Scatter factor corrections for elongated fields
International Nuclear Information System (INIS)
Higgins, P.D.; Sohn, W.H.; Sibata, C.H.; McCarthy, W.A.
1989-01-01
Measurements have been made to determine scatter factor corrections for elongated fields of Cobalt-60 and for nominal linear accelerator energies of 6 MV (Siemens Mevatron 67) and 18 MV (AECL Therac 20). It was found that for every energy the collimator scatter factor varies by 2% or more as the field length-to-width ratio increases beyond 3:1. The phantom scatter factor is independent of which collimator pair is elongated at these energies. For 18 MV photons it was found that the collimator scatter factor is complicated by field-size-dependent backscatter into the beam monitor
Improved scatter correction with factor analysis for planar and SPECT imaging
Knoll, Peter; Rahmim, Arman; Gültekin, Selma; Šámal, Martin; Ljungberg, Michael; Mirzaei, Siroos; Segars, Paul; Szczupak, Boguslaw
2017-09-01
Quantitative nuclear medicine imaging is an increasingly important frontier. In order to achieve quantitative imaging, various interactions of photons with matter have to be modeled and compensated. Although correction for photon attenuation has been addressed by including x-ray CT scans (accurate), correction for Compton scatter remains an open issue. The inclusion of scattered photons within the energy window used for planar or SPECT data acquisition decreases the contrast of the image. While a number of methods for scatter correction have been proposed in the past, in this work, we propose and assess a novel, user-independent framework applying factor analysis (FA). Extensive Monte Carlo simulations for planar and tomographic imaging were performed using the SIMIND software. Furthermore, planar acquisition of two Petri dishes filled with 99mTc solutions and a Jaszczak phantom study (Data Spectrum Corporation, Durham, NC, USA) using a dual head gamma camera were performed. In order to use FA for scatter correction, we subdivided the applied energy window into a number of sub-windows, serving as input data. FA results in two factor images (photo-peak, scatter) and two corresponding factor curves (energy spectra). Planar and tomographic Jaszczak phantom gamma camera measurements were recorded. The tomographic data (simulations and measurements) were processed for each angular position resulting in a photo-peak and a scatter data set. The reconstructed transaxial slices of the Jaszczak phantom were quantified using an ImageJ plugin. The data obtained by FA showed good agreement with the energy spectra, photo-peak, and scatter images obtained in all Monte Carlo simulated data sets. For comparison, the standard dual-energy window (DEW) approach was additionally applied for scatter correction. FA in comparison with the DEW method results in significant improvements in image accuracy for both planar and tomographic data sets. FA can be used as a user
Cross plane scattering correction
International Nuclear Information System (INIS)
Shao, L.; Karp, J.S.
1990-01-01
Most previous scattering correction techniques for PET are based on assumptions made for a single transaxial plane and are independent of axial variations. These techniques will incorrectly estimate the scattering fraction for volumetric PET imaging systems since they do not take the cross-plane scattering into account. In this paper, the authors propose a new point source scattering deconvolution method (2-D). The cross-plane scattering is incorporated into the algorithm by modeling a scattering point source function. In the model, the scattering dependence both on axial and transaxial directions is reflected in the exponential fitting parameters and these parameters are directly estimated from a limited number of measured point response functions. The authors' results comparing the standard in-plane point source deconvolution to the authors' cross-plane source deconvolution show that for a small source, the former technique overestimates the scatter fraction in the plane of the source and underestimate the scatter fraction in adjacent planes. In addition, the authors also propose a simple approximation technique for deconvolution
Mohammadi, S. M.; Tavakoli-Anbaran, H.; Zeinali, H. Z.
2017-02-01
The parallel-plate free-air ionization chamber termed FAC-IR-300 was designed at the Atomic Energy Organization of Iran, AEOI. This chamber is used for low and medium X-ray dosimetry on the primary standard level. In order to evaluate the air-kerma, some correction factors such as electron-loss correction factor (ke) and photon scattering correction factor (ksc) are needed. ke factor corrects the charge loss from the collecting volume and ksc factor corrects the scattering of photons into collecting volume. In this work ke and ksc were estimated by Monte Carlo simulation. These correction factors are calculated for mono-energy photon. As a result of the simulation data, the ke and ksc values for FAC-IR-300 ionization chamber are 1.0704 and 0.9982, respectively.
International Nuclear Information System (INIS)
Mohammadi, S.M.; Tavakoli-Anbaran, H.; Zeinali, H.Z.
2017-01-01
The parallel-plate free-air ionization chamber termed FAC-IR-300 was designed at the Atomic Energy Organization of Iran, AEOI. This chamber is used for low and medium X-ray dosimetry on the primary standard level. In order to evaluate the air-kerma, some correction factors such as electron-loss correction factor (k e ) and photon scattering correction factor (k sc ) are needed. k e factor corrects the charge loss from the collecting volume and k sc factor corrects the scattering of photons into collecting volume. In this work k e and k sc were estimated by Monte Carlo simulation. These correction factors are calculated for mono-energy photon. As a result of the simulation data, the k e and k sc values for FAC-IR-300 ionization chamber are 1.0704 and 0.9982, respectively.
Parametrisation of the collimator scatter correction factors of square and rectangular photon beams
International Nuclear Information System (INIS)
Jager, H.N.; Heukelom, S.; Kleffens, H.J. van; Gasteren, J.J.M. van; Laarse, R. van der; Venselaar, J.L.M.; Westermann, C.F.
1995-01-01
Collimator scatter correction factors S c have been measured with a cylindrical mini-phantom for five types of dual photon energy accelerators with energies between 6 and 25 MV. Using these S c -data three methods to parametrize S c of square fields have been compared including a third-order polynomial of the natural logarithm of the fieldsize normalised by the fieldsize of 10 cm 2 . Also six methods to calculate S c of rectangular fields have been compared including a new one which determines the equivalent fieldsize by extending Sterling's method. The deviation between measured and calculated S c for every accelerator, energy and all methods are determined resulting in the maximum and average deviation per method. Applied to square fields the maximum and average deviation were for the method of Chen 0.64% and 0.15%, of Szymzcyk 0.98% and 0.21%, and of this work 0.41% and 0.10%. For the rectangular fields the deviations were for the method of Sterling 1.89% and 0.50%, of Vadash 1.60% and 0.28%, of Szymczyk et al. 1.21% and 0.25%, of Chen 1.84% and 0.31% and of this work 0.79% and 0.20%. Finally, a recommendation is given how to limit the number of fields at which S c should be measured
Li, Runze; Peng, Tong; Liang, Yansheng; Yang, Yanlong; Yao, Baoli; Yu, Xianghua; Min, Junwei; Lei, Ming; Yan, Shaohui; Zhang, Chunmin; Ye, Tong
2017-10-01
Focusing and imaging through scattering media has been proved possible with high resolution wavefront shaping. A completely scrambled scattering field can be corrected by applying a correction phase mask on a phase only spatial light modulator (SLM) and thereby the focusing quality can be improved. The correction phase is often found by global searching algorithms, among which Genetic Algorithm (GA) stands out for its parallel optimization process and high performance in noisy environment. However, the convergence of GA slows down gradually with the progression of optimization, causing the improvement factor of optimization to reach a plateau eventually. In this report, we propose an interleaved segment correction (ISC) method that can significantly boost the improvement factor with the same number of iterations comparing with the conventional all segment correction method. In the ISC method, all the phase segments are divided into a number of interleaved groups; GA optimization procedures are performed individually and sequentially among each group of segments. The final correction phase mask is formed by applying correction phases of all interleaved groups together on the SLM. The ISC method has been proved significantly useful in practice because of its ability to achieve better improvement factors when noise is present in the system. We have also demonstrated that the imaging quality is improved as better correction phases are found and applied on the SLM. Additionally, the ISC method lowers the demand of dynamic ranges of detection devices. The proposed method holds potential in applications, such as high-resolution imaging in deep tissue.
International Nuclear Information System (INIS)
Martin, G.; Coca, M.; Capote, R.
1996-01-01
Using Monte Carlo method technique , a computer code which simulates the time of flight experiment to measure double differential cross section was developed. The correction factor for flux attenuation and multiple scattering, that make a deformation to the measured spectrum, were calculated. The energy dependence of the correction factor was determined and a comparison with other works is shown. Calculations for Fe 56 at two different scattering angles were made. We also reproduce the experiment performed at the Nuclear Analysis Laboratory for C 12 at 25 celsius degree and the calculated correction factor for the is measured is shown. We found a linear relation between the scatter size and the correction factor for flux attenuation
Atmospheric scattering corrections to solar radiometry
International Nuclear Information System (INIS)
Box, M.A.; Deepak, A.
1979-01-01
Whenever a solar radiometer is used to measure direct solar radiation, some diffuse sky radiation invariably enters the detector's field of view along with the direct beam. Therefore, the atmospheric optical depth obtained by the use of Bouguer's transmission law (also called Beer-Lambert's law), that is valid only for direct radiation, needs to be corrected by taking account of the scattered radiation. In this paper we shall discuss the correction factors needed to account for the diffuse (i.e., singly and multiply scattered) radiation and the algorithms developed for retrieving aerosol size distribution from such measurements. For a radiometer with a small field of view (half-cone angle 0 ) and relatively clear skies (optical depths <0.4), it is shown that the total diffuse contributions represents approximately l% of the total intensity. It is assumed here that the main contributions to the diffuse radiation within the detector's view cone are due to single scattering by molecules and aerosols and multiple scattering by molecules alone, aerosol multiple scattering contributions being treated as negligibly small. The theory and the numerical results discussed in this paper will be helpful not only in making corrections to the measured optical depth data but also in designing improved solar radiometers
Source distribution dependent scatter correction for PVI
International Nuclear Information System (INIS)
Barney, J.S.; Harrop, R.; Dykstra, C.J.
1993-01-01
Source distribution dependent scatter correction methods which incorporate different amounts of information about the source position and material distribution have been developed and tested. The techniques use image to projection integral transformation incorporating varying degrees of information on the distribution of scattering material, or convolution subtraction methods, with some information about the scattering material included in one of the convolution methods. To test the techniques, the authors apply them to data generated by Monte Carlo simulations which use geometric shapes or a voxelized density map to model the scattering material. Source position and material distribution have been found to have some effect on scatter correction. An image to projection method which incorporates a density map produces accurate scatter correction but is computationally expensive. Simpler methods, both image to projection and convolution, can also provide effective scatter correction
Compton scatter correction for planner scintigraphic imaging
Energy Technology Data Exchange (ETDEWEB)
Vaan Steelandt, E; Dobbeleir, A; Vanregemorter, J [Algemeen Ziekenhuis Middelheim, Antwerp (Belgium). Dept. of Nuclear Medicine and Radiotherapy
1995-12-01
A major problem in nuclear medicine is the image degradation due to Compton scatter in the patient. Photons emitted by the radioactive tracer scatter in collision with electrons of the surrounding tissue. Due to the resulting loss of energy and change in direction, the scattered photons induce an object dependant background on the images. This results in a degradation of the contrast of warm and cold lesions. Although theoretically interesting, most of the techniques proposed in literature like the use of symmetrical photopeaks can not be implemented on the commonly used gamma camera due to the energy/linearity/sensitivity corrections applied in the detector. A method for a single energy isotope based on existing methods with adjustments towards daily practice and clinical situations is proposed. It is assumed that the scatter image, recorded from photons collected within a scatter window adjacent to the photo peak, is a reasonable close approximation of the true scatter component of the image reconstructed from the photo peak window. A fraction `k` of the image using the scatter window is subtracted from the image recorded in the photo peak window to produce the compensated image. The principal matter of the method is the right value for the factor `k`, which is determined in a mathematical way and confirmed by experiments. To determine `k`, different kinds of scatter media are used and are positioned in different ways in order to simulate a clinical situation. For a secondary energy window from 100 to 124 keV below a photo peak window from 126 to 154 keV, a value of 0.7 is found. This value has been verified using both an antropomorph thyroid phantom and the Rollo contrast phantom.
Scatter and attenuation correction in SPECT
International Nuclear Information System (INIS)
Ljungberg, Michael
2004-01-01
The adsorbed dose is related to the activity uptake in the organ and its temporal distribution. Measured count rate with scintillation cameras is related to activity through the system sensitivity, cps/MBq. By accounting for physical processes and imaging limitations we can measure the activity at different time points. Correction for physical factor, such as attenuation and scatter is required for accurate quantitation. Both planar and SPECT imaging can be used to estimate activities for radiopharmaceutical dosimetry. Planar methods have been the most widely used but is a 2D technique. With accurate modelling for imagine in iterative reconstruction, SPECT methods will prove to be more accurate
Schowalter, M; Müller, K; Rosenauer, A
2012-01-01
Modified atomic scattering amplitudes (MASAs), taking into account the redistribution of charge due to bonds, and the respective correction factors considering the effect of static atomic displacements were computed for the chemically sensitive 002 reflection for ternary III-V and II-VI semiconductors. MASAs were derived from computations within the density functional theory formalism. Binary eight-atom unit cells were strained according to each strain state s (thin, intermediate, thick and fully relaxed electron microscopic specimen) and each concentration (x = 0, …, 1 in 0.01 steps), where the lattice parameters for composition x in strain state s were calculated using continuum elasticity theory. The concentration dependence was derived by computing MASAs for each of these binary cells. Correction factors for static atomic displacements were computed from relaxed atom positions by generating 50 × 50 × 50 supercells using the lattice parameter of the eight-atom unit cells. Atoms were randomly distributed according to the required composition. Polynomials were fitted to the composition dependence of the MASAs and the correction factors for the different strain states. Fit parameters are given in the paper.
Generalised Batho correction factor
International Nuclear Information System (INIS)
Siddon, R.L.
1984-01-01
There are various approximate algorithms available to calculate the radiation dose in the presence of a heterogeneous medium. The Webb and Fox product over layers formulation of the generalised Batho correction factor requires determination of the number of layers and the layer densities for each ray path. It has been shown that the Webb and Fox expression is inefficient for the heterogeneous medium which is expressed as regions of inhomogeneity rather than layers. The inefficiency of the layer formulation is identified as the repeated problem of determining for each ray path which inhomogeneity region corresponds to a particular layer. It has been shown that the formulation of the Batho correction factor as a product over inhomogeneity regions avoids that topological problem entirely. The formulation in terms of a product over regions simplifies the computer code and reduces the time required to calculate the Batho correction factor for the general heterogeneous medium. (U.K.)
Model Correction Factor Method
DEFF Research Database (Denmark)
Christensen, Claus; Randrup-Thomsen, Søren; Morsing Johannesen, Johannes
1997-01-01
The model correction factor method is proposed as an alternative to traditional polynomial based response surface techniques in structural reliability considering a computationally time consuming limit state procedure as a 'black box'. The class of polynomial functions is replaced by a limit...... of the model correction factor method, is that in simpler form not using gradient information on the original limit state function or only using this information once, a drastic reduction of the number of limit state evaluation is obtained together with good approximations on the reliability. Methods...
First order correction to quasiclassical scattering amplitude
International Nuclear Information System (INIS)
Kuz'menko, A.V.
1978-01-01
First order (with respect to h) correction to quasiclassical with the aid of scattering amplitude in nonrelativistic quantum mechanics is considered. This correction is represented by two-loop diagrams and includes the double integrals. With the aid of classical equations of motion, the sum of the contributions of the two-loop diagrams is transformed into the expression which includes one-dimensional integrals only. The specific property of the expression obtained is that the integrand does not possess any singularities in the focal points of the classical trajectory. The general formula takes much simpler form in the case of one-dimensional systems
Radiative corrections to deep inelastic muon scattering
International Nuclear Information System (INIS)
Akhundov, A.A.; Bardin, D.Yu.; Lohman, W.
1986-01-01
A summary is given of the most recent results for the calculaion of radiative corrections to deep inelastic muon-nucleon scattering. Contributions from leptonic electromagnetic processes up to the order a 4 , vacuum polarization by leptons and hadrons, hadronic electromagnetic processes approximately a 3 and γZ interference have been taken into account. The dependence of the individual contributions on kinematical variables is studied. Contributions, not considered in earlier calculations of radiative corrections, reach in certain kinematical regions several per cent at energies above 100 GeV
Mass corrections in deep-inelastic scattering
International Nuclear Information System (INIS)
Gross, D.J.; Treiman, S.B.; Wilczek, F.A.
1977-01-01
The moment sum rules for deep-inelastic lepton scattering are expected for asymptotically free field theories to display a characteristic pattern of logarithmic departures from scaling at large enough Q 2 . In the large-Q 2 limit these patterns do not depend on hadron or quark masses m. For modest values of Q 2 one expects corrections at the level of powers of m 2 /Q 2 . We discuss the question whether these mass effects are accessible in perturbation theory, as applied to the twist-2 Wilson coefficients and more generally. Our conclusion is that some part of the mass effects must arise from a nonperturbative origin. We also discuss the corrections which arise from higher orders in perturbation theory for very large Q 2 , where mass effects can perhaps be ignored. The emphasis here is on a characterization of the Q 2 , x domain where higher-order corrections are likely to be unimportant
Holographic corrections to meson scattering amplitudes
Energy Technology Data Exchange (ETDEWEB)
Armoni, Adi; Ireson, Edwin, E-mail: 746616@swansea.ac.uk
2017-06-15
We compute meson scattering amplitudes using the holographic duality between confining gauge theories and string theory, in order to consider holographic corrections to the Veneziano amplitude and associated higher-point functions. The generic nature of such computations is explained, thanks to the well-understood nature of confining string backgrounds, and two different examples of the calculation in given backgrounds are used to illustrate the details. The effect we discover, whilst only qualitative, is re-obtainable in many such examples, in four-point but also higher point amplitudes.
Hadron mass corrections in semi-inclusive deep inelastic scattering
International Nuclear Information System (INIS)
Accardi, A.; Hobbs, T.; Melnitchouk, W.
2009-01-01
We derive mass corrections for semi-inclusive deep inelastic scattering of leptons from nucleons using a collinear factorization framework which incorporates the initial state mass of the target nucleon and the final state mass of the produced hadron h. The hadron mass correction is made by introducing a generalized, finite-Q 2 scaling variable ζ h for the hadron fragmentation function, which approaches the usual energy fraction z h = E h /ν in the Bjorken limit. We systematically examine the kinematic dependencies of the mass corrections to semi-inclusive cross sections, and find that these are even larger than for inclusive structure functions. The hadron mass corrections compete with the experimental uncertainties at kinematics typical of current facilities, Q 2 2 and intermediate x B > 0.3, and will be important to efforts at extracting parton distributions from semi-inclusive processes at intermediate energies.
Scattering Correction For Image Reconstruction In Flash Radiography
International Nuclear Information System (INIS)
Cao, Liangzhi; Wang, Mengqi; Wu, Hongchun; Liu, Zhouyu; Cheng, Yuxiong; Zhang, Hongbo
2013-01-01
Scattered photons cause blurring and distortions in flash radiography, reducing the accuracy of image reconstruction significantly. The effect of the scattered photons is taken into account and an iterative deduction of the scattered photons is proposed to amend the scattering effect for image restoration. In order to deduct the scattering contribution, the flux of scattered photons is estimated as the sum of two components. The single scattered component is calculated accurately together with the uncollided flux along the characteristic ray, while the multiple scattered component is evaluated using correction coefficients pre-obtained from Monte Carlo simulations.The arbitrary geometry pretreatment and ray tracing are carried out based on the customization of AutoCAD. With the above model, an Iterative Procedure for image restORation code, IPOR, is developed. Numerical results demonstrate that the IPOR code is much more accurate than the direct reconstruction solution without scattering correction and it has a very high computational efficiency
Scattering Correction For Image Reconstruction In Flash Radiography
Energy Technology Data Exchange (ETDEWEB)
Cao, Liangzhi; Wang, Mengqi; Wu, Hongchun; Liu, Zhouyu; Cheng, Yuxiong; Zhang, Hongbo [Xi' an Jiaotong Univ., Xi' an (China)
2013-08-15
Scattered photons cause blurring and distortions in flash radiography, reducing the accuracy of image reconstruction significantly. The effect of the scattered photons is taken into account and an iterative deduction of the scattered photons is proposed to amend the scattering effect for image restoration. In order to deduct the scattering contribution, the flux of scattered photons is estimated as the sum of two components. The single scattered component is calculated accurately together with the uncollided flux along the characteristic ray, while the multiple scattered component is evaluated using correction coefficients pre-obtained from Monte Carlo simulations.The arbitrary geometry pretreatment and ray tracing are carried out based on the customization of AutoCAD. With the above model, an Iterative Procedure for image restORation code, IPOR, is developed. Numerical results demonstrate that the IPOR code is much more accurate than the direct reconstruction solution without scattering correction and it has a very high computational efficiency.
How to simplify transmission-based scatter correction for clinical application
International Nuclear Information System (INIS)
Baccarne, V.; Hutton, B.F.
1998-01-01
Full text: The performances of ordered subsets (OS) EM reconstruction including attenuation, scatter and spatial resolution correction are evaluated using cardiac Monte Carlo data. We demonstrate how simplifications in the scatter model allow one to correct SPECT data for scatter in terms of quantitation and quality in a reasonable time. Initial reconstruction of the 20% window is performed including attenuation correction (broad beam μ values), to estimate the activity quantitatively (accuracy 3%), but not spatially. A rough reconstruction with 2 iterations (subset size: 8) is sufficient for subsequent scatter correction. Estimation of primary photons is obtained by projecting the previous distribution including attenuation (narrow beam μ values). Estimation of the scatter is obtained by convolving the primary estimates by a depth dependent scatter kernel, and scaling the result by a factor calculated from the attenuation map. The correction can be accelerated by convolving several adjacent planes with the same kernel, and using an average scaling factor. Simulation of the effects of the collimator during the scatter correction was demonstrated to be unnecessary. Final reconstruction is performed using 6 iterations OSEM, including attenuation (narrow beam μ values) and spatial resolution correction. Scatter correction is implemented by incorporating the estimated scatter as a constant offset in the forward projection step. The total correction + reconstruction (64 proj. 40x128 pixel) takes 38 minutes on a Sun Sparc 20. Quantitatively, the accuracy is 7% in a reconstructed slice. The SNR inside the whole myocardium (defined from the original object), is equal to 2.1 and 2.3 - in the corrected and the primary slices respectively. The scatter correction preserves the myocardium to ventricle contrast (primary: 0.79, corrected: 0.82). These simplifications allow acceleration of correction without influencing the quality of the result
Software correction of scatter coincidence in positron CT
International Nuclear Information System (INIS)
Endo, M.; Iinuma, T.A.
1984-01-01
This paper describes a software correction of scatter coincidence in positron CT which is based on an estimation of scatter projections from true projections by an integral transform. Kernels for the integral transform are projected distributions of scatter coincidences for a line source at different positions in a water phantom and are calculated by Klein-Nishina's formula. True projections of any composite object can be determined from measured projections by iterative applications of the integral transform. The correction method was tested in computer simulations and phantom experiments with Positologica. The results showed that effects of scatter coincidence are not negligible in the quantitation of images, but the correction reduces them significantly. (orig.)
Real-time scatter measurement and correction in film radiography
International Nuclear Information System (INIS)
Shaw, C.G.
1987-01-01
A technique for real-time scatter measurement and correction in scanning film radiography is described. With this technique, collimated x-ray fan beams are used to partially reject scattered radiation. Photodiodes are attached to the aft-collimator for sampled scatter measurement. Such measurement allows the scatter distribution to be reconstructed and subtracted from digitized film image data for accurate transmission measurement. In this presentation the authors discuss the physical and technical considerations of this scatter correction technique. Examples are shown that demonstrate the feasibility of the technique. Improved x-ray transmission measurement and dual-energy subtraction imaging are demonstrated with phantoms
Derivation of Batho's correction factor for heterogeneities
International Nuclear Information System (INIS)
Lulu, B.A.; Bjaerngard, B.E.
1982-01-01
Batho's correction factor for dose in a heterogeneous, layered medium is derived from the tissue--air ratio method (TARM). The reason why the Batho factor is superior to the TARM factor at low energy is ascribed to the fact that it accounts for the distribution of the scatter-generating matter along the centerline. The poor behavior of the Batho factor at high energies is explained as a consequence of the lack of electron equilibrium at appreciable depth below the surface. Key words: Batho factor, heterogeneity, inhomogeneity, tissue--air ratio method
Disadvantage factor for anisotropic scattering
International Nuclear Information System (INIS)
Saad, E.A.; Abdel Krim, M.S.; EL-Dimerdash, A.A.
1990-01-01
The invariant embedding method is used to solve the problem for a two region reactor with anisotropic scattering and to compute the disadvantage factor necessary for calculating some reactor parameters
Output factors and scatter ratios
Energy Technology Data Exchange (ETDEWEB)
Shrivastava, P N; Summers, R E; Samulski, T V; Baird, L C [Allegheny General Hospital, Pittsburgh, PA (USA); Ahuja, A S; Dubuque, G L; Hendee, W R; Chhabra, A S
1979-07-01
Reference is made to a previous publication on output factors and scatter ratios for radiotherapy units in which it was suggested that the output factor should be included in the definitions of scatter-air ratio and tissue-maximum ratio. In the present correspondence from other authors and from the authors of the previous publication, the original definitions and the proposed changes are discussed. Radiation scatter from source and collimator degradation of beam energy and calculation of dose in tissue are considered in relation to the objective of accurate dosimetry.
Attenuation and scatter correction in SPECT
International Nuclear Information System (INIS)
Pant, G.S.; Pandey, A.K.
2000-01-01
While passing through matter, photons undergo various types of interactions. In the process, some photons are completely absorbed, some are scattered in different directions with or without any change in their energy and some pass through unattenuated. These unattenuated photons carry the information with them. However, the image data gets corrupted with attenuation and scatter processes. This paper deals with the effect of these two processes in nuclear medicine images and suggests the methods to overcome them
Neural network scatter correction technique for digital radiography
International Nuclear Information System (INIS)
Boone, J.M.
1990-01-01
This paper presents a scatter correction technique based on artificial neural networks. The technique utilizes the acquisition of a conventional digital radiographic image, coupled with the acquisition of a multiple pencil beam (micro-aperture) digital image. Image subtraction results in a sparsely sampled estimate of the scatter component in the image. The neural network is trained to develop a causal relationship between image data on the low-pass filtered open field image and the sparsely sampled scatter image, and then the trained network is used to correct the entire image (pixel by pixel) in a manner which is operationally similar to but potentially more powerful than convolution. The technique is described and is illustrated using clinical primary component images combined with scatter component images that are realistically simulated using the results from previously reported Monte Carlo investigations. The results indicate that an accurate scatter correction can be realized using this technique
Radiative corrections to neutrino deep inelastic scattering revisited
International Nuclear Information System (INIS)
Arbuzov, Andrej B.; Bardin, Dmitry Yu.; Kalinovskaya, Lidia V.
2005-01-01
Radiative corrections to neutrino deep inelastic scattering are revisited. One-loop electroweak corrections are re-calculated within the automatic SANC system. Terms with mass singularities are treated including higher order leading logarithmic corrections. Scheme dependence of corrections due to weak interactions is investigated. The results are implemented into the data analysis of the NOMAD experiment. The present theoretical accuracy in description of the process is discussed
International Nuclear Information System (INIS)
Yang, J.; Kuikka, J.T.; Vanninen, E.; Laensimies, E.; Kauppinen, T.; Patomaeki, L.
1999-01-01
Photon scatter is one of the most important factors degrading the quantitative accuracy of SPECT images. Many scatter correction methods have been proposed. The single isotope method was proposed by us. Aim: We evaluate the scatter correction method of improving the quality of images by acquiring emission and transmission data simultaneously with single isotope scan. Method: To evaluate the proposed scatter correction method, a contrast and linearity phantom was studied. Four female patients with fibromyalgia (FM) syndrome and four with chronic back pain (BP) were imaged. Grey-to-cerebellum (G/C) and grey-to-white matter (G/W) ratios were determined by one skilled operator for 12 regions of interest (ROIs) in each subject. Results: The linearity of activity response was improved after the scatter correction (r=0.999). The y-intercept value of the regression line was 0.036 (p [de
Research of scatter correction on industry computed tomography
International Nuclear Information System (INIS)
Sun Shaohua; Gao Wenhuan; Zhang Li; Chen Zhiqiang
2002-01-01
In the scanning process of industry computer tomography, scatter blurs the reconstructed image. The grey values of pixels in the reconstructed image are away from what is true and such effect need to be corrected. If the authors use the conventional method of deconvolution, many steps of iteration are needed and the computing time is not satisfactory. The author discusses a method combining Ordered Subsets Convex algorithm and scatter model to implement scatter correction and promising results are obtained in both speed and image quality
Relativistic corrections to the elastic electron scattering from 208Pb
International Nuclear Information System (INIS)
Chandra, H.; Sauer, G.
1976-01-01
In the present work we have calculated the differential cross sections for the elastic electron scattering from 208 Pb using the charge distributions resulting from various corrections. The point proton and neutron mass distributions have been calculated from the spherical wave functions for 208 Pb obtained by Kolb et al. The relativistic correction to the nuclear charge distribution coming from the electromagnetic structure of the nucleon has been accomplished by assuming a linear superposition of Gaussian shapes for the proton and the neutron charge form factor. Results of this calculation are quite similar to an earlier calculation by Bertozzi et al., who have used a different wave function for 208 Pb and have assumed exponential smearing for the proton corresponding to the dipole fit for the form factor. Also in the present work, reason for the small spin orbit contribution to the effective charge distribution is discussed in some detail. It is also shown that the use of a single Gaussian shape for the proton smearing usually underestimates the actual theoretical cross section
QED radiative corrections to impact factors
International Nuclear Information System (INIS)
Kuraev, E.A.; Lipatov, L.N.; Shishkina, T.V.
2001-01-01
We consider radiative corrections to the electron and photon impact factors. The generalized eikonal representation for the e + e - scattering amplitude at high energies and fixed momentum transfers is violated by nonplanar diagrams. An additional contribution to the two-loop approximation appears from the Bethe-Heitler mechanism of fermion pair production with the identity of the fermions in the final state taken into account. The violation of the generalized eikonal representation is also related to the charge parity conservation in QED. A one-loop correction to the photon impact factor for small virtualities of the exchanged photon is obtained using the known results for the cross section of the e + e - production during photon-nuclei interactions
Higher Order Heavy Quark Corrections to Deep-Inelastic Scattering
Blümlein, Johannes; DeFreitas, Abilio; Schneider, Carsten
2015-04-01
The 3-loop heavy flavor corrections to deep-inelastic scattering are essential for consistent next-to-next-to-leading order QCD analyses. We report on the present status of the calculation of these corrections at large virtualities Q2. We also describe a series of mathematical, computer-algebraic and combinatorial methods and special function spaces, needed to perform these calculations. Finally, we briefly discuss the status of measuring αs (MZ), the charm quark mass mc, and the parton distribution functions at next-to-next-to-leading order from the world precision data on deep-inelastic scattering.
Higher order heavy quark corrections to deep-inelastic scattering
International Nuclear Information System (INIS)
Bluemlein, J.; Freitas, A. de; Johannes Kepler Univ., Linz; Schneider, C.
2014-11-01
The 3-loop heavy flavor corrections to deep-inelastic scattering are essential for consistent next-to-next-to-leading order QCD analyses. We report on the present status of the calculation of these corrections at large virtualities Q 2 . We also describe a series of mathematical, computer-algebraic and combinatorial methods and special function spaces, needed to perform these calculations. Finally, we briefly discuss the status of measuring α s (M Z ), the charm quark mass m c , and the parton distribution functions at next-to-next-to-leading order from the world precision data on deep-inelastic scattering.
Method for measuring multiple scattering corrections between liquid scintillators
Energy Technology Data Exchange (ETDEWEB)
Verbeke, J.M., E-mail: verbeke2@llnl.gov; Glenn, A.M., E-mail: glenn22@llnl.gov; Keefer, G.J., E-mail: keefer1@llnl.gov; Wurtz, R.E., E-mail: wurtz1@llnl.gov
2016-07-21
A time-of-flight method is proposed to experimentally quantify the fractions of neutrons scattering between scintillators. An array of scintillators is characterized in terms of crosstalk with this method by measuring a californium source, for different neutron energy thresholds. The spectral information recorded by the scintillators can be used to estimate the fractions of neutrons multiple scattering. With the help of a correction to Feynman's point model theory to account for multiple scattering, these fractions can in turn improve the mass reconstruction of fissile materials under investigation.
Corrections to the large-angle scattering amplitude
International Nuclear Information System (INIS)
Goloskokov, S.V.; Kudinov, A.V.; Kuleshov, S.P.
1979-01-01
High-energy behaviour of scattering amplitudes is considered within the frames of Logunov-Tavchelidze quasipotential approach. The representation of scattering amplitude of two scalar particles, convenient for the study of its asymptotic properties is given. Obtained are corrections of the main value of scattering amplitude of the first and the second orders in 1/p, where p is the pulse of colliding particles in the system of the inertia centre. An example of the obtained formulas use for a concrete quasipotential is given
Evaluation of a scattering correction method for high energy tomography
Tisseur, David; Bhatia, Navnina; Estre, Nicolas; Berge, Léonie; Eck, Daniel; Payan, Emmanuel
2018-01-01
One of the main drawbacks of Cone Beam Computed Tomography (CBCT) is the contribution of the scattered photons due to the object and the detector. Scattered photons are deflected from their original path after their interaction with the object. This additional contribution of the scattered photons results in increased measured intensities, since the scattered intensity simply adds to the transmitted intensity. This effect is seen as an overestimation in the measured intensity thus corresponding to an underestimation of absorption. This results in artifacts like cupping, shading, streaks etc. on the reconstructed images. Moreover, the scattered radiation provides a bias for the quantitative tomography reconstruction (for example atomic number and volumic mass measurement with dual-energy technique). The effect can be significant and difficult in the range of MeV energy using large objects due to higher Scatter to Primary Ratio (SPR). Additionally, the incident high energy photons which are scattered by the Compton effect are more forward directed and hence more likely to reach the detector. Moreover, for MeV energy range, the contribution of the photons produced by pair production and Bremsstrahlung process also becomes important. We propose an evaluation of a scattering correction technique based on the method named Scatter Kernel Superposition (SKS). The algorithm uses a continuously thickness-adapted kernels method. The analytical parameterizations of the scatter kernels are derived in terms of material thickness, to form continuously thickness-adapted kernel maps in order to correct the projections. This approach has proved to be efficient in producing better sampling of the kernels with respect to the object thickness. This technique offers applicability over a wide range of imaging conditions and gives users an additional advantage. Moreover, since no extra hardware is required by this approach, it forms a major advantage especially in those cases where
International Nuclear Information System (INIS)
Shidahara, Miho; Kato, Takashi; Kawatsu, Shoji; Yoshimura, Kumiko; Ito, Kengo; Watabe, Hiroshi; Kim, Kyeong Min; Iida, Hidehiro; Kato, Rikio
2005-01-01
An image-based scatter correction (IBSC) method was developed to convert scatter-uncorrected into scatter-corrected SPECT images. The purpose of this study was to validate this method by means of phantom simulations and human studies with 99m Tc-labeled tracers, based on comparison with the conventional triple energy window (TEW) method. The IBSC method corrects scatter on the reconstructed image I AC μb with Chang's attenuation correction factor. The scatter component image is estimated by convolving I AC μb with a scatter function followed by multiplication with an image-based scatter fraction function. The IBSC method was evaluated with Monte Carlo simulations and 99m Tc-ethyl cysteinate dimer SPECT human brain perfusion studies obtained from five volunteers. The image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were compared. Using data obtained from the simulations, the image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were found to be nearly identical for both gray and white matter. In human brain images, no significant differences in image contrast were observed between the IBSC and TEW methods. The IBSC method is a simple scatter correction technique feasible for use in clinical routine. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Shidahara, Miho; Kato, Takashi; Kawatsu, Shoji; Yoshimura, Kumiko; Ito, Kengo [National Center for Geriatrics and Gerontology Research Institute, Department of Brain Science and Molecular Imaging, Obu, Aichi (Japan); Watabe, Hiroshi; Kim, Kyeong Min; Iida, Hidehiro [National Cardiovascular Center Research Institute, Department of Investigative Radiology, Suita (Japan); Kato, Rikio [National Center for Geriatrics and Gerontology, Department of Radiology, Obu (Japan)
2005-10-01
An image-based scatter correction (IBSC) method was developed to convert scatter-uncorrected into scatter-corrected SPECT images. The purpose of this study was to validate this method by means of phantom simulations and human studies with {sup 99m}Tc-labeled tracers, based on comparison with the conventional triple energy window (TEW) method. The IBSC method corrects scatter on the reconstructed image I{sub AC}{sup {mu}}{sup b} with Chang's attenuation correction factor. The scatter component image is estimated by convolving I{sub AC}{sup {mu}}{sup b} with a scatter function followed by multiplication with an image-based scatter fraction function. The IBSC method was evaluated with Monte Carlo simulations and {sup 99m}Tc-ethyl cysteinate dimer SPECT human brain perfusion studies obtained from five volunteers. The image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were compared. Using data obtained from the simulations, the image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were found to be nearly identical for both gray and white matter. In human brain images, no significant differences in image contrast were observed between the IBSC and TEW methods. The IBSC method is a simple scatter correction technique feasible for use in clinical routine. (orig.)
Shidahara, Miho; Watabe, Hiroshi; Kim, Kyeong Min; Kato, Takashi; Kawatsu, Shoji; Kato, Rikio; Yoshimura, Kumiko; Iida, Hidehiro; Ito, Kengo
2005-10-01
An image-based scatter correction (IBSC) method was developed to convert scatter-uncorrected into scatter-corrected SPECT images. The purpose of this study was to validate this method by means of phantom simulations and human studies with 99mTc-labeled tracers, based on comparison with the conventional triple energy window (TEW) method. The IBSC method corrects scatter on the reconstructed image I(mub)AC with Chang's attenuation correction factor. The scatter component image is estimated by convolving I(mub)AC with a scatter function followed by multiplication with an image-based scatter fraction function. The IBSC method was evaluated with Monte Carlo simulations and 99mTc-ethyl cysteinate dimer SPECT human brain perfusion studies obtained from five volunteers. The image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were compared. Using data obtained from the simulations, the image counts and contrast of the scatter-corrected images obtained by the IBSC and TEW methods were found to be nearly identical for both gray and white matter. In human brain images, no significant differences in image contrast were observed between the IBSC and TEW methods. The IBSC method is a simple scatter correction technique feasible for use in clinical routine.
A model of diffraction scattering with unitary corrections
International Nuclear Information System (INIS)
Etim, E.; Malecki, A.; Satta, L.
1989-01-01
The inability of the multiple scattering model of Glauber and similar geometrical picture models to fit data at Collider energies, to fit low energy data at large momentum transfers and to explain the absence of multiple diffraction dips in the data is noted. It is argued and shown that a unitary correction to the multiple scattering amplitude gives rise to a better model and allows to fit all available data on nucleon-nucleon and nucleus-nucleus collisions at all energies and all momentum transfers. There are no multiple diffraction dips
Multiple scattering and attenuation corrections in Deep Inelastic Neutron Scattering experiments
International Nuclear Information System (INIS)
Dawidowski, J; Blostein, J J; Granada, J R
2006-01-01
Multiple scattering and attenuation corrections in Deep Inelastic Neutron Scattering experiments are analyzed. The theoretical basis of the method is stated, and a Monte Carlo procedure to perform the calculation is presented. The results are compared with experimental data. The importance of the accuracy in the description of the experimental parameters is tested, and the implications of the present results on the data analysis procedures is examined
Complete $O(\\alpha)$ QED corrections to polarized Compton scattering
Denner, Ansgar
1999-01-01
The complete QED corrections of O(alpha) to polarized Compton scattering are calculated for finite electron mass and including the real corrections induced by the processes e^- gamma -> e^- gamma gamma and e^- gamma -> e^- e^- e^+. All relevant formulas are listed in a form that is well suited for a direct implementation in computer codes. We present a detailed numerical discussion of the O(alpha)-corrected cross section and the left-right asymmetry in the energy range of present and future Compton polarimeters, which are used to determine the beam polarization of high-energetic e^+- beams. For photons with energies of a few eV and electrons with SLC energies or smaller, the corrections are of the order of a few per mille. In the energy range of future e^+e^- colliders, however, they reach 1-2% and cannot be neglected in a precision polarization measurement.
Automatic Power Factor Correction Using Capacitive Bank
Mr.Anant Kumar Tiwari,; Mrs. Durga Sharma
2014-01-01
The power factor correction of electrical loads is a problem common to all industrial companies. Earlier the power factor correction was done by adjusting the capacitive bank manually [1]. The automated power factor corrector (APFC) using capacitive load bank is helpful in providing the power factor correction. Proposed automated project involves measuring the power factor value from the load using microcontroller. The design of this auto-adjustable power factor correction is ...
Coastal Zone Color Scanner atmospheric correction algorithm - Multiple scattering effects
Gordon, Howard R.; Castano, Diego J.
1987-01-01
Errors due to multiple scattering which are expected to be encountered in application of the current Coastal Zone Color Scanner (CZCS) atmospheric correction algorithm are analyzed. The analysis is based on radiative transfer computations in model atmospheres, in which the aerosols and molecules are distributed vertically in an exponential manner, with most of the aerosol scattering located below the molecular scattering. A unique feature of the analysis is that it is carried out in scan coordinates rather than typical earth-sun coordinates, making it possible to determine the errors along typical CZCS scan lines. Information provided by the analysis makes it possible to judge the efficacy of the current algorithm with the current sensor and to estimate the impact of the algorithm-induced errors on a variety of applications.
On the radiative corrections to the neutrino deep inelastic scattering
International Nuclear Information System (INIS)
Bardin, D.Yu.; Dokuchaeva, V.A.
1986-01-01
A unique set of formulae is presented for the radiative corrections to the double differential cross section of deep inelastic neutrino scattering in channels of charged and neutral currents within a simple quark parton model in a renormalization scheme on mass-shell. It is shown that these cross sections when being integrated up to the one-dimensional distribution or up to the total cross section reproduce many results existing in the literature
Comparative evaluation of scatter correction techniques in 3D positron emission tomography
Zaidi, H
2000-01-01
Much research and development has been concentrated on the scatter compensation required for quantitative 3D PET. Increasingly sophisticated scatter correction procedures are under investigation, particularly those based on accurate scatter models, and iterative reconstruction-based scatter compensation approaches. The main difference among the correction methods is the way in which the scatter component in the selected energy window is estimated. Monte Carlo methods give further insight and might in themselves offer a possible correction procedure. Methods: Five scatter correction methods are compared in this paper where applicable. The dual-energy window (DEW) technique, the convolution-subtraction (CVS) method, two variants of the Monte Carlo-based scatter correction technique (MCBSC1 and MCBSC2) and our newly developed statistical reconstruction-based scatter correction (SRBSC) method. These scatter correction techniques are evaluated using Monte Carlo simulation studies, experimental phantom measurements...
An empirical correction for moderate multiple scattering in super-heterodyne light scattering.
Botin, Denis; Mapa, Ludmila Marotta; Schweinfurth, Holger; Sieber, Bastian; Wittenberg, Christopher; Palberg, Thomas
2017-05-28
Frequency domain super-heterodyne laser light scattering is utilized in a low angle integral measurement configuration to determine flow and diffusion in charged sphere suspensions showing moderate to strong multiple scattering. We introduce an empirical correction to subtract the multiple scattering background and isolate the singly scattered light. We demonstrate the excellent feasibility of this simple approach for turbid suspensions of transmittance T ≥ 0.4. We study the particle concentration dependence of the electro-kinetic mobility in low salt aqueous suspension over an extended concentration regime and observe a maximum at intermediate concentrations. We further use our scheme for measurements of the self-diffusion coefficients in the fluid samples in the absence or presence of shear, as well as in polycrystalline samples during crystallization and coarsening. We discuss the scope and limits of our approach as well as possible future applications.
Fully 3D iterative scatter-corrected OSEM for HRRT PET using a GPU
Energy Technology Data Exchange (ETDEWEB)
Kim, Kyung Sang; Ye, Jong Chul, E-mail: kssigari@kaist.ac.kr, E-mail: jong.ye@kaist.ac.kr [Bio-Imaging and Signal Processing Lab., Department of Bio and Brain Engineering, Korea Advanced Institute of Science and Technology (KAIST), 335 Gwahak-no, Yuseong-gu, Daejon 305-701 (Korea, Republic of)
2011-08-07
Accurate scatter correction is especially important for high-resolution 3D positron emission tomographies (PETs) such as high-resolution research tomograph (HRRT) due to large scatter fraction in the data. To address this problem, a fully 3D iterative scatter-corrected ordered subset expectation maximization (OSEM) in which a 3D single scatter simulation (SSS) is alternatively performed with a 3D OSEM reconstruction was recently proposed. However, due to the computational complexity of both SSS and OSEM algorithms for a high-resolution 3D PET, it has not been widely used in practice. The main objective of this paper is, therefore, to accelerate the fully 3D iterative scatter-corrected OSEM using a graphics processing unit (GPU) and verify its performance for an HRRT. We show that to exploit the massive thread structures of the GPU, several algorithmic modifications are necessary. For SSS implementation, a sinogram-driven approach is found to be more appropriate compared to a detector-driven approach, as fast linear interpolation can be performed in the sinogram domain through the use of texture memory. Furthermore, a pixel-driven backprojector and a ray-driven projector can be significantly accelerated by assigning threads to voxels and sinograms, respectively. Using Nvidia's GPU and compute unified device architecture (CUDA), the execution time of a SSS is less than 6 s, a single iteration of OSEM with 16 subsets takes 16 s, and a single iteration of the fully 3D scatter-corrected OSEM composed of a SSS and six iterations of OSEM takes under 105 s for the HRRT geometry, which corresponds to acceleration factors of 125x and 141x for OSEM and SSS, respectively. The fully 3D iterative scatter-corrected OSEM algorithm is validated in simulations using Geant4 application for tomographic emission and in actual experiments using an HRRT.
Coulomb corrections to scattering length and effective radius
International Nuclear Information System (INIS)
Mur, V.D.; Kudryavtsev, A.E.; Popov, V.S.
1983-01-01
The problem considered is extraction of the ''purely nuclear'' scattering length asub(s) (corresponding to the strong potential Vsub(s) at the Coulomb interaction switched off) from the Coulomb-nuclear scattering length asub(cs), which is an object of experimental measurement. The difference between asub(s) and asub(cs) is especially large if the potential Vsub(s) has a level (real or virtual) with an energy close to zero. For this case formulae are obtained relating the scattering lengths asub(s) and asub(cs), as well as the effective radii rsub(s) and rsub(cs). The results are extended to states with arbitrary angular momenta l. It is shown that the Coulomb correction is especially large for the coefficient with ksup(2l) in the expansion of the effective radius; in this case the correction contains a large logarithm ln(asub(B)/rsub(0)). The Coulomb renormalization of other terms in the effective radius espansion is of order (rsub(0)/asub(B)), where r 0 is the nuclear force radius, asub(B) is the Bohr radius. The obtained formulae are tried on a number of model potentials Vsub(s), used in nuclear physics
The analysis and correction of neutron scattering effects in neutron imaging
International Nuclear Information System (INIS)
Raine, D.A.; Brenizer, J.S.
1997-01-01
A method of correcting for the scattering effects present in neutron radiographic and computed tomographic imaging has been developed. Prior work has shown that beam, object, and imaging system geometry factors, such as the L/D ratio and angular divergence, are the primary sources contributing to the degradation of neutron images. With objects smaller than 20--40 mm in width, a parallel beam approximation can be made where the effects from geometry are negligible. Factors which remain important in the image formation process are the pixel size of the imaging system, neutron scattering, the size of the object, the conversion material, and the beam energy spectrum. The Monte Carlo N-Particle transport code, version 4A (MCNP4A), was used to separate and evaluate the effect that each of these parameters has on neutron image data. The simulations were used to develop a correction algorithm which is easy to implement and requires no a priori knowledge of the object. The correction algorithm is based on the determination of the object scatter function (OSF) using available data outside the object to estimate the shape and magnitude of the OSF based on a Gaussian functional form. For objects smaller than 1 mm (0.04 in.) in width, the correction function can be well approximated by a constant function. Errors in the determination and correction of the MCNP simulated neutron scattering component were under 5% and larger errors were only noted in objects which were at the extreme high end of the range of object sizes simulated. The Monte Carlo data also indicated that scattering does not play a significant role in the blurring of neutron radiographic and tomographic images. The effect of neutron scattering on computed tomography is shown to be minimal at best, with the most serious effect resulting when the basic backprojection method is used
Two-loop fermionic corrections to massive Bhabha scattering
Energy Technology Data Exchange (ETDEWEB)
Actis, S.; Riemann, T. [Deutsches Elektronen-Synchrotron (DESY), Zeuthen (Germany); Czakon, M. [Wuerzburg Univ. (Germany). Inst. fuer Theoretische Physik und Astrophysik]|[Institute of Nuclear Physics, NSCR DEMOKRITOS, Athens (Greece); Gluza, J. [Silesia Univ., Katowice (Poland). Inst. of Physics
2007-05-15
We evaluate the two-loop corrections to Bhabha scattering from fermion loops in the context of pure Quantum Electrodynamics. The differential cross section is expressed by a small number of Master Integrals with exact dependence on the fermion masses m{sub e}, m{sub f} and the Mandelstam invariants s, t, u. We determine the limit of fixed scattering angle and high energy, assuming the hierarchy of scales m{sup 2}{sub e}<
Multiple-scattering corrections to the Beer-Lambert law
International Nuclear Information System (INIS)
Zardecki, A.
1983-01-01
The effect of multiple scattering on the validity of the Beer-Lambert law is discussed for a wide range of particle-size parameters and optical depths. To predict the amount of received radiant power, appropriate correction terms are introduced. For particles larger than or comparable to the wavelength of radiation, the small-angle approximation is adequate; whereas for small densely packed particles, the diffusion theory is advantageously employed. These two approaches are used in the context of the problem of laser-beam propagation in a dense aerosol medium. In addition, preliminary results obtained by using a two-dimensional finite-element discrete-ordinates transport code are described. Multiple-scattering effects for laser propagation in fog, cloud, rain, and aerosol cloud are modeled
Meson exchange corrections in deep inelastic scattering on deuteron
International Nuclear Information System (INIS)
Kaptari, L.P.; Titov, A.I.
1989-01-01
Starting with the general equations of motion of the nucleons interacting with the mesons the one-particle Schroedinger-like equation for the nucleon wave function and the deep inelastic scattering amplitude with the meson-exchange currents are obtained. Effective pion-, sigma-, and omega-meson exchanges are considered. It is found that the mesonic corrections only partially (about 60%) restore the energy sum rule breaking because of the nucleon off-mass-shell effects in nuclei. This results contradicts with the prediction based on the calculation of the energy sum rule limited by the second order of the nucleon-meson vertex and static approximation. 17 refs.; 3 figs
Energy Technology Data Exchange (ETDEWEB)
Bootsma, G. J., E-mail: Gregory.Bootsma@rmp.uhn.on.ca [Radiation Medicine Program, Princess Margaret Cancer Centre, Toronto, Ontario M5G 2M9 (Canada); Verhaegen, F. [Department of Radiation Oncology - MAASTRO, GROW—School for Oncology and Developmental Biology, Maastricht University Medical Center, Maastricht 6201 BN (Netherlands); Medical Physics Unit, Department of Oncology, McGill University, Montreal, Quebec H3G 1A4 (Canada); Jaffray, D. A. [Radiation Medicine Program, Princess Margaret Cancer Centre, Toronto, Ontario M5G 2M9 (Canada); Ontario Cancer Institute, Princess Margaret Cancer Centre, Toronto, Ontario M5G 2M9 (Canada); Department of Radiation Oncology, University of Toronto, Toronto, Ontario M5G 2M9 (Canada)
2015-01-15
suitable GOF metric with strong correlation with the actual error of the scatter fit, S{sub F}. Fitting the scatter distribution to a limited sum of sine and cosine functions using a low-pass filtered fast Fourier transform provided a computationally efficient and accurate fit. The CMCF algorithm reduces the number of photon histories required by over four orders of magnitude. The simulated experiments showed that using a compensator reduced the computational time by a factor between 1.5 and 1.75. The scatter estimates for the simulated and measured data were computed between 35–93 s and 114–122 s, respectively, using 16 Intel Xeon cores (3.0 GHz). The CMCF scatter correction improved the contrast-to-noise ratio by 10%–50% and reduced the reconstruction error to under 3% for the simulated phantoms. Conclusions: The novel CMCF algorithm significantly reduces the computation time required to estimate the scatter distribution by reducing the statistical noise in the MC scatter estimate and limiting the number of projection angles that must be simulated. Using the scatter estimate provided by the CMCF algorithm to correct both simulated and real projection data showed improved reconstruction image quality.
DEFF Research Database (Denmark)
Keller, Sune H; Svarer, Claus; Sibomana, Merence
2013-01-01
scatter correction in the μ-map reconstruction and total variation filtering to the transmission processing. Results: Comparing MAP-TR and the new TXTV with gold standard CT-based attenuation correction, we found that TXTV has less bias as compared to MAP-TR. We also compared images acquired at the HRRT......In the standard software for the Siemens high-resolution research tomograph (HRRT) positron emission tomography (PET) scanner the most commonly used segmentation in the μ -map reconstruction for human brain scans is maximum a posteriori for transmission (MAP-TR). Bias in the lower cerebellum...
Scatter correction method for x-ray CT using primary modulation: Phantom studies
International Nuclear Information System (INIS)
Gao Hewei; Fahrig, Rebecca; Bennett, N. Robert; Sun Mingshan; Star-Lack, Josh; Zhu Lei
2010-01-01
elliptical annulus (30 cm in the minor axis and 38 cm in the major axis) and with a circular annulus (38 cm in diameter). Conclusions: On the three phantom studies, good scatter correction performance of the proposed method has been demonstrated using both image comparisons and quantitative analysis. The theory and experiments demonstrate that a strong primary modulation that possesses a low transmission factor and a high modulation frequency is preferred for high scatter correction accuracy.
Monte Carlo evaluation of scattering correction methods in 131I studies using pinhole collimator
International Nuclear Information System (INIS)
López Díaz, Adlin; San Pedro, Aley Palau; Martín Escuela, Juan Miguel; Rodríguez Pérez, Sunay; Díaz García, Angelina
2017-01-01
Scattering is quite important for image activity quantification. In order to study the scattering factors and the efficacy of 3 multiple window energy scatter correction methods during 131 I thyroid studies with a pinhole collimator (5 mm hole) a Monte Carlo simulation (MC) was developed. The GAMOS MC code was used to model the gamma camera and the thyroid source geometry. First, to validate the MC gamma camera pinhole-source model, sensibility in air and water of the simulated and measured thyroid phantom geometries were compared. Next, simulations to investigate scattering and the result of triple energy (TEW), Double energy (DW) and Reduced double (RDW) energy windows correction methods were performed for different thyroid sizes and depth thicknesses. The relative discrepancies to MC real event were evaluated. Results: The accuracy of the GAMOS MC model was verified and validated. The image’s scattering contribution was significant, between 27-40 %. The discrepancies between 3 multiple window energy correction method results were significant (between 9-86 %). The Reduce Double Window methods (15%) provide discrepancies of 9-16 %. Conclusions: For the simulated thyroid geometry with pinhole, the RDW (15 %) was the most effective. (author)
Investigating the effect and photon scattering correction in isotopic scanning with gamma and SPECT
International Nuclear Information System (INIS)
Movafeghi, Amir
1997-01-01
Nowdays medical imaging systems has been become a very important tool in medicine, both in diagnosis and treatment. With the fast improvement in the computer sciences in the last three decades, three dimensional imaging systems or topographic systems has been developed for the daily applications. Among the different methods, for now X-ray Computerized tomography scanning, Magnetic Resonance Imaging, Single Photon Emission Computerized Tomography and Positron Emission tomography have been found many clinical application. SPECT and PET imaging systems are working with the use of emitting photons from special radioisotopes. In these two systems, image is reconstructed from a distribution of radioisotope in the human body's organs. In SPECT accuracy of data quantification for image reconstruction has influenced from photon attenuation, photon scattering, statistical noises and variation in detector response due to distance. Except scattering other three factors could be modeled and compensated with relatively simple models. Photon scattering is a complex process and usually semiemperical methods is used for its modeling. The effect of scattering photons on images was considered. This survey was done in both lab and clinical cases. Radioisotopes were 192 Ir and 99m Tc. 192 Ir is a solid source with the half-life of 73 days and is used at industrial radiography application. At the beginning, models and methods, were established by the help of 192 Ir. Then at the final stage, they were developed to use for 99m Tc. There are different methods for the error correction of scattered photons. A method from the 'window subtraction' group has been developed for lab cases. Generally, in this method with the use of adjacent window of the photopeak window, scattered photons are subtracted from the original count. A Monte Carlo simulation is used for better evaluation of results. In the clinical section , a dual head SPECT system was (ADAC system of Shariati hospital at Tehran). The
Aethalometer multiple scattering correction Cref for mineral dust aerosols
Di Biagio, Claudia; Formenti, Paola; Cazaunau, Mathieu; Pangui, Edouard; Marchand, Nicolas; Doussin, Jean-François
2017-08-01
In this study we provide a first estimate of the Aethalometer multiple scattering correction Cref for mineral dust aerosols. Cref is an empirical constant used to correct the aerosol absorption coefficient measurements for the multiple scattering artefact of the Aethalometer; i.e. the filter fibres on which aerosols are deposited scatter light and this is miscounted as absorption. The Cref at 450 and 660 nm was obtained from the direct comparison of Aethalometer data (Magee Sci. AE31) with (i) the absorption coefficient calculated as the difference between the extinction and scattering coefficients measured by a Cavity Attenuated Phase Shift Extinction analyser (CAPS PMex) and a nephelometer respectively at 450 nm and (ii) the absorption coefficient from a MAAP (Multi-Angle Absorption Photometer) at 660 nm. Measurements were performed on seven dust aerosol samples generated in the laboratory by the mechanical shaking of natural parent soils issued from different source regions worldwide. The single scattering albedo (SSA) at 450 and 660 nm and the size distribution of the aerosols were also measured. Cref for mineral dust varies between 1.81 and 2.56 for a SSA of 0.85-0.96 at 450 nm and between 1.75 and 2.28 for a SSA of 0.98-0.99 at 660 nm. The calculated mean for dust is 2.09 (±0.22) at 450 nm and 1.92 (±0.17) at 660 nm. With this new Cref the dust absorption coefficient by the Aethalometer is about 2 % (450 nm) and 11 % (660 nm) higher than that obtained by using Cref = 2.14 at both 450 and 660 nm, as usually assumed in the literature. This difference induces a change of up to 3 % in the dust SSA at 660 nm. The Cref seems to be independent of the fine and coarse particle size fractions, and so the obtained Cref can be applied to dust both close to sources and following transport. Additional experiments performed with pure kaolinite minerals and polluted ambient aerosols indicate Cref of 2.49 (±0.02) and 2.32 (±0.01) at 450 and 660 nm (SSA = 0.96-0.97) for
Mentrup, Detlef; Jockel, Sascha; Menser, Bernd; Neitzel, Ulrich
2016-06-01
The aim of this work was to experimentally compare the contrast improvement factors (CIFs) of a newly developed software-based scatter correction to the CIFs achieved by an antiscatter grid. To this end, three aluminium discs were placed in the lung, the retrocardial and the abdominal areas of a thorax phantom, and digital radiographs of the phantom were acquired both with and without a stationary grid. The contrast generated by the discs was measured in both images, and the CIFs achieved by grid usage were determined for each disc. Additionally, the non-grid images were processed with a scatter correction software. The contrasts generated by the discs were determined in the scatter-corrected images, and the corresponding CIFs were calculated. The CIFs obtained with the grid and with the software were in good agreement. In conclusion, the experiment demonstrates quantitatively that software-based scatter correction allows restoring the image contrast of a non-grid image in a manner comparable with an antiscatter grid. © The Author 2015. Published by Oxford University Press. All rights reserved. For Permissions, please email: journals.permissions@oup.com.
Energy Technology Data Exchange (ETDEWEB)
Yang, J.; Kuikka, J.T.; Vanninen, E.; Laensimies, E. [Kuopio Univ. Hospital (Finland). Dept. of Clinical Physiology and Nuclear Medicine; Kauppinen, T.; Patomaeki, L. [Kuopio Univ. (Finland). Dept. of Applied Physics
1999-05-01
Photon scatter is one of the most important factors degrading the quantitative accuracy of SPECT images. Many scatter correction methods have been proposed. The single isotope method was proposed by us. Aim: We evaluate the scatter correction method of improving the quality of images by acquiring emission and transmission data simultaneously with single isotope scan. Method: To evaluate the proposed scatter correction method, a contrast and linearity phantom was studied. Four female patients with fibromyalgia (FM) syndrome and four with chronic back pain (BP) were imaged. Grey-to-cerebellum (G/C) and grey-to-white matter (G/W) ratios were determined by one skilled operator for 12 regions of interest (ROIs) in each subject. Results: The linearity of activity response was improved after the scatter correction (r=0.999). The y-intercept value of the regression line was 0.036 (p<0.0001) after scatter correction and the slope was 0.954. Pairwise correlation indicated the agreement between nonscatter corrected and scatter corrected images. Reconstructed slices before and after scatter correction demonstrate a good correlation in the quantitative accuracy of radionuclide concentration. G/C values have significant correlation coefficients between original and corrected data. Conclusion: The transaxial images of human brain studies show that the scatter correction using single isotope in simultaneous transmission and emission tomography provides a good scatter compensation. The contrasts were increased on all 12 ROIs. The scatter compensation enhanced details of physiological lesions. (orig.) [Deutsch] Die Photonenstreuung gehoert zu den wichtigsten Faktoren, die die quantitative Genauigkeit von SPECT-Bildern vermindern. Es wurde eine ganze Reihe von Methoden zur Streuungskorrektur vorgeschlagen. Von uns wurde die Einzelisotopen-Methode empfohlen. Ziel: Wir untersuchten die Streuungskorrektur-Methode zur Verbesserung der Bildqualitaet durch simultane Gewinnung von Emissions
Library based x-ray scatter correction for dedicated cone beam breast CT
International Nuclear Information System (INIS)
Shi, Linxi; Zhu, Lei; Vedantham, Srinivasan; Karellas, Andrew
2016-01-01
Purpose: The image quality of dedicated cone beam breast CT (CBBCT) is limited by substantial scatter contamination, resulting in cupping artifacts and contrast-loss in reconstructed images. Such effects obscure the visibility of soft-tissue lesions and calcifications, which hinders breast cancer detection and diagnosis. In this work, we propose a library-based software approach to suppress scatter on CBBCT images with high efficiency, accuracy, and reliability. Methods: The authors precompute a scatter library on simplified breast models with different sizes using the GEANT4-based Monte Carlo (MC) toolkit. The breast is approximated as a semiellipsoid with homogeneous glandular/adipose tissue mixture. For scatter correction on real clinical data, the authors estimate the breast size from a first-pass breast CT reconstruction and then select the corresponding scatter distribution from the library. The selected scatter distribution from simplified breast models is spatially translated to match the projection data from the clinical scan and is subtracted from the measured projection for effective scatter correction. The method performance was evaluated using 15 sets of patient data, with a wide range of breast sizes representing about 95% of general population. Spatial nonuniformity (SNU) and contrast to signal deviation ratio (CDR) were used as metrics for evaluation. Results: Since the time-consuming MC simulation for library generation is precomputed, the authors’ method efficiently corrects for scatter with minimal processing time. Furthermore, the authors find that a scatter library on a simple breast model with only one input parameter, i.e., the breast diameter, sufficiently guarantees improvements in SNU and CDR. For the 15 clinical datasets, the authors’ method reduces the average SNU from 7.14% to 2.47% in coronal views and from 10.14% to 3.02% in sagittal views. On average, the CDR is improved by a factor of 1.49 in coronal views and 2.12 in sagittal
Library based x-ray scatter correction for dedicated cone beam breast CT
Energy Technology Data Exchange (ETDEWEB)
Shi, Linxi; Zhu, Lei, E-mail: leizhu@gatech.edu [Nuclear and Radiological Engineering and Medical Physics Programs, The George W. Woodruff School of Mechanical Engineering, Georgia Institute of Technology, Atlanta, Georgia 30332 (United States); Vedantham, Srinivasan; Karellas, Andrew [Department of Radiology, University of Massachusetts Medical School, Worcester, Massachusetts 01655 (United States)
2016-08-15
Purpose: The image quality of dedicated cone beam breast CT (CBBCT) is limited by substantial scatter contamination, resulting in cupping artifacts and contrast-loss in reconstructed images. Such effects obscure the visibility of soft-tissue lesions and calcifications, which hinders breast cancer detection and diagnosis. In this work, we propose a library-based software approach to suppress scatter on CBBCT images with high efficiency, accuracy, and reliability. Methods: The authors precompute a scatter library on simplified breast models with different sizes using the GEANT4-based Monte Carlo (MC) toolkit. The breast is approximated as a semiellipsoid with homogeneous glandular/adipose tissue mixture. For scatter correction on real clinical data, the authors estimate the breast size from a first-pass breast CT reconstruction and then select the corresponding scatter distribution from the library. The selected scatter distribution from simplified breast models is spatially translated to match the projection data from the clinical scan and is subtracted from the measured projection for effective scatter correction. The method performance was evaluated using 15 sets of patient data, with a wide range of breast sizes representing about 95% of general population. Spatial nonuniformity (SNU) and contrast to signal deviation ratio (CDR) were used as metrics for evaluation. Results: Since the time-consuming MC simulation for library generation is precomputed, the authors’ method efficiently corrects for scatter with minimal processing time. Furthermore, the authors find that a scatter library on a simple breast model with only one input parameter, i.e., the breast diameter, sufficiently guarantees improvements in SNU and CDR. For the 15 clinical datasets, the authors’ method reduces the average SNU from 7.14% to 2.47% in coronal views and from 10.14% to 3.02% in sagittal views. On average, the CDR is improved by a factor of 1.49 in coronal views and 2.12 in sagittal
Factorization and non-factorization in diffractive hard scattering
International Nuclear Information System (INIS)
Berera, Arjun
1997-01-01
Factorization, in the sense defined for inclusive hard scattering, is discussed for diffractive hard scattering. A factorization theorem similar to its inclusive counterpart is presented for diffractive DIS. For hadron-hadron diffractive hard scattering, in contrast to its inclusive counterpart, the expected breakdown of factorization is discussed. Cross section estimates are given from a simple field theory model for non-factorizing double-pomeron-exchange (DPE) dijet production with and without account for Sudakov suppression
Wall attenuation and scatter corrections for ion chambers: measurements versus calculations
Energy Technology Data Exchange (ETDEWEB)
Rogers, D W.O.; Bielajew, A F [National Research Council of Canada, Ottawa, ON (Canada). Div. of Physics
1990-08-01
In precision ion chamber dosimetry in air, wall attenuation and scatter are corrected for A{sub wall} (K{sub att} in IAEA terminology, K{sub w}{sup -1} in standards laboratory terminology). Using the EGS4 system the authors show that Monte Carlo calculated A{sub wall} factors predict relative variations in detector response with wall thickness which agree with all available experimental data within a statistical uncertainty of less than 0.1%. They calculated correction factors for use in exposure and air kerma standards are different by up to 1% from those obtained by extrapolating these same measurements. Using calculated correction factors would imply increases of 0.7-1.0% in the exposure and air kerma standards based on spherical and large diameter, large length cylindrical chambers and decreases of 0.3-0.5% for standards based on large diameter pancake chambers. (author).
