference to the different models...
Rf Discharge Impedance Measurements Using a New Method to Determine the Stray Impedances
Bakker, L.P.; Kroesen, G.M.W.; Hoog, de F.J.
1999-01-01
The impedance of a capacitively coupled radio frequency discharge in a tubular fluorescent lamp filled with neon and mercury is measured. The stray impedances in the electrical network are determined using a new method that requires no extra instruments. The reflection of power is used to determine
Czech Academy of Sciences Publication Activity Database
Jurák, Pavel; Halámek, Josef; Vondra, Vlastimil; Viščor, Ivo; Lipoldová, J.; Plachý, M.
2010-01-01
Roč. 37, - (2010), s. 1051-1054 ISSN 0276-6574 R&D Projects: GA AV ČR IAA200650801 Institutional research plan: CEZ:AV0Z20650511 Keywords : respiratory effect * thoracic impedance signal * impedance cardiography Subject RIV: JA - Electronics ; Optoelectronics, Electrical Engineering http://cinc.mit.edu/archives/2010/pdf/1051.pdf
Temporary over voltages in the high voltage networks
International Nuclear Information System (INIS)
Vukelja, Petar; Naumov, Radomir; Mrvic, Jovan; Minovski, Risto
2001-01-01
The paper treats the temporary over voltages that may arise in the high voltage networks as a result of: ground faults, loss of load, loss of one or two phases and switching operation. Based on the analysis, the measures for their limitation are proposed. (Original)
Directory of Open Access Journals (Sweden)
Shuyu Lin
2017-02-01
Full Text Available The input electrical impedance behaves as a capacitive when a piezoelectric transducer is excited near its resonance frequency. In order to increase the energy transmission efficiency, a series or parallel inductor should be used to compensate the capacitive impedance of the piezoelectric transducer. In this paper, the effect of the series matching inductor on the electromechanical characteristics of the piezoelectric transducer is analyzed. The dependency of the resonance/anti-resonance frequency, the effective electromechanical coupling coefficient, the electrical quality factor and the electro-acoustical efficiency on the matching inductor is obtained. It is shown that apart from compensating the capacitive impedance of the piezoelectric transducer, the series matching inductor can also change the electromechanical characteristics of the piezoelectric transducer. When series matching inductor is increased, the resonance frequency is decreased and the anti-resonance unchanged; the effective electromechanical coupling coefficient is increased. For the electrical quality factor and the electroacoustic efficiency, the dependency on the matching inductor is different when the transducer is operated at the resonance and the anti-resonance frequency. The electromechanical characteristics of the piezoelectric transducer with series matching inductor are measured. It is shown that the theoretically predicted relationship between the electromechanical characteristics and the series matching inductor is in good agreement with the experimental results.
Lin, Shuyu; Xu, Jie
2017-02-10
The input electrical impedance behaves as a capacitive when a piezoelectric transducer is excited near its resonance frequency. In order to increase the energy transmission efficiency, a series or parallel inductor should be used to compensate the capacitive impedance of the piezoelectric transducer. In this paper, the effect of the series matching inductor on the electromechanical characteristics of the piezoelectric transducer is analyzed. The dependency of the resonance/anti-resonance frequency, the effective electromechanical coupling coefficient, the electrical quality factor and the electro-acoustical efficiency on the matching inductor is obtained. It is shown that apart from compensating the capacitive impedance of the piezoelectric transducer, the series matching inductor can also change the electromechanical characteristics of the piezoelectric transducer. When series matching inductor is increased, the resonance frequency is decreased and the anti-resonance unchanged; the effective electromechanical coupling coefficient is increased. For the electrical quality factor and the electroacoustic efficiency, the dependency on the matching inductor is different when the transducer is operated at the resonance and the anti-resonance frequency. The electromechanical characteristics of the piezoelectric transducer with series matching inductor are measured. It is shown that the theoretically predicted relationship between the electromechanical characteristics and the series matching inductor is in good agreement with the experimental results.
Matching with transfer matrices
International Nuclear Information System (INIS)
Perez-Alvarez, R.; Velasco, V.R.; Garcia-Moliner, F.; Rodriguez-Coppola, H.
1987-10-01
An ABC configuration - which corresponds to various systems of physical interest, such as a barrier or a quantum well - is studied by combining a surface Green function matching analysis of the entire system with a description of the intermediate (B) region in terms of a transfer matrix in the sense of Mora et al. (1985). This hybrid approach proves very useful when it is very difficult to construct the corresponding Green function G B . An application is made to the calculation of quantised subband levels in a parabolic quantum well. Further possibilities of extension of this approach are pointed out. (author). 27 refs, 1 tab
Coaxial pulse matching transformer
International Nuclear Information System (INIS)
Ledenev, V.V.; Khimenko, L.T.
1986-01-01
This paper describes a coaxial pulse matching transformer with comparatively simple design, increased mechanical strength, and low stray inductance. The transformer design makes it easy to change the turns ratio. The circuit of the device and an expression for the current multiplication factor are presented; experiments confirm the efficiency of the transformer. Apparatus with a coaxial transformer for producing high-power pulsed magnetic fields is designed (current pulses of 1-10 MA into a load and a natural frequency of 100 kHz)
Energy Technology Data Exchange (ETDEWEB)
Lee, Ki-Soo; Sun, Yang-Kook; Kim, Dong-Won [Department of Chemical Engineering, Hanyang University, Seungdong-gu, Seoul 133-791 (Korea); Noh, Jaegeun [Department of Chemistry, Hanyang University, Seungdong-gu, Seoul 133-791 (Korea); Song, Kwang Soup [Advanced Medical Device Center, Korea Electrotechnology, Research Institute, Ansan, Gyeonggi-do 426-170 (Korea)
2009-10-15
This study demonstrates that the addition of thiophene improves the cycle life of lithium-ion cells at high voltage. Electrochemical impedance spectroscopy results suggest that addition of thiophene significantly suppresses the increase of the charge transfer resistance that occurs during cycling up to high voltage. Differential scanning calorimetric studies showed that the thermal stability of fully charged LiCoO{sub 2} cathode was also enhanced in the presence of thiophene. (author)
Impedance measurements of components for the ALS
International Nuclear Information System (INIS)
Corlett, J.N.; Rimmer, R.A.
1993-05-01
The high current and short bunch length of the ALS beam make the machine susceptible to beam instabilities over a frequency range extending to 13 GHz and beyond. All components of the storage ring have been carefully designed to minimize the impedance presented to the beam, and assemblies have been laid out to avoid resonant enclosures between components. Novel bellows shields allowing considerable mechanical movement while maintaining a low impedance are described. Results are presented of impedance measurements of ALS components and assemblies of components, using a precision coaxial wire technique in frequency domain, extending to frequencies beyond cut-off. All measurements were performed at the Lambertson Beam Electrodynamics Laboratory of the Center for Beam Physics at LBL
Bunch length and impedance measurements in SPEAR
International Nuclear Information System (INIS)
Bane, K.; Donald, M.; Hofmann, A.; Jowett, J.; Lockman, W.; Morton, P.; Stege, R.; Spence, W.; Wilson, P.
