Algorithm Development for Multi-Energy SXR based Electron Temperature Profile Reconstruction
Clayton, D. J.; Tritz, K.; Finkenthal, M.; Kumar, D.; Stutman, D.
2012-10-01
New techniques utilizing computational tools such as neural networks and genetic algorithms are being developed to infer plasma electron temperature profiles on fast time scales (> 10 kHz) from multi-energy soft-x-ray (ME-SXR) diagnostics. Traditionally, a two-foil SXR technique, using the ratio of filtered continuum emission measured by two SXR detectors, has been employed on fusion devices as an indirect method of measuring electron temperature. However, these measurements can be susceptible to large errors due to uncertainties in time-evolving impurity density profiles, leading to unreliable temperature measurements. To correct this problem, measurements using ME-SXR diagnostics, which use three or more filtered SXR arrays to distinguish line and continuum emission from various impurities, in conjunction with constraints from spectroscopic diagnostics, can be used to account for unknown or time evolving impurity profiles [K. Tritz et al, Bull. Am. Phys. Soc. Vol. 56, No. 12 (2011), PP9.00067]. On NSTX, ME-SXR diagnostics can be used for fast (10-100 kHz) temperature profile measurements, using a Thomson scattering diagnostic (60 Hz) for periodic normalization. The use of more advanced algorithms, such as neural network processing, can decouple the reconstruction of the temperature profile from spectral modeling.
International Nuclear Information System (INIS)
Qian Jinping; Ren Qilong; Wan Baonian; Liu Haiqin; Zeng Long; Luo Zhengping; Chen Dalong; Shi Tonghui; Sun Youwen; Shen Biao; Xiao Bingjia; Lao, L. L.; Hanada, K.
2015-01-01
The method of plasma current profile reconstruction using the polarimeter/interferometer (POINT) data from a simulated equilibrium is explored and validated. It is shown that the safety factor (q) profile can be generally reconstructed from the external magnetic and POINT data. The reconstructed q profile is found to reasonably agree with the initial equilibriums. Comparisons of reconstructed q and density profiles using the magnetic data and the POINT data with 3%, 5% and 10% random errors are investigated. The result shows that the POINT data could be used to a reasonably accurate determination of the q profile. (fusion engineering)
Zheng, Shijie; Zhang, Nan; Xia, Yanjun; Wang, Hongtao
2014-03-01
A new heuristic strategy for the non-uniform strain profile reconstruction along Fiber Bragg Gratings is proposed in this paper, which is based on the modified transfer matrix and Genetic Programming(GP) algorithm. The present method uses Genetic Programming to determine the applied strain field as a function of position along the fiber length. The structures that undergo adaptation in genetic programming are hierarchical structures which are different from that of conventional genetic algorithm operating on strings. GP regress the strain profile function which matches the 'measured' spectrum best and makes space resolution of strain reconstruction arbitrarily high, or even infinite. This paper also presents an experimental verification of the reconstruction of non-homogeneous strain fields using GP. The results are compared with numerical calculations of finite element method. Both the simulation examples and experimental results demonstrate that Genetic Programming can effectively reconstruct continuous profile expression along the whole FBG, and greatly improves its computational efficiency and accuracy.
Zhou, Chaojie; Ding, Xiaohua; Zhang, Jie; Yang, Jungang; Ma, Qiang
2017-12-01
While global oceanic surface information with large-scale, real-time, high-resolution data is collected by satellite remote sensing instrumentation, three-dimensional (3D) observations are usually obtained from in situ measurements, but with minimal coverage and spatial resolution. To meet the needs of 3D ocean investigations, we have developed a new algorithm to reconstruct the 3D ocean temperature field based on the Array for Real-time Geostrophic Oceanography (Argo) profiles and sea surface temperature (SST) data. The Argo temperature profiles are first optimally fitted to generate a series of temperature functions of depth, with the vertical temperature structure represented continuously. By calculating the derivatives of the fitted functions, the calculation of the vertical temperature gradient of the Argo profiles at an arbitrary depth is accomplished. A gridded 3D temperature gradient field is then found by applying inverse distance weighting interpolation in the horizontal direction. Combined with the processed SST, the 3D temperature field reconstruction is realized below the surface using the gridded temperature gradient. Finally, to confirm the effectiveness of the algorithm, an experiment in the Pacific Ocean south of Japan is conducted, for which a 3D temperature field is generated. Compared with other similar gridded products, the reconstructed 3D temperature field derived by the proposed algorithm achieves satisfactory accuracy, with correlation coefficients of 0.99 obtained, including a higher spatial resolution (0.25° × 0.25°), resulting in the capture of smaller-scale characteristics. Finally, both the accuracy and the superiority of the algorithm are validated.
UV Reconstruction Algorithm And Diurnal Cycle Variability
Curylo, Aleksander; Litynska, Zenobia; Krzyscin, Janusz; Bogdanska, Barbara
2009-03-01
UV reconstruction is a method of estimation of surface UV with the use of available actinometrical and aerological measurements. UV reconstruction is necessary for the study of long-term UV change. A typical series of UV measurements is not longer than 15 years, which is too short for trend estimation. The essential problem in the reconstruction algorithm is the good parameterization of clouds. In our previous algorithm we used an empirical relation between Cloud Modification Factor (CMF) in global radiation and CMF in UV. The CMF is defined as the ratio between measured and modelled irradiances. Clear sky irradiance was calculated with a solar radiative transfer model. In the proposed algorithm, the time variability of global radiation during the diurnal cycle is used as an additional source of information. For elaborating an improved reconstruction algorithm relevant data from Legionowo [52.4 N, 21.0 E, 96 m a.s.l], Poland were collected with the following instruments: NILU-UV multi channel radiometer, Kipp&Zonen pyranometer, radiosonde profiles of ozone, humidity and temperature. The proposed algorithm has been used for reconstruction of UV at four Polish sites: Mikolajki, Kolobrzeg, Warszawa-Bielany and Zakopane since the early 1960s. Krzyscin's reconstruction of total ozone has been used in the calculations.