Virtual two-loop corrections to Bhabha scattering
International Nuclear Information System (INIS)
Bjoerkevoll, K.S.
1992-03-01
The author has developed methods for the calculation of contributions from six ladder-like diagrams to Bhabha scattering. The leading terms both for separate diagrams and for the sum of the gauge-invariant set of all diagrams have been calculated. The study has been limited to contributions from Feynman diagrams without real photons, and all calculations have been done with s>> |t| >>m 2 , where s is the center of mass energy squared, t is the square of the transferred four-momentum, and m is the electron mass. For the separate diagrams the results depend upon how λ 2 is related to s, |t| and m 2 , whereas the leading term of the sum of the six diagrams is the same in the cases that have been considered. The methods described should be valuable for calculations of contributions from other Feynman diagrams, in particular QED corrections to Bhabha scattering or pair production at small angles. 23 refs., 5 figs., 5 tabs
Jo, Byung-Du; Lee, Young-Jin; Kim, Dae-Hong; Jeon, Pil-Hyun; Kim, Hee-Joung
2014-03-01
In conventional digital radiography (DR) using a dual energy subtraction technique, a significant fraction of the detected photons are scattered within the body, resulting in the scatter component. Scattered radiation can significantly deteriorate image quality in diagnostic X-ray imaging systems. Various methods of scatter correction, including both measurement and non-measurement-based methods have been proposed in the past. Both methods can reduce scatter artifacts in images. However, non-measurement-based methods require a homogeneous object and have insufficient scatter component correction. Therefore, we employed a measurement-based method to correct for the scatter component of inhomogeneous objects from dual energy DR (DEDR) images. We performed a simulation study using a Monte Carlo simulation with a primary modulator, which is a measurement-based method for the DEDR system. The primary modulator, which has a checkerboard pattern, was used to modulate primary radiation. Cylindrical phantoms of variable size were used to quantify imaging performance. For scatter estimation, we used Discrete Fourier Transform filtering. The primary modulation method was evaluated using a cylindrical phantom in the DEDR system. The scatter components were accurately removed using a primary modulator. When the results acquired with scatter correction and without correction were compared, the average contrast-to-noise ratio (CNR) with the correction was 1.35 times higher than that obtained without correction, and the average root mean square error (RMSE) with the correction was 38.00% better than that without correction. In the subtraction study, the average CNR with correction was 2.04 (aluminum subtraction) and 1.38 (polymethyl methacrylate (PMMA) subtraction) times higher than that obtained without the correction. The analysis demonstrated the accuracy of scatter correction and the improvement of image quality using a primary modulator and showed the feasibility of
Aethalometer multiple scattering correction Cref for mineral dust aerosols
Directory of Open Access Journals (Sweden)
C. Di Biagio
2017-08-01
Full Text Available In this study we provide a first estimate of the Aethalometer multiple scattering correction Cref for mineral dust aerosols. Cref is an empirical constant used to correct the aerosol absorption coefficient measurements for the multiple scattering artefact of the Aethalometer; i.e. the filter fibres on which aerosols are deposited scatter light and this is miscounted as absorption. The Cref at 450 and 660 nm was obtained from the direct comparison of Aethalometer data (Magee Sci. AE31 with (i the absorption coefficient calculated as the difference between the extinction and scattering coefficients measured by a Cavity Attenuated Phase Shift Extinction analyser (CAPS PMex and a nephelometer respectively at 450 nm and (ii the absorption coefficient from a MAAP (Multi-Angle Absorption Photometer at 660 nm. Measurements were performed on seven dust aerosol samples generated in the laboratory by the mechanical shaking of natural parent soils issued from different source regions worldwide. The single scattering albedo (SSA at 450 and 660 nm and the size distribution of the aerosols were also measured. Cref for mineral dust varies between 1.81 and 2.56 for a SSA of 0.85–0.96 at 450 nm and between 1.75 and 2.28 for a SSA of 0.98–0.99 at 660 nm. The calculated mean for dust is 2.09 (±0.22 at 450 nm and 1.92 (±0.17 at 660 nm. With this new Cref the dust absorption coefficient by the Aethalometer is about 2 % (450 nm and 11 % (660 nm higher than that obtained by using Cref = 2.14 at both 450 and 660 nm, as usually assumed in the literature. This difference induces a change of up to 3 % in the dust SSA at 660 nm. The Cref seems to be independent of the fine and coarse particle size fractions, and so the obtained Cref can be applied to dust both close to sources and following transport. Additional experiments performed with pure kaolinite minerals and polluted ambient aerosols indicate Cref of 2.49 (±0.02 and 2
International Nuclear Information System (INIS)
Mayers, J.; Cywinski, R.
1985-03-01
Some of the approximations commonly used for the analytical estimation of multiple scattering corrections to thermal neutron elastic scattering data from cylindrical and plane slab samples have been tested using a Monte Carlo program. It is shown that the approximations are accurate for a wide range of sample geometries and scattering cross-sections. Neutron polarisation analysis provides the most stringent test of multiple scattering calculations as multiply scattered neutrons may be redistributed not only geometrically but also between the spin flip and non spin flip scattering channels. A very simple analytical technique for correcting for multiple scattering in neutron polarisation analysis has been tested using the Monte Carlo program and has been shown to work remarkably well in most circumstances. (author)
Coherent scattering and matrix correction in bone-lead measurements
International Nuclear Information System (INIS)
Todd, A.C.
2000-01-01
The technique of K-shell x-ray fluorescence of lead in bone has been used in many studies of the health effects of lead. This paper addresses one aspect of the technique, namely the coherent conversion factor (CCF) which converts between the matrix of the calibration standards and those of human bone. The CCF is conventionally considered a constant but is a function of scattering angle, energy and the elemental composition of the matrices. The aims of this study were to quantify the effect on the CCF of several assumptions which may not have been tested adequately and to compare the CCFs for plaster of Paris (the present matrix of calibration standards) and a synthetic apatite matrix. The CCF was calculated, using relativistic form factors, for published compositions of bone, both assumed and assessed compositions of plaster, and the synthetic apatite. The main findings of the study were, first, that impurities in plaster, lead in the plaster or bone matrices, coherent scatter from non-bone tissues and the individual subject's measurement geometry are all minor or negligible effects; and, second, that the synthetic apatite matrix is more representative of bone mineral than is plaster of Paris. (author)
International Nuclear Information System (INIS)
Cheng, J-C; Rahmim, Arman; Blinder, Stephan; Camborde, Marie-Laure; Raywood, Kelvin; Sossi, Vesna
2007-01-01
We describe an ordinary Poisson list-mode expectation maximization (OP-LMEM) algorithm with a sinogram-based scatter correction method based on the single scatter simulation (SSS) technique and a random correction method based on the variance-reduced delayed-coincidence technique. We also describe a practical approximate scatter and random-estimation approach for dynamic PET studies based on a time-averaged scatter and random estimate followed by scaling according to the global numbers of true coincidences and randoms for each temporal frame. The quantitative accuracy achieved using OP-LMEM was compared to that obtained using the histogram-mode 3D ordinary Poisson ordered subset expectation maximization (3D-OP) algorithm with similar scatter and random correction methods, and they showed excellent agreement. The accuracy of the approximated scatter and random estimates was tested by comparing time activity curves (TACs) as well as the spatial scatter distribution from dynamic non-human primate studies obtained from the conventional (frame-based) approach and those obtained from the approximate approach. An excellent agreement was found, and the time required for the calculation of scatter and random estimates in the dynamic studies became much less dependent on the number of frames (we achieved a nearly four times faster performance on the scatter and random estimates by applying the proposed method). The precision of the scatter fraction was also demonstrated for the conventional and the approximate approach using phantom studies
Efficiency factors in Mie scattering
International Nuclear Information System (INIS)
Nussenzveig, H.M.
1980-04-01
Asymptotic approximation to the Mie efficiency factors for extinction, absorption and radiation pressure, derived from complex angular momentum theory and averaged over Δβ approximately π (β = size parameter), are given and compared with the exact results. For complex refractive indices N = n + i kappa with 1.1 -2 - 10 -3 % between β = 10 and β = 1000, and computing time is reduced by a factor of order β, so that the Mie formulae can advantageously by replaced by the asymptotic ones in most applications. (Author) [pt
Keller, Sune H; Svarer, Claus; Sibomana, Merence
2013-09-01
In the standard software for the Siemens high-resolution research tomograph (HRRT) positron emission tomography (PET) scanner the most commonly used segmentation in the μ -map reconstruction for human brain scans is maximum a posteriori for transmission (MAP-TR). Bias in the lower cerebellum and pons in HRRT brain images have been reported. The two main sources of the problem with MAP-TR are poor bone/soft tissue segmentation below the brain and overestimation of bone mass in the skull. We developed the new transmission processing with total variation (TXTV) method that introduces scatter correction in the μ-map reconstruction and total variation filtering to the transmission processing. Comparing MAP-TR and the new TXTV with gold standard CT-based attenuation correction, we found that TXTV has less bias as compared to MAP-TR. We also compared images acquired at the HRRT scanner using TXTV to the GE Advance scanner images and found high quantitative correspondence. TXTV has been used to reconstruct more than 4000 HRRT scans at seven different sites with no reports of biases. TXTV-based reconstruction is recommended for human brain scans on the HRRT.
Energy Technology Data Exchange (ETDEWEB)
Larcos, G.; Hutton, B.F.; Farlow, D.C.; Campbell- Rodgers, N.; Gruenewald, S.M.; Lau, Y.H. [Westmead Hospital, Westmead, Sydney, NSW (Australia). Departments of Nuclear Medicine and Ultrasound and Medical Physics
1998-06-01
Full text: The introduction of transmission based attenuation correction (AC) has increased the diagnostic accuracy of Tc-99m MIBI myocardial perfusion SPECT. The aim of this study is to evaluate recent developments, including scatter correction (SC) and resolution recovery (RR). We reviewed 13 patients who underwent Tc-99m MIBI SPECT (two day protocol) and coronary angiography and 4 manufacturer supplied studies assigned a low pretest likelihood of coronary artery disease (CAD). Patients had a mean age of 59 years (range: 41-78). Data were reconstructed using filtered backprojection (FBP; method 1), maximum likelihood (ML) incorporating AC (method 2), ADAC software using sinogram based SC+RR followed by ML with AC (method 3) and ordered subset ML incorporating AC,SC and RR (method 4). Images were reported by two of three blinded experienced physicians using a standard semiquantitative scoring scheme. Fixed or reversible perfusion defects were considered abnormal; CAD was considered present with stenoses > 50%. Patients had normal coronary anatomy (n=9), single (n=4) or two vessel CAD (n=4) (four in each of LAD, RCA and LCX). There were no statistically significant differences for any combination. Normalcy rate = 100% for all methods. Physicians graded 3/17 (methods 2,4) and 1/17 (method 3) images as fair or poor in quality. Thus, AC or AC+SC+RR produce good quality images in most patients; there is potential for improvement in sensitivity over standard FBP with no significant change in normalcy or specificity
Brunner, Stephen; Nett, Brian E; Tolakanahalli, Ranjini; Chen, Guang-Hong
2011-02-21
X-ray scatter is a significant problem in cone-beam computed tomography when thicker objects and larger cone angles are used, as scattered radiation can lead to reduced contrast and CT number inaccuracy. Advances have been made in x-ray computed tomography (CT) by incorporating a high quality prior image into the image reconstruction process. In this paper, we extend this idea to correct scatter-induced shading artifacts in cone-beam CT image-guided radiation therapy. Specifically, this paper presents a new scatter correction algorithm which uses a prior image with low scatter artifacts to reduce shading artifacts in cone-beam CT images acquired under conditions of high scatter. The proposed correction algorithm begins with an empirical hypothesis that the target image can be written as a weighted summation of a series of basis images that are generated by raising the raw cone-beam projection data to different powers, and then, reconstructing using the standard filtered backprojection algorithm. The weight for each basis image is calculated by minimizing the difference between the target image and the prior image. The performance of the scatter correction algorithm is qualitatively and quantitatively evaluated through phantom studies using a Varian 2100 EX System with an on-board imager. Results show that the proposed scatter correction algorithm using a prior image with low scatter artifacts can substantially mitigate scatter-induced shading artifacts in both full-fan and half-fan modes.
Physics Model-Based Scatter Correction in Multi-Source Interior Computed Tomography.
Gong, Hao; Li, Bin; Jia, Xun; Cao, Guohua
2018-02-01
Multi-source interior computed tomography (CT) has a great potential to provide ultra-fast and organ-oriented imaging at low radiation dose. However, X-ray cross scattering from multiple simultaneously activated X-ray imaging chains compromises imaging quality. Previously, we published two hardware-based scatter correction methods for multi-source interior CT. Here, we propose a software-based scatter correction method, with the benefit of no need for hardware modifications. The new method is based on a physics model and an iterative framework. The physics model was derived analytically, and was used to calculate X-ray scattering signals in both forward direction and cross directions in multi-source interior CT. The physics model was integrated to an iterative scatter correction framework to reduce scatter artifacts. The method was applied to phantom data from both Monte Carlo simulations and physical experimentation that were designed to emulate the image acquisition in a multi-source interior CT architecture recently proposed by our team. The proposed scatter correction method reduced scatter artifacts significantly, even with only one iteration. Within a few iterations, the reconstructed images fast converged toward the "scatter-free" reference images. After applying the scatter correction method, the maximum CT number error at the region-of-interests (ROIs) was reduced to 46 HU in numerical phantom dataset and 48 HU in physical phantom dataset respectively, and the contrast-noise-ratio at those ROIs increased by up to 44.3% and up to 19.7%, respectively. The proposed physics model-based iterative scatter correction method could be useful for scatter correction in dual-source or multi-source CT.
Jo, Byung-Du; Lee, Young-Jin; Kim, Dae-Hong; Kim, Hee-Joung
2014-08-01
In conventional digital radiography (DR) using a dual energy subtraction technique, a significant fraction of the detected photons are scattered within the body, making up the scatter component. Scattered radiation can significantly deteriorate image quality in diagnostic X-ray imaging systems. Various methods of scatter correction, including both measurement- and non-measurement-based methods, have been proposed in the past. Both methods can reduce scatter artifacts in images. However, non-measurement-based methods require a homogeneous object and have insufficient scatter component correction. Therefore, we employed a measurement-based method to correct for the scatter component of inhomogeneous objects from dual energy DR (DEDR) images. We performed a simulation study using a Monte Carlo simulation with a primary modulator, which is a measurement-based method for the DEDR system. The primary modulator, which has a checkerboard pattern, was used to modulate the primary radiation. Cylindrical phantoms of variable size were used to quantify the imaging performance. For scatter estimates, we used discrete Fourier transform filtering, e.g., a Gaussian low-high pass filter with a cut-off frequency. The primary modulation method was evaluated using a cylindrical phantom in the DEDR system. The scatter components were accurately removed using a primary modulator. When the results acquired with scatter correction and without scatter correction were compared, the average contrast-to-noise ratio (CNR) with the correction was 1.35 times higher than that obtained without the correction, and the average root mean square error (RMSE) with the correction was 38.00% better than that without the correction. In the subtraction study, the average CNR with the correction was 2.04 (aluminum subtraction) and 1.38 (polymethyl methacrylate (PMMA) subtraction) times higher than that obtained without the correction. The analysis demonstrated the accuracy of the scatter correction and the
TH-A-18C-04: Ultrafast Cone-Beam CT Scatter Correction with GPU-Based Monte Carlo Simulation
Energy Technology Data Exchange (ETDEWEB)
Xu, Y [UT Southwestern Medical Center, Dallas, TX (United States); Southern Medical University, Guangzhou (China); Bai, T [UT Southwestern Medical Center, Dallas, TX (United States); Xi' an Jiaotong University, Xi' an (China); Yan, H; Ouyang, L; Wang, J; Pompos, A; Jiang, S; Jia, X [UT Southwestern Medical Center, Dallas, TX (United States); Zhou, L [Southern Medical University, Guangzhou (China)
2014-06-15
Purpose: Scatter artifacts severely degrade image quality of cone-beam CT (CBCT). We present an ultrafast scatter correction framework by using GPU-based Monte Carlo (MC) simulation and prior patient CT image, aiming at automatically finish the whole process including both scatter correction and reconstructions within 30 seconds. Methods: The method consists of six steps: 1) FDK reconstruction using raw projection data; 2) Rigid Registration of planning CT to the FDK results; 3) MC scatter calculation at sparse view angles using the planning CT; 4) Interpolation of the calculated scatter signals to other angles; 5) Removal of scatter from the raw projections; 6) FDK reconstruction using the scatter-corrected projections. In addition to using GPU to accelerate MC photon simulations, we also use a small number of photons and a down-sampled CT image in simulation to further reduce computation time. A novel denoising algorithm is used to eliminate MC scatter noise caused by low photon numbers. The method is validated on head-and-neck cases with simulated and clinical data. Results: We have studied impacts of photo histories, volume down sampling factors on the accuracy of scatter estimation. The Fourier analysis was conducted to show that scatter images calculated at 31 angles are sufficient to restore those at all angles with <0.1% error. For the simulated case with a resolution of 512×512×100, we simulated 10M photons per angle. The total computation time is 23.77 seconds on a Nvidia GTX Titan GPU. The scatter-induced shading/cupping artifacts are substantially reduced, and the average HU error of a region-of-interest is reduced from 75.9 to 19.0 HU. Similar results were found for a real patient case. Conclusion: A practical ultrafast MC-based CBCT scatter correction scheme is developed. The whole process of scatter correction and reconstruction is accomplished within 30 seconds. This study is supported in part by NIH (1R01CA154747-01), The Core Technology Research
Correction factor for hair analysis by PIXE
International Nuclear Information System (INIS)
Montenegro, E.C.; Baptista, G.B.; Castro Faria, L.V. de; Paschoa, A.S.
1980-01-01
The application of the Particle Induced X-ray Emission (PIXE) technique to analyse quantitatively the elemental composition of hair specimens brings about some difficulties in the interpretation of the data. The present paper proposes a correction factor to account for the effects of the energy loss of the incident particle with penetration depth, and X-ray self-absorption when a particular geometrical distribution of elements in hair is assumed for calculational purposes. The correction factor has been applied to the analysis of hair contents Zn, Cu and Ca as a function of the energy of the incident particle. (orig.)
Correction factor for hair analysis by PIXE
International Nuclear Information System (INIS)
Montenegro, E.C.; Baptista, G.B.; Castro Faria, L.V. de; Paschoa, A.S.
1979-06-01
The application of the Particle Induced X-ray Emission (PIXE) technique to analyse quantitatively the elemental composition of hair specimens brings about some difficulties in the interpretation of the data. The present paper proposes a correction factor to account for the effects of energy loss of the incident particle with penetration depth, and x-ray self-absorption when a particular geometrical distribution of elements in hair is assumed for calculational purposes. The correction factor has been applied to the analysis of hair contents Zn, Cu and Ca as a function of the energy of the incident particle.(Author) [pt
International Nuclear Information System (INIS)
Ruehrnschopf and, Ernst-Peter; Klingenbeck, Klaus
2011-01-01
The main components of scatter correction procedures are scatter estimation and a scatter compensation algorithm. This paper completes a previous paper where a general framework for scatter compensation was presented under the prerequisite that a scatter estimation method is already available. In the current paper, the authors give a systematic review of the variety of scatter estimation approaches. Scatter estimation methods are based on measurements, mathematical-physical models, or combinations of both. For completeness they present an overview of measurement-based methods, but the main topic is the theoretically more demanding models, as analytical, Monte-Carlo, and hybrid models. Further classifications are 3D image-based and 2D projection-based approaches. The authors present a system-theoretic framework, which allows to proceed top-down from a general 3D formulation, by successive approximations, to efficient 2D approaches. A widely useful method is the beam-scatter-kernel superposition approach. Together with the review of standard methods, the authors discuss their limitations and how to take into account the issues of object dependency, spatial variance, deformation of scatter kernels, external and internal absorbers. Open questions for further investigations are indicated. Finally, the authors refer on some special issues and applications, such as bow-tie filter, offset detector, truncated data, and dual-source CT.
International Nuclear Information System (INIS)
Nielson, K.K.; Garcia, S.R.
1976-09-01
Two methods are described for computing multielement x-ray absorption corrections for aerosol samples collected in IPC-1478 and Whatman 41 filters. The first relies on scatter peak intensities and scattering cross sections to estimate the mass of light elements (Z less than 14) in the sample. This mass is used with the measured heavy element (Z greater than or equal to 14) masses to iteratively compute sample absorption corrections. The second method utilizes a linear function of ln(μ) vs ln(E) determined from the scatter peak ratios and estimates sample mass from the scatter peak intensities. Both methods assume a homogeneous depth distribution of aerosol in a fraction of the front of the filters, and the assumption is evaluated with respect to an exponential aerosol depth distribution. Penetration depths for various real, synthethic and liquid aerosols were measured. Aerosol penetration appeared constant over a 1.1 mg/cm 2 range of sample loading for IPC filters, while absorption corrections for Si and S varied by a factor of two over the same loading range. Corrections computed by the two methods were compared with measured absorption corrections and with atomic absorption analyses of the same samples
A software-based x-ray scatter correction method for breast tomosynthesis
Jia Feng, Steve Si; Sechopoulos, Ioannis
2011-01-01
Purpose: To develop a software-based scatter correction method for digital breast tomosynthesis (DBT) imaging and investigate its impact on the image quality of tomosynthesis reconstructions of both phantoms and patients.
Binding and Pauli principle corrections in subthreshold pion-nucleus scattering
International Nuclear Information System (INIS)
Kam, J. de
1981-01-01
In this investigation I develop a three-body model for the single scattering optical potential in which the nucleon binding and the Pauli principle are accounted for. A unitarity pole approximation is used for the nucleon-core interaction. Calculations are presented for the π- 4 He elastic scattering cross sections at energies below the inelastic threshold and for the real part of the π- 4 He scattering length by solving the three-body equations. Off-shell kinematics and the Pauli principle are carefully taken into account. The binding correction and the Pauli principle correction each have an important effect on the differential cross sections and the scattering length. However, large cancellations occur between these two effects. I find an increase in the π- 4 He scattering length by 100%; an increase in the cross sections by 20-30% and shift of the minimum in π - - 4 He scattering to forward angles by 10 0 . (orig.)
International Nuclear Information System (INIS)
Siewerdsen, J.H.; Daly, M.J.; Bakhtiar, B.
2006-01-01
X-ray scatter poses a significant limitation to image quality in cone-beam CT (CBCT), resulting in contrast reduction, image artifacts, and lack of CT number accuracy. We report the performance of a simple scatter correction method in which scatter fluence is estimated directly in each projection from pixel values near the edge of the detector behind the collimator leaves. The algorithm operates on the simple assumption that signal in the collimator shadow is attributable to x-ray scatter, and the 2D scatter fluence is estimated by interpolating between pixel values measured along the top and bottom edges of the detector behind the collimator leaves. The resulting scatter fluence estimate is subtracted from each projection to yield an estimate of the primary-only images for CBCT reconstruction. Performance was investigated in phantom experiments on an experimental CBCT benchtop, and the effect on image quality was demonstrated in patient images (head, abdomen, and pelvis sites) obtained on a preclinical system for CBCT-guided radiation therapy. The algorithm provides significant reduction in scatter artifacts without compromise in contrast-to-noise ratio (CNR). For example, in a head phantom, cupping artifact was essentially eliminated, CT number accuracy was restored to within 3%, and CNR (breast-to-water) was improved by up to 50%. Similarly in a body phantom, cupping artifact was reduced by at least a factor of 2 without loss in CNR. Patient images demonstrate significantly increased uniformity, accuracy, and contrast, with an overall improvement in image quality in all sites investigated. Qualitative evaluation illustrates that soft-tissue structures that are otherwise undetectable are clearly delineated in scatter-corrected reconstructions. Since scatter is estimated directly in each projection, the algorithm is robust with respect to system geometry, patient size and heterogeneity, patient motion, etc. Operating without prior information, analytical modeling
SU-E-I-07: An Improved Technique for Scatter Correction in PET
International Nuclear Information System (INIS)
Lin, S; Wang, Y; Lue, K; Lin, H; Chuang, K
2014-01-01
Purpose: In positron emission tomography (PET), the single scatter simulation (SSS) algorithm is widely used for scatter estimation in clinical scans. However, bias usually occurs at the essential steps of scaling the computed SSS distribution to real scatter amounts by employing the scatter-only projection tail. The bias can be amplified when the scatter-only projection tail is too small, resulting in incorrect scatter correction. To this end, we propose a novel scatter calibration technique to accurately estimate the amount of scatter using pre-determined scatter fraction (SF) function instead of the employment of scatter-only tail information. Methods: As the SF depends on the radioactivity distribution and the attenuating material of the patient, an accurate theoretical relation cannot be devised. Instead, we constructed an empirical transformation function between SFs and average attenuation coefficients based on a serious of phantom studies with different sizes and materials. From the average attenuation coefficient, the predicted SFs were calculated using empirical transformation function. Hence, real scatter amount can be obtained by scaling the SSS distribution with the predicted SFs. The simulation was conducted using the SimSET. The Siemens Biograph™ 6 PET scanner was modeled in this study. The Software for Tomographic Image Reconstruction (STIR) was employed to estimate the scatter and reconstruct images. The EEC phantom was adopted to evaluate the performance of our proposed technique. Results: The scatter-corrected image of our method demonstrated improved image contrast over that of SSS. For our technique and SSS of the reconstructed images, the normalized standard deviation were 0.053 and 0.182, respectively; the root mean squared errors were 11.852 and 13.767, respectively. Conclusion: We have proposed an alternative method to calibrate SSS (C-SSS) to the absolute scatter amounts using SF. This method can avoid the bias caused by the insufficient
SU-E-I-07: An Improved Technique for Scatter Correction in PET
Energy Technology Data Exchange (ETDEWEB)
Lin, S; Wang, Y; Lue, K; Lin, H; Chuang, K [Chuang, National Tsing Hua University, Hsichu, Taiwan (China)
2014-06-01
Purpose: In positron emission tomography (PET), the single scatter simulation (SSS) algorithm is widely used for scatter estimation in clinical scans. However, bias usually occurs at the essential steps of scaling the computed SSS distribution to real scatter amounts by employing the scatter-only projection tail. The bias can be amplified when the scatter-only projection tail is too small, resulting in incorrect scatter correction. To this end, we propose a novel scatter calibration technique to accurately estimate the amount of scatter using pre-determined scatter fraction (SF) function instead of the employment of scatter-only tail information. Methods: As the SF depends on the radioactivity distribution and the attenuating material of the patient, an accurate theoretical relation cannot be devised. Instead, we constructed an empirical transformation function between SFs and average attenuation coefficients based on a serious of phantom studies with different sizes and materials. From the average attenuation coefficient, the predicted SFs were calculated using empirical transformation function. Hence, real scatter amount can be obtained by scaling the SSS distribution with the predicted SFs. The simulation was conducted using the SimSET. The Siemens Biograph™ 6 PET scanner was modeled in this study. The Software for Tomographic Image Reconstruction (STIR) was employed to estimate the scatter and reconstruct images. The EEC phantom was adopted to evaluate the performance of our proposed technique. Results: The scatter-corrected image of our method demonstrated improved image contrast over that of SSS. For our technique and SSS of the reconstructed images, the normalized standard deviation were 0.053 and 0.182, respectively; the root mean squared errors were 11.852 and 13.767, respectively. Conclusion: We have proposed an alternative method to calibrate SSS (C-SSS) to the absolute scatter amounts using SF. This method can avoid the bias caused by the insufficient
Non perturbative method for radiative corrections applied to lepton-proton scattering
International Nuclear Information System (INIS)
Chahine, C.
1979-01-01
We present a new, non perturbative method to effect radiative corrections in lepton (electron or muon)-nucleon scattering, useful for existing or planned experiments. This method relies on a spectral function derived in a previous paper, which takes into account both real soft photons and virtual ones and hence is free from infrared divergence. Hard effects are computed perturbatively and then included in the form of 'hard factors' in the non peturbative soft formulas. Practical computations are effected using the Gauss-Jacobi integration method which reduce the relevant integrals to a rapidly converging sequence. For the simple problem of the radiative quasi-elastic peak, we get an exponentiated form conjectured by Schwinger and found by Yennie, Frautschi and Suura. We compare also our results with the peaking approximation, which we derive independantly, and with the exact one-photon emission formula of Mo and Tsai. Applications of our method to the continuous spectrum include the radiative tail of the Δ 33 resonance in e + p scattering and radiative corrections to the Feynman scale invariant F 2 structure function for the kinematics of two recent high energy muon experiments
Qattan, I. A.
2017-06-01
I present a prediction of the e± elastic scattering cross-section ratio, Re+e-, as determined using a new parametrization of the two-photon exchange (TPE) corrections to electron-proton elastic scattering cross section σR. The extracted ratio is compared to several previous phenomenological extractions, TPE hadronic calculations, and direct measurements from the comparison of electron and positron scattering. The TPE corrections and the ratio Re+e- show a clear change of sign at low Q2, which is necessary to explain the high-Q2 form factors discrepancy while being consistent with the known Q2→0 limit. While my predictions are in generally good agreement with previous extractions, TPE hadronic calculations, and existing world data including the recent two measurements from the CLAS and VEPP-3 Novosibirsk experiments, they are larger than the new OLYMPUS measurements at larger Q2 values.
Scatter correction using a primary modulator on a clinical angiography C-arm CT system.
Bier, Bastian; Berger, Martin; Maier, Andreas; Kachelrieß, Marc; Ritschl, Ludwig; Müller, Kerstin; Choi, Jang-Hwan; Fahrig, Rebecca
2017-09-01
Cone beam computed tomography (CBCT) suffers from a large amount of scatter, resulting in severe scatter artifacts in the reconstructions. Recently, a new scatter correction approach, called improved primary modulator scatter estimation (iPMSE), was introduced. That approach utilizes a primary modulator that is inserted between the X-ray source and the object. This modulation enables estimation of the scatter in the projection domain by optimizing an objective function with respect to the scatter estimate. Up to now the approach has not been implemented on a clinical angiography C-arm CT system. In our work, the iPMSE method is transferred to a clinical C-arm CBCT. Additional processing steps are added in order to compensate for the C-arm scanner motion and the automatic X-ray tube current modulation. These challenges were overcome by establishing a reference modulator database and a block-matching algorithm. Experiments with phantom and experimental in vivo data were performed to evaluate the method. We show that scatter correction using primary modulation is possible on a clinical C-arm CBCT. Scatter artifacts in the reconstructions are reduced with the newly extended method. Compared to a scan with a narrow collimation, our approach showed superior results with an improvement of the contrast and the contrast-to-noise ratio for the phantom experiments. In vivo data are evaluated by comparing the results with a scan with a narrow collimation and with a constant scatter correction approach. Scatter correction using primary modulation is possible on a clinical CBCT by compensating for the scanner motion and the tube current modulation. Scatter artifacts could be reduced in the reconstructions of phantom scans and in experimental in vivo data. © 2017 American Association of Physicists in Medicine.
Ultrafast cone-beam CT scatter correction with GPU-based Monte Carlo simulation
Directory of Open Access Journals (Sweden)
Yuan Xu
2014-03-01
Full Text Available Purpose: Scatter artifacts severely degrade image quality of cone-beam CT (CBCT. We present an ultrafast scatter correction framework by using GPU-based Monte Carlo (MC simulation and prior patient CT image, aiming at automatically finish the whole process including both scatter correction and reconstruction within 30 seconds.Methods: The method consists of six steps: 1 FDK reconstruction using raw projection data; 2 Rigid Registration of planning CT to the FDK results; 3 MC scatter calculation at sparse view angles using the planning CT; 4 Interpolation of the calculated scatter signals to other angles; 5 Removal of scatter from the raw projections; 6 FDK reconstruction using the scatter-corrected projections. In addition to using GPU to accelerate MC photon simulations, we also use a small number of photons and a down-sampled CT image in simulation to further reduce computation time. A novel denoising algorithm is used to eliminate MC noise from the simulated scatter images caused by low photon numbers. The method is validated on one simulated head-and-neck case with 364 projection angles.Results: We have examined variation of the scatter signal among projection angles using Fourier analysis. It is found that scatter images at 31 angles are sufficient to restore those at all angles with < 0.1% error. For the simulated patient case with a resolution of 512 × 512 × 100, we simulated 5 × 106 photons per angle. The total computation time is 20.52 seconds on a Nvidia GTX Titan GPU, and the time at each step is 2.53, 0.64, 14.78, 0.13, 0.19, and 2.25 seconds, respectively. The scatter-induced shading/cupping artifacts are substantially reduced, and the average HU error of a region-of-interest is reduced from 75.9 to 19.0 HU.Conclusion: A practical ultrafast MC-based CBCT scatter correction scheme is developed. It accomplished the whole procedure of scatter correction and reconstruction within 30 seconds.----------------------------Cite this
Non-eikonal corrections for the scattering of spin-one particles
Energy Technology Data Exchange (ETDEWEB)
Gaber, M.W.; Wilkin, C. [Department of Physics and Astronomy, University College London, WC1E 6BT, London (United Kingdom); Al-Khalili, J.S. [Department of Physics, University of Surrey, GU2 7XH, Guildford, Surrey (United Kingdom)
2004-08-01
The Wallace Fourier-Bessel expansion of the scattering amplitude is generalised to the case of the scattering of a spin-one particle from a potential with a single tensor coupling as well as central and spin-orbit terms. A generating function for the eikonal-phase (quantum) corrections is evaluated in closed form. For medium-energy deuteron-nucleus scattering, the first-order correction is dominant and is shown to be significant in the interpretation of analysing power measurements. This conclusion is supported by a numerical comparison of the eikonal observables, evaluated with and without corrections, with those obtained from a numerical resolution of the Schroedinger equation for d-{sup 58}Ni scattering at incident deuteron energies of 400 and 700 MeV. (orig.)
International Nuclear Information System (INIS)
Gomez Facenda, A.; Castillo Lopez, J. P.; Torres Aroche, L. A.; Coca Perez, M. A.
2013-01-01
Activity quantification in nuclear medicine imaging is highly desirable, particularly for dosimetry and biodistribution studies of radiopharmaceuticals. Quantitative 111 In imaging is increasingly important with the current interest in therapy using 90 Y-radiolabeled compounds. Photons scattered in the patient are one of the major problems in quantification, which leads to degradation of image quality. The aim of this work was to assess the configuration of energy windows and the best weight factor for the scatter correction in 111 In images. All images were obtained using the Monte Carlo simulation code, Simind, configured to emulate the gamma camera Nucline SPIRIT DH-V. Simulations were validated by a positive agreement between experimental and simulated line-spread functions (LSF) of 99 mTc. It was examined the sensitivity, the scatter-to-total ratio, the contrast and the spatial resolution for scatter-compensated images obtained from six different multi-windows scatter corrections. Taking into consideration the results, the best energy-window setting was two 20% windows centered at 171 and 245keV, together with a 10% scatter window located between the photo peaks at 209keV. (Author)
Radiative corrections to high-energy neutrino scattering
International Nuclear Information System (INIS)
Rujula, A. de; Petronzio, R.; Savoy-Navarro, A.
1979-01-01
Motivated by precise neutrino experiments, the electromagnetic radiative corrections to the data are reconsidered. The usefulness is investigated and the simplicity demonstrated of the 'leading log' approximation: the calculation to order α ln (Q/μ), α ln (Q/msub(q)). Here Q is an energy scale of the overall process, μ is the lepton mass and msub(q) is a hadronic mass, the effective quark mass in a parton model. The leading log radiative corrections to dsigma/dy distributions and to suitably interpreted dsigma/dx distributions are quark-mass independent. The authors improve upon the conventional leading log approximation and compute explicitly the largest terms that lie beyond the leading log level. In practice this means that the model-independent formulae, though approximate, are likely to be excellent estimates everywhere except at low energy or very large y. It is pointed out that radiative corrections to measurements of deviations from the Callan-Gross relation and to measurements of the 'sea' constituency of nucleons are gigantic. The QCD inspired study of deviations from scaling is of particular interest. The authors compute, beyond the leading log level, the radiative corrections of the QCD predictions. (Auth.)
International Nuclear Information System (INIS)
Ljungberg, M.
1990-05-01
Quantitative scintigrafic images, obtained by NaI(Tl) scintillation cameras, are limited by photon attenuation and contribution from scattered photons. A Monte Carlo program was developed in order to evaluate these effects. Simple source-phantom geometries and more complex nonhomogeneous cases can be simulated. Comparisons with experimental data for both homogeneous and nonhomogeneous regions and with published results have shown good agreement. The usefulness for simulation of parameters in scintillation camera systems, stationary as well as in SPECT systems, has also been demonstrated. An attenuation correction method based on density maps and build-up functions has been developed. The maps were obtained from a transmission measurement using an external 57 Co flood source and the build-up was simulated by the Monte Carlo code. Two scatter correction methods, the dual-window method and the convolution-subtraction method, have been compared using the Monte Carlo method. The aim was to compare the estimated scatter with the true scatter in the photo-peak window. It was concluded that accurate depth-dependent scatter functions are essential for a proper scatter correction. A new scatter and attenuation correction method has been developed based on scatter line-spread functions (SLSF) obtained for different depths and lateral positions in the phantom. An emission image is used to determine the source location in order to estimate the scatter in the photo-peak window. Simulation studies of a clinically realistic source in different positions in cylindrical water phantoms were made for three photon energies. The SLSF-correction method was also evaluated by simulation studies for 1. a myocardial source, 2. uniform source in the lungs and 3. a tumour located in the lungs in a realistic, nonhomogeneous computer phantom. The results showed that quantitative images could be obtained in nonhomogeneous regions. (67 refs.)
DEFF Research Database (Denmark)
de Nijs, Robin; Lagerburg, Vera; Klausen, Thomas L
2014-01-01
and the activity, which depends on the collimator type, the utilized energy windows and the applied scatter correction techniques. In this study, energy window subtraction-based scatter correction methods are compared experimentally and quantitatively. MATERIALS AND METHODS: (177)Lu SPECT images of a phantom...... technique, the measured ratio was close to the real ratio, and the differences between spheres were small. CONCLUSION: For quantitative (177)Lu imaging MEGP collimators are advised. Both energy peaks can be utilized when the ESSE correction technique is applied. The difference between the calculated...
Evaluation of a method for correction of scatter radiation in thorax cone beam CT
International Nuclear Information System (INIS)
Rinkel, J.; Dinten, J.M.; Esteve, F.
2004-01-01
Purpose: Cone beam CT (CBCT) enables three-dimensional imaging with isotropic resolution. X-ray scatter estimation is a big challenge for quantitative CBCT imaging of thorax: scatter level is significantly higher on cone beam systems compared to collimated fan beam systems. The effects of this scattered radiation are cupping artefacts, streaks, and quantification inaccuracies. The beam stops conventional scatter estimation approach can be used for CBCT but leads to a significant increase in terms of dose and acquisition time. At CEA-LETI has been developed an original scatter management process without supplementary acquisition. Methods and Materials: This Analytical Plus Indexing-based method (API) of scatter correction in CBCT is based on scatter calibration through offline acquisitions with beam stops on lucite plates, combined to an analytical transformation issued from physical equations. This approach has been applied with success in bone densitometry and mammography. To evaluate this method in CBCT, acquisitions from a thorax phantom with and without beam stops have been performed. To compare different scatter correction approaches, Feldkamp algorithm has been applied on rough data corrected from scatter by API and by beam stops approaches. Results: The API method provides results in good agreement with the beam stops array approach, suppressing cupping artefact. Otherwise influence of the scatter correction method on the noise in the reconstructed images has been evaluated. Conclusion: The results indicate that the API method is effective for quantitative CBCT imaging of thorax. Compared to a beam stops array method it needs a lower x-ray dose and shortens acquisition time. (authors)
Three-loop corrections to the soft anomalous dimension in multileg scattering
Almelid, Øyvind; Gardi, Einan
2016-01-01
We present the three-loop result for the soft anomalous dimension governing long-distance singularities of multi-leg gauge-theory scattering amplitudes of massless partons. We compute all contributing webs involving semi-infinite Wilson lines at three loops and obtain the complete three-loop correction to the dipole formula. We find that non-dipole corrections appear already for three coloured partons, where the correction is a constant without kinematic dependence. Kinematic dependence appears only through conformally-invariant cross ratios for four coloured partons or more, and the result can be expressed in terms of single-valued harmonic polylogarithms of weight five. While the non-dipole three-loop term does not vanish in two-particle collinear limits, its contribution to the splitting amplitude anomalous dimension reduces to a constant, and it only depends on the colour charges of the collinear pair, thereby preserving strict collinear factorization properties. Finally we verify that our result is consi...
A software-based x-ray scatter correction method for breast tomosynthesis
International Nuclear Information System (INIS)
Jia Feng, Steve Si; Sechopoulos, Ioannis
2011-01-01
Purpose: To develop a software-based scatter correction method for digital breast tomosynthesis (DBT) imaging and investigate its impact on the image quality of tomosynthesis reconstructions of both phantoms and patients. Methods: A Monte Carlo (MC) simulation of x-ray scatter, with geometry matching that of the cranio-caudal (CC) view of a DBT clinical prototype, was developed using the Geant4 toolkit and used to generate maps of the scatter-to-primary ratio (SPR) of a number of homogeneous standard-shaped breasts of varying sizes. Dimension-matched SPR maps were then deformed and registered to DBT acquisition projections, allowing for the estimation of the primary x-ray signal acquired by the imaging system. Noise filtering of the estimated projections was then performed to reduce the impact of the quantum noise of the x-ray scatter. Three dimensional (3D) reconstruction was then performed using the maximum likelihood-expectation maximization (MLEM) method. This process was tested on acquisitions of a heterogeneous 50/50 adipose/glandular tomosynthesis phantom with embedded masses, fibers, and microcalcifications and on acquisitions of patients. The image quality of the reconstructions of the scatter-corrected and uncorrected projections was analyzed by studying the signal-difference-to-noise ratio (SDNR), the integral of the signal in each mass lesion (integrated mass signal, IMS), and the modulation transfer function (MTF). Results: The reconstructions of the scatter-corrected projections demonstrated superior image quality. The SDNR of masses embedded in a 5 cm thick tomosynthesis phantom improved 60%-66%, while the SDNR of the smallest mass in an 8 cm thick phantom improved by 59% (p < 0.01). The IMS of the masses in the 5 cm thick phantom also improved by 15%-29%, while the IMS of the masses in the 8 cm thick phantom improved by 26%-62% (p < 0.01). Some embedded microcalcifications in the tomosynthesis phantoms were visible only in the scatter-corrected
Scatter measurement and correction method for cone-beam CT based on single grating scan
Huang, Kuidong; Shi, Wenlong; Wang, Xinyu; Dong, Yin; Chang, Taoqi; Zhang, Hua; Zhang, Dinghua
2017-06-01
In cone-beam computed tomography (CBCT) systems based on flat-panel detector imaging, the presence of scatter significantly reduces the quality of slices. Based on the concept of collimation, this paper presents a scatter measurement and correction method based on single grating scan. First, according to the characteristics of CBCT imaging, the scan method using single grating and the design requirements of the grating are analyzed and figured out. Second, by analyzing the composition of object projection images and object-and-grating projection images, the processing method for the scatter image at single projection angle is proposed. In addition, to avoid additional scan, this paper proposes an angle interpolation method of scatter images to reduce scan cost. Finally, the experimental results show that the scatter images obtained by this method are accurate and reliable, and the effect of scatter correction is obvious. When the additional object-and-grating projection images are collected and interpolated at intervals of 30 deg, the scatter correction error of slices can still be controlled within 3%.
International Nuclear Information System (INIS)
Mukai, T.; Torizuka, K.; Douglass, K.H.; Wagner, H.N.
1985-01-01
Quantitative assessment of tracer distribution with single photon emission computed tomography (SPECT) is difficult because of attenuation and scattering of gamma rays within the object. A method considering the source geometry was developed, and effects of attenuation and scatter on SPECT quantitation were studied using phantoms with non-uniform attenuation. The distribution of attenuation coefficients (μ) within the source were obtained by transmission CT. The attenuation correction was performed by an iterative reprojection technique. The scatter correction was done by convolution of the attenuation corrected image and an appropriate filter made by line source studies. The filter characteristics depended on μ and SPEC measurement at each pixel. The SPECT obtained by this method showed the most reasonable results than the images reconstructed by other methods. The scatter correction could compensate completely for a 28% scatter components from a long line source, and a 61% component for thick and extended source. Consideration of source geometries was necessary for effective corrections. The present method is expected to be valuable for the quantitative assessment of regional tracer activity
Inverse scattering and refraction corrected reflection for breast cancer imaging
Wiskin, J.; Borup, D.; Johnson, S.; Berggren, M.; Robinson, D.; Smith, J.; Chen, J.; Parisky, Y.; Klock, John
2010-03-01
Reflection ultrasound (US) has been utilized as an adjunct imaging modality for over 30 years. TechniScan, Inc. has developed unique, transmission and concomitant reflection algorithms which are used to reconstruct images from data gathered during a tomographic breast scanning process called Warm Bath Ultrasound (WBU™). The transmission algorithm yields high resolution, 3D, attenuation and speed of sound (SOS) images. The reflection algorithm is based on canonical ray tracing utilizing refraction correction via the SOS and attenuation reconstructions. The refraction correction reflection algorithm allows 360 degree compounding resulting in the reflection image. The requisite data are collected when scanning the entire breast in a 33° C water bath, on average in 8 minutes. This presentation explains how the data are collected and processed by the 3D transmission and reflection imaging mode algorithms. The processing is carried out using two NVIDIA® Tesla™ GPU processors, accessing data on a 4-TeraByte RAID. The WBU™ images are displayed in a DICOM viewer that allows registration of all three modalities. Several representative cases are presented to demonstrate potential diagnostic capability including: a cyst, fibroadenoma, and a carcinoma. WBU™ images (SOS, attenuation, and reflection modalities) are shown along with their respective mammograms and standard ultrasound images. In addition, anatomical studies are shown comparing WBU™ images and MRI images of a cadaver breast. This innovative technology is designed to provide additional tools in the armamentarium for diagnosis of breast disease.
Development of the Heated Length Correction Factor
International Nuclear Information System (INIS)
Park, Ho-Young; Kim, Kang-Hoon; Nahm, Kee-Yil; Jung, Yil-Sup; Park, Eung-Jun
2008-01-01
The Critical Heat Flux (CHF) on a nuclear fuel is defined by the function of flow channel geometry and flow condition. According to the selection of the explanatory variable, there are three hypotheses to explain CHF at uniformly heated vertical rod (inlet condition hypothesis, exit condition hypothesis, local condition hypothesis). For inlet condition hypothesis, CHF is characterized by function of system pressure, rod diameter, rod length, mass flow and inlet subcooling. For exit condition hypothesis, exit quality substitutes for inlet subcooling. Generally the heated length effect on CHF in exit condition hypothesis is smaller than that of other variables. Heated length is usually excluded in local condition hypothesis to describe the CHF with only local fluid conditions. Most of commercial plants currently use the empirical CHF correlation based on local condition hypothesis. Empirical CHF correlation is developed by the method of fitting the selected sensitive local variables to CHF test data using the multiple non-linear regression. Because this kind of method can not explain physical meaning, it is difficult to reflect the proper effect of complex geometry. So the recent CHF correlation development strategy of nuclear fuel vendor is making the basic CHF correlation which consists of basic flow variables (local fluid conditions) at first, and then the geometrical correction factors are compensated additionally. Because the functional forms of correction factors are determined from the independent test data which represent the corresponding geometry separately, it can be applied to other CHF correlation directly only with minor coefficient modification
Effects of scatter and attenuation corrections on phantom and clinical brain SPECT
International Nuclear Information System (INIS)
Prando, S.; Robilotta, C.C.R.; Oliveira, M.A.; Alves, T.C.; Busatto Filho, G.
2002-01-01
Aim: The present work evaluated the effects of combinations of scatter and attenuation corrections on the analysis of brain SPECT. Materials and Methods: We studied images of the 3D Hoffman brain phantom and from a group of 20 depressive patients with confirmed cardiac insufficiency (CI) and 14 matched healthy controls (HC). Data were acquired with a Sophy-DST/SMV-GE dual-head camera after venous injection of 1110MBq 99m Tc-HMPAO. Two energy windows, 15% on 140keV and 30% centered on 108keV of the Compton distribution, were used to obtain corresponding sets of 128x128x128 projections. Tomograms were reconstructed using OSEM (2 iterations, 8 sub-sets) and Metz filter (order 8, 4 pixels FWHM psf) and FBP with Butterworth filter (order 10, frequency 0.7 Nyquist). Ten combinations of Jaszczak correction (factors 0.3, 0.4 and 0.5) and the 1st order Chang correction (u=0.12cm -1 and 0.159cm -1 ) were applied on the phantom data. In all the phantom images, contrast and signal-noise ratio between 3 ROIs (ventricle, occipital and thalamus) and cerebellum, as well as the ratio between activities in gray and white matters, were calculated and compared with the expected values. The patients images were corrected with k=0.5 and u=0.159cm -1 and reconstructed with OSEM and Metz filter. The images were inspected visually and blood flow comparisons between the CI and the HC groups were performed using Statistical Parametric Mapping (SPM). Results: The best results in the analysis of the contrast and activities ratio were obtained with k=0.5 and u=0.159cm -1 . The results of the activities ratio obtained with OSEM e Metz filter are similar to those published by Laere et al.[J.Nucl.Med 2000;41:2051-2062]. The method of correction using effective attenuation coefficient produced results visually acceptable, but inadequate for the quantitative evaluation. The results of signal-noise ratio are better with OSEM than FBP reconstruction method. The corrections in the CI patients studies
Investigation of Compton scattering correction methods in cardiac SPECT by Monte Carlo simulations
International Nuclear Information System (INIS)
Silva, A.M. Marques da; Furlan, A.M.; Robilotta, C.C.
2001-01-01
The goal of this work was the use of Monte Carlo simulations to investigate the effects of two scattering correction methods: dual energy window (DEW) and dual photopeak window (DPW), in quantitative cardiac SPECT reconstruction. MCAT torso-cardiac phantom, with 99m Tc and non-uniform attenuation map was simulated. Two different photopeak windows were evaluated in DEW method: 15% and 20%. Two 10% wide subwindows centered symmetrically within the photopeak were used in DPW method. Iterative ML-EM reconstruction with modified projector-backprojector for attenuation correction was applied. Results indicated that the choice of the scattering and photopeak windows determines the correction accuracy. For the 15% window, fitted scatter fraction gives better results than k = 0.5. For the 20% window, DPW is the best method, but it requires parameters estimation using Monte Carlo simulations. (author)
Corrections to the leading eikonal amplitude for high-energy scattering and quasipotential approach
International Nuclear Information System (INIS)
Nguyen Suan Hani; Nguyen Duy Hung
2003-12-01
Asymptotic behaviour of the scattering amplitude for two scalar particle at high energy and fixed momentum transfers is reconsidered in quantum field theory. In the framework of the quasipotential approach and the modified perturbation theory a systematic scheme of finding the leading eikonal scattering amplitudes and its corrections is developed and constructed. The connection between the solutions obtained by quasipotential and functional approaches is also discussed. (author)
Multiple scattering corrections to the Beer-Lambert law. 1: Open detector.
Tam, W G; Zardecki, A
1982-07-01
Multiple scattering corrections to the Beer-Lambert law are analyzed by means of a rigorous small-angle solution to the radiative transfer equation. Transmission functions for predicting the received radiant power-a directly measured quantity in contrast to the spectral radiance in the Beer-Lambert law-are derived. Numerical algorithms and results relating to the multiple scattering effects for laser propagation in fog, cloud, and rain are presented.
Energy Technology Data Exchange (ETDEWEB)
Shi, L; Zhu, L [Georgia Institute of Technology, Atlanta, GA (Georgia); Vedantham, S; Karellas, A [University of Massachusetts Medical School, Worcester, MA (United States)
2016-06-15
Purpose: The image quality of dedicated cone-beam breast CT (CBBCT) is fundamentally limited by substantial x-ray scatter contamination, resulting in cupping artifacts and contrast-loss in reconstructed images. Such effects obscure the visibility of soft-tissue lesions and calcifications, which hinders breast cancer detection and diagnosis. In this work, we propose to suppress x-ray scatter in CBBCT images using a deterministic forward projection model. Method: We first use the 1st-pass FDK-reconstructed CBBCT images to segment fibroglandular and adipose tissue. Attenuation coefficients are assigned to the two tissues based on the x-ray spectrum used for imaging acquisition, and is forward projected to simulate scatter-free primary projections. We estimate the scatter by subtracting the simulated primary projection from the measured projection, and then the resultant scatter map is further refined by a Fourier-domain fitting algorithm after discarding untrusted scatter information. The final scatter estimate is subtracted from the measured projection for effective scatter correction. In our implementation, the proposed scatter correction takes 0.5 seconds for each projection. The method was evaluated using the overall image spatial non-uniformity (SNU) metric and the contrast-to-noise ratio (CNR) with 5 clinical datasets of BI-RADS 4/5 subjects. Results: For the 5 clinical datasets, our method reduced the SNU from 7.79% to 1.68% in coronal view and from 6.71% to 3.20% in sagittal view. The average CNR is improved by a factor of 1.38 in coronal view and 1.26 in sagittal view. Conclusion: The proposed scatter correction approach requires no additional scans or prior images and uses a deterministic model for efficient calculation. Evaluation with clinical datasets demonstrates the feasibility and stability of the method. These features are attractive for clinical CBBCT and make our method distinct from other approaches. Supported partly by NIH R21EB019597, R21CA134128
Atomic form factors, incoherent scattering functions, and photon scattering cross sections
International Nuclear Information System (INIS)
Hubbell, J.H.; Veigele, W.J.; Briggs, E.A.; Brown, R.T.; Cromer, D.T.; Howerton, R.J.
1975-01-01
Tabulations are presented of the atomic form factor, F (α,Z), and the incoherent scattering function, S (x,Z), for values of x (=sin theta/2)/lambda) from 0.005 A -1 to 10 9 A -1 , for all elements A=1 to 100. These tables are constructed from available state-of-the-art theoretical data, including the Pirenne formulas for Z=1, configuration-into action results by Brown using Brown-Fontana and Weiss correlated wavefunctions for Z=2 to 6 non-relativistic Hartree-Fock results by Cromer for Z=7 to 100 and a relativistic K-shell analytic expression for F (x,Z) by Bethe Levinger for x>10 A -1 for all elements Z=2 to 100. These tabulated values are graphically compared with available photon scattering angular distribution measurements. Tables of coherent (Rayleigh) and incoherent (Compton) total scattering cross sections obtained by nummerical integration over combinations of F 2 (x,Z) with the Thomson formula and S (x,Z) with the Klum-Nishina Formual, respectively, are presented for all elements Z=1 to 100, for photon energies 100 eV (lambda=124 A) to 100 MeV (0.000124 A). The incoherent scattering cross sections also include the radiative and double-Compton corrections as given by Mork. Similar tables are presented for the special cases of terminally-bonded hydrogen and for the H 2 molecule, interpolated and extrapolated from values calculated by Stewart et al., and by Bentley and Stewart using Kolos-Roothaan wavefunctions
The modular small-angle X-ray scattering data correction sequence.
Pauw, B R; Smith, A J; Snow, T; Terrill, N J; Thünemann, A F
2017-12-01
Data correction is probably the least favourite activity amongst users experimenting with small-angle X-ray scattering: if it is not done sufficiently well, this may become evident only during the data analysis stage, necessitating the repetition of the data corrections from scratch. A recommended comprehensive sequence of elementary data correction steps is presented here to alleviate the difficulties associated with data correction, both in the laboratory and at the synchrotron. When applied in the proposed order to the raw signals, the resulting absolute scattering cross section will provide a high degree of accuracy for a very wide range of samples, with its values accompanied by uncertainty estimates. The method can be applied without modification to any pinhole-collimated instruments with photon-counting direct-detection area detectors.
Magnetic corrections to π -π scattering lengths in the linear sigma model
Loewe, M.; Monje, L.; Zamora, R.
2018-03-01
In this article, we consider the magnetic corrections to π -π scattering lengths in the frame of the linear sigma model. For this, we consider all the one-loop corrections in the s , t , and u channels, associated to the insertion of a Schwinger propagator for charged pions, working in the region of small values of the magnetic field. Our calculation relies on an appropriate expansion for the propagator. It turns out that the leading scattering length, l =0 in the S channel, increases for an increasing value of the magnetic field, in the isospin I =2 case, whereas the opposite effect is found for the I =0 case. The isospin symmetry is valid because the insertion of the magnetic field occurs through the absolute value of the electric charges. The channel I =1 does not receive any corrections. These results, for the channels I =0 and I =2 , are opposite with respect to the thermal corrections found previously in the literature.
In-medium effects in K+ scattering versus Glauber model with noneikonal corrections
International Nuclear Information System (INIS)
Eliseev, S.M.; Rihan, T.H.
1996-01-01
The discrepancy between the experimental and the theoretical ratio R of the total cross sections, R=σ(K + - 12 C)/6σ(K + - d), at momenta up to 800 MeV/c is discussed in the framework of the Glauber multiple scattering approach. It is shown that various corrections such as adopting relativistic K + -N amplitudes as well as noneikonal corrections seem to fail in reproducing the experimental data especially at higher momenta. 17 refs., 1 fig
Parity Violation in Elastic Electron-Proton Scattering and the Proton's Strange Magnetic Form Factor
International Nuclear Information System (INIS)
Spayde, D. T.; Averett, T.; Barkhuff, D.; Beck, D. H.; Beise, E. J.; Benson, C.; Breuer, H.; Carr, R.; Covrig, S.; DelCorso, J.
2000-01-01
We report a new measurement of the parity-violating asymmetry in elastic electron scattering from the proton at backward scattering angles. This asymmetry is sensitive to the strange magnetic form factor of the proton as well as electroweak axial radiative corrections. The new measurement of A=-4.92±0.61±0.73 ppm provides a significant constraint on these quantities. The implications for the strange magnetic form factor are discussed in the context of theoretical estimates for the axial corrections. (c) 2000 The American Physical Society
International Nuclear Information System (INIS)
Guerin, Bastien
2010-01-01
We developed and validated a fast Monte Carlo simulation of PET acquisitions based on the SimSET program modeling accurately the propagation of gamma photons in the patient as well as the block-based PET detector. Comparison of our simulation with another well validated code, GATE, and measurements on two GE Discovery ST PET scanners showed that it models accurately energy spectra (errors smaller than 4.6%), the spatial resolution of block-based PET scanners (6.1%), scatter fraction (3.5%), sensitivity (2.3%) and count rates (12.7%). Next, we developed a novel scatter correction incorporating the energy and position of photons detected in list-mode. Our approach is based on the reformulation of the list-mode likelihood function containing the energy distribution of detected coincidences in addition to their spatial distribution, yielding an EM reconstruction algorithm containing spatial and energy dependent correction terms. We also proposed using the energy in addition to the position of gamma photons in the normalization of the scatter sinogram. Finally, we developed a method for estimating primary and scatter photons energy spectra from total spectra detected in different sectors of the PET scanner. We evaluated the accuracy and precision of our new spatio-spectral scatter correction and that of the standard spatial correction using realistic Monte Carlo simulations. These results showed that incorporating the energy in the scatter correction reduces bias in the estimation of the absolute activity level by ∼ 60% in the cold regions of the largest patients and yields quantification errors less than 13% in all regions. (author)
Evaluation of the ICS and DEW scatter correction methods for low statistical content scans in 3D PET
International Nuclear Information System (INIS)
Sossi, V.; Oakes, T.R.; Ruth, T.J.
1996-01-01
The performance of the Integral Convolution and the Dual Energy Window scatter correction methods in 3D PET has been evaluated over a wide range of statistical content of acquired data (1M to 400M events) The order in which scatter correction and detector normalization should be applied has also been investigated. Phantom and human neuroreceptor studies were used with the following figures of merit: axial and radial uniformity, sinogram and image noise, contrast accuracy and contrast accuracy uniformity. Both scatter correction methods perform reliably in the range of number of events examined. Normalization applied after scatter correction yields better radial uniformity and fewer image artifacts
Evaluation of attenuation and scatter correction requirements in small animal PET and SPECT imaging
Konik, Arda Bekir
Positron emission tomography (PET) and single photon emission tomography (SPECT) are two nuclear emission-imaging modalities that rely on the detection of high-energy photons emitted from radiotracers administered to the subject. The majority of these photons are attenuated (absorbed or scattered) in the body, resulting in count losses or deviations from true detection, which in turn degrades the accuracy of images. In clinical emission tomography, sophisticated correction methods are often required employing additional x-ray CT or radionuclide transmission scans. Having proven their potential in both clinical and research areas, both PET and SPECT are being adapted for small animal imaging. However, despite the growing interest in small animal emission tomography, little scientific information exists about the accuracy of these correction methods on smaller size objects, and what level of correction is required. The purpose of this work is to determine the role of attenuation and scatter corrections as a function of object size through simulations. The simulations were performed using Interactive Data Language (IDL) and a Monte Carlo based package, Geant4 application for emission tomography (GATE). In IDL simulations, PET and SPECT data acquisition were modeled in the presence of attenuation. A mathematical emission and attenuation phantom approximating a thorax slice and slices from real PET/CT data were scaled to 5 different sizes (i.e., human, dog, rabbit, rat and mouse). The simulated emission data collected from these objects were reconstructed. The reconstructed images, with and without attenuation correction, were compared to the ideal (i.e., non-attenuated) reconstruction. Next, using GATE, scatter fraction values (the ratio of the scatter counts to the total counts) of PET and SPECT scanners were measured for various sizes of NEMA (cylindrical phantoms representing small animals and human), MOBY (realistic mouse/rat model) and XCAT (realistic human model
QED corrections in deep-inelastic scattering from tensor polarized deuteron target
Gakh, G I
2001-01-01
The QED correction in the deep inelastic scattering from the polarized tensor of the deuteron target is considered. The calculations are based on the covariant parametrization of the deuteron quadrupole polarization tensor. The Drell-Yan representations in the electrodynamics are used for describing the radiation real and virtual particles
Coulomb correction to the screening angle of the Moliere multiple scattering theory
International Nuclear Information System (INIS)
Kuraev, E.A.; Voskresenskaya, O.O.; Tarasov, A.V.
2012-01-01
Coulomb correction to the screening angular parameter of the Moliere multiple scattering theory is found. Numerical calculations are presented in the range of nuclear charge 4 ≤ Z ≤ 82. Comparison with the Moliere result for the screening angle reveals up to 30% deviation from it for sufficiently heavy elements of the target material
Study of radiative corrections with application to the electron-neutrino scattering
International Nuclear Information System (INIS)
Oliveira, L.C.S. de.
1977-01-01
The radiative correction method is studied which appears in Quantum Field Theory, for some weak interaction processes. e.g., Beta decay and muon decay. Such a method is then applied to calculate transition probability for the electron-neutrino scattering using the U-A theory as a base. The calculations of infrared and ultraviolet divergences are also discussed. (L.C.) [pt
A library least-squares approach for scatter correction in gamma-ray tomography
International Nuclear Information System (INIS)
Meric, Ilker; Anton Johansen, Geir; Valgueiro Malta Moreira, Icaro
2015-01-01
Scattered radiation is known to lead to distortion in reconstructed images in Computed Tomography (CT). The effects of scattered radiation are especially more pronounced in non-scanning, multiple source systems which are preferred for flow imaging where the instantaneous density distribution of the flow components is of interest. In this work, a new method based on a library least-squares (LLS) approach is proposed as a means of estimating the scatter contribution and correcting for this. The validity of the proposed method is tested using the 85-channel industrial gamma-ray tomograph previously developed at the University of Bergen (UoB). The results presented here confirm that the LLS approach can effectively estimate the amounts of transmission and scatter components in any given detector in the UoB gamma-ray tomography system. - Highlights: • A LLS approach is proposed for scatter correction in gamma-ray tomography. • The validity of the LLS approach is tested through experiments. • Gain shift and pulse pile-up affect the accuracy of the LLS approach. • The LLS approach successfully estimates scatter profiles
International Nuclear Information System (INIS)
Kiselev, M.A.; Aksenov, V.L.; Lombardo, D.; Kisselev, A.M.; Lesieur, P.
2003-01-01
Small-angle X-ray scattering (SAXS) experiments have been performed on dimyristoylphosphatidylcholine (DMPC) unilamellar vesicles in 40% aqueous sucrose solution. Model of separated form factors was applied for the evaluation of SAXS curves from large unilamellar vesicles. For the first time vesicle structure factor, polydispersity, average radius and membrane thickness were calculated simultaneously from the SAXS curves at T=30 deg C for DMPC concentrations in the range from 15 to 75 mM (1-5% w/w). Structure factor correction to the scattering curve was shown to be negligibly small for the lipid concentration of 15 mM (1% w/w). It was proved to be necessary to introduce structure factor correction to the scattering curves for lipid concentrations ≥ 30 mM (2% w/w)
Kiselev, M A; Kisselev, A M; Lesieur, P; Aksenov, V L
2003-01-01
Small-angle X-ray scattering (SAXS) experiments have been performed on dimyristoylphosphatidylcholine (DMPC) unilamellar vesicles in 40 % aqueous sucrose solution. Model of separated form factors was applied for the evaluation of SAXS curves from large unilamellar vesicles. For the first time vesicle structure factor, polydispersity, average radius and membrane thickness were calculated simultaneously from the SAXS curves at T=306{\\circ}C for DMPC concentrations in the range from 15 to 75 mM (1-5 % w/w). Structure factor correction to the scattering curve was shown to be negligibly small for the lipid concentration of 15 mM (1 % w/w). It was proved to be necessary to introduce structure factor correction to the scattering curves for lipid concentrations {\\ge}30 mM (2 % w/w).