1988-05-01
Subsequent to an extensive smoothing of the vacuum chamber a comprehensive study of the SPEAR impedance was undertaken. Bunch length, synchrotron quadrupole mode frequency, and parasitic mode loss were measured as functions of beam current. The results showed that, although the gross longitudinal impedance had indeed been reduced, the 'capacitive' component had also decreased relative to the 'inductive'--to the extent that previously compensated potential well distortion now induced bunch lengthening at low currents, and the turbulent threshold had actually been lowered. A specially designed multi-cell disc-loaded 'capacitor' cavity was shown to be capable of removing this effect by restoring the original compensation. A model of the new SPEAR impedance is also obtained. 7 refs., 6 figs
Bunch length and impedance measurements in SPEAR
International Nuclear Information System (INIS)
Bane, K.; Donald, M.; Morton, P.; Stege, R.; Spence, W.; Wilson, P.; Hofmann, A.; Jowett, J.; Lockman, W.
1988-01-01
Subsequent to an extensive smoothing of the vacuum chamber a comprehensive study of the SPEAR impedance was undertaken. Bunch length, synchrotron quadrupole mode frequency, and parasitic mode loss were measured as functions of beam current. This paper shows that although the gross longitudinal impedance had indeed been reduced, the capacitive component had also decreased relative to the inductive - to the extent that previously compensated potential well distortion now induced bunch lengthening at low currents, and the turbulent threshold had actually been lowered. A specially designed multi-cell disc-loaded capacitor cavity was shown to be capable of removing this effect by restoring the original compensation. A model of the new SPEAR impedance is also obtained
Wavelet analysis of the impedance cardiogram waveforms
Podtaev, S.; Stepanov, R.; Dumler, A.; Chugainov, S.; Tziberkin, K.
2012-12-01
Impedance cardiography has been used for diagnosing atrial and ventricular dysfunctions, valve disorders, aortic stenosis, and vascular diseases. Almost all the applications of impedance cardiography require determination of some of the characteristic points of the ICG waveform. The ICG waveform has a set of characteristic points known as A, B, E ((dZ/dt)max) X, Y, O and Z. These points are related to distinct physiological events in the cardiac cycle. Objective of this work is an approbation of a new method of processing and interpretation of the impedance cardiogram waveforms using wavelet analysis. A method of computer thoracic tetrapolar polyrheocardiography is used for hemodynamic registrations. Use of original wavelet differentiation algorithm allows combining filtration and calculation of the derivatives of rheocardiogram. The proposed approach can be used in clinical practice for early diagnostics of cardiovascular system remodelling in the course of different pathologies.
Wavelet analysis of the impedance cardiogram waveforms
International Nuclear Information System (INIS)
Podtaev, S; Stepanov, R; Dumler, A; Chugainov, S; Tziberkin, K
2012-01-01
Impedance cardiography has been used for diagnosing atrial and ventricular dysfunctions, valve disorders, aortic stenosis, and vascular diseases. Almost all the applications of impedance cardiography require determination of some of the characteristic points of the ICG waveform. The ICG waveform has a set of characteristic points known as A, B, E ((dZ/dt) max ) X, Y, O and Z. These points are related to distinct physiological events in the cardiac cycle. Objective of this work is an approbation of a new method of processing and interpretation of the impedance cardiogram waveforms using wavelet analysis. A method of computer thoracic tetrapolar polyrheocardiography is used for hemodynamic registrations. Use of original wavelet differentiation algorithm allows combining filtration and calculation of the derivatives of rheocardiogram. The proposed approach can be used in clinical practice for early diagnostics of cardiovascular system remodelling in the course of different pathologies.
Coupling slots without shunt impedance drop
International Nuclear Information System (INIS)
Balleyguier, P.
1996-01-01
It is well known that coupling slots between adjacent cells in a π-mode structure reduce shunt impedance per unit length with respect to single cell cavities. To design optimized coupling slots, one has to answer the following question: for a given coupling factor, what shape, dimension, position and number of slots lead to the lowest shunt impedance drop? A numerical study using the 3D code MAFIA has been carried out. The aim was to design the 352 MHz cavities for the high intensity proton accelerator of the TRISPAL project. The result is an unexpected set of four 'petal' slots. Such slots should lead to a quasi-negligible drop in shunt impedance: about -1% on average, for particle velocity from 0.4 c to 0.8 c. (author)
Measurements of the dynamic input impedance of a dc SQUID
International Nuclear Information System (INIS)
Hilbert, C.; Clarke, J.
1985-01-01
The impedance of a circuit coupled magnetically via a mutual inductance M/sub i/ to a dc SQUID of geometric inductance L is modified by the dynamic input impedance of the SQUID, which can be characterized by the flux-to-current transfer function J/sub Phi/approx. =partialJ/partialPhi; J is the current circulating in the SQUID loop and ∫ is the flux applied to the loop. At the same time, the SQUID is modified by the presence of the input circuit in the lumped circuit approximation, one expects its inductance to be reduced to L'(1-α/sub e/ 2 )L, where α/sub e/ is an effective coupling coefficient. Calculations of J/sub Phi/ using an analog simulator are described and presented in the form of a dynamic inductance L and a dynamic resistance R versus bias current I and Phi. Experimental measurements of L and R were made on a planar, thin-film SQUID tightly coupled to a spiral input coil that was connected in series with a capacitor C/sub i/ to form a resonant circuit. Thus, J/sub Phi/ was determined from the change in the resonant frequency and quality factor of this circuit as a function of I and Phi. At low bias currents (low Josephson frequencies) the measured values of L were in reasonable agreement with values simulated for the reduced SQUID, while at higher bias currents (higher Josephson frequencies) the measured values were in better agreement with values simulated for the unscreened SQUID. Similar conclusions were reached in the comparison of the experimental and simulated values of the flux-to-voltage transfer function V/sub Phi/
Lactate threshold by muscle electrical impedance in professional rowers
Jotta, B.; Coutinho, A. B. B.; Pino, A. V.; Souza, M. N.
2017-04-01
Lactate threshold (LT) is one of the physiological parameters usually used in rowing sport training prescription because it indicates the transitions from aerobic to anaerobic metabolism. Assessment of LT is classically based on a series of values of blood lactate concentrations obtained during progressive exercise tests and thus has an invasive aspect. The feasibility of noninvasive LT estimative through bioelectrical impedance spectroscopy (BIS) data collected in thigh muscles during rowing ergometer exercise tests was investigated. Nineteen professional rowers, age 19 (mean) ± 4.8 (standard deviation) yr, height 187.3 ± 6.6 cm, body mass 83 ± 7.7 kg, and training experience of 7 ± 4 yr, were evaluated in a rowing ergometer progressive test with paired measures of blood lactate concentration and BIS in thigh muscles. Bioelectrical impedance data were obtained by using a bipolar method of spectroscopy based on the current response to a voltage step. An electrical model was used to interpret BIS data and to derive parameters that were investigated to estimate LT noninvasively. From the serial blood lactate measurements, LT was also determined through Dmax method (LTDmax). The zero crossing of the second derivative of kinetic of the capacitance electrode (Ce), one of the BIS parameters, was used to estimate LT. The agreement between the LT estimates through BIS (LTBIS) and through Dmax method (LTDmax) was evaluated using Bland-Altman plots, leading to a mean difference between the estimates of just 0.07 W and a Pearson correlation coefficient r = 0.85. This result supports the utilization of the proposed method based on BIS parameters for estimating noninvasively the lactate threshold in rowing.