Computed laminography and reconstruction algorithm
International Nuclear Information System (INIS)
Que Jiemin; Cao Daquan; Zhao Wei; Tang Xiao
2012-01-01
Computed laminography (CL) is an alternative to computed tomography if large objects are to be inspected with high resolution. This is especially true for planar objects. In this paper, we set up a new scanning geometry for CL, and study the algebraic reconstruction technique (ART) for CL imaging. We compare the results of ART with variant weighted functions by computer simulation with a digital phantom. It proves that ART algorithm is a good choice for the CL system. (authors)
Beard reconstruction: A surgical algorithm.
Ninkovic, M; Heidekrueger, P I; Ehrl, D; von Spiegel, F; Broer, P N
2016-06-01
Facial defects with loss of hair-bearing regions can be caused by trauma, infection, tumor excision, or burn injury. The presented analysis evaluates a series of different surgical approaches with a focus on male beard reconstruction, emphasizing the role of tissue expansion of regional and free flaps. Locoregional and free flap reconstructions were performed in 11 male patients with 14 facial defects affecting the hair-bearing bucco-mandibular or perioral region. In order to minimize donor-site morbidity and obtain large amounts of thin, pliable, hair-bearing tissue, pre-expansion was performed in five of 14 patients. Eight of 14 patients were treated with locoregional flap reconstructions and six with free flap reconstructions. Algorithms regarding pre- and intraoperative decision making are discussed and long-term (mean follow-up 1.5 years) results analyzed. Major complications, including tissue expander infection with the need for removal or exchange, partial or full flap loss, occurred in 0% (0/8) of patients with locoregional flaps and in 17% (1/6) of patients undergoing free flap reconstructions. Secondary refinement surgery was performed in 25% (2/8) of locoregional flaps and in 67% (4/6) of free flaps. Both locoregional and distant tissue transfers play a role in beard reconstruction, while pre-expansion remains an invaluable tool. Paying attention to the presented principles and considering the significance of aesthetic facial subunits, range of motion, aesthetics, and patient satisfaction were improved long term in all our patients while minimizing donor-site morbidity. Copyright © 2016 British Association of Plastic, Reconstructive and Aesthetic Surgeons. Published by Elsevier Ltd. All rights reserved.
Algorithms for reconstructing images for industrial applications
International Nuclear Information System (INIS)
Lopes, R.T.; Crispim, V.R.
1986-01-01
Several algorithms for reconstructing objects from their projections are being studied in our Laboratory, for industrial applications. Such algorithms are useful locating the position and shape of different composition of materials in the object. A Comparative study of two algorithms is made. The two investigated algorithsm are: The MART (Multiplicative - Algebraic Reconstruction Technique) and the Convolution Method. The comparison are carried out from the point view of the quality of the image reconstructed, number of views and cost. (Author) [pt
Park, Doo Y; Fessier, Jeffrey A; Yost, Michael G; Levine, Steven P
2000-03-01
Computed tomographic (CT) reconstructions of air contaminant concentration fields were conducted in a room-sized chamber employing a single open-path Fourier transform infrared (OP-FTIR) instrument and a combination of 52 flat mirrors and 4 retroreflectors. A total of 56 beam path data were repeatedly collected for around 1 hr while maintaining a stable concentration gradient. The plane of the room was divided into 195 pixels (13 × 15) for reconstruction. The algebraic reconstruction technique (ART) failed to reconstruct the original concentration gradient patterns for most cases. These poor results were caused by the "highly underdetermined condition" in which the number of unknown values (156 pixels) exceeds that of known data (56 path integral concentrations) in the experimental setting. A new CT algorithm, called the penalized weighted least-squares (PWLS), was applied to remedy this condition. The peak locations were correctly positioned in the PWLS-CT reconstructions. A notable feature of the PWLS-CT reconstructions was a significant reduction of highly irregular noise peaks found in the ART-CT reconstructions. However, the peak heights were slightly reduced in the PWLS-CT reconstructions due to the nature of the PWLS algorithm. PWLS could converge on the original concentration gradient even when a fairly high error was embedded into some experimentally measured path integral concentrations. It was also found in the simulation tests that the PWLS algorithm was very robust with respect to random errors in the path integral concentrations. This beam geometry and the use of a single OP-FTIR scanning system, in combination with the PWLS algorithm, is a system applicable to both environmental and industrial settings.
A combinational fast algorithm for image reconstruction
International Nuclear Information System (INIS)
Wu Zhongquan
1987-01-01
A combinational fast algorithm has been developed in order to increase the speed of reconstruction. First, an interpolation method based on B-spline functions is used in image reconstruction. Next, the influence of the boundary conditions assumed here on the interpolation of filtered projections and on the image reconstruction is discussed. It is shown that this boundary condition has almost no influence on the image in the central region of the image space, because the error of interpolation rapidly decreases by a factor of ten in shifting two pixels from the edge toward the center. In addition, a fast algorithm for computing the detecting angle has been used with the mentioned interpolation algorithm, and the cost for detecting angle computaton is reduced by a factor of two. The implementation results show that in the same subjective and objective fidelity, the computational cost for the interpolation using this algorithm is about one-twelfth of the conventional algorithm
Tau reconstruction and identification algorithm
Indian Academy of Sciences (India)
CMS has developed sophisticated tau identification algorithms for tau hadronic decay modes. Production of tau lepton decaying to hadrons are studied at 7 TeV centre-of-mass energy with 2011 collision data collected by CMS detector and has been used to measure the performance of tau identification algorithms by ...