Energy Technology Data Exchange (ETDEWEB)
Kim, Sang In; Kim, Bong Hwan; Kim, Jang Lyul; Lee, Jung Il [Health Physics Team, Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of)
2015-12-15
The calibration methods of neutron-measuring devices such as the neutron survey meter have advantages and disadvantages. To compare the calibration factors obtained by the shadow cone method and semi-empirical method, 10 neutron survey meters of five different types were used in this study. This experiment was performed at the Korea Atomic Energy Research Institute (KAERI; Daejeon, South Korea), and the calibration neutron fields were constructed using a {sup 252}Californium ({sup 252}Cf) neutron source, which was positioned in the center of the neutron irradiation room. The neutron spectra of the calibration neutron fields were measured by a europium-activated lithium iodide scintillator in combination with KAERI's Bonner sphere system. When the shadow cone method was used, 10 single moderator-based survey meters exhibited a smaller calibration factor by as much as 3.1 - 9.3% than that of the semi-empirical method. This finding indicates that neutron survey meters underestimated the scattered neutrons and attenuated neutrons (i.e., the total scatter corrections). This underestimation of the calibration factor was attributed to the fact that single moderator-based survey meters have an under-ambient dose equivalent response in the thermal or thermal-dominant neutron field. As a result, when the shadow cone method is used for a single moderator-based survey meter, an additional correction and the International Organization for Standardization standard 8529-2 for room-scattered neutrons should be considered.
International Nuclear Information System (INIS)
Kim, Sang In; Kim, Bong Hwan; Kim, Jang Lyul; Lee, Jung Il
2015-01-01
The calibration methods of neutron-measuring devices such as the neutron survey meter have advantages and disadvantages. To compare the calibration factors obtained by the shadow cone method and semi-empirical method, 10 neutron survey meters of five different types were used in this study. This experiment was performed at the Korea Atomic Energy Research Institute (KAERI; Daejeon, South Korea), and the calibration neutron fields were constructed using a 252 Californium ( 252 Cf) neutron source, which was positioned in the center of the neutron irradiation room. The neutron spectra of the calibration neutron fields were measured by a europium-activated lithium iodide scintillator in combination with KAERI's Bonner sphere system. When the shadow cone method was used, 10 single moderator-based survey meters exhibited a smaller calibration factor by as much as 3.1 - 9.3% than that of the semi-empirical method. This finding indicates that neutron survey meters underestimated the scattered neutrons and attenuated neutrons (i.e., the total scatter corrections). This underestimation of the calibration factor was attributed to the fact that single moderator-based survey meters have an under-ambient dose equivalent response in the thermal or thermal-dominant neutron field. As a result, when the shadow cone method is used for a single moderator-based survey meter, an additional correction and the International Organization for Standardization standard 8529-2 for room-scattered neutrons should be considered
A library least-squares approach for scatter correction in gamma-ray tomography
Meric, Ilker; Anton Johansen, Geir; Valgueiro Malta Moreira, Icaro
2015-03-01
Scattered radiation is known to lead to distortion in reconstructed images in Computed Tomography (CT). The effects of scattered radiation are especially more pronounced in non-scanning, multiple source systems which are preferred for flow imaging where the instantaneous density distribution of the flow components is of interest. In this work, a new method based on a library least-squares (LLS) approach is proposed as a means of estimating the scatter contribution and correcting for this. The validity of the proposed method is tested using the 85-channel industrial gamma-ray tomograph previously developed at the University of Bergen (UoB). The results presented here confirm that the LLS approach can effectively estimate the amounts of transmission and scatter components in any given detector in the UoB gamma-ray tomography system.
Two-photon exchange corrections in elastic lepton-proton scattering
Energy Technology Data Exchange (ETDEWEB)
Tomalak, Oleksandr; Vanderhaeghen, Marc [Johannes Gutenberg Universitaet Mainz (Germany)
2015-07-01
The measured value of the proton charge radius from the Lamb shift of energy levels in muonic hydrogen is in strong contradiction, by 7-8 standard deviations, with the value obtained from electronic hydrogen spectroscopy and the value extracted from unpolarized electron-proton scattering data. The dominant unaccounted higher order contribution in scattering experiments corresponds to the two photon exchange (TPE) diagram. The elastic contribution to the TPE correction was studied with the fixed momentum transfer dispersion relations and compared to the hadronic model with off-shell photon-nucleon vertices. A dispersion relation formalism with one subtraction was proposed. Theoretical predictions of the TPE elastic contribution to the unpolarized elastic electron-proton scattering and polarization transfer observables in the low momentum transfer region were made. The TPE formalism was generalized to the case of massive leptons and the elastic contribution was evaluated for the kinematics of upcoming muon-proton scattering experiment (MUSE).
Pseudoscalar form factors in tau-neutrino nucleon scattering
International Nuclear Information System (INIS)
Hagiwara, K.; Mawatari, K.; Yokoya, H.
2004-01-01
We investigate the pseudoscalar transition form factors of nucleon for quasi-elastic scattering and Δ resonance production in tau-neutrino nucleon scattering via the charged current interactions. Although the pseudoscalar form factors play an important role for the τ production in neutrino-nucleon scattering, these are not known well. In this Letter, we examine their effects in quasi-elastic scattering and Δ resonance production and find that the cross section, Q 2 distribution, and spin polarization of the produced τ ± leptons are quite sensitive to the pseudoscalar form factors
Energy Technology Data Exchange (ETDEWEB)
Park, Yang-Kyun, E-mail: ykpark@mgh.harvard.edu; Sharp, Gregory C.; Phillips, Justin; Winey, Brian A. [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, Massachusetts 02114 (United States)
2015-08-15
Purpose: To demonstrate the feasibility of proton dose calculation on scatter-corrected cone-beam computed tomographic (CBCT) images for the purpose of adaptive proton therapy. Methods: CBCT projection images were acquired from anthropomorphic phantoms and a prostate patient using an on-board imaging system of an Elekta infinity linear accelerator. Two previously introduced techniques were used to correct the scattered x-rays in the raw projection images: uniform scatter correction (CBCT{sub us}) and a priori CT-based scatter correction (CBCT{sub ap}). CBCT images were reconstructed using a standard FDK algorithm and GPU-based reconstruction toolkit. Soft tissue ROI-based HU shifting was used to improve HU accuracy of the uncorrected CBCT images and CBCT{sub us}, while no HU change was applied to the CBCT{sub ap}. The degree of equivalence of the corrected CBCT images with respect to the reference CT image (CT{sub ref}) was evaluated by using angular profiles of water equivalent path length (WEPL) and passively scattered proton treatment plans. The CBCT{sub ap} was further evaluated in more realistic scenarios such as rectal filling and weight loss to assess the effect of mismatched prior information on the corrected images. Results: The uncorrected CBCT and CBCT{sub us} images demonstrated substantial WEPL discrepancies (7.3 ± 5.3 mm and 11.1 ± 6.6 mm, respectively) with respect to the CT{sub ref}, while the CBCT{sub ap} images showed substantially reduced WEPL errors (2.4 ± 2.0 mm). Similarly, the CBCT{sub ap}-based treatment plans demonstrated a high pass rate (96.0% ± 2.5% in 2 mm/2% criteria) in a 3D gamma analysis. Conclusions: A priori CT-based scatter correction technique was shown to be promising for adaptive proton therapy, as it achieved equivalent proton dose distributions and water equivalent path lengths compared to those of a reference CT in a selection of anthropomorphic phantoms.
NLO corrections to the photon impact factor: Combining real and virtual corrections
International Nuclear Information System (INIS)
Bartels, J.; Colferai, D.; Kyrieleis, A.; Gieseke, S.
2002-08-01
In this third part of our calculation of the QCD NLO corrections to the photon impact factor we combine our previous results for the real corrections with the singular pieces of the virtual corrections and present finite analytic expressions for the quark-antiquark-gluon intermediate state inside the photon impact factor. We begin with a list of the infrared singular pieces of the virtual correction, obtained in the first step of our program. We then list the complete results for the real corrections (longitudinal and transverse photon polarization). In the next step we defined, for the real corrections, the collinear and soft singular regions and calculate their contributions to the impact factor. We then subtract the contribution due to the central region. Finally, we combine the real corrections with the singular pieces of the virtual corrections and obtain our finite results. (orig.)
49 CFR 325.73 - Microphone distance correction factors. 1
2010-10-01
... 49 Transportation 5 2010-10-01 2010-10-01 false Microphone distance correction factors. 1 325.73 Section 325.73 Transportation Other Regulations Relating to Transportation (Continued) FEDERAL MOTOR... MOTOR CARRIER NOISE EMISSION STANDARDS Correction Factors § 325.73 Microphone distance correction...
Monte Carlo evaluation of accuracy and noise properties of two scatter correction methods
International Nuclear Information System (INIS)
Narita, Y.; Eberl, S.; Nakamura, T.
1996-01-01
Two independent scatter correction techniques, transmission dependent convolution subtraction (TDCS) and triple-energy window (TEW) method, were evaluated in terms of quantitative accuracy and noise properties using Monte Carlo simulation (EGS4). Emission projections (primary, scatter and scatter plus primary) were simulated for 99m Tc and 201 Tl for numerical chest phantoms. Data were reconstructed with ordered-subset ML-EM algorithm including attenuation correction using the transmission data. In the chest phantom simulation, TDCS provided better S/N than TEW, and better accuracy, i.e., 1.0% vs -7.2% in myocardium, and -3.7% vs -30.1% in the ventricular chamber for 99m Tc with TDCS and TEW, respectively. For 201 Tl, TDCS provided good visual and quantitative agreement with simulated true primary image without noticeably increasing the noise after scatter correction. Overall TDCS proved to be more accurate and less noisy than TEW, facilitating quantitative assessment of physiological functions with SPECT
Compton scatter and randoms corrections for origin ensembles 3D PET reconstructions
Energy Technology Data Exchange (ETDEWEB)
Sitek, Arkadiusz [Harvard Medical School, Boston, MA (United States). Dept. of Radiology; Brigham and Women' s Hospital, Boston, MA (United States); Kadrmas, Dan J. [Utah Univ., Salt Lake City, UT (United States). Utah Center for Advanced Imaging Research (UCAIR)
2011-07-01
In this work we develop a novel approach to correction for scatter and randoms in reconstruction of data acquired by 3D positron emission tomography (PET) applicable to tomographic reconstruction done by the origin ensemble (OE) approach. The statistical image reconstruction using OE is based on calculation of expectations of the numbers of emitted events per voxel based on complete-data space. Since the OE estimation is fundamentally different than regular statistical estimators such those based on the maximum likelihoods, the standard methods of implementation of scatter and randoms corrections cannot be used. Based on prompts, scatter, and random rates, each detected event is graded in terms of a probability of being a true event. These grades are utilized by the Markov Chain Monte Carlo (MCMC) algorithm used in OE approach for calculation of the expectation over the complete-data space of the number of emitted events per voxel (OE estimator). We show that the results obtained with the OE are almost identical to results obtained by the maximum likelihood-expectation maximization (ML-EM) algorithm for reconstruction for experimental phantom data acquired using Siemens Biograph mCT 3D PET/CT scanner. The developed correction removes artifacts due to scatter and randoms in investigated 3D PET datasets. (orig.)
Virtual hadronic and heavy-fermion O({alpha}{sup 2}) corrections to Bhabha scattering
Energy Technology Data Exchange (ETDEWEB)
Actis, Stefano [Inst. fuer Theoretische Physik E, RWTH Aachen (Germany); Czakon, Michal [Wuerzburg Univ. (Germany). Inst. fuer Theoretische Physik und Astrophysik]|[Uniwersytet Slaski, Katowice (Poland). Inst. of Physics and Chemistry of Metals; Gluza, Janusz [Uniwersytet Slaski, Katowice (Poland). Inst. of Physics and Chemistry of Metals; Riemann, Tord [Deutsches Elektronen-Synchrotron (DESY), Zeuthen (Germany)
2008-07-15
Effects of vacuum polarization by hadronic and heavy-fermion insertions were the last unknown two-loop QED corrections to high-energy Bhabha scattering. Here we describe the corrections in detail and explore their numerical influence. The hadronic contributions to the virtual O({alpha}{sup 2}) QED corrections to the Bhabha-scattering cross-section are evaluated using dispersion relations and computing the convolution of hadronic data with perturbatively calculated kernel functions. The technique of dispersion integrals is also employed to derive the virtual O({alpha}{sup 2}) corrections generated by muon-, tau- and top-quark loops in the small electron-mass limit for arbitrary values of the internal-fermion masses. At a meson factory with 1 GeV center-of-mass energy the complete effect of hadronic and heavy-fermion corrections amounts to less than 0.5 per mille and reaches, at 10 GeV, up to about 2 per mille. At the Z resonance it amounts to 2.3 per mille at 3 degrees; overall, hadronic corrections are less than 4 per mille. For ILC energies (500 GeV or above), the combined effect of hadrons and heavy fermions becomes 6 per mille at 3 degrees; hadrons contribute less than 20 per mille in the whole angular region. (orig.)
Violation of the factorization theorem in large-angle radiative Bhabha scattering
International Nuclear Information System (INIS)
Arbuzov, A.B.; Kuraev, Eh.A.; Shajkhatdenov, B.G.
1998-01-01
The lowest order QED radiative corrections to the radiative large-angle Bhabha scattering process in the region where all the kinematical invariants are large compared to the electron mass are considered. We show that the leading logarithmic corrections do not factor before the Born cross section, contrary to the picture assumed in the renormalization group approach. Estimation of the leading and nonleading contributions for typical kinematics of the hard process for energy of Φ factory is done
The equivalent square concept for the head scatter factor based on scatter from flattening filter
International Nuclear Information System (INIS)
Kim, Siyong; Palta, Jatinder R.; Zhu, Timothy C.
1998-01-01
The equivalent field relationship between square and circular fields for the head scatter factor was evaluated at the source plane. The method was based on integrating the head scatter parameter for projected shaped fields in the source plane and finding a field that produced the same ratio of head scatter to primary dose on the central axis. A value of σ/R≅0.9 was obtained, where σ was one-half of the side length of the equivalent square and R was the radius of the circular field. The assumptions were that the equivalent field relationship for head scatter depends primarily on the characteristics of scatter from the flattening filter, and that the differential scatter-to-primary ratio of scatter from the flattening filter decreases linearly with the radius, within the physical radius of the flattening filter. Lam and co-workers showed empirically that the area-to-perimeter ratio formula, when applied to an equivalent square formula at the flattening filter plane, gave an accurate prediction of the head scatter factor. We have analytically investigated the validity of the area-to-perimeter ratio formula. Our results support the fact that the area-to-perimeter ratio formula can also be used as the equivalent field formula for head scatter at the source plane. The equivalent field relationships for wedge and tertiary collimator scatter were also evaluated. (author)
The equivalent square concept for the head scatter factor based on scatter from flattening filter
Energy Technology Data Exchange (ETDEWEB)
Kim, Siyong; Palta, Jatinder R.; Zhu, Timothy C. [Department of Radiation Oncology, University of Florida College of Medicine, Gainesville, Florida (United States)
1998-06-01
The equivalent field relationship between square and circular fields for the head scatter factor was evaluated at the source plane. The method was based on integrating the head scatter parameter for projected shaped fields in the source plane and finding a field that produced the same ratio of head scatter to primary dose on the central axis. A value of {sigma}/R{approx_equal}0.9 was obtained, where {sigma} was one-half of the side length of the equivalent square and R was the radius of the circular field. The assumptions were that the equivalent field relationship for head scatter depends primarily on the characteristics of scatter from the flattening filter, and that the differential scatter-to-primary ratio of scatter from the flattening filter decreases linearly with the radius, within the physical radius of the flattening filter. Lam and co-workers showed empirically that the area-to-perimeter ratio formula, when applied to an equivalent square formula at the flattening filter plane, gave an accurate prediction of the head scatter factor. We have analytically investigated the validity of the area-to-perimeter ratio formula. Our results support the fact that the area-to-perimeter ratio formula can also be used as the equivalent field formula for head scatter at the source plane. The equivalent field relationships for wedge and tertiary collimator scatter were also evaluated. (author)
International Nuclear Information System (INIS)
Boros, C.
1999-01-01
Recent measurement of the structure function F 2 υ in neutrino deep inelastic scattering allows us to compare structure functions measured in neutrino and charged lepton scattering for the first time with reasonable precision. The comparison between neutrino and muon structure functions made by the CCFR Collaboration indicates that there is a discrepancy between these structure functions at small Bjorken x values. In this talk I examine two effects which might account for this experimental discrepancy: nuclear shadowing corrections for neutrinos and contributions from strange and anti-strange quarks. Copyright (1999) World Scientific Publishing Co. Pte. Ltd
Flesia, C.; Schwendimann, P.
1992-01-01
The contribution of the multiple scattering to the lidar signal is dependent on the optical depth tau. Therefore, the radar analysis, based on the assumption that the multiple scattering can be neglected is limited to cases characterized by low values of the optical depth (tau less than or equal to 0.1) and hence it exclude scattering from most clouds. Moreover, all inversion methods relating lidar signal to number densities and particle size must be modified since the multiple scattering affects the direct analysis. The essential requests of a realistic model for lidar measurements which include the multiple scattering and which can be applied to practical situations follow. (1) Requested are not only a correction term or a rough approximation describing results of a certain experiment, but a general theory of multiple scattering tying together the relevant physical parameter we seek to measure. (2) An analytical generalization of the lidar equation which can be applied in the case of a realistic aerosol is requested. A pure analytical formulation is important in order to avoid the convergency and stability problems which, in the case of numerical approach, are due to the large number of events that have to be taken into account in the presence of large depth and/or a strong experimental noise.
Debye-Waller Factor in Neutron Scattering by Ferromagnetic Metals
Paradezhenko, G. V.; Melnikov, N. B.; Reser, B. I.
2018-04-01
We obtain an expression for the neutron scattering cross section in the case of an arbitrary interaction of the neutron with the crystal. We give a concise, simple derivation of the Debye-Waller factor as a function of the scattering vector and the temperature. For ferromagnetic metals above the Curie temperature, we estimate the Debye-Waller factor in the range of scattering vectors characteristic of polarized magnetic neutron scattering experiments. In the example of iron, we compare the results of harmonic and anharmonic approximations.
Teuho, Jarmo; Saunavaara, Virva; Tolvanen, Tuula; Tuokkola, Terhi; Karlsson, Antti; Tuisku, Jouni; Teräs, Mika
2017-10-01
In PET, corrections for photon scatter and attenuation are essential for visual and quantitative consistency. MR attenuation correction (MRAC) is generally conducted by image segmentation and assignment of discrete attenuation coefficients, which offer limited accuracy compared with CT attenuation correction. Potential inaccuracies in MRAC may affect scatter correction, because the attenuation image (μ-map) is used in single scatter simulation (SSS) to calculate the scatter estimate. We assessed the impact of MRAC to scatter correction using 2 scatter-correction techniques and 3 μ-maps for MRAC. Methods: The tail-fitted SSS (TF-SSS) and a Monte Carlo-based single scatter simulation (MC-SSS) algorithm implementations on the Philips Ingenuity TF PET/MR were used with 1 CT-based and 2 MR-based μ-maps. Data from 7 subjects were used in the clinical evaluation, and a phantom study using an anatomic brain phantom was conducted. Scatter-correction sinograms were evaluated for each scatter correction method and μ-map. Absolute image quantification was investigated with the phantom data. Quantitative assessment of PET images was performed by volume-of-interest and ratio image analysis. Results: MRAC did not result in large differences in scatter algorithm performance, especially with TF-SSS. Scatter sinograms and scatter fractions did not reveal large differences regardless of the μ-map used. TF-SSS showed slightly higher absolute quantification. The differences in volume-of-interest analysis between TF-SSS and MC-SSS were 3% at maximum in the phantom and 4% in the patient study. Both algorithms showed excellent correlation with each other with no visual differences between PET images. MC-SSS showed a slight dependency on the μ-map used, with a difference of 2% on average and 4% at maximum when a μ-map without bone was used. Conclusion: The effect of different MR-based μ-maps on the performance of scatter correction was minimal in non-time-of-flight 18 F-FDG PET
International Nuclear Information System (INIS)
Ito, Hiroshi; Iida, Hidehiro; Kinoshita, Toshibumi; Hatazawa, Jun; Okudera, Toshio; Uemura, Kazuo
1999-01-01
The transmission dependent convolution subtraction method which is one of the methods for scatter correction of SPECT was applied to the assessment of CBF using SPECT and I-123-IMP. The effects of scatter correction on regional distribution of CBF were evaluated on a pixel by pixel basis by means of an anatomic standardization technique. SPECT scan was performed on six healthy men. Image reconstruction was carried out with and without the scatter correction. All reconstructed images were globally normalized for the radioactivity of each pixel, and transformed into a standard brain anatomy. After anatomic standardization, the average SPECT images were calculated for scatter corrected and uncorrected groups, and these groups were compared on pixel by pixel basis. In the scatter uncorrected group, a significant overestimation of CBF was observed in the deep cerebral white matter, pons, thalamus, putamen, hippocampal region and cingulate gyrus as compared with scatter corrected group. A significant underestimation was observed in all neocortical regions, especially in the occipital and parietal lobes, and the cerebellar cortex. The regional distribution of CBF obtained by scatter corrected SPECT was similar to that obtained by O-15 water PET. The scatter correction is needed for the assessment of CBF using SPECT. (author)
Energy Technology Data Exchange (ETDEWEB)
Carloni Calame, C. [Southampton Univ. (United Kingdom). School of Physics; Czyz, H.; Gluza, J.; Gunia, M. [Silesia Univ., Katowice (Poland). Dept. of Field Theory and Particle Physics; Montagna, G. [Pavia Univ. (Italy). Dipt. di Fisica Nucleare e Teorica; INFN, Sezione di Pavia (Italy); Nicrosini, O.; Piccinini, F. [INFN, Sezione di Pavia (Italy); Riemann, T. [Deutsches Elektronen-Synchrotron (DESY), Zeuthen (Germany); Worek, M. [Wuppertal Univ. (Germany). Fachbereich C Physik
2011-07-15
Virtual fermionic N{sub f}=1 and N{sub f}=2 contributions to Bhabha scattering are combined with realistic real corrections at next-to-next-to-leading order in QED. The virtual corrections are determined by the package BHANNLOHF, and real corrections with the Monte Carlo generators BHAGEN-1PH, HELAC-PHEGAS and EKHARA. Numerical results are discussed at the energies of and with realistic cuts used at the {phi} factory DA{phi}NE, at the B factories PEP-II and KEK, and at the charm/{tau} factory BEPC II. We compare these complete calculations with the approximate ones realized in the generator BABAYAGA rate at NLO used at meson factories to evaluate their luminosities. For realistic reference event selections we find agreement for the NNLO leptonic and hadronic corrections within 0.07% or better and conclude that they are well accounted for in the generator by comparison with the present experimental accuracy. (orig.)
International Nuclear Information System (INIS)
Stodilka, Robert Z.; Msaki, Peter; Prato, Frank S.; Nicholson, Richard L.; Kemp, B.J.
1998-01-01
Mounting evidence indicates that scatter and attenuation are major confounds to objective diagnosis of brain disease by quantitative SPECT. There is considerable debate, however, as to the relative importance of scatter correction (SC) and attenuation correction (AC), and how they should be implemented. The efficacy of SC and AC for 99m Tc brain SPECT was evaluated using a two-compartment fully tissue-equivalent anthropomorphic head phantom. Four correction schemes were implemented: uniform broad-beam AC, non-uniform broad-beam AC, uniform SC+AC, and non-uniform SC+AC. SC was based on non-stationary deconvolution scatter subtraction, modified to incorporate a priori knowledge of either the head contour (uniform SC) or transmission map (non-uniform SC). The quantitative accuracy of the correction schemes was evaluated in terms of contrast recovery, relative quantification (cortical:cerebellar activity), uniformity ((coefficient of variation of 230 macro-voxels) x100%), and bias (relative to a calibration scan). Our results were: uniform broad-beam (μ=0.12cm -1 ) AC (the most popular correction): 71% contrast recovery, 112% relative quantification, 7.0% uniformity, +23% bias. Non-uniform broad-beam (soft tissue μ=0.12cm -1 ) AC: 73%, 114%, 6.0%, +21%, respectively. Uniform SC+AC: 90%, 99%, 4.9%, +12%, respectively. Non-uniform SC+AC: 93%, 101%, 4.0%, +10%, respectively. SC and AC achieved the best quantification; however, non-uniform corrections produce only small improvements over their uniform counterparts. SC+AC was found to be superior to AC; this advantage is distinct and consistent across all four quantification indices. (author)
TU-F-18C-03: X-Ray Scatter Correction in Breast CT: Advances and Patient Testing
International Nuclear Information System (INIS)
Ramamurthy, S; Sechopoulos, I
2014-01-01
Purpose: To further develop and perform patient testing of an x-ray scatter correction algorithm for dedicated breast computed tomography (BCT). Methods: A previously proposed algorithm for x-ray scatter signal reduction in BCT imaging was modified and tested with a phantom and on patients. A wireless electronic positioner system was designed and added to the BCT system that positions a tungsten plate in and out of the x-ray beam. The interpolation used by the algorithm was replaced with a radial basis function-based algorithm, with automated exclusion of non-valid sampled points due to patient motion or other factors. A 3D adaptive noise reduction filter was also introduced to reduce the impact of scatter quantum noise post-reconstruction. The impact on image quality of the improved algorithm was evaluated using a breast phantom and seven patient breasts, using quantitative metrics such signal difference (SD) and signal difference-to-noise ratios (SDNR) and qualitatively using image profiles. Results: The improvements in the algorithm resulted in a more robust interpolation step, with no introduction of image artifacts, especially at the imaged object boundaries, which was an issue in the previous implementation. Qualitative evaluation of the reconstructed slices and corresponding profiles show excellent homogeneity of both the background and the higher density features throughout the whole imaged object, as well as increased accuracy in the Hounsfield Units (HU) values of the tissues. Profiles also demonstrate substantial increase in both SD and SDNR between glandular and adipose regions compared to both the uncorrected and system-corrected images. Conclusion: The improved scatter correction algorithm can be reliably used during patient BCT acquisitions with no introduction of artifacts, resulting in substantial improvement in image quality. Its impact on actual clinical performance needs to be evaluated in the future. Research Agreement, Koning Corp., Hologic
Forward two-photon exchange in elastic lepton-proton scattering and hyperfine-splitting correction
Energy Technology Data Exchange (ETDEWEB)
Tomalak, Oleksandr [Johannes Gutenberg Universitaet, Institut fuer Kernphysik and PRISMA Cluster of Excellence, Mainz (Germany)
2017-08-15
We relate the forward two-photon exchange (TPE) amplitudes to integrals of the inclusive lepton-proton scattering cross sections. These relations yield an alternative way for the evaluation of the TPE correction to hyperfine-splitting (HFS) in the hydrogen-like atoms with an equivalent to the standard approach (Iddings, Drell and Sullivan) result implying the Burkhardt-Cottingham sum rule. For evaluation of the individual effects (e.g., elastic contribution) our approach yields a distinct result. We compare both methods numerically on examples of the elastic contribution and the full TPE correction to HFS in electronic and muonic hydrogen. (orig.)
International Nuclear Information System (INIS)
Shaw, C.G.; Ergun, D.L.; Myerowitz, P.D.; Van Lysel, M.S.; Mistretta, C.A.; Zarnstorff, W.C.; Crummy, A.B.
1982-01-01
The logarithmic amplification of video signals and the availability of data in digital form make digital subtraction videoangiography a suitable tool for videodensitometric estimation of physiological quantities. A system for this purpose was implemented with a digital video image processor. However, it was found that the radiation scattering and veiling glare present in the image-intensified video must be removed to make meaningful quantitations. An algorithm to make such a correction was developed and is presented. With this correction, the videodensitometry system was calibrated with phantoms and used to measure the left ventricular ejection fraction of a canine heart
Compton scatter correction in case of multiple crosstalks in SPECT imaging.
Sychra, J J; Blend, M J; Jobe, T H
1996-02-01
A strategy for Compton scatter correction in brain SPECT images was proposed recently. It assumes that two radioisotopes are used and that a significant portion of photons of one radioisotope (for example, Tc99m) spills over into the low energy acquisition window of the other radioisotope (for example, Tl201). We are extending this approach to cases of several radioisotopes with mutual, multiple and significant photon spillover. In the example above, one may correct not only the Tl201 image but also the Tc99m image corrupted by the Compton scatter originating from the small component of high energy Tl201 photons. The proposed extension is applicable to other anatomical domains (cardiac imaging).
Corrections on energy spectrum and scattering for fast neutron radiography at NECTAR facility
International Nuclear Information System (INIS)
Liu Shuquan; Thomas, Boucherl; Li Hang; Zou Yubin; Lu Yuanrong; Guo Zhiyu
2013-01-01
Distortions caused by the neutron spectrum and scattered neutrons are major problems in fast neutron radiography and should be considered for improving the image quality. This paper puts emphasis on the removal of these image distortions and deviations for fast neutron radiography performed at the NECTAR facility of the research reactor FRM-Ⅱ in Technische Universitaet Mounchen (TUM), Germany. The NECTAR energy spectrum is analyzed and established to modify the influence caused by the neutron spectrum, and the Point Scattered Function (PScF) simulated by the Monte-Carlo program MCNPX is used to evaluate scattering effects from the object and improve image quality. Good analysis results prove the sound effects of the above two corrections. (authors)
Corrections on energy spectrum and scatterings for fast neutron radiography at NECTAR facility
Liu, Shu-Quan; Bücherl, Thomas; Li, Hang; Zou, Yu-Bin; Lu, Yuan-Rong; Guo, Zhi-Yu
2013-11-01
Distortions caused by the neutron spectrum and scattered neutrons are major problems in fast neutron radiography and should be considered for improving the image quality. This paper puts emphasis on the removal of these image distortions and deviations for fast neutron radiography performed at the NECTAR facility of the research reactor FRM- II in Technische Universität München (TUM), Germany. The NECTAR energy spectrum is analyzed and established to modify the influence caused by the neutron spectrum, and the Point Scattered Function (PScF) simulated by the Monte-Carlo program MCNPX is used to evaluate scattering effects from the object and improve image quality. Good analysis results prove the sound effects of the above two corrections.
Scattering at low energies by potentials containing power-law corrections to the Coulomb interaction
International Nuclear Information System (INIS)
Kuitsinskii, A.A.
1986-01-01
The low-energy asymptotic behavior is found for the phase shifts and scattering amplitudes in the case of central potentials which decrease at infinity as n/r+ar /sup -a/,a 1. In problems of atomic and nuclear physics one is generally interested in collisions of clusters consisting of several charged particles. The effective interaction potential of such clusters contains long-range power law corrections to the Coulomb interaction that is presented
On the radiative corrections of deep inelastic scattering of muon neutrino on nucleon
International Nuclear Information System (INIS)
So Sang Guk
1986-01-01
The radiative corrections of deep inelastic scattering process VΜP→ ΜN are considered. Matrix element which takes Feynman one photon exchange diagrams into account at high transfer momentum are used. Based on calculation of the matrix element one can obtain matrix element for given process. It is shown that the effective cross section which takes one photon exchange into account is obtained. (author)
Role of humidity and other correction factors in the AAPM TG-21 dosimetry protocol
International Nuclear Information System (INIS)
Rogers, D.W.; Ross, C.K.
1988-01-01
A detailed derivation is presented of the formulas required to determine Ngas and Dmed in the AAPM TG-21 dosimetry protocol. This protocol specifies how to determine the absorbed dose in an electron or photon beam when using exposure or absorbed dose calibrated ion chambers. It is shown that the expression given in TG-21's recent letter of clarification is incorrect. Accounting for humidity correctly increases, by 0.4%, all absorbed dose determinations using an exposure calibrated ion chamber. Taking into account other correction factors in the equation for exposure could also have varying, but significant effects (possibly over 1%). These are the stem scatter correction, the axial nonuniformity correction and the electrode correction for electrodes made of different materials from the wall. Attention is drawn to differences in the definitions of the exposure and absorbed dose calibration factors, Nx and ND, respectively, as supplied by the NBS and the NRCC
Attenuation correction of myocardial SPECT by scatter-photopeak window method in normal subjects
International Nuclear Information System (INIS)
Okuda, Koichi; Nakajima, Kenichi; Matsuo, Shinro; Kinuya, Seigo; Motomura, Nobutoku; Kubota, Masahiro; Yamaki, Noriyasu; Maeda, Hisato
2009-01-01
Segmentation with scatter and photopeak window data using attenuation correction (SSPAC) method can provide a patient-specific non-uniform attenuation coefficient map only by using photopeak and scatter images without X-ray computed tomography (CT). The purpose of this study is to evaluate the performance of attenuation correction (AC) by the SSPAC method on normal myocardial perfusion database. A total of 32 sets of exercise-rest myocardial images with Tc-99m-sestamibi were acquired in both photopeak (140 keV±10%) and scatter (7% of lower side of the photopeak window) energy windows. Myocardial perfusion databases by the SSPAC method and non-AC (NC) were created from 15 female and 17 male subjects with low likelihood of cardiac disease using quantitative perfusion SPECT software. Segmental myocardial counts of a 17-segment model from these databases were compared on the basis of paired t test. AC average myocardial perfusion count was significantly higher than that in NC in the septal and inferior regions (P<0.02). On the contrary, AC average count was significantly lower in the anterolateral and apical regions (P<0.01). Coefficient variation of the AC count in the mid, apical and apex regions was lower than that of NC. The SSPAC method can improve average myocardial perfusion uptake in the septal and inferior regions and provide uniform distribution of myocardial perfusion. The SSPAC method could be a practical method of attenuation correction without X-ray CT. (author)
A new method for x-ray scatter correction: first assessment on a cone-beam CT experimental setup
International Nuclear Information System (INIS)
Rinkel, J; Gerfault, L; Esteve, F; Dinten, J-M
2007-01-01
Cone-beam computed tomography (CBCT) enables three-dimensional imaging with isotropic resolution and a shorter acquisition time compared to a helical CT scanner. Because a larger object volume is exposed for each projection, scatter levels are much higher than in collimated fan-beam systems, resulting in cupping artifacts, streaks and quantification inaccuracies. In this paper, a general method to correct for scatter in CBCT, without supplementary on-line acquisition, is presented. This method is based on scatter calibration through off-line acquisition combined with on-line analytical transformation based on physical equations, to adapt calibration to the object observed. The method was tested on a PMMA phantom and on an anthropomorphic thorax phantom. The results were validated by comparison to simulation for the PMMA phantom and by comparison to scans obtained on a commercial multi-slice CT scanner for the thorax phantom. Finally, the improvements achieved with the new method were compared to those obtained using a standard beam-stop method. The new method provided results that closely agreed with the simulation and with the conventional CT scanner, eliminating cupping artifacts and significantly improving quantification. Compared to the beam-stop method, lower x-ray doses and shorter acquisition times were needed, both divided by a factor of 9 for the same scatter estimation accuracy
Quality correction factors of composite IMRT beam deliveries: Theoretical considerations
International Nuclear Information System (INIS)
Bouchard, Hugo
2012-01-01
Purpose: In the scope of intensity modulated radiation therapy (IMRT) dosimetry using ionization chambers, quality correction factors of plan-class-specific reference (PCSR) fields are theoretically investigated. The symmetry of the problem is studied to provide recommendable criteria for composite beam deliveries where correction factors are minimal and also to establish a theoretical limit for PCSR delivery k Q factors. Methods: The concept of virtual symmetric collapsed (VSC) beam, being associated to a given modulated composite delivery, is defined in the scope of this investigation. Under symmetrical measurement conditions, any composite delivery has the property of having a k Q factor identical to its associated VSC beam. Using this concept of VSC, a fundamental property of IMRT k Q factors is demonstrated in the form of a theorem. The sensitivity to the conditions required by the theorem is thoroughly examined. Results: The theorem states that if a composite modulated beam delivery produces a uniform dose distribution in a volume V cyl which is symmetric with the cylindrical delivery and all beams fulfills two conditions in V cyl : (1) the dose modulation function is unchanged along the beam axis, and (2) the dose gradient in the beam direction is constant for a given lateral position; then its associated VSC beam produces no lateral dose gradient in V cyl , no matter what beam modulation or gantry angles are being used. The examination of the conditions required by the theorem lead to the following results. The effect of the depth-dose gradient not being perfectly constant with depth on the VSC beam lateral dose gradient is found negligible. The effect of the dose modulation function being degraded with depth on the VSC beam lateral dose gradient is found to be only related to scatter and beam hardening, as the theorem holds also for diverging beams. Conclusions: The use of the symmetry of the problem in the present paper leads to a valuable theorem showing
SU-D-206-04: Iterative CBCT Scatter Shading Correction Without Prior Information
International Nuclear Information System (INIS)
Bai, Y; Wu, P; Mao, T; Gong, S; Wang, J; Niu, T; Sheng, K; Xie, Y
2016-01-01
Purpose: To estimate and remove the scatter contamination in the acquired projection of cone-beam CT (CBCT), to suppress the shading artifacts and improve the image quality without prior information. Methods: The uncorrected CBCT images containing shading artifacts are reconstructed by applying the standard FDK algorithm on CBCT raw projections. The uncorrected image is then segmented to generate an initial template image. To estimate scatter signal, the differences are calculated by subtracting the simulated projections of the template image from the raw projections. Since scatter signals are dominantly continuous and low-frequency in the projection domain, they are estimated by low-pass filtering the difference signals and subtracted from the raw CBCT projections to achieve the scatter correction. Finally, the corrected CBCT image is reconstructed from the corrected projection data. Since an accurate template image is not readily segmented from the uncorrected CBCT image, the proposed scheme is iterated until the produced template is not altered. Results: The proposed scheme is evaluated on the Catphan©600 phantom data and CBCT images acquired from a pelvis patient. The result shows that shading artifacts have been effectively suppressed by the proposed method. Using multi-detector CT (MDCT) images as reference, quantitative analysis is operated to measure the quality of corrected images. Compared to images without correction, the method proposed reduces the overall CT number error from over 200 HU to be less than 50 HU and can increase the spatial uniformity. Conclusion: An iterative strategy without relying on the prior information is proposed in this work to remove the shading artifacts due to scatter contamination in the projection domain. The method is evaluated in phantom and patient studies and the result shows that the image quality is remarkably improved. The proposed method is efficient and practical to address the poor image quality issue of CBCT
SU-D-206-04: Iterative CBCT Scatter Shading Correction Without Prior Information
Energy Technology Data Exchange (ETDEWEB)
Bai, Y; Wu, P; Mao, T; Gong, S; Wang, J; Niu, T [Sir Run Run Shaw Hospital, Zhejiang University School of Medicine, Institute of Translational Medicine, Zhejiang University, Hangzhou, Zhejiang (China); Sheng, K [Department of Radiation Oncology, University of California, Los Angeles, School of Medicine, Los Angeles, CA (United States); Xie, Y [Institute of Biomedical and Health Engineering, Shenzhen Institutes of Advanced Technology, Chinese Academy of Sciences, Shenzhen, Guangdong (China)
2016-06-15
Purpose: To estimate and remove the scatter contamination in the acquired projection of cone-beam CT (CBCT), to suppress the shading artifacts and improve the image quality without prior information. Methods: The uncorrected CBCT images containing shading artifacts are reconstructed by applying the standard FDK algorithm on CBCT raw projections. The uncorrected image is then segmented to generate an initial template image. To estimate scatter signal, the differences are calculated by subtracting the simulated projections of the template image from the raw projections. Since scatter signals are dominantly continuous and low-frequency in the projection domain, they are estimated by low-pass filtering the difference signals and subtracted from the raw CBCT projections to achieve the scatter correction. Finally, the corrected CBCT image is reconstructed from the corrected projection data. Since an accurate template image is not readily segmented from the uncorrected CBCT image, the proposed scheme is iterated until the produced template is not altered. Results: The proposed scheme is evaluated on the Catphan©600 phantom data and CBCT images acquired from a pelvis patient. The result shows that shading artifacts have been effectively suppressed by the proposed method. Using multi-detector CT (MDCT) images as reference, quantitative analysis is operated to measure the quality of corrected images. Compared to images without correction, the method proposed reduces the overall CT number error from over 200 HU to be less than 50 HU and can increase the spatial uniformity. Conclusion: An iterative strategy without relying on the prior information is proposed in this work to remove the shading artifacts due to scatter contamination in the projection domain. The method is evaluated in phantom and patient studies and the result shows that the image quality is remarkably improved. The proposed method is efficient and practical to address the poor image quality issue of CBCT
Robust parameterization of elastic and absorptive electron atomic scattering factors
International Nuclear Information System (INIS)
Peng, L.M.; Ren, G.; Dudarev, S.L.; Whelan, M.J.
1996-01-01
A robust algorithm and computer program have been developed for the parameterization of elastic and absorptive electron atomic scattering factors. The algorithm is based on a combined modified simulated-annealing and least-squares method, and the computer program works well for fitting both elastic and absorptive atomic scattering factors with five Gaussians. As an application of this program, the elastic electron atomic scattering factors have been parameterized for all neutral atoms and for s up to 6 A -1 . Error analysis shows that the present results are considerably more accurate than the previous analytical fits in terms of the mean square value of the deviation between the numerical and fitted scattering factors. Parameterization for absorptive atomic scattering factors has been made for 17 important materials with the zinc blende structure over the temperature range 1 to 1000 K, where appropriate, and for temperature ranges for which accurate Debye-Waller factors are available. For other materials, the parameterization of the absorptive electron atomic scattering factors can be made using the program by supplying the atomic number of the element, the Debye-Waller factor and the acceleration voltage. For ions or when more accurate numerical results for neutral atoms are available, the program can read in the numerical values of the elastic scattering factors and return the parameters for both the elastic and absorptive scattering factors. The computer routines developed have been tested both on computer workstations and desktop PC computers, and will be made freely available via electronic mail or on floppy disk upon request. (orig.)
NNLO massive corrections to Bhabha scattering and theoretical precision of BabaYaga rate at NLO
International Nuclear Information System (INIS)
Carloni Calame, C.M.; Nicrosini, O.; Piccinini, F.; Riemann, T.; Worek, M.
2011-12-01
We provide an exact calculation of next-to-next-to-leading order (NNLO) massive corrections to Bhabha scattering in QED, relevant for precision luminosity monitoring at meson factories. Using realistic reference event selections, exact numerical results for leptonic and hadronic corrections are given and compared with the corresponding approximate predictions of the event generator BabaYaga rate at NLO. It is shown that the NNLO massive corrections are necessary for luminosity measurements with per mille precision. At the same time they are found to be well accounted for in the generator at an accuracy level below the one per mille. An update of the total theoretical precision of BabaYaga rate at NLO is presented and possible directions for a further error reduction are sketched. (orig.)
Next-to-soft corrections to high energy scattering in QCD and gravity
Energy Technology Data Exchange (ETDEWEB)
Luna, A.; Melville, S. [SUPA, School of Physics and Astronomy, University of Glasgow,Glasgow G12 8QQ, Scotland (United Kingdom); Naculich, S.G. [Department of Physics, Bowdoin College,Brunswick, ME 04011 (United States); White, C.D. [Centre for Research in String Theory, School of Physics and Astronomy,Queen Mary University of London,327 Mile End Road, London E1 4NS (United Kingdom)
2017-01-12
We examine the Regge (high energy) limit of 4-point scattering in both QCD and gravity, using recently developed techniques to systematically compute all corrections up to next-to-leading power in the exchanged momentum i.e. beyond the eikonal approximation. We consider the situation of two scalar particles of arbitrary mass, thus generalising previous calculations in the literature. In QCD, our calculation describes power-suppressed corrections to the Reggeisation of the gluon. In gravity, we confirm a previous conjecture that next-to-soft corrections correspond to two independent deflection angles for the incoming particles. Our calculations in QCD and gravity are consistent with the well-known double copy relating amplitudes in the two theories.
Verification of a table of phantom scatter factors for radiotherapy treatment planning
International Nuclear Information System (INIS)
Arts, J.K.; Bailey, M.J.; Hill, R.
2004-01-01
Full text: Many commercially available treatment planning systems require the medical physicist to measure and enter significant quantities of data for the verification of physics based algorithms. The CMS XiO (St. Louis, USA) treatment planning system requires a table of phantom scatter factors amongst other data. In a previous paper by Storchi et al, a table of phantom scatter factors is described. This table gives the phantom scatter factor as a function of field size and quality index determined from a collection of measured data for the total scatter factor and the collimator scatter factor from 25 different beam qualities ranging from 4MV up to 25MV. These factors have been determined at a fixed reference depth of 10cm for square fields of various sizes. This work investigates the claim that this table can be used as an alternative to calculated phantom scatter curve from measured data of a particular treatment unit. According to definition, it is difficult to directly measure the phantom scatter correction factor (Sp). This problem can be solved using the relation; S cp (A) = S c (A)S p (A) where S cp (A)) is the measured total scatter factor for a field size of square side dimension, A and S c (A) is the measured collimator scatter factor for a field size of square side dimension, A (Khan et al 1980, van Gasteren et al 1991). The total scatter correction factor (Sc,p) was measured in a full phantom, and the collimator scatter factor (Sc) measured using an ESTRO mini-phantom. These factors were measured on three Siemens linear accelerators (Concord, USA) with energies 6MV and 18MV and square field sizes ranging from 4x4cm to 40x40cm. The Primus and KD Mevatron produced 6 and 18MV X-rays and the MXE Mevatron produced 6Mv X-rays only. The values for Sp were calculated by rearranging equation (1). Phantom scatter factors were calculated from the data provided by Storchi et al using the quality index of each beam. For comparison, a set of Sp values was
Activated human neutrophils release hepatocyte growth factor/scatter factor.
LENUS (Irish Health Repository)
McCourt, M
2012-02-03
BACKGROUND: Hepatocyte growth factor or scatter factor (HGF\\/SF) is a pleiotropic cytokine that has potent angiogenic properties. We have previously demonstrated that neutrophils (PMN) are directly angiogenic by releasing vascular endothelial growth factor (VEGF). We hypothesized that the acute inflammatory response can stimulate PMN to release HGF. AIMS: To examine the effects of inflammatory mediators on PMN HGF release and the effect of recombinant human HGF (rhHGF) on PMN adhesion receptor expression and PMN VEGF release. METHODS: In the first experiment, PMN were isolated from healthy volunteers and stimulated with tumour necrosis factor-alpha (TNF-alpha), lipopolysaccharide (LPS), interleukin-8 (IL-8), and formyl methionyl-leucyl-phenylalanine (fMLP). Culture supernatants were assayed for HGF using ELISA. In the second experiment, PMN were lysed to measure total HGF release and HGF expression in the PMN was detected by Western immunoblotting. Finally, PMN were stimulated with rhHGF. PMN CD 11a, CD 11b, and CD 18 receptor expression and VEGF release was measured using flow cytometry and ELISA respectively. RESULTS: TNF-alpha, LPS and fMLP stimulation resulted in significantly increased release of PMN HGF (755+\\/-216, 484+\\/-221 and 565+\\/-278 pg\\/ml, respectively) compared to controls (118+\\/-42 pg\\/ml). IL-8 had no effect. Total HGF release following cell lysis and Western blot suggests that HGF is released from intracellular stores. Recombinant human HGF did not alter PMN adhesion receptor expression and had no effect on PMN VEGF release. CONCLUSIONS: This study demonstrates that pro-inflammatory mediators can stimulate HGF release from a PMN intracellular store and that activated PMN in addition to secreting VEGF have further angiogenic potential by releasing HGF.
Economic benefits of power factor correction at a nuclear facility
International Nuclear Information System (INIS)
Boger, R.M.; Dalos, W.; Juguilon, M.E.
1986-01-01
The economic benefits of correcting poor power factor at an operating nuclear facility are shown. A project approach for achieving rapid return of investment without disrupting plant availability is described. Examples of technical problems associated with using capacitors for power factor correction are presented
Coulomb corrections to nuclear scattering lengths and effective ranges for weakly bound systems
International Nuclear Information System (INIS)
Mur, V.D.; Popov, V.S.; Sergeev, A.V.
1996-01-01
A procedure is considered for extracting the purely nuclear scattering length as and effective range rs (which correspond to a strong-interaction potential Vs with disregarded Coulomb interaction) from the experimentally determined nuclear quantities acs and rcs, which are modified by Coulomb interaction. The Coulomb renormalization of as and rs is especially strong if the system under study involves a level with energy close to zero (on the nuclear scale). This applies to formulas that determine the Coulomb renormalization of the low-energy parameters of s scattering (l=0). Detailed numerical calculations are performed for coefficients appearing in the equations that determine Coulomb corrections for various models of the potential Vs(r). This makes it possible to draw qualitative conclusions that the dependence of Coulomb corrections on the form of the strong-interaction potential and, in particular, on its small-distance behavior. A considerable enhancement of Coulomb corrections to the effective range rs is found for potentials with a barrier
Energy Technology Data Exchange (ETDEWEB)
Kauppinen, T.; Vanninen, E.; Kuikka, J.T. [Kuopio Central Hospital (Finland). Dept. of Clinical Physiology; Koskinen, M.O. [Dept. of Clinical Physiology and Nuclear Medicine, Tampere Univ. Hospital, Tampere (Finland); Alenius, S. [Signal Processing Lab., Tampere Univ. of Technology, Tampere (Finland)
2000-09-01
Filtered back-projection (FBP) is generally used as the reconstruction method for single-photon emission tomography although it produces noisy images with apparent streak artefacts. It is possible to improve the image quality by using an algorithm with iterative correction steps. The iterative reconstruction technique also has an additional benefit in that computation of attenuation correction can be included in the process. A commonly used iterative method, maximum-likelihood expectation maximisation (ML-EM), can be accelerated using ordered subsets (OS-EM). We have applied to the OS-EM algorithm a Bayesian one-step late correction method utilising median root prior (MRP). Methodological comparison was performed by means of measurements obtained with a brain perfusion phantom and using patient data. The aim of this work was to quantitate the accuracy of iterative reconstruction with scatter and non-uniform attenuation corrections and post-filtering in SPET brain perfusion imaging. SPET imaging was performed using a triple-head gamma camera with fan-beam collimators. Transmission and emission scans were acquired simultaneously. The brain phantom used was a high-resolution three-dimensional anthropomorphic JB003 phantom. Patient studies were performed in ten chronic pain syndrome patients. The images were reconstructed using conventional FBP and iterative OS-EM and MRP techniques including scatter and nonuniform attenuation corrections. Iterative reconstructions were individually post-filtered. The quantitative results obtained with the brain perfusion phantom were compared with the known actual contrast ratios. The calculated difference from the true values was largest with the FBP method; iteratively reconstructed images proved closer to the reality. Similar findings were obtained in the patient studies. The plain OS-EM method improved the contrast whereas in the case of the MRP technique the improvement in contrast was not so evident with post-filtering. (orig.)
International Nuclear Information System (INIS)
Kauppinen, T.; Vanninen, E.; Kuikka, J.T.; Alenius, S.
2000-01-01
Filtered back-projection (FBP) is generally used as the reconstruction method for single-photon emission tomography although it produces noisy images with apparent streak artefacts. It is possible to improve the image quality by using an algorithm with iterative correction steps. The iterative reconstruction technique also has an additional benefit in that computation of attenuation correction can be included in the process. A commonly used iterative method, maximum-likelihood expectation maximisation (ML-EM), can be accelerated using ordered subsets (OS-EM). We have applied to the OS-EM algorithm a Bayesian one-step late correction method utilising median root prior (MRP). Methodological comparison was performed by means of measurements obtained with a brain perfusion phantom and using patient data. The aim of this work was to quantitate the accuracy of iterative reconstruction with scatter and non-uniform attenuation corrections and post-filtering in SPET brain perfusion imaging. SPET imaging was performed using a triple-head gamma camera with fan-beam collimators. Transmission and emission scans were acquired simultaneously. The brain phantom used was a high-resolution three-dimensional anthropomorphic JB003 phantom. Patient studies were performed in ten chronic pain syndrome patients. The images were reconstructed using conventional FBP and iterative OS-EM and MRP techniques including scatter and nonuniform attenuation corrections. Iterative reconstructions were individually post-filtered. The quantitative results obtained with the brain perfusion phantom were compared with the known actual contrast ratios. The calculated difference from the true values was largest with the FBP method; iteratively reconstructed images proved closer to the reality. Similar findings were obtained in the patient studies. The plain OS-EM method improved the contrast whereas in the case of the MRP technique the improvement in contrast was not so evident with post-filtering. (orig.)
Directory of Open Access Journals (Sweden)
Mehravar Rafati
2017-01-01
Conclusion: The simulation and the clinical studies showed that the new approach could be better performance than DEW, TEW methods, according to values of the contrast, and the SNR for scatter correction.
GEO-LEO reflectance band inter-comparison with BRDF and atmospheric scattering corrections
Chang, Tiejun; Xiong, Xiaoxiong Jack; Keller, Graziela; Wu, Xiangqian
2017-09-01
The inter-comparison of the reflective solar bands between the instruments onboard a geostationary orbit satellite and onboard a low Earth orbit satellite is very helpful to assess their calibration consistency. GOES-R was launched on November 19, 2016 and Himawari 8 was launched October 7, 2014. Unlike the previous GOES instruments, the Advanced Baseline Imager on GOES-16 (GOES-R became GOES-16 after November 29 when it reached orbit) and the Advanced Himawari Imager (AHI) on Himawari 8 have onboard calibrators for the reflective solar bands. The assessment of calibration is important for their product quality enhancement. MODIS and VIIRS, with their stringent calibration requirements and excellent on-orbit calibration performance, provide good references. The simultaneous nadir overpass (SNO) and ray-matching are widely used inter-comparison methods for reflective solar bands. In this work, the inter-comparisons are performed over a pseudo-invariant target. The use of stable and uniform calibration sites provides comparison with appropriate reflectance level, accurate adjustment for band spectral coverage difference, reduction of impact from pixel mismatching, and consistency of BRDF and atmospheric correction. The site in this work is a desert site in Australia (latitude -29.0 South; longitude 139.8 East). Due to the difference in solar and view angles, two corrections are applied to have comparable measurements. The first is the atmospheric scattering correction. The satellite sensor measurements are top of atmosphere reflectance. The scattering, especially Rayleigh scattering, should be removed allowing the ground reflectance to be derived. Secondly, the angle differences magnify the BRDF effect. The ground reflectance should be corrected to have comparable measurements. The atmospheric correction is performed using a vector version of the Second Simulation of a Satellite Signal in the Solar Spectrum modeling and BRDF correction is performed using a semi
International Nuclear Information System (INIS)
Laitinen, T.; Dalla, S.; Huttunen-Heikinmaa, K.; Valtonen, E.
2015-01-01
To understand the origin of Solar Energetic Particles (SEPs), we must study their injection time relative to other solar eruption manifestations. Traditionally the injection time is determined using the Velocity Dispersion Analysis (VDA) where a linear fit of the observed event onset times at 1 AU to the inverse velocities of SEPs is used to derive the injection time and path length of the first-arriving particles. VDA does not, however, take into account that the particles that produce a statistically observable onset at 1 AU have scattered in the interplanetary space. We use Monte Carlo test particle simulations of energetic protons to study the effect of particle scattering on the observable SEP event onset above pre-event background, and consequently on VDA results. We find that the VDA results are sensitive to the properties of the pre-event and event particle spectra as well as SEP injection and scattering parameters. In particular, a VDA-obtained path length that is close to the nominal Parker spiral length does not imply that the VDA injection time is correct. We study the delay to the observed onset caused by scattering of the particles and derive a simple estimate for the delay time by using the rate of intensity increase at the SEP onset as a parameter. We apply the correction to a magnetically well-connected SEP event of 2000 June 10, and show it to improve both the path length and injection time estimates, while also increasing the error limits to better reflect the inherent uncertainties of VDA
International Nuclear Information System (INIS)
Bardin, D.Yu.
1979-01-01
Basing on the simple quark-parton model of strong interaction and on the Weinberg-Salam theory compact formulae are derived for the radiative correction to the charged current induced deep inelastic scattering of neutrinos on nucleons. The radiative correction is found to be around 20-30%, i.e., the value typical for deep inelastic lN-scattering. The results obtained are rather different from the presently available estimations of the effect under consideration
Energy Technology Data Exchange (ETDEWEB)
Williams, Robert W. [Department of Biomedical Informatics, Uniformed Services University, 4301 Jones Bridge Road, Bethesda, MD 20815 (United States)], E-mail: bob@bob.usuhs.mil; Schluecker, Sebastian [Institute of Physical Chemistry, University of Wuerzburg, Wuerzburg (Germany); Hudson, Bruce S. [Department of Chemistry, Syracuse University, Syracuse, NY (United States)
2008-01-22
A scaled quantum mechanical harmonic force field (SQMFF) corrected for anharmonicity is obtained for the 23 K L-alanine crystal structure using van der Waals corrected periodic boundary condition density functional theory (DFT) calculations with the PBE functional. Scale factors are obtained with comparisons to inelastic neutron scattering (INS), Raman, and FT-IR spectra of polycrystalline L-alanine at 15-23 K. Calculated frequencies for all 153 normal modes differ from observed frequencies with a standard deviation of 6 wavenumbers. Non-bonded external k = 0 lattice modes are included, but assignments to these modes are presently ambiguous. The extension of SQMFF methodology to lattice modes is new, as are the procedures used here for providing corrections for anharmonicity and van der Waals interactions in DFT calculations on crystals. First principles Born-Oppenheimer molecular dynamics (BOMD) calculations are performed on the L-alanine crystal structure at a series of classical temperatures ranging from 23 K to 600 K. Corrections for zero-point energy (ZPE) are estimated by finding the classical temperature that reproduces the mean square displacements (MSDs) measured from the diffraction data at 23 K. External k = 0 lattice motions are weakly coupled to bonded internal modes.
International Nuclear Information System (INIS)
Williams, Robert W.; Schluecker, Sebastian; Hudson, Bruce S.
2008-01-01
A scaled quantum mechanical harmonic force field (SQMFF) corrected for anharmonicity is obtained for the 23 K L-alanine crystal structure using van der Waals corrected periodic boundary condition density functional theory (DFT) calculations with the PBE functional. Scale factors are obtained with comparisons to inelastic neutron scattering (INS), Raman, and FT-IR spectra of polycrystalline L-alanine at 15-23 K. Calculated frequencies for all 153 normal modes differ from observed frequencies with a standard deviation of 6 wavenumbers. Non-bonded external k = 0 lattice modes are included, but assignments to these modes are presently ambiguous. The extension of SQMFF methodology to lattice modes is new, as are the procedures used here for providing corrections for anharmonicity and van der Waals interactions in DFT calculations on crystals. First principles Born-Oppenheimer molecular dynamics (BOMD) calculations are performed on the L-alanine crystal structure at a series of classical temperatures ranging from 23 K to 600 K. Corrections for zero-point energy (ZPE) are estimated by finding the classical temperature that reproduces the mean square displacements (MSDs) measured from the diffraction data at 23 K. External k = 0 lattice motions are weakly coupled to bonded internal modes
Determination of epithermal flux correction factor (α) for irradiation ...
African Journals Online (AJOL)
Due to resonance that occur in the epithermal energy region of a reactor, the flux spectra in that region deviates strongly from the ideal I/E law to a I/E1+α with alpha as the correction factor. The factor has to be determined if zirconium as monitor pairs to determine the correction factor for inner irradiation channel 5 and outer ...
Energy Technology Data Exchange (ETDEWEB)
Kim, Ye-Seul; Park, Hye-Suk; Kim, Hee-Joung [Yonsei University, Wonju (Korea, Republic of); Choi, Young-Wook; Choi, Jae-Gu [Korea Electrotechnology Research Institute, Ansan (Korea, Republic of)
2014-12-15
Digital breast tomosynthesis (DBT) is a technique that was developed to overcome the limitations of conventional digital mammography by reconstructing slices through the breast from projections acquired at different angles. In developing and optimizing DBT, The x-ray scatter reduction technique remains a significant challenge due to projection geometry and radiation dose limitations. The most common approach to scatter reduction is a beam-stop-array (BSA) algorithm; however, this method raises concerns regarding the additional exposure involved in acquiring the scatter distribution. The compressed breast is roughly symmetric, and the scatter profiles from projections acquired at axially opposite angles are similar to mirror images. The purpose of this study was to apply the BSA algorithm with only two scans with a beam stop array, which estimates the scatter distribution with minimum additional exposure. The results of the scatter correction with angular interpolation were comparable to those of the scatter correction with all scatter distributions at each angle. The exposure increase was less than 13%. This study demonstrated the influence of the scatter correction obtained by using the BSA algorithm with minimum exposure, which indicates its potential for practical applications.
Numerical tables of anomalous scattering factors calculated by the Cromer and Liberman's method
International Nuclear Information System (INIS)
Sasaki, Satoshi.
1989-02-01
Anomalous scattering factors f' and f'' have been calculated for the atoms Li through Bi, plus U, using the relativistic treatment described by Cromer and Liberman. The final f' value does not include the Jensen's correction term on the magnetic scattering. The tables are presented with the f' and f'' values (i) at 0.01 A intervals in the wavelength range from 0.1 to 2.89 A and (ii) at 0.0001 A intervals in the neighborhood of the K, L 1 , L 2 , and L 3 absorption edges. (author)
Power factor correction (PFC) converters feeding brushless DC ...
African Journals Online (AJOL)
DR OKE
1Department of Electrical Engineering, Indian Institute of Technology Delhi, INDIA ... Hence, power factor correction (PFC) converters are used for achieving a unity ...... He is currently working as a Systems Engineer (Power IC) in AvantGarde ...
Kaneta, Tomohiro; Kurihara, Hideyuki; Hakamatsuka, Takashi; Ito, Hiroshi; Maruoka, Shin; Fukuda, Hiroshi; Takahashi, Shoki; Yamada, Shogo
2004-12-01
123I-15-(p-iodophenyl)-3-(R,S)-methylpentadecanoic acid (BMIPP) and 99mTc-tetrofosmin (TET) are widely used for evaluation of myocardial fatty acid metabolism and perfusion, respectively. ECG-gated TET SPECT is also used for evaluation of myocardial wall motion. These tests are often performed on the same day to minimize both the time required and inconvenience to patients and medical staff. However, as 123I and 99mTc have similar emission energies (159 keV and 140 keV, respectively), it is necessary to consider not only scattered photons, but also primary photons of each radionuclide detected in the wrong window (cross-talk). In this study, we developed and evaluated the effectiveness of a new scatter and cross-talk correction imaging protocol. Fourteen patients with ischemic heart disease or heart failure (8 men and 6 women with a mean age of 69.4 yr, ranging from 45 to 94 yr) were enrolled in this study. In the routine one-day acquisition protocol, BMIPP SPECT was performed in the morning, with TET SPECT performed 4 h later. An additional SPECT was performed just before injection of TET with the energy window for 99mTc. These data correspond to the scatter and cross-talk factor of the next TET SPECT. The correction was performed by subtraction of the scatter and cross-talk factor from TET SPECT. Data are presented as means +/- S.E. Statistical analyses were performed using Wilcoxon's matched-pairs signed-ranks test, and p corrected total count was 26.0 +/- 5.3%. EDV and ESV after correction were significantly greater than those before correction (p = 0.019 and 0.016, respectively). After correction, EF was smaller than that before correction, but the difference was not significant. Perfusion scores (17 segments per heart) were significantly lower after as compared with those before correction (p correction revealed significant differences in EDV, ESV, and perfusion scores. These observations indicate that scatter and cross-talk correction is required for one
Journal Impact Factor: Do the Numerator and Denominator Need Correction?
Liu, Xue-Li; Gai, Shuang-Shuang; Zhou, Jing
2016-01-01
To correct the incongruence of document types between the numerator and denominator in the traditional impact factor (IF), we make a corresponding adjustment to its formula and present five corrective IFs: IFTotal/Total, IFTotal/AREL, IFAR/AR, IFAREL/AR, and IFAREL/AREL. Based on a survey of researchers in the fields of ophthalmology and mathematics, we obtained the real impact ranking of sample journals in the minds of peer experts. The correlations between various IFs and questionnaire score were analyzed to verify their journal evaluation effects. The results show that it is scientific and reasonable to use five corrective IFs for journal evaluation for both ophthalmology and mathematics. For ophthalmology, the journal evaluation effects of the five corrective IFs are superior than those of traditional IF: the corrective effect of IFAR/AR is the best, IFAREL/AR is better than IFTotal/Total, followed by IFTotal/AREL, and IFAREL/AREL. For mathematics, the journal evaluation effect of traditional IF is superior than those of the five corrective IFs: the corrective effect of IFTotal/Total is best, IFAREL/AR is better than IFTotal/AREL and IFAREL/AREL, and the corrective effect of IFAR/AR is the worst. In conclusion, not all disciplinary journal IF need correction. The results in the current paper show that to correct the IF of ophthalmologic journals may be valuable, but it seems to be meaningless for mathematic journals. PMID:26977697
International Nuclear Information System (INIS)
Groiselle, C.; Rocchisani, J.M.; Moretti, J.L.; Dreuille, O. de; Gaillard, J.F.; Bendriem, B.
1997-01-01
SPECT quantification: a review of the different correction methods with Compton scatter attenuation and spatial deterioration effects. The improvement of gamma-cameras, acquisition and reconstruction software opens new perspectives in term of image quantification in nuclear medicine. In order to meet the challenge, numerous works have been undertaken in recent years to correct for the different physical phenomena that prevent an exact estimation of the radioactivity distribution. The main phenomena that have to betaken into account are scatter, attenuation and resolution. In this work, authors present the physical basis of each issue, its consequences on quantification and the main methods proposed to correct them. (authors)
DEFF Research Database (Denmark)
Ploquin, N.; Kertzscher, Gustavo; Vandervoort, E.
2015-01-01
A dosimetry system based on Al2O3:C radioluminescence (RL), and RADPOS, a novel 4D dosimetry system using microMOSFETs, were used to measure total scatter factors, (Sc,p)fclindet, for the CyberKnife robotic radiosugery system. New Monte Carlo calculated correction factors are presented and applied...
Volume correction factor in time dose relationships in brachytherapy
International Nuclear Information System (INIS)
Supe, S.J.; Sasane, J.B.