Impedance spectroscopy study of N719-sensitized ZnO-based solar cells
Energy Technology Data Exchange (ETDEWEB)
Pauporté, Th., E-mail: thierry-pauporte@chimie-paristech.fr [Laboratoire d' Electrochimie, Chimie des Interfaces et Modélisation pour l' Energie, Ecole Nationale Supérieure de Chimie de Paris, 11 rue P. et M. Curie, 75231 Paris cedex 05 (France); Magne, C. [Laboratoire d' Electrochimie, Chimie des Interfaces et Modélisation pour l' Energie, Ecole Nationale Supérieure de Chimie de Paris, 11 rue P. et M. Curie, 75231 Paris cedex 05 (France); Saint-Gobain Recherche, 39 quai Lucien Lefranc, 93303 Aubervilliers Cedex (France)
2014-06-02
ZnO porous films prepared by electrodeposition and by sol–gel techniques have been sensitized by the N719 dye and used as photoelectrode in dye-sensitized solar cells. Similar stationary current–voltage characteristics have been measured for films having two drastically different morphologies and nanostructures. The solar cells have been studied by impedance spectroscopy over large frequency and applied voltage ranges in order to elucidate the inherent correlation between the photoelectrode structure and properties. The electrical response has been analyzed using an ad-hoc electrical model to extrapolate the electronic structure and charge carrier kinetic properties of the photoelectrodes. The two films exhibit very different density distributions of their trap states below the conduction band edge. Moreover, their electron lifetimes and transport times vary differently with the DOS due to different surface and conduction properties. The charge collection efficiency has been calculated by two different approaches, one being derived from first principles. Very high values are reported in all cases. Globally, the limitation of N719/ZnO solar cell performances is related to their open circuit voltage and their short circuit current. The latter point is discussed in the light of our impedance results and photoelectrode light absorption data. - Highlights: • Exponential trap state distribution • Electron transport and recombination depend on oxide preparation route. • Charge collection efficiency modeling • Very high charge collection efficiency in N719/ZnO dye-sensitized solar cells • ZnO cell performance limitations.
DEFF Research Database (Denmark)
Micallef, Alexander; Apap, Maurice; Spiteri-Staines, Cyril
2017-01-01
Optimization of the islanded and grid-connected operation of microgrids is important to achieve a high degree of reliability. In this paper, the authors consider the effect of current harmonics in single phase microgrids during both modes of operation. A detailed analysis of the effect of the out......Optimization of the islanded and grid-connected operation of microgrids is important to achieve a high degree of reliability. In this paper, the authors consider the effect of current harmonics in single phase microgrids during both modes of operation. A detailed analysis of the effect...... of the output impedance of the considered primary control loops on the harmonic output of the considered voltage source inverters is initially carried out. A virtual admittance loop is proposed to attenuate the current harmonic output in grid-connected operation that is generated due to the grid voltage...... distortion present at the point of common coupling (PCC) and due to local non-linear loads. This paper also considers the harmonic current sharing and resulting voltage harmonics at the PCC during islanded operation of the microgrid. A capacitive virtual impedance loop was implemented to improve the harmonic...
Optimal background matching camouflage.
Michalis, Constantine; Scott-Samuel, Nicholas E; Gibson, David P; Cuthill, Innes C
2017-07-12
Background matching is the most familiar and widespread camouflage strategy: avoiding detection by having a similar colour and pattern to the background. Optimizing background matching is straightforward in a homogeneous environment, or when the habitat has very distinct sub-types and there is divergent selection leading to polymorphism. However, most backgrounds have continuous variation in colour and texture, so what is the best solution? Not all samples of the background are likely to be equally inconspicuous, and laboratory experiments on birds and humans support this view. Theory suggests that the most probable background sample (in the statistical sense), at the size of the prey, would, on average, be the most cryptic. We present an analysis, based on realistic assumptions about low-level vision, that estimates the distribution of background colours and visual textures, and predicts the best camouflage. We present data from a field experiment that tests and supports our predictions, using artificial moth-like targets under bird predation. Additionally, we present analogous data for humans, under tightly controlled viewing conditions, searching for targets on a computer screen. These data show that, in the absence of predator learning, the best single camouflage pattern for heterogeneous backgrounds is the most probable sample. © 2017 The Authors.
Characterization of Porous WO3 Electrochromic Device by Electrochemical Impedance Spectroscopy
Directory of Open Access Journals (Sweden)
Chien Chon Chen
2013-01-01
Full Text Available This paper concerns the microstructure of the anodic tungsten oxide (WO3 and its use in an electrochromic (EC glass device. When voltages between 100 V and 160 V were applied to tungsten film for 1 h under 0.4 wt. % NaF electrolyte, porous WO3 film was formed. The film, which had a large surface area, was used as electrochromic film for EC glass. The average transmittance in a visible region of the spectrum for a 144 cm2 EC device was above 75% in the bleached state and below 40% in the colored state, respectively. Repeatability using of the colored/bleached cycles was tested good by a cyclic voltammograms method. The internal impedance values under colored and bleached states were detected and simulated using an electrical impedance spectra (EIS technique. The EC glass impedance characteristics were simulated using resistors, capacitors, and Warburg impedance. The ITO/WO3, WO3/electrolyte, electrolyte/NiO, and NiO/ITO interfaces can be simulated using a resistance capacitance (RC parallel circuits, and bulk materials such as the indium tin oxide (ITO and conducting wire can be simulated by using a series of resisters.
Simple Wide Frequency Range Impedance Meter Based on AD5933 Integrated Circuit
Directory of Open Access Journals (Sweden)
Chabowski Konrad
2015-03-01
Full Text Available As it contains elements of complete digital impedance meter, the AD5933 integrated circuit is an interesting solution for impedance measurements. However, its use for measurements in a wide range of impedances and frequencies requires an additional digital and analogue circuitry. This paper presents the design and performance of a simple impedance meter based on the AD5933 IC. Apart from the AD5933 IC it consists of a clock generator with a programmable prescaler, a novel DC offset canceller for the excitation signal based on peak detectors and a current to voltage converter with switchable conversion ratios. The authors proposed a simple method for choosing the measurement frequency to minimalize errors resulting from the spectral leakage and distortion caused by a lack of an anti-aliasing filter in the DDS generator. Additionally, a novel method for the AD5933 IC calibration was proposed. It consists in a mathematical compensation of the systematic error occurring in the argument of the value returned from the AD5933 IC as a result. The performance of the whole system is demonstrated in an exemplary measurement.