Wind reconstruction algorithm for Viking Lander 1
Kynkäänniemi, Tuomas; Kemppinen, Osku; Harri, Ari-Matti; Schmidt, Walter
2017-06-01
The wind measurement sensors of Viking Lander 1 (VL1) were only fully operational for the first 45 sols of the mission. We have developed an algorithm for reconstructing the wind measurement data after the wind measurement sensor failures. The algorithm for wind reconstruction enables the processing of wind data during the complete VL1 mission. The heater element of the quadrant sensor, which provided auxiliary measurement for wind direction, failed during the 45th sol of the VL1 mission. Additionally, one of the wind sensors of VL1 broke down during sol 378. Regardless of the failures, it was still possible to reconstruct the wind measurement data, because the failed components of the sensors did not prevent the determination of the wind direction and speed, as some of the components of the wind measurement setup remained intact for the complete mission. This article concentrates on presenting the wind reconstruction algorithm and methods for validating the operation of the algorithm. The algorithm enables the reconstruction of wind measurements for the complete VL1 mission. The amount of available sols is extended from 350 to 2245 sols.
Wind reconstruction algorithm for Viking Lander 1
Directory of Open Access Journals (Sweden)
T. Kynkäänniemi
2017-06-01
Full Text Available The wind measurement sensors of Viking Lander 1 (VL1 were only fully operational for the first 45 sols of the mission. We have developed an algorithm for reconstructing the wind measurement data after the wind measurement sensor failures. The algorithm for wind reconstruction enables the processing of wind data during the complete VL1 mission. The heater element of the quadrant sensor, which provided auxiliary measurement for wind direction, failed during the 45th sol of the VL1 mission. Additionally, one of the wind sensors of VL1 broke down during sol 378. Regardless of the failures, it was still possible to reconstruct the wind measurement data, because the failed components of the sensors did not prevent the determination of the wind direction and speed, as some of the components of the wind measurement setup remained intact for the complete mission. This article concentrates on presenting the wind reconstruction algorithm and methods for validating the operation of the algorithm. The algorithm enables the reconstruction of wind measurements for the complete VL1 mission. The amount of available sols is extended from 350 to 2245 sols.
Multichannel algorithm for fast 3D reconstruction
International Nuclear Information System (INIS)
Rodet, Thomas; Grangeat, Pierre; Desbat, Laurent
2002-01-01
Some recent medical imaging applications such as functional imaging (PET and SPECT) or interventional imaging (CT fluoroscopy) involve increasing amounts of data. In order to reduce the image reconstruction time, we develop a new fast 3D reconstruction algorithm based on a divide and conquer approach. The proposed multichannel algorithm performs an indirect frequential subband decomposition of the image f to be reconstructed (f=Σf j ) through the filtering of the projections Rf. The subband images f j are reconstructed on a downsampled grid without information suppression. In order to reduce the computation time, we do not backproject the null filtered projections and we downsample the number of projections according to the Shannon conditions associated with the subband image. Our algorithm is based on filtering and backprojection operators. Using the same algorithms for these basic operators, our approach is three and a half times faster than a classical FBP algorithm for a 2D image 512x512 and six times faster for a 3D image 32x512x512. (author)
Backtracking algorithm for lepton reconstruction with HADES
International Nuclear Information System (INIS)
Sellheim, P
2015-01-01
The High Acceptance Di-Electron Spectrometer (HADES) at the GSI Helmholtzzentrum für Schwerionenforschung investigates dilepton and strangeness production in elementary and heavy-ion collisions. In April - May 2012 HADES recorded 7 billion Au+Au events at a beam energy of 1.23 GeV/u with the highest multiplicities measured so far. The track reconstruction and particle identification in the high track density environment are challenging. The most important detector component for lepton identification is the Ring Imaging Cherenkov detector. Its main purpose is the separation of electrons and positrons from large background of charged hadrons produced in heavy-ion collisions. In order to improve lepton identification this backtracking algorithm was developed. In this contribution we will show the results of the algorithm compared to the currently applied method for e +/- identification. Efficiency and purity of a reconstructed e +/- sample will be discussed as well. (paper)
A very fast implementation of 2D iterative reconstruction algorithms
DEFF Research Database (Denmark)
Toft, Peter Aundal; Jensen, Peter James
1996-01-01
that iterative reconstruction algorithms can be implemented and run almost as fast as direct reconstruction algorithms. The method has been implemented in a software package that is available for free, providing reconstruction algorithms using ART, EM, and the Least Squares Conjugate Gradient Method...
Quantitative tomography simulations and reconstruction algorithms
International Nuclear Information System (INIS)
Martz, H.E.; Aufderheide, M.B.; Goodman, D.; Schach von Wittenau, A.; Logan, C.; Hall, J.; Jackson, J.; Slone, D.
2000-01-01
X-ray, neutron and proton transmission radiography and computed tomography (CT) are important diagnostic tools that are at the heart of LLNL's effort to meet the goals of the DOE's Advanced Radiography Campaign. This campaign seeks to improve radiographic simulation and analysis so that radiography can be a useful quantitative diagnostic tool for stockpile stewardship. Current radiographic accuracy does not allow satisfactory separation of experimental effects from the true features of an object's tomographically reconstructed image. This can lead to difficult and sometimes incorrect interpretation of the results. By improving our ability to simulate the whole radiographic and CT system, it will be possible to examine the contribution of system components to various experimental effects, with the goal of removing or reducing them. In this project, we are merging this simulation capability with a maximum-likelihood (constrained-conjugate-gradient-CCG) reconstruction technique yielding a physics-based, forward-model image-reconstruction code. In addition, we seek to improve the accuracy of computed tomography from transmission radiographs by studying what physics is needed in the forward model. During FY 2000, an improved version of the LLNL ray-tracing code called HADES has been coupled with a recently developed LLNL CT algorithm known as CCG. The problem of image reconstruction is expressed as a large matrix equation relating a model for the object being reconstructed to its projections (radiographs). Using a constrained-conjugate-gradient search algorithm, a maximum likelihood solution is sought. This search continues until the difference between the input measured radiographs or projections and the simulated or calculated projections is satisfactorily small
A subzone reconstruction algorithm for efficient staggered compatible remapping
Energy Technology Data Exchange (ETDEWEB)
Starinshak, D.P., E-mail: starinshak1@llnl.gov; Owen, J.M., E-mail: mikeowen@llnl.gov
2015-09-01
Staggered-grid Lagrangian hydrodynamics algorithms frequently make use of subzonal discretization of state variables for the purposes of improved numerical accuracy, generality to unstructured meshes, and exact conservation of mass, momentum, and energy. For Arbitrary Lagrangian–Eulerian (ALE) methods using a geometric overlay, it is difficult to remap subzonal variables in an accurate and efficient manner due to the number of subzone–subzone intersections that must be computed. This becomes prohibitive in the case of 3D, unstructured, polyhedral meshes. A new procedure is outlined in this paper to avoid direct subzonal remapping. The new algorithm reconstructs the spatial profile of a subzonal variable using remapped zonal and nodal representations of the data. The reconstruction procedure is cast as an under-constrained optimization problem. Enforcing conservation at each zone and node on the remapped mesh provides the set of equality constraints; the objective function corresponds to a quadratic variation per subzone between the values to be reconstructed and a set of target reference values. Numerical results for various pure-remapping and hydrodynamics tests are provided. Ideas for extending the algorithm to staggered-grid radiation-hydrodynamics are discussed as well as ideas for generalizing the algorithm to include inequality constraints.