1987-01-01
Paterson's clinical data about the maximum tolerance doses for various volumes of interstitial implants with Ra-226 delivered in seven days was made use of in deriving volume correction factors for TDF and CRE concepts respectively for brachytherapy. The derived volume correction factors for TDF and for CRE differ fromthe one assumed for CRE by Kirk et al. and implied for TDF by Goitein. A normalising volume of 70 cc has been suggested for both CRE and TDF concepts for brachytherapy. A table showing the volume corrected TDF is presented for various volumes and dose rates for continuous irradiation. The use of this table is illustrated with examples. (orig.) [de
International Nuclear Information System (INIS)
Ramamurthy, Senthil; D’Orsi, Carl J; Sechopoulos, Ioannis
2016-01-01
A previously proposed x-ray scatter correction method for dedicated breast computed tomography was further developed and implemented so as to allow for initial patient testing. The method involves the acquisition of a complete second set of breast CT projections covering 360° with a perforated tungsten plate in the path of the x-ray beam. To make patient testing feasible, a wirelessly controlled electronic positioner for the tungsten plate was designed and added to a breast CT system. Other improvements to the algorithm were implemented, including automated exclusion of non-valid primary estimate points and the use of a different approximation method to estimate the full scatter signal. To evaluate the effectiveness of the algorithm, evaluation of the resulting image quality was performed with a breast phantom and with nine patient images. The improvements in the algorithm resulted in the avoidance of introduction of artifacts, especially at the object borders, which was an issue in the previous implementation in some cases. Both contrast, in terms of signal difference and signal difference-to-noise ratio were improved with the proposed method, as opposed to with the correction algorithm incorporated in the system, which does not recover contrast. Patient image evaluation also showed enhanced contrast, better cupping correction, and more consistent voxel values for the different tissues. The algorithm also reduces artifacts present in reconstructions of non-regularly shaped breasts. With the implemented hardware and software improvements, the proposed method can be reliably used during patient breast CT imaging, resulting in improvement of image quality, no introduction of artifacts, and in some cases reduction of artifacts already present. The impact of the algorithm on actual clinical performance for detection, diagnosis and other clinical tasks in breast imaging remains to be evaluated. (paper)
Correction factor for the experimental prompt neutron decay constant
International Nuclear Information System (INIS)
Talamo, Alberto; Gohar, Y.; Sadovich, S.; Kiyavitskaya, H.; Bournos, V.; Fokov, Y.; Routkovskaya, C.
2013-01-01
Highlights: • Definition of a spatial correction factor for the experimental prompt neutron decay constant. • Introduction of a MCNP6 calculation methodology to simulate Rossi-alpha distribution for pulsed neutron sources. • Comparison of MCNP6 results with experimental data for count rate, Rossi-alpha, and Feynman-alpha distributions. • Improvement of the comparison between numerical and experimental results by taking into account the dead-time effect. - Abstract: This study introduces a new correction factor to obtain the experimental effective multiplication factor of subcritical assemblies by the point kinetics formulation. The correction factor is defined as the ratio between the MCNP6 prompt neutron decay constant obtained in criticality mode and the one obtained in source mode. The correction factor mainly takes into account the longer neutron lifetime in the reflector region and the effects of the external neutron source. For the YALINA Thermal facility, the comparison between the experimental and computational effective multiplication factors noticeably improves after the application of the correction factor. The accuracy of the MCNP6 computational model of the YALINA Thermal subcritical assembly has been verified by reproducing the neutron count rate, Rossi-α, and Feynman-α distributions obtained from the experimental data
Multiple scattering corrections to the Beer-Lambert law. 2: Detector with a variable field of view.
Zardecki, A; Tam, W G
1982-07-01
The multiple scattering corrections to the Beer-Lambert law in the case of a detector with a variable field of view are analyzed. We introduce transmission functions relating the received radiant power to reference power levels relevant to two different experimental situations. In the first case, the transmission function relates the received power to a reference power level appropriate to a nonattenuating medium. In the second case, the reference power level is established by bringing the receiver to the close-up position with respect to the source. To examine the effect of the variation of the detector field of view the behavior of the gain factor is studied. Numerical results modeling the laser beam propagation in fog, cloud, and rain are presented.
SU-D-206-07: CBCT Scatter Correction Based On Rotating Collimator
International Nuclear Information System (INIS)
Yu, G; Feng, Z; Yin, Y; Qiang, L; Li, B; Huang, P; Li, D
2016-01-01
Purpose: Scatter correction in cone-beam computed tomography (CBCT) has obvious effect on the removal of image noise, the cup artifact and the increase of image contrast. Several methods using a beam blocker for the estimation and subtraction of scatter have been proposed. However, the inconvenience of mechanics and propensity to residual artifacts limited the further evolution of basic and clinical research. Here, we propose a rotating collimator-based approach, in conjunction with reconstruction based on a discrete Radon transform and Tchebichef moments algorithm, to correct scatter-induced artifacts. Methods: A rotating-collimator, comprising round tungsten alloy strips, was mounted on a linear actuator. The rotating-collimator is divided into 6 portions equally. The round strips space is evenly spaced on each portion but staggered between different portions. A step motor connected to the rotating collimator drove the blocker to around x-ray source during the CBCT acquisition. The CBCT reconstruction based on a discrete Radon transform and Tchebichef moments algorithm is performed. Experimental studies using water phantom and Catphan504 were carried out to evaluate the performance of the proposed scheme. Results: The proposed algorithm was tested on both the Monte Carlo simulation and actual experiments with the Catphan504 phantom. From the simulation result, the mean square error of the reconstruction error decreases from 16% to 1.18%, the cupping (τcup) from 14.005% to 0.66%, and the peak signal-to-noise ratio increase from 16.9594 to 31.45. From the actual experiments, the induced visual artifacts are significantly reduced. Conclusion: We conducted an experiment on CBCT imaging system with a rotating collimator to develop and optimize x-ray scatter control and reduction technique. The proposed method is attractive in applications where a high CBCT image quality is critical, for example, dose calculation in adaptive radiation therapy. We want to thank Dr. Lei
Asgari, Afrouz; Ashoor, Mansour; Sohrabpour, Mostafa; Shokrani, Parvaneh; Rezaei, Ali
2015-05-01
Improving signal to noise ratio (SNR) and qualified images by the various methods is very important for detecting the abnormalities at the body organs. Scatter and attenuation of photons by the organs lead to errors in radiopharmaceutical estimation as well as degradation of images. The choice of suitable energy window and the radionuclide have a key role in nuclear medicine which appearing the lowest scatter fraction as well as having a nearly constant linear attenuation coefficient as a function of phantom thickness. The energy windows of symmetrical window (SW), asymmetric window (ASW), high window (WH) and low window (WL) using Tc-99m and Sm-153 radionuclide with solid water slab phantom (RW3) and Teflon bone phantoms have been compared, and Matlab software and Monte Carlo N-Particle (MCNP4C) code were modified to simulate these methods and obtaining the amounts of FWHM and full width at tenth maximum (FWTM) using line spread functions (LSFs). The experimental data were obtained from the Orbiter Scintron gamma camera. Based on the results of the simulation as well as experimental work, the performance of WH and ASW display of the results, lowest scatter fraction as well as constant linear attenuation coefficient as a function of phantom thickness. WH and ASW were optimal windows in nuclear medicine imaging for Tc-99m in RW3 phantom and Sm-153 in Teflon bone phantom. Attenuation correction was done for WH and ASW optimal windows and for these radionuclides using filtered back projection algorithm. Results of simulation and experimental show that very good agreement between the set of experimental with simulation as well as theoretical values with simulation data were obtained which was nominally less than 7.07 % for Tc-99m and less than 8.00 % for Sm-153. Corrected counts were not affected by the thickness of scattering material. The Simulated results of Line Spread Function (LSF) for Sm-153 and Tc-99m in phantom based on four windows and TEW method were
SU-D-206-07: CBCT Scatter Correction Based On Rotating Collimator
Energy Technology Data Exchange (ETDEWEB)
Yu, G; Feng, Z [Shandong Normal University, Jinan, Shandong (China); Yin, Y [Shandong Cancer Hospital and Institute, China, Jinan, Shandong (China); Qiang, L [Zhang Jiagang STFK Medical Device Co, Zhangjiangkang, Suzhou (China); Li, B [Shandong Academy of Medical Sciences, Jinan, Shandong provice (China); Huang, P [Shandong Province Key Laboratory of Medical Physics and Image Processing Te, Ji’nan, Shandong province (China); Li, D [School of Physics and Electronics, Shandong Normal University, Jinan, Shandong (China)
2016-06-15
Purpose: Scatter correction in cone-beam computed tomography (CBCT) has obvious effect on the removal of image noise, the cup artifact and the increase of image contrast. Several methods using a beam blocker for the estimation and subtraction of scatter have been proposed. However, the inconvenience of mechanics and propensity to residual artifacts limited the further evolution of basic and clinical research. Here, we propose a rotating collimator-based approach, in conjunction with reconstruction based on a discrete Radon transform and Tchebichef moments algorithm, to correct scatter-induced artifacts. Methods: A rotating-collimator, comprising round tungsten alloy strips, was mounted on a linear actuator. The rotating-collimator is divided into 6 portions equally. The round strips space is evenly spaced on each portion but staggered between different portions. A step motor connected to the rotating collimator drove the blocker to around x-ray source during the CBCT acquisition. The CBCT reconstruction based on a discrete Radon transform and Tchebichef moments algorithm is performed. Experimental studies using water phantom and Catphan504 were carried out to evaluate the performance of the proposed scheme. Results: The proposed algorithm was tested on both the Monte Carlo simulation and actual experiments with the Catphan504 phantom. From the simulation result, the mean square error of the reconstruction error decreases from 16% to 1.18%, the cupping (τcup) from 14.005% to 0.66%, and the peak signal-to-noise ratio increase from 16.9594 to 31.45. From the actual experiments, the induced visual artifacts are significantly reduced. Conclusion: We conducted an experiment on CBCT imaging system with a rotating collimator to develop and optimize x-ray scatter control and reduction technique. The proposed method is attractive in applications where a high CBCT image quality is critical, for example, dose calculation in adaptive radiation therapy. We want to thank Dr. Lei
Meyer, Michael; Kalender, Willi A.; Kyriakou, Yiannis
2010-01-01
Scattered radiation is a major source of artifacts in flat detector computed tomography (FDCT) due to the increased irradiated volumes. We propose a fast projection-based algorithm for correction of scatter artifacts. The presented algorithm combines a convolution method to determine the spatial distribution of the scatter intensity distribution with an object-size-dependent scaling of the scatter intensity distributions using a priori information generated by Monte Carlo simulations. A projection-based (PBSE) and an image-based (IBSE) strategy for size estimation of the scanned object are presented. Both strategies provide good correction and comparable results; the faster PBSE strategy is recommended. Even with such a fast and simple algorithm that in the PBSE variant does not rely on reconstructed volumes or scatter measurements, it is possible to provide a reasonable scatter correction even for truncated scans. For both simulations and measurements, scatter artifacts were significantly reduced and the algorithm showed stable behavior in the z-direction. For simulated voxelized head, hip and thorax phantoms, a figure of merit Q of 0.82, 0.76 and 0.77 was reached, respectively (Q = 0 for uncorrected, Q = 1 for ideal). For a water phantom with 15 cm diameter, for example, a cupping reduction from 10.8% down to 2.1% was achieved. The performance of the correction method has limitations in the case of measurements using non-ideal detectors, intensity calibration, etc. An iterative approach to overcome most of these limitations was proposed. This approach is based on root finding of a cupping metric and may be useful for other scatter correction methods as well. By this optimization, cupping of the measured water phantom was further reduced down to 0.9%. The algorithm was evaluated on a commercial system including truncated and non-homogeneous clinically relevant objects.
Correction factors for clinical dosemeters used in large field dosimetry
International Nuclear Information System (INIS)
Campos, L.L.; Caldas, L.
1989-08-01
The determination of the absorbed dose in high-energy photon and electron beams by the user is carried out as a two-step procedure. First the ionization chamber is calibrated at a reference quality by the user at a standard laboratory, and then the chamber is used to determine the absorbed dose with the user's beam. A number of conversion and correction factors have to be applied. Different sets of factors are needed depending on the physical quantity the calibration refers to, the calibration geometry and the chamber design. Another correction factor to be introduced for the absorbed dose determination in large fields refers to radiation effects on the stem, cable and sometimes connectors. A simple method was developed to be suggested to hospital physicists to be followed during large radiation field dosimetry, in order to evaluate the radiation effects of cables and connectors and to determine correction factors for each system or geometry. (author) [pt
Heavy flavour corrections to polarised and unpolarised deep-inelastic scattering at 3-loop order
International Nuclear Information System (INIS)
Ablinger, J.; Round, M.; Schneider, C.; Hasselhuhn, A.
2016-11-01
We report on progress in the calculation of 3-loop corrections to the deep-inelastic structure functions from massive quarks in the asymptotic region of large momentum transfer Q"2. Recently completed results allow us to obtain the O(a"3_s) contributions to several heavy flavour Wilson coefficients which enter both polarised and unpolarised structure functions for lepton-nucleon scattering. In particular, we obtain the non-singlet contributions to the unpolarised structure functions F_2(x,Q"2) and xF_3(x,Q"2) and the polarised structure function g_1(x,Q"2). From these results we also obtain the heavy flavour contributions to the Gross-Llewellyn-Smith and the Bjorken sum rules.
Collinear factorization for deep inelastic scattering structure functions at large Bjorken xB
International Nuclear Information System (INIS)
Accardi, Alberto; Qiu, Jian-Wei
2008-01-01
http://dx.doi.org/10.1088/1126-6708/2008/07/090 We examine the uncertainty of perturbative QCD factorization for hadron structure functions in deep inelastic scattering at a large value of the Bjorken variable xB. We analyze the target mass correction to the structure functions by using the collinear factorization approach in the momentum space. We express the long distance physics of structure functions and the leading target mass corrections in terms of parton distribution functions with the standard operator definition. We compare our result with existing work on the target mass correction. We also discuss the impact of a final-state jet function on the extraction of parton distributions at large fractional momentum x.
Dual-energy digital mammography for calcification imaging: Scatter and nonuniformity corrections
International Nuclear Information System (INIS)
Kappadath, S. Cheenu; Shaw, Chris C.
2005-01-01
Mammographic images of small calcifications, which are often the earliest signs of breast cancer, can be obscured by overlapping fibroglandular tissue. We have developed and implemented a dual-energy digital mammography (DEDM) technique for calcification imaging under full-field imaging conditions using a commercially available aSi:H/CsI:Tl flat-panel based digital mammography system. The low- and high-energy images were combined using a nonlinear mapping function to cancel the tissue structures and generate the dual-energy (DE) calcification images. The total entrance-skin exposure and mean-glandular dose from the low- and high-energy images were constrained so that they were similar to screening-examination levels. To evaluate the DE calcification image, we designed a phantom using calcium carbonate crystals to simulate calcifications of various sizes (212-425 μm) overlaid with breast-tissue-equivalent material 5 cm thick with a continuously varying glandular-tissue ratio from 0% to 100%. We report on the effects of scatter radiation and nonuniformity in x-ray intensity and detector response on the DE calcification images. The nonuniformity was corrected by normalizing the low- and high-energy images with full-field reference images. Correction of scatter in the low- and high-energy images significantly reduced the background signal in the DE calcification image. Under the current implementation of DEDM, utilizing the mammography system and dose level tested, calcifications in the 300-355 μm size range were clearly visible in DE calcification images. Calcification threshold sizes decreased to the 250-280 μm size range when the visibility criteria were lowered to barely visible. Calcifications smaller than ∼250 μm were usually not visible in most cases. The visibility of calcifications with our DEDM imaging technique was limited by quantum noise, not system noise
Impact on dose and image quality of a software-based scatter correction in mammography.
Monserrat, Teresa; Prieto, Elena; Barbés, Benigno; Pina, Luis; Elizalde, Arlette; Fernández, Belén
2017-01-01
Background In 2014, Siemens developed a new software-based scatter correction (Progressive Reconstruction Intelligently Minimizing Exposure [PRIME]), enabling grid-less digital mammography. Purpose To compare doses and image quality between PRIME (grid-less) and standard (with anti-scatter grid) modes. Material and Methods Contrast-to-noise ratio (CNR) was measured for various polymethylmethacrylate (PMMA) thicknesses and dose values provided by the mammograph were recorded. CDMAM phantom images were acquired for various PMMA thicknesses and inverse Image Quality Figure (IQF inv ) was calculated. Values of incident entrance surface air kerma (ESAK) and average glandular dose (AGD) were obtained from the DICOM header for a total of 1088 pairs of clinical cases. Two experienced radiologists compared subjectively the image quality of a total of 149 pairs of clinical cases. Results CNR values were higher and doses were lower in PRIME mode for all thicknesses. IQF inv values in PRIME mode were lower for all thicknesses except for 40 mm of PMMA equivalent, in which IQF inv was slightly greater in PRIME mode. A mean reduction of 10% in ESAK and 12% in AGD in PRIME mode with respect to standard mode was obtained. The clinical image quality in PRIME and standard acquisitions resulted to be similar in most of the cases (84% for the first radiologist and 67% for the second one). Conclusion The use of PRIME software reduces, in average, the dose of radiation to the breast without affecting image quality. This reduction is greater for thinner and denser breasts.
An inter-crystal scatter correction method for DOI PET image reconstruction
International Nuclear Information System (INIS)
Lam, Chih Fung; Hagiwara, Naoki; Obi, Takashi; Yamaguchi, Masahiro; Yamaya, Taiga; Murayama, Hideo
2006-01-01
New positron emission tomography (PET) scanners utilize depth-of-interaction (DOI) information to improve image resolution, particularly at the edge of field-of-view while maintaining high detector sensitivity. However, the inter-crystal scatter (ICS) effect cannot be neglected in DOI scanners due to the use of smaller crystals. ICS is the phenomenon wherein there are multiple scintillations for irradiation of a gamma photon due to Compton scatter in detecting crystals. In the case of ICS, only one scintillation position is approximated for detectors with Anger-type logic calculation. This causes an error in position detection and ICS worsens the image contrast, particularly for smaller hotspots. In this study, we propose to model an ICS probability by using a Monte Carlo simulator. The probability is given as a statistical relationship between the gamma photon first interaction crystal pair and the detected crystal pair. It is then used to improve the system matrix of a statistical image reconstruction algorithm, such as maximum likehood expectation maximization (ML-EM) in order to correct for the position error caused by ICS. We apply the proposed method to simulated data of the jPET-D4, which is a four-layer DOI PET being developed at the National Institute of Radiological Sciences. Our computer simulations show that image contrast is recovered successfully by the proposed method. (author)
Reconstruction of atomic effective potentials from isotropic scattering factors
International Nuclear Information System (INIS)
Romera, E.; Angulo, J.C.; Torres, J.J.
2002-01-01
We present a method for the approximate determination of one-electron effective potentials of many-electron systems from a finite number of values of the isotropic scattering factor. The method is based on the minimum cross-entropy technique. An application to some neutral ground-state atomic systems has been done within a Hartree-Fock framework
Energy Technology Data Exchange (ETDEWEB)
Bluemlein, J.; Hasselhuhn, A.; Kovacikova, P.; Moch, S.
2011-04-15
We provide a fast and precise Mellin-space implementation of the O({alpha}{sub s}) heavy flavor Wilson coefficients for charged current deep inelastic scattering processes. They are of importance for the extraction of the strange quark distribution in neutrino-nucleon scattering and the QCD analyses of the HERA charged current data. Errors in the literature are corrected. We also discuss a series of more general parton parameterizations in Mellin space. (orig.)
International Nuclear Information System (INIS)
Sanchez Catasus, C.; Morales, L.; Aguila, A.
2002-01-01
Aim: It is well known that some patients with temporal lobe epilepsy (TLE) show normal perfusion during interictal SPECT study. The aim of this research was to evaluate if the scatter radiation has some influence on this kind of result. Materials and Methods: We studied 15 patients with TLE by clinical diagnosis and by video-EEG monitoring with surface electrodes (11 left TLE, 4 right TLE), which showed normal perfusion during interictal brain 99m Tc-HMPAO SPECT. The SPECT data were reconstructed by filtered backprojection without scatter correction (A). The same SPECT data were reconstructed after the projections were corrected by dual energy window method of scatter correction (B). Attenuation was corrected in all cases using first order Chang Method. For A and B images groups, cerebellum perfusion ratios were calculated on irregular regions of interest (ROI) drawn on anterior (ATL), lateral (LTL), mesial (MTL) and whole temporal lobe (WTL). To evaluate the influence of scatter radiation, the cerebellum perfusion ratios of each subject were compared with a normal database of 10 normal subjects, with and without scatter correction, using z-score analysis. Results: In group A, the z-score was less than 2 in all cases. In group B, the z-score was more than 2 in 6 cases, 4 in MTL (3 left, 1 right) and 2 in left LTL, which were coincident with the EEG localization. All images of group B showed better contrast than images of group A. Conclusions: These results suggest that scatter correction could improve the sensitivity of interictal brain SPECT to identify epileptic focus in patients with TLE
International Nuclear Information System (INIS)
Kojima, Akihiro; Matsumoto, Masanori; Ohyama, Yoichi; Tomiguchi, Seiji; Kira, Mitsuko; Takahashi, Mutsumasa.
1997-01-01
To investigate validity of scatter correction by the TEW method in 201 Tl imaging, we performed an experimental study using the gamma camera with the capability to perform the TEW method and a plate source with a defect. Images were acquired with the triple energy window which is recommended by the gamma camera manufacturer. The result of the energy spectrum showed that backscattered photons were included within the lower sub-energy window and main energy window, and the spectral shapes in the upper half region of the photopeak (70 keV) were not changed greatly by the source shape and the thickness of scattering materials. The scatter fraction calculated using energy spectra and, visual observation and the contrast values measured at the defect using planar images also showed that substantial primary photons were included in the upper sub-energy window. In TEW method (for scatter correction), two sub-energy windows are expected to be defined on the part of energy region in which total counts mainly consist of scattered photons. Therefore, it is necessary to investigate the use of the upper sub-energy window on scatter correction by the TEW method in 201 Tl imaging. (author)
Busi, Matteo; Olsen, Ulrik L.; Knudsen, Erik B.; Frisvad, Jeppe R.; Kehres, Jan; Dreier, Erik S.; Khalil, Mohamad; Haldrup, Kristoffer
2018-03-01
Spectral computed tomography is an emerging imaging method that involves using recently developed energy discriminating photon-counting detectors (PCDs). This technique enables measurements at isolated high-energy ranges, in which the dominating undergoing interaction between the x-ray and the sample is the incoherent scattering. The scattered radiation causes a loss of contrast in the results, and its correction has proven to be a complex problem, due to its dependence on energy, material composition, and geometry. Monte Carlo simulations can utilize a physical model to estimate the scattering contribution to the signal, at the cost of high computational time. We present a fast Monte Carlo simulation tool, based on McXtrace, to predict the energy resolved radiation being scattered and absorbed by objects of complex shapes. We validate the tool through measurements using a CdTe single PCD (Multix ME-100) and use it for scattering correction in a simulation of a spectral CT. We found the correction to account for up to 7% relative amplification in the reconstructed linear attenuation. It is a useful tool for x-ray CT to obtain a more accurate material discrimination, especially in the high-energy range, where the incoherent scattering interactions become prevailing (>50 keV).
Absorption correction factor in X-ray fluorescent quantitative analysis
International Nuclear Information System (INIS)
Pimjun, S.
1994-01-01
An experiment on absorption correction factor in X-ray fluorescent quantitative analysis were carried out. Standard samples were prepared from the mixture of Fe 2 O 3 and tapioca flour at various concentration of Fe 2 O 3 ranging from 5% to 25%. Unknown samples were kaolin containing 3.5% to-50% of Fe 2 O 3 Kaolin samples were diluted with tapioca flour in order to reduce the absorption of FeK α and make them easy to prepare. Pressed samples with 0.150 /cm 2 and 2.76 cm in diameter, were used in the experiment. Absorption correction factor is related to total mass absorption coefficient (χ) which varied with sample composition. In known sample, χ can be calculated by conveniently the formula. However in unknown sample, χ can be determined by Emission-Transmission method. It was found that the relationship between corrected FeK α intensity and contents of Fe 2 O 3 in these samples was linear. This result indicate that this correction factor can be used to adjust the accuracy of X-ray intensity. Therefore, this correction factor is essential in quantitative analysis of elements comprising in any sample by X-ray fluorescent technique
QCD and power corrections to sum rules in deep-inelastic lepton-nucleon scattering
International Nuclear Information System (INIS)
Ravindran, V.; Neerven, W.L. van
2001-01-01
In this paper we study QCD and power corrections to sum rules which show up in deep-inelastic lepton-hadron scattering. Furthermore we will make a distinction between fundamental sum rules which can be derived from quantum field theory and those which are of a phenomenological origin. Using current algebra techniques the fundamental sum rules can be expressed into expectation values of (partially) conserved (axial-)vector currents sandwiched between hadronic states. These expectation values yield the quantum numbers of the corresponding hadron which are determined by the underlying flavour group SU(n) F . In this case one can show that there exist an intimate relation between the appearance of power and QCD corrections. The above features do not hold for phenomenological sum rules, hereafter called non-fundamental. They have no foundation in quantum field theory and they mostly depend on certain assumptions made for the structure functions like super-convergence relations or the parton model. Therefore only the fundamental sum rules provide us with a stringent test of QCD
Distance factor on reducing scattered radiation risk during interventional fluoroscopy
International Nuclear Information System (INIS)
Husaini Salleh; Mohd Khalid Matori; Muhammad Jamal Mat Isa; Zainal Jamaluddin; Mohd Firdaus Abdul Rahman; Mohd Khairusalih Mohd Zin
2012-01-01
Interventional Radiology (IR) is subspecialty of diagnostic radiology where minimally invasive procedures are performed using an x-ray as a guidance. This procedure can deliver high radiation doses to patient and medical staff compared with other radiological method due to long screening time. The use of proper shielding, shorten the exposure time and keep the distance are the practices to reduce scattered radiation risks to staff involve in this procedure. This project is to study the distance factor on reducing the scattered radiation effect to the medical staff. It also may provide the useful information which can be use to establish the scattered radiation profile during the IR for the sake of radiation protection and safety to the medical staff involved. (author)
Analytical evaluation of atomic form factors: Application to Rayleigh scattering
Energy Technology Data Exchange (ETDEWEB)
Safari, L., E-mail: laleh.safari@ist.ac.at [IST Austria (Institute of Science and Technology Austria), Am Campus 1, 3400 Klosterneuburg (Austria); Department of Physics, University of Oulu, Box 3000, FI-90014 Oulu (Finland); Santos, J. P. [Laboratório de Instrumentação, Engenharia Biomédica e Física da Radiação (LIBPhys-UNL), Departamento de Física, Faculdade de Ciências e Tecnologia, FCT, Universidade Nova de Lisboa, 2829-516 Caparica (Portugal); Amaro, P. [Laboratório de Instrumentação, Engenharia Biomédica e Física da Radiação (LIBPhys-UNL), Departamento de Física, Faculdade de Ciências e Tecnologia, FCT, Universidade Nova de Lisboa, 2829-516 Caparica (Portugal); Physikalisches Institut, Universität Heidelberg, D-69120 Heidelberg (Germany); Jänkälä, K. [Department of Physics, University of Oulu, Box 3000, FI-90014 Oulu (Finland); Fratini, F. [Department of Physics, University of Oulu, Box 3000, FI-90014 Oulu (Finland); Institute of Atomic and Subatomic Physics, TU Wien, Stadionallee 2, 1020 Wien (Austria); Departamento de Física, Instituto de Ciências Exatas, Universidade Federal de Minas Gerais, 31270-901 Belo Horizonte, MG (Brazil)
2015-05-15
Atomic form factors are widely used for the characterization of targets and specimens, from crystallography to biology. By using recent mathematical results, here we derive an analytical expression for the atomic form factor within the independent particle model constructed from nonrelativistic screened hydrogenic wave functions. The range of validity of this analytical expression is checked by comparing the analytically obtained form factors with the ones obtained within the Hartee-Fock method. As an example, we apply our analytical expression for the atomic form factor to evaluate the differential cross section for Rayleigh scattering off neutral atoms.
Energy Technology Data Exchange (ETDEWEB)
Appaji Gowda, S.B. [Department of Studies in Physics, Manasagangothri, University of Mysore, Mysore 570006 (India); Umesh, T.K. [Department of Studies in Physics, Manasagangothri, University of Mysore, Mysore 570006 (India)]. E-mail: tku@physics.uni-mysore.ac.in
2006-01-15
Dispersion corrections to the forward Rayleigh scattering amplitudes of tantalum, mercury and lead in the photon energy range 24-136 keV have been determined by a numerical evaluation of the dispersion integral that relates them through optical theorem to the photo effect cross sections. The photo effect cross sections have been extracted by subtracting the coherent and incoherent scattering contribution from the measured total attenuation cross section, using high-resolution high-purity germanium detector in a narrow beam good geometry set up. The real part of the dispersion correction to which the relativistic corrections calculated by Kissel and Pratt (S-matrix approach) or Creagh and McAuley (multipole corrections) have been included are in better agreement with the available theoretical values.
Power factor correction (PFC) converters feeding brushless DC ...
African Journals Online (AJOL)
This paper presents a comprehensive study of power factor correction (PFC) converters for feeding brushless DC (BLDC) motor drive. This work explores various configurations of PFC converters which are classified into five different categories of non-isolated, bridgeless (BL) non-isolated, isolated, BL-isolated PFC ...
Status of the NLO Corrections to the Photon Impact Factor
Gieseke, Stefan
2002-01-01
We present the status of the programme of calculating the next-to-leading order corrections to the virtual photon impact factor. In particular, we discuss new results for the transversely polarized photon. We briefly outline the definition of infrared finite terms and the subtraction of the leading logarithmic parts.
Power Factor Correction for Thyristor Equipment in Glass Industry ...
African Journals Online (AJOL)
Thyristor power controllers are now widely used in the glass industry for controlling furnace temperature. While offering a number of operational advantages, they operate at lagging power factors which require correction for minimum power cost. Harmonic resonance with the utility feed, however, complicate the use of ...
International Nuclear Information System (INIS)
Bai, J.; Hashimoto, J.; Suzuki, T.; Nakahara, T.; Kubo, A.; Ohira, M.; Takao, M.; Ogawa, K.
2007-01-01
The aims of this study were to elucidate the feasibility of scatter correction in improving the quantitative accuracy of the Heart-to-Mediastinum (H/M) ratio in I-123 MIBG imaging and to clarify whether the H/M ratio calculated from the scatter corrected image improves the accuracy of differentiating patients with Parkinsonism from other neurological disorders. The H/M ratio was calculated using the counts from planar images processed with and without scatter correction in the phantom and on patients. The triple energy window (TEW) method was used for scatter correction. Fifty five patients were enrolled in the clinical study. The Receiver Operating Characteristic (ROC) Curve analysis was used to evaluate diagnostic performance. The H/M ratio was found to be increased after scatter correction in the phantom simulating normal cardiac uptake, while no changes were observed in the phantom simulating no uptake. It was observed that scatter correction stabilized the H/M ratio by eliminating the influence of scatter photons originating from the liver, especially in the condition of no cardiac uptake. Similarly, scatter correction increased the H/M ratio in conditions other than Parkinson's disease but did not show any change in Parkinson's disease itself to widen the differences in the H/M ratios between the two groups. The overall power of the test did not show any significant improvement after scatter correction in differentiating patients with Parkinsonism. Based on the results of this study it has been concluded that scatter correction improves the quantitative accuracy of H/M ratio in MIBG imaging, but it does not offer any significant incremental diagnostic value over conventional imaging (without scatter correction). Nevertheless it is felt that the scatter correction technique deserves special consideration in order to make the test more robust and obtain stable H/M ratios. (author)
Miwa, Kenta; Umeda, Takuro; Murata, Taisuke; Wagatsuma, Kei; Miyaji, Noriaki; Terauchi, Takashi; Koizumi, Mitsuru; Sasaki, Masayuki
2016-02-01
Overcorrection of scatter caused by patient motion during whole-body PET/computed tomography (CT) imaging can induce the appearance of photopenic artifacts in the PET images. The present study aimed to quantify the accuracy of scatter limitation correction (SLC) for eliminating photopenic artifacts. This study analyzed photopenic artifacts in (18)F-fluorodeoxyglucose ((18)F-FDG) PET/CT images acquired from 12 patients and from a National Electrical Manufacturers Association phantom with two peripheral plastic bottles that simulated the human body and arms, respectively. The phantom comprised a sphere (diameter, 10 or 37 mm) containing fluorine-18 solutions with target-to-background ratios of 2, 4, and 8. The plastic bottles were moved 10 cm posteriorly between CT and PET acquisitions. All PET data were reconstructed using model-based scatter correction (SC), no scatter correction (NSC), and SLC, and the presence or absence of artifacts on the PET images was visually evaluated. The SC and SLC images were also semiquantitatively evaluated using standardized uptake values (SUVs). Photopenic artifacts were not recognizable in any NSC and SLC image from all 12 patients in the clinical study. The SUVmax of mismatched SLC PET/CT images were almost equal to those of matched SC and SLC PET/CT images. Applying NSC and SLC substantially eliminated the photopenic artifacts on SC PET images in the phantom study. SLC improved the activity concentration of the sphere for all target-to-background ratios. The highest %errors of the 10 and 37-mm spheres were 93.3 and 58.3%, respectively, for mismatched SC, and 73.2 and 22.0%, respectively, for mismatched SLC. Photopenic artifacts caused by SC error induced by CT and PET image misalignment were corrected using SLC, indicating that this method is useful and practical for clinical qualitative and quantitative PET/CT assessment.
Energy Technology Data Exchange (ETDEWEB)
Kim, Kyungsang; Ye, Jong Chul, E-mail: jong.ye@kaist.ac.kr [Bio Imaging and Signal Processing Laboratory, Department of Bio and Brain Engineering, KAIST 291, Daehak-ro, Yuseong-gu, Daejeon 34141 (Korea, Republic of); Lee, Taewon; Cho, Seungryong [Medical Imaging and Radiotherapeutics Laboratory, Department of Nuclear and Quantum Engineering, KAIST 291, Daehak-ro, Yuseong-gu, Daejeon 34141 (Korea, Republic of); Seong, Younghun; Lee, Jongha; Jang, Kwang Eun [Samsung Advanced Institute of Technology, Samsung Electronics, 130, Samsung-ro, Yeongtong-gu, Suwon-si, Gyeonggi-do, 443-803 (Korea, Republic of); Choi, Jaegu; Choi, Young Wook [Korea Electrotechnology Research Institute (KERI), 111, Hanggaul-ro, Sangnok-gu, Ansan-si, Gyeonggi-do, 426-170 (Korea, Republic of); Kim, Hak Hee; Shin, Hee Jung; Cha, Joo Hee [Department of Radiology and Research Institute of Radiology, Asan Medical Center, University of Ulsan College of Medicine, 88 Olympic-ro, 43-gil, Songpa-gu, Seoul, 138-736 (Korea, Republic of)
2015-09-15
Purpose: In digital breast tomosynthesis (DBT), scatter correction is highly desirable, as it improves image quality at low doses. Because the DBT detector panel is typically stationary during the source rotation, antiscatter grids are not generally compatible with DBT; thus, a software-based scatter correction is required. This work proposes a fully iterative scatter correction method that uses a novel fast Monte Carlo simulation (MCS) with a tissue-composition ratio estimation technique for DBT imaging. Methods: To apply MCS to scatter estimation, the material composition in each voxel should be known. To overcome the lack of prior accurate knowledge of tissue composition for DBT, a tissue-composition ratio is estimated based on the observation that the breast tissues are principally composed of adipose and glandular tissues. Using this approximation, the composition ratio can be estimated from the reconstructed attenuation coefficients, and the scatter distribution can then be estimated by MCS using the composition ratio. The scatter estimation and image reconstruction procedures can be performed iteratively until an acceptable accuracy is achieved. For practical use, (i) the authors have implemented a fast MCS using a graphics processing unit (GPU), (ii) the MCS is simplified to transport only x-rays in the energy range of 10–50 keV, modeling Rayleigh and Compton scattering and the photoelectric effect using the tissue-composition ratio of adipose and glandular tissues, and (iii) downsampling is used because the scatter distribution varies rather smoothly. Results: The authors have demonstrated that the proposed method can accurately estimate the scatter distribution, and that the contrast-to-noise ratio of the final reconstructed image is significantly improved. The authors validated the performance of the MCS by changing the tissue thickness, composition ratio, and x-ray energy. The authors confirmed that the tissue-composition ratio estimation was quite
Perturbative corrections to B → D form factors in QCD
Wang, Yu-Ming; Wei, Yan-Bing; Shen, Yue-Long; Lü, Cai-Dian
2017-06-01
We compute perturbative QCD corrections to B → D form factors at leading power in Λ/ m b , at large hadronic recoil, from the light-cone sum rules (LCSR) with B-meson distribution amplitudes in HQET. QCD factorization for the vacuum-to- B-meson correlation function with an interpolating current for the D-meson is demonstrated explicitly at one loop with the power counting scheme {m}_c˜ O(√{Λ {m}_b}) . The jet functions encoding information of the hard-collinear dynamics in the above-mentioned correlation function are complicated by the appearance of an additional hard-collinear scale m c , compared to the counterparts entering the factorization formula of the vacuum-to- B-meson correction function for the construction of B → π from factors. Inspecting the next-to-leading-logarithmic sum rules for the form factors of B → Dℓν indicates that perturbative corrections to the hard-collinear functions are more profound than that for the hard functions, with the default theory inputs, in the physical kinematic region. We further compute the subleading power correction induced by the three-particle quark-gluon distribution amplitudes of the B-meson at tree level employing the background gluon field approach. The LCSR predictions for the semileptonic B → Dℓν form factors are then extrapolated to the entire kinematic region with the z-series parametrization. Phenomenological implications of our determinations for the form factors f BD +,0 ( q 2) are explored by investigating the (differential) branching fractions and the R( D) ratio of B → Dℓν and by determining the CKM matrix element |V cb | from the total decay rate of B → Dμν μ .
DEFF Research Database (Denmark)
Pinkevych, Mykola; Cromer, Deborah; Tolstrup, Martin
2016-01-01
[This corrects the article DOI: 10.1371/journal.ppat.1005000.][This corrects the article DOI: 10.1371/journal.ppat.1005740.][This corrects the article DOI: 10.1371/journal.ppat.1005679.].......[This corrects the article DOI: 10.1371/journal.ppat.1005000.][This corrects the article DOI: 10.1371/journal.ppat.1005740.][This corrects the article DOI: 10.1371/journal.ppat.1005679.]....
International Nuclear Information System (INIS)
Hambye, A.S.; Vervaet, A.; Dobbeleir, A.
2002-01-01
Compared to other non invasive testings for CAD diagnosis, myocardial perfusion imaging (MPI) is considered as a very sensitive method which accuracy is however often dimmed by a certain lack of specificity, especially in patients with a small heart. With gated SPECT MPI, use of end-diastolic instead of summed images has been presented as an interesting approach for increasing specificity. Since scatter correction is reported to improve image contrast, it might potentially constitute another way to ameliorate MPI accuracy. We aimed at comparing the value of both approaches, either separate or combined, for CAD diagnosis. Methods. Hundred patients addressed for gated 99m-Tc sestamibi SPECT MPI were prospectively included (Group A). Thirty-five had an end-systolic volume <30ml by QGS-analysis (Group B). All had a coronary angiogram within 3 months of the MPI. Four polar maps (non-corrected and scatter-corrected summed, and non-corrected and scatter-corrected end-diastolic) were created to quantify the extent (EXT) and severity (TDS) of the perfusion defects if any. ROC-curve analysis was applied to define the optimal thresholds of EXT and TDS separating non-CAD from CAD-patients, using a 50%-stenosis on coronary angiogram as cutoff for disease positivity. Results. Significant CAD was present in 86 patients (25 in Group B). In Group A, assessment of EXT and TDS of perfusion defects on scatter-corrected summed images demonstrated the highest accuracy (76% for EXT; sens: 77%; spec: 71%, and 74% for TDS, sens: 73%, spec: 79%). Accuracy of EXT and TDS calculated from the other data sets was slightly but not significantly lower, especially because of a lower sensitivity. As a comparison, visual analysis was 90% accurate for the diagnosis of CAD (sens: 94%, spec: 64%). In group B, overall results were worse mainly due to a decreased sensitivity, with accuracies ranging between 51 and 63%. Again scatter-corrected summed data were the most accurate (EXT: 60%, TDS: 63%, visual
Energy Technology Data Exchange (ETDEWEB)
Kitamura, Keishi [Shimadzu Corporation, 1 Nishinokyo-Kuwabaracho, Nakagyo-ku, Kyoto-shi, Kyoto 604-8511 (Japan)]. E-mail: kitam@shimadzu.co.jp; Ishikawa, Akihiro [Shimadzu Corporation, 1 Nishinokyo-Kuwabaracho, Nakagyo-ku, Kyoto-shi, Kyoto 604-8511 (Japan); Mizuta, Tetsuro [Shimadzu Corporation, 1 Nishinokyo-Kuwabaracho, Nakagyo-ku, Kyoto-shi, Kyoto 604-8511 (Japan); Yamaya, Taiga [National Institute of Radiological Sciences, 9-1 Anagawa-4, Inage-ku, Chiba-shi, Chiba 263-8555 (Japan); Yoshida, Eiji [National Institute of Radiological Sciences, 9-1 Anagawa-4, Inage-ku, Chiba-shi, Chiba 263-8555 (Japan); Murayama, Hideo [National Institute of Radiological Sciences, 9-1 Anagawa-4, Inage-ku, Chiba-shi, Chiba 263-8555 (Japan)
2007-02-01
The jPET-D4 is a brain positron emission tomography (PET) scanner composed of 4-layer depth-of-interaction (DOI) detectors with a large number of GSO crystals, which achieves both high spatial resolution and high scanner sensitivity. Since the sensitivity of each crystal element is highly dependent on DOI layer depth and incidental {gamma} ray energy, it is difficult to estimate normalization factors and scatter components with high statistical accuracy. In this work, we implemented a hybrid scatter correction method combined with component-based normalization, which estimates scatter components with a dual energy acquisition using a convolution subtraction-method for an estimation of trues from an upper energy window. In order to reduce statistical noise in sinograms, the implemented scheme uses the DOI compression (DOIC) method, that combines deep pairs of DOI layers into the nearest shallow pairs of DOI layers with natural detector samplings. Since the compressed data preserve the block detector configuration, as if the data are acquired using 'virtual' detectors with high {gamma}-ray stopping power, these correction methods can be applied directly to DOIC sinograms. The proposed method provides high-quality corrected images with low statistical noise, even for a multi-layer DOI-PET.
Sramek, Benjamin Koerner
The ability to deliver conformal dose distributions in radiation therapy through intensity modulation and the potential for tumor dose escalation to improve treatment outcome has necessitated an increase in localization accuracy of inter- and intra-fractional patient geometry. Megavoltage cone-beam CT imaging using the treatment beam and onboard electronic portal imaging device is one option currently being studied for implementation in image-guided radiation therapy. However, routine clinical use is predicated upon continued improvements in image quality and patient dose delivered during acquisition. The formal statement of hypothesis for this investigation was that the conformity of planned to delivered dose distributions in image-guided radiation therapy could be further enhanced through the application of kilovoltage scatter correction and intermediate view estimation techniques to megavoltage cone-beam CT imaging, and that normalized dose measurements could be acquired and inter-compared between multiple imaging geometries. The specific aims of this investigation were to: (1) incorporate the Feldkamp, Davis and Kress filtered backprojection algorithm into a program to reconstruct a voxelized linear attenuation coefficient dataset from a set of acquired megavoltage cone-beam CT projections, (2) characterize the effects on megavoltage cone-beam CT image quality resulting from the application of Intermediate View Interpolation and Intermediate View Reprojection techniques to limited-projection datasets, (3) incorporate the Scatter and Primary Estimation from Collimator Shadows (SPECS) algorithm into megavoltage cone-beam CT image reconstruction and determine the set of SPECS parameters which maximize image quality and quantitative accuracy, and (4) evaluate the normalized axial dose distributions received during megavoltage cone-beam CT image acquisition using radiochromic film and thermoluminescent dosimeter measurements in anthropomorphic pelvic and head and
International Nuclear Information System (INIS)
Shumeiko, N.M.; Timoshin, S.I.
1991-01-01
Compact formulae for a total 1-loop electromagnetic corrections, including the contribution of electromagnetic hadron effects to the deep inelastic scattering of polarized leptons on polarized nucleons in the quark-parton model have been obtained. The cases of longitudinal and transverse nucleon polarization are considered in detail. A thorough numerical calculation of corrections to cross sections and polarization asymmetries at muon (electron) energies over the range of 200-2000 GeV (10-16 GeV) has been made. It has been established that the contribution of corrections to the hadron current considerably affects the behaviour of longitudinal asymmetry. A satisfactory agreement is found between the model calculations of corrections to the lepton current and the phenomenological calculation results, which makes it possible to find the total 1-loop correction within the framework of a common approach. (Author)
Soft factors for double parton scattering at NNLO
Energy Technology Data Exchange (ETDEWEB)
Vladimirov, Alexey [Institut für Theoretische Physik, Universität Regensburg,D-93040 Regensburg (Germany)
2016-12-13
We show at NNLO that the soft factors for double parton scattering (DPS) for both integrated and unintegrated kinematics, can be presented entirely in the terms of the soft factor for single Drell-Yan process, i.e. the transverse momentum dependent (TMD) soft factor. Using the linearity of the logarithm of TMD soft factor in rapidity divergences, we decompose the DPS soft factor matrices into a product of matrices with rapidity divergences in given sectors, and thus, define individual double parton distributions at NNLO. The rapidity anomalous dimension matrices for double parton distributions are presented in the terms of TMD rapidity anomalous dimension. The analysis is done using the generating function approach to web diagrams. Significant part of the result is obtained from the symmetry properties of web diagrams without referring to explicit expressions or a particular rapidity regularization scheme. Additionally, we present NNLO expression for the web diagram generating function for Wilson lines with two light-like directions.
Raylman, R. R.; Majewski, S.; Wojcik, R.; Weisenberger, A. G.; Kross, B.; Popov, V.
2001-06-01
Positron emission mammography (PEM) has begun to show promise as an effective method for the detection of breast lesions. Due to its utilization of tumor-avid radiopharmaceuticals labeled with positron-emitting radionuclides, this technique may be especially useful in imaging of women with radiodense or fibrocystic breasts. While the use of these radiotracers affords PEM unique capabilities, it also introduces some limitations. Specifically, acceptance of accidental and Compton-scattered coincidence events can decrease lesion detectability. The authors studied the effect of accidental coincidence events on PEM images produced by the presence of /sup 18/F-Fluorodeoxyglucose in the organs of a subject using an anthropomorphic phantom. A delayed-coincidence technique was tested as a method for correcting PEM images for the occurrence of accidental events. Also, a Compton scatter correction algorithm designed specifically for PEM was developed and tested using a compressed breast phantom. Finally, the effect of object size on image counts and a correction for this effect were explored. The imager used in this study consisted of two PEM detector heads mounted 20 cm apart on a Lorad biopsy apparatus. The results demonstrated that a majority of the accidental coincidence events (/spl sim/80%) detected by this system were produced by radiotracer uptake in the adipose and muscle tissue of the torso. The presence of accidental coincidence events was shown to reduce lesion detectability. Much of this effect was eliminated by correction of the images utilizing estimates of accidental-coincidence contamination acquired with delayed coincidence circuitry built into the PEM system. The Compton scatter fraction for this system was /spl sim/14%. Utilization of a new scatter correction algorithm reduced the scatter fraction to /spl sim/1.5%. Finally, reduction of count recovery due to object size was measured and a correction to the data applied. Application of correction techniques
Attenuation correction factors for cylindrical, disc and box geometry
International Nuclear Information System (INIS)
Agarwal, Chhavi; Poi, Sanhita; Mhatre, Amol; Goswami, A.; Gathibandhe, M.
2009-01-01
In the present study, attenuation correction factors have been experimentally determined for samples having cylindrical, disc and box geometry and compared with the attenuation correction factors calculated by Hybrid Monte Carlo (HMC) method [ C. Agarwal, S. Poi, A. Goswami, M. Gathibandhe, R.A. Agrawal, Nucl. Instr. and. Meth. A 597 (2008) 198] and with the near-field and far-field formulations available in literature. It has been observed that the near-field formulae, although said to be applicable at close sample-detector geometry, does not work at very close sample-detector configuration. The advantage of the HMC method is that it is found to be valid for all sample-detector geometries.
International Nuclear Information System (INIS)
Afari, F.
2011-01-01
The use of equivalent squares is of great value and importance when determining output and depth dose data for rectangular fields. The variation with field shape of collimator scatter factors (S c ), phantom scatter factors (S c,p ) were studied using measurements on GWGP 80 cobalt - 60 teletherapy machine at the National Centre of Radiotherapy and Nuclear Medicine in the Korle-Bu Teaching Hospital. Measurements of the collimator scatter factors (S c ), phantom scatter factors (S p ) and total scatter factors (S c, p) were made at the depth of 5 cm, 10 cm, 15 cm and 20 cm in full scatter water phantom for square field side and rectangular fields of varying dimensions. The measurements were done using the source - axis distance (Sad) technique. The values of total scatter factor (S c,p ), phantom scatter factor and collimator scatter factor (S c ) obtained were used to estimate equivalent squares for the rectangular fields at the various depths. The equivalent squares were computed using the method of interpolation which is based on the scatter analysis of these scatter factors and these estimated equivalent squares were then compared with equivalent squares were then compared with equivalent square fields from BJR (supplement 21) tables of equivalent squares. The research revealed that there were average deviation of 1.5% for smaller rectangular field sizes and 8.8% for elongated rectangular field sizes between the estimated square field sizes and the equivalent square field from BJR (supplement 21) Table of equivalent square fields. The 8.8% for the elongated rectangular fields is not accepted, though such fields are rarely used in our Hospitals. It was found that the values of the equivalent square at the various depth were very consistent and do not vary with reference depth. These findings confirm that the clinical use of the BJR (supplement 21) Table of equivalent squares for total scatter factors and phantom scatter related quantities of rectangular fields is
Correction factors for assessing immersion suits under harsh conditions.
Power, Jonathan; Tikuisis, Peter; Ré, António Simões; Barwood, Martin; Tipton, Michael
2016-03-01
Many immersion suit standards require testing of thermal protective properties in calm, circulating water while these suits are typically used in harsher environments where they often underperform. Yet it can be expensive and logistically challenging to test immersion suits in realistic conditions. The goal of this work was to develop a set of correction factors that would allow suits to be tested in calm water yet ensure they will offer sufficient protection in harsher conditions. Two immersion studies, one dry and the other with 500 mL of water within the suit, were conducted in wind and waves to measure the change in suit insulation. In both studies, wind and waves resulted in a significantly lower immersed insulation value compared to calm water. The minimum required thermal insulation for maintaining heat balance can be calculated for a given mean skin temperature, metabolic heat production, and water temperature. Combining the physiological limits of sustainable cold water immersion and actual suit insulation, correction factors can be deduced for harsh conditions compared to calm. The minimum in-situ suit insulation to maintain thermal balance is 1.553-0.0624·TW + 0.00018·TW(2) for a dry calm condition. Multiplicative correction factors to the above equation are 1.37, 1.25, and 1.72 for wind + waves, 500 mL suit wetness, and both combined, respectively. Calm water certification tests of suit insulation should meet or exceed the minimum in-situ requirements to maintain thermal balance, and correction factors should be applied for a more realistic determination of minimum insulation for harsh conditions. Crown Copyright © 2015. Published by Elsevier Ltd. All rights reserved.
DEFF Research Database (Denmark)
Slot Thing, Rune; Bernchou, Uffe; Mainegra-Hing, Ernesto
2013-01-01
Abstract Purpose. Cone beam computed tomography (CBCT) image quality is limited by scattered photons. Monte Carlo (MC) simulations provide the ability of predicting the patient-specific scatter contamination in clinical CBCT imaging. Lengthy simulations prevent MC-based scatter correction from...
International Nuclear Information System (INIS)
Adachi, Itaru; Doi, Kenji; Komori, Tsuyoshi; Hou, Nobuyoshi; Tabuchi, Koujirou; Matsui, Ritsuo; Sueyoshi, Kouzou; Utsunomiya, Keita; Narabayashi, Isamu
1998-01-01
This investigation was undertaken to study clinical usefulness of scatter and attenuation correction (SAC) of brain SPECT in infants to compare the standard reconstruction (STD). The brain SPECT was performed in 31 patients with 19 epilepsy, 5 cerebro-vascular disease, 2 brain tumor, 3 meningitis, 1 hydrocephalus and psychosis (mean age 5.0±4.9 years old). Many patients was necessary to be injected sedatives for restraining body motion after Technetium-99m hexamethylpropylene amine oxime ( 99m Tc-HMPAO) was injected at the convulsion or rest. Brain SPECT data were acquired with triple detector gamma camera (GCA-9300 Toshiba Japan). These data were reconstructed by filtered backprojection after the raw data were corrected by triple energy windows method of scatter correction and Chang filtered method of attenuation correction. The same data was reconstructed by filtered backprojection without these corrections. Both SAC and STD SPECT images were analyzed by the visual interpretation. The uptake ratio of cerebral basal nuclei was calculated by the counts of the thalamus or lenticular nuclei divided by the cortex. All images of SAC method were excellent than that of STD method. The thalamic uptake ratio in SAC method was higher than that of STD method (1.22±0.09>0.87±0.22 p 1.02±0.16 p<0.01). Transmission scan is the most suitable method of absorption correction. But the transmission scan is not adequate for examination of children, because this scan needs a lot of time and the infants are exposed by the line source radioisotope. It was concluded that these scatter and absorption corrections were most suitable method for brain SPECT in pediatrics. (author)
Energy Technology Data Exchange (ETDEWEB)
Adachi, Itaru; Doi, Kenji; Komori, Tsuyoshi; Hou, Nobuyoshi; Tabuchi, Koujirou; Matsui, Ritsuo; Sueyoshi, Kouzou; Utsunomiya, Keita; Narabayashi, Isamu [Osaka Medical Coll., Takatsuki (Japan)
1998-01-01
This investigation was undertaken to study clinical usefulness of scatter and attenuation correction (SAC) of brain SPECT in infants to compare the standard reconstruction (STD). The brain SPECT was performed in 31 patients with 19 epilepsy, 5 cerebro-vascular disease, 2 brain tumor, 3 meningitis, 1 hydrocephalus and psychosis (mean age 5.0{+-}4.9 years old). Many patients was necessary to be injected sedatives for restraining body motion after Technetium-99m hexamethylpropylene amine oxime ({sup 99m}Tc-HMPAO) was injected at the convulsion or rest. Brain SPECT data were acquired with triple detector gamma camera (GCA-9300 Toshiba Japan). These data were reconstructed by filtered backprojection after the raw data were corrected by triple energy windows method of scatter correction and Chang filtered method of attenuation correction. The same data was reconstructed by filtered backprojection without these corrections. Both SAC and STD SPECT images were analyzed by the visual interpretation. The uptake ratio of cerebral basal nuclei was calculated by the counts of the thalamus or lenticular nuclei divided by the cortex. All images of SAC method were excellent than that of STD method. The thalamic uptake ratio in SAC method was higher than that of STD method (1.22{+-}0.09>0.87{+-}0.22 p<0.01). The lenticular nuclear uptake ratio in SAC method was higher than that of STD method (1.26{+-}0.15>1.02{+-}0.16 p<0.01). Transmission scan is the most suitable method of absorption correction. But the transmission scan is not adequate for examination of children, because this scan needs a lot of time and the infants are exposed by the line source radioisotope. It was concluded that these scatter and absorption corrections were most suitable method for brain SPECT in pediatrics. (author)
Correction factors for photon spectrometry in nuclear parameters study
International Nuclear Information System (INIS)
Patrao, Karla Cristina de Souza
2004-10-01
The goal of this work was the determination, using metrologic severity, the factors of correction for coincidences XX, Xγ and γγ and the factors of transference of efficiency for use in gamma spectrometry. On this way, it was carried through by determination of nuclear parameters of a nuclide used in medicine diagnostic ( 201 Tl) and the standardization of two environmental samples, of regular and irregular geometry, proceeding from the residual (ashes and slag) from the nuclear industry. The results shows that this adopted methodology is valid, and it allows its application for many different nuclides, including complex decay schema nuclides, using only photons spectrometry techniques on semiconductor detectors. (author)
Parity-Violating Electron Deuteron Scattering and the Proton's Neutral Axial Vector Form Factor
International Nuclear Information System (INIS)
Ito, T.
2003-01-01
The authors report on a new measurement of the parity-violating asymmetry in quasielastic electron scattering from the deuteron at the backward angles at electron beam energy of 125 MeV [Q 2 =0.038 (GeV/c) 2 ]. This quantity provides a determination of the neutral weak axial vector form factor of the nucleon. In addition to the tree level amplitude associated with Z-exchange, the neutral weak axial vector form factor as measured in electron scattering can potentially receive large electroweak corrections, including the anapole moment, that are absent in neutrino scattering. The measured asymmetry A -3.51 ± 0.57 (stat) ± 0.58 (sys) ppm is consistent with theoretical predictions. We also report on updated results of the previous experiment at 200 MeV [Q 2 = 0.091 (GeV/c) 2 ] on a deuterium target. The updated results are also consistent with theoretical predictions on the neutral weal axial vector form factor
The effect of correction factors on internal bremsstrahlung spectrum
International Nuclear Information System (INIS)
Elias, M.M.
1985-01-01
Correction factors affecting the experimental spectrum of internal bremsstrahlung (IB) accompanying B-decay have been studied. A wide survey of previous experimental studies, show a discrepancy between experimental results and the available theories of IB production especially for heavy nuclides and forbidden B-transitions, with some agreement in bounded regions in IB spectrum. This is due to the uncertainity in evaluating correction factors and neglecting others. The experimental distributions of IB from 147 Pm, 204 Tl and 90 Y are measured by using two different crystals of NaI(Tl) scintillators (2''x2'' and 3'''x3'') with suitable geometrical arrangements designed to minimize the effect of several factors. The present study covers overlapped ranges of energy from 0.05 to 2.2 MeV to observe the effect of these factors along this wide region of energy. The experimental probabilities are compared with theories. It agrees fairly well with Lewis and Ford theory in case of 14 '7Pm, 204 Tl, and with detour transitions theory in the intermediate region of energies of IB spectrum from 90 Y. Deviation of our results from theories is less than all previous experimental work. 20 tabs.; 43 figs.; 84 refs.; 5 apps
Energy Technology Data Exchange (ETDEWEB)
Raymond Raylman; Stanislaw Majewski; Randolph Wojcik; Andrew Weisenberger; Brian Kross; Vladimir Popov
2001-06-01
Positron emission mammography (PEM) has begun to show promise as an effective method for the detection of breast lesions. Due to its utilization of tumor-avid radiopharmaceuticals labeled with positron-emitting radionuclides, this technique may be especially useful in imaging of women with radiodense or fibrocystic breasts. While the use of these radiotracers affords PEM unique capabilities, it also introduces some limitations. Specifically, acceptance of accidental and Compton-scattered coincidence events can decrease lesion detectability. The authors studied the effect of accidental coincidence events on PEM images produced by the presence of 18F-Fluorodeoxyglucose in the organs of a subject using an anthropomorphic phantom. A delayed-coincidence technique was tested as a method for correcting PEM images for the occurrence of accidental events. Also, a Compton scatter correction algorithm designed specifically for PEM was developed and tested using a compressed breast phantom.
International Nuclear Information System (INIS)
Raymond Raylman; Stanislaw Majewski; Randolph Wojcik; Andrew Weisenberger; Brian Kross; Vladimir Popov
2001-01-01
Positron emission mammography (PEM) has begun to show promise as an effective method for the detection of breast lesions. Due to its utilization of tumor-avid radiopharmaceuticals labeled with positron-emitting radionuclides, this technique may be especially useful in imaging of women with radiodense or fibrocystic breasts. While the use of these radiotracers affords PEM unique capabilities, it also introduces some limitations. Specifically, acceptance of accidental and Compton-scattered coincidence events can decrease lesion detectability. The authors studied the effect of accidental coincidence events on PEM images produced by the presence of 18F-Fluorodeoxyglucose in the organs of a subject using an anthropomorphic phantom. A delayed-coincidence technique was tested as a method for correcting PEM images for the occurrence of accidental events. Also, a Compton scatter correction algorithm designed specifically for PEM was developed and tested using a compressed breast phantom
Energy Technology Data Exchange (ETDEWEB)
Shi, L; Zhu, L [Georgia Institute of Technology, Atlanta, GA (Georgia); Vedantham, S; Karellas, A [University of Massachusetts Medical School, Worcester, MA (United States)
2016-06-15
Purpose: Scatter contamination is detrimental to image quality in dedicated cone-beam breast CT (CBBCT), resulting in cupping artifacts and loss of contrast in reconstructed images. Such effects impede visualization of breast lesions and the quantitative accuracy. Previously, we proposed a library-based software approach to suppress scatter on CBBCT images. In this work, we quantify the efficacy and stability of this approach using datasets from 15 human subjects. Methods: A pre-computed scatter library is generated using Monte Carlo simulations for semi-ellipsoid breast models and homogeneous fibroglandular/adipose tissue mixture encompassing the range reported in literature. Projection datasets from 15 human subjects that cover 95 percentile of breast dimensions and fibroglandular volume fraction were included in the analysis. Our investigations indicate that it is sufficient to consider the breast dimensions alone and variation in fibroglandular fraction does not significantly affect the scatter-to-primary ratio. The breast diameter is measured from a first-pass reconstruction; the appropriate scatter distribution is selected from the library; and, deformed by considering the discrepancy in total projection intensity between the clinical dataset and the simulated semi-ellipsoidal breast. The deformed scatter-distribution is subtracted from the measured projections for scatter correction. Spatial non-uniformity (SNU) and contrast-to-noise ratio (CNR) were used as quantitative metrics to evaluate the results. Results: On the 15 patient cases, our method reduced the overall image spatial non-uniformity (SNU) from 7.14%±2.94% (mean ± standard deviation) to 2.47%±0.68% in coronal view and from 10.14%±4.1% to 3.02% ±1.26% in sagittal view. The average contrast to noise ratio (CNR) improved by a factor of 1.49±0.40 in coronal view and by 2.12±1.54 in sagittal view. Conclusion: We demonstrate the robustness and effectiveness of a library-based scatter correction
Monte Carlo simulation and scatter correction of the GE Advance PET scanner with SimSET and Geant4
International Nuclear Information System (INIS)
Barret, Olivier; Carpenter, T Adrian; Clark, John C; Ansorge, Richard E; Fryer, Tim D
2005-01-01
For Monte Carlo simulations to be used as an alternative solution to perform scatter correction, accurate modelling of the scanner as well as speed is paramount. General-purpose Monte Carlo packages (Geant4, EGS, MCNP) allow a detailed description of the scanner but are not efficient at simulating voxel-based geometries (patient images). On the other hand, dedicated codes (SimSET, PETSIM) will perform well for voxel-based objects but will be poor in their capacity of simulating complex geometries such as a PET scanner. The approach adopted in this work was to couple a dedicated code (SimSET) with a general-purpose package (Geant4) to have the efficiency of the former and the capabilities of the latter. The combined SimSET+Geant4 code (SimG4) was assessed on the GE Advance PET scanner and compared to the use of SimSET only. A better description of the resolution and sensitivity of the scanner and of the scatter fraction was obtained with SimG4. The accuracy of scatter correction performed with SimG4 and SimSET was also assessed from data acquired with the 20 cm NEMA phantom. SimG4 was found to outperform SimSET and to give slightly better results than the GE scatter correction methods installed on the Advance scanner (curve fitting and scatter modelling for the 300-650 keV and 375-650 keV energy windows, respectively). In the presence of a hot source close to the edge of the field of view (as found in oxygen scans), the GE curve-fitting method was found to fail whereas SimG4 maintained its performance
International Nuclear Information System (INIS)
Narita, Y.; Eberl, S.; Bautovich, G.; Iida, H.; Hutton, B.F.; Braun, M.; Nakamura, T.
1996-01-01
Scatter correction is a prerequisite for quantitative SPECT, but potentially increases noise. Monte Carlo simulations (EGS4) and physical phantom measurements were used to compare accuracy and noise properties of two scatter correction techniques: the triple-energy window (TEW), and the transmission dependent convolution subtraction (TDCS) techniques. Two scatter functions were investigated for TDCS: (i) the originally proposed mono-exponential function (TDCS mono ) and (ii) an exponential plus Gaussian scatter function (TDCS Gauss ) demonstrated to be superior from our Monte Carlo simulations. Signal to noise ratio (S/N) and accuracy were investigated in cylindrical phantoms and a chest phantom. Results from each method were compared to the true primary counts (simulations), or known activity concentrations (phantom studies). 99m Tc was used in all cases. The optimized TDCS Gauss method overall performed best, with an accuracy of better than 4% for all simulations and physical phantom studies. Maximum errors for TEW and TDCS mono of -30 and -22%, respectively, were observed in the heart chamber of the simulated chest phantom. TEW had the worst S/N ratio of the three techniques. The S/N ratios of the two TDCS methods were similar and only slightly lower than those of simulated true primary data. Thus, accurate quantitation can be obtained with TDCS Gauss , with a relatively small reduction in S/N ratio. (author)
Energy Technology Data Exchange (ETDEWEB)
Bai, J. [Keio Univ., Tokyo (Japan). 21st Century Center of Excellence Program; Hashimoto, J.; Kubo, A. [Keio Univ., Tokyo (Japan). Dept. of Radiology; Ogawa, K. [Hosei Univ., Tokyo (Japan). Dept. of Electronic Informatics; Fukunaga, A.; Onozuka, S. [Keio Univ., Tokyo (Japan). Dept. of Neurosurgery
2007-07-01
The aim of this study was to evaluate the effect of scatter and attenuation correction in region of interest (ROI) analysis of brain perfusion single-photon emission tomography (SPECT), and to assess the influence of selecting the reference area on the calculation of lesion-to-reference count ratios. Patients, methods: Data were collected from a brain phantom and ten patients with unilateral internal carotid artery stenosis. A simultaneous emission and transmission scan was performed after injecting {sup 123}I-iodoamphetamine. We reconstructed three SPECT images from common projection data: with scatter correction and nonuniform attenuation correction, with scatter correction and uniform attenuation correction, and with uniform attenuation correction applied to data without scatter correction. Regional count ratios were calculated by using four different reference areas (contralateral intact side, ipsilateral cerebellum, whole brain and hemisphere). Results: Scatter correction improved the accuracy of measuring the count ratios in the phantom experiment. It also yielded marked difference in the count ratio in the clinical study when using the cerebellum, whole brain or hemisphere as the reference. Difference between nonuniform and uniform attenuation correction was not significant in the phantom and clinical studies except when the cerebellar reference was used. Calculation of the lesion-to-normal count ratios referring the same site in the contralateral hemisphere was not dependent on the use of scatter correction or transmission scan-based attenuation correction. Conclusion: Scatter correction was indispensable for accurate measurement in most of the ROI analyses. Nonuniform attenuation correction is not necessary when using the reference area other than the cerebellum. (orig.)