Electrochemical impedance characterization of FeSn2 electrodes for Li-ion batteries
International Nuclear Information System (INIS)
Chamas, M.; Lippens, P-E.; Jumas, J-C.; Hassoun, J.; Panero, S.; Scrosati, B.
2011-01-01
Highlights: → In this paper we study a tin based, FeSn 2 , high capacity lithium-alloying electrode. → The electrochemical performance of this electrode in lithium batteries is remarkably influenced by the current rate. → This aspect is investigated by electrochemical techniques such as galvanostatic cycling and impedance spectroscopy. → The results demonstrated that the good electrochemical behavior of the electrode at the higher currents is due to the formation of a stable solid electrolyte interphase (SEI) film. - Abstract: This work reports the electrochemical characterization of a micro-scale FeSn 2 electrode in a lithium battery. The electrode is proposed as anode material for advanced lithium ion batteries due to its characteristics of high capacity (500 mAh g -1 ) and low working voltage (0.6 V vs. Li). The electrochemical alloying process is studied by cyclic voltammetry and galvanostatic cycling while the interfacial properties are investigated by electrochemical impedance spectroscopy. The impedance measurements in combination with the galvanostatic cycling tests reveal relatively low overall impedance values and good electrochemical performance for the electrode, both in terms of delivered capacity and cycling stability, even at the higher C-rate regimes.
Electrochemical Impedance Spectroscopy Of Metal Alloys
Macdowell, L. G.; Calle, L. M.
1993-01-01
Report describes use of electrochemical impedance spectroscopy (EIS) to investigate resistances of 19 alloys to corrosion under conditions similar to those of corrosive, chloride-laden seaside environment of Space Transportation System launch site. Alloys investigated: Hastelloy C-4, C-22, C-276, and B-2; Inconel(R) 600, 625, and 825; Inco(R) G-3; Monel 400; Zirconium 702; Stainless Steel 304L, 304LN, 316L, 317L, and 904L; 20Cb-3; 7Mo+N; ES2205; and Ferralium 255. Results suggest electrochemical impedance spectroscopy used to predict corrosion performances of metal alloys.
Study of Body Composition by Impedance Analysis
González-Solís, J. L.; Vargas-Luna, M.; Sosa-Aquino, M.; Bernal-Alvarado, J.; Gutiérrez-Juárez, G.; Huerta-Franco, R.; Sanchis-Sabater, A.
2002-08-01
This work presents a set of impedance measurements and preliminary results on the analysis of body composition using impedance spectroscopy. This study is made using a pork meat sample and spectra from fat and flesh region were independently obtained using the same electrodes array. From these measurements, and theoretical considerations, it is possible to explain the behavior of the composite sample flesh-fat-flesh and, fitting the electrical parameters of the model, it shows the plausibility of a physical and quantitative application to human corporal composition.
Real-Time Thevenin Impedance Computation
DEFF Research Database (Denmark)
Sommer, Stefan Horst; Jóhannsson, Hjörtur
2013-01-01
operating state, and strict time constraints are difficult to adhere to as the complexity of the grid increases. Several suggested approaches for real-time stability assessment require Thevenin impedances to be determined for the observed system conditions. By combining matrix factorization, graph reduction......, and parallelization, we develop an algorithm for computing Thevenin impedances an order of magnitude faster than previous approaches. We test the factor-and-solve algorithm with data from several power grids of varying complexity, and we show how the algorithm allows realtime stability assessment of complex power...
Current-voltage-temperature characteristics of DNA origami
Energy Technology Data Exchange (ETDEWEB)
Bellido, Edson P; Bobadilla, Alfredo D; Rangel, Norma L; Seminario, Jorge M [Department of Chemical Engineering, Texas A and M University, College Station, TX 77843 (United States); Zhong Hong; Norton, Michael L [Department of Chemistry, Marshall University, Huntington, WV 25755 (United States); Sinitskii, Alexander [Department of Chemistry, Rice University, Houston, TX 77005 (United States)
2009-04-29
The temperature dependences of the current-voltage characteristics of a sample of triangular DNA origami deposited in a 100 nm gap between platinum electrodes are measured using a probe station. Below 240 K, the sample shows high impedance, similar to that of the substrate. Near room temperature the current shows exponential behavior with respect to the inverse of temperature. Sweep times of 1 s do not yield a steady state; however sweep times of 450 s for the bias voltage secure a steady state. The thermionic emission and hopping conduction models yield similar barriers of {approx}0.7 eV at low voltages. For high voltages, the hopping conduction mechanism yields a barrier of 0.9 eV and the thermionic emission yields 1.1 eV. The experimental data set suggests that the dominant conduction mechanism is hopping in the range 280-320 K. The results are consistent with theoretical and experimental estimates of the barrier for related molecules.
Current-voltage-temperature characteristics of DNA origami
International Nuclear Information System (INIS)
Bellido, Edson P; Bobadilla, Alfredo D; Rangel, Norma L; Seminario, Jorge M; Zhong Hong; Norton, Michael L; Sinitskii, Alexander
2009-01-01
The temperature dependences of the current-voltage characteristics of a sample of triangular DNA origami deposited in a 100 nm gap between platinum electrodes are measured using a probe station. Below 240 K, the sample shows high impedance, similar to that of the substrate. Near room temperature the current shows exponential behavior with respect to the inverse of temperature. Sweep times of 1 s do not yield a steady state; however sweep times of 450 s for the bias voltage secure a steady state. The thermionic emission and hopping conduction models yield similar barriers of ∼0.7 eV at low voltages. For high voltages, the hopping conduction mechanism yields a barrier of 0.9 eV and the thermionic emission yields 1.1 eV. The experimental data set suggests that the dominant conduction mechanism is hopping in the range 280-320 K. The results are consistent with theoretical and experimental estimates of the barrier for related molecules.
Computer controlled high voltage system
Energy Technology Data Exchange (ETDEWEB)
Kunov, B; Georgiev, G; Dimitrov, L [and others
1996-12-31
A multichannel computer controlled high-voltage power supply system is developed. The basic technical parameters of the system are: output voltage -100-3000 V, output current - 0-3 mA, maximum number of channels in one crate - 78. 3 refs.
A Voltage Quality Detection Method
DEFF Research Database (Denmark)
Chen, Zhe; Wei, Mu
2008-01-01
This paper presents a voltage quality detection method based on a phase-locked loop (PLL) technique. The technique can detect the voltage magnitude and phase angle of each individual phase under both normal and fault power system conditions. The proposed method has the potential to evaluate various...