Selected event reconstruction algorithms for the CBM experiment at FAIR
International Nuclear Information System (INIS)
Lebedev, Semen; Höhne, Claudia; Lebedev, Andrey; Ososkov, Gennady
2014-01-01
Development of fast and efficient event reconstruction algorithms is an important and challenging task in the Compressed Baryonic Matter (CBM) experiment at the future FAIR facility. The event reconstruction algorithms have to process terabytes of input data produced in particle collisions. In this contribution, several event reconstruction algorithms are presented. Optimization of the algorithms in the following CBM detectors are discussed: Ring Imaging Cherenkov (RICH) detector, Transition Radiation Detectors (TRD) and Muon Chamber (MUCH). The ring reconstruction algorithm in the RICH is discussed. In TRD and MUCH track reconstruction algorithms are based on track following and Kalman Filter methods. All algorithms were significantly optimized to achieve maximum speed up and minimum memory consumption. Obtained results showed that a significant speed up factor for all algorithms was achieved and the reconstruction efficiency stays at high level.
Greedy algorithms for diffuse optical tomography reconstruction
Dileep, B. P. V.; Das, Tapan; Dutta, Pranab K.
2018-03-01
Diffuse optical tomography (DOT) is a noninvasive imaging modality that reconstructs the optical parameters of a highly scattering medium. However, the inverse problem of DOT is ill-posed and highly nonlinear due to the zig-zag propagation of photons that diffuses through the cross section of tissue. The conventional DOT imaging methods iteratively compute the solution of forward diffusion equation solver which makes the problem computationally expensive. Also, these methods fail when the geometry is complex. Recently, the theory of compressive sensing (CS) has received considerable attention because of its efficient use in biomedical imaging applications. The objective of this paper is to solve a given DOT inverse problem by using compressive sensing framework and various Greedy algorithms such as orthogonal matching pursuit (OMP), compressive sampling matching pursuit (CoSaMP), and stagewise orthogonal matching pursuit (StOMP), regularized orthogonal matching pursuit (ROMP) and simultaneous orthogonal matching pursuit (S-OMP) have been studied to reconstruct the change in the absorption parameter i.e, Δα from the boundary data. Also, the Greedy algorithms have been validated experimentally on a paraffin wax rectangular phantom through a well designed experimental set up. We also have studied the conventional DOT methods like least square method and truncated singular value decomposition (TSVD) for comparison. One of the main features of this work is the usage of less number of source-detector pairs, which can facilitate the use of DOT in routine applications of screening. The performance metrics such as mean square error (MSE), normalized mean square error (NMSE), structural similarity index (SSIM), and peak signal to noise ratio (PSNR) have been used to evaluate the performance of the algorithms mentioned in this paper. Extensive simulation results confirm that CS based DOT reconstruction outperforms the conventional DOT imaging methods in terms of
Advanced reconstruction algorithms for electron tomography: From comparison to combination
Energy Technology Data Exchange (ETDEWEB)
Goris, B. [EMAT, University of Antwerp, Groenenborgerlaan 171, B-2020 Antwerp (Belgium); Roelandts, T. [Vision Lab, University of Antwerp, Universiteitsplein 1, B-2610 Wilrijk (Belgium); Batenburg, K.J. [Vision Lab, University of Antwerp, Universiteitsplein 1, B-2610 Wilrijk (Belgium); Centrum Wiskunde and Informatica, Science Park 123, NL-1098XG Amsterdam (Netherlands); Heidari Mezerji, H. [EMAT, University of Antwerp, Groenenborgerlaan 171, B-2020 Antwerp (Belgium); Bals, S., E-mail: sara.bals@ua.ac.be [EMAT, University of Antwerp, Groenenborgerlaan 171, B-2020 Antwerp (Belgium)
2013-04-15
In this work, the simultaneous iterative reconstruction technique (SIRT), the total variation minimization (TVM) reconstruction technique and the discrete algebraic reconstruction technique (DART) for electron tomography are compared and the advantages and disadvantages are discussed. Furthermore, we describe how the result of a three dimensional (3D) reconstruction based on TVM can provide objective information that is needed as the input for a DART reconstruction. This approach results in a tomographic reconstruction of which the segmentation is carried out in an objective manner. - Highlights: ► A comparative study between different reconstruction algorithms for tomography is performed. ► Reconstruction algorithms that uses prior knowledge about the specimen have a superior result. ► One reconstruction algorithm can provide the prior knowledge for a second algorithm.
Strategies of reconstruction algorithms for computerized tomography
International Nuclear Information System (INIS)
Garderet, P.