International Nuclear Information System (INIS)
Miller, C.; Filipow, L.; Jackson, S.; Riauka, T.
1996-01-01
A new method to correct for attenuation and the buildup of scatter in planar imaging quantification is presented. The method is based on the combined use of 3D density information provided by computed tomography to correct for attenuation and the application of Monte Carlo simulated buildup factors to correct for buildup in the projection pixels. CT and nuclear medicine images were obtained for a purpose-built nonhomogeneous phantom that models the human anatomy in the thoracic and abdominal regions. The CT transverse slices of the phantom were converted to a set of consecutive density maps. An algorithm was developed that projects the 3D information contained in the set of density maps to create opposing pairs of accurate 2D correction maps that were subsequently applied to planar images acquired from a dual-head gamma camera. A comparison of results obtained by the new method and the geometric mean approach based on published techniques is presented for some of the source arrangements used. Excellent results were obtained for various source - phantom configurations used to evaluate the method. Activity quantification of a line source at most locations in the nonhomogeneous phantom produced errors of less than 2%. Additionally, knowledge of the actual source depth is not required for accurate activity quantification. Quantification of volume sources placed in foam, Perspex and aluminium produced errors of less than 7% for the abdominal and thoracic configurations of the phantom. (author)
International Nuclear Information System (INIS)
Gu Yi; Xiong Shengqing; Zhou Jianxin; Fan Zhengguo; Ge Liangquan
2014-01-01
γ-ray released by the radon daughter has severe impact on airborne γ-ray spectrometry. The spectral-ratio method is one of the best mathematical methods for radon background deduction in airborne γ-ray spectrometry. In this paper, an advanced spectral-ratio method was proposed which deducts Compton scattering ray by the fast Fourier transform rather than tripping ratios, the relationship between survey height and correction coefficient of the advanced spectral-ratio radon background correction method was studied, the advanced spectral-ratio radon background correction mathematic model was established, and the ground saturation model calibrating technology for correction coefficient was proposed. As for the advanced spectral-ratio radon background correction method, its applicability and correction efficiency are improved, and the application cost is saved. Furthermore, it can prevent the physical meaning lost and avoid the possible errors caused by matrix computation and mathematical fitting based on spectrum shape which is applied in traditional correction coefficient. (authors)
The real corrections to the virtual photon impact factor
International Nuclear Information System (INIS)
Kyrieleis, A.
2003-10-01
We investigate the interaction of two virtual photons in the high energy limit of quantum chromodynamics (QCD). We are motivated by two closely linked problems: the calculation of the γ * γ * total cross section, σ γ * γ * , in the framework NLO BFKL and the NLO extensions to the colour dipole picture. We calculate the squared amplitude for the process γ * + q → qqg + q in arbitrary space-time dimensions for both longitudinal and transverse photon polarisation. In transverse configuration space the result is found to factorise supporting in this way the photon wave function interpretation. This is the first step in the calculation of the qqg Fock component of the γ * wave function. The squared vertex for the transition Reggeon-γ * → qqg is extracted in order to calculate the NLO real corrections to the γ * impact factor. Together with the virtual corrections this will allow for the calculation of σ γ * γ * . The infrared divergences are dimensionally regularised and their cancellation against those from the virtual corrections is performed. After the introduction of Feynman parameters in each Feynman diagram, part of the qqg phase space integration is performed analytically. The various divergences emerging due to the independent treatment of the diagrams are regularised. A procedure is worked out to obtain for each diagram a finite phase space integral. In the case of longitudinal photon polarisation, this method is implemented in a computer program and the full phase space integration is carried out numerically. First numerical tests of the calculation are successfully performed. (orig.)
International Nuclear Information System (INIS)
Torres-Espallardo, I; Spanoudaki, V; Ziegler, S I; Rafecas, M; McElroy, D P
2008-01-01
Random coincidences can contribute substantially to the background in positron emission tomography (PET). Several estimation methods are being used for correcting them. The goal of this study was to investigate the validity of techniques for random coincidence estimation, with various low-energy thresholds (LETs). Simulated singles list-mode data of the MADPET-II small animal PET scanner were used as input. The simulations have been performed using the GATE simulation toolkit. Several sources with different geometries have been employed. We evaluated the number of random events using three methods: delayed window (DW), singles rate (SR) and time histogram fitting (TH). Since the GATE simulations allow random and true coincidences to be distinguished, a comparison between the number of random coincidences estimated using the standard methods and the number obtained using GATE was performed. An overestimation in the number of random events was observed using the DW and SR methods. This overestimation decreases for LETs higher than 255 keV. It is additionally reduced when the single events which have undergone a Compton interaction in crystals before being detected are removed from the data. These two observations lead us to infer that the overestimation is due to inter-crystal scatter. The effect of this mismatch in the reconstructed images is important for quantification because it leads to an underestimation of activity. This was shown using a hot-cold-background source with 3.7 MBq total activity in the background region and a 1.59 MBq total activity in the hot region. For both 200 keV and 400 keV LET, an overestimation of random coincidences for the DW and SR methods was observed, resulting in approximately 1.5% or more (at 200 keV LET: 1.7% for DW and 7% for SR) and less than 1% (at 400 keV LET: both methods) underestimation of activity within the background region. In almost all cases, images obtained by compensating for random events in the reconstruction
International Nuclear Information System (INIS)
Nakajima, Kenichi; Matsudaira, Masamichi; Yamada, Masato; Taki, Junichi; Tonami, Norihisa; Hisada, Kinichi
1995-01-01
Triple-energy window (TEW) method is a simple and practical approach for correcting Compton scatter in single-photon emission tracer studies. The fraction of scatter correction, with a point source or 30 ml-syringe placed under the camera, was measured by the TEW method. The scatter fraction was 55% for 201 Tl, 29% for 99m Tc and 57% for 123 I. Composite energy spectra were generated and separated by the TEW method. Combination of 99m Tc and 201 Tl was well separated, and 201 Tl and 123 I were separated within an error of 10%; whereas asymmetric photopeak energy window was necessary for separating 123 I and 99m Tc. By applying this method to myocardial SPECT study, the effect of scatter elimination was investigated in each myocardial wall by polar map and profile curve analysis. The effect of scatter was higher in the septum and the inferior wall. The count ratio relative to the anterior wall including scatter was 9% higher in 123 I, 7-8% higher in 99m Tc and 6% higher in 201 Tl. Apparent count loss after scatter correction was 30% for 123 I, 13% for 99m Tc and 38% for 201 Tl. Image contrast, as defined myocardium-to-left ventricular cavity count ratio, improved by scatter correction. Since the influence of Compton scatter was significant in cardiac planar and SPECT studies; the degree of scatter fraction should be kept in mind both in quantification and visual interpretation. (author)
Steponavičius, Raimundas; Thennadil, Suresh N
2013-05-01
Sample-to-sample photon path length variations that arise due to multiple scattering can be removed by decoupling absorption and scattering effects by using the radiative transfer theory, with a suitable set of measurements. For samples where particles both scatter and absorb light, the extracted bulk absorption spectrum is not completely free from nonlinear particle effects, since it is related to the absorption cross-section of particles that changes nonlinearly with particle size and shape. For the quantitative analysis of absorbing-only (i.e., nonscattering) species present in a matrix that contains a particulate species that absorbs and scatters light, a method to eliminate particle effects completely is proposed here, which utilizes the particle size information contained in the bulk scattering coefficient extracted by using the Mie theory to carry out an additional correction step to remove particle effects from bulk absorption spectra. This should result in spectra that are equivalent to spectra collected with only the liquid species in the mixture. Such an approach has the potential to significantly reduce the number of calibration samples as well as improve calibration performance. The proposed method was tested with both simulated and experimental data from a four-component model system.
Directory of Open Access Journals (Sweden)
I. S. Timchenko
2015-07-01
Full Text Available For calculating the radiative tails in the spectra of inelastic electron scattering by nuclei, the approximation, namely, the equivalent radiator method (ERM, is used. However, the applicability of this method for evaluating the radiative tail from the elastic scattering peak has been little investigated, and therefore, it has become the subject of the present study for the case of light nuclei. As a result, spectral regions were found, where a significant discrepancy between the ERM calculation and the exact-formula calculation was observed. A link was established between this phenomenon and the diffraction minimum of the squared form-factor of the nuclear ground state. Varieties of calculations were carried out for different kinematics of electron scattering by nuclei. The analysis of the calculation results has shown the conditions, at which the equivalent radiator method can be applied for adequately evaluating the radiative tail of the elastic scattering peak.
Zaidi, H; Slosman, D O
2003-01-01
Reliable attenuation correction represents an essential component of the long chain of modules required for the reconstruction of artifact-free, quantitative brain positron emission tomography (PET) images. In this work we demonstrate the proof of principle of segmented magnetic resonance imaging (MRI)-guided attenuation and scatter corrections in 3D brain PET. We have developed a method for attenuation correction based on registered T1-weighted MRI, eliminating the need of an additional transmission (TX) scan. The MR images were realigned to preliminary reconstructions of PET data using an automatic algorithm and then segmented by means of a fuzzy clustering technique which identifies tissues of significantly different density and composition. The voxels belonging to different regions were classified into air, skull, brain tissue and nasal sinuses. These voxels were then assigned theoretical tissue-dependent attenuation coefficients as reported in the ICRU 44 report followed by Gaussian smoothing and additio...
Window selection for dual photopeak window scatter correction in Tc-99m imaging
International Nuclear Information System (INIS)
Vries, D.J. de; King, M.A.
1994-01-01
The width and placement of the windows for the dual photopeak window (DPW) scatter subtraction method for Tc-99m imaging is investigated in order to obtain a method that is stable on a multihead detector system for single photon emission computed tomography (SPECT) and is capable of providing a good scatter estimate for extended objects. For various window pairs, stability and noise were examined with experiments using a SPECT system, while Monte Carlo simulations were used to predict the accuracy of scatter estimates for a variety of objects and to guide the development of regression relations for various window pairs. The DPW method that resulted from this study was implemented with a symmetric 20% photopeak window composed of a 15% asymmetric photopeak window and a 5% lower window abutted at 7 keV below the peak. A power function regression was used to relate the scatter-to-total ratio to the lower window-to-total ratio at each pixel, from which an estimated scatter image was calculated. DPW demonstrated good stability, achieved by abutting the two windows away from the peak. Performance was assessed and compared with Compton window subtraction (CWS). For simulated extended objects, DPW generally produced a less biased scatter estimate than the commonly used CWS method with k = 0.5. In acquisitions of a clinical SPECT phantom, contrast recovery was comparable for both DPW and CWS; however, DPW showed greater visual contrast in clinical SPECT bone studies
Photon energy-fluence correction factor in low energy brachytherapy
International Nuclear Information System (INIS)
Antunes, Paula C.G.; Yoriyaz, Hélio; Vijande, Javier; Giménez-Alventosa, Vicent; Ballester, Facundo
2017-01-01
The AAPM TG-43 brachytherapy dosimetry formalism has become a standard for brachytherapy dosimetry worldwide; it implicitly assumes that charged-particle equilibrium (CPE) exists for the determination of absorbed dose to water at different locations. At the time of relating dose to tissue and dose to water, or vice versa, it is usually assumed that the photon fluence in water and in tissues are practically identical, so that the absorbed dose in the two media can be related by their ratio of mass energy-absorption coefficients. The purpose of this work is to study the influence of photon energy-fluence in different media and to evaluate a proposal for energy-fluence correction factors for the conversion between dose-to-tissue (D tis ) and dose-to-water (D w ). State-of-the art Monte Carlo (MC) calculations are used to score photon fluence differential in energy in water and in various human tissues (muscle, adipose and bone) in two different codes, MCNP and PENELOPE, which in all cases include a realistic modeling of the 125 I low-energy brachytherapy seed in order to benchmark the formalism proposed. A correction is introduced that is based on the ratio of the water-to-tissue photon energy-fluences using the large-cavity theory. In this work, an efficient way to correlate absorbed dose to water and absorbed dose to tissue in brachytherapy calculations at clinically relevant distances for low-energy photon emitting seed is proposed. The energy-fluence based corrections given in this work are able to correlate absorbed dose to tissue and absorbed dose to water with an accuracy better than 0.5% in the most critical cases. (author)
Photon energy-fluence correction factor in low energy brachytherapy
Energy Technology Data Exchange (ETDEWEB)
Antunes, Paula C.G.; Yoriyaz, Hélio [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil); Vijande, Javier; Giménez-Alventosa, Vicent; Ballester, Facundo, E-mail: pacrisguian@gmail.com [Department of Atomic, Molecular, and Nuclear Physics and Instituto de Física Corpuscular (UV-CSIC), University of Valencia (Spain)
2017-07-01
The AAPM TG-43 brachytherapy dosimetry formalism has become a standard for brachytherapy dosimetry worldwide; it implicitly assumes that charged-particle equilibrium (CPE) exists for the determination of absorbed dose to water at different locations. At the time of relating dose to tissue and dose to water, or vice versa, it is usually assumed that the photon fluence in water and in tissues are practically identical, so that the absorbed dose in the two media can be related by their ratio of mass energy-absorption coefficients. The purpose of this work is to study the influence of photon energy-fluence in different media and to evaluate a proposal for energy-fluence correction factors for the conversion between dose-to-tissue (D{sub tis}) and dose-to-water (D{sub w}). State-of-the art Monte Carlo (MC) calculations are used to score photon fluence differential in energy in water and in various human tissues (muscle, adipose and bone) in two different codes, MCNP and PENELOPE, which in all cases include a realistic modeling of the {sup 125}I low-energy brachytherapy seed in order to benchmark the formalism proposed. A correction is introduced that is based on the ratio of the water-to-tissue photon energy-fluences using the large-cavity theory. In this work, an efficient way to correlate absorbed dose to water and absorbed dose to tissue in brachytherapy calculations at clinically relevant distances for low-energy photon emitting seed is proposed. The energy-fluence based corrections given in this work are able to correlate absorbed dose to tissue and absorbed dose to water with an accuracy better than 0.5% in the most critical cases. (author)
International Nuclear Information System (INIS)
Broome, J.
1965-11-01
The programme SCATTER is a KDF9 programme in the Egtran dialect of Fortran to generate normalized angular distributions for elastically scattered neutrons from data input as the coefficients of a Legendre polynomial series, or from differential cross-section data. Also, differential cross-section data may be analysed to produce Legendre polynomial coefficients. Output on cards punched in the format of the U.K. A. E. A. Nuclear Data Library is optional. (author)
Theoretical evaluation of self-shielding factors due to scattering resonances in foils
International Nuclear Information System (INIS)
Selander, W.N.
1960-06-01
A semi-analytical method is given for evaluating self-shielding factors for activation measurements which use thin foils having neutron scattering resonances. The energy loss by scattering in the foil is taken into account. The energy-dependent neutron angular distribution is expanded as a double series, the coefficients of which are (energy dependent) solutions of an infinite set of coupled integral equations. These are truncated in some suitable manner and solved numerically. The leading term of the series is proportional to the average, or effective flux in the activation sample. The product of this terra and the neutron capture cross-section is integrated numerically over the resonance to give the resonance self-shielding correction. Figure 4 shows resonance self-shielding factors derived in this mariner for the 132ev resonance in Co-59 and figure 5 shows similar results for the two Mn-55 resonances at 337ev and 1080ev. Self-shielding factors for 1/v capture are not significantly different from unity. (author)
Wide angle Compton scattering within the SCET factorization framework
International Nuclear Information System (INIS)
Kivel, N.
2016-01-01
Existing data for the electromagnetic proton form factors and for the cross section of the wide angle Compton scattering (WACS) show that the hard two-gluon exchange mechanism (collinear factorization) is still not applicable in the kinematical region where Mandelstam variables s ∼ -t ∼ -u are about a few GeV 2 . On the other hand these observables can be described in phenomenological models where spectator quarks are soft which assumes a large contribution due to the soft-overlap mechanism. It turns out that the simple QCD factorization picture is not complete and must also include the soft-overlap contribution which can be described as a certain matrix element in the soft collinear effective theory (SCET). Then the leading power contribution to WACS amplitude is described as a sum of the hard- and soft-spectator contributions. The existing experimental data allows one to check certain conclusions based on the assumption about dominant role of the soft-spectator mechanism. (author)
Magota, Keiichi; Shiga, Tohru; Asano, Yukari; Shinyama, Daiki; Ye, Jinghan; Perkins, Amy E; Maniawski, Piotr J; Toyonaga, Takuya; Kobayashi, Kentaro; Hirata, Kenji; Katoh, Chietsugu; Hattori, Naoya; Tamaki, Nagara
2017-12-01
In 3-dimensional PET/CT imaging of the brain with 15 O-gas inhalation, high radioactivity in the face mask creates cold artifacts and affects the quantitative accuracy when scatter is corrected by conventional methods (e.g., single-scatter simulation [SSS] with tail-fitting scaling [TFS-SSS]). Here we examined the validity of a newly developed scatter-correction method that combines SSS with a scaling factor calculated by Monte Carlo simulation (MCS-SSS). Methods: We performed phantom experiments and patient studies. In the phantom experiments, a plastic bottle simulating a face mask was attached to a cylindric phantom simulating the brain. The cylindric phantom was filled with 18 F-FDG solution (3.8-7.0 kBq/mL). The bottle was filled with nonradioactive air or various levels of 18 F-FDG (0-170 kBq/mL). Images were corrected either by TFS-SSS or MCS-SSS using the CT data of the bottle filled with nonradioactive air. We compared the image activity concentration in the cylindric phantom with the true activity concentration. We also performed 15 O-gas brain PET based on the steady-state method on patients with cerebrovascular disease to obtain quantitative images of cerebral blood flow and oxygen metabolism. Results: In the phantom experiments, a cold artifact was observed immediately next to the bottle on TFS-SSS images, where the image activity concentrations in the cylindric phantom were underestimated by 18%, 36%, and 70% at the bottle radioactivity levels of 2.4, 5.1, and 9.7 kBq/mL, respectively. At higher bottle radioactivity, the image activity concentrations in the cylindric phantom were greater than 98% underestimated. For the MCS-SSS, in contrast, the error was within 5% at each bottle radioactivity level, although the image generated slight high-activity artifacts around the bottle when the bottle contained significantly high radioactivity. In the patient imaging with 15 O 2 and C 15 O 2 inhalation, cold artifacts were observed on TFS-SSS images, whereas
International Nuclear Information System (INIS)
Lehtinen, Ossi; Geiger, Dorin; Lee, Zhongbo; Whitwick, Michael Brian; Chen, Ming-Wei; Kis, Andras; Kaiser, Ute
2015-01-01
Here, we present a numerical post-processing method for removing the effect of anti-symmetric residual aberrations in high-resolution transmission electron microscopy (HRTEM) images of weakly scattering 2D-objects. The method is based on applying the same aberrations with the opposite phase to the Fourier transform of the recorded image intensity and subsequently inverting the Fourier transform. We present the theoretical justification of the method, and its verification based on simulated images in the case of low-order anti-symmetric aberrations. Ultimately the method is applied to experimental hardware aberration-corrected HRTEM images of single-layer graphene and MoSe 2 resulting in images with strongly reduced residual low-order aberrations, and consequently improved interpretability. Alternatively, this method can be used to estimate by trial and error the residual anti-symmetric aberrations in HRTEM images of weakly scattering objects
N3LO corrections to jet production in deep inelastic scattering using the Projection-to-Born method
Currie, J.; Gehrmann, T.; Glover, E. W. N.; Huss, A.; Niehues, J.; Vogt, A.
2018-05-01
Computations of higher-order QCD corrections for processes with exclusive final states require a subtraction method for real-radiation contributions. We present the first-ever generalisation of a subtraction method for third-order (N3LO) QCD corrections. The Projection-to-Born method is used to combine inclusive N3LO coefficient functions with an exclusive second-order (NNLO) calculation for a final state with an extra jet. The input requirements, advantages, and potential applications of the method are discussed, and validations at lower orders are performed. As a test case, we compute the N3LO corrections to kinematical distributions and production rates for single-jet production in deep inelastic scattering in the laboratory frame, and compare them with data from the ZEUS experiment at HERA. The corrections are small in the central rapidity region, where they stabilize the predictions to sub per-cent level. The corrections increase substantially towards forward rapidity where large logarithmic effects are expected, thereby yielding an improved description of the data in this region.
NNLO QCD corrections to jet production at hadron colliders from gluon scattering
International Nuclear Information System (INIS)
Currie, James; Ridder, Aude Gehrmann-De; Glover, E.W.N.; Pires, João
2014-01-01
We present the next-to-next-to-leading order (NNLO) QCD corrections to dijet production in the purely gluonic channel retaining the full dependence on the number of colours. The sub-leading colour contribution in this channel first appears at NNLO and increases the NNLO correction by around 10% and exhibits a p T dependence, rising from 8% at low p T to 15% at high p T . The present calculation demonstrates the utility of the antenna subtraction method for computing the full colour NNLO corrections to dijet production at the Large Hadron Collider
International Nuclear Information System (INIS)
Arrington, John; Sick, Ingo
2007-01-01
The extraction of the strangeness form factors from parity-violating elastic electron-proton scattering is sensitive to the electromagnetic form factors at low Q 2 . We provide parametrizations for the form factors and uncertainties, including the effects of two-photon exchange corrections to the extracted electromagnetic form factors. We study effect of the correlations between different form factors, in particular as they impact the parity-violating asymmetry and the extraction of the strangeness form factors. We provide a prescription to extract the strangeness form factors from the asymmetry that provides an excellent approximation of the full two-photon correction. The corrected form factors are also appropriate as input for other low-Q analyses, although the effects of correlations and two-photon exchange corrections may be different
Calculation of beam quality correction factor using Monte Carlo simulation
International Nuclear Information System (INIS)
Kawachi, T.; Saitoh, H.; Myojoyama, A.; Katayose, T.; Kojima, T.; Fukuda, K.; Inoue, M.
2005-01-01
In recent years, a number of the CyberKnife systems (Accuray C., U.S.) have been increasing significantly. However, the CyberKnife has unique treatment head structure and beam collimating system. Therefore, the global standard protocols can not be adopted for absolute absorbed dose dosimetry in CyberKnife beam. In this work, the energy spectrum of photon and electron from CyberKnife treatment head at 80 cm SSD and several depths in water are simulated with conscientious geometry using by the EGS Monte Carlo method. Furthermore, for calculation of the beam quality correction factor k Q , the mean restricted mass stopping power and the mass energy absorption coefficient of air, water and several chamber wall and waterproofing sleeve materials are calculated. As a result, the factors k Q CyberKnife beam for several ionization chambers are determined. And the relationship between the beam quality index PDD(10) x in CyberKnife beam and k Q is described in this report. (author)
International Nuclear Information System (INIS)
Poux, J.P.
1972-06-01
In this research thesis, after a recall of processes of elastic scattering of positrons on electrons (kinematics and cross section), and of involved radiative corrections, the author describes the experimental installation (positron beam, ionization chamber, targets, spectrometer, electronic logics associated with the counter telescope) which has been used to measure the differential cross section of recoil electrons, and the methods which have been used. In a third part, the author reports the calculation of corrections and the obtained spectra. In the next part, the author reports the interpretation of results and their comparison with the experiment performed by Browman, Grossetete and Yount. The author shows that both experiments are complementary to each other, and are in agreement with the calculation performed by Yennie, Hearn and Kuo
International Nuclear Information System (INIS)
Bergstrom, P
2016-01-01
Purpose: The National Institute of Standards and Technology (NIST) uses 3 free-air chambers to establish primary standards for radiation dosimetry at x-ray energies. For medium-energy × rays, the Ritz free-air chamber is the main measurement device. In order to convert the charge or current collected by the chamber to the radiation quantities air kerma or air kerma rate, a number of correction factors specific to the chamber must be applied. Methods: We used the Monte Carlo codes EGSnrc and PENELOPE. Results: Among these correction factors are the diaphragm correction (which accounts for interactions of photons from the x-ray source in the beam-defining diaphragm of the chamber), the scatter correction (which accounts for the effects of photons scattered out of the primary beam), the electron-loss correction (which accounts for electrons that only partially expend their energy in the collection region), the fluorescence correction (which accounts for ionization due to reabsorption ffluorescence photons and the bremsstrahlung correction (which accounts for the reabsorption of bremsstrahlung photons). We have computed monoenergetic corrections for the NIST Ritz chamber for the 1 cm, 3 cm and 7 cm collection plates. Conclusion: We find good agreement with other’s results for the 7 cm plate. The data used to obtain these correction factors will be used to establish air kerma and it’s uncertainty in the standard NIST x-ray beams.
Energy Technology Data Exchange (ETDEWEB)
Bergstrom, P [National Institute of Standards and Technology, 100 Bureau Drive, Gaithersburg, MD 20899 (United States)
2016-06-15
Purpose: The National Institute of Standards and Technology (NIST) uses 3 free-air chambers to establish primary standards for radiation dosimetry at x-ray energies. For medium-energy × rays, the Ritz free-air chamber is the main measurement device. In order to convert the charge or current collected by the chamber to the radiation quantities air kerma or air kerma rate, a number of correction factors specific to the chamber must be applied. Methods: We used the Monte Carlo codes EGSnrc and PENELOPE. Results: Among these correction factors are the diaphragm correction (which accounts for interactions of photons from the x-ray source in the beam-defining diaphragm of the chamber), the scatter correction (which accounts for the effects of photons scattered out of the primary beam), the electron-loss correction (which accounts for electrons that only partially expend their energy in the collection region), the fluorescence correction (which accounts for ionization due to reabsorption ffluorescence photons and the bremsstrahlung correction (which accounts for the reabsorption of bremsstrahlung photons). We have computed monoenergetic corrections for the NIST Ritz chamber for the 1 cm, 3 cm and 7 cm collection plates. Conclusion: We find good agreement with other’s results for the 7 cm plate. The data used to obtain these correction factors will be used to establish air kerma and it’s uncertainty in the standard NIST x-ray beams.
International Nuclear Information System (INIS)
Sorriaux, J; Paganetti, H; Testa, M; Giantsoudi, D; Schuemann, J; Bertrand, D; Orban de Xivry, J.; Lee, J; Palmans, H; Vynckier, S; Sterpin, E
2014-01-01
Purpose: In current practice, most proton therapy centers apply IAEA TRS-398 reference dosimetry protocol. Quality correction factors (kQ) take into account in the dose determination process the differences in beam qualities used for calibration unit and for treatment unit. These quality correction factors are valid for specific reference conditions. TRS-398 reference conditions should be achievable in both scattered proton beams (i.e. DS) and scanned proton beams (i.e. PBS). However, it is not a priori clear if TRS-398 kQ data, which are based on Monte Carlo (MC) calculations in scattered beams, can be used for scanned beams. Using TOPAS-Geant4 MC simulations, the study aims to determine whether broad beam quality correction factors calculated in TRS-398 can be directly applied to PBS delivery modality. Methods: As reference conditions, we consider a 10×10×10 cm 3 homogeneous dose distribution delivered by PBS system in a water phantom (32/10 cm range/modulation) and an air cavity placed at the center of the spread-out-Bragg-peak. In order to isolate beam differences, a hypothetical broad beam is simulated. This hypothetical beam reproduces exactly the same range modulation, and uses the same energy layers than the PBS field. Ion chamber responses are computed for the PBS and hypothetical beams and then compared. Results: For an air cavity of 2×2×0.2 cm 3 , the ratio of ion chamber responses for the PBS and hypothetical beam qualities is 0.9991 ± 0.0016. Conclusion: Quality correction factors are insensitive to the delivery pattern of the beam (broad beam or PBS), as long as similar dose distributions are achieved. This investigation, for an air cavity, suggests that broad beam quality correction factors published in TRS-398 can be applied for scanned beams. J. Sorriaux is financially supported by a public-private partnership involving the company Ion Beam Applications (IBA)
Sayed, Sadeed Bin; Uysal, Ismail Enes; Bagci, Hakan; Ulku, H. Arda
2018-01-01
Quantum tunneling is observed between two nanostructures that are separated by a sub-nanometer gap. Electrons “jumping” from one structure to another create an additional current path. An auxiliary tunnel is introduced between the two structures as a support for this so that a classical electromagnetic solver can account for the effects of quantum tunneling. The dispersive permittivity of the tunnel is represented by a Drude model, whose parameters are obtained from the electron tunneling probability. The transient scattering from the connected nanostructures (i.e., nanostructures plus auxiliary tunnel) is analyzed using a time domain volume integral equation solver. Numerical results demonstrating the effect of quantum tunneling on the scattered fields are provided.
Energy Technology Data Exchange (ETDEWEB)
Rinkel, J.; Dinten, J.M. [CEA Grenoble (DTBS/STD), Lab. d' Electronique et de Technologie de l' Informatique, LETI, 38 (France); Esteve, F. [European Synchrotron Radiation Facility (ESRF), 38 - Grenoble (France)
2004-07-01
Purpose: Cone beam CT (CBCT) enables three-dimensional imaging with isotropic resolution. X-ray scatter estimation is a big challenge for quantitative CBCT imaging of thorax: scatter level is significantly higher on cone beam systems compared to collimated fan beam systems. The effects of this scattered radiation are cupping artefacts, streaks, and quantification inaccuracies. The beam stops conventional scatter estimation approach can be used for CBCT but leads to a significant increase in terms of dose and acquisition time. At CEA-LETI has been developed an original scatter management process without supplementary acquisition. Methods and Materials: This Analytical Plus Indexing-based method (API) of scatter correction in CBCT is based on scatter calibration through offline acquisitions with beam stops on lucite plates, combined to an analytical transformation issued from physical equations. This approach has been applied with success in bone densitometry and mammography. To evaluate this method in CBCT, acquisitions from a thorax phantom with and without beam stops have been performed. To compare different scatter correction approaches, Feldkamp algorithm has been applied on rough data corrected from scatter by API and by beam stops approaches. Results: The API method provides results in good agreement with the beam stops array approach, suppressing cupping artefact. Otherwise influence of the scatter correction method on the noise in the reconstructed images has been evaluated. Conclusion: The results indicate that the API method is effective for quantitative CBCT imaging of thorax. Compared to a beam stops array method it needs a lower x-ray dose and shortens acquisition time. (authors)
International Nuclear Information System (INIS)
Maltman, K.
1998-01-01
Using the framework of effective chiral Lagrangians, we show that, in order to correctly implement electromagnetism (EM), as generated from the Standard Model, into effective hadronic theories (such as meson-exchange models) it is insufficient to consider only graphs in the low-energy effective theory containing explicit photon lines. The Standard Model requires the presence of contact interactions in the effective theory which are electromagnetic in origin, but which involve no photons in the effective theory. We illustrate the problems which can result from a ''standard'' EM subtraction: i.e., from assuming that removing all contributions in the effective theory generated by graphs with explicit photon lines fully removes EM effects, by considering the case of the s-wave ππ scattering lengths. In this case it is shown that such a subtraction procedure would lead to the incorrect conclusion that the strong interaction isospin-breaking contributions to these quantities were large when, in fact, they are known to vanish at leading order in m d -m u . The leading EM contact corrections for the channels employed in the extraction of the I=0,2 s-wave ππ scattering lengths from experiment are also evaluated. (orig.)
International Nuclear Information System (INIS)
Ouyang, Luo; Lee, Huichen Pam; Wang, Jing
2015-01-01
Purpose: To evaluate a moving blocker-based approach in estimating and correcting megavoltage (MV) and kilovoltage (kV) scatter contamination in kV cone-beam computed tomography (CBCT) acquired during volumetric modulated arc therapy (VMAT). Methods and materials: During the concurrent CBCT/VMAT acquisition, a physical attenuator (i.e., “blocker”) consisting of equally spaced lead strips was mounted and moved constantly between the CBCT source and patient. Both kV and MV scatter signals were estimated from the blocked region of the imaging panel, and interpolated into the unblocked region. A scatter corrected CBCT was then reconstructed from the unblocked projections after scatter subtraction using an iterative image reconstruction algorithm based on constraint optimization. Experimental studies were performed on a Catphan® phantom and an anthropomorphic pelvis phantom to demonstrate the feasibility of using a moving blocker for kV–MV scatter correction. Results: Scatter induced cupping artifacts were substantially reduced in the moving blocker corrected CBCT images. Quantitatively, the root mean square error of Hounsfield units (HU) in seven density inserts of the Catphan phantom was reduced from 395 to 40. Conclusions: The proposed moving blocker strategy greatly improves the image quality of CBCT acquired with concurrent VMAT by reducing the kV–MV scatter induced HU inaccuracy and cupping artifacts
Energy Technology Data Exchange (ETDEWEB)
Ferguson, S; Ahmad, S; Chen, Y; Ferreira, C; Islam, M; Lau, A; Jin, H [University of Oklahoma Health Sciences Center, Oklahoma City, OK (United States); Keeling, V [Carti, Inc., Little Rock, AR (United States)
2016-06-15
Purpose: To commission and investigate the accuracy of an output (cGy/MU) prediction model for a compact passively scattered proton therapy system. Methods: A previously published output prediction model (Sahoo et al, Med Phys, 35, 5088–5097, 2008) was commissioned for our Mevion S250 proton therapy system. This model is a correction-based model that multiplies correction factors (d/MUwnc=ROFxSOBPF xRSFxSOBPOCFxOCRxFSFxISF). These factors accounted for changes in output due to options (12 large, 5 deep, and 7 small), modulation width M, range R, off-center, off-axis, field-size, and off-isocenter. In this study, the model was modified to ROFxSOBPFxRSFxOCRxFSFxISF-OCFxGACF by merging SOBPOCF and ISF for simplicity and introducing a gantry angle correction factor (GACF). To commission the model, outputs over 1,000 data points were taken at the time of the system commissioning. The output was predicted by interpolation (1D for SOBPF, FSF, and GACF; 2D for RSF and OCR) with inverse-square calculation (ISF-OCR). The outputs of 273 combinations of R and M covering total 24 options were measured to test the model. To minimize fluence perturbation, scattered dose from range compensator and patient was not considered. The percent differences between the predicted (P) and measured (M) outputs were calculated to test the prediction accuracy ([P-M]/Mx100%). Results: GACF was required because of up to 3.5% output variation dependence on the gantry angle. A 2D interpolation was required for OCR because the dose distribution was not radially symmetric especially for the deep options. The average percent differences were −0.03±0.98% (mean±SD) and the differences of all the measurements fell within ±3%. Conclusion: It is concluded that the model can be clinically used for the compact passively scattered proton therapy system. However, great care should be taken when the field-size is less than 5×5 cm{sup 2} where a direct output measurement is required due to substantial
International Nuclear Information System (INIS)
Fan, Peng; Hutton, Brian F.; Holstensson, Maria; Ljungberg, Michael; Hendrik Pretorius, P.; Prasad, Rameshwar; Liu, Chi; Ma, Tianyu; Liu, Yaqiang; Wang, Shi; Thorn, Stephanie L.; Stacy, Mitchel R.; Sinusas, Albert J.
2015-01-01
Purpose: The energy spectrum for a cadmium zinc telluride (CZT) detector has a low energy tail due to incomplete charge collection and intercrystal scattering. Due to these solid-state detector effects, scatter would be overestimated if the conventional triple-energy window (TEW) method is used for scatter and crosstalk corrections in CZT-based imaging systems. The objective of this work is to develop a scatter and crosstalk correction method for 99m Tc/ 123 I dual-radionuclide imaging for a CZT-based dedicated cardiac SPECT system with pinhole collimators (GE Discovery NM 530c/570c). Methods: A tailing model was developed to account for the low energy tail effects of the CZT detector. The parameters of the model were obtained using 99m Tc and 123 I point source measurements. A scatter model was defined to characterize the relationship between down-scatter and self-scatter projections. The parameters for this model were obtained from Monte Carlo simulation using SIMIND. The tailing and scatter models were further incorporated into a projection count model, and the primary and self-scatter projections of each radionuclide were determined with a maximum likelihood expectation maximization (MLEM) iterative estimation approach. The extracted scatter and crosstalk projections were then incorporated into MLEM image reconstruction as an additive term in forward projection to obtain scatter- and crosstalk-corrected images. The proposed method was validated using Monte Carlo simulation, line source experiment, anthropomorphic torso phantom studies, and patient studies. The performance of the proposed method was also compared to that obtained with the conventional TEW method. Results: Monte Carlo simulations and line source experiment demonstrated that the TEW method overestimated scatter while their proposed method provided more accurate scatter estimation by considering the low energy tail effect. In the phantom study, improved defect contrasts were observed with both
Energy Technology Data Exchange (ETDEWEB)
Fan, Peng [Department of Diagnostic Radiology, Yale University, New Haven, Connecticut 06520 and Department of Engineering Physics, Tsinghua University, Beijing 100084 (China); Hutton, Brian F. [Institute of Nuclear Medicine, University College London, London WC1E 6BT, United Kingdom and Centre for Medical Radiation Physics, University of Wollongong, New South Wales 2522 (Australia); Holstensson, Maria [Department of Nuclear Medicine, Karolinska University Hospital, Stockholm 14186 (Sweden); Ljungberg, Michael [Department of Medical Radiation Physics, Lund University, Lund 222 41 (Sweden); Hendrik Pretorius, P. [Department of Radiology, University of Massachusetts Medical School, Worcester, Massachusetts 01655 (United States); Prasad, Rameshwar; Liu, Chi, E-mail: chi.liu@yale.edu [Department of Diagnostic Radiology, Yale University, New Haven, Connecticut 06520 (United States); Ma, Tianyu; Liu, Yaqiang; Wang, Shi [Department of Engineering Physics, Tsinghua University, Beijing 100084 (China); Thorn, Stephanie L.; Stacy, Mitchel R.; Sinusas, Albert J. [Department of Internal Medicine, Yale Translational Research Imaging Center, Yale University, New Haven, Connecticut 06520 (United States)
2015-12-15
Purpose: The energy spectrum for a cadmium zinc telluride (CZT) detector has a low energy tail due to incomplete charge collection and intercrystal scattering. Due to these solid-state detector effects, scatter would be overestimated if the conventional triple-energy window (TEW) method is used for scatter and crosstalk corrections in CZT-based imaging systems. The objective of this work is to develop a scatter and crosstalk correction method for {sup 99m}Tc/{sup 123}I dual-radionuclide imaging for a CZT-based dedicated cardiac SPECT system with pinhole collimators (GE Discovery NM 530c/570c). Methods: A tailing model was developed to account for the low energy tail effects of the CZT detector. The parameters of the model were obtained using {sup 99m}Tc and {sup 123}I point source measurements. A scatter model was defined to characterize the relationship between down-scatter and self-scatter projections. The parameters for this model were obtained from Monte Carlo simulation using SIMIND. The tailing and scatter models were further incorporated into a projection count model, and the primary and self-scatter projections of each radionuclide were determined with a maximum likelihood expectation maximization (MLEM) iterative estimation approach. The extracted scatter and crosstalk projections were then incorporated into MLEM image reconstruction as an additive term in forward projection to obtain scatter- and crosstalk-corrected images. The proposed method was validated using Monte Carlo simulation, line source experiment, anthropomorphic torso phantom studies, and patient studies. The performance of the proposed method was also compared to that obtained with the conventional TEW method. Results: Monte Carlo simulations and line source experiment demonstrated that the TEW method overestimated scatter while their proposed method provided more accurate scatter estimation by considering the low energy tail effect. In the phantom study, improved defect contrasts were
Resistivity Correction Factor for the Four-Probe Method: Experiment II
Yamashita, Masato; Yamaguchi, Shoji; Nishii, Toshifumi; Kurihara, Hiroshi; Enjoji, Hideo
1989-05-01
Experimental verification of the theoretically derived resistivity correction factor F is presented. Factor F can be applied to a system consisting of a disk sample and a four-probe array. Measurements are made on isotropic graphite disks and crystalline ITO films. Factor F can correct the apparent variations of the data and lead to reasonable resistivities and sheet resistances. Here factor F is compared to other correction factors; i.e. FASTM and FJIS.
Energy Technology Data Exchange (ETDEWEB)
Liang, X; Zhang, Z; Xie, Y [Shenzhen Institutes of Advanced Technology, Chinese Academy of Sciences, Shenzhen, GuangDong (China); Gong, S; Niu, T [Department of Radiation Oncology, Sir Run Run Shaw Hospital, Zhejiang University School of Medicine, Hangzhou, Zhejiang (China); Institute of Translational Medicine, Zhejiang University, Hangzhou, Zhejiang (China); Zhou, Q [Department of Radiation Oncology, Sir Run Run Shaw Hospital, Zhejiang University School of Medicine, Hangzhou, Zhejiang (China)
2016-06-15
Purpose: X-ray scatter photons result in significant image quality degradation of cone-beam CT (CBCT). Measurement based algorithms using beam blocker directly acquire the scatter samples and achieve significant improvement on the quality of CBCT image. Within existing algorithms, single-scan and stationary beam blocker proposed previously is promising due to its simplicity and practicability. Although demonstrated effectively on tabletop system, the blocker fails to estimate the scatter distribution on clinical CBCT system mainly due to the gantry wobble. In addition, the uniform distributed blocker strips in our previous design results in primary data loss in the CBCT system and leads to the image artifacts due to data insufficiency. Methods: We investigate the motion behavior of the beam blocker in each projection and design an optimized non-uniform blocker strip distribution which accounts for the data insufficiency issue. An accurate scatter estimation is then achieved from the wobble modeling. Blocker wobble curve is estimated using threshold-based segmentation algorithms in each projection. In the blocker design optimization, the quality of final image is quantified using the number of the primary data loss voxels and the mesh adaptive direct search algorithm is applied to minimize the objective function. Scatter-corrected CT images are obtained using the optimized blocker. Results: The proposed method is evaluated using Catphan@504 phantom and a head patient. On the Catphan©504, our approach reduces the average CT number error from 115 Hounsfield unit (HU) to 11 HU in the selected regions of interest, and improves the image contrast by a factor of 1.45 in the high-contrast regions. On the head patient, the CT number error is reduced from 97 HU to 6 HU in the soft tissue region and image spatial non-uniformity is decreased from 27% to 5% after correction. Conclusion: The proposed optimized blocker design is practical and attractive for CBCT guided radiation
International Nuclear Information System (INIS)
Liang, X; Zhang, Z; Xie, Y; Gong, S; Niu, T; Zhou, Q
2016-01-01
Purpose: X-ray scatter photons result in significant image quality degradation of cone-beam CT (CBCT). Measurement based algorithms using beam blocker directly acquire the scatter samples and achieve significant improvement on the quality of CBCT image. Within existing algorithms, single-scan and stationary beam blocker proposed previously is promising due to its simplicity and practicability. Although demonstrated effectively on tabletop system, the blocker fails to estimate the scatter distribution on clinical CBCT system mainly due to the gantry wobble. In addition, the uniform distributed blocker strips in our previous design results in primary data loss in the CBCT system and leads to the image artifacts due to data insufficiency. Methods: We investigate the motion behavior of the beam blocker in each projection and design an optimized non-uniform blocker strip distribution which accounts for the data insufficiency issue. An accurate scatter estimation is then achieved from the wobble modeling. Blocker wobble curve is estimated using threshold-based segmentation algorithms in each projection. In the blocker design optimization, the quality of final image is quantified using the number of the primary data loss voxels and the mesh adaptive direct search algorithm is applied to minimize the objective function. Scatter-corrected CT images are obtained using the optimized blocker. Results: The proposed method is evaluated using Catphan@504 phantom and a head patient. On the Catphan©504, our approach reduces the average CT number error from 115 Hounsfield unit (HU) to 11 HU in the selected regions of interest, and improves the image contrast by a factor of 1.45 in the high-contrast regions. On the head patient, the CT number error is reduced from 97 HU to 6 HU in the soft tissue region and image spatial non-uniformity is decreased from 27% to 5% after correction. Conclusion: The proposed optimized blocker design is practical and attractive for CBCT guided radiation
International Nuclear Information System (INIS)
Blanco, F; Garcia, G
2009-01-01
A simplified form of the well-known screening-corrected additivity rule procedure for the calculation of electron-molecule cross sections is proposed for the treatment of some very large macro-molecules. While the comparison of the standard and simplified treatments for a DNA dodecamer reveals very similar results, the new treatment presents some important advantages for large molecules.
New results on the 3-loop heavy flavor corrections in deep-inelastic scattering
Energy Technology Data Exchange (ETDEWEB)
Behring, A.; Bluemlein, J.; Freitas, A. de [Deutsches Elektronen-Synchrotron, Zeuthen (Germany); and others
2013-12-15
We report on recent progress in the calculation of the 3-loop massiveWilson coefficients in deep inelastic scattering at general values of N for neutral- and charged-current reactions in the asymptotic region Q{sup 2}>>m{sup 2}. Four new out of eight massive operator matrix elements and Wilson coefficients have been obtained recently. We also discuss recent results on Feynman graphs containing two massive fermion lines and present complete results for the bubble topologies for all processes.
International Nuclear Information System (INIS)
Thing, Rune S.; Bernchou, Uffe; Brink, Carsten; Mainegra-Hing, Ernesto
2013-01-01
Purpose: Cone beam computed tomography (CBCT) image quality is limited by scattered photons. Monte Carlo (MC) simulations provide the ability of predicting the patient-specific scatter contamination in clinical CBCT imaging. Lengthy simulations prevent MC-based scatter correction from being fully implemented in a clinical setting. This study investigates the combination of using fast MC simulations to predict scatter distributions with a ray tracing algorithm to allow calibration between simulated and clinical CBCT images. Material and methods: An EGSnrc-based user code (egs c bct), was used to perform MC simulations of an Elekta XVI CBCT imaging system. A 60keV x-ray source was used, and air kerma scored at the detector plane. Several variance reduction techniques (VRTs) were used to increase the scatter calculation efficiency. Three patient phantoms based on CT scans were simulated, namely a brain, a thorax and a pelvis scan. A ray tracing algorithm was used to calculate the detector signal due to primary photons. A total of 288 projections were simulated, one for each thread on the computer cluster used for the investigation. Results: Scatter distributions for the brain, thorax and pelvis scan were simulated within 2 % statistical uncertainty in two hours per scan. Within the same time, the ray tracing algorithm provided the primary signal for each of the projections. Thus, all the data needed for MC-based scatter correction in clinical CBCT imaging was obtained within two hours per patient, using a full simulation of the clinical CBCT geometry. Conclusions: This study shows that use of MC-based scatter corrections in CBCT imaging has a great potential to improve CBCT image quality. By use of powerful VRTs to predict scatter distributions and a ray tracing algorithm to calculate the primary signal, it is possible to obtain the necessary data for patient specific MC scatter correction within two hours per patient
2002-01-01
Tile Calorimeter modules stored at CERN. The larger modules belong to the Barrel, whereas the smaller ones are for the two Extended Barrels. (The article was about the completion of the 64 modules for one of the latter.) The photo on the first page of the Bulletin n°26/2002, from 24 July 2002, illustrating the article «The ATLAS Tile Calorimeter gets into shape» was published with a wrong caption. We would like to apologise for this mistake and so publish it again with the correct caption.
Haze-factor maakt plaats voor F-scatter
Swinkels, G.L.A.M.
2014-01-01
Uitgelicht licht & scherming - Wageningen UR Glastuinbouw deed vorig jaar onderzoek naar een methode om de lichtspreiding onder diffuus glas te karakteriseren. dit resulteerde in een nieuwe waarde, de zogenaamde F-scatter voor voorwaartse lichtspreiding. Leveranciers van glas en
Kim, Kio; Habas, Piotr A; Rajagopalan, Vidya; Scott, Julia A; Corbett-Detig, James M; Rousseau, Francois; Barkovich, A James; Glenn, Orit A; Studholme, Colin
2011-09-01
A common solution to clinical MR imaging in the presence of large anatomical motion is to use fast multislice 2D studies to reduce slice acquisition time and provide clinically usable slice data. Recently, techniques have been developed which retrospectively correct large scale 3D motion between individual slices allowing the formation of a geometrically correct 3D volume from the multiple slice stacks. One challenge, however, in the final reconstruction process is the possibility of varying intensity bias in the slice data, typically due to the motion of the anatomy relative to imaging coils. As a result, slices which cover the same region of anatomy at different times may exhibit different sensitivity. This bias field inconsistency can induce artifacts in the final 3D reconstruction that can impact both clinical interpretation of key tissue boundaries and the automated analysis of the data. Here we describe a framework to estimate and correct the bias field inconsistency in each slice collectively across all motion corrupted image slices. Experiments using synthetic and clinical data show that the proposed method reduces intensity variability in tissues and improves the distinction between key tissue types.
75 FR 5536 - Pipeline Safety: Control Room Management/Human Factors, Correction
2010-02-03
... DEPARTMENT OF TRANSPORTATION Pipeline and Hazardous Materials Safety Administration 49 CFR Parts...: Control Room Management/Human Factors, Correction AGENCY: Pipeline and Hazardous Materials Safety... following correcting amendments: PART 192--TRANSPORTATION OF NATURAL AND OTHER GAS BY PIPELINE: MINIMUM...
Directory of Open Access Journals (Sweden)
2012-01-01
Full Text Available Regarding Gorelik, G., & Shackelford, T.K. (2011. Human sexual conflict from molecules to culture. Evolutionary Psychology, 9, 564–587: The authors wish to correct an omission in citation to the existing literature. In the final paragraph on p. 570, we neglected to cite Burch and Gallup (2006 [Burch, R. L., & Gallup, G. G., Jr. (2006. The psychobiology of human semen. In S. M. Platek & T. K. Shackelford (Eds., Female infidelity and paternal uncertainty (pp. 141–172. New York: Cambridge University Press.]. Burch and Gallup (2006 reviewed the relevant literature on FSH and LH discussed in this paragraph, and should have been cited accordingly. In addition, Burch and Gallup (2006 should have been cited as the originators of the hypothesis regarding the role of FSH and LH in the semen of rapists. The authors apologize for this oversight.
2002-01-01
The photo on the second page of the Bulletin n°48/2002, from 25 November 2002, illustrating the article «Spanish Visit to CERN» was published with a wrong caption. We would like to apologise for this mistake and so publish it again with the correct caption. The Spanish delegation, accompanied by Spanish scientists at CERN, also visited the LHC superconducting magnet test hall (photo). From left to right: Felix Rodriguez Mateos of CERN LHC Division, Josep Piqué i Camps, Spanish Minister of Science and Technology, César Dopazo, Director-General of CIEMAT (Spanish Research Centre for Energy, Environment and Technology), Juan Antonio Rubio, ETT Division Leader at CERN, Manuel Aguilar-Benitez, Spanish Delegate to Council, Manuel Delfino, IT Division Leader at CERN, and Gonzalo León, Secretary-General of Scientific Policy to the Minister.
Ionization chamber correction factors for MR-linacs.
Pojtinger, Stefan; Dohm, Oliver Steffen; Kapsch, Ralf-Peter; Thorwarth, Daniela
2018-06-07
Previously, readings of air-filled ionization chambers have been described as being influenced by magnetic fields. To use these chambers for dosimetry in magnetic resonance guided radiotherapy (MRgRT), this effect must be taken into account by introducing a correction factor k B . The purpose of this study is to systematically investigate k B for a typical reference setup for commercially available ionization chambers with different magnetic field strengths. The Monte Carlo simulation tool EGSnrc was used to simulate eight commercially available ionization chambers in magnetic fields whose magnetic flux density was in the range of 0-2.5 T. To validate the simulation, the influence of the magnetic field was experimentally determined for a PTW30013 Farmer-type chamber for magnetic flux densities between 0 and 1.425 T. Changes in the detector response of up to 8% depending on the magnetic flux density, on the chamber geometry and on the chamber orientation were obtained. In the experimental setup, a maximum deviation of less than 2% was observed when comparing measured values with simulated values. Dedicated values for two MR-linac systems (ViewRay MRIdian, ViewRay Inc, Cleveland, United States, 0.35 T/ 6 MV and Elekta Unity, Elekta AB, Stockholm, Sweden, 1.5 T/7 MV) were determined for future use in reference dosimetry. Simulated values for thimble-type chambers are in good agreement with experiments as well as with the results of previous publications. After further experimental validation, the results can be considered for definition of standard protocols for purposes of reference dosimetry in MRgRT.
Ionization chamber correction factors for MR-linacs
Pojtinger, Stefan; Steffen Dohm, Oliver; Kapsch, Ralf-Peter; Thorwarth, Daniela
2018-06-01
Previously, readings of air-filled ionization chambers have been described as being influenced by magnetic fields. To use these chambers for dosimetry in magnetic resonance guided radiotherapy (MRgRT), this effect must be taken into account by introducing a correction factor k B. The purpose of this study is to systematically investigate k B for a typical reference setup for commercially available ionization chambers with different magnetic field strengths. The Monte Carlo simulation tool EGSnrc was used to simulate eight commercially available ionization chambers in magnetic fields whose magnetic flux density was in the range of 0–2.5 T. To validate the simulation, the influence of the magnetic field was experimentally determined for a PTW30013 Farmer-type chamber for magnetic flux densities between 0 and 1.425 T. Changes in the detector response of up to 8% depending on the magnetic flux density, on the chamber geometry and on the chamber orientation were obtained. In the experimental setup, a maximum deviation of less than 2% was observed when comparing measured values with simulated values. Dedicated values for two MR-linac systems (ViewRay MRIdian, ViewRay Inc, Cleveland, United States, 0.35 T/ 6 MV and Elekta Unity, Elekta AB, Stockholm, Sweden, 1.5 T/7 MV) were determined for future use in reference dosimetry. Simulated values for thimble-type chambers are in good agreement with experiments as well as with the results of previous publications. After further experimental validation, the results can be considered for definition of standard protocols for purposes of reference dosimetry in MRgRT.
Directory of Open Access Journals (Sweden)
2014-01-01
Full Text Available Regarding Tagler, M. J., and Jeffers, H. M. (2013. Sex differences in attitudes toward partner infidelity. Evolutionary Psychology, 11, 821–832: The authors wish to correct values in the originally published manuscript. Specifically, incorrect 95% confidence intervals around the Cohen's d values were reported on page 826 of the manuscript where we reported the within-sex simple effects for the significant Participant Sex × Infidelity Type interaction (first paragraph, and for attitudes toward partner infidelity (second paragraph. Corrected values are presented in bold below. The authors would like to thank Dr. Bernard Beins at Ithaca College for bringing these errors to our attention. Men rated sexual infidelity significantly more distressing (M = 4.69, SD = 0.74 than they rated emotional infidelity (M = 4.32, SD = 0.92, F(1, 322 = 23.96, p < .001, d = 0.44, 95% CI [0.23, 0.65], but there was little difference between women's ratings of sexual (M = 4.80, SD = 0.48 and emotional infidelity (M = 4.76, SD = 0.57, F(1, 322 = 0.48, p = .29, d = 0.08, 95% CI [−0.10, 0.26]. As expected, men rated sexual infidelity (M = 1.44, SD = 0.70 more negatively than they rated emotional infidelity (M = 2.66, SD = 1.37, F(1, 322 = 120.00, p < .001, d = 1.12, 95% CI [0.85, 1.39]. Although women also rated sexual infidelity (M = 1.40, SD = 0.62 more negatively than they rated emotional infidelity (M = 2.09, SD = 1.10, this difference was not as large and thus in the evolutionary theory supportive direction, F(1, 322 = 72.03, p < .001, d = 0.77, 95% CI [0.60, 0.94].
Zhao, Cong; Zhong, Yuncheng; Duan, Xinhui; Zhang, You; Huang, Xiaokun; Wang, Jing; Jin, Mingwu
2018-06-01
Four-dimensional (4D) x-ray cone-beam computed tomography (CBCT) is important for a precise radiation therapy for lung cancer. Due to the repeated use and 4D acquisition over a course of radiotherapy, the radiation dose becomes a concern. Meanwhile, the scatter contamination in CBCT deteriorates image quality for treatment tasks. In this work, we propose the use of a moving blocker (MB) during the 4D CBCT acquisition (‘4D MB’) and to combine motion-compensated reconstruction to address these two issues simultaneously. In 4D MB CBCT, the moving blocker reduces the x-ray flux passing through the patient and collects the scatter information in the blocked region at the same time. The scatter signal is estimated from the blocked region for correction. Even though the number of projection views and projection data in each view are not complete for conventional reconstruction, 4D reconstruction with a total-variation (TV) constraint and a motion-compensated temporal constraint can utilize both spatial gradient sparsity and temporal correlations among different phases to overcome the missing data problem. The feasibility simulation studies using the 4D NCAT phantom showed that 4D MB with motion-compensated reconstruction with 1/3 imaging dose reduction could produce satisfactory images and achieve 37% improvement on structural similarity (SSIM) index and 55% improvement on root mean square error (RMSE), compared to 4D reconstruction at the regular imaging dose without scatter correction. For the same 4D MB data, 4D reconstruction outperformed 3D TV reconstruction by 28% on SSIM and 34% on RMSE. A study of synthetic patient data also demonstrated the potential of 4D MB to reduce the radiation dose by 1/3 without compromising the image quality. This work paves the way for more comprehensive studies to investigate the dose reduction limit offered by this novel 4D MB method using physical phantom experiments and real patient data based on clinical relevant metrics.
Coherence factors in a high-tc cuprate probed by quasi-particle scattering off vortices.
Hanaguri, T; Kohsaka, Y; Ono, M; Maltseva, M; Coleman, P; Yamada, I; Azuma, M; Takano, M; Ohishi, K; Takagi, H
2009-02-13
When electrons pair in a superconductor, quasi-particles develop an acute sensitivity to different types of scattering potential that is described by the appearance of coherence factors in the scattering amplitudes. Although the effects of coherence factors are well established in isotropic superconductors, they are much harder to detect in their anisotropic counterparts, such as high-superconducting-transition-temperature cuprates. We demonstrate an approach that highlights the momentum-dependent coherence factors in Ca2-xNaxCuO2Cl2. We used Fourier-transform scanning tunneling spectroscopy to reveal a magnetic-field dependence in quasi-particle scattering interference patterns that is sensitive to the sign of the anisotropic gap. This result is associated with the d-wave coherence factors and quasi-particle scattering off vortices. Our technique thus provides insights into the nature of electron pairing as well as quasi-particle scattering processes in unconventional superconductors.
Energy Technology Data Exchange (ETDEWEB)
Chun, Se Young [School of Electrical and Computer Engineering, Ulsan National Institute of Science and Technology (UNIST), Ulsan (Korea, Republic of)
2016-03-15
PET and SPECT are important tools for providing valuable molecular information about patients to clinicians. Advances in nuclear medicine hardware technologies and statistical image reconstruction algorithms enabled significantly improved image quality. Sequentially or simultaneously acquired anatomical images such as CT and MRI from hybrid scanners are also important ingredients for improving the image quality of PET or SPECT further. High-quality anatomical information has been used and investigated for attenuation and scatter corrections, motion compensation, and noise reduction via post-reconstruction filtering and regularization in inverse problems. In this article, we will review works using anatomical information for molecular image reconstruction algorithms for better image quality by describing mathematical models, discussing sources of anatomical information for different cases, and showing some examples.
Energy Technology Data Exchange (ETDEWEB)
Conti, C.C., E-mail: ccconti@ird.gov.br [Institute for Radioprotection and Dosimetry – IRD/CNEN, Rio de Janeiro (Brazil); Physics Institute, State University of Rio de Janeiro – UERJ, Rio de Janeiro (Brazil); Anjos, M.J. [Physics Institute, State University of Rio de Janeiro – UERJ, Rio de Janeiro (Brazil); Salgado, C.M. [Nuclear Engineering Institute – IEN/CNEN, Rio de Janeiro (Brazil)
2014-09-15
Highlights: •This work describes a procedure for sample self-absorption correction. •The use of Monte Carlo simulation to calculate the mass attenuation coefficients curve was effective. •No need for transmission measurement, saving time, financial resources and effort. •This article provides de curves for the 90° scattering angle. •Calculation on-line at (www.macx.net.br). -- Abstract: X-ray fluorescence technique plays an important role in nondestructive analysis nowadays. The development of equipment, including portable ones, enables a wide assortment of possibilities for analysis of stable elements, even in trace concentrations. Nevertheless, despite of the advantages, one important drawback is radiation self-attenuation in the sample being measured, which needs to be considered in the calculation for the proper determination of elemental concentration. The mass attenuation coefficient can be determined by transmission measurement, but, in this case, the sample must be in slab shape geometry and demands two different setups and measurements. The Rayleigh to Compton scattering ratio, determined from the X-ray fluorescence spectrum, provides a link to the mass attenuation coefficient by means of a polynomial type equation. This work presents a way to construct a Rayleigh to Compton scattering ratio versus mass attenuation coefficient curve by using the MCNP5 Monte Carlo computer code. The comparison between the calculated and literature values of the mass attenuation coefficient for some known samples showed to be within 15%. This calculation procedure is available on-line at (www.macx.net.br)
Lee, Ho; Fahimian, Benjamin P.; Xing, Lei
2017-03-01
This paper proposes a binary moving-blocker (BMB)-based technique for scatter correction in cone-beam computed tomography (CBCT). In concept, a beam blocker consisting of lead strips, mounted in front of the x-ray tube, moves rapidly in and out of the beam during a single gantry rotation. The projections are acquired in alternating phases of blocked and unblocked cone beams, where the blocked phase results in a stripe pattern in the width direction. To derive the scatter map from the blocked projections, 1D B-Spline interpolation/extrapolation is applied by using the detected information in the shaded regions. The scatter map of the unblocked projections is corrected by averaging two scatter maps that correspond to their adjacent blocked projections. The scatter-corrected projections are obtained by subtracting the corresponding scatter maps from the projection data and are utilized to generate the CBCT image by a compressed-sensing (CS)-based iterative reconstruction algorithm. Catphan504 and pelvis phantoms were used to evaluate the method’s performance. The proposed BMB-based technique provided an effective method to enhance the image quality by suppressing scatter-induced artifacts, such as ring artifacts around the bowtie area. Compared to CBCT without a blocker, the spatial nonuniformity was reduced from 9.1% to 3.1%. The root-mean-square error of the CT numbers in the regions of interest (ROIs) was reduced from 30.2 HU to 3.8 HU. In addition to high resolution, comparable to that of the benchmark image, the CS-based reconstruction also led to a better contrast-to-noise ratio in seven ROIs. The proposed technique enables complete scatter-corrected CBCT imaging with width-truncated projections and allows reducing the acquisition time to approximately half. This work may have significant implications for image-guided or adaptive radiation therapy, where CBCT is often used.
Block, Martin M
2002-01-01
Using an eikonal structure for the scattering amplitude, factorization theorems for nucleon-nucleon, gamma p and gamma gamma scattering at high energies have been derived, using only some very general assumptions. Using a QCD-inspired eikonal analysis of nucleon-nucleon scattering, we present here experimental confirmation for factorization of cross sections, nuclear slope parameters B and rho -values (ratio of real to imaginary portion of forward scattering amplitudes), showing that: 1) the three factorization theorems of Block and Kaidalov [2000] hold, 2) the additive quark model holds to approximately=1%, and 3) vector dominance holds to better than approximately=4%. Predictions for the total cross section, elastic cross section and other forward scattering parameters at the LHC (14 TeV) are given. (12 refs).
Universality of soft and collinear factors in hard-scattering factorization
International Nuclear Information System (INIS)
Collins, John C.; Metz, Andreas
2004-01-01
Universality in QCD factorization of parton densities, fragmentation functions, and soft factors is endangered by the process dependence of the directions of Wilson lines in their definitions. We find a choice of directions that is consistent with factorization and that gives universality between e + e - annihilation, semi-inclusive deep-inelastic scattering, and the Drell-Yan process. Universality is only modified by a time-reversal transformation of the soft function and parton densities between Drell-Yan and the other processes, whose only effect is the known reversal of sign for T-odd parton densities such as the Sivers function. The modifications of the definitions needed to remove rapidity divergences with lightlike Wilson lines do not affect the results
Tyler, D K
2002-01-01
The activity assay of a radiopharmaceutical administration to a patient is normally achieved via the use of a radionuclide calibrator. Because of the different geometries and elemental compositions between plastic syringes and glass vials, the calibration factors for syringes may well be significantly different from those for the glass containers. The magnitude of these differences depends on the energies of the emitted photons. For some radionuclides variations have been observed of 70 %, it is therefore important to recalibrate for syringes or use syringe calibration factors. Calibration factors and volume correction factors have been derived for the NPL secondary standard radionuclide calibrator, for a variety of commonly used syringes and needles, for the most commonly used medical radionuclide.
Macroscopic folded form factors for 12C + 12C inelastic scattering
International Nuclear Information System (INIS)
Rickertsen, L.D.; Satchler, G.R.; Stokstad, R.G.; Wieland, R.M.
1976-01-01
The angular distributions for the scattering of carbon-12 from carbon-12 at 117.1 MeV are shown as is also the result of coupled-channel calculations for the elastic and inelastic scattering using these folded form factors
Effects of attenuation and scatter corrections in cat brain PET images using microPET R4 scanner
International Nuclear Information System (INIS)
Kim, Jin Su; Lee, Jae Sung; Lee, Jong Jin
2006-01-01
The aim of this study was to examine the effects of attenuation correction (AC) and scatter correction (SC) on the quantification of PET count rates. To assess the effects of AC and SC, 18 F-FDG PET images of phantom and cat brain were acquired using microPET R4 scanner. Thirty-minute transmission images using 68 Ge source and emission images after injection of FDG were acquired. PET images were reconstructed using. 2D OSEM. AC and SC were applied. Regional count rates were measured using ROls drawn on cerebral cortex including frontal, parietal, and latral temporal lobes and deep gray matter including head of caudate nucleus, putamen and thalamus for pre- and post-AC and SC images. The count rates were then normalized with the injected dose per body weight. To assess the effects of AC, count ratio of 'deep gray matter/cerebral cortex' was calculated. To assess the effects of SC, ROls were also drawn on the gray matter (GM) and white matter (WM), and contrast between them ((GM-WM)/GM) was measured. After the AC, count ratio of 'deep gray matter/cerebral cortex' was increased by 17±7%. After the SC, contrast was also increased by 12±3%. Relative count of deep gray matter and contrast between gray and white matters were increased after AC and SC, suggesting that the AC would be critical for the quantitative analysis of cat brain PET data
International Nuclear Information System (INIS)
Williams, W.G.