Stability of high current diode under 100-nanosecond-pulse voltage
International Nuclear Information System (INIS)
Lai Dingguo; Qiu Aici; Zhang Yongmin; Huang Jianjun; Ren Shuqing; Yang Li
2012-01-01
Stability of high current diode under pulse voltage with 80 ns and 34 ns rise time was studied on the flash Ⅱ accelerator. Influence of rise time of diode voltage on startup time and cathode emission uniformity and repeatability of diode impedance was analyzed by comparing the experimental results with numerically simulated results, and the influence mechanism was discussed. The startup time of diode increases with the increasing of rise time of voltage, and the repeatability of diode impedance decreases. Discal plane cathode is prone to emit rays intensely in the center area, the time that plasma covers the surface of the cathode increases and the shielding effect has more impact on cathode emission according to the increase of rise time. Local intense emission on the cathode increases expansion speed of plasma and reduces the effective emission area. The stability of characteristic impedance of diode under a pulse voltage with slow rise time is decreased by the combined action of expansion speed of plasma and the effective emission area. (authors)
Large impedances and Majorana bound states in superconducting circuits
International Nuclear Information System (INIS)
Ulrich, Jascha
2017-01-01
Superconducting circuits offer the opportunity to study quantum mechanics on mesoscopic scales unimpeded by dissipation. This fact and the nonlinearity of the Josephson inductance make it possible to use superconducting circuits as artificial atoms whose long-lived states can be selectively addressed and studied. A pronounced nonlinearity of the energy spectrum, however, requires quantum fluctuations of the flux across the Josephson junction which are large on the scale of the superconducting flux quantum Φ Q =h/2e. This implies charge fluctuations below the single Cooper-pair limit via flux-charge duality. The localization of charge leads to a strong susceptibility to interactions with charges in the environment which has motivated the search for schemes to decouple charges from their environment. This thesis is concerned with theoretical challenges arising from two complementary approaches to this problem: the realization of large impedances and the fractionalization of electrons by means of Majorana bound states. In recent years, the decoupling of charges from the environment through reactive large impedances, so-called ''superinductances'' L, has attracted much interest. These inductances feature small parasitic capacitance C such that the characteristic impedance √(L/C) is much larger than the superconducting resistance quantum R Q =h/4e 2 . Superinductances have various applications ranging from qubit designs such as the 0-π qubit or the fluxonium to impedance matching, Bloch oscillations and the stabilization of phase slips in superconducting nanowires. Although there exists a well-established formalism for the quantization of superconducting circuits in terms of node fluxes, this formalism is ill-suited for the description of fast flux transport with localized charges in large-impedance environments. In particular, the nonlinear capacitive behavior of phase slip junctions cannot be modeled in a straightforward way using node fluxes
Voltage Controlled Dynamic Demand Response
DEFF Research Database (Denmark)
Bhattarai, Bishnu Prasad; Bak-Jensen, Birgitte; Mahat, Pukar
2013-01-01
Future power system is expected to be characterized by increased penetration of intermittent sources. Random and rapid fluctuations in demands together with intermittency in generation impose new challenges for power balancing in the existing system. Conventional techniques of balancing by large...... central or dispersed generations might not be sufficient for future scenario. One of the effective methods to cope with this scenario is to enable demand response. This paper proposes a dynamic voltage regulation based demand response technique to be applied in low voltage (LV) distribution feeders....... An adaptive dynamic model has been developed to determine composite voltage dependency of an aggregated load on feeder level. Following the demand dispatch or control signal, optimum voltage setting at the LV substation is determined based on the voltage dependency of the load. Furthermore, a new technique...
Transient voltage oscillations in coils
International Nuclear Information System (INIS)
Chowdhuri, P.
1985-01-01
Magnet coils may be excited into internal voltage oscillations by transient voltages. Such oscillations may electrically stress the magnet's dielectric components to many times its normal stress. This may precipitate a dielectric failure, and the attendant prolonged loss of service and costly repair work. Therefore, it is important to know the natural frequencies of oscillations of a magnet during the design stage, and to determine whether the expected switching transient voltages can excite the magnet into high-voltage internal oscillations. The series capacitance of a winding significantly affects its natural frequencies. However, the series capacitance is difficult to calculate, because it may comprise complex capacitance network, consisting of intra- and inter-coil turn-to-turn capacitances of the coil sections. A method of calculating the series capacitance of a winding is proposed. This method is rigorous but simple to execute. The time-varying transient voltages along the winding are also calculated
Analysis of Magnetically-Coupled Impedance Source Three-Phase Four-Switch Inverters
DEFF Research Database (Denmark)
Li, Kerui; Abdelhakim, Ahmed; Yang, Yongheng
2017-01-01
In this paper, magnetically-coupled impedance source (MCIS) three-phase four-switch inverters are introduced for renewable energy applications. This inverter utilizes two single-phase MCIS inverters to obtain a three-phase output, where less component-count is achieved in order to reduce the inve......In this paper, magnetically-coupled impedance source (MCIS) three-phase four-switch inverters are introduced for renewable energy applications. This inverter utilizes two single-phase MCIS inverters to obtain a three-phase output, where less component-count is achieved in order to reduce...... the inverter volume. On the other hand, as a result of the reduced number of switches, the proposed inverter suffers from higher voltage stresses. The MCIS inverters are first reviewed. Then, the modulation and operation principle of the MCIS three-phase inverter topology are introduced before sizing...
DEFF Research Database (Denmark)
Mosbæk, Rasmus Rode; Hjelm, Johan; Barfod, R.
2013-01-01
As solid oxide fuel cell (SOFC) technology is moving closer to a commercial break through, lifetime limiting factors, and methods to measure the “state-of-health” of operating cells and stacks are becoming of increasing interest. This requires application of advanced methods for detailed...... electrochemical characterization during operation. An experimental stack with low ohmic resistance from Topsoe Fuel Cell A/S was characterized in detail using electrochemical impedance spectroscopy (EIS). An investigation of the optimal geometrical placement of the current feeds and voltage probes was carried out...... with hydrogen as fuel with 52% fuel utilization and constant current load (0.2 A cm–2) at 750 °C. Stack interconnects were coated with six different coatings to prevent chromium poisoning on the cathode side. Four repeating units (RUs) with different coatings were selected for detailed impedance analysis. EIS...
Simultaneous reconstruction of shape and generalized impedance functions in electrostatic imaging
International Nuclear Information System (INIS)
Cakoni, Fioralba; Hu, Yuqing; Kress, Rainer
2014-01-01
Determining the geometry and the physical nature of an inclusion within a conducting medium from voltage and current measurements on the accessible boundary of the medium can be modeled as an inverse boundary value problem for the Laplace equation subject to appropriate boundary conditions on the inclusion. We continue the investigations on the particular inverse problem with a generalized impedance condition started in Cakoni and Kress (2013 Inverse Problems 29 015005) by presenting an inverse algorithm for the simultaneous reconstruction of both the shape of the inclusion and the two impedance functions via a boundary integral equation approach. In addition to describing the reconstruction algorithm and illustrating its feasibility by numerical examples we also provide some extensions to the uniqueness results in Cakoni and Kress (2013 Inverse Problems 29 015005). (paper)
Job Searchers, Job Matches and the Elasticity of Matching
Broersma, L.; van Ours, J.C.