1984-10-01
Image reconstruction from projections has progressively spread out over all fields of medical imaging. As the mathematical aspects of the problem become more and more comprehensively explored a great variety of numerical solutions have been developed best suited to such-and-such imaging medical application and taking into account the physical phenomena related to data collection (a priori properties for signal and noise). The purpose of that survey is to present the general mathematical frame and the fundamental assumptions of various strategies; Fourier methods approximate explicit deterministic inversion formula for the Radon transform. Algebraic reconstruction techniques set up an a priori discrete model through a series expansion approach of the solution. The numerical system to be solved is huge when a fine grid of pixels is to be reconstructed; iterative solutions may then be found. Recently some least square procedures have been shown to be tractable which avoid the use of iterative methods. Finally maximum like hood approach incorporates accurately the Poisson nature of photon noise and are well adapted to emission computed tomography. The various strategies will be analysed from both aspects of theoretical assumptions needed for suitable use and of computing facilities, actual performance and cost. In the end we take a glimpse of the extension of the algorithms from two dimensional imaging to fully three dimensional volume analysis in preparation of the future medical imaging technologies
Versatility of the CFR algorithm for limited angle reconstruction
International Nuclear Information System (INIS)
Fujieda, I.; Heiskanen, K.; Perez-Mendez, V.
1990-01-01
The constrained Fourier reconstruction (CFR) algorithm and the iterative reconstruction-reprojection (IRR) algorithm are evaluated based on their accuracy for three types of limited angle reconstruction problems. The cFR algorithm performs better for problems such as Xray CT imaging of a nuclear reactor core with one large data gap due to structural blocking of the source and detector pair. For gated heart imaging by Xray CT, radioisotope distribution imaging by PET or SPECT, using a polygonal array of gamma cameras with insensitive gaps between camera boundaries, the IRR algorithm has a slight advantage over the CFR algorithm but the difference is not significant
Low dose reconstruction algorithm for differential phase contrast imaging.
Wang, Zhentian; Huang, Zhifeng; Zhang, Li; Chen, Zhiqiang; Kang, Kejun; Yin, Hongxia; Wang, Zhenchang; Marco, Stampanoni
2011-01-01
Differential phase contrast imaging computed tomography (DPCI-CT) is a novel x-ray inspection method to reconstruct the distribution of refraction index rather than the attenuation coefficient in weakly absorbing samples. In this paper, we propose an iterative reconstruction algorithm for DPCI-CT which benefits from the new compressed sensing theory. We first realize a differential algebraic reconstruction technique (DART) by discretizing the projection process of the differential phase contrast imaging into a linear partial derivative matrix. In this way the compressed sensing reconstruction problem of DPCI reconstruction can be transformed to a resolved problem in the transmission imaging CT. Our algorithm has the potential to reconstruct the refraction index distribution of the sample from highly undersampled projection data. Thus it can significantly reduce the dose and inspection time. The proposed algorithm has been validated by numerical simulations and actual experiments.
A new simple iterative reconstruction algorithm for SPECT transmission measurement
International Nuclear Information System (INIS)
Hwang, D.S.; Zeng, G.L.
2005-01-01
This paper proposes a new iterative reconstruction algorithm for transmission tomography and compares this algorithm with several other methods. The new algorithm is simple and resembles the emission ML-EM algorithm in form. Due to its simplicity, it is easy to implement and fast to compute a new update at each iteration. The algorithm also always guarantees non-negative solutions. Evaluations are performed using simulation studies and real phantom data. Comparisons with other algorithms such as convex, gradient, and logMLEM show that the proposed algorithm is as good as others and performs better in some cases
A new method for depth profiling reconstruction in confocal microscopy
Esposito, Rosario; Scherillo, Giuseppe; Mensitieri, Giuseppe
2018-05-01
Confocal microscopy is commonly used to reconstruct depth profiles of chemical species in multicomponent systems and to image nuclear and cellular details in human tissues via image intensity measurements of optical sections. However, the performance of this technique is reduced by inherent effects related to wave diffraction phenomena, refractive index mismatch and finite beam spot size. All these effects distort the optical wave and cause an image to be captured of a small volume around the desired illuminated focal point within the specimen rather than an image of the focal point itself. The size of this small volume increases with depth, thus causing a further loss of resolution and distortion of the profile. Recently, we proposed a theoretical model that accounts for the above wave distortion and allows for a correct reconstruction of the depth profiles for homogeneous samples. In this paper, this theoretical approach has been adapted for describing the profiles measured from non-homogeneous distributions of emitters inside the investigated samples. The intensity image is built by summing the intensities collected from each of the emitters planes belonging to the illuminated volume, weighed by the emitters concentration. The true distribution of the emitters concentration is recovered by a new approach that implements this theoretical model in a numerical algorithm based on the Maximum Entropy Method. Comparisons with experimental data and numerical simulations show that this new approach is able to recover the real unknown concentration distribution from experimental profiles with an accuracy better than 3%.
Duality reconstruction algorithm for use in electrical impedance tomography
International Nuclear Information System (INIS)
Abdullah, M.Z.; Dickin, F.J.
1996-01-01
A duality reconstruction algorithm for solving the inverse problem in electrical impedance tomography (EIT) is described. In this method, an algorithm based on the Geselowitz compensation (GC) theorem is used first to reconstruct an approximate version of the image. It is then fed as a first guessed data to the modified Newton-Raphson (MNR) algorithm which iteratively correct the image until a final acceptable solution is reached. The implementation of the GC and MNR based algorithms using the finite element method will be discussed. Reconstructed images produced by the algorithm will also be presented. Consideration is also given to the most computationally intensive aspects of the algorithm, namely the inversion of the large and sparse matrices. The methods taken to approximately compute the inverse ot those matrices will be outlined. (author)
Compressively sampled MR image reconstruction using generalized thresholding iterative algorithm
Elahi, Sana; kaleem, Muhammad; Omer, Hammad
2018-01-01
Compressed sensing (CS) is an emerging area of interest in Magnetic Resonance Imaging (MRI). CS is used for the reconstruction of the images from a very limited number of samples in k-space. This significantly reduces the MRI data acquisition time. One important requirement for signal recovery in CS is the use of an appropriate non-linear reconstruction algorithm. It is a challenging task to choose a reconstruction algorithm that would accurately reconstruct the MR images from the under-sampled k-space data. Various algorithms have been used to solve the system of non-linear equations for better image quality and reconstruction speed in CS. In the recent past, iterative soft thresholding algorithm (ISTA) has been introduced in CS-MRI. This algorithm directly cancels the incoherent artifacts produced because of the undersampling in k -space. This paper introduces an improved iterative algorithm based on p -thresholding technique for CS-MRI image reconstruction. The use of p -thresholding function promotes sparsity in the image which is a key factor for CS based image reconstruction. The p -thresholding based iterative algorithm is a modification of ISTA, and minimizes non-convex functions. It has been shown that the proposed p -thresholding iterative algorithm can be used effectively to recover fully sampled image from the under-sampled data in MRI. The performance of the proposed method is verified using simulated and actual MRI data taken at St. Mary's Hospital, London. The quality of the reconstructed images is measured in terms of peak signal-to-noise ratio (PSNR), artifact power (AP), and structural similarity index measure (SSIM). The proposed approach shows improved performance when compared to other iterative algorithms based on log thresholding, soft thresholding and hard thresholding techniques at different reduction factors.