1975-01-01
The use of the polarization analysis technique to separate spin-flip from non-spin-flip thermal neutron scattering is especially important in determining magnetic scattering cross-sections. In order to identify a spin-flip ratio in the scattering with a particular scattering process, it is necessary to correct the experimentally observed 'flipping-ratio' to allow for the efficiencies of the vital instrument components (polarizers and spin-flippers), as well as multiple scattering effects in the sample. Analytical expressions for these corections are presented and their magnitudes in typical cases estimated. The errors in measurement depend strongly on the uncertainties in the calibration of the efficiencies of the polarizers and the spin-flipper. The final section is devoted to a discussion of polarization analysis instruments
Definition of correcting factors for absolute radon content measurement formula
International Nuclear Information System (INIS)
Ji Changsong; Xiao Ziyun; Yang Jianfeng
1992-01-01
The absolute method of radio content measurement is based on thomas radon measurement formula. It was found in experiment that the systematic error existed in radon content measurement by means of thomas formula. By the analysis on the behaviour of radon daughter five factors including filter efficiency, detector construction factor, self-absorbance, energy spectrum factor, and gravity factor were introduced into the thomas formula, so that the systematic error was eliminated. The measuring methods of the five factors are given
Directory of Open Access Journals (Sweden)
Mahsa Noori Asl
2013-01-01
Full Text Available Compton-scattered photons included within the photopeak pulse-height window result in the degradation of SPECT images both qualitatively and quantitatively. The purpose of this study is to evaluate and compare six scatter correction methods based on setting the energy windows in 99m Tc spectrum. SIMIND Monte Carlo simulation is used to generate the projection images from a cold-sphere hot-background phantom. For evaluation of different scatter correction methods, three assessment criteria including image contrast, signal-to-noise ratio (SNR and relative noise of the background (RNB are considered. Except for the dual-photopeak window (DPW method, the image contrast of the five cold spheres is improved in the range of 2.7-26%. Among methods considered, two methods show a nonuniform correction performance. The RNB for all of the scatter correction methods is ranged from minimum 0.03 for DPW method to maximum 0.0727 for the three energy window (TEW method using trapezoidal approximation. The TEW method using triangular approximation because of ease of implementation, good improvement of the image contrast and the SNR for the five cold spheres, and the low noise level is proposed as most appropriate correction method.
Energy Technology Data Exchange (ETDEWEB)
Aspelund, O; Gustafsson, B
1967-05-15
After an introductory discussion of various methods for correction of experimental left-right ratios for polarized multiple-scattering and finite-geometry effects necessary and sufficient formulas for consistent tracking of polarization effects in successive scattering orders are derived. The simplifying assumptions are then made that the scattering is purely elastic and nuclear, and that in the description of the kinematics of the arbitrary Scattering {mu}, only one triple-parameter - the so-called spin rotation parameter {beta}{sup ({mu})} - is required. Based upon these formulas a general discussion of the importance of the correct inclusion of polarization effects in any scattering order is presented. Special attention is then paid to the question of depolarization of an already polarized beam. Subsequently, the afore-mentioned formulas are incorporated in the comprehensive Monte Carlo program MULTPOL, which has been designed so as to correctly account for finite-geometry effects in the sense that both the scattering sample and the detectors (both having cylindrical shapes) are objects of finite dimensions located at finite distances from each other and from the source of polarized fast-neutrons. A special feature of MULTPOL is the application of the method of correlated sampling for reduction of the standard deviations .of the results of the simulated experiment. Typical data of performance of MULTPOL have been obtained by the application of this program to the correction of experimental polarization data observed in n + '{sup 12}C elastic scattering between 1 and 2 MeV. Finally, in the concluding remarks the possible modification of MULTPOL to other experimental geometries is briefly discussed.
Energy Technology Data Exchange (ETDEWEB)
Blümlein, Johannes, E-mail: Johannes.Bluemlein@desy.de [Deutsches Elektronen–Synchrotron, DESY, Platanenallee 6, D-15738 Zeuthen (Germany); Hasselhuhn, Alexander [Deutsches Elektronen–Synchrotron, DESY, Platanenallee 6, D-15738 Zeuthen (Germany); Research Institute for Symbolic Computation (RISC), Johannes Kepler University, Altenbergerstraße 69, A-4040 Linz (Austria); Pfoh, Torsten [Deutsches Elektronen–Synchrotron, DESY, Platanenallee 6, D-15738 Zeuthen (Germany)
2014-04-15
We calculate the O(α{sub s}{sup 2}) heavy flavor corrections to charged current deep-inelastic scattering at large scales Q{sup 2}≫m{sup 2}. The contributing Wilson coefficients are given as convolutions between massive operator matrix elements and massless Wilson coefficients. Foregoing results in the literature are extended and corrected. Numerical results are presented for the kinematic region of the HERA data.
Spatial correction factors for YALINA Booster facility loaded with medium and low enriched fuels
International Nuclear Information System (INIS)
Talamo, A.; Gohar, Y.; Bournos, V.; Fokov, Y.; Kiyavitskaya, H.; Routkovskaya, C.
2012-01-01
The Bell and Glasstone spatial correction factor is used in analyses of subcritical assemblies to correct the experimental reactivity as function of the detector position. Besides the detector position, several other parameters affect the correction factor: the energy weighting function of the detector, the detector size, the energy-angle distribution of source neutrons, and the reactivity of the subcritical assembly. This work focuses on the dependency of the correction factor on the detector material and it investigates the YALINA Booster subcritical assembly loaded with medium (36%) and low (10%) enriched fuels. (authors)
Spatial correction factors for YALINA Booster facility loaded with medium and low enriched fuels
Energy Technology Data Exchange (ETDEWEB)
Talamo, A.; Gohar, Y. [Argonne National Laboratory, 9700 S. Cass Ave, Argonne, IL 60439 (United States); Bournos, V.; Fokov, Y.; Kiyavitskaya, H.; Routkovskaya, C. [Joint Inst. for Power and Nuclear Research-Sosny, 99 Academician A.K.Krasin Str, Minsk 220109 (Belarus)
2012-07-01
The Bell and Glasstone spatial correction factor is used in analyses of subcritical assemblies to correct the experimental reactivity as function of the detector position. Besides the detector position, several other parameters affect the correction factor: the energy weighting function of the detector, the detector size, the energy-angle distribution of source neutrons, and the reactivity of the subcritical assembly. This work focuses on the dependency of the correction factor on the detector material and it investigates the YALINA Booster subcritical assembly loaded with medium (36%) and low (10%) enriched fuels. (authors)
Inelastic magnetic electron scattering form factors of the 26 Mg nucleus
Indian Academy of Sciences (India)
Magnetic electron scattering (3) form factors with core polarization effects, ... to 3+ states of the 26Mg nucleus have been studied using shell model calculations. ... The wave functions of the radial single-particle matrix elements have been ...
International Nuclear Information System (INIS)
Fujita, Masahiro; Seneca, Nicholas; Innis, Robert B.; Varrone, Andrea; Kim, Kyeong Min; Watabe, Hiroshi; Iida, Hidehiro; Zoghbi, Sami S.; Tipre, Dnyanesh; Seibyl, John P.
2004-01-01
Prior studies with anthropomorphic phantoms and single, static in vivo brain images have demonstrated that scatter correction significantly improves the accuracy of regional quantitation of single-photon emission tomography (SPET) brain images. Since the regional distribution of activity changes following a bolus injection of a typical neuroreceptor ligand, we examined the effect of scatter correction on the compartmental modeling of serial dynamic images of striatal and extrastriatal dopamine D 2 receptors using [ 123 I]epidepride. Eight healthy human subjects [age 30±8 (range 22-46) years] participated in a study with a bolus injection of 373±12 (354-389) MBq [ 123 I]epidepride and data acquisition over a period of 14 h. A transmission scan was obtained in each study for attenuation and scatter correction. Distribution volumes were calculated by means of compartmental nonlinear least-squares analysis using metabolite-corrected arterial input function and brain data processed with scatter correction using narrow-beam geometry μ (SC) and without scatter correction using broad-beam μ (NoSC). Effects of SC were markedly different among brain regions. SC increased activities in the putamen and thalamus after 1-1.5 h while it decreased activity during the entire experiment in the temporal cortex and cerebellum. Compared with NoSC, SC significantly increased specific distribution volume in the putamen (58%, P=0.0001) and thalamus (23%, P=0.0297). Compared with NoSC, SC made regional distribution of the specific distribution volume closer to that of [ 18 F]fallypride. It is concluded that SC is required for accurate quantification of distribution volumes of receptor ligands in SPET studies. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Fujita, Masahiro; Seneca, Nicholas; Innis, Robert B. [Department of Psychiatry, Yale University School of Medicine and VA Connecticut Healthcare System, West Haven, CT (United States); Molecular Imaging Branch, National Institute of Mental Health, Bethesda, MD (United States); Varrone, Andrea [Department of Psychiatry, Yale University School of Medicine and VA Connecticut Healthcare System, West Haven, CT (United States); Biostructure and Bioimaging Institute, National Research Council, Napoli (Italy); Kim, Kyeong Min; Watabe, Hiroshi; Iida, Hidehiro [Department of Investigative Radiology, National Cardiovascular Center Research Institute, Osaka (Japan); Zoghbi, Sami S. [Department of Psychiatry, Yale University School of Medicine and VA Connecticut Healthcare System, West Haven, CT (United States); Molecular Imaging Branch, National Institute of Mental Health, Bethesda, MD (United States); Department of Radiology, Yale University School of Medicine and VA Connecticut Healthcare System, West Haven, CT (United States); Tipre, Dnyanesh [Molecular Imaging Branch, National Institute of Mental Health, Bethesda, MD (United States); Seibyl, John P. [Institute for Neurodegenerative Disorders, New Haven, CT (United States)
2004-05-01
Prior studies with anthropomorphic phantoms and single, static in vivo brain images have demonstrated that scatter correction significantly improves the accuracy of regional quantitation of single-photon emission tomography (SPET) brain images. Since the regional distribution of activity changes following a bolus injection of a typical neuroreceptor ligand, we examined the effect of scatter correction on the compartmental modeling of serial dynamic images of striatal and extrastriatal dopamine D{sub 2} receptors using [{sup 123}I]epidepride. Eight healthy human subjects [age 30{+-}8 (range 22-46) years] participated in a study with a bolus injection of 373{+-}12 (354-389) MBq [{sup 123}I]epidepride and data acquisition over a period of 14 h. A transmission scan was obtained in each study for attenuation and scatter correction. Distribution volumes were calculated by means of compartmental nonlinear least-squares analysis using metabolite-corrected arterial input function and brain data processed with scatter correction using narrow-beam geometry {mu} (SC) and without scatter correction using broad-beam {mu} (NoSC). Effects of SC were markedly different among brain regions. SC increased activities in the putamen and thalamus after 1-1.5 h while it decreased activity during the entire experiment in the temporal cortex and cerebellum. Compared with NoSC, SC significantly increased specific distribution volume in the putamen (58%, P=0.0001) and thalamus (23%, P=0.0297). Compared with NoSC, SC made regional distribution of the specific distribution volume closer to that of [{sup 18}F]fallypride. It is concluded that SC is required for accurate quantification of distribution volumes of receptor ligands in SPET studies. (orig.)
Chi, Zhijun; Du, Yingchao; Huang, Wenhui; Tang, Chuanxiang
2017-12-01
The necessity for compact and relatively low cost x-ray sources with monochromaticity, continuous tunability of x-ray energy, high spatial coherence, straightforward polarization control, and high brightness has led to the rapid development of Thomson scattering x-ray sources. To meet the requirement of in-situ monochromatic computed tomography (CT) for large-scale and/or high-attenuation materials based on this type of x-ray source, there is an increasing demand for effective algorithms to correct the energy-angle correlation. In this paper, we take advantage of the parametrization of the x-ray attenuation coefficient to resolve this problem. The linear attenuation coefficient of a material can be decomposed into a linear combination of the energy-dependent photoelectric and Compton cross-sections in the keV energy regime without K-edge discontinuities, and the line integrals of the decomposition coefficients of the above two parts can be determined by performing two spectrally different measurements. After that, the line integral of the linear attenuation coefficient of an imaging object at a certain interested energy can be derived through the above parametrization formula, and monochromatic CT can be reconstructed at this energy using traditional reconstruction methods, e.g., filtered back projection or algebraic reconstruction technique. Not only can monochromatic CT be realized, but also the distributions of the effective atomic number and electron density of the imaging object can be retrieved at the expense of dual-energy CT scan. Simulation results validate our proposal and will be shown in this paper. Our results will further expand the scope of application for Thomson scattering x-ray sources.
DEFF Research Database (Denmark)
Kjærgaard, Magnus; Poulsen, Flemming Martin
2011-01-01
Random coil chemical shifts are necessary for secondary chemical shift analysis, which is the main NMR method for identification of secondary structure in proteins. One of the largest challenges in the determination of random coil chemical shifts is accounting for the effect of neighboring residues....... The contributions from the neighboring residues are typically removed by using neighbor correction factors determined based on each residue's effect on glycine chemical shifts. Due to its unusual conformational freedom, glycine may be particularly unrepresentative for the remaining residue types. In this study, we...... in the conformational ensemble are an important source of neighbor effects in disordered proteins. Glutamine derived random coil chemical shifts and correction factors modestly improve our ability to predict (13)C chemical shifts of intrinsically disordered proteins compared to existing datasets, and may thus improve...
PMBLDC motor drive with power factor correction controller
DEFF Research Database (Denmark)
George, G.J.; Ramachandran, Rakesh; Arun, N.
2012-01-01
reliability, and low maintenance requirements. The proposed Power Factor Controller topology improves power quality by improving performance of PMBLDCM drive, such as reduction of AC main current harmonics, near unity power factor. PFC converter forces the drive to draw sinusoidal supply current in phase...
International Nuclear Information System (INIS)
Cho, Sung Koo; Choi, Sang Hyoun; Kim, Chan Hyeong; Na, Seong Ho
2006-01-01
In recent years, the MOSFET dosimeter has been widely used in various medical applications such as dose verification in radiation therapeutic and diagnostic applications. The MOSFET dosimeter is, however, mainly made of silicon and shows some energy dependence for low energy photons. Therefore, the MOSFET dosimeter tends to overestimate the dose for low energy scattered photons in a phantom. This study determines the correction factors to compensate these dependences of the MOSFET dosimeter in ATOM phantom. For this, we first constructed a computational model of the ATOM phantom based on the 3D CT image data of the phantom. The voxel phantom was then implemented in a Monte Carlo simulation code and used to calculate the energy spectrum of the photon field at each of the MOSFET dosimeter locations in the phantom. Finally, the correction factors were calculated based on the energy spectrum of the photon field at the dosimeter locations and the pre-determined energy and directional dependence of the MOSFET dosimeter. Our result for 60 Co and 137 Cs photon fields shows that the correction factors are distributed within the range of 0.89 and 0.97 considering all the MOSFET dosimeter locations in the phantom
New method in obtaining correction factor of power confirming
International Nuclear Information System (INIS)
Deng Yongjun; Li Rundong; Liu Yongkang; Zhou Wei
2010-01-01
Westcott theory is the most widely used method in reactor power calibration, which particularly suited to research reactor. But this method is very fussy because lots of correction parameters which rely on empirical formula to special reactor type are needed. The incidence coefficient between foil activity and reactor power was obtained by Monte-Carlo calculation, which was carried out with precise description of the reactor core and the foil arrangement position by MCNP input card. So the reactor power was determined by the core neutron fluence profile and the foil activity placed in the position for normalization use. The characteristic of this new method is simpler, more flexible and accurate than Westcott theory. In this paper, the results of SPRR-300 obtained by the new method in theory were compared with the experimental results, which verified the possibility of this new method. (authors)
Analysis of the factors that affect photon counts in Compton scattering
International Nuclear Information System (INIS)
Luo, Guang; Xiao, Guangyu
2015-01-01
Compton scattering has been applied in a variety of fields. The factors that affect Compton scattering have been studied extensively in the literature. However, the factors that affect the measured photon counts in Compton scattering are rarely considered. In this paper, we make a detailed discussion on those factors. First, Compton scattering experiments of some alloy series and powder mixture series are explored. Second, the electron density is researched in terms of atom and lattice constants. Third, the factor of attenuation coefficient is discussed. And then, the active degree of electrons is discussed based on the DFT theory. Lastly, the conclusions are made, that the factors affecting Compton scattering photon counts include mainly electron number density, attenuation coefficient and active degree of electrons. - Highlights: • Compton scattering experiments of some alloy series and powder mixture series are explored. • The influence of electron density is researched in terms of atom and lattice constants. • The influence of attenuation coefficient is discussed. • The active degree of electrons is discussed detailedly based on DFT theory
Wuhrer, R.; Moran, K.
2014-03-01
Quantitative X-ray mapping with silicon drift detectors and multi-EDS detector systems have become an invaluable analysis technique and one of the most useful methods of X-ray microanalysis today. The time to perform an X-ray map has reduced considerably with the ability to map minor and trace elements very accurately due to the larger detector area and higher count rate detectors. Live X-ray imaging can now be performed with a significant amount of data collected in a matter of minutes. A great deal of information can be obtained from X-ray maps. This includes; elemental relationship or scatter diagram creation, elemental ratio mapping, chemical phase mapping (CPM) and quantitative X-ray maps. In obtaining quantitative x-ray maps, we are able to easily generate atomic number (Z), absorption (A), fluorescence (F), theoretical back scatter coefficient (η), and quantitative total maps from each pixel in the image. This allows us to generate an image corresponding to each factor (for each element present). These images allow the user to predict and verify where they are likely to have problems in our images, and are especially helpful to look at possible interface artefacts. The post-processing techniques to improve the quantitation of X-ray map data and the development of post processing techniques for improved characterisation are covered in this paper.
International Nuclear Information System (INIS)
Wuhrer, R; Moran, K
2014-01-01
Quantitative X-ray mapping with silicon drift detectors and multi-EDS detector systems have become an invaluable analysis technique and one of the most useful methods of X-ray microanalysis today. The time to perform an X-ray map has reduced considerably with the ability to map minor and trace elements very accurately due to the larger detector area and higher count rate detectors. Live X-ray imaging can now be performed with a significant amount of data collected in a matter of minutes. A great deal of information can be obtained from X-ray maps. This includes; elemental relationship or scatter diagram creation, elemental ratio mapping, chemical phase mapping (CPM) and quantitative X-ray maps. In obtaining quantitative x-ray maps, we are able to easily generate atomic number (Z), absorption (A), fluorescence (F), theoretical back scatter coefficient (η), and quantitative total maps from each pixel in the image. This allows us to generate an image corresponding to each factor (for each element present). These images allow the user to predict and verify where they are likely to have problems in our images, and are especially helpful to look at possible interface artefacts. The post-processing techniques to improve the quantitation of X-ray map data and the development of post processing techniques for improved characterisation are covered in this paper
International Nuclear Information System (INIS)
Palmans, H.; Verhaegen, F.
1995-01-01
In the last decade, several clinical proton beam therapy facilities have been developed. To satisfy the demand for uniformity in clinical (routine) proton beam dosimetry two dosimetry protocols (ECHED and AAPM) have been published. Both protocols neglect the influence of ion chamber dependent parameters on dose determination in proton beams because of the scatter properties of these beams, although the problem has not been studied thoroughly yet. A comparison between water calorimetry and ionisation chamber dosimetry showed a discrepancy of 2.6% between the former method and ionometry following the ECHED protocol. Possibly, a small part of this difference can be attributed to chamber dependent correction factors. Indications for this possibility are found in ionometry measurements. To allow the simulation of complex geometries with different media necessary for the study of those corrections, an existing proton Monte Carlo code (PTRAN, Berger) has been modified. The original code, that applies Mollire's multiple scattering theory and Vavilov's energy straggling theory, calculates depth dose profiles, energy distributions and radial distributions for pencil beams in water. Comparisons with measurements and calculations reported in the literature are done to test the program's accuracy. Preliminary results of the influence of chamber design and chamber materials on dose to water determination are presented
Energy Technology Data Exchange (ETDEWEB)
Palmans, H [Ghent Univ. (Belgium). Dept. of Biomedical Physics; Verhaegen, F
1995-12-01
In the last decade, several clinical proton beam therapy facilities have been developed. To satisfy the demand for uniformity in clinical (routine) proton beam dosimetry two dosimetry protocols (ECHED and AAPM) have been published. Both protocols neglect the influence of ion chamber dependent parameters on dose determination in proton beams because of the scatter properties of these beams, although the problem has not been studied thoroughly yet. A comparison between water calorimetry and ionisation chamber dosimetry showed a discrepancy of 2.6% between the former method and ionometry following the ECHED protocol. Possibly, a small part of this difference can be attributed to chamber dependent correction factors. Indications for this possibility are found in ionometry measurements. To allow the simulation of complex geometries with different media necessary for the study of those corrections, an existing proton Monte Carlo code (PTRAN, Berger) has been modified. The original code, that applies Mollire`s multiple scattering theory and Vavilov`s energy straggling theory, calculates depth dose profiles, energy distributions and radial distributions for pencil beams in water. Comparisons with measurements and calculations reported in the literature are done to test the program`s accuracy. Preliminary results of the influence of chamber design and chamber materials on dose to water determination are presented.
Factorization for short distance hadron-hadron scattering
International Nuclear Information System (INIS)
Collins, J.C.; Illinois Inst. of Tech., Chicago; Soper, D.E.; Sterman, G.
1985-01-01
We show that factorization holds at leading twist in the Drell-Yang cross section dsigma/dQ 2 dy and related inclusive hadron-hadron cross sections. We review the heuristic arguments for factorization, as well as the difficulties which must be overcome in a proof. We go on to give detailed arguments for the all order cancellation of soft gluons, and to show how this leads to factorization. (orig.)
International Nuclear Information System (INIS)
Nasrun, S; Syamsul-Hadi, M; Sumardi
2000-01-01
Gamma-ray Spectrometer Exploranium GR-320 is the instrument radiometric survey which is able to measure radioelement content directly in field based on partial gamma-ray energy of elements. Because of the instrument is new and it was granted from the lAEA, so it is necessarily to create a correction factor for the instrument due to be gaining the better data. Correction factor was generated from comparing gamma spectrometer's radioelement content to those of chemical analysed data of calibration pad. The correction factor for Potassium (K) is 1.31, uranium is 1.46, and thorium is 0.39
International Nuclear Information System (INIS)
Haddad, Kh.
2009-02-01
Gamma spectrometry forms the most important and capable tool for measuring radioactive materials. Determination of the efficiency and attenuation correction factors is the most tedious problem in the gamma spectrometric assay of bulk samples. A new experimental and easy method for these correction factors determination using self radiation was proposed in this work. An experimental study of the correlation between self attenuation correction factor and sample thickness and its practical application was also introduced. The work was performed on NORM and uranyl nitrate bulk sample. The results of proposed methods agreed with those of traditional ones.(author)
Active Power Factor Correction Using a Sliding Mode Controller
Directory of Open Access Journals (Sweden)
Korhan KAYIŞLI
2008-03-01
Full Text Available In this paper, a sliding mode controller is designed for active shaping of the input current in the boost converter. Robustness of the designed controller is tested with variable output voltage references, different loads and network voltage variations. For the simulations, MATLAB/Simulink programme is used. From simulation results, the same phase was provided between input current and input voltage and nearly unity power factor was obtained.
Resistivity Correction Factor for the Four-Probe Method: Experiment I
Yamashita, Masato; Yamaguchi, Shoji; Enjoji, Hideo
1988-05-01
Experimental verification of the theoretically derived resistivity correction factor (RCF) is presented. Resistivity and sheet resistance measurements by the four-probe method are made on three samples: isotropic graphite, ITO film and Au film. It is indicated that the RCF can correct the apparent variations of experimental data to yield reasonable resistivities and sheet resistances.
MUTIL, Asymmetry Factor of Mott Cross-Sections for Electron, Positron Scattering
International Nuclear Information System (INIS)
Idoeta, R.; Legarda, F.
2002-01-01
1 - Description of program or function: The asymmetry factor S of Mott's differential cross section for the scattering of electrons and positrons by point nuclei without screening is calculated for any energy, atomic number and angle of scattering. 2 - Method of solution: We have summed the conditionally convergent series, F and G, appearing in the asymmetry factor using two consecutive transformations: The one of Yennie, Ravenhall and Wilson and that of Euler till we have seven times six significant figures repeated in the factor S. 3 - Restrictions on the complexity of the problem: Those appearing in the use of Mott's cross section for unscreened point nuclei
International Nuclear Information System (INIS)
Häggström, I; Karlsson, M; Larsson, A; Schmidtlein, C
2014-01-01
Purpose: To investigate the effects of corrections for random and scattered coincidences on kinetic parameters in brain tumors, by using ten Monte Carlo (MC) simulated dynamic FLT-PET brain scans. Methods: The GATE MC software was used to simulate ten repetitions of a 1 hour dynamic FLT-PET scan of a voxelized head phantom. The phantom comprised six normal head tissues, plus inserted regions for blood and tumor tissue. Different time-activity-curves (TACs) for all eight tissue types were used in the simulation and were generated in Matlab using a 2-tissue model with preset parameter values (K1,k2,k3,k4,Va,Ki). The PET data was reconstructed into 28 frames by both ordered-subset expectation maximization (OSEM) and 3D filtered back-projection (3DFBP). Five image sets were reconstructed, all with normalization and different additional corrections C (A=attenuation, R=random, S=scatter): Trues (AC), trues+randoms (ARC), trues+scatters (ASC), total counts (ARSC) and total counts (AC). Corrections for randoms and scatters were based on real random and scatter sinograms that were back-projected, blurred and then forward projected and scaled to match the real counts. Weighted non-linearleast- squares fitting of TACs from the blood and tumor regions was used to obtain parameter estimates. Results: The bias was not significantly different for trues (AC), trues+randoms (ARC), trues+scatters (ASC) and total counts (ARSC) for either 3DFBP or OSEM (p<0.05). Total counts with only AC stood out however, with an up to 160% larger bias. In general, there was no difference in bias found between 3DFBP and OSEM, except in parameter Va and Ki. Conclusion: According to our results, the methodology of correcting the PET data for randoms and scatters performed well for the dynamic images where frames have much lower counts compared to static images. Generally, no bias was introduced by the corrections and their importance was emphasized since omitting them increased bias extensively
Tuneable diode laser spectroscopy correction factor investigation on ammonia measurement
Li, Nilton; El-Hamalawi, Ashraf; Baxter, Jim; Barrett, Richard; Wheatley, Andrew
2018-01-01
Current diesel engine aftertreatment systems, such as Selective Catalyst Reduction (SCR) use ammonia (NH3) to reduce Nitrogen Oxides (NOx) into Nitrogen (N2) and water (H2O). However, if the reaction between NH3 and NOx is unbalanced, it can lead either NH3 or NOx being released into the environment. As NH3 is classified as a dangerous compound in the environment, its accurate measurement is essential. Tuneable Diode Laser (TDL) spectroscopy is one of the methods used to measure raw emissions inside engine exhaust pipes, especially NH3. This instrument requires a real-time exhaust temperature, pressure and other interference compounds in order to adjust itself to reduce the error in NH3 readings. Most researchers believed that exhaust temperature and pressure were the most influential factors in TDL when measuring NH3 inside exhaust pipes. The aim of this paper was to quantify these interference effects on TDL when undertaking NH3 measurement. Surprisingly, the results show that pressure was the least influential factor when compared to temperature, H2O, CO2 and O2 when undertaking NH3 measurement using TDL.
Kos, L; Aronzon, A; Takayama, H; Maina, F; Ponzetto, C; Merlino, G; Pavan, W
1999-02-01
The mechanisms governing development of neural crest-derived melanocytes, and how alterations in these pathways lead to hypopigmentation disorders, are not completely understood. Hepatocyte growth factor/scatter factor (HGF/SF) signaling through the tyrosine-kinase receptor, MET, is capable of promoting the proliferation, increasing the motility, and maintaining high tyrosinase activity and melanin synthesis of melanocytes in vitro. In addition, transgenic mice that ubiquitously overexpress HGF/SF demonstrate hyperpigmentation in the skin and leptomenigenes and develop melanomas. To investigate whether HGF/ SF-MET signaling is involved in the development of neural crest-derived melanocytes, transgenic embryos, ubiquitously overexpressing HGF/SF, were analyzed. In HGF/SF transgenic embryos, the distribution of melanoblasts along the characteristic migratory pathway was not affected. However, additional ectopically localized melanoblasts were also observed in the dorsal root ganglia and neural tube, as early as 11.5 days post coitus (p.c.). We utilized an in vitro neural crest culture assay to further explore the role of HGF/SF-MET signaling in neural crest development. HGF/SF added to neural crest cultures increased melanoblast number, permitted differentiation into pigmented melanocytes, promoted melanoblast survival, and could replace mast-cell growth factor/Steel factor (MGF) in explant cultures. To examine whether HGF/SF-MET signaling is required for the proper development of melanocytes, embryos with a targeted Met null mutation (Met-/-) were analysed. In Met-/- embryos, melanoblast number and location were not overtly affected up to 14 days p.c. These results demonstrate that HGF/SF-MET signaling influences, but is not required for, the initial development of neural crest-derived melanocytes in vivo and in vitro.
Palasantzas, G.
2008-01-01
In this work we study the quality factor associated with dissipation due to scattering of shear horizontal surface acoustic waves by random self-affine roughness. It is shown that the quality factor is strongly influenced by both the surface roughness exponent H and the roughness amplitude w to
International Nuclear Information System (INIS)
Stimpson, Shane; Collins, Benjamin; Kochunas, Brendan
2017-01-01
The MPACT code, being developed collaboratively by the University of Michigan and Oak Ridge National Laboratory, is the primary deterministic neutron transport solver being deployed within the Virtual Environment for Reactor Applications (VERA) as part of the Consortium for Advanced Simulation of Light Water Reactors (CASL). In many applications of the MPACT code, transport-corrected scattering has proven to be an obstacle in terms of stability, and considerable effort has been made to try to resolve the convergence issues that arise from it. Most of the convergence problems seem related to the transport-corrected cross sections, particularly when used in the 2D method of characteristics (MOC) solver, which is the focus of this work. Here in this paper, the stability and performance of the 2-D MOC solver in MPACT is evaluated for two iteration schemes: Gauss-Seidel and Jacobi. With the Gauss-Seidel approach, as the MOC solver loops over groups, it uses the flux solution from the previous group to construct the inscatter source for the next group. Alternatively, the Jacobi approach uses only the fluxes from the previous outer iteration to determine the inscatter source for each group. Consequently for the Jacobi iteration, the loop over groups can be moved from the outermost loop-as is the case with the Gauss-Seidel sweeper-to the innermost loop, allowing for a substantial increase in efficiency by minimizing the overhead of retrieving segment, region, and surface index information from the ray tracing data. Several test problems are assessed: (1) Babcock & Wilcox 1810 Core I, (2) Dimple S01A-Sq, (3) VERA Progression Problem 5a, and (4) VERA Problem 2a. The Jacobi iteration exhibits better stability than Gauss-Seidel, allowing for converged solutions to be obtained over a much wider range of iteration control parameters. Additionally, the MOC solve time with the Jacobi approach is roughly 2.0-2.5× faster per sweep. While the performance and stability of the Jacobi
Improvements to the Chebyshev expansion of attenuation correction factors for cylindrical samples
International Nuclear Information System (INIS)
Mildner, D.F.R.; Carpenter, J.M.
1990-01-01
The accuracy of the Chebyshev expansion coefficients used for the calculation of attenuation correction factors for cylinderical samples has been improved. An increased order of expansion allows the method to be useful over a greater range of attenuation. It is shown that many of these coefficients are exactly zero, others are rational numbers, and others are rational frations of π -1 . The assumptions of Sears in his asymptotic expression of the attenuation correction factor are also examined. (orig.)
Evaluation of the uniformity of wide circular reference source and application of correction factors
International Nuclear Information System (INIS)
Silva Junior, I.A.; Xavier, M.; Siqueira, P.T.D.; Sordi, G.A.A.; Potiens, M.P.A.
2017-01-01
In this work the uniformity of wide circular reference sources is evaluated. This kind of reference source is still widely used in Brazil. In previous works wide rectangular reference sources were analyzed and it was shown the importance of the application of correction factors in calibration procedures of radiation monitors. Now a transposition of the methods used formerly is performed, evaluating the uniformities of circular reference sources and calculating the associated correction factors. (author)
Chen, X. W.; Zhao, C. Y.; Wang, B. X.
2018-05-01
Thermal barrier coatings are common porous materials coated on the surface of devices operating under high temperatures and designed for heat insulation. This study presents a comprehensive investigation on the microstructural effect on radiative scattering coefficient and asymmetry factor of anisotropic thermal barrier coatings. Based on the quartet structure generation set algorithm, the finite-difference-time-domain method is applied to calculate angular scattering intensity distribution of complicated random microstructure, which takes wave nature into account. Combining Monte Carlo method with Particle Swarm Optimization, asymmetry factor, scattering coefficient and absorption coefficient are retrieved simultaneously. The retrieved radiative properties are identified with the angular scattering intensity distribution under different pore shapes, which takes dependent scattering and anisotropic pore shape into account implicitly. It has been found that microstructure significantly affects the radiative properties in thermal barrier coatings. Compared with spherical shape, irregular anisotropic pore shape reduces the forward scattering peak. The method used in this paper can also be applied to other porous media, which designs a frame work for further quantitative study on porous media.
Correction factors for 13C-labelled substrate oxidation at whole-body and muscle level
DEFF Research Database (Denmark)
Van Hall, Gerrit
1999-01-01
acid cycle. Changes in metabolic rate induced, for example, by feeding, hormonal changes and physical activity, as well as infusion time, have been shown to affect both correction factors. The present paper explains the theoretical and physiological basis of these correction factors and makes...... for the proportion of labelled CO2 that is produced via oxidation but not excreted. Furthermore, depending on the substrate and position of the C label(s), there may also be a need to correct for labelled C from the metabolized substrate that does not appear as CO2, but rather becomes temporarily fixed in other...
International Nuclear Information System (INIS)
Freitas, B.M.; Silva, A.X. da
2014-01-01
The Instituto de Radioprotecao e Dosimetria (IRD) runs a neutron individual monitoring service with albedo type monitor and thermoluminescent detectors (TLD). Moreover the largest number of workers exposed to neutrons in Brazil is exposed to 241 Am-Be fields. Therefore a study of the response of albedo dosemeter due to neutron scattering from 241 Am-Be source is important for a proper calibration. In this work, it has been evaluated the influence of the scattering correction in two distances at the Low Scattering Laboratory of the Neutron Laboratory of the Brazilian National Laboratory (Lab. Nacional de Metrologia Brasileira de Radiacoes Ionizantes) in the calibration of that albedo dosemeter for a 241 Am-Be source. (author)
Energy Technology Data Exchange (ETDEWEB)
Abdallah, A M; Elsherbiny, E M; Sobhy, M [Reactor departement, nuclear research centre, Inshaas, (Egypt)
1995-10-01
The P{sub n}-spatial expansion method has been used for calculating the one speed transport utilization factor in heterogenous slab cells in which neutrons may scatter anisotropically; by considering the P{sup 1-} approximation with a two-term scattering kernel in both the fuel and moderator regions, an analytical expression for the disadvantage factor has been derived. The numerical results obtained have been shown to be much better than those calculated by the usual P{sup 1-} and P{sup 3-} approximations and comparable with those obtained by some exact methods. 3 tabs.
Robert, Mark E; Linthicum, Fred H
2016-01-01
Profile count method for estimating cell number in sectioned tissue applies a correction factor for double count (resulting from transection during sectioning) of count units selected to represent the cell. For human spiral ganglion cell counts, we attempted to address apparent confusion between published correction factors for nucleus and nucleolus count units that are identical despite the role of count unit diameter in a commonly used correction factor formula. We examined a portion of human cochlea to empirically derive correction factors for the 2 count units, using 3-dimensional reconstruction software to identify double counts. The Neurotology and House Histological Temporal Bone Laboratory at University of California at Los Angeles. Using a fully sectioned and stained human temporal bone, we identified and generated digital images of sections of the modiolar region of the lower first turn of cochlea, identified count units with a light microscope, labeled them on corresponding digital sections, and used 3-dimensional reconstruction software to identify double-counted count units. For 25 consecutive sections, we determined that double-count correction factors for nucleus count unit (0.91) and nucleolus count unit (0.92) matched the published factors. We discovered that nuclei and, therefore, spiral ganglion cells were undercounted by 6.3% when using nucleolus count units. We determined that correction factors for count units must include an element for undercounting spiral ganglion cells as well as the double-count element. We recommend a correction factor of 0.91 for the nucleus count unit and 0.98 for the nucleolus count unit when using 20-µm sections. © American Academy of Otolaryngology—Head and Neck Surgery Foundation 2015.
International Nuclear Information System (INIS)
Hayashi, Naoki; Shibamoto, Yuta; Obata, Yasunori; Kimura, Takashi; Nakazawa, Hisato; Hagiwara, Masahiro; Hashizume, Chisa I.; Mori, Yoshimasa; Kobayashi, Tatsuya
2010-01-01
The purpose of this study was to evaluate the effect of megavoltage photon beam attenuation (PBA) by couch tops and to propose a method for correction of PBA. Four series of phantom measurements were carried out. First, PBA by the exact couch top (ECT, Varian) and Imaging Couch Top (ICT, BrainLAB) was evaluated using a water-equivalent phantom. Second, PBA by Type-S system (Med-Tec), ECT and ICT was compared with a spherical phantom. Third, percentage depth dose (PDD) after passing through ICT was measured to compare with control data of PDD. Forth, the gantry angle dependency of PBA by ICT was evaluated. Then, an equation for PBA correction was elaborated and correction factors for PBA at isocenter were obtained. Finally, this method was applied to a patient with hepatoma. PBA of perpendicular beams by ICT was 4.7% on average. With the increase in field size, the measured values became higher. PBA by ICT was greater than that by Type-S system and ECT. PBA increased significantly as the angle of incidence increased, ranging from 4.3% at 180 deg to 11.2% at 120 deg. Calculated doses obtained by the equation and correction factors agreed quite well with the measured doses between 120 deg and 180 deg of angles of incidence. Also in the patient, PBA by ICT was corrected quite well by the equation and correction factors. In conclusion, PBA and its gantry angle dependency by ICT were observed. This simple method using the equation and correction factors appeared useful to correct the isocenter dose when the PBA effect cannot be corrected by a treatment planning system. (author)
The determination of beam quality correction factors: Monte Carlo simulations and measurements.
González-Castaño, D M; Hartmann, G H; Sánchez-Doblado, F; Gómez, F; Kapsch, R-P; Pena, J; Capote, R
2009-08-07
Modern dosimetry protocols are based on the use of ionization chambers provided with a calibration factor in terms of absorbed dose to water. The basic formula to determine the absorbed dose at a user's beam contains the well-known beam quality correction factor that is required whenever the quality of radiation used at calibration differs from that of the user's radiation. The dosimetry protocols describe the whole ionization chamber calibration procedure and include tabulated beam quality correction factors which refer to 60Co gamma radiation used as calibration quality. They have been calculated for a series of ionization chambers and radiation qualities based on formulae, which are also described in the protocols. In the case of high-energy photon beams, the relative standard uncertainty of the beam quality correction factor is estimated to amount to 1%. In the present work, two alternative methods to determine beam quality correction factors are prescribed-Monte Carlo simulation using the EGSnrc system and an experimental method based on a comparison with a reference chamber. Both Monte Carlo calculations and ratio measurements were carried out for nine chambers at several radiation beams. Four chamber types are not included in the current dosimetry protocols. Beam quality corrections for the reference chamber at two beam qualities were also measured using a calorimeter at a PTB Primary Standards Dosimetry Laboratory. Good agreement between the Monte Carlo calculated (1% uncertainty) and measured (0.5% uncertainty) beam quality correction factors was obtained. Based on these results we propose that beam quality correction factors can be generated both by measurements and by the Monte Carlo simulations with an uncertainty at least comparable to that given in current dosimetry protocols.
Separating form factor and nuclear model effects in quasielastic neutrino-nucleus scattering
Wieske, Joseph
2017-09-01
When studying neutrino oscillations an understanding of charged current quasielastic (CCQE) neutrino-nucleus scattering is imperative. This interaction depends on a nuclear model as well as knowledge of form factors. In the past, CCQE data from the MiniBooNE experiment was analyzed assuming the Relativistic Fermi Gas (RFG) nuclear model, an axial dipole form factor in, and using the the z-expansion for the axial form factor in. We present the first analysis that combines a non-RFG nuclear model, in particular the Correlated Fermi Gas nuclear model (CFG) of, and the z expansion for the axial form factor. This will allow us to separate form factor and nuclear model effects in CCQE scattering. This project was supported through the Wayne State University REU program under NSF Grant PHY-1460853 and by the DOE Grant DE-SC0007983.
An investigation of accelerator head scatter and output factor in air
International Nuclear Information System (INIS)
Ding, George X.
2004-01-01
Our purpose in this study was to investigate whether the Monte Carlo simulation can accurately predict output factors in air. Secondary goals were to study the head scatter components and investigate the collimator exchange effect. The Monte Carlo code, BEAMnrc, was used in the study. Photon beams of 6 and 18 MV were from a Varian Clinac 2100EX accelerator and the measurements were performed using an ionization chamber in a mini-phantom. The Monte Carlo calculated in air output factors was within 1% of measured values. The simulation provided information of the origin and the magnitude of the collimator exchange effect. It was shown that the collimator backscatter to the beam monitor chamber played a significant role in the beam output factors. However the magnitude of the scattered dose contributions from the collimator at the isocenter is negligible. The maximum scattered dose contribution from the collimators was about 0.15% and 0.4% of the total dose at the isocenter for a 6 and 18 MV beam, respectively. The scattered dose contributions from the flattening filter at the isocenter were about 0.9-3% and 0.2-6% of the total dose for field sizes of 4x4 cm 2 -40x40 cm 2 for the 6 and 18 MV beam, respectively. The study suggests that measurements of head scatter factors be done at large depth well beyond the depth of electron contamination. The insight information may have some implications for developing generalized empirical models to calculate the head scatter
Relativistic and QED corrections to the g factor of Li-like ions
International Nuclear Information System (INIS)
Glazov, D.A.; Shabaev, V.M.; Volotka, A.V.; Tupitsyn, I.I.; Yerokhin, V.A.; Plunien, G.; Soff, G.
2004-01-01
Calculations of various corrections to the g factor of Li-like ions are presented, which result in a significant improvement of the theoretical accuracy in the region Z=6-92. The configuration-interaction Dirac-Fock method is employed for the evaluation of the interelectronic-interaction correction of order 1/Z 2 and higher. This correction is combined with the 1/Z interelectronic-interaction term derived within a rigorous QED approach. The one-electron QED correction of first order in α is obtained by employing our recent results for the self-energy term and by evaluating the vacuum-polarization contribution. The screening of QED corrections is taken into account to the leading orders in αZ and 1/Z
International Nuclear Information System (INIS)
Yan Guanghua; Liu, Chihray; Lu Bo; Palta, Jatinder R; Li, Jonathan G
2008-01-01
The purpose of this study was to choose an appropriate head scatter source model for the fast and accurate independent planar dose calculation for intensity-modulated radiation therapy (IMRT) with MLC. The performance of three different head scatter source models regarding their ability to model head scatter and facilitate planar dose calculation was evaluated. A three-source model, a two-source model and a single-source model were compared in this study. In the planar dose calculation algorithm, in-air fluence distribution was derived from each of the head scatter source models while considering the combination of Jaw and MLC opening. Fluence perturbations due to tongue-and-groove effect, rounded leaf end and leaf transmission were taken into account explicitly. The dose distribution was calculated by convolving the in-air fluence distribution with an experimentally determined pencil-beam kernel. The results were compared with measurements using a diode array and passing rates with 2%/2 mm and 3%/3 mm criteria were reported. It was found that the two-source model achieved the best agreement on head scatter factor calculation. The three-source model and single-source model underestimated head scatter factors for certain symmetric rectangular fields and asymmetric fields, but similar good agreement could be achieved when monitor back scatter effect was incorporated explicitly. All the three source models resulted in comparable average passing rates (>97%) when the 3%/3 mm criterion was selected. The calculation with the single-source model and two-source model was slightly faster than the three-source model due to their simplicity
Energy Technology Data Exchange (ETDEWEB)
Yan Guanghua [Department of Nuclear and Radiological Engineering, University of Florida, Gainesville, FL 32611 (United States); Liu, Chihray; Lu Bo; Palta, Jatinder R; Li, Jonathan G [Department of Radiation Oncology, University of Florida, Gainesville, FL 32610-0385 (United States)
2008-04-21
The purpose of this study was to choose an appropriate head scatter source model for the fast and accurate independent planar dose calculation for intensity-modulated radiation therapy (IMRT) with MLC. The performance of three different head scatter source models regarding their ability to model head scatter and facilitate planar dose calculation was evaluated. A three-source model, a two-source model and a single-source model were compared in this study. In the planar dose calculation algorithm, in-air fluence distribution was derived from each of the head scatter source models while considering the combination of Jaw and MLC opening. Fluence perturbations due to tongue-and-groove effect, rounded leaf end and leaf transmission were taken into account explicitly. The dose distribution was calculated by convolving the in-air fluence distribution with an experimentally determined pencil-beam kernel. The results were compared with measurements using a diode array and passing rates with 2%/2 mm and 3%/3 mm criteria were reported. It was found that the two-source model achieved the best agreement on head scatter factor calculation. The three-source model and single-source model underestimated head scatter factors for certain symmetric rectangular fields and asymmetric fields, but similar good agreement could be achieved when monitor back scatter effect was incorporated explicitly. All the three source models resulted in comparable average passing rates (>97%) when the 3%/3 mm criterion was selected. The calculation with the single-source model and two-source model was slightly faster than the three-source model due to their simplicity.
Relativistic corrections to the form factors of Bc into P-wave orbitally excited charmonium
Zhu, Ruilin
2018-06-01
We investigated the form factors of the Bc meson into P-wave orbitally excited charmonium using the nonrelativistic QCD effective theory. Through the analytic computation, the next-to-leading order relativistic corrections to the form factors were obtained, and the asymptotic expressions were studied in the infinite bottom quark mass limit. Employing the general form factors, we discussed the exclusive decays of the Bc meson into P-wave orbitally excited charmonium and a light meson. We found that the relativistic corrections lead to a large correction for the form factors, which makes the branching ratios of the decay channels B (Bc ± →χcJ (hc) +π± (K±)) larger. These results are useful for the phenomenological analysis of the Bc meson decays into P-wave charmonium, which shall be tested in the LHCb experiments.
International Nuclear Information System (INIS)
Singh, Sukhpal; Kumar, Ashok; Singh, Charanjeet; Thind, Kulwant Singh; Mudahar, Gurmel S.
2008-01-01
The simultaneous variation of gamma ray buildup factors with absorber thickness (up to 6.5 mfp) and total scatter acceptance angle (which is the sum of incidence and exit beam divergence) in the media of high volume flyash concrete and water was studied experimentally using a point isotropic 137 Cs source
Atti, C. Ciofi degli; Pace, E.; Salmé, G.
1993-01-01
The theory of inclusive lepton scattering of polarized leptons by polarized J = 1/2 hadrons is presented and the origin of different expressions for the polarized nuclear response function appearing in the literature is explained. The sensitivity of the longitudinal asymmetry upon the neutron form factors is investigated.
SU-F-T-67: Correction Factors for Monitor Unit Verification of Clinical Electron Beams
Energy Technology Data Exchange (ETDEWEB)
Haywood, J [Mercy Health Partners, Muskegon, MI (United States)
2016-06-15
Purpose: Monitor units calculated by electron Monte Carlo treatment planning systems are often higher than TG-71 hand calculations for a majority of patients. Here I’ve calculated tables of geometry and heterogeneity correction factors for correcting electron hand calculations. Method: A flat water phantom with spherical volumes having radii ranging from 3 to 15 cm was created. The spheres were centered with respect to the flat water phantom, and all shapes shared a surface at 100 cm SSD. D{sub max} dose at 100 cm SSD was calculated for each cone and energy on the flat phantom and for the spherical volumes in the absence of the flat phantom. The ratio of dose in the sphere to dose in the flat phantom defined the geometrical correction factor. The heterogeneity factors were then calculated from the unrestricted collisional stopping power for tissues encountered in electron beam treatments. These factors were then used in patient second check calculations. Patient curvature was estimated by the largest sphere that aligns to the patient contour, and appropriate tissue density was read from the physical properties provided by the CT. The resulting MU were compared to those calculated by the treatment planning system and TG-71 hand calculations. Results: The geometry and heterogeneity correction factors range from ∼(0.8–1.0) and ∼(0.9–1.01) respectively for the energies and cones presented. Percent differences for TG-71 hand calculations drop from ∼(3–14)% to ∼(0–2)%. Conclusion: Monitor units calculated with the correction factors typically decrease the percent difference to under actionable levels, < 5%. While these correction factors work for a majority of patients, there are some patient anatomies that do not fit the assumptions made. Using these factors in hand calculations is a first step in bringing the verification monitor units into agreement with the treatment planning system MU.
Energy Technology Data Exchange (ETDEWEB)
Sorriaux, J; Lee, J [Molecular Imaging Radiotherapy & Oncology, Universite Catholique de Louvain, Brussels (Belgium); ICTEAM Institute, Universite catholique de Louvain, Louvain-la-Neuve (Belgium); Testa, M; Paganetti, H [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, MA 02114, Massachusetts (United States); Bertrand, D; Orban de Xivry, J [Ion Beam Applications, Louvain-la-neuve, Brabant Wallon (Belgium); Palmans, H [EBG MedAustron GmbH, Wiener Neustadt (Austria); National Physical Laboratory, Teddington (United Kingdom); Vynckier, S [Cliniques Universitaires Saint-Luc, Brussels (Belgium); Sterpin, E [Molecular Imaging Radiotherapy & Oncology, Universite Catholique de Louvain, Brussels (Belgium)
2015-06-15
Purpose: The IAEA TRS-398 code of practice details the reference conditions for reference dosimetry of proton beams using ionization chambers and the required beam quality correction factors (kQ). Pencil beam scanning (PBS) requires multiple spots to reproduce the reference conditions. The objective is to demonstrate, using Monte Carlo (MC) calculations, that kQ factors for broad beams can be used for scanned beams under the same reference conditions with no significant additional uncertainty. We consider hereafter the general Alfonso formalism (Alfonso et al, 2008) for non-standard beam. Methods: To approach the reference conditions and the associated dose distributions, PBS must combine many pencil beams with range modulation and shaping techniques different than those used in passive systems (broad beams). This might lead to a different energy spectrum at the measurement point. In order to evaluate the impact of these differences on kQ factors, ion chamber responses are computed with MC (Geant4 9.6) in a dedicated scanned pencil beam (Q-pcsr) producing a 10×10cm2 composite field with a flat dose distribution from 10 to 16 cm depth. Ion chamber responses are also computed by MC in a broad beam with quality Q-ds (double scattering). The dose distribution of Q -pcsr matches the dose distribution of Q-ds. k-(Q-pcsr,Q-ds) is computed for a 2×2×0.2cm{sup 3} idealized air cavity and a realistic plane-parallel ion chamber (IC). Results: Under reference conditions, quality correction factors for a scanned composite field versus a broad beam are the same for air cavity dose response, k-(Q-pcsr,Q-ds) =1.001±0.001 and for a Roos IC, k-(Q-pcsr,Q-ds) =0.999±0.005. Conclusion: Quality correction factors for ion chamber response in scanned and broad proton therapy beams are identical under reference conditions within the calculation uncertainties. The results indicate that quality correction factors published in IAEA TRS-398 can be used for scanned beams in the SOBP of a
International Nuclear Information System (INIS)
Sorriaux, J; Lee, J; Testa, M; Paganetti, H; Bertrand, D; Orban de Xivry, J; Palmans, H; Vynckier, S; Sterpin, E
2015-01-01
Purpose: The IAEA TRS-398 code of practice details the reference conditions for reference dosimetry of proton beams using ionization chambers and the required beam quality correction factors (kQ). Pencil beam scanning (PBS) requires multiple spots to reproduce the reference conditions. The objective is to demonstrate, using Monte Carlo (MC) calculations, that kQ factors for broad beams can be used for scanned beams under the same reference conditions with no significant additional uncertainty. We consider hereafter the general Alfonso formalism (Alfonso et al, 2008) for non-standard beam. Methods: To approach the reference conditions and the associated dose distributions, PBS must combine many pencil beams with range modulation and shaping techniques different than those used in passive systems (broad beams). This might lead to a different energy spectrum at the measurement point. In order to evaluate the impact of these differences on kQ factors, ion chamber responses are computed with MC (Geant4 9.6) in a dedicated scanned pencil beam (Q-pcsr) producing a 10×10cm2 composite field with a flat dose distribution from 10 to 16 cm depth. Ion chamber responses are also computed by MC in a broad beam with quality Q-ds (double scattering). The dose distribution of Q -pcsr matches the dose distribution of Q-ds. k-(Q-pcsr,Q-ds) is computed for a 2×2×0.2cm 3 idealized air cavity and a realistic plane-parallel ion chamber (IC). Results: Under reference conditions, quality correction factors for a scanned composite field versus a broad beam are the same for air cavity dose response, k-(Q-pcsr,Q-ds) =1.001±0.001 and for a Roos IC, k-(Q-pcsr,Q-ds) =0.999±0.005. Conclusion: Quality correction factors for ion chamber response in scanned and broad proton therapy beams are identical under reference conditions within the calculation uncertainties. The results indicate that quality correction factors published in IAEA TRS-398 can be used for scanned beams in the SOBP of a high
A study of coarse mesh collision probability correction factors in slab lattices
International Nuclear Information System (INIS)
Buckler, A.N.
1975-07-01
Calculations of collision probability leakage estimates are performed in one dimensional slab geometry with one neutron group to gain some insight into methods of correction for the coarseness of the mesh H. The chief result is that the correction factor, beta, can be written as CD/H where C → 4 for the diffusion limit. An explicit expression for C is derived in terms of the E 3 function, for a linear flux variation across the slabs. (author)
Zhou, Lihong; Yuan, Liming; Thomas, Rick; Iannacchione, Anthony
2017-01-01
When there are installations of air velocity sensors in the mining industry for real-time airflow monitoring, a problem exists with how the monitored air velocity at a fixed location corresponds to the average air velocity, which is used to determine the volume flow rate of air in an entry with the cross-sectional area. Correction factors have been practically employed to convert a measured centerline air velocity to the average air velocity. However, studies on the recommended correction fac...
International Nuclear Information System (INIS)
Peschar, R.; Schenk, H.
1991-01-01
A method to derive joint probability distributions of structure factors is presented which incorporates anomalous-scattering and isomorphous-replacement data in a unified procedure. The structure factors F H and F -H , whose magnitudes are different due to anomalous scattering, are shown to be isomorphously related. This leads to a definition of isomorphism by means of which isomorphous-replacement and anomalous-scattering data can be handled simultaneously. The definition and calculation of the general term of the joint probability distribution for isomorphous structure factors turns out to be crucial. Its analytical form leads to an algorithm by means of which any particular joint probability distribution of structure factors can be constructed. The calculation of the general term is discussed for the case of four isomorphous structure factors in P1, assuming the atoms to be independently and uniformly distributed. A main result is the construction of the probability distribution of the 64 triplet phase sums present in space group P1 amongst four isomorphous structure factors F H , four isomorphous F K and four isomorphous F -H-K . The procedure is readily generalized in the case where an arbitrary number of isomorphous structure factors are available for F H , F K and F -H-K . (orig.)
Energy Technology Data Exchange (ETDEWEB)
Shiga, Tohru; Takano, Akihiro; Tsukamoto, Eriko; Tamaki, Nagara [Department of Nuclear Medicine, Hokkaido University School of Medicine, Sapporo (Japan); Kubo, Naoki [Department of Radiological Technology, College of Medical Technology, Hokkaido University, Sapporo (Japan); Kobayashi, Junko; Takeda, Yoji; Nakamura, Fumihiro; Koyama, Tsukasa [Department of Psychiatry and Neurology, Hokkaido University School of Medicine, Sapporo (Japan); Katoh, Chietsugu [Department of Tracer Kinetics, Hokkaido University School of Medicine, Sapporo (Japan)
2002-03-01
Scatter correction (SC) using the triple energy window method (TEW) has recently been applied for brain perfusion single-photon emission tomography (SPET). The aim of this study was to investigate the effect of scatter correction using TEW on N-isopropyl-p-[{sup 123}I]iodoamphetamine ({sup 123}I-IMP) SPET in normal subjects. The study population consisted of 15 right-handed normal subjects. SPET data were acquired from 20 min to 40 min after the injection of 167 MBq of IMP, using a triple-head gamma camera. Images were reconstructed with and without SC. 3D T1-weighted magnetic resonance (MR) images were also obtained with a 1.5-Tesla scanner. First, IMP images with and without SC were co-registered to the 3D MRI. Second, the two co-registered IMP images were normalised using SPM96. A t statistic image for the contrast condition effect was constructed. We investigated areas using a voxel-level threshold of 0.001, with a corrected threshold of 0.05. Compared with results obtained without SC, the IMP distribution with SC was significantly decreased in the peripheral areas of the cerebellum, the cortex and the ventricle, and also in the lateral occipital cortex and the base of the temporal lobe. On the other hand, the IMP distribution with SC was significantly increased in the anterior and posterior cingulate cortex, the insular cortex and the medial part of the thalamus. It is concluded that differences in the IMP distribution with and without SC exist not only in the peripheral areas of the cerebellum, the cortex and the ventricle but also in the occipital lobe, the base of the temporal lobe, the insular cortex, the medial part of the thalamus, and the anterior and posterior cingulate cortex. This needs to be recognised for adequate interpretation of IMP brain perfusion SPET after scatter correction. (orig.)
Corrections to the bag factor in B- anti B-mixing
Energy Technology Data Exchange (ETDEWEB)
Grozin, Andrey G. [BINP, Novosibirsk (Russian Federation); Klein, Rebecca; Mannel, Thomas; Pivovarov, Alexei A. [Universitaet Siegen (Germany)
2016-07-01
B- anti B-Mixing is parameterized by the matrix elements of local operators O{sub i}. For the computation of these matrix elements a bag factor B{sub i} can be introduced left angle anti B vertical stroke O{sub i} vertical stroke B right angle =B{sub i} left angle anti B vertical stroke O{sub i} vertical stroke B right angle {sup fac}, which is unity in naive factorization. Any deviation from unity describes the accuracy of the naive factorization prescription. Corrections to B{sub i} emerge from QCD radiative corrections and from nonperturbative contributions at order 1/m{sub b}. We present the current status of these corrections to B{sub i}.
Calculation of the Pitot tube correction factor for Newtonian and non-Newtonian fluids.
Etemad, S Gh; Thibault, J; Hashemabadi, S H
2003-10-01
This paper presents the numerical investigation performed to calculate the correction factor for Pitot tubes. The purely viscous non-Newtonian fluids with the power-law model constitutive equation were considered. It was shown that the power-law index, the Reynolds number, and the distance between the impact and static tubes have a major influence on the Pitot tube correction factor. The problem was solved for a wide range of these parameters. It was shown that employing Bernoulli's equation could lead to large errors, which depend on the magnitude of the kinetic energy and energy friction loss terms. A neural network model was used to correlate the correction factor of a Pitot tube as a function of these three parameters. This correlation is valid for most Newtonian, pseudoplastic, and dilatant fluids at low Reynolds number.
Energy Technology Data Exchange (ETDEWEB)
Talamo, Alberto, E-mail: alby@anl.gov [Argonne National Laboratory, 9700S. Cass Avenue, Argonne, IL 60439 (United States); Gohar, Y.; Cao, Y.; Zhong, Z. [Argonne National Laboratory, 9700S. Cass Avenue, Argonne, IL 60439 (United States); Kiyavitskaya, H.; Bournos, V.; Fokov, Y.; Routkovskaya, C. [Joint Institute for Power and Nuclear Research-Sosny, National Academy of Sciences of Belarus, 99 acad. Krasin str., Minsk 220109 (Belarus)
2012-03-11
In subcritical assemblies, the Bell and Glasstone spatial correction factor is used to correct the measured reactivity from different detector positions. In addition to the measuring position, several other parameters affect the correction factor: the detector material, the detector size, and the energy-angle distribution of source neutrons. The effective multiplication factor calculated by computer codes in criticality mode slightly differs from the average value obtained from the measurements in the different experimental channels of the subcritical assembly, which are corrected by the Bell and Glasstone spatial correction factor. Generally, this difference is due to (1) neutron counting errors; (2) geometrical imperfections, which are not simulated in the calculational model, and (3) quantities and distributions of material impurities, which are missing from the material definitions. This work examines these issues and it focuses on the detector choice and the calculation methodologies. The work investigated the YALINA Booster subcritical assembly of Belarus, which has been operated with three different fuel enrichments in the fast zone either: high (90%) and medium (36%), medium (36%), or low (21%) enriched uranium fuel.
International Nuclear Information System (INIS)
Talamo, Alberto; Gohar, Y.; Cao, Y.; Zhong, Z.; Kiyavitskaya, H.; Bournos, V.; Fokov, Y.; Routkovskaya, C.
2012-01-01
In subcritical assemblies, the Bell and Glasstone spatial correction factor is used to correct the measured reactivity from different detector positions. In addition to the measuring position, several other parameters affect the correction factor: the detector material, the detector size, and the energy-angle distribution of source neutrons. The effective multiplication factor calculated by computer codes in criticality mode slightly differs from the average value obtained from the measurements in the different experimental channels of the subcritical assembly, which are corrected by the Bell and Glasstone spatial correction factor. Generally, this difference is due to (1) neutron counting errors; (2) geometrical imperfections, which are not simulated in the calculational model, and (3) quantities and distributions of material impurities, which are missing from the material definitions. This work examines these issues and it focuses on the detector choice and the calculation methodologies. The work investigated the YALINA Booster subcritical assembly of Belarus, which has been operated with three different fuel enrichments in the fast zone either: high (90%) and medium (36%), medium (36%), or low (21%) enriched uranium fuel.
New look at displacement factor and point of measurement corrections in ionization chamber dosimetry
International Nuclear Information System (INIS)
Awschalom, M.; Rosenberg, I.; Ten Haken, R.K.
1983-01-01
A new technique is presented for determination of the effective point of measurement when cavity ionization chambers are used to measure the absorbed dose due to ionizing radiation in a dense medium. An algorithm is derived relating the effective point of measurement to the displacement correction factor. This algorithm relates variations of the displacement factor to the radiation field gradient. The technique is applied to derive the magnitudes of the corrections for several chambers in a p(66)Be(49) neutron therapy beam. 30 references, 4 figures, 1 table
Determination of the air attenuation correction factor for a free air ionization chamber
International Nuclear Information System (INIS)
Silva, Natalia F.; Cintra, Felipe B.; Castro, Maysa C. de; Caldas, Linda V.E.
2016-01-01
The objective of this work is to present the experimental and simulation results for the air attenuation correction factor for a free air ionization chamber with concentric cylinders of Victoreen, model 481-5. This correction factor was obtained for the standard mammography qualities established in the Instrument Calibration Laboratory (LCI) of IPEN. The values were compared with the results from the German primary standard laboratory Physikalisch- Technische Bundesanstalt (PTB), and maximum differences of 0.40% in relation to the experimental value and 0.31% in relation to the simulated value were obtained. (author)
International Nuclear Information System (INIS)
Ross, A.W.; Fink, M.
1986-01-01
An improved set of scattering factors for all neutral elements has bee completed for inclusion in the new edition of the International Tables of X-ray Crystallography . These calculations are compared with the former electron scattering factors and the deviations between the two are discussed
Braun, Frank; Schalk, Robert; Heintz, Annabell; Feike, Patrick; Firmowski, Sebastian; Beuermann, Thomas; Methner, Frank-Jürgen; Kränzlin, Bettina; Gretz, Norbert; Rädle, Matthias
2017-07-01
In this report, a quantitative nicotinamide adenine dinucleotide hydrate (NADH) fluorescence measurement algorithm in a liquid tissue phantom using a fiber-optic needle probe is presented. To determine the absolute concentrations of NADH in this phantom, the fluorescence emission spectra at 465 nm were corrected using diffuse reflectance spectroscopy between 600 nm and 940 nm. The patented autoclavable Nitinol needle probe enables the acquisition of multispectral backscattering measurements of ultraviolet, visible, near-infrared and fluorescence spectra. As a phantom, a suspension of calcium carbonate (Calcilit) and water with physiological NADH concentrations between 0 mmol l-1 and 2.0 mmol l-1 were used to mimic human tissue. The light scattering characteristics were adjusted to match the backscattering attributes of human skin by modifying the concentration of Calcilit. To correct the scattering effects caused by the matrices of the samples, an algorithm based on the backscattered remission spectrum was employed to compensate the influence of multiscattering on the optical pathway through the dispersed phase. The monitored backscattered visible light was used to correct the fluorescence spectra and thereby to determine the true NADH concentrations at unknown Calcilit concentrations. Despite the simplicity of the presented algorithm, the root-mean-square error of prediction (RMSEP) was 0.093 mmol l-1.
Cashmore, Aaron W; Indig, Devon; Hampton, Stephen E; Hegney, Desley G; Jalaludin, Bin B
2016-11-01
Little is known about the environmental and organisational determinants of workplace violence in correctional health settings. This paper describes the views of health professionals working in these settings on the factors influencing workplace violence risk. All employees of a large correctional health service in New South Wales, Australia, were invited to complete an online survey. The survey included an open-ended question seeking the views of participants about the factors influencing workplace violence in correctional health settings. Responses to this question were analysed using qualitative thematic analysis. Participants identified several factors that they felt reduced the risk of violence in their workplace, including: appropriate workplace health and safety policies and procedures; professionalism among health staff; the presence of prison guards and the quality of security provided; and physical barriers within clinics. Conversely, participants perceived workplace violence risk to be increased by: low health staff-to-patient and correctional officer-to-patient ratios; high workloads; insufficient or underperforming security staff; and poor management of violence, especially horizontal violence. The views of these participants should inform efforts to prevent workplace violence among correctional health professionals.