1998-01-01
This paper stresses the importance of a specification of the matching function in which the measure of job matches corresponds to the measure of job searchers. In many empirical studies on the matching function this requirement has not been fulfilled because it is difficult to find information about
Mark, T A; Gallistel, C R
1994-01-01
Rats responded on concurrent variable interval schedules of brain stimulation reward in 2-trial sessions. Between trials, there was a 16-fold reversal in the relative rate of reward. In successive, narrow time windows, the authors compared the ratio of the times spent on the 2 levers to the ratio of the rewards received. Time-allocation ratios tracked wide, random fluctuations in the reward ratio. The adjustment to the midsession reversal in relative rate of reward was largely completed within 1 interreward interval on the leaner schedule. Both results were unaffected by a 16-fold change in the combined rates of reward. The large, rapid, scale-invariant shifts in time-allocation ratios that underlie matching behavior imply that the subjective relative rate of reward can be determined by a very few of the most recent interreward intervals and that this estimate can directly determine the ratio of the expected stay durations.
Electromagnetic wave matching device
International Nuclear Information System (INIS)
Hirata, Yosuke; Mitsunaka, Yoshika; Hayashi, Ken-ichi; Ito, Yasuyuki.
1997-01-01
The present invention provides an electromagnetic wave matching capable of reducing a cost for the transmission system in a system of using electromagnetic waves for plasma heating of a thermonuclear reactor. Namely, incident electromagnetic waves are reflected by using a plurality of phase correction mirrors. The reflected electromagnetic waves are connected to an external transmission system through an exit. The phase correction mirrors have such a shape to receive a plurality of beam-like electromagnetic waves and output electromagnetic waves by the number different from the number of the received electromagnetic wave beams having a predetermined distribution. Further, at least two of the phase correction mirrors have such a shape to change the phase of the electromagnetic waves beams incident to the reflection surface of the phase correction mirrors by a predetermined amount corresponding to the position of the reflection surface. Then, the cost for transmission system can greatly be reduced. (I.S.)
Rong, Taotao; Yang, Lin-An; Yang, Lin; Hao, Yue
2018-01-01
In this work, we report an investigation of resonant tunneling diodes (RTDs) with lattice-matched and polarization-matched AlInN/GaN heterostructures using the numerical simulation. Compared with the lattice-matched AlInN/GaN RTDs, the RTDs based on polarization-matched AlInN/GaN hetero-structures exhibit symmetrical conduction band profiles due to eliminating the polarization charge discontinuity, which achieve the equivalence of double barrier transmission coefficients, thereby the relatively high driving current, the high symmetry of current density, and the high peak-to-valley current ratio (PVCR) under the condition of the positive and the negative sweeping voltages. Simulations show that the peak current density approaches 1.2 × 107 A/cm2 at the bias voltage of 0.72 V and the PVCR approaches 1.37 at both sweeping voltages. It also shows that under the condition of the same shallow energy level, when the trap density reaches 1 × 1019 cm-3, the polarization-matched RTDs still have acceptable negative differential resistance (NDR) characteristics, while the NDR characteristics of lattice-matched RTDs become irregular. After introducing the deeper energy level of 1 eV into the polarization-matched and lattice-matched RTDs, 60 scans are performed under the same trap density. Simulation results show that the degradation of the polarization-matched RTDs is 22%, while lattice-matched RTDs have a degradation of 55%. It can be found that the polarization-matched RTDs have a greater defect tolerance than the lattice-matched RTDs, which is beneficial to the available manufacture of actual terahertz RTD devices.
Impedance analysis on organic ultrathin layers
Energy Technology Data Exchange (ETDEWEB)
Bom, Sidhant; Wagner, Veit [Jacobs University Bremen, School of Engineering and Science, Campus Ring 8, 28759 Bremen (Germany)
2008-07-01
Impedance spectroscopy is a standard technique for thin film analysis to obtain important information as thicknesses, diffusion properties of mobile ions and leakage currents. The measured electrical impedance of a sample is modeled by a physical equivalent circuit of resistors and capacitors. In the present work this information is obtained as a function of frequency also for ultrathin organic layers in the monolayer regime. A series of semiconducting and insulating polymers (regioregular poly-3-hexylthiophene (rr-P3HT), polymethylmethacrylate (PMMA)) and self assembled monolayers (octadecyltrichlorosilane (OTS), hexamethyldisilazane (HMDS), thiolated phospholipids) were deposited either on highly n-doped silicon wafers or on gold surfaces. E.g. ultrathin layers were obtained by dip coating a silicon wafer in rr-P3HT solution in chloroform. The thickness of 2 nm determined for this system by impedance measurement agrees well with the atomic force microscopy analysis and corresponds to a single layer of polymer chains. The leakage current is seen as an ohmic contribution at low frequencies and allows a systematic optimization of process parameters. In summary, impedance spectroscopy allows very fast and convenient analysis of thin organic layers even down to the monolayer regime.
Impedances in lossy elliptical vacuum chambers
International Nuclear Information System (INIS)
Piwinski, A.
1994-04-01
The wake fields of a bunched beam caused by the resistivity of the chamber walls are investigated for a vacuum chamber with elliptical cross section. The longitudinal and transverse impedances are calculated for arbitrary energies and for an arbitrary position of the beam in the chamber. (orig.)
Electrical impedance spectroscopy and diagnosis of tendinitis
International Nuclear Information System (INIS)
Yoon, Kisung; Lee, Kyeong Woo; Kim, Sang Beom; Lee, Jong Hwa; Han, Tai Ryoon; Jung, Dong Keun; Roh, Mee Sook
2010-01-01
There have been a number of studies that investigate the usefulness of bioelectric signals in diagnoses and treatment in the medical field. Tendinitis is a musculoskeletal disorder with a very high rate of occurrence. This study attempts to examine whether electrical impedance spectroscopy (EIS) can detect pathological changes in a tendon and find the exact location of the lesion. Experimental tendinitis was induced by injecting collagenase into one side of the patellar tendons in rabbits, while the other side was used as the control. After measuring the impedance in the tendinitis and intact tendon tissue, the dissipation factor was computed. The real component of impedance and the dissipation factor turned out to be lower in tendinitis than in intact tissues. Moreover, the tendinitis dissipation factor spectrum showed a clear difference from that of the intact tendon, indicating its usefulness as a tool for detecting the location of the lesion. Pathologic findings from the tissues that were obtained after measuring the impedance confirmed the presence of characteristics of tendinitis. In conclusion, EIS is a useful method for diagnosing tendinitis and detecting the lesion location in invasive treatment
Electrical Impedance Tomography Technology (EITT) Project
Oliva-Buisson, Yvette J.
2014-01-01
Development of a portable, lightweight device providing two-dimensional tomographic imaging of the human body using impedance mapping. This technology can be developed to evaluate health risks and provide appropriate medical care on the ISS, during space travel and on the ground.