Kim, Ye-seul; Park, Hye-suk; Lee, Haeng-Hwa; Choi, Young-Wook; Choi, Jae-Gu; Kim, Hak Hee; Kim, Hee-Joung
2016-02-01
Digital breast tomosynthesis (DBT) is a recently developed system for three-dimensional imaging that offers the potential to reduce the false positives of mammography by preventing tissue overlap. Many qualitative evaluations of digital breast tomosynthesis were previously performed by using a phantom with an unrealistic model and with heterogeneous background and noise, which is not representative of real breasts. The purpose of the present work was to compare reconstruction algorithms for DBT by using various breast phantoms; validation was also performed by using patient images. DBT was performed by using a prototype unit that was optimized for very low exposures and rapid readout. Three algorithms were compared: a back-projection (BP) algorithm, a filtered BP (FBP) algorithm, and an iterative expectation maximization (EM) algorithm. To compare the algorithms, three types of breast phantoms (homogeneous background phantom, heterogeneous background phantom, and anthropomorphic breast phantom) were evaluated, and clinical images were also reconstructed by using the different reconstruction algorithms. The in-plane image quality was evaluated based on the line profile and the contrast-to-noise ratio (CNR), and out-of-plane artifacts were evaluated by means of the artifact spread function (ASF). Parenchymal texture features of contrast and homogeneity were computed based on reconstructed images of an anthropomorphic breast phantom. The clinical images were studied to validate the effect of reconstruction algorithms. The results showed that the CNRs of masses reconstructed by using the EM algorithm were slightly higher than those obtained by using the BP algorithm, whereas the FBP algorithm yielded much lower CNR due to its high fluctuations of background noise. The FBP algorithm provides the best conspicuity for larger calcifications by enhancing their contrast and sharpness more than the other algorithms; however, in the case of small-size and low
Parallel Algorithm for Reconstruction of TAC Images
International Nuclear Information System (INIS)
Vidal Gimeno, V.
2012-01-01
The algebraic reconstruction methods are based on solving a system of linear equations. In a previous study, was used and showed as the PETSc library, was and is a scientific computing tool, which facilitates and enables the optimal use of a computer system in the image reconstruction process.
Real time equilibrium reconstruction algorithm in EAST tokamak
International Nuclear Information System (INIS)
Wang Huazhong; Luo Jiarong; Huang Qinchao
2004-01-01
The EAST (HT-7U) superconducting tokamak is a national project of China on fusion research, with a capability of long-pulse (∼1000 s) operation. In order to realize a long-duration steady-state operation of EAST, some significant capability of real-time control is required. It would be very crucial to obtain the current profile parameters and the plasma shapes in real time by a flexible control system. As those discharge parameters cannot be directly measured, so a current profile consistent with the magnetohydrodynamic equilibrium should be evaluated from external magnetic measurements, based on a linearized iterative least square method, which can meet the requirements of the measurements. The arithmetic that the EFIT (equilibrium fitting code) is used for reference will be given in this paper and the computational efforts are reduced by parameterizing the current profile linearly in terms of a number of physical parameters. In order to introduce this reconstruction algorithm clearly, the main hardware design will be listed also. (authors)
DART: a practical reconstruction algorithm for discrete tomography.
Batenburg, Kees Joost; Sijbers, Jan
2011-09-01
In this paper, we present an iterative reconstruction algorithm for discrete tomography, called discrete algebraic reconstruction technique (DART). DART can be applied if the scanned object is known to consist of only a few different compositions, each corresponding to a constant gray value in the reconstruction. Prior knowledge of the gray values for each of the compositions is exploited to steer the current reconstruction towards a reconstruction that contains only these gray values. Based on experiments with both simulated CT data and experimental μCT data, it is shown that DART is capable of computing more accurate reconstructions from a small number of projection images, or from a small angular range, than alternative methods. It is also shown that DART can deal effectively with noisy projection data and that the algorithm is robust with respect to errors in the estimation of the gray values.
Directory of Open Access Journals (Sweden)
P. Xia
2013-10-01
Full Text Available Traditionally, balloon-based radiosonde soundings are used to study the spatial distribution of atmospheric water vapour. However, this approach cannot be frequently employed due to its high cost. In contrast, GPS tomography technique can obtain water vapour in a high temporal resolution. In the tomography technique, an iterative or non-iterative reconstruction algorithm is usually utilised to overcome rank deficiency of observation equations for water vapour inversion. However, the single iterative or non-iterative reconstruction algorithm has their limitations. For instance, the iterative reconstruction algorithm requires accurate initial values of water vapour while the non-iterative reconstruction algorithm needs proper constraint conditions. To overcome these drawbacks, we present a combined iterative and non-iterative reconstruction approach for the three-dimensional (3-D water vapour inversion using GPS observations and COSMIC profiles. In this approach, the non-iterative reconstruction algorithm is first used to estimate water vapour density based on a priori water vapour information derived from COSMIC radio occultation data. The estimates are then employed as initial values in the iterative reconstruction algorithm. The largest advantage of this approach is that precise initial values of water vapour density that are essential in the iterative reconstruction algorithm can be obtained. This combined reconstruction algorithm (CRA is evaluated using 10-day GPS observations in Hong Kong and COSMIC profiles. The test results indicate that the water vapor accuracy from CRA is 16 and 14% higher than that of iterative and non-iterative reconstruction approaches, respectively. In addition, the tomography results obtained from the CRA are further validated using radiosonde data. Results indicate that water vapour densities derived from the CRA agree with radiosonde results very well at altitudes above 2.5 km. The average RMS value of their
Reconstruction of fiber grating refractive-index profiles from complex bragg reflection spectra.