The Proton Coulomb Form Factor from Polarized Inclusive e-p Scattering
Energy Technology Data Exchange (ETDEWEB)
Harris, Christopher Matthew [Univ. of Virginia, Charlottesville, VA (United States)
2001-05-01
The proton form factors provide information on the fundamental properties of the proton and provide a test for models based on QCD. In 1998 at Jefferson Lab (JLAB) in Newport News, VA, experiment E93026 measured the inclusive e-p scattering cross section from a polarized ammonia (^{15}NH_{3}) target at a four momentum transfer squared of Q^{2} = 0.5 (GeV/c)^{2}. Longitudinally polarized electrons were scattered from the polarized target and the scattered electron was detected. Data has been analyzed to obtain the asymmetry from elastically scattered electrons from hydrogen in ^{15}NH_{3}. The asymmetry, A_{p}, has been used to determine the proton elastic form factor G_{Ep}. The result is consistent with the dipole model and data from previous experiments. However, due to the choice of kinematics, the uncertainty in the measurement is large.
Minet, Olaf; Scheibe, Patrick; Beuthan, Jürgen; Zabarylo, Urszula
2010-02-01
State-of-the-art image processing methods offer new possibilities for diagnosing diseases using scattered light. The optical diagnosis of rheumatism is taken as an example to show that the diagnostic sensitivity can be improved using overlapped pseudo-coloured images of different wavelengths, provided that multispectral images are recorded to compensate for any motion related artefacts which occur during examination.
International Nuclear Information System (INIS)
Hashimoto, Jun; Kubo, Atsushi; Ogawa, Koichi; Ichihara, Takashi; Motomura, Nobutoku; Takayama, Takuzo; Iwanaga, Shiro; Mitamura, Hideo; Ogawa, Satoshi
1998-01-01
A practical method for scatter and attenuation compensation was employed in thallium-201 myocardial single-photon emission tomography (SPET or ECT) with the triple-energy-window (TEW) technique and an iterative attenuation correction method by using a measured attenuation map. The map was reconstructed from technetium-99m transmission CT (TCT) data. A dual-headed SPET gamma camera system equipped with parallel-hole collimators was used for ECT/TCT data acquisition and a new type of external source named ''sheet line source'' was designed for TCT data acquisition. This sheet line source was composed of a narrow long fluoroplastic tube embedded in a rectangular acrylic board. After injection of 99m Tc solution into the tube by an automatic injector, the board was attached in front of the collimator surface of one of the two detectors. After acquiring emission and transmission data separately or simultaneously, we eliminated scattered photons in the transmission and emission data with the TEW method, and reconstructed both images. Then, the effect of attenuation in the scatter-corrected ECT images was compensated with Chang's iterative method by using measured attenuation maps. Our method was validated by several phantom studies and clinical cardiac studies. The method offered improved homogeneity in distribution of myocardial activity and accurate measurements of myocardial tracer uptake. We conclude that the above correction method is feasible because a new type of 99m Tc external source may not produce truncation in TCT images and is cost-effective and easy to prepare in clinical situations. (orig.)
Applicability of perturbative QCD and NLO power corrections for the pion form factor
International Nuclear Information System (INIS)
Yeh Tsungwen
2002-01-01
As is well recognized, the asymptotic of the perturbative QCD prediction for the pion form factor is much smaller than the upper end of the data. We investigate this problem. We first evaluate the next-to-leading-order (NLO) power correction for the pion form factor. The corrected form factor contains nonperturbative parameters which are determined from a χ 2 fit to the data. Interpreting these parameters leads to the fact that the involved strong interaction coupling constant should be identified as an effective coupling constant under a nonperturbative QCD vacuum. If the scale associated with the effective coupling constant is identified as 2 Q 2 , then Q 2 , the momentum transfer square for the pion form factor to be measured, can have a value about 1 GeV 2 , and , the averaged momentum fraction variable, can locate around 0.5. This circumstance is consistent with the asymptotic model for the pion wave function
International Nuclear Information System (INIS)
Ofori, Y. T.
2013-07-01
Ghana Research Reactor-1 (GHARR-1), an MNSR (Miniature Neutron Source Reactor) is to be converted from HEU (Highly Enriched Uranium) to LEU (Low Enriched Uranium) fuel, along with all current MNSRs in various other countries. The purpose of the conversion is to minimize the use of HEU for non-proliferation of high-grade nuclear fuel. In this research work, a comparative study has been performed for the determination of the Dancoff, thermal utilization and thermal disadvantage factors of highly enriched uranium (HEU) and potential low enriched uranium (LEU) cores of GHARR-1. A one group transport theory and collision probability based methodologies was used to develop mathematical formulations for thermal utilization factor and thermal disadvantage factor assuming isotropic scattering. This methodology was implemented in a FORTRAN 95 based computer program THERMCALC, which uses Bessell and BesselK as subroutines developed to calculate the modified Bessel functions I n and K n respectively using the polynomial approximation method. Furthermore, a Dancoff correction factor of 0.1519 thermal utilization factor of 0.9767 and a thermal disadvantage factor of 1.894 were obtained for the 90.2% highly enriched Uranium core of GHARR-1. The results compare favorably with literature. Thus THERMCALC can be used as a reliable tool for the calculation of Dancoff, thermal utilization and disadvantage factors of MNSR cores. Other potential LEU cores; UO 2 (with different fuel meat densities and enrichments) and U 3 Si 2 have also been analysed. UO 2 with 12.6% of Uranium-235 was chosen as the most potential LEU core for the GHARR-1. (au)
A Novel Bridgeless Power Factor Correction with Interleaved Boost Stages in Continous Current Mode
DEFF Research Database (Denmark)
Li, Qingnan; Andersen, Michael A. E.; Thomsen, Ole Cornelius
2011-01-01
The operation and trade-off of Bridgeless Power Factor Correction (BPFC) circuit with interleaved Boost stages are investigated. By using interleaved BPFC, an overall reduction of the size of EMI filter can be achieved without increasing the switching frequency of the converter. And higher...
75 FR 72739 - Compliance Testing Procedures: Correction Factor for Room Air Conditioners
2010-11-26
...: Correction Factor for Room Air Conditioners AGENCY: Office of the General Counsel, Department of Energy (DOE... air conditioners. The petition seeks temporary enforcement forbearance, or in the alternative, a... procedures for room air conditioners. Public comment is requested on whether DOE should grant the petition...
Conduction Losses and Common Mode EMI Analysis on Bridgeless Power Factor Correction
DEFF Research Database (Denmark)
Li, Qingnan; Andersen, Michael Andreas E.; Thomsen, Ole Cornelius
2009-01-01
In this paper, a review of Bridgeless Boost power factor correction (PFC) converters is presented at first. Performance comparison on conduction losses and common mode electromagnetic interference (EMI) are analyzed between conventional Boost PFC converter and members of Bridgeless PFC family...
Integrals of random fields treated by the model correction factor method
DEFF Research Database (Denmark)
Franchin, P.; Ditlevsen, Ove Dalager; Kiureghian, Armen Der
2002-01-01
The model correction factor method (MCFM) is used in conjunction with the first-order reliability method (FORM) to solve structural reliability problems involving integrals of non-Gaussian random fields. The approach replaces the limit-state function with an idealized one, in which the integrals ...
DEFF Research Database (Denmark)
Franchin, P.; Ditlevsen, Ove Dalager; Kiureghian, Armen Der
2002-01-01
The model correction factor method (MCFM) is used in conjunction with the first-order reliability method (FORM) to solve structural reliability problems involving integrals of non-Gaussian random fields. The approach replaces the limit-state function with an idealized one, in which the integrals ...
High efficiency three-phase power factor correction rectifier using SiC switches
DEFF Research Database (Denmark)
Kouchaki, Alireza; Nymand, Morten
2017-01-01
This paper presents designing procedure of a high efficiency 5 kW silicon-carbide (SiC) based threephase power factor correction (PFC). SiC switches present low capacitive switching loss compared to the alternative Si switches. Therefore, the switching frequency can be increased and hence the siz...
Dynamics of liquid N2 studied by neutron inelastic scattering
DEFF Research Database (Denmark)
Pedersen, Karen Schou; Carneiro, Kim; Hansen, Flemming Yssing
1982-01-01
Neutron inelastic-scattering data from liquid N2 at wave-vector transfer κ between 0.18 and 2.1 Å-1 and temperatures ranging from T=65-77 K are presented. The data are corrected for the contribution from multiple scattering and incoherent scattering. The resulting dynamic structure factor S (κ,ω)...
International Nuclear Information System (INIS)
Siegel, E.
1982-01-01
The generalized-disorder collective-boson mode-softening universality-principle (GDCBMSUP) for collective-boson mode dispersion in disordered systems (liquids, quantum liquids, glasses, powders, disordered magnets, plasmas...), a unified qualitative and semi-qualitative and semi-quantitative descriptive prescription for treating the properties of very differently disordered systems, is directly dependent upon a measurement (or calculation) of the static structure factor S(k) determined from a frequency average of the dynamic structure factor S(k,w), a multiple of the inelastic differential neutron scattering cross section d 2 sigma/dwdOMEGA. The prescription for this principle is given and, because of its universal applicability to disordered systems of any type with any type and/or degree of disorder, the neutron scattering determination of S(k) takes on renewed importance
On the Debye-Waller factor in atom-surface scattering
International Nuclear Information System (INIS)
Garcia, N.; Maradudin, A.A.; Celli, V.
1982-01-01
A theory for the Debye-Waller factor in atom-surface scattering is presented, to lowest order in the phonon contributions. Multiple-scattering effects as well as the cross-correlated surface atom displacements are included. The theory accounts for experimental data without the necessity of introducing the Armand effect, which is due to the finite size of the incident atom. The work presented here implies that the Kirchhoff approximation fails when the energy of the incident particle is in the energy range of the phonon spectrum. The results of the calculation are presented in the high-temperature limit, and it is observed that the Rayleigh surface phonons contribute three-quarters of the Debye-Waller factor, while the bulk phonons account for the rest. This result is interesting because the calculation of the former contribution is simpler than that of the latter. (author)
Advanteges of using Two-Switch Forward in Single-Stage Power Factor Corrected Power Supplies
DEFF Research Database (Denmark)
Petersen, Lars
2000-01-01
A single-Stage power factor corrected power supply using a two-switch forward is proposed to increase efficiency. The converter is operated in the DCM (Discontinues Conduction Mode). This will insure the intermediate DC-bus to be controlled only by means of circuit parameters and therefore...... power supply has been implemented. The measured efficiency and power factor are about 87% and 0.96 respectively....
International Nuclear Information System (INIS)
Alejo Luque, L.; Rodriguez Romero, R.; Castro Tejero, P.; Fandino Lareo, J. M.
2011-01-01
This paper discusses the measures and uncertainties of the correction factors for dose-polarization (k, 1) and recombination (k,) of different ionization chambers plane-parallel and cylindrical. The values ??have been obtained using photon and electron beams of various energies generated by linear accelerators nominal Varian 21EX CLJNAC Tomotherapy Hi-Art and JI. We study the cases in which you can avoid the application of the factors obtained, according to the criteria proposed
Baryon scattering at high energies. Wave function, impact factor, and gluon radiation
International Nuclear Information System (INIS)
Bartels, J.; Motyka, L.; Jagellonian Univ., Krakow
2007-11-01
The scattering of a baryon consisting of three massive quarks is investigated in the high energy limit of perturbative QCD. A model of a relativistic proton-like wave function, dependent on valence quark longitudinal and transverse momenta and on quark helicities, is proposed, and we derive the baryon impact factors for two, three and four t-channel gluons. We find that the baryonic impact factor can be written as a sum of three pieces: in the first one a subsystem consisting of two of the three quarks behaves very much like the quark-antiquark pair in γ * scattering, whereas the third quark acts as a spectator. The second term belongs to the odderon, whereas in the third (C-even) piece all three quarks participate in the scattering. This term is new and has no analogue in γ * scattering. We also study the small x evolution of gluon radiation for each of these three terms. The first term follows the same pattern of gluon radiation as the γ * -initiated quark-antiquark dipole, and, in particular, it contains the BFKL evolution followed by the 2→4 transition vertex (triple Pomeron vertex). The odderon-term is described by the standard BKP evolution, and the baryon couples to both known odderon solutions, the Janik-Wosiek solution and the BLV solution. Finally, the t-channel evolution of the third term starts with a three reggeized gluon state which then, via a new 3→4 transition vertex, couples to the four gluon (two-Pomeron) state. We briefly discuss a few consequences of these findings, in particular the pattern of unitarization of high energy baryon scattering amplitudes. (orig.)
Baryon scattering at high energies. Wave function, impact factor, and gluon radiation
Energy Technology Data Exchange (ETDEWEB)
Bartels, J. [Hamburg Univ. (Germany). 2. Inst. fuer Theoretische Physik; Motyka, L. [Hamburg Univ. (Germany). 2. Inst. fuer Theoretische Physik]|[Jagellonian Univ., Krakow (Poland). Inst. of Physics
2007-11-15
The scattering of a baryon consisting of three massive quarks is investigated in the high energy limit of perturbative QCD. A model of a relativistic proton-like wave function, dependent on valence quark longitudinal and transverse momenta and on quark helicities, is proposed, and we derive the baryon impact factors for two, three and four t-channel gluons. We find that the baryonic impact factor can be written as a sum of three pieces: in the first one a subsystem consisting of two of the three quarks behaves very much like the quark-antiquark pair in {gamma}{sup *} scattering, whereas the third quark acts as a spectator. The second term belongs to the odderon, whereas in the third (C-even) piece all three quarks participate in the scattering. This term is new and has no analogue in {gamma}{sup *} scattering. We also study the small x evolution of gluon radiation for each of these three terms. The first term follows the same pattern of gluon radiation as the {gamma}{sup *}-initiated quark-antiquark dipole, and, in particular, it contains the BFKL evolution followed by the 2{yields}4 transition vertex (triple Pomeron vertex). The odderon-term is described by the standard BKP evolution, and the baryon couples to both known odderon solutions, the Janik-Wosiek solution and the BLV solution. Finally, the t-channel evolution of the third term starts with a three reggeized gluon state which then, via a new 3{yields}4 transition vertex, couples to the four gluon (two-Pomeron) state. We briefly discuss a few consequences of these findings, in particular the pattern of unitarization of high energy baryon scattering amplitudes. (orig.)
Directory of Open Access Journals (Sweden)
R. Fierz-Schmidhauser
2010-03-01
Full Text Available Ambient relative humidity (RH determines the water content of atmospheric aerosol particles and thus has an important influence on the amount of visible light scattered by particles. The RH dependence of the particle light scattering coefficient (σ_{sp} is therefore an important variable for climate forcing calculations. We used a humidification system for a nephelometer which allows for the measurement of σ_{sp} at a defined RH in the range of 20–95%. In this paper we present measurements of light scattering enhancement factors f(RH=σ_{sp}(RH/σ_{sp}(dry from a 1-month campaign (May 2008 at the high alpine site Jungfraujoch (3580 m a.s.l., Switzerland. Measurements at the Jungfraujoch are representative for the lower free troposphere above Central Europe. For this aerosol type hardly any information about the f(RH is available so far. At this site, f(RH=85% varied between 1.2 and 3.3. Measured f(RH agreed well with f(RH calculated with Mie theory using measurements of the size distribution, chemical composition and hygroscopic diameter growth factors as input. Good f(RH predictions at RH<85% were also obtained with a simplified model, which uses the Ångström exponent of σ_{sp}(dry as input. RH influences further intensive optical aerosol properties. The backscatter fraction decreased by about 30% from 0.128 to 0.089, and the single scattering albedo increased on average by 0.05 at 85% RH compared to dry conditions. These changes in σ_{sp}, backscatter fraction and single scattering albedo have a distinct impact on the radiative forcing of the Jungfraujoch aerosol.
International Nuclear Information System (INIS)
Sasaki, Satoshi.
1984-01-01
Anomalous scattering factors f' and f'' have been calculated for the atoms Li through Bi, plus U, using the relativistic treatment described by Cromer and Liberman (1970, 1981). The tables presented in this paper include values (i) in the wavelength range from 0.1 to 2.89 A in 0.01 A intervals and (ii) in the neighborhood of the K,L 1 ,L 2 , and L 3 absorption edges in 0.0001 A intervals. (author)
Thermodynamic model for the elastic form factor in diffraction scattering of protons
International Nuclear Information System (INIS)
Grashin, A.F.; Evstratenko, A.S.; Lepeshkin, M.V.
1988-01-01
An explicit expression is obtained for the differential pp(p-bar)-scattering cross section in the diffraction-cone region by employing the thermodynamic model for the elastic form factor previously proposed in Ref. 4. Data for the energy region 16.3≤(s)/sup 1/2/ ≤546 GeV have been analyzed and significant deviations have been discovered from the commonly used approximations in the form of linear or quadratic exponentials
Off-critical statistical models: factorized scattering theories and bootstrap program
International Nuclear Information System (INIS)
Mussardo, G.
1992-01-01
We analyze those integrable statistical systems which originate from some relevant perturbations of the minimal models of conformal field theories. When only massive excitations are present, the systems can be efficiently characterized in terms of the relativistic scattering data. We review the general properties of the factorizable S-matrix in two dimensions with particular emphasis on the bootstrap principle. The classification program of the allowed spins of conserved currents and of the non-degenerate S-matrices is discussed and illustrated by means of some significant examples. The scattering theories of several massive perturbations of the minimal models are fully discussed. Among them are the Ising model, the tricritical Ising model, the Potts models, the series of the non-unitary minimal models M 2,2n+3 , the non-unitary model M 3,5 and the scaling limit of the polymer system. The ultraviolet limit of these massive integrable theories can be exploited by the thermodynamics Bethe ansatz, in particular the central charge of the original conformal theories can be recovered from the scattering data. We also consider the numerical method based on the so-called conformal space truncated approach which confirms the theoretical results and allows a direct measurement of the scattering data, i.e. the masses and the S-matrix of the particles in bootstrap interaction. The problem of computing the off-critical correlation functions is discussed in terms of the form-factor approach
Atomic scattering factor of the ASTRO-H (Hitomi) SXT reflector around the gold's L edges
DEFF Research Database (Denmark)
Kikuchi, Naomichi; Kurashima, Sho; Ishida, Manabu
2016-01-01
The atomic scattering factor in the energy range of 11.2-15.4 keV for the ASTRO-H Soft X-ray Telescope (SXT) is reported. The large effective area of the SXT makes use of photon spectra above 10 keV viable, unlike most other X-ray satellites with total-reflection mirror optics. Presence of gold's L-edges...... in the energy band is a major issue, as it complicates the function of the effective area. In order to model the area, the reflectivity measurements in the 11.2-15.4 keV band with the energy pitch of 0.4-0.7 eV were made in the synchrotron beamline Spring-8 BL01B1. We obtained atomic scattering factors f1 and f......2 by the curve fitting to the reflectivities of our witness sample. The edges associated with the L-I, II, and III transitions are identified, of which the depths are found to be roughly 60% shallower than those expected from the Henke's atomic scattering factor....
Current s - quark mass corrections to the form factors of D - meson semileptonic decays
International Nuclear Information System (INIS)
Hussain, F.; Ivanov, A.N.; Troitskaya, N.I.
1994-11-01
The infinite mass effective theory, when a heavy quark mass tends to infinity, and Chiral perturbation theory at the quark level, based on the extended Nambu - Jona - Lasinio model with linear realization of chiral U(3) x U(3) symmetry, are applied to the calculations of current s - quark mass corrections to the form factors of the D → K-bar e + ν e and D → K-bar * e + ν e decays. These corrections turn out to be quite significant, of the order of 7 - 20%. The theoretical results are compared with experimental data. (author). 17 refs
Relationship between γ detection dead-time and count correction factor
International Nuclear Information System (INIS)
Wu Huailong; Zhang Jianhua; Chu Chengsheng; Hu Guangchun; Zhang Changfan; Hu Gen; Gong Jian; Tian Dongfeng
2015-01-01
The relationship between dead-time and count correction factor was investigated by using interference source for purpose of high γ activity measurement. The count rates maintain several 10 s"-"l with γ energy of 0.3-1.3 MeV for 10"4-10"5 Bq radioactive source. It is proved that the relationship between count loss and dead-time is unconcerned at various energy and various count intensities. The same correction formula can be used for any nuclide measurement. (authors)
Energy Technology Data Exchange (ETDEWEB)
Semchishen, A V; Seminogov, V N; Semchishen, V A [Institute of Laser and Information Technologies, Russian Academy of Sciences, Troitsk, Moscow Region (Russian Federation)
2012-04-30
Forward scattering of light passing through large-scale irregularities of the interface between two media having different refractive indices is considered. An analytical expression for the ratio of intensities of directional and diffusion components of scattered light in the far-field zone is derived. It is theoretically shown that the critical depth of possible interface relief irregularities, starting from which the intensity of the diffuse component in the passing light flow becomes comparable with the directional light component, responsible for the image formation on the eye retina, is 3 - 4 {mu}m, with the increase in the refractive index in the postoperational zone taken into account. These profile depth values agree with the experimentally measured ones and may affect the contrast sensitivity of vision.
International Nuclear Information System (INIS)
Semchishen, A V; Seminogov, V N; Semchishen, V A
2012-01-01
Forward scattering of light passing through large-scale irregularities of the interface between two media having different refractive indices is considered. An analytical expression for the ratio of intensities of directional and diffusion components of scattered light in the far-field zone is derived. It is theoretically shown that the critical depth of possible interface relief irregularities, starting from which the intensity of the diffuse component in the passing light flow becomes comparable with the directional light component, responsible for the image formation on the eye retina, is 3 - 4 μm, with the increase in the refractive index in the postoperational zone taken into account. These profile depth values agree with the experimentally measured ones and may affect the contrast sensitivity of vision.
Neutron activation analysis of copper traces: a study for sodium correction factor
International Nuclear Information System (INIS)
Tripathi, A.B.R.; Bhadkambekar, C.A.; Basu, A.K.; Chattopadhyay, N.
2007-01-01
Peak ratio correction factors for accurate quantitative determination of copper by NAA via 64 Cu radioisotope in presence of high 24 Na radioactivities has been established. Copper is the principal element as a marker of bullet residues on targets in connection to forensic ballistics cases. Reliable and precise estimation of copper by NAA either via non-destructive way or by resorting to radiochemical separation is of importance in forensic analysis for arriving at definitive inferences. However, majority of samples originating from wearing apparels, paper, leather, skin, glass or any other metal exhibit matrices contain high levels of sodium. The NAA scheme for determination of copper rests on measurements of net counts at 511 KeV which is the positron annihilation peak of 64 Cu. 24 Na also contributes significantly exactly at 511 KeV of gamma energy albeit by different mechanism i.e., by pair production. Therefore, total signal at 511 KeV is contributed by both. The easiest approach for correct estimation of copper traces has been established by the peak ratio correction factor. This has significance as both 64 Cu and 24 Na have comparable half lives, hence, as such time gap measurements cannot improve the situation. The consistency of peak ratio correction factor could be established for a particular geometry. (author)
Design of The High Efficiency Power Factor Correction Circuit for Power Supply
Directory of Open Access Journals (Sweden)
Atiye Hülya OBDAN
2017-12-01
Full Text Available Designing power factor correction circuits for switched power supplies has become important in recent years in terms of efficient use of energy. Power factor correction techniques play a significant role in high power density and energy efficiency. For these purposes, bridgeless PFC topologies and control strategies have been developed alongside basic boost PFC circuits. The power density can be increased using bridgeless structures by means of reducing losses in the circuit. This article examines bridgeless PFC structures and compares their performances in terms of losses and power factor. A semi-bridgeless PFC, which is widely used at high power levels, was analyzed and simulated. The designed circuit simulation using the current mode control method was performed in the PSIM program. A prototype of a 900 W semi-bridgeless PFC circuit was implemented and the results obtained from the circuit are presented
1/mQ corrections to form factors and extraction of |Vcb|
International Nuclear Information System (INIS)
Liu, J.; Chao, K.
1997-01-01
Form factors for 0 - →0 - and 0 - →1 - mesonic transitions in the heavy quark limit and the 1/m Q corrections are analyzed model independently within the Bethe-Salpeter (BS) formalism. The analysis shows that the BS formalism has spin-flavor symmetry in the heavy quark limit and respects Luke's theorem when the 1/m Q corrections are taken into account. All form factors for B→D (*) transitions beyond the zero recoil point are estimated in a relativistic constituent quark model based on the BS formalism. Using these form factors we calculate the branching ratios for the semileptonic decays B→D (*) l + ν l and extract the Cabibbo-Kobayashi-Maskawa matrix element |V cb |. We get |V cb |=0.042±0.003 which is consistent with the current world average. copyright 1997 The American Physical Society
Self-Excited Single-Stage Power Factor Correction Driving Circuit for LED Lighting
Directory of Open Access Journals (Sweden)
Yong-Nong Chang
2014-01-01
Full Text Available This pa\tper proposes a self-excited single-stage high power factor LED lighting driving circuit. Being featured with power factor correction capability without needing any control devices, the proposed circuit structure is with low cost and suitable for commercial production. The power factor correction function is accomplished by using inductor in combination with a half-bridge quasi resonant converter to achieve active switching and yield out voltage regulation according to load requirement. Furthermore, the zero-voltage switching in the half-bridge converter can be attained to promote the overall performance efficiency of the proposed circuit. Finally, the validity and production availability of the proposed circuit will be verified as well.
Zoros, E.; Moutsatsos, A.; Pappas, E. P.; Georgiou, E.; Kollias, G.; Karaiskos, P.; Pantelis, E.
2017-09-01
Detector-, field size- and machine-specific correction factors are required for precise dosimetry measurements in small and non-standard photon fields. In this work, Monte Carlo (MC) simulation techniques were used to calculate the k{{Qmsr},{{Q}0}}{{fmsr},{{f}ref}} and k{{Qclin},{{Q}msr}}{{fclin},{{f}msr}} correction factors for a series of ionization chambers, a synthetic microDiamond and diode dosimeters, used for reference and/or output factor (OF) measurements in the Gamma Knife Perfexion photon fields. Calculations were performed for the solid water (SW) and ABS plastic phantoms, as well as for a water phantom of the same geometry. MC calculations for the k{{Qclin},{{Q}msr}}{{fclin},{{f}msr}} correction factors in SW were compared against corresponding experimental results for a subset of ionization chambers and diode detectors. Reference experimental OF data were obtained through the weighted average of corresponding measurements using TLDs, EBT-2 films and alanine pellets. k{{Qmsr},{{Q}0}}{{fmsr},{{f}ref}} values close to unity (within 1%) were calculated for most of ionization chambers in water. Greater corrections of up to 6.0% were observed for chambers with relatively large air-cavity dimensions and steel central electrode. A phantom correction of 1.006 and 1.024 (breaking down to 1.014 from the ABS sphere and 1.010 from the accompanying ABS phantom adapter) were calculated for the SW and ABS phantoms, respectively, adding up to k{{Qmsr},{{Q}0}}{{fmsr},{{f}ref}} corrections in water. Both measurements and MC calculations for the diode and microDiamond detectors resulted in lower than unit k{{Qclin},{{Q}msr}}{{fclin},{{f}msr}} correction factors, due to their denser sensitive volume and encapsulation materials. In comparison, higher than unit k{{Qclin},{{Q}msr}}{{fclin},{{f}msr}} results for the ionization chambers suggested field size depended dose underestimations (being significant for the 4 mm field), with magnitude depending on the combination of
International Nuclear Information System (INIS)
Vivanco, M.G. Bernui de; Cardenas R, A.
2006-01-01
The ocular brachytherapy many times unique alternative to conserve the visual organ in patients of ocular cancer, one comes carrying out in the National Institute of Neoplastic Illnesses (INEN) using threads of Iridium 192; those which, they are placed in radial form on the interior surface of a spherical cap of gold of 18 K; the cap remains in the eye until reaching the prescribed dose by the doctor. The main objective of this work is to be able to calculate in a correct and practical way the one time that the treatment of ocular brachytherapy should last to reach the dose prescribed by the doctor. To reach this objective I use the Sievert integral corrected by attenuation effects and scattering (Meisberg polynomials); calculating it by the Simpson method. In the calculations by means of the Sievert integral doesn't take into account the scattering produced by the gold cap neither the variation of the constant of frequency of exposure with the distance. The calculations by means of Sievert integral are compared with those obtained using the Monte Carlo Penelope simulation code, where it is observed that they agree at distances of the surface of the cap greater or equal to 2mm. (Author)
International Nuclear Information System (INIS)
Marchand, D.
1998-11-01
This thesis presents the radiative corrections to the virtual compton scattering and the magnetic method adopted in the Hall A at Jefferson Laboratory, to measure the electrons beam energy with an accuracy of 10 4 . The virtual compton scattering experiments allow the access to the generalised polarizabilities of the protons. The extraction of these polarizabilities is obtained by the experimental and theoretical cross sections comparison. That's why the systematic errors and the radiative effects of the experiments have to be controlled very seriously. In this scope, a whole calculation of the internal radiative corrections has been realised in the framework of the quantum electrodynamic. The method of the dimensional regularisation has been used to the treatment of the ultraviolet and infra-red divergences. The absolute measure method of the energy, takes into account the magnetic deviation, made up of eight identical dipoles. The energy is determined from the deviation angle calculation of the beam and the measure of the magnetic field integral along the deviation
Synchronous scattering and diffraction from gold nanotextured surfaces with structure factors
Gu, Min-Jhong; Lee, Ming-Tsang; Huang, Chien-Hsun; Wu, Chi-Chun; Chen, Yu-Bin
2018-05-01
Synchronous scattering and diffraction were demonstrated using reflectance from gold nanotextured surfaces at oblique (θi = 15° and 60°) incidence of wavelength λ = 405 nm. Two samples of unique auto-correlation functions were cost-effectively fabricated. Multiple structure factors of their profiles were confirmed with Fourier expansions. Bi-directional reflectance function (BRDF) from these samples provided experimental proofs. On the other hand, standard deviation of height and unique auto-correlation function of each sample were used to generate surfaces numerically. Comparing their BRDF with those of totally random rough surfaces further suggested that structure factors in profile could reduce specular reflection more than totally random roughness.
Q resolution calculation of small angle neutron scattering spectrometer and analysis of form factor
International Nuclear Information System (INIS)
Chen Liang; Peng Mei; Wang Yan; Sun Liangwei; Chen Bo
2011-01-01
The calculational methods of Small Angle Neutron Scattering (SANS) spectrometer Q resolution function and its correlative Q standard difference were introduced. The effects of Q standard difference were analysed with the geometry lay out of spectrometer and the spread of neutron wavelength. The one dimension Q resolution Gaussian function were analysed. The form factor curve of ideal solid sphere and two different instrument arrangement parameter was convoluted respectively and the different smearing curve of form factor was obtained. The combination of using the Q resolution function to more accurately analysis SANS data. (authors)
The factorization method for inverse acoustic scattering in a layered medium
International Nuclear Information System (INIS)
Bondarenko, Oleksandr; Kirsch, Andreas; Liu, Xiaodong
2013-01-01
In this paper, we consider a problem of inverse acoustic scattering by an impenetrable obstacle embedded in a layered medium. We will show that the factorization method can be applied to recover the embedded obstacle; that is, the equation F-tilde g =φ z is solvable if and only if the sampling point z is in the interior of the unknown obstacle. Here, F-tilde is a self-adjoint operator related to the far field operator and ϕ z is the far field pattern of the Green function with respect to the problem of scattering by the background medium for point z. The validity of the factorization method is proven with the help of a mixed reciprocity principle and an application of the scattering operator. Due to the established mixed reciprocity principle, knowledge of the Green function for the background medium is no longer required, which makes the method attractive from the computational point of view. The paper is only concerned with sound-soft obstacles, but the analysis can be easily extended for sound-hard obstacles, or obstacles with separated sound-soft and sound-hard parts. Finally, we provide an explicit example for a radially symmetric case and present some numerical examples. (paper)
Towards a Resolution of the Proton Form Factor Problem: New Electron and Positron Scattering Data
Adikaram, D.; Rimal, D.; Weinstein, L. B.; Raue, B.; Khetarpal, P.; Bennett, R. P.; Arrington, J.; Brooks, W. K.; Adhikari, K. P.; Afanasev, A. V.; Amaryan, M. J.; Anderson, M. D.; Anefalos Pereira, S.; Avakian, H.; Ball, J.; Battaglieri, M.; Bedlinskiy, I.; Biselli, A. S.; Bono, J.; Boiarinov, S.; Briscoe, W. J.; Burkert, V. D.; Carman, D. S.; Careccia, S.; Celentano, A.; Chandavar, S.; Charles, G.; Colaneri, L.; Cole, P. L.; Contalbrigo, M.; Crede, V.; D'Angelo, A.; Dashyan, N.; De Vita, R.; De Sanctis, E.; Deur, A.; Djalali, C.; Dodge, G. E.; Dupre, R.; Egiyan, H.; El Alaoui, A.; El Fassi, L.; Elouadrhiri, L.; Eugenio, P.; Fedotov, G.; Fegan, S.; Filippi, A.; Fleming, J. A.; Fradi, A.; Garillon, B.; Gilfoyle, G. P.; Giovanetti, K. L.; Girod, F. X.; Goetz, J. T.; Gohn, W.; Golovatch, E.; Gothe, R. W.; Griffioen, K. A.; Guegan, B.; Guidal, M.; Guo, L.; Hafidi, K.; Hakobyan, H.; Hanretty, C.; Harrison, N.; Hattawy, M.; Hicks, K.; Holtrop, M.; Hughes, S. M.; Hyde, C. E.; Ilieva, Y.; Ireland, D. G.; Ishkhanov, B. S.; Jenkins, D.; Jiang, H.; Jo, H. S.; Joo, K.; Joosten, S.; Kalantarians, N.; Keller, D.; Khandaker, M.; Kim, A.; Kim, W.; Klein, A.; Klein, F. J.; Koirala, S.; Kubarovsky, V.; Kuhn, S. E.; Livingston, K.; Lu, H. Y.; MacGregor, I. J. D.; Markov, N.; Mattione, P.; Mayer, M.; McKinnon, B.; Mestayer, M. D.; Meyer, C. A.; Mirazita, M.; Mokeev, V.; Montgomery, R. A.; Moody, C. I.; Moutarde, H.; Movsisyan, A.; Camacho, C. Munoz; Nadel-Turonski, P.; Niccolai, S.; Niculescu, G.; Osipenko, M.; Ostrovidov, A. I.; Park, K.; Pasyuk, E.; Peña, C.; Pisano, S.; Pogorelko, O.; Price, J. W.; Procureur, S.; Prok, Y.; Protopopescu, D.; Puckett, A. J. R.; Ripani, M.; Rizzo, A.; Rosner, G.; Rossi, P.; Roy, P.; Sabatié, F.; Salgado, C.; Schott, D.; Schumacher, R. A.; Seder, E.; Sharabian, Y. G.; Simonyan, A.; Skorodumina, I.; Smith, E. S.; Smith, G. D.; Sober, D. I.; Sokhan, D.; Sparveris, N.; Stepanyan, S.; Stoler, P.; Strauch, S.; Sytnik, V.; Taiuti, M.; Tian, Ye; Trivedi, A.; Ungaro, M.; Voskanyan, H.; Voutier, E.; Walford, N. K.; Watts, D. P.; Wei, X.; Wood, M. H.; Zachariou, N.; Zana, L.; Zhang, J.; Zhao, Z. W.; Zonta, I.; CLAS Collaboration
2015-02-01
There is a significant discrepancy between the values of the proton electric form factor, GEp, extracted using unpolarized and polarized electron scattering. Calculations predict that small two-photon exchange (TPE) contributions can significantly affect the extraction of GEp from the unpolarized electron-proton cross sections. We determined the TPE contribution by measuring the ratio of positron-proton to electron-proton elastic scattering cross sections using a simultaneous, tertiary electron-positron beam incident on a liquid hydrogen target and detecting the scattered particles in the Jefferson Lab CLAS detector. This novel technique allowed us to cover a wide range in virtual photon polarization (ɛ ) and momentum transfer (Q2) simultaneously, as well as to cancel luminosity-related systematic errors. The cross section ratio increases with decreasing ɛ at Q2=1.45 GeV2 . This measurement is consistent with the size of the form factor discrepancy at Q2≈1.75 GeV2 and with hadronic calculations including nucleon and Δ intermediate states, which have been shown to resolve the discrepancy up to 2 - 3 GeV2 .
Energy Technology Data Exchange (ETDEWEB)
Hwang, Dong Hyeon; Cho, Sung-San [Hongik University, Seoul (Korea, Republic of)
2017-03-15
A fretting fatigue life estimation method that takes into account the stress gradient effect was developed by the authors [Journal of Mechanical Science and Technology, 28 (2014) 2153-2159]. In the developed method, fatigue damage value at the cracking location is corrected with fatigue damage gradient and the corrected value is compared directly with the plain fatigue data for life estimation. In other words, the correction factor is the ratio of plain fatigue damage to fretting fatigue damage at the same life and a function of fatigue damage gradient. Since reliability of the method was verified only for cylinder-on-flat contact configuration in the previous study, the present study extends application of the method to flat-on-flat contact configurations by developing the correction factor for both the contact configuration. Fretting fatigue experiments were conducted to obtain fatigue life data for various fretting pads. Finite element analyses were conducted to evaluate the Smith-Watson-Topper (SWT) fatigue damage parameter in the cracking region. It is revealed that the SWT parameter in fat-on-flat contact configuration decreases exponentially away from the surface as in cylinder-on-flat contact configuration, and thus the SWT gradient at the surface can be evaluated reliably. Moreover, it is found that decrease in the SWT parameter around the cracking location can be expressed by piecewise exponential curves. If the gradient of SWT at the surface is used as a representative value of SWT gradient, it is impossible to establish functional relationship between the SWT gradient and the correction factor for both the contact configurations although it was possible for cylinder-on-flat contact configuration. However, if weighted average of the SWT gradient values obtained from each exponential curve in the piecewise exponential curve is used as a representative value, the correction factor for both the contact configurations becomes a function of the SWT gradient
O(α2L2) radiative corrections to deep inelastic ep scattering for different kinematical variables
International Nuclear Information System (INIS)
Bluemlein, J.
1994-03-01
The QED radiative corrections are calculated in the leading log approximation up to O(α 2 ) for different definitions of the kinematical variables using jet measurement, the 'mixed' variables, the double angle method, and a measurement based on θ e and y JB . Higher order contributions due to exponentiation of soft radiation are included. (orig.)
Calculations of electron fluence correction factors using the Monte Carlo code PENELOPE
International Nuclear Information System (INIS)
Siegbahn, E A; Nilsson, B; Fernandez-Varea, J M; Andreo, P
2003-01-01
In electron-beam dosimetry, plastic phantom materials may be used instead of water for the determination of absorbed dose to water. A correction factor φ water plastic is then needed for converting the electron fluence in the plastic phantom to the fluence at an equivalent depth in water. The recommended values for this factor given by AAPM TG-25 (1991 Med. Phys. 18 73-109) and the IAEA protocols TRS-381 (1997) and TRS-398 (2000) disagree, in particular at large depths. Calculations of the electron fluence have been done, using the Monte Carlo code PENELOPE, in semi-infinite phantoms of water and common plastic materials (PMMA, clear polystyrene, A-150, polyethylene, Plastic water TM and Solid water TM (WT1)). The simulations have been carried out for monoenergetic electron beams of 6, 10 and 20 MeV, as well as for a realistic clinical beam. The simulated fluence correction factors differ from the values in the AAPM and IAEA recommendations by up to 2%, and are in better agreement with factors obtained by Ding et al (1997 Med. Phys. 24 161-76) using EGS4. Our Monte Carlo calculations are also in good accordance with φ water plastic values measured by using an almost perturbation-free ion chamber. The important interdependence between depth- and fluence-scaling corrections for plastic phantoms is discussed. Discrepancies between the measured and the recommended values of φ water plastic may then be explained considering the different depth-scaling rules used
International Nuclear Information System (INIS)
Etter, S.
1982-01-01
By current ultrasonic flow measuring equipment (UFME) the mean velocity is measured for one or two measuring paths. This mean velocity is not equal to the velocity averaged over the flow cross-section, by means of which the flow rate is calculated. This difference will be found already for axially symmetrical, fully developed velocity profiles and, to a larger extent, for disturbed profiles varying in flow direction and for nonsteady flow. Corrective factors are defined for steady and nonsteady flows. These factors can be derived from the flow profiles within the UFME. By mathematical simulation of the entrainment effect the influence of cross and swirl flows on various ultrasonic measuring methods is studied. The applied UFME with crossed measuring paths is shown to be largely independent of cross and swirl flows. For evaluation in a computer of velocity network measurements in circular cross-sections the equations for interpolation and integration are derived. Results of the mathematical method are the isotach profile, the flow rate and, for fully developed flow, directly the corrective factor. In the experimental part corrective factors are determined in nonsteady flow in a measuring plane before and in form measuring planes behind a perturbation. (orig./RW) [de
International Nuclear Information System (INIS)
Zemlyanaya, E.V.; Kiselev, M.A.
2002-01-01
On the basis of the model of separated form-factors, a code for fitting of small-angle neutron scattering spectra of the polydispersed vesicle population has been developed with corrections to the resolution function of the YuMO spectrometer. Vesicle and membrane bilayer parameters have been analyzed for various hierarchical models of the neutron scattering length density across the membrane. It was shown that hydration of vesicle can be described by the linear distribution function of water molecules. For the first time from the small-angle experiment, without additional methods, the average radius and polydispersity of the vesicle population, thickness of the membrane bilayer, thickness of hydrophobic and hydrophilic parts of bilayer, water distribution function and number of water molecules in the hydrophilic part have been calculated. (author)
Energy Technology Data Exchange (ETDEWEB)
Chen, X; Ouyang, L; Jia, X; Zhang, Y; Wang, J [UT Southwestern Medical Center, Dallas, TX (United States); Yan, H [Cyber Medical Corporation, Xi’an (China)
2016-06-15
Purpose: A moving blocker based strategy has shown promising results for scatter correction in cone-beam computed tomography (CBCT). Different geometry designs and moving speeds of the blocker affect its performance in image reconstruction accuracy. The goal of this work is to optimize the geometric design and moving speed of the moving blocker system through experimental evaluations. Methods: An Elekta Synergy XVI system and an anthropomorphic pelvis phantom CIRS 801-P were used for our experiment. A blocker consisting of lead strips was inserted between the x-ray source and the phantom moving back and forth along rotation axis to measure the scatter signal. Accoriding to our Monte Carlo simulation results, three blockers were used, which have the same lead strip width 3.2mm and different gap between neighboring lead strips, 3.2, 6.4 and 9.6mm. For each blocker, three moving speeds were evaluated, 10, 20 and 30 pixels per projection (on the detector plane). Scatter signal in the unblocked region was estimated by cubic B-spline based interpolation from the blocked region. CBCT image was reconstructed by a total variation (TV) based algebraic iterative reconstruction (ART) algorithm from the partially blocked projection data. Reconstruction accuracy in each condition is quantified as CT number error of region of interest (ROI) by comparing to a CBCT reconstructed image from analytically simulated unblocked and scatter free projection data. Results: Highest reconstruction accuracy is achieved when the blocker width is 3.2 mm, the gap between neighboring lead strips is 9.6 mm and the moving speed is 20 pixels per projection. RMSE of the CT number of ROIs can be reduced from 436 to 27. Conclusions: Image reconstruction accuracy is greatly affected by the geometry design of the blocker. The moving speed does not have a very strong effect on reconstruction result if it is over 20 pixels per projection.
International Nuclear Information System (INIS)
Lim, Teik-Cheng
2016-01-01
For moderately thick plates, the use of First order Shear Deformation Theory (FSDT) with a constant shear correction factor of 5/6 is sufficient to take into account the plate deflection arising from transverse shear deformation. For very thick plates, the use of Third order Shear Deformation Theory (TSDT) is preferred as it allows the shear strain distribution to be varied through the plate thickness. Therefore no correction factor is required in TSDT, unlike FSDT. Due to the complexity involved in TSDT, this paper obtains a more accurate shear correction factor for use in FSDT of very thick simply supported and uniformly loaded isosceles right triangular plates based on the TSDT. By matching the maximum deflections for this plate according to FSDT and TSDT, a variable shear correction factor is obtained. Results show that the shear correction factor for the simplified TSDT, i.e. 14/17, is least accurate. The commonly adopted shear correction factor of 5/6 in FSDT is valid only for very thin or highly auxetic plates. This paper provides a variable shear correction for FSDT deflection that matches the plate deflection by TSDT. This variable shear correction factor allows designers to justify the use of a commonly adopted shear correction factor of 5/6 even for very thick plates as long as the Poisson’s ratio of the plate material is sufficiently negative. (paper)
Energy Technology Data Exchange (ETDEWEB)
Fernandez, B [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1963-07-01
A calculation for double scattering and absorption corrections in fast neutron scattering experiments using Monte-Carlo method is given. Application to cylindrical target is presented in FORTRAN symbolic language. (author) [French] Un calcul des corrections de double diffusion et d'absorption dans les experiences de diffusion de neutrons rapides par la methode de Monte-Carlo est presente. L'application au cas d'une cible cylindrique est traitee en langage symbolique FORTRAN. (auteur)
International Nuclear Information System (INIS)
Grimbergen, T.W.M.; Dijk, E. van; Vries, W. de
1998-01-01
A new method is described for the determination of x-ray quality dependent correction factors for free-air ionization chambers. The method is based on weighting correction factors for mono-energetic photons, which are calculated using the Monte Carlo method, with measured air kerma spectra. With this method, correction factors for electron loss, scatter inside the chamber and transmission through the diaphragm and front wall have been calculated for the NMi free-air chamber for medium-energy x-rays for a wide range of x-ray qualities in use at NMi. The newly obtained correction factors were compared with the values in use at present, which are based on interpolation of experimental data for a specific set of x-ray qualities. For x-ray qualities which are similar to this specific set, the agreement between the correction factors determined with the new method and those based on the experimental data is better than 0.1%, except for heavily filtered x-rays generated at 250 kV. For x-ray qualities dissimilar to the specific set, differences up to 0.4% exist, which can be explained by uncertainties in the interpolation procedure of the experimental data. Since the new method does not depend on experimental data for a specific set of x-ray qualities, the new method allows for a more flexible use of the free-air chamber as a primary standard for air kerma for any x-ray quality in the medium-energy x-ray range. (author)
Evaluation on correction factor for in-line X-ray phase contrast computed tomography
Energy Technology Data Exchange (ETDEWEB)
Jin, Mingli; Huang, Zhifeng; Zhang, Li; Zhang, Ran [Tsinghua Univ., Beijing (China). Dept. of Engineering Physics; Ministry of Education, Beijing (China). Key Laboratory of Particle and Radiation Imaging; Yin, Hongxia; Liu, Yunfu; Wang, Zhenchang [Capital Medical Univ., Beijing (China). Medical Imaging Center; Xiao, Tiqiao [Chinese Academy of Sciences, Shanghai (China). Shanghai Inst. of Applied Physics
2011-07-01
X-ray in-line phase contrast computed tomography (CT) is an effective nondestructive tool, providing 3D distribution of the refractive index of weakly absorbing low-Z object with high resolution and image contrast, especially with high-brilliance third-generation synchrotron radiation sources. Modified Bronnikov's algorithm (MBA), one of the in-line phase contrast CT reconstruction algorithms, can reconstruct the refractive index distribution of a pure phase object with a single computed tomographic data set. The key idea of the MBA is to use a correction factor in the filter function to stabilize the behavior at low frequencies. In this paper, we evaluate the influences of the correction factor to the final reconstruction results of the absorption-phase-mixed objects with analytical simulation and actual experiments. The limitations of the MBA are discussed finally. (orig.)
An improved correlated sampling method for calculating correction factor of detector
International Nuclear Information System (INIS)
Wu Zhen; Li Junli; Cheng Jianping
2006-01-01
In the case of a small size detector lying inside a bulk of medium, there are two problems in the correction factors calculation of the detectors. One is that the detector is too small for the particles to arrive at and collide in; the other is that the ratio of two quantities is not accurate enough. The method discussed in this paper, which combines correlated sampling with modified particle collision auto-importance sampling, and has been realized on the MCNP-4C platform, can solve these two problems. Besides, other 3 variance reduction techniques are also combined with correlated sampling respectively to calculate a simple calculating model of the correction factors of detectors. The results prove that, although all the variance reduction techniques combined with correlated sampling can improve the calculating efficiency, the method combining the modified particle collision auto-importance sampling with the correlated sampling is the most efficient one. (authors)
Low Capacitive Inductors for Fast Switching Devices in Active Power Factor Correction Applications
DEFF Research Database (Denmark)
Hernandez Botella, Juan Carlos; Petersen, Lars Press; Andersen, Michael A. E.
2014-01-01
This paper examines different winding strategies for reduced capacitance inductors in active power factor correction circuits (PFC). The effect of the parasitic capacitance is analyzed from an electro magnetic compatibility (EMI) and efficiency point of views. The purpose of this work is to inves......This paper examines different winding strategies for reduced capacitance inductors in active power factor correction circuits (PFC). The effect of the parasitic capacitance is analyzed from an electro magnetic compatibility (EMI) and efficiency point of views. The purpose of this work...... is to investigate different winding approaches and identify suitable solutions for high switching frequency/high speed transition PFC designs. A low parasitic capacitance PCB based inductor design is proposed to address the challenges imposed by high switching frequency PFC Boost converters....
International Nuclear Information System (INIS)
Santos, Talita O.; Rocha, Zildete; Knupp, Eliana A.N.; Kastner, Geraldo F.; Oliveira, Arno H. de; Oliveira, Arno H. de
2015-01-01
Gamma spectrometry technique has been used in order to obtain the activity concentrations of natural and artificial radionuclides in environmental samples of different origins, compositions and densities. These samples characteristics may influence the calibration condition by the self-attenuation effect. The sample density has been considered the most important factor. For reliable results, it is necessary to determine self-attenuation correction factor which has been subject of great interest due to its effect on activity concentration. In this context, the aim of this work is to show the calibration process considering the correction by self-attenuation in the evaluation of the concentration of each radionuclide to a gamma HPGEe detector spectrometry system. (author)
Power correction to the asymptotics of the pion electromagnetic form factor
International Nuclear Information System (INIS)
Geshkenbein, B.V.; Terentyev, M.V.
1982-01-01
The contribution of the power correction approximately (μ 2 /Q 2 ) 2 enhanced by the factor approximately μ 2 /anti m 2 , to the pion form factor (FF) is calculated (here μ is the pion mass, anti m=1/2(msub(u)+msub(α)) is the mean value of the u- and d-quark masses, Q 2 =-(p-p') 2 > 0, where p, p' are meson momenta at initial and final state. It is shown that the only source of large corrections is due to the contribution of the local pseudoscalar current. The main (approximately 1/Q 2 ) asymptotics of FF associated with the axial current contribution, is derived. The contribution (approximately 1/Q 4 ) of the pseudoscalar current is calculated
International Nuclear Information System (INIS)
Pereira, Marcelo O.; Anjos, Marcelino J.; Lopes, Ricardo T.
2009-01-01
Non-destructive techniques with X-ray, such as tomography, radiography and X-ray fluorescence are sensitive to the attenuation coefficient and have a large field of applications in medical as well as industrial area. In the case of X-ray fluorescence analysis the knowledge of photon X-ray attenuation coefficients provides important information to obtain the elemental concentration. On the other hand, the mass attenuation coefficient values are determined by transmission methods. So, the use of X-ray scattering can be considered as an alternative to transmission methods. This work proposes a new method for obtain the X-ray absorption curve through superposition peak Rayleigh and Compton scattering of the lines L a e L β of Tungsten (Tungsten L lines of an X-ray tube with W anode). The absorption curve was obtained using standard samples with effective atomic number in the range from 6 to 16. The method were applied in certified samples of bovine liver (NIST 1577B) , milk powder and V-10. The experimental measurements were obtained using the portable system EDXRF of the Nuclear Instrumentation Laboratory (LIN-COPPE/UFRJ) with Tungsten (W) anode. (author)
Conti, C. C.; Anjos, M. J.; Salgado, C. M.
2014-09-01
X-ray fluorescence technique plays an important role in nondestructive analysis nowadays. The development of equipment, including portable ones, enables a wide assortment of possibilities for analysis of stable elements, even in trace concentrations. Nevertheless, despite of the advantages, one important drawback is radiation self-attenuation in the sample being measured, which needs to be considered in the calculation for the proper determination of elemental concentration. The mass attenuation coefficient can be determined by transmission measurement, but, in this case, the sample must be in slab shape geometry and demands two different setups and measurements. The Rayleigh to Compton scattering ratio, determined from the X-ray fluorescence spectrum, provides a link to the mass attenuation coefficient by means of a polynomial type equation. This work presents a way to construct a Rayleigh to Compton scattering ratio versus mass attenuation coefficient curve by using the MCNP5 Monte Carlo computer code. The comparison between the calculated and literature values of the mass attenuation coefficient for some known samples showed to be within 15%. This calculation procedure is available on-line at www.macx.net.br.
Study of the electromagnetic form factors of Helium-3 and Tritium nuclei by electron scattering
International Nuclear Information System (INIS)
Amroun, A.
1989-01-01
Accurate measurements of the tritium electromagnetic form factor demonstrated that, when the exchange currents are included, the theoretical and the experimental data are in agreement. Similar calculations carried out on helium-3 were not satisfactory. In this investigation, a new electromagnetic form factor of helium-3 is measured. The transfer zone of the diffraction spectra concerning the first minimum and the second maximum is considered. The aim of the study is to test on both nuclei the validity and the uncertainties of the models. The scattering of electrons on helium-3 is analyzed. The experiment was performed in the Saclay linear accelerator. The isoscalar and isovector form factors could be differentiated. By comparing the theoretical and the experimental data, it is demonstrated that the use of three body forces in the calculations has no effect on the form factor results [fr
Structure of rapidity divergences in multi-parton scattering soft factors
Vladimirov, Alexey
2018-04-01
We discuss the structure of rapidity divergences that are presented in the soft factors of transverse momentum dependent (TMD) factorization theorems. To provide the discussion on the most general level we consider soft factors for multi-parton scattering. We show that the rapidity divergences are result of the gluon exchanges with the distant transverse plane, and are structurally equivalent to the ultraviolet divergences. It allows to formulate and to prove the renormalization theorem for rapidity divergences. The proof is made with the help the conformal transformation which maps rapidity divergences to ultraviolet divergences. The theorem is the systematic form of the factorization of rapidity divergences, which is required for the definition of TMD parton distributions. In particular, the definition of multi parton distributions is presented. The equivalence of ultraviolet and rapidity divergences leads to the exact relation between soft and rapidity anomalous dimensions. Using this relation we derive the rapidity anomalous dimension at the three-loop order.
Measured attenuation correction methods
International Nuclear Information System (INIS)
Ostertag, H.; Kuebler, W.K.; Doll, J.; Lorenz, W.J.
1989-01-01
Accurate attenuation correction is a prerequisite for the determination of exact local radioactivity concentrations in positron emission tomography. Attenuation correction factors range from 4-5 in brain studies to 50-100 in whole body measurements. This report gives an overview of the different methods of determining the attenuation correction factors by transmission measurements using an external positron emitting source. The long-lived generator nuclide 68 Ge/ 68 Ga is commonly used for this purpose. The additional patient dose from the transmission source is usually a small fraction of the dose due to the subsequent emission measurement. Ring-shaped transmission sources as well as rotating point or line sources are employed in modern positron tomographs. By masking a rotating line or point source, random and scattered events in the transmission scans can be effectively suppressed. The problems of measured attenuation correction are discussed: Transmission/emission mismatch, random and scattered event contamination, counting statistics, transmission/emission scatter compensation, transmission scan after administration of activity to the patient. By using a double masking technique simultaneous emission and transmission scans become feasible. (orig.)
Seasonal correction factors in radon exposure assessment: are they help or hindrance?
International Nuclear Information System (INIS)
Denman, A.R.; Groves-Kirkby, C.J.; Phillips, P.S.; Woolridge, A.C.; Crockett, R.G.M.
2008-01-01
Northern Hemisphere radon levels are generally higher in Winter than in Summer, primarily due to the increased interior/exterior temperature difference during the heating season, which results in greater atmospheric pressure differential and enhanced radon ingress. Following a survey of domestic radon levels in the United Kingdom (UK), the former National Radiological Protection Board (NRPB) established measurement protocols and promulgated nationally-applicable Seasonal Correction Factors (SCF). These factors convert a one-month or three-month radon concentration measurement, commencing in any month of the year, to an annual mean radon concentration. Subsequent study suggests that this approach may not be sufficiently sensitive to local conditions, and a major independent investigation reported seasonal correction factors specific to nine geographic regions, together with a composite set applicable to all regions. Similar geographical variability has been observed in other countries. In a recent evaluation of the applicability of short-term exposures in quantifying long-term domestic radon levels, radon levels in 34 houses were monitored over a 12-month period with 1-week, 1-month and 3-month exposures. Radon concentration variation departed significantly from that expected on the basis of the recommended SCFs, with year-end discontinuities at all exposure durations. Weekly monitoring with electrets was continued in three of these locations for four years. Short-term variations in radon levels were observed, particularly during the shorter exposures, and this dataset has also shown year-on-year variations. Overall, SCFs derived from this dataset are significantly lower than those recommended, but are comparable with other results from the UK and elsewhere, particularly those that recognise geological diversity and are consequently prepared on a regional rather than a national basis. These findings call into question the validity of using nationally
Proximity formulae for folding potentials. [Saxon-Woods form factors, first order corrections
Energy Technology Data Exchange (ETDEWEB)
Schechter, H; Canto, L F [Rio de Janeiro Univ. (Brazil). Inst. de Fisica
1979-03-05
The proximity formulae of Brink and Stancu are applied to folding potentials. A numerical study is made for the case of single folding potentials with Saxon-Woods form factors. It is found that a proximity formula is accurate to 1-2% at separations of the order of the radius of the Coulomb barrier and that first order corrections due to first curvature are important. The approximations involved are discussed.
International Nuclear Information System (INIS)
Blanco, F.; Rosado, J.; Illana, A.; Garcia, G.
2010-01-01
The SCAR and EGAR procedures have been proposed in order to extend to lower energies the applicability of the additivity rule for calculation of electron-molecule total cross sections. Both those approximate treatments arise after considering geometrical screening corrections due to partial overlapping of atoms in the molecule, as seen by the incident electrons. The main features, results and limitations of both treatments are put here in comparison by means of their application to some different sized species.
Determination of true coincidence correction factors using Monte-Carlo simulation techniques
Directory of Open Access Journals (Sweden)
Chionis Dionysios A.
2014-01-01
Full Text Available Aim of this work is the numerical calculation of the true coincidence correction factors by means of Monte-Carlo simulation techniques. For this purpose, the Monte Carlo computer code PENELOPE was used and the main program PENMAIN was properly modified in order to include the effect of the true coincidence phenomenon. The modified main program that takes into consideration the true coincidence phenomenon was used for the full energy peak efficiency determination of an XtRa Ge detector with relative efficiency 104% and the results obtained for the 1173 keV and 1332 keV photons of 60Co were found consistent with respective experimental ones. The true coincidence correction factors were calculated as the ratio of the full energy peak efficiencies was determined from the original main program PENMAIN and the modified main program PENMAIN. The developed technique was applied for 57Co, 88Y, and 134Cs and for two source-to-detector geometries. The results obtained were compared with true coincidence correction factors calculated from the "TrueCoinc" program and the relative bias was found to be less than 2%, 4%, and 8% for 57Co, 88Y, and 134Cs, respectively.
Zhou, Lihong; Yuan, Liming; Thomas, Rick; Iannacchione, Anthony
2017-12-01
When there are installations of air velocity sensors in the mining industry for real-time airflow monitoring, a problem exists with how the monitored air velocity at a fixed location corresponds to the average air velocity, which is used to determine the volume flow rate of air in an entry with the cross-sectional area. Correction factors have been practically employed to convert a measured centerline air velocity to the average air velocity. However, studies on the recommended correction factors of the sensor-measured air velocity to the average air velocity at cross sections are still lacking. A comprehensive airflow measurement was made at the Safety Research Coal Mine, Bruceton, PA, using three measuring methods including single-point reading, moving traverse, and fixed-point traverse. The air velocity distribution at each measuring station was analyzed using an air velocity contour map generated with Surfer ® . The correction factors at each measuring station for both the centerline and the sensor location were calculated and are discussed.
International Nuclear Information System (INIS)
Christillin, P.
1986-01-01
The theory of nuclear Compton scattering is reformulated with explicit consideration of both virtual and real pionic degrees of freedom. The effects due to low-lying nuclear states, to seagull terms, to pion condensation and to the Δ dynamics in the nucleus and their interplay in the different energy regions are examined. It is shown that all corrections to the one-body terms, of diffractive behaviour determined by the nuclear form factor, have an effective two-body character. The possibility of using Compton scattering as a complementary source of information about nuclear dynamics is restressed. (author)
Scattering factor evaluation for internal dose assessment due to 60Co
International Nuclear Information System (INIS)
Gautam, Y.P.; Kumar, A.; Sharma, S.; Sharma, A.K.; Dube, B.; Hegde, A.G.
2008-01-01
Guidelines for the assessment of internal doses from monitoring suggest default measurement of uncertainties (i.e. lognormal scattering factor, SF) to be used for different types of monitoring data. In this paper, SF values have been evaluated for internal contamination due to 60 Co in two cases using whole body counting data. SF values of 1.04 and 1.03 were obtained for case I and II respectively while SF value of 1.03 was obtained using bioassay data for case I. SF evaluated is in good agreement with the default values given by IDEAS guidelines. (author)
Factorization properties and spurious solutions in N-body scattering theories
International Nuclear Information System (INIS)
Vanzani, V.
1979-01-01
The origin of spurious solutions in N-body scattering equations is discussed. It is shown that spurious solutions are expected because of specific factorization properties of the homogeneous equations. The equations proposed by Rosenberg, by Mitra, Gillespie, Sugar and Panchapakesan, by Takahashi and Mishima, by Alessandrini, by Sasakawa, by Sloan, Bencze and Redish, by Weinberg and van Winter and by Avishai are considered. It is explicitly shown that spurious multipliers arise from repeated employment of resolvent equations or, equiValently, from generalized iteration procedure
DetOx: a program for determining anomalous scattering factors of mixed-oxidation-state species.
Sutton, Karim J; Barnett, Sarah A; Christensen, Kirsten E; Nowell, Harriott; Thompson, Amber L; Allan, David R; Cooper, Richard I
2013-01-01
Overlapping absorption edges will occur when an element is present in multiple oxidation states within a material. DetOx is a program for partitioning overlapping X-ray absorption spectra into contributions from individual atomic species and computing the dependence of the anomalous scattering factors on X-ray energy. It is demonstrated how these results can be used in combination with X-ray diffraction data to determine the oxidation state of ions at specific sites in a mixed-valance material, GaCl(2).
X-ray atomic scattering factors of low-Z ions with a core hole
International Nuclear Information System (INIS)
Hau-Riege, Stefan P.
2007-01-01
Short and intense x-ray pulses may be used for atomic-resolution diffraction imaging of single biological molecules. One of the dominant damage mechanisms is atomic ionization, resulting in a large fraction of atoms with core holes. We calculated the atomic scattering factor of atoms with atomic charge numbers between 3 and 10 in different ionization states with and without a core hole. Our results show that orbital occupation and the change of the orbitals upon core ionization (core relaxation) have a significant impact on the diffraction pattern
Asymptotic energy scale factors for pseudoscalar meson scattering and charmed meson couplings
International Nuclear Information System (INIS)
Thews, R.L.
1977-01-01
Energy scale factors ν 0 for PP → PP scattering amplitudes are related via absence of exotic resonances of ratios of tensor to vector coupling strengths. These same ratios are extracted from FESR's for non-exotic reactions. The scale factors obtained are all of the order of 1.0 GeV 2 or less, indepedent of quantum numbers. This contradicts the expectations of dual amplitudes in which ν 0 =1/α', and trajectory slopes are smaller for charmed mesons. Decay widths for tensor mesons are predicted. An observed SU(3) violation for the ratio A 2 → KantiK/K** → Kπ is shown to be consistent with the FESR results. Charmed meson decays are predicted to be factors of 2 to 3 larger than those predicted by SU(4). (author)
Influence factors and corrections of low-energy γ-ray penetration in ash analysis
International Nuclear Information System (INIS)
Cheng Bo; Tuo Xianguo; Zhou Jianbin; Tong Yunfu
2002-01-01
The author introduces the system of the coal ash analyzer. This system is based on the low-energy γ-ray source 241 Am emitted two kinds of energy peaks 26.4 keV and 59.6 keV to analyze the ash in coal with the penetration way. The author also offers the factors to influence the accuracy of ash analysis, such as the size of coal, the environmental temperature, the important elements in coal, and water in coal too. At the same time, depending on the cause of the factors, it offer some methods of correction such as the way of the auto-hold energy peak, the way of the auto-compensation way, and so on. The author also mentions the other influence factors of the measurement accuracy to be noticed during the experiment. All these aim at clearing off the influence factors of measurement accuracy through the experiments
Real-time correction of tsunami site effect by frequency-dependent tsunami-amplification factor
Tsushima, H.
2017-12-01
For tsunami early warning, I developed frequency-dependent tsunami-amplification factor and used it to design a recursive digital filter that can be applicable for real-time correction of tsunami site response. In this study, I assumed that a tsunami waveform at an observing point could be modeled by convolution of source, path and site effects in time domain. Under this assumption, spectral ratio between offshore and the nearby coast can be regarded as site response (i.e. frequency-dependent amplification factor). If the amplification factor can be prepared before tsunamigenic earthquakes, its temporal convolution to offshore tsunami waveform provides tsunami prediction at coast in real time. In this study, tsunami waveforms calculated by tsunami numerical simulations were used to develop frequency-dependent tsunami-amplification factor. Firstly, I performed numerical tsunami simulations based on nonlinear shallow-water theory from many tsuanmigenic earthquake scenarios by varying the seismic magnitudes and locations. The resultant tsunami waveforms at offshore and the nearby coastal observing points were then used in spectral-ratio analysis. An average of the resulted spectral ratios from the tsunamigenic-earthquake scenarios is regarded as frequency-dependent amplification factor. Finally, the estimated amplification factor is used in design of a recursive digital filter that can be applicable in time domain. The above procedure is applied to Miyako bay at the Pacific coast of northeastern Japan. The averaged tsunami-height spectral ratio (i.e. amplification factor) between the location at the center of the bay and the outside show a peak at wave-period of 20 min. A recursive digital filter based on the estimated amplification factor shows good performance in real-time correction of tsunami-height amplification due to the site effect. This study is supported by Japan Society for the Promotion of Science (JSPS) KAKENHI grant 15K16309.