Impedance and collective effects in the KEKB
Energy Technology Data Exchange (ETDEWEB)
Chin, Yongho [National Lab. for High Energy Physics, Tsukuba, Ibaraki (Japan); Oide, Katsunobu
1996-08-01
This paper focuses on beam instabilities due to single-beam collective effects, impedances from various beamline elements, ion trapping, photo-electrons, and other issues in the KEKB. We will also discuss the power deposition generated by a beam in the form of the Higher-Order-Mode (HOM) losses by interacting with its surroundings. (author)
Bioelectrical impedance analysis--part I
DEFF Research Database (Denmark)
Kyle, Ursula G; Bosaeus, Ingvar; De Lorenzo, Antonio D
2004-01-01
The use of bioelectrical impedance analysis (BIA) is widespread both in healthy subjects and patients, but suffers from a lack of standardized method and quality control procedures. BIA allows the determination of the fat-free mass (FFM) and total body water (TBW) in subjects without significant...
Battery impedance spectroscopy using bidirectional grid connected ...
Indian Academy of Sciences (India)
Shimul Kumar Dam
Keywords. Impedance spectroscopy; grid connection; battery converter; state of charge; health monitoring. 1. Introduction .... the load should be within the safe range of operation specified by the ... A split capacitor damping scheme is adopted here as shown in ...... spectroscopy testing on the Advanced Technology Devel-.
Introduction to impedance for short relativistic bunches
International Nuclear Information System (INIS)
Morton, P.L.
1993-02-01
The purpose of this paper is to introduce the concept of impedance to calculate the wake field forces left behind by a short bunch which travels at relativistic speed through a structure with discontinuities. We will try to be as intuitive as possible and leave the more rigorous derivations to the second paper on this subject by J. Wang
Impedance Control of a Redundant Parallel Manipulator
DEFF Research Database (Denmark)
Méndez, Juan de Dios Flores; Schiøler, Henrik; Madsen, Ole
2017-01-01
This paper presents the design of Impedance Control to a redundantly actuated Parallel Kinematic Manipulator. The proposed control is based on treating each limb as a single system and their connection through the internal interaction forces. The controller introduces a stiffness and damping...
Microwave transmission-line impedance data
Gunston, M A R
1996-01-01
A compendium of data for computing the characteristic impedance of transmission lines based on physical dimensions. Covers both conventional structures and unusual geometries, including coaxial, eccentric and elliptic coaxial, twin-wire, wire-above-ground, microstrip and derivatives, stripline, slabline and trough line. Also details numerous configurations of coupled lines.
Generalized impedances and wakes in asymmetric structures
International Nuclear Information System (INIS)
Heifets, S.; Wagner, A.; Zotter, B.
1998-01-01
In rotationally structures, the dominant m = 0 longitudinal impedance does not depend on the offsets of either the leading or the trailing particles, while the dominant m = 1 transverse impedance is proportional to the offset of the leading particles, while it is still independent of the offsets of the trailing ones. This behavior is no longer true in rotationally non-symmetric structures, where in general all impedances depend on the offsets of both the leading and the trailing particles. The same behavior is shown by wake functions and wake potentials. The concept of generalized impedances or generalized wake functions must be used to calculate the effect of leading particles on trailing ones with different offsets, each described by two transverse coordinates. This dependence of wake potentials on four additional parameters (two for each offset) would make their use very cumbersome. Fortunately, it was found that the transverse wake potentials can be separated into superpositions of dipolar components, which are proportional to the offset of the leading bunch, and quadrupolar components, which are proportional to the offset of the trailing particles. Higher multipole components are much smaller, and can be neglected for most structures without rotational symmetry. In this report, the authors derive analytical expressions for these multipolar components, which permits estimates of the size of the neglected terms. In particular, when structures have one or two transverse symmetry planes, the expressions simplify and explain the behavior of wake potentials which had been computed for rotationally non-symmetric structures
Detection of irradiated potatoes by impedance measurement
International Nuclear Information System (INIS)
Hayashi, T.; Todoriki, S.; Otobe, K.; Sugiyama, J.
1996-01-01
Potato is one of the major food items to be treated with ionising radiation and potatoes are irradiated on a large scale in several countries. Every year around 15,000 t of potatoes are irradiated at doses of 60 to 150 Gy (average dose is about 100 Gy) in Japan. Although various methods to detect irradiated potatoes have been investigated, no established method has been reported. Measuring electrical conductivity or impedance of potatoes has been reported as a promising method for the detection of irradiated potatoes. In previous studies it has been found that the ratio of impedance magnitude at 50 kHz to that at 5 kHz, measured immediately after puncturing a potato tuber, is dependent upon the dose applied to the tuber, independent of storage temperature and stable during storage after irradiation. The aim of this study was to establish the optimum conditions for impedance measurement and to examine the applicability of the impedance measuring method to various cultivars (cv.) of potatoes. (author)
Quantity precommitment and price matching
DEFF Research Database (Denmark)
Tumennasan, Norovsambuu
We revisit the question of whether price matching is anti-competitive in a capacity constrained duopoly setting. We show that the effect of price matching depends on capacity. Specifically, price matching has no effect when capacity is relatively low, but it benefits the firms when capacity...... is relatively high. Interestingly, when capacity is in an intermediate range, price matching benefits only the small firm but does not affect the large firm in any way. Therefore, one has to consider capacity seriously when evaluating if price matching is anti-competitive. If the firms choose their capacities...... simultaneously before pricing decisions, then the effect of price matching is either pro-competitive or ambiguous. We show that if the cost of capacity is high, then price matching can only (weakly) decrease the market price. On the other hand, if the cost of capacity is low, then the effect of price matching...
DEFF Research Database (Denmark)
Zhao, Xin; Guerrero, Josep M.; Savaghebi, Mehdi
2017-01-01
of the utility grid during grid faults. In this paper, a LVRT control strategy based on positive/negative sequence droop control is proposed for grid-interactive MGs to ride-through voltage sags with not only inductive/resistive, but also complex line impedance. By using the proposed control strategy, MGs can......Due to the increasing penetration level of microgrids (MGs), it becomes a critical issue for MGs to help sustaining power system stability. Therefore, ancillary services, such as the low-voltage ride-through (LVRT) capability should be incorporated in MGs in order to guarantee stable operation...... support the grid voltage, make profits, and also ride-through the voltage dip during the whole fault period. A two layer hierarchical control strategy is proposed in this paper. The primary controller consists of voltage and current inner loops, a conventional droop control and a virtual impedance loop...
Comparative study of 0° X-cut and Y + 36°-cut lithium niobate high-voltage sensing
Patel, N.; Branch, D. W.; Schamiloglu, E.; Cular, S.