Huang, D W; Yang, C C
1999-07-20
Reconstruction of the refractive-index profiles of fiber gratings from their complex Bragg reflection spectra is experimentally demonstrated. The amplitude and phase of the complex reflection spectrum were measured with a balanced Michelson interferometer. By integrating the coupled-mode equations, we built the relationship between the complex coupling coefficient and the complex reflection spectrum as an iterative algorithm for reconstructing the index profile. This method is expected to be useful for reconstructing the index profiles of fiber gratings with any apodization, chirp, or dc structures. An apodized chirped grating and a uniform grating with a depression of index modulation were used to demonstrate the technique.
Performance of the ATLAS primary vertex reconstruction algorithms
Zhang, Matt
2017-01-01
The reconstruction of primary vertices in the busy, high pile up environment of the LHC is a challenging task. The challenges and novel methods developed by the ATLAS experiment to reconstruct vertices in such environments will be presented. Such advances in vertex seeding include methods taken from medical imagining, which allow for reconstruction of very nearby vertices will be highlighted. The performance of the current vertexing algorithms using early Run-2 data will be presented and compared to results from simulation.
A fast sparse reconstruction algorithm for electrical tomography
International Nuclear Information System (INIS)
Zhao, Jia; Xu, Yanbin; Tan, Chao; Dong, Feng
2014-01-01
Electrical tomography (ET) has been widely investigated due to its advantages of being non-radiative, low-cost and high-speed. However, the image reconstruction of ET is a nonlinear and ill-posed inverse problem and the imaging results are easily affected by measurement noise. A sparse reconstruction algorithm based on L 1 regularization is robust to noise and consequently provides a high quality of reconstructed images. In this paper, a sparse reconstruction by separable approximation algorithm (SpaRSA) is extended to solve the ET inverse problem. The algorithm is competitive with the fastest state-of-the-art algorithms in solving the standard L 2 −L 1 problem. However, it is computationally expensive when the dimension of the matrix is large. To further improve the calculation speed of solving inverse problems, a projection method based on the Krylov subspace is employed and combined with the SpaRSA algorithm. The proposed algorithm is tested with image reconstruction of electrical resistance tomography (ERT). Both simulation and experimental results demonstrate that the proposed method can reduce the computational time and improve the noise robustness for the image reconstruction. (paper)
Search for 'Little Higgs' and reconstruction algorithms developments in Atlas
International Nuclear Information System (INIS)
Rousseau, D.
2007-05-01
This document summarizes developments of framework and reconstruction algorithms for the ATLAS detector at the LHC. A library of reconstruction algorithms has been developed in a more and more complex environment. The reconstruction software originally designed on an optimistic Monte-Carlo simulation, has been confronted with a more detailed 'as-built' simulation. The 'Little Higgs' is an effective theory which can be taken for granted, or as an opportunity to study heavy resonances. In several cases, these resonances can be detected in original channels like tZ, ZH or WH. (author)
A filtered backprojection reconstruction algorithm for Compton camera
Energy Technology Data Exchange (ETDEWEB)
Lojacono, Xavier; Maxim, Voichita; Peyrin, Francoise; Prost, Remy [Lyon Univ., Villeurbanne (France). CNRS, Inserm, INSA-Lyon, CREATIS, UMR5220; Zoglauer, Andreas [California Univ., Berkeley, CA (United States). Space Sciences Lab.
2011-07-01
In this paper we present a filtered backprojection reconstruction algorithm for Compton Camera detectors of particles. Compared to iterative methods, widely used for the reconstruction of images from Compton camera data, analytical methods are fast, easy to implement and avoid convergence issues. The method we propose is exact for an idealized Compton camera composed of two parallel plates of infinite dimension. We show that it copes well with low number of detected photons simulated from a realistic device. Images reconstructed from both synthetic data and realistic ones obtained with Monte Carlo simulations demonstrate the efficiency of the algorithm. (orig.)
A new algorithm for 3D reconstruction from support functions
DEFF Research Database (Denmark)
Gardner, Richard; Kiderlen, Markus
2009-01-01
We introduce a new algorithm for reconstructing an unknown shape from a finite number of noisy measurements of its support function. The algorithm, based on a least squares procedure, is very easy to program in standard software such as Matlab and allows, for the first time, good 3D reconstructio...
Convergence of Algorithms for Reconstructing Convex Bodies and Directional Measures
DEFF Research Database (Denmark)
Gardner, Richard; Kiderlen, Markus; Milanfar, Peyman
2006-01-01
We investigate algorithms for reconstructing a convex body K in Rn from noisy measurements of its support function or its brightness function in k directions u1, . . . , uk. The key idea of these algorithms is to construct a convex polytope Pk whose support function (or brightness function) best...
Iterative concurrent reconstruction algorithms for emission computed tomography
International Nuclear Information System (INIS)
Brown, J.K.; Hasegawa, B.H.; Lang, T.F.