Perturbative corrections to Λ_b→Λ form factors from QCD light-cone sum rules
International Nuclear Information System (INIS)
Wang, Yu-Ming; Shen, Yue-Long
2016-01-01
We compute radiative corrections to Λ_b→Λ from factors, at next-to-leading logarithmic accuracy, from QCD light-cone sum rules with Λ_b-baryon distribution amplitudes. Employing the diagrammatic approach factorization of the vacuum-to-Λ_b-baryon correlation function is justified at leading power in Λ/m_b, with the aid of the method of regions. Hard functions entering the factorization formulae are identical to the corresponding matching coefficients of heavy-to-light currents from QCD onto soft-collinear effective theory. The universal jet function from integrating out the hard-collinear fluctuations exhibits richer structures compared with the one involved in the factorization expressions of the vacuum-to-B-meson correlation function. Based upon the QCD resummation improved sum rules we observe that the perturbative corrections at O(α_s) shift the Λ_b→Λ from factors at large recoil significantly and the dominant contribution originates from the next-to-leading order jet function instead of the hard coefficient functions. Having at hand the sum rule predictions for the Λ_b→Λ from factors we further investigate several decay observables in the electro-weak penguin Λ_b→Λ ℓ"+ℓ"− transitions in the factorization limit (i.e., ignoring the “non-factorizable' hadronic effects which cannot be expressed in terms of the Λ_b→Λ from factors), including the invariant mass distribution of the lepton pair, the forward-backward asymmetry in the dilepton system and the longitudinal polarization fraction of the leptonic sector.
International Nuclear Information System (INIS)
Lee, K.W.; Sheu, R.J.
2015-01-01
High-energy neutrons (>10 MeV) contribute substantially to the dose fraction but result in only a small or negligible response in most conventional moderated-type neutron detectors. Neutron dosemeters used for radiation protection purpose are commonly calibrated with 252 Cf neutron sources and are used in various workplace. A workplace-specific correction factor is suggested. In this study, the effect of the neutron spectrum on the accuracy of dose measurements was investigated. A set of neutron spectra representing various neutron environments was selected to study the dose responses of a series of Bonner spheres, including standard and extended-range spheres. By comparing 252 Cf-calibrated dose responses with reference values based on fluence-to-dose conversion coefficients, this paper presents recommendations for neutron field characterisation and appropriate correction factors for responses of conventional neutron dosemeters used in environments with high-energy neutrons. The correction depends on the estimated percentage of high-energy neutrons in the spectrum or the ratio between the measured responses of two Bonner spheres (the 4P6-8 extended-range sphere versus the 6'' standard sphere). (authors)
Correction factors for safe performance of API X65 pipeline steel
International Nuclear Information System (INIS)
Hashemi, Sayyed H.
2009-01-01
Prediction of required Charpy energy for fracture arrest is vital for safe performance of gas transportation pipelines. This is commonly estimated through failure models calibrated in the past on fracture data from combined Charpy tests and full-thickness burst experiments. Unfortunately, such pipeline failure models are unable to correctly predict the minimum arrest toughness of thermo-mechanical controlled rolled (TMCR) steels. To refine the existing failure models, different empirical adjustments have been proposed in recent years. In this paper, similar correction factors were derived from fracture information of instrumented Charpy impact tests on API X65 steel. The contribution of different fracture mechanisms of impact test specimens was determined through energy partitioning analysis. Parts of the energy contribution were correlated then to the source of uncertainty observed in similar experiments. The applied technique was similar to that of previous studies on X70 and X100 steels, and proved to be encouraging in giving consistent results compared to available test data.
Form factors and scattering amplitudes in N=4 SYM in dimensional and massive regularizations
Energy Technology Data Exchange (ETDEWEB)
Henn, Johannes M. [Humboldt-Universitaet, Berlin (Germany). Inst. fuer Physik; California Univ., Santa Barbara, CA (United States). Kavli Inst. for Theoretical Physics; Moch, Sven [California Univ., Santa Barbara, CA (United States). Kavli Inst. for Theoretical Physics; Deutsches Elektronen-Synchrotron (DESY), Zeuthen (Germany); Naculich, Stephen G. [California Univ., Santa Barbara, CA (United States). Kavli Inst. for Theoretical Physics; Bowdoin College, Brunswick, ME (United States). Dept. of Physics
2011-09-15
The IR-divergent scattering amplitudes of N=4 supersymmetric Yang-Mills theory can be regulated in a variety of ways, including dimensional regularization and massive (or Higgs) regularization. The IR-finite part of an amplitude in different regularizations generally differs by an additive constant at each loop order, due to the ambiguity in separating finite and divergent contributions. We give a prescription for defining an unambiguous, regulator-independent finite part of the amplitude by factoring off a product of IR-divergent ''wedge'' functions. For the cases of dimensional regularization and the common-mass Higgs regulator, we define the wedge function in terms of a form factor, and demonstrate the regularization independence of the n-point amplitude through two loops. We also deduce the form of the wedge function for the more general differential-mass Higgs regulator, although we lack an explicit operator definition in this case. Finally, using extended dual conformal symmetry, we demonstrate the link between the differential-mass wedge function and the anomalous dual conformal Ward identity for the finite part of the scattering amplitude. (orig.)
Form factors and scattering amplitudes in N=4 SYM in dimensional and massive regularizations
International Nuclear Information System (INIS)
Henn, Johannes M.; Naculich, Stephen G.; Bowdoin College, Brunswick, ME
2011-09-01
The IR-divergent scattering amplitudes of N=4 supersymmetric Yang-Mills theory can be regulated in a variety of ways, including dimensional regularization and massive (or Higgs) regularization. The IR-finite part of an amplitude in different regularizations generally differs by an additive constant at each loop order, due to the ambiguity in separating finite and divergent contributions. We give a prescription for defining an unambiguous, regulator-independent finite part of the amplitude by factoring off a product of IR-divergent ''wedge'' functions. For the cases of dimensional regularization and the common-mass Higgs regulator, we define the wedge function in terms of a form factor, and demonstrate the regularization independence of the n-point amplitude through two loops. We also deduce the form of the wedge function for the more general differential-mass Higgs regulator, although we lack an explicit operator definition in this case. Finally, using extended dual conformal symmetry, we demonstrate the link between the differential-mass wedge function and the anomalous dual conformal Ward identity for the finite part of the scattering amplitude. (orig.)
Wang, Lilie
In ionization chamber radiation dosimetry, the introduction of the ion chamber into medium will unavoidably distort the radiation field near the chamber because the chamber cavity material (air) is different from the medium. A replacement correction factor, Prepl was introduced in order to correct the chamber readings to give an accurate radiation dose in the medium without the presence of the chamber. Generally it is very hard to measure the values of Prepl since they are intertwined with the chamber wall effect. In addition, the P repl values always come together with the stopping-power ratio of the two media involved. This makes the problem of determining the P repl values even more complicated. Monte Carlo simulation is an ideal method to investigate the replacement correction factors. In this study, four different methods of calculating the values of Prepl by Monte Carlo simulation are discussed. Two of the methods are designated as 'direct' methods in the sense that the evaluation of the stopping-power ratio is not necessary. The systematic uncertainties of the two direct methods are estimated to be about 0.1-0.2% which comes from the ambiguous definition of the energy cutoff Delta used in the Spencer-Attix cavity theory. The two direct methods are used to calculate the values of P repl for both plane-parallel chambers and cylindrical thimble chambers in either electron beams or photon beams. The calculation results are compared to measurements. For electron beams, good agreements are obtained. For thimble chambers in photon beams, significant discrepancies are observed between calculations and measurements. The experiments are thus investigated and the procedures are simulated by the Monte Carlo method. It is found that the interpretation of the measured data as the replacement correction factors in dosimetry protocols are not correct. In applying the calculation to the BIPM graphite chamber in a 60Co beam, the calculated values of P repl differ from those
Development of a correction factor for Xe-133 vials for use with a dose calibrator
International Nuclear Information System (INIS)
Gels, G.L.; Piltingsrud, H.V.
1982-01-01
Manufacturers of dose calibrators who give calibration settings for various radionuclies sometimes do not specify the type of radionuclide container the calibration is for. The container, moreover, may not be of the same type as those a user might purchase. When these factors are not considered, the activity administered to the patient may be significantly different from that intended. An experiment is described in which calibration factors are determined for measurement of Xe-133 activity in vials in a dose calibrator. This was accomplished by transferring the Xe-133 from the commercial vials to standard NBS calibration ampules. Based on ten such transfers, the resulting correction factor for the dose calibrator was 1.22
International Nuclear Information System (INIS)
Rodrigues, L.L.C.
1988-01-01
A simple method was developed to be suggested to hospital physicists in order to be followed during large radiation field dosimetry, to evaluate the effects of cables, connectors and extension cables irradiation and to determine correction factors for each system or geometry. All quality control tests were performed according to the International Electrotechnical Commission for three clinical dosimeters. Photon and electron irradiation effects for cables, connectors and extention cables were investigated under different experimental conditions by means of measurements of chamber sensitivity to a standard radiation source of 90 Sr. The radiation induced leakage current was also measured for cables, connectors and extension cables irradiated by photons and electrons. All measurements were performed at standard dosimetry conditions. Finally, measurements were performed in large fields. Cable factors and leakage factors were determined by the relation between chamber responses for irradiated and unirradiated cables. (author) [pt
International Nuclear Information System (INIS)
Calderon, E; Siergiej, D
2014-01-01
Purpose: Output factor determination for small fields (less than 20 mm) presents significant challenges due to ion chamber volume averaging and diode over-response. Measured output factor values between detectors are known to have large deviations as field sizes are decreased. No set standard to resolve this difference in measurement exists. We observed differences between measured output factors of up to 14% using two different detectors. Published Monte Carlo derived correction factors were used to address this challenge and decrease the output factor deviation between detectors. Methods: Output factors for Elekta's linac-based stereotactic cone system were measured using the EDGE detector (Sun Nuclear) and the A16 ion chamber (Standard Imaging). Measurements conditions were 100 cm SSD (source to surface distance) and 1.5 cm depth. Output factors were first normalized to a 10.4 cm × 10.4 cm field size using a daisy-chaining technique to minimize the dependence of field size on detector response. An equation expressing the relation between published Monte Carlo correction factors as a function of field size for each detector was derived. The measured output factors were then multiplied by the calculated correction factors. EBT3 gafchromic film dosimetry was used to independently validate the corrected output factors. Results: Without correction, the deviation in output factors between the EDGE and A16 detectors ranged from 1.3 to 14.8%, depending on cone size. After applying the calculated correction factors, this deviation fell to 0 to 3.4%. Output factors determined with film agree within 3.5% of the corrected output factors. Conclusion: We present a practical approach to applying published Monte Carlo derived correction factors to measured small field output factors for the EDGE and A16 detectors. Using this method, we were able to decrease the percent deviation between both detectors from 14.8% to 3.4% agreement
International Nuclear Information System (INIS)
Nakagawa, Masayuki; Ishiguro, Yukio; Tokuno, Yukio.
1978-01-01
The self-shielding factors for elastic removal cross sections of light and medium weight nuclides were calculated for the parameter, σ 0 within the conventional concept of the group constant sets. The numerical study were performed for obtaining a simple and accurate method. The present results were compared with the exact values and the conventional ones, and shown to be remarkably improved. It became apparent that the anisotropy of the elastic scattering did not affect to the self-shielding factors though it did to the infinite dilution cross sections. With use of the present revised set, the neutron flux were calculated in an iron medium and in a prototype FBR and compared with those by the fine spectrum calculations and the conventional set. The present set showed the considerable improvement in the vicinity of the large resonance regions of sodium, iron and oxygen. (auth.)
A transient, Hex-Z nodal code corrected by discontinuity factors
International Nuclear Information System (INIS)
Shatilla, Y.A.M.; Henry, A.F.
1993-01-01
This document constitutes Volume 1 of the Final Report of a three-year study supported by the special Research Grant Program for Nuclear Energy Research set up by the US Department of Energy. The original motivation for the work was to provide a fast and accurate computer program for the analysis of transients in heavy water or graphite-moderated reactors being considered as candidates for the New Production Reactor. Thus, part of the funding was by way of pass-through money from the Savannah River Laboratory. With this intent in mind, a three-dimensional (Hex-Z), general-energy-group transient, nodal code was created, programmed, and tested. In order to improve accuracy, correction terms, called open-quotes discontinuity factors,close quotes were incorporated into the nodal equations. Ideal values of these factors force the nodal equations to provide node-integrated reaction rates and leakage rates across nodal surfaces that match exactly those edited from a more exact reference calculation. Since the exact reference solution is needed to compute the ideal discontinuity factors, the fact that they result in exact nodal equations would be of little practical interest were it not that approximate discontinuity factors, found at a greatly reduced cost, often yield very accurate results. For example, for light-water reactors, discontinuity factors found from two-dimensional, fine-mesh, multigroup transport solutions for two-dimensional cuts of a fuel assembly provide very accurate predictions of three-dimensional, full-core power distributions. The present document (volume 1) deals primarily with the specification, programming and testing of the three-dimensional, Hex-Z computer program. The program solves both the static (eigenvalue) and transient, general-energy-group, nodal equations corrected by user-supplied discontinuity factors
Correction of X-ray diffraction profiles in linear-type PSPC by position factor
International Nuclear Information System (INIS)
Takahashi, Toshio
1992-01-01
PSPC (Position Sensitive Proportional Counter) makes it possible to obtain one-dimentional diffraction profiles without mechanical scanning. In a linear-type PSPC, the obtained profiles need correcting, because the position factor influences the intensity of the diffracted X-ray beam and the counting rate at each position on PSPC. The distances from the specimen are not the same at the center and at the edge of the detector, and the intensity decreases at the edge because of radiation and absorption. The counting rate varies with the incident angle of the diffracted beam at each position on PSPC. The position factor f i at channel i of the multichannel-analyser is given by f i = cos 4 α i ·exp{-μR(1/cosα i -1)} where R is the distance between the specimen and the center of PSPC, μ is the linear absorption coefficient and α i is the incident angle of the diffracted beam at channel i. The background profiles of silica gel powder were measured with CrKα and CuKα. The parameters of the model function were fitted to the profiles by the non-linear least squares method. The agreement between these parameters and the calculated values shows that the position factor can correct the measured profiles properly. (author)
International Nuclear Information System (INIS)
Correia, Amanda Ribeiro
2011-01-01
In order to reach therapy and diagnosis objectives, the activity must be determined with high accuracy to administer a radiopharmaceutical to a patient. Initially, a glass vial with the radiopharmaceutical is placed into the radionuclide calibrator to determine its activity. Subsequently, an aliquot is transferred to a syringe and again its activity is measured on the calibrator before being administered to the patient. The glass vial and the syringe are different in many aspects as the calibration factors too, which may cause incorrect activities administered to the patient. This study aims to determine the correction factors, as well as the values of the uncertainties associated to two distinct models of calibrators: one that uses ionization chamber and another Geiger-Mueller as detectors. The radionuclides chosen were 99 Tc m and 123 1 and the containers were glass vials (type lOR and P6) and plastic syringes of 3 and 5 mL. The correction factors for each type of vials or syringe were determined as a function of volume and type of calibrator. Activity measurements comparison was also made involving several radionuclide calibrators of different models belonging to four nuclear medicine hospitals and to National Metrology Laboratory of lionizing Radiation (LNMRI). In the measurements of activity values larger than allowed by CNEN NN-3.05 norm, results have shown deviations for syringes in calibrator with Geiger-Mueller detectors and for both radionuclides. (author)
Directory of Open Access Journals (Sweden)
Adhika Prajna Nandiwardhana
2017-01-01
Full Text Available Penggunaan motor brushless DC telah banyak digunakan dalam berbagai bidang seperti peralatan rumah tangga maupun industri dikarenakan motor ini memiliki struktur yang sederhana, efisiensi dan torsi yang tinggi, serta menggunakan konsep komutasi elektris yang berbeda dari motor DC lainnya. Namun pengoperasian pada umumnya yang menggunakan sumber AC, penyearah serta inverter membuat tingginya nilai harmonisa arus (THD sebesar 73,33% dan power factor sebesar 0,803 dimana nilai ini kurang baik dalam pengaplikasiannya. Pada penelitian ini akan dikaji mengenai proses power factor correction yang mereduksi harmonisa arus (THD sumber AC dengan menggunakan zeta converter dalam pengaplikasian motor brushless DC, serta pengoperasian motor dengan mengamati respon motor terhadap kecepatan referensi yang berubah-ubah dan mengamati kestabilan motor terhadap pembebanan yang bervariasi. Dalam menerapkan metode yang dilakukan pada penelitian ini, pengoperasian motor brushless DC yang telah dirancang dapat bekerja dengan baik meliputi respon motor yang dapat mengikuti kecepatan referensi yang berubah-ubah, serta kestabilan motor dalam mempertahankan kecepatannya pada pembebanan yang bervariasi. Proses power factor correction dapat meningkatkan kualitas daya pada berbagai kecepatan dan mode penerapan yang berbeda-beda, dimana peningkatan tersebut membuktikan kinerja yang baik dalam sistem ini dan memiliki nilai kualitas daya yang baik.
Correction factors for {gamma}-ray relative intensities in the {sup 66}Ga radioisotope
Energy Technology Data Exchange (ETDEWEB)
Schmid, G.J. [Duke Univ., Durham, NC (United States)]|[Triangle Universities Nuclear Lab., Durham, NC (United States); Chasteler, R.M. [Duke Univ., Durham, NC (United States)]|[Triangle Universities Nuclear Lab., Durham, NC (United States); Laymon, C.M. [Duke Univ., Durham, NC (United States)]|[Triangle Universities Nuclear Lab., Durham, NC (United States); Weller, H.R. [Duke Univ., Durham, NC (United States)]|[Triangle Universities Nuclear Lab., Durham, NC (United States); Moore, E.F. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States); Bybee, C.R. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States); Drake, J.M. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States); Tilley, D.R. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States); Vavrina, G. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States); Wallace, P.M. [Triangle Universities Nuclear Lab., Durham, NC (United States)]|[North Carolina State Univ., Raleigh, NC (United States)
1996-09-16
We present here strong evidence that recently published values for the relative intensities of {gamma}-ray lines in the {sup 66}Ga({beta}{sup +}+EC){sup 66}Zn decay are incorrect at the higher energies ({proportional_to}30% too low at 4.8 MeV). In particular, we find that our current results are consistent with a set of correction factors which were first suggested 20 years ago, but have gone largely ignored until now. Our validation of these little known correction factors will have bearing on experiments which use the {sup 66}Ga radioisotope to extrapolate absolute detector efficiencies to higher energies. In particular, we discuss the conclusions of a recent D(p, {gamma}){sup 3}He experiment which will be strongly affected by our current results. The astrophysical S-factor data derived from this D(p, {gamma}){sup 3}He experiment are now seen to be systematically too low by {proportional_to}30%. (orig.).
High Efficiency Three-phase Power Factor Correction Rectifier using Wide Band-Gap Devices
DEFF Research Database (Denmark)
Kouchaki, Alireza
Improving the conversion efficiency of power factor correction (PFC) rectifiers has become compelling due to their wide applications such as adjustable speed drives, uninterruptible power supplies (UPS), and battery chargers for electric vehicles (EVs). The attention to PFCs has increased even more....... Therefore, current controllers are also important to be investigated in this project. In this PhD research work, a comprehensive design of a two-level three-phase PFC rectifier using silicon-carbide (SiC) switches to achieve high efficiency is presented. The work is divided into two main parts: 1) Optimum...
DEFF Research Database (Denmark)
Li, Qingnan; Andersen, Michael A. E.; Thomsen, Ole Cornelius
2011-01-01
Nowadays, efficiency and power density are the most important issues for Power Factor Correction (PFC) converters development. However, it is a challenge to reach both high efficiency and power density in a system at the same time. In this paper, taking a Bridgeless PFC (BPFC) as an example......, a useful compromise between efficiency and power density of the Boost inductors on 3.2kW is achieved using an optimized design procedure. The experimental verifications based on the optimized inductor are carried out from 300W to 3.2kW at 220Vac input....
Three-loop massive form factors: complete light-fermion corrections for the vector current
Lee, Roman N.; Smirnov, Alexander V.; Smirnov, Vladimir A.; Steinhauser, Matthias
2018-03-01
We compute the three-loop QCD corrections to the massive quark-anti-quark-photon form factors F 1 and F 2 involving a closed loop of massless fermions. This subset is gauge invariant and contains both planar and non-planar contributions. We perform the reduction using FIRE and compute the master integrals with the help of differential equations. Our analytic results can be expressed in terms of Goncharov polylogarithms. We provide analytic results for all master integrals which are not present in the large- N c calculation considered in refs. [1, 2].
Factors related to stability following the surgical correction of skeletal open bite.
Ito, Goshi; Koh, Myongsun; Fujita, Tadashi; Shirakura, Maya; Ueda, Hiroshi; Tanne, Kazuo
2014-05-01
If a skeletal anterior open bite malocclusion is treated by orthognathic surgery directed only at the mandible, the lower jaw is repositioned upward in a counter-clockwise rotation. However, this procedure has a high risk of relapse. In the present study, the key factors associated with post-surgical stability of corrected skeletal anterior open bite malocclusions were investigated. Eighteen orthognathic patients were subjected to cephalometric analysis to assess the dental and skeletal changes following mandibular surgery for the correction of an anterior open bite. The patients were divided into two groups, determined by an increase or decrease in nasion-menton (N-Me) distance as a consequence of surgery. Changes in overbite, the displacements of molars and positional changes in Menton were evaluated immediately before and after surgery and after a minimum of one year post-operatively. The group with a decreased N-Me distance exhibited a significantly greater backward positioning of the mandible. The group with an increased N-Me distance experienced significantly greater dentoalveolar extrusion of the lower molars. A sufficient mandibular backward repositioning is an effective technique in the prevention of open bite relapse. In addition, it is important not to induce molar extrusion during post-surgical orthodontic treatment to preserve stability of the surgical open bite correction.
Two-loop master integrals for the mixed EW-QCD virtual corrections to Drell-Yan scattering
Energy Technology Data Exchange (ETDEWEB)
Bonciani, Roberto [' ' La Sapienza' ' Univ., Rome (Italy). Dipt. di Fisica; INFN Sezione Roma (Italy); Di Vita, Stefano [Deutsches Elektronen-Synchrotron (DESY), Hamburg (Germany); Mastrolia, Pierpaolo [Max-Planck-Institut fuer Physik, Muenchen (Germany); Padova Univ. (Italy). Dipt. di Fisica e Astronomia; INFN Sezione di Padova (Italy); Schubert, Ulrich [Max-Planck-Institut fuer Physik, Muenchen (Germany)
2016-04-15
We present the calculation of the master integrals needed for the two-loop QCD x EW corrections to q+ anti q → l{sup -}+l{sup +} and q+ anti q{sup '} → l{sup -}+ anti ν, for massless external particles. We treat W and Z bosons as degenerate in mass. We identify three types of diagrams, according to the presence of massive internal lines: the no-mass type, the one-mass type, and the two-mass type, where all massive propagators, when occurring, contain the same mass value. We find a basis of 49 master integrals and evaluate them with the method of the differential equations. The Magnus exponential is employed to choose a set of master integrals that obeys a canonical system of differential equations. Boundary conditions are found either by matching the solutions onto simpler integrals in special kinematic configurations, or by requiring the regularity of the solution at pseudo-thresholds. The canonical master integrals are finally given as Taylor series around d=4 space-time dimensions, up to order four, with coefficients given in terms of iterated integrals, respectively up to weight four.
International Nuclear Information System (INIS)
Majumdar, Bishnu; Patel, Narayan Prasad; Vijayan, V.
2006-01-01
The aim of this study is to derive the non-uniformity correction factor for the two therapy ionization chambers for the dose measurement near the brachytherapy source. The two ionization chambers of 0.6 cc and 0.1 cc volume were used. The measurement in air was performed for distances between 0.8 cm and 20 cm from the source in specially designed measurement jig. The non-uniformity correction factors were derived from the measured values. The experimentally derived factors were compared with the theoretically calculated non-uniformity correction factors and a close agreement was found between these two studies. The experimentally derived non-uniformity correction factor supports the anisotropic theory. (author)
Directory of Open Access Journals (Sweden)
Francesco Chirico
2017-03-01
Full Text Available Introduction: Over the past two decades, numerous studies on indoor air and the Sick Building Syndrome (SBS have been conducted, mostly in office environments. However, there is little knowledge about SBS in police officers. This study was aimed to fill this gap. Methods: A cross-sectional questionnaire survey was conducted in 2016 at the Triveneto Penitentiary Center, Northern Italy. Chi-square was used to test the difference of prevalence between office workers (OWs and correctional officers (COs of personal characteristics, cases of SBS, and general and mucocutaneous symptoms associated with SBS. A binary logistic regression was used to identify among individual, environmental, and psychosocial characteristics, factors associated with correctional officers’ Sick Building Syndrome. Results: Chi-squared analyses revealed that there were statistically significant differences in the estimated prevalence of SBS general symptoms (χ2 (1 = 12.22, P < .05, SBS mucocutaneous symptoms (χ2 (1 = 9.04, P < .05, and cases of SBS (χ2 (1 = 4.39, P <.05 between COs and OWs. COs reported that their health had been affected by the passive smoking (β = 2.34, P < .05 and unpleasant odour (β = 2.51, P < .05 as environmental risk factors; work-family conflict (β = 2.14, P < .05, psychological and physical isolation (β = 2.07, P < .05, and negative public image (β = 2.06, P < .05 as psychosocial risk factors. Finally, atopy (β = 2.02, P < .05 and to be current smoker (β = 2.02, P < .05 were statistically significant behavioral predictors of SBS among correctional officers. Discussion: Our survey showed that symptoms compatible with the sick building syndrome are common in correctional officers and that psychosocial work climate and exposure to passive smoking could have a strong influence on the prevalence of both general and mucocutaneous symptoms associated with SBS. A health policy for passive tobacco smoking within prisons, and for work-related stress
Electron fluence correction factors for various materials in clinical electron beams
International Nuclear Information System (INIS)
Olivares, M.; Blois, F. de; Podgorsak, E.B.; Seuntjens, J.P.
2001-01-01
Relative to solid water, electron fluence correction factors at the depth of dose maximum in bone, lung, aluminum, and copper for nominal electron beam energies of 9 MeV and 15 MeV of the Clinac 18 accelerator have been determined experimentally and by Monte Carlo calculation. Thermoluminescent dosimeters were used to measure depth doses in these materials. The measured relative dose at d max in the various materials versus that of solid water, when irradiated with the same number of monitor units, has been used to calculate the ratio of electron fluence for the various materials to that of solid water. The beams of the Clinac 18 were fully characterized using the EGS4/BEAM system. EGSnrc with the relativistic spin option turned on was used to optimize the primary electron energy at the exit window, and to calculate depth doses in the five phantom materials using the optimized phase-space data. Normalizing all depth doses to the dose maximum in solid water stopping power ratio corrected, measured depth doses and calculated depth doses differ by less than ±1% at the depth of dose maximum and by less than 4% elsewhere. Monte Carlo calculated ratios of doses in each material to dose in LiF were used to convert the TLD measurements at the dose maximum into dose at the center of the TLD in the phantom material. Fluence perturbation correction factors for a LiF TLD at the depth of dose maximum deduced from these calculations amount to less than 1% for 0.15 mm thick TLDs in low Z materials and are between 1% and 3% for TLDs in Al and Cu phantoms. Electron fluence ratios of the studied materials relative to solid water vary between 0.83±0.01 and 1.55±0.02 for materials varying in density from 0.27 g/cm3 (lung) to 8.96 g/cm3 (Cu). The difference in electron fluence ratios derived from measurements and calculations ranges from -1.6% to +0.2% at 9 MeV and from -1.9% to +0.2% at 15 MeV and is not significant at the 1σ level. Excluding the data for Cu, electron fluence
Estimation of the self-attenuation correction factor for gamma rays emission from nuclear materials
International Nuclear Information System (INIS)
Badawy, A.; El-Gammal, W.A.
2001-01-01
This work presents an investigation of the self-attenuation of gamma-rays emission from nuclear materials (NMs) for measuring the U-235 enrichment, U-235 mass content and isotopic composition of NMs by non-destructive assay technique [NDA]. The measurements then would not need the use of suitable NM Standards which may not be available in many situations. The self-attenuation correction factor (F) may be estimated by the use of the linear attenuation factor of the assayed sample, the geometrical configuration of the assay set-up and the position of the assayed sample relative to the detector. A developed mathematical analysis makes use of specific parameters which affect the estimation of the self-attenuation of the source-detector system which emits passive gamma-rays at certain prominent signatures
Power Factor Correction Capacitors for Multiple Parallel Three-Phase ASD Systems
DEFF Research Database (Denmark)
Yang, Yongheng; Blaabjerg, Frede
2017-01-01
Today’s three-phase Adjustable Speed Drive (ASD) systems still employ Diode Rectifiers (DRs) and Silicon-Controlled Rectifiers (SCRs) as the front-end converters due to structural and control simplicity, small volume, low cost, and high reliability. However, the uncontrollable DRs and phase......-controllable SCRs bring side-effects by injecting high harmonics to the grid, which will degrade the system performance in terms of lowering the overall efficiency and overheating the system if remain uncontrolled or unattenuated. For multiple ASD systems, certain harmonics in the entire system can be mitigated...... the power factor, passive capacitors can be installed, which yet can trigger the system resonance. Hence, this paper analyzes the resonant issues in multiple ASD systems with power factor correction capacitors. Potential damping solutions are summarized. Simulations are carried out, while laboratory tests...
DEFF Research Database (Denmark)
Kouchaki, Alireza; Niroumand, Farideh Javidi; Haase, Frerk
2015-01-01
This paper presents an analytical method for designing the inductor of three-phase power factor correction converters (PFCs). The complex behavior of the inductor current complicates the inductor design procedure as well as the core loss and copper loss calculations. Therefore, this paper analyze...... to calculate the core loss in the PFC application. To investigate the impact of the dc link voltage level, two inductors for different dc voltage levels are designed and the results are compared.......This paper presents an analytical method for designing the inductor of three-phase power factor correction converters (PFCs). The complex behavior of the inductor current complicates the inductor design procedure as well as the core loss and copper loss calculations. Therefore, this paper analyzes...... circuit is used to provide the inductor current harmonic spectrum. Therefore, using the harmonic spectrum, the low and high frequency copper losses are calculated. The high frequency minor B-H loops in one switching cycle are also analyzed. Then, the loss map provided by the measurement setup is used...
Analytic representation of the backscatter correction factor at the exit of high energy photon beams
International Nuclear Information System (INIS)
Kappas, K.; Rosenwald, J.C.
1991-01-01
In high-energy X-ray beams, the dose calculated near the exit surface under electronic equilibrium conditions is generally over-estimated since it is derived from measurements performed in water with large thickness of backscattering material. The resulting error depends on a number of parameters such as beam energy, field dimension, thickness of overlying and underlying material. The authors have systematically measured for 4 different energies and for different para- meters and for different combinations of the above parameters, the reduction of dose due to backscatter. This correction is expressed as a multiplicative factor, called 'Backscatter Correction Factor' (BCF). This BCF is larger for lower energies, larger field sizes and larger depths. The BCF has been represented by an analytical expression which involves an exponential function of the backscattering thickness and linear relationships with depth field size and beam quality index. Using this expression, the BCF can be calculated within 0.5% for any conditions in the energy range investigated. (author). 14 refs.; 4 figs.; 3 tabs
Home radon levels and seasonal correction factors for the Isle of Man
International Nuclear Information System (INIS)
Grainger, P.; Preece, A.W.; Goodfellow, S.A.
2000-01-01
Ionizing radiation dose levels due to home radon can rise to levels that would be illegal for workers in the nuclear industry. It is well known that radon levels within homes and from home to home, and also from month to month, vary considerably. To define an Isle of Man radon seasonal correction factor, readings were taken in eight homes over a 12 month period. An average island indoor exposure of 48 Bq m -3 (range 4-518 Bq m -3 ) was determined from 285 homes selected from a cohort of 1300 families participating in the European Longitudinal Study of Pregnancy and Childhood (ELSPAC) in the Isle of Man. This compares with a UK home average of 20 Bq m -3 and a European Union average (excluding UK) of 68 Bq m -3 . Ten homes of those measured were found to have radon levels above the National Radiological Protection Board 200 Bq m -3 action level. There are 29 377 homes on the Isle of Man, suggesting that there could be some 900 or more homes above the action level. No statistical difference was found between the NRPB and Isle of Man seasonal correction factors. (author)
Application of geometry correction factors for low-level waste package dose measurements. Revision 1
International Nuclear Information System (INIS)
Chandler, M.C.; Parish, B.
1995-01-01
Plans are to determine the Cs-137 content of low-level waste packages generated in High-Level Waste by measuring the radiation level at a specified distance from the package with a hand-held radiation instrument. The measurement taken at this specified distance, either 3 or 5 feet, is called the far-field measurement. This report documents a method for adjusting the gamma exposure rate (mR/hr) reading used in dose-to-curie determinations when the far-field measurement equals the background reading. This adjustment is necessary to reduce the conservatism resulting from using a minimum detection limit exposure rate for the dose-to-curie determination for the far-field measurement position. To accomplish this adjustment, the near-field (5 cm) measurement is multiplied by a geometry correction factor to obtain an estimate of the far field exposure rate (which is below instrument sensitivity). This estimate of the far field exposure rate is used to estimate the Cs-137 curie content of the package. This report establishes the geometry correction factors for the dose-to-curie determination when the far-field gamma exposure measurement equals the background reading. This report also provides a means of demonstrating compliance to 1S Manual requirements for exposure rate readings at different locations from waste packages while specifying only two measurement positions. This demonstration of compliance is necessary to minimize the number of locations exposure rate measurements that are required, i.e., ALARA
Hydrophone area-averaging correction factors in nonlinearly generated ultrasonic beams
International Nuclear Information System (INIS)
Cooling, M P; Humphrey, V F; Wilkens, V
2011-01-01
The nonlinear propagation of an ultrasonic wave can be used to produce a wavefield rich in higher frequency components that is ideally suited to the calibration, or inter-calibration, of hydrophones. These techniques usually use a tone-burst signal, limiting the measurements to harmonics of the fundamental calibration frequency. Alternatively, using a short pulse enables calibration at a continuous spectrum of frequencies. Such a technique is used at PTB in conjunction with an optical measurement technique to calibrate devices. Experimental findings indicate that the area-averaging correction factor for a hydrophone in such a field demonstrates a complex behaviour, most notably varying periodically between frequencies that are harmonics of the centre frequency of the original pulse and frequencies that lie midway between these harmonics. The beam characteristics of such nonlinearly generated fields have been investigated using a finite difference solution to the nonlinear Khokhlov-Zabolotskaya-Kuznetsov (KZK) equation for a focused field. The simulation results are used to calculate the hydrophone area-averaging correction factors for 0.2 mm and 0.5 mm devices. The results clearly demonstrate a number of significant features observed in the experimental investigations, including the variation with frequency, drive level and hydrophone element size. An explanation for these effects is also proposed.
Hydrophone area-averaging correction factors in nonlinearly generated ultrasonic beams
Cooling, M. P.; Humphrey, V. F.; Wilkens, V.
2011-02-01
The nonlinear propagation of an ultrasonic wave can be used to produce a wavefield rich in higher frequency components that is ideally suited to the calibration, or inter-calibration, of hydrophones. These techniques usually use a tone-burst signal, limiting the measurements to harmonics of the fundamental calibration frequency. Alternatively, using a short pulse enables calibration at a continuous spectrum of frequencies. Such a technique is used at PTB in conjunction with an optical measurement technique to calibrate devices. Experimental findings indicate that the area-averaging correction factor for a hydrophone in such a field demonstrates a complex behaviour, most notably varying periodically between frequencies that are harmonics of the centre frequency of the original pulse and frequencies that lie midway between these harmonics. The beam characteristics of such nonlinearly generated fields have been investigated using a finite difference solution to the nonlinear Khokhlov-Zabolotskaya-Kuznetsov (KZK) equation for a focused field. The simulation results are used to calculate the hydrophone area-averaging correction factors for 0.2 mm and 0.5 mm devices. The results clearly demonstrate a number of significant features observed in the experimental investigations, including the variation with frequency, drive level and hydrophone element size. An explanation for these effects is also proposed.
Directory of Open Access Journals (Sweden)
mandana Saki
2009-04-01
Full Text Available Background: Social damage is one of the problems which destroys the active manpower and young population in the country and also prevents social development. Family is the first environment in which social ground for the child is provided.Any dispute, controversy or inferiority complex in the family may lead to wrong doings The present study aims at the effective family and personality factors on felony in felon adolescents kept in Khorramabad bouse of correction. Materials and Methods: This descriptive- analytic study was carried out on all of the adolescents in Khorramabad house of correction as case group selected using census random sampling and high school students as control group selected by cluster random sampling. Data gathering tool was a 2-part questionnaire including demographic information and standard SCL 90 questionnaire. Results: Results showed that there is a significant relationship between the place of residence, failure in school program, economic status of the family, parents presence, fathers job and education, mothers job and education, parents inter personal relationship and family conflicting. The results also showed that aggression, anxiety, depression, somatic complain, obsessive compulsive, inter personality sensitivity, Psychotic, paranoid and phobia had a significant difference in the case and control groups. Conclusion: Regarding the fact that family environment is considered as the most important educative factor and no social damage has not occurred out of family influence, so no society can claim to be healthy unless to have healthy families.
Validation of head scatter factor for an Elekta synergy platform linear accelerator
International Nuclear Information System (INIS)
Johannes, N.B.
2013-07-01
A semi-empirical method has been proposed and developed to model and compute head or collimator scatter factors for 6 and 15 MV photon beams from Elekta Synergy platform linear accelerator at the radiation oncology centre of 'Sweden-Ghana Medical Centre Limited', East Legon Hills in Accra. The proposed model was based on two dimensional Gaussian distribution, whose output was compared to measured head scatter factor data for the linear accelerator obtained during commissioning of the teletherapy machine. The two dimensions Gaussian distribution model used physical specifications and configuration of the head unit (collimator system) of the linear accelerator, which were obtained from the user manual provided by the manufacturer of the linear accelerator. The algorithm for the model was implemented using Matlab software in the Microsoft windows environment. The model was done for both square and rectangular fields, and the output compared with corresponding measured data. The comparisons for the square fields were used to establish an error term in the Gaussian distribution function. The error term was determined by plotting the difference between the output factors from MatLab and the corresponding measured data as function of one side of a square field (equivalent square field). The correlation equation of the curve obtained was chosen as the error term, which was incorporated into the Gaussian distribution function. This was repeated for two photon beam energies (6 and 15 MV). The refined Gaussian distributions were then used to determine head scatter factors for square and rectangular fields. For the rectangular fields, Sterling's proposed formula was used to find equivalent square used to obtain the equivalent square fields found in the error terms of the proposed formula was sed to find equivalent square used to obtain the equivalent square fields found in the error terms of the proposed and developed model. The output of the 2D Gaussian distribution without
Lifescience Database Archive (English)
Full Text Available 16979567 Type I interferon [corrected] gene induction by the interferon regulatory factorfamily...ng) (.svg) (.html) (.csml) Show Type I interferon [corrected] gene induction by the interferon regulatory factorfamily...orrected] gene induction by the interferon regulatory factorfamily of transcription factors. Authors Honda K
International Nuclear Information System (INIS)
Li Lijun; Zhu Haijun; Zhang Xinzhong; Li Feizhou; Song Hongyu
2004-01-01
Objective: To evaluate the method of measurement of total scatter calibrate factor (Sc, p). Methods: To measure the Sc, p at different depths on central axis of 6MV, 15MV photon beams through different ways. Results: It was found that the measured data of Sc, p changed with the different depths to a range of 1% - 7%. Using the direct method, the Sc, p measured depth should be the same as the depth in dose normalization point of the prescription dose. If the Sc, p (fsz, d) was measured at the other depths, it could be obtained indirectly by the calculation formula. Conclusions: The Sc, p in the prescription dose can be obtained either by the direct measure method or the indirect calculation formula. But emphasis should be laid on the proper measure depth. (authors)
Diffractive Production of Jets and Weak Bosons, and Tests of Hard-Scattering Factorization
Alvero, L; Terrón, J; Whitmore, J; Alvero, Lyndon; Collins, John C.; Terron, Juan; Whitmore, Jim
1999-01-01
We extract diffractive parton densities from diffractive, deep inelastic (DIS) ep data from the ZEUS experiment. Then we use these fits to predict the diffractive production of jets and of W's and Z's in p\\bar p collisions at the Tevatron. Although the DIS data require a hard quark density in the pomeron, we find fairly low rates for the Tevatron processes (a few percent of the inclusive cross section). This results from the combined effects of Q^{2} evolution and of a normalization of the parton densities to the data. The calculated rates for W production are generally consistent with the preliminary data from the Tevatron. However, the jet data from CDF with a ``Roman pot'' trigger are substantially lower than the results of our calculations; if confirmed, this would signal a breakdown of hard-scattering factorization.
DEFF Research Database (Denmark)
Larsen, Ole H; Stentoft, Jesper; Radia, Deepti
2013-01-01
Haemostatic treatment modalities alternative to platelet transfusion are desirable to control serious acute bleeds in primary immune thrombocytopenia (ITP). This study challenged the hypothesis that recombinant activated factor VII (rFVIIa) combined with fibrinogen concentrate may correct whole b...
Correction factors to convert microdosimetry measurements in silicon to tissue in 12C ion therapy.
Bolst, David; Guatelli, Susanna; Tran, Linh T; Chartier, Lachlan; Lerch, Michael L F; Matsufuji, Naruhiro; Rosenfeld, Anatoly B
2017-03-21
Silicon microdosimetry is a promising technology for heavy ion therapy (HIT) quality assurance, because of its sub-mm spatial resolution and capability to determine radiation effects at a cellular level in a mixed radiation field. A drawback of silicon is not being tissue-equivalent, thus the need to convert the detector response obtained in silicon to tissue. This paper presents a method for converting silicon microdosimetric spectra to tissue for a therapeutic 12 C beam, based on Monte Carlo simulations. The energy deposition spectra in a 10 μm sized silicon cylindrical sensitive volume (SV) were found to be equivalent to those measured in a tissue SV, with the same shape, but with dimensions scaled by a factor κ equal to 0.57 and 0.54 for muscle and water, respectively. A low energy correction factor was determined to account for the enhanced response in silicon at low energy depositions, produced by electrons. The concept of the mean path length [Formula: see text] to calculate the lineal energy was introduced as an alternative to the mean chord length [Formula: see text] because it was found that adopting Cauchy's formula for the [Formula: see text] was not appropriate for the radiation field typical of HIT as it is very directional. [Formula: see text] can be determined based on the peak of the lineal energy distribution produced by the incident carbon beam. Furthermore it was demonstrated that the thickness of the SV along the direction of the incident 12 C ion beam can be adopted as [Formula: see text]. The tissue equivalence conversion method and [Formula: see text] were adopted to determine the RBE 10 , calculated using a modified microdosimetric kinetic model, applied to the microdosimetric spectra resulting from the simulation study. Comparison of the RBE 10 along the Bragg peak to experimental TEPC measurements at HIMAC, NIRS, showed good agreement. Such agreement demonstrates the validity of the developed tissue equivalence correction factors and of
International Nuclear Information System (INIS)
Burns, D.T.
2002-01-01
Traditionally, the correction factor k wall for attenuation and scatter in the walls of cavity ionization chamber primary standards has been evaluated experimentally using an extrapolation method. During the past decade, there have been a number of Monte Carlo calculations of k wall indicating that for certain ionization chamber types the extrapolation method may not be valid. In particular, values for k wall have been proposed that, if adopted by each laboratory concerned, would have a significant effect on the results of international comparisons of air-kerma primary standards. The calculations have also proposed new values for the axial component k an of the point-source uniformity correction. Central to the results of international comparisons is the BIPM air-kerma standard. Unlike most others, the BIPM standard is of the parallel-plate design for which the extrapolation method for evaluating k wall should be valid. The value in use at present is k wall =1.0026 (standard uncertainty 0.0008). Rogers and Treurniet calculated the value k wall =1.0014 for the BIPM standard, which is in moderate agreement with the value in use (no overall uncertainty was given). However, they also calculated k an =1.0024 (statistical uncertainty 0.0003) which is very different from the value k an =0.9964 (0.0007) in use at present for the BIPM standard. A new 60 Co facility has recently been installed at the BIPM and the opportunity was taken to re-evaluate the correction factors for the BIPM standard in this new beam. Given that almost all of the Monte Carlo work to date has used the EGS Monte Carlo code, it was decided to use the code PENELOPE. The new source, container, head and collimating jaws were simulated in detail with more that fifty components being modelled, as shown. This model was used to create a phase-space file in the plane 90 cm from the source. The normalized distribution of photon number with energy is shown, where the various sources of scattered photons are
International Nuclear Information System (INIS)
Manduci, L.; Tenailleau, L.; Trolet, J.L.; De Vismes, A.; Lopez, G.; Piccione, M.
2010-01-01
The mass attenuation coefficients for a number of marine and terrestrial bioindicators were measured using γ spectrometry for energies between 22 and 80 keV. These values were then used to find the correction factor k for the apparent radioactivity. The experimental results were compared with a Monte Carlo simulation performed using PENELOPE in order to evaluate the reliability of the simplified calculation and to determine the correction factors.
A novel correction factor based on extended volume to complement the conformity index.
Jin, F; Wang, Y; Wu, Y-Z
2012-08-01
We propose a modified conformity index (MCI), based on extended volume, that improves on existing indices by correcting for the insensitivity of previous conformity indices to reference dose shape to assess the quality of high-precision radiation therapy and present an evaluation of its application. In this paper, the MCI is similar to the conformity index suggested by Paddick (CI(Paddick)), but with a different correction factor. It is shown for three cases: with an extended target volume, with an extended reference dose volume and without an extended volume. Extended volume is generated by expanding the original volume by 0.1-1.1 cm isotropically. Focusing on the simulation model, measurements of MCI employ a sphere target and three types of reference doses: a sphere, an ellipsoid and a cube. We can constrain the potential advantage of the new index by comparing MCI with CI(Paddick). The measurements of MCI in head-neck cancers treated with intensity-modulated radiation therapy and volumetric-modulated arc therapy provide a window on its clinical practice. The results of MCI for a simulation model and clinical practice are presented and the measurements are corrected for limited spatial resolution. The three types of MCI agree with each other, and comparisons between the MCI and CI(Paddick) are also provided. The results from our analysis show that the proposed MCI can provide more objective and accurate conformity measurement for high-precision radiation therapy. In combination with a dose-volume histogram, it will be a more useful conformity index.
International Nuclear Information System (INIS)
Araki, Fujio
2007-01-01
In some recent dosimetry protocols, plastic is allowed as a phantom material for the determination of an absorbed dose to water in electron beams, especially for low energy with beam qualities R 50 2 . In electron dosimetry with plastic, a depth-scaling factor, c pl , and a chamber-dependent fluence correction factor, h pl , are needed to convert the dose measured at a water-equivalent reference depth in plastic to a dose at a reference depth in water. The purpose of this study is to calculate correction factors for the use of plastic phantoms for clinical electron dosimetry using the EGSnrc Monte Carlo code system. RMI-457 and WE-211 were investigated as phantom materials. First the c pl values for plastic materials were calculated as a function of a half-value depth of maximum ionization, I 50 , in plastic. The c pl values for RMI-457 and WE-211 varied from 0.992 to 1.002 and from 0.971 to 0.979, respectively, in a range of nominal energies from 4 MeV to 18 MeV, and varied slightly as a function of I 50 in plastic. Since h pl values depend on the wall correction factor, P wall , of the chamber used, they are evaluated using a pure electron fluence correction factor, φ pl w , and P wall w and P wall pl for a combination of water or plastic phantoms and plane-parallel ionization chambers (NACP-02, Markus and Roos). The φ pl w and P wall (P wall w and P wall pl ) values were calculated as a function of the water-equivalent depth in plastic materials and at a reference depth as a function of R 50 in water, respectively. The φ pl w values varied from 1.024 at 4 MeV to 1.013 at 18 MeV for RMI-457, and from 1.025 to 1.016 for WE-211. P wall w values for plane-parallel chambers showed values in the order of 1.5% to 2% larger than unity at 4 MeV, consistent with earlier results. The P wall pl values of RMI-457 and WE-211 were close to unity for all the energy beams. Finally, calculated h pl values of RMI-457 ranged from 1.009 to 1.005, from 1.010 to 1.003 and from 1
Energy Technology Data Exchange (ETDEWEB)
Wang, A; Paysan, P; Brehm, M; Maslowski, A; Lehmann, M; Messmer, P; Munro, P; Yoon, S; Star-Lack, J; Seghers, D [Varian Medical Systems, Palo Alto, CA (United States)
2016-06-15
Purpose: To improve CBCT image quality for image-guided radiotherapy by applying advanced reconstruction algorithms to overcome scatter, noise, and artifact limitations Methods: CBCT is used extensively for patient setup in radiotherapy. However, image quality generally falls short of diagnostic CT, limiting soft-tissue based positioning and potential applications such as adaptive radiotherapy. The conventional TrueBeam CBCT reconstructor uses a basic scatter correction and FDK reconstruction, resulting in residual scatter artifacts, suboptimal image noise characteristics, and other artifacts like cone-beam artifacts. We have developed an advanced scatter correction that uses a finite-element solver (AcurosCTS) to model the behavior of photons as they pass (and scatter) through the object. Furthermore, iterative reconstruction is applied to the scatter-corrected projections, enforcing data consistency with statistical weighting and applying an edge-preserving image regularizer to reduce image noise. The combined algorithms have been implemented on a GPU. CBCT projections from clinically operating TrueBeam systems have been used to compare image quality between the conventional and improved reconstruction methods. Planning CT images of the same patients have also been compared. Results: The advanced scatter correction removes shading and inhomogeneity artifacts, reducing the scatter artifact from 99.5 HU to 13.7 HU in a typical pelvis case. Iterative reconstruction provides further benefit by reducing image noise and eliminating streak artifacts, thereby improving soft-tissue visualization. In a clinical head and pelvis CBCT, the noise was reduced by 43% and 48%, respectively, with no change in spatial resolution (assessed visually). Additional benefits include reduction of cone-beam artifacts and reduction of metal artifacts due to intrinsic downweighting of corrupted rays. Conclusion: The combination of an advanced scatter correction with iterative reconstruction
Real-Gas Correction Factors for Hypersonic Flow Parameters in Helium
Erickson, Wayne D.
1960-01-01
The real-gas hypersonic flow parameters for helium have been calculated for stagnation temperatures from 0 F to 600 F and stagnation pressures up to 6,000 pounds per square inch absolute. The results of these calculations are presented in the form of simple correction factors which must be applied to the tabulated ideal-gas parameters. It has been shown that the deviations from the ideal-gas law which exist at high pressures may cause a corresponding significant error in the hypersonic flow parameters when calculated as an ideal gas. For example the ratio of the free-stream static to stagnation pressure as calculated from the thermodynamic properties of helium for a stagnation temperature of 80 F and pressure of 4,000 pounds per square inch absolute was found to be approximately 13 percent greater than that determined from the ideal-gas tabulation with a specific heat ratio of 5/3.
International Nuclear Information System (INIS)
Kuahara, Lilian T.; Correa, Eduardo L.; Potiens, Maria da Penha A.
2015-01-01
The aim of this study is to present the geometry correction factors and their respective uncertainties to P6 and 10R vials and 3 mL and 5 mL plastic syringes using the reference activity meters present at the Laboratorio de Calibracao de Instrumentos (LCI), IPEN the secondary standard system Capintec NPL-CRC radionuclide calibrator, with traceability to the National Physics Laboratory (NPL), England and the work standard Capintec CRC-15BT and the Capintec CRC-25R. The procedure was made using 99m Tc, which is responsible for about 80 % of the routine procedures in a nuclear medicine service. Variations of up to 40 % between different vials using the same radionuclide were found. (author)
On the evaluation of the correction factor μ (rho', tau') for the periodic pulse method
International Nuclear Information System (INIS)
Mueller, J.W.
1976-01-01
The inconveniences associated with the purely numerical approach we have chosen to solve some of the problems which arise in connection with the source-pulser method are twofold. On the one hand, there is the trouble of calculating the tables for μ, requiring several nights of computer time. On the other hand, apart from some simple limiting values as μ = 1 for tau' = 0 or 1, μ = 1/0.5 + /0.5 - tau'/ for rho' → 0 (and 0 > 1, no appropriate analytical form for the correction factor μ of sufficient precision is known for the moment. This drawback, we hope, is partly removed by a tabulation which should cover the whole region of practical interest. The computer programs for both the evaluation of μ and the Monte Carlo simulation are available upon request
Correction factors and performance of a 4 degrees C sealed water calorimeter.
Seuntjens, J; Palmans, H
1999-03-01
In the past two decades, the water calorimetry technique for determination of absorbed dose to water in several types of radiation beams has moved significantly closer to being a recognized method. In this paper we summarize the constructional details of a 4 degrees C sealed water calorimeter currently in operation at the University of Gent. This sealed water (SW) calorimeter is of the Domen type and has been improved in several aspects compared with its original design. The relevant correction factors for heat transport and for field perturbation are described. Using relative response measurements in 60Co, we experimentally verified the relative heat defect for two distinct chemical systems, using two different detection vessel arrangements. The overall 1sigma uncertainty on the absorbed dose to water at 60Co based on this system amounts to 0.7%.
Mitigation of voltage sag, swell and power factor correction using solid-state transformer b
Directory of Open Access Journals (Sweden)
M.R. Banaei
2014-09-01
Full Text Available This paper presents a novel topology of solid-state transformer (SST. In the design process, the AC/DC, DC/AC and AC/AC converters have been integrated to achieve higher efficiency. To obtain higher efficiency from other SST with DC-link topologies, the AC/DC and DC/AC converters have been integrated in one matrix converter. The proposed SST performs typical functions and has advantages such as power factor correction, voltage sag and swell elimination, voltage flicker reduction and protection capability in fault situations. In addition, it has other benefits such as light weight, low volume and elimination of hazardous liquid dielectrics because it uses medium frequency transformer. The operation and some performances of the proposed SST have been verified by the simulation results.
Description of corrections on electrode polarization impedance using isopotential interface factor
Directory of Open Access Journals (Sweden)
John Alexander Gomez Sanchez
2012-08-01
Full Text Available In this paper, we propose an equation and define the Isopotential Interface Factor (IIF to quantify the contribution of electrode polarization impedance in two tetrapolar electrode shapes. The first tetrapolar electrode geometry shape was adjacent and the second axial concentric, both probes were made of stainless steel (AISI 304. The experiments were carried out with an impedance analyzer (Solartron 1260 using a frequency range between 0.1 Hz and 8 MHz. Based on a theoretical simplification, the experimental results show a lower value of the IIF in the axial concentric tetrapolar electrode system which caused a lower correction of interface value. The higher value of the IIF in the adjacent electrode system was KEEI (1Hz, 0.28 mS/cm = 1.41 and decreased when the frequency and conductance were increased, whereas in the axial concentric electrode system was KEEI (1Hz, 0.28 mS/cm = 0.08. The average isopotential interface factor throughout the whole range of conductivities and frequencies was 0.23 in the adjacent electrode system and 0.02 in the axial concentric electrode system. The index of inherent electrical anisotropy (IEA was used to present an analysis of electrical anisotropy of biceps brachii muscle in vitro using the corrections of both tetrapolar electrode systems. A higher IEA was present in lower frequency where the variation below 1 kHz was 15 % in adjacent electrode configuration and 26 % in the axial concentric probe with respect to full range. The IIF is then shown that it can be used to describe the quality of an electrode system.
Mann, Kulwinder Singh
2018-01-01
Scattered photon's influence on measured values of attenuation coefficients (μm, cm2g-1) for six low-Z (effective atomic number) building materials, at three photon energies has been estimated. Narrow-beam transmission geometry has been used for the measurements. Samples of commonly used engineering materials (Cements, Clay, Lime-Stone, Plaster of Paris) have been selected for the present study. Standard radioactive sources Cs137 and Co60 have been used for obtaining γ-ray energies 661.66, 1173.24 and 1332.50 keV. The optical thickness (OT) of 0.5 mfp (mean free path) has been found the optimum optical thickness (OOT) for μm-measurement in the selected energy range (661.66-1332.50 keV). The aim of this investigation is to provide neglected information regarding subsistence of scattered photons in narrow beam geometry measurements for low-Z materials. The measurements have been performed for a wide range of sample-thickness (2-26 cm) such that their OT varies between 0.2-3.5 mfp in selected energy range. A computer program (GRIC2-toolkit) has been used for various theoretical computations required in this investigation. It has been concluded that in selected energy-range, good accuracy in μm-measurement of low-Z materials can be achieved by keeping their sample's OT below 0.5 mfp. The exposure buildup factors have been measured with the help of mathematical-model developed in this investigation.
Energy Technology Data Exchange (ETDEWEB)
Hess-Flores, Mauricio [Univ. of California, Davis, CA (United States)
2011-11-10
Scene reconstruction from video sequences has become a prominent computer vision research area in recent years, due to its large number of applications in fields such as security, robotics and virtual reality. Despite recent progress in this field, there are still a number of issues that manifest as incomplete, incorrect or computationally-expensive reconstructions. The engine behind achieving reconstruction is the matching of features between images, where common conditions such as occlusions, lighting changes and texture-less regions can all affect matching accuracy. Subsequent processes that rely on matching accuracy, such as camera parameter estimation, structure computation and non-linear parameter optimization, are also vulnerable to additional sources of error, such as degeneracies and mathematical instability. Detection and correction of errors, along with robustness in parameter solvers, are a must in order to achieve a very accurate final scene reconstruction. However, error detection is in general difficult due to the lack of ground-truth information about the given scene, such as the absolute position of scene points or GPS/IMU coordinates for the camera(s) viewing the scene. In this dissertation, methods are presented for the detection, factorization and correction of error sources present in all stages of a scene reconstruction pipeline from video, in the absence of ground-truth knowledge. Two main applications are discussed. The first set of algorithms derive total structural error measurements after an initial scene structure computation and factorize errors into those related to the underlying feature matching process and those related to camera parameter estimation. A brute-force local correction of inaccurate feature matches is presented, as well as an improved conditioning scheme for non-linear parameter optimization which applies weights on input parameters in proportion to estimated camera parameter errors. Another application is in
Energy Technology Data Exchange (ETDEWEB)
Rutsch, G. [Pittsburgh Univ., PA (United States). Dept. of Physics and Astronomy]|[Dept. of Electrical Engineering, Univ. of Pittsburgh, PA (United States); Devaty, R.P.; Choyke, W.J. [Pittsburgh Univ., PA (United States). Dept. of Physics and Astronomy; Langer, D.W. [Dept. of Electrical Engineering, Univ. of Pittsburgh, PA (United States); Rowland, L.B. [Northrop Grumman Science and Technology Center, Pittsburgh, PA (United States)
1998-06-01
In order to accurately extract material properties from temperature dependent Hall measurements, it is necessary to know the Hall scattering factor (r{sub H}), among other material properties. We present measurements of the Hall scattering factor on nitrogen doped 4H SiC epitaxial layers from 40 K to room temperature in magnetic fields up to 9 Tesla. The measured effective Hall scattering factor varies from 0.91 to 1.21. (orig.) 6 refs.
Gordon, Howard R.; Wang, Menghua
1992-01-01
The first step in the Coastal Zone Color Scanner (CZCS) atmospheric-correction algorithm is the computation of the Rayleigh-scattering (RS) contribution, L sub r, to the radiance leaving the top of the atmosphere over the ocean. In the present algorithm, L sub r is computed by assuming that the ocean surface is flat. Calculations of the radiance leaving an RS atmosphere overlying a rough Fresnel-reflecting ocean are presented to evaluate the radiance error caused by the flat-ocean assumption. Simulations are carried out to evaluate the error incurred when the CZCS-type algorithm is applied to a realistic ocean in which the surface is roughened by the wind. In situations where there is no direct sun glitter, it is concluded that the error induced by ignoring the Rayleigh-aerosol interaction is usually larger than that caused by ignoring the surface roughness. This suggests that, in refining algorithms for future sensors, more effort should be focused on dealing with the Rayleigh-aerosol interaction than on the roughness of the sea surface.
International Nuclear Information System (INIS)
Mathew, D; Tanny, S; Parsai, E; Sperling, N
2015-01-01
Purpose: The current small field dosimetry formalism utilizes quality correction factors to compensate for the difference in detector response relative to dose deposited in water. The correction factors are defined on a machine-specific basis for each beam quality and detector combination. Some research has suggested that the correction factors may only be weakly dependent on machine-to-machine variations, allowing for determinations of class-specific correction factors for various accelerator models. This research examines the differences in small field correction factors for three detectors across two Varian Truebeam accelerators to determine the correction factor dependence on machine-specific characteristics. Methods: Output factors were measured on two Varian Truebeam accelerators for equivalently tuned 6 MV and 6 FFF beams. Measurements were obtained using a commercial plastic scintillation detector (PSD), two ion chambers, and a diode detector. Measurements were made at a depth of 10 cm with an SSD of 100 cm for jaw-defined field sizes ranging from 3×3 cm 2 to 0.6×0.6 cm 2 , normalized to values at 5×5cm 2 . Correction factors for each field on each machine were calculated as the ratio of the detector response to the PSD response. Percent change of correction factors for the chambers are presented relative to the primary machine. Results: The Exradin A26 demonstrates a difference of 9% for 6×6mm 2 fields in both the 6FFF and 6MV beams. The A16 chamber demonstrates a 5%, and 3% difference in 6FFF and 6MV fields at the same field size respectively. The Edge diode exhibits less than 1.5% difference across both evaluated energies. Field sizes larger than 1.4×1.4cm2 demonstrated less than 1% difference for all detectors. Conclusion: Preliminary results suggest that class-specific correction may not be appropriate for micro-ionization chamber. For diode systems, the correction factor was substantially similar and may be useful for class-specific reference
Energy Technology Data Exchange (ETDEWEB)
Mathew, D; Tanny, S; Parsai, E; Sperling, N [University of Toledo Medical Center, Toledo, OH (United States)
2015-06-15
Purpose: The current small field dosimetry formalism utilizes quality correction factors to compensate for the difference in detector response relative to dose deposited in water. The correction factors are defined on a machine-specific basis for each beam quality and detector combination. Some research has suggested that the correction factors may only be weakly dependent on machine-to-machine variations, allowing for determinations of class-specific correction factors for various accelerator models. This research examines the differences in small field correction factors for three detectors across two Varian Truebeam accelerators to determine the correction factor dependence on machine-specific characteristics. Methods: Output factors were measured on two Varian Truebeam accelerators for equivalently tuned 6 MV and 6 FFF beams. Measurements were obtained using a commercial plastic scintillation detector (PSD), two ion chambers, and a diode detector. Measurements were made at a depth of 10 cm with an SSD of 100 cm for jaw-defined field sizes ranging from 3×3 cm{sup 2} to 0.6×0.6 cm{sup 2}, normalized to values at 5×5cm{sup 2}. Correction factors for each field on each machine were calculated as the ratio of the detector response to the PSD response. Percent change of correction factors for the chambers are presented relative to the primary machine. Results: The Exradin A26 demonstrates a difference of 9% for 6×6mm{sup 2} fields in both the 6FFF and 6MV beams. The A16 chamber demonstrates a 5%, and 3% difference in 6FFF and 6MV fields at the same field size respectively. The Edge diode exhibits less than 1.5% difference across both evaluated energies. Field sizes larger than 1.4×1.4cm2 demonstrated less than 1% difference for all detectors. Conclusion: Preliminary results suggest that class-specific correction may not be appropriate for micro-ionization chamber. For diode systems, the correction factor was substantially similar and may be useful for class
International Nuclear Information System (INIS)
Arbuzov, A.; Kalinovskaya, L.; Bardin, D.; Deutsches Elektronen-Synchrotron; Bluemlein, J.; Riemann, T.
1995-11-01
A description of the Fortran program HECTOR for a variety of semi-analytical calculations of radiative QED, QCD, and electroweak corrections to the double-differential cross sections of NC and CC deep inelastic charged lepton proton (or lepton deuteron) scattering is presented. HECTOR originates from the substantially improved and extended earlier programs HELIOS and TERAD91. It is mainly intended for applications at HERA or LEP x LHC, but may be used also for μN scattering in fixed target experiments. The QED corrections may be calculated in different sets of variables: leptonic, hadronic, mixed, Jaquet-Blondel, double angle etc. Besides the leading logarithmic approximation up to order O(α 2 ), exact O(α) corrections and inclusive soft photon exponentiation are taken into account. The photoproduction region is also covered. (orig.)
International Nuclear Information System (INIS)
Laitano, R.F.; Toni, M.P.; Pimpinella, M.; Bovi, M.
2002-01-01
The factor K wall to correct for photon attenuation and scatter in the wall of ionization chambers for 60 Co air-kerma measurement has been traditionally determined by a procedure based on a linear extrapolation of the chamber current to zero wall thickness. Monte Carlo calculations by Rogers and Bielajew (1990 Phys. Med. Biol. 35 1065-78) provided evidence, mostly for chambers of cylindrical and spherical geometry, of appreciable deviations between the calculated values of K wall and those obtained by the traditional extrapolation procedure. In the present work an experimental method other than the traditional extrapolation procedure was used to determine the K wall factor. In this method the dependence of the ionization current in a cylindrical chamber was analysed as a function of an effective wall thickness in place of the physical (radial) wall thickness traditionally considered in this type of measurement. To this end the chamber wall was ideally divided into distinct regions and for each region an effective thickness to which the chamber current correlates was determined. A Monte Carlo calculation of attenuation and scatter effects in the different regions of the chamber wall was also made to compare calculation to measurement results. The K wall values experimentally determined in this work agree within 0.2% with the Monte Carlo calculation. The agreement between these independent methods and the appreciable deviation (up to about 1%) between the results of both these methods and those obtained by the traditional extrapolation procedure support the conclusion that the two independent methods providing comparable results are correct and the traditional extrapolation procedure is likely to be wrong. The numerical results of the present study refer to a cylindrical cavity chamber like that adopted as the Italian national air-kerma standard at INMRI-ENEA (Italy). The method used in this study applies, however, to any other chamber of the same type. (author)