2015-08-01
A comparison study between Y + 36° and 0° X-cut lithium niobate (LiNbO3) was performed to evaluate the influence of crystal cut on the acoustic propagation to realize a piezoelectric high-voltage sensor. The acoustic time-of-flight for each crystal cut was measured when applying direct current (DC), alternating current (AC), and pulsed voltages. Results show that the voltage-induced shift in the acoustic wave propagation time scaled quadratically with voltage for DC and AC voltages applied to X-cut crystals. For the Y + 36° crystal, the voltage-induced shift scales linearly with DC voltages and quadratically with AC voltages. When applying 5 μs voltage pulses to both crystals, the voltage-induced shift scaled linearly with voltage. For the Y + 36° cut, the voltage-induced shift from applying DC voltages ranged from 10 to 54 ps and 35 to 778 ps for AC voltages at 640 V over the frequency range of 100 Hz-100 kHz. Using the same conditions as the Y + 36° cut, the 0° X-cut crystal sensed a shift of 10-273 ps for DC voltages and 189-813 ps for AC voltage application. For 5 μs voltage pulses, the 0° X-cut crystal sensed a voltage induced shift of 0.250-2 ns and the Y + 36°-cut crystal sensed a time shift of 0.115-1.6 ns. This suggests a frequency sensitive response to voltage where the influence of the crystal cut was not a significant contributor under DC, AC, or pulsed voltage conditions. The measured DC data were compared to a 1-D impedance matrix model where the predicted incremental length changed as a function of voltage. When the voltage source error was eliminated through physical modeling from the uncertainty budget, the combined uncertainty of the sensor (within a 95% confidence interval) decreased to 0.0033% using a Y + 36°-cut crystal and 0.0032% using an X-cut crystal for all the voltage conditions used in this experiment.
Comparative study of 0° X-cut and Y + 36°-cut lithium niobate high-voltage sensing
International Nuclear Information System (INIS)
Patel, N.; Branch, D. W.; Cular, S.; Schamiloglu, E.
2015-01-01
A comparison study between Y + 36° and 0° X-cut lithium niobate (LiNbO 3 ) was performed to evaluate the influence of crystal cut on the acoustic propagation to realize a piezoelectric high-voltage sensor. The acoustic time-of-flight for each crystal cut was measured when applying direct current (DC), alternating current (AC), and pulsed voltages. Results show that the voltage-induced shift in the acoustic wave propagation time scaled quadratically with voltage for DC and AC voltages applied to X-cut crystals. For the Y + 36° crystal, the voltage-induced shift scales linearly with DC voltages and quadratically with AC voltages. When applying 5 μs voltage pulses to both crystals, the voltage-induced shift scaled linearly with voltage. For the Y + 36° cut, the voltage-induced shift from applying DC voltages ranged from 10 to 54 ps and 35 to 778 ps for AC voltages at 640 V over the frequency range of 100 Hz–100 kHz. Using the same conditions as the Y + 36° cut, the 0° X-cut crystal sensed a shift of 10–273 ps for DC voltages and 189–813 ps for AC voltage application. For 5 μs voltage pulses, the 0° X-cut crystal sensed a voltage induced shift of 0.250–2 ns and the Y + 36°-cut crystal sensed a time shift of 0.115–1.6 ns. This suggests a frequency sensitive response to voltage where the influence of the crystal cut was not a significant contributor under DC, AC, or pulsed voltage conditions. The measured DC data were compared to a 1-D impedance matrix model where the predicted incremental length changed as a function of voltage. When the voltage source error was eliminated through physical modeling from the uncertainty budget, the combined uncertainty of the sensor (within a 95% confidence interval) decreased to 0.0033% using a Y + 36°-cut crystal and 0.0032% using an X-cut crystal for all the voltage conditions used in this experiment
International Nuclear Information System (INIS)
Gower, E J; Sullivan, J S
2002-01-01
High voltage, solid state, inductive adder, pulse generators have found increasing application as fast kicker pulse modulators for charged particle beams. The solid state, inductive adder, pulse generator is similar in operation to the linear induction accelerator. The main difference is that the solid state, adder couples energy by transformer action from multiple primaries to a voltage summing stalk, instead of an electron beam. Ideally, the inductive adder produces a rectangular voltage pulse at the load. In reality, there is usually some voltage variation at the load due to droop on primary circuit storage capacitors, or, temporal variations in the load impedance. Power MOSFET circuits have been developed to provide analog modulation of the output voltage amplitude of a solid state, inductive adder, pulse generator. The modulation is achieved by including MOSFET based, variable subtraction circuits in the multiple primary stack. The subtraction circuits can be used to compensate for voltage droop, or, to tailor the output pulse amplitude to provide a desired effect in the load. Power MOSFET subtraction circuits have been developed to modulate short, temporal (60-400 ns), voltage and current pulses. MOSFET devices have been tested up to 20 amps and 800 Volts with a band pass of 50 MHz. An analog modulation cell has been tested in a five cell high, voltage adder stack
Directory of Open Access Journals (Sweden)
D. Maurath
2008-05-01
Full Text Available This paper presents novel circuit concepts for integrated rectifiers and voltage converting interfaces for energy harvesting micro-generators. In the context of energy harvesting, usually only small voltages are supplied by vibration-driven generators. Therefore, rectification with minimum voltage losses and low reverse currents is an important issue. This is realized by novel integrated rectifiers which were fabricated and are presented in this article. Additionally, there is a crucial need for dynamic load adaptation as well as voltage up-conversion. A circuit concept is presented, which is able to obtain both requirements. This generator interface adapts its input impedance for an optimal energy transfer efficiency. Furthermore, this generator interface provides implicit voltage up-conversion, whereas the generator output energy is stored on a buffer, which is connected to the output of the voltage converting interface. As simulations express, this fully integrated converter is able to boost ac-voltages greater than |0.35 V| to an output dc-voltage of 2.0 V–2.5 V. Thereby, high harvesting efficiencies above 80% are possible within the entire operational range.
Energy Technology Data Exchange (ETDEWEB)
Rittinghaus, D
1981-09-01
The complex short-circuit impedances of energized low-voltage main systems were measured with a double-bridge in underground mining operation. The magnitude of the short-circuit currents depends on these impedances. Customary calculations of such currents depend on empirical approximations. To verify the accuracy of these approximations, the measured impedances of 61 nodes in three different main systems were compared with the results of the calculations. The comparison made between the short-circuit currents determined by measurable quantities and the values calculated according to VDE 0118 shows that the stipulated coefficients for calculating the minimum short-circuit currents lie very far on the safe side. An amendment for calculating the short-circuit in accordance with VDE 0118 is therefore suggested.
Development of Electromechanical Architectures for AC Voltage Metrology
Directory of Open Access Journals (Sweden)
Alexandre BOUNOUH
2010-12-01
Full Text Available This paper presents results of work undertaken for exploring MEMS capabilities to fabricate AC voltage references for electrical metrology and high precision instrumentation through the mechanical-electrical coupling in MEMS. From first MEMS test structures previously realized, a second set of devices with improved characteristics has been developed and fabricated with Silicon on Insulator (SOI Surface Micromachining process. These MEMS exhibit pull-in voltages of 5 V and 10 V to match with the best performance of the read-out electronics developed for driving the MEMS. Deep Level Transient Spectroscopy measurements carried out on the new design show resonance frequencies of about only some kHz, and the stability of the MEMS output voltage measured at 100 kHz has been found very promising for the best samples where the relative deviation from the mean value over almost 12 hours showed a standard deviation of about 6.3 ppm.