1994-01-01
Direct reconstruction techniques, such as those based on filtered backprojection, are typically used for emission computed tomography (ECT), even though it has been argued that iterative reconstruction methods may produce better clinical images. The major disadvantage of iterative reconstruction algorithms, and a significant reason for their lack of clinical acceptance, is their computational burden. We outline a new class of ''concurrent'' iterative reconstruction techniques for ECT in which the reconstruction process is reorganized such that a significant fraction of the computational processing occurs concurrently with the acquisition of ECT projection data. These new algorithms use the 10-30 min required for acquisition of a typical SPECT scan to iteratively process the available projection data, significantly reducing the requirements for post-acquisition processing. These algorithms are tested on SPECT projection data from a Hoffman brain phantom acquired with a 2 x 10 5 counts in 64 views each having 64 projections. The SPECT images are reconstructed as 64 x 64 tomograms, starting with six angular views. Other angular views are added to the reconstruction process sequentially, in a manner that reflects their availability for a typical acquisition protocol. The results suggest that if T s of concurrent processing are used, the reconstruction processing time required after completion of the data acquisition can be reduced by at least 1/3 T s. (Author)
Reconstruction Algorithms in Undersampled AFM Imaging
DEFF Research Database (Denmark)
Arildsen, Thomas; Oxvig, Christian Schou; Pedersen, Patrick Steffen
2016-01-01
This paper provides a study of spatial undersampling in atomic force microscopy (AFM) imaging followed by different image reconstruction techniques based on sparse approximation as well as interpolation. The main reasons for using undersampling is that it reduces the path length and thereby...... the scanning time as well as the amount of interaction between the AFM probe and the specimen. It can easily be applied on conventional AFM hardware. Due to undersampling, it is then necessary to further process the acquired image in order to reconstruct an approximation of the image. Based on real AFM cell...... images, our simulations reveal that using a simple raster scanning pattern in combination with conventional image interpolation performs very well. Moreover, this combination enables a reduction by a factor 10 of the scanning time while retaining an average reconstruction quality around 36 dB PSNR...
A new iterative algorithm to reconstruct the refractive index.
Liu, Y J; Zhu, P P; Chen, B; Wang, J Y; Yuan, Q X; Huang, W X; Shu, H; Li, E R; Liu, X S; Zhang, K; Ming, H; Wu, Z Y
2007-06-21
The latest developments in x-ray imaging are associated with techniques based on the phase contrast. However, the image reconstruction procedures demand significant improvements of the traditional methods, and/or new algorithms have to be introduced to take advantage of the high contrast and sensitivity of the new experimental techniques. In this letter, an improved iterative reconstruction algorithm based on the maximum likelihood expectation maximization technique is presented and discussed in order to reconstruct the distribution of the refractive index from data collected by an analyzer-based imaging setup. The technique considered probes the partial derivative of the refractive index with respect to an axis lying in the meridional plane and perpendicular to the propagation direction. Computer simulations confirm the reliability of the proposed algorithm. In addition, the comparison between an analytical reconstruction algorithm and the iterative method has been also discussed together with the convergent characteristic of this latter algorithm. Finally, we will show how the proposed algorithm may be applied to reconstruct the distribution of the refractive index of an epoxy cylinder containing small air bubbles of about 300 micro of diameter.
Fast parallel algorithm for CT image reconstruction.
Flores, Liubov A; Vidal, Vicent; Mayo, Patricia; Rodenas, Francisco; Verdú, Gumersindo
2012-01-01
In X-ray computed tomography (CT) the X rays are used to obtain the projection data needed to generate an image of the inside of an object. The image can be generated with different techniques. Iterative methods are more suitable for the reconstruction of images with high contrast and precision in noisy conditions and from a small number of projections. Their use may be important in portable scanners for their functionality in emergency situations. However, in practice, these methods are not widely used due to the high computational cost of their implementation. In this work we analyze iterative parallel image reconstruction with the Portable Extensive Toolkit for Scientific computation (PETSc).
New vertex reconstruction algorithms for CMS
Frühwirth, R; Prokofiev, Kirill; Speer, T.; Vanlaer, P.; Chabanat, E.; Estre, N.
2003-01-01
The reconstruction of interaction vertices can be decomposed into a pattern recognition problem (``vertex finding'') and a statistical problem (``vertex fitting''). We briefly review classical methods. We introduce novel approaches and motivate them in the framework of high-luminosity experiments like at the LHC. We then show comparisons with the classical methods in relevant physics channels
Profile reconstruction from neutron reflectivity data and a priori knowledge
International Nuclear Information System (INIS)
Leeb, H.
2008-01-01
The problem of incomplete and noisy information in profile reconstruction from neutron reflectometry data is considered. In particular methods of Bayesian statistics in combination with modelling or inverse scattering techniques are considered in order to properly include the required a priori knowledge to obtain quantitatively reliable estimates of the reconstructed profiles. Applying Bayes theorem the results of different experiments on the same sample can be consistently included in the profile reconstruction
Efficient reconstruction of dispersive dielectric profiles using time domain reflectometry (TDR
Directory of Open Access Journals (Sweden)
P. Leidenberger
2006-01-01
Full Text Available We present a numerical model for time domain reflectometry (TDR signal propagation in dispersive dielectric materials. The numerical probe model is terminated with a parallel circuit, consisting of an ohmic resistor and an ideal capacitance. We derive analytical approximations for the capacitance, the inductance and the conductance of three-wire probes. We couple the time domain model with global optimization in order to reconstruct water content profiles from TDR traces. For efficiently solving the inverse problem we use genetic algorithms combined with a hierarchical parameterization. We investigate the performance of the method by reconstructing synthetically generated profiles. The algorithm is then applied to retrieve dielectric profiles from TDR traces measured in the field. We succeed in reconstructing dielectric and ohmic profiles where conventional methods, based on travel time extraction, fail.
Image-reconstruction algorithms for positron-emission tomography systems
International Nuclear Information System (INIS)
Cheng, S.N.C.
1982-01-01
The positional uncertainty in the time-of-flight measurement of a positron-emission tomography system is modelled as a Gaussian distributed random variable and the image is assumed to be piecewise constant on a rectilinear lattice. A reconstruction algorithm using maximum-likelihood estimation is derived for the situation in which time-of-flight data are sorted as the most-likely-position array. The algorithm is formulated as a linear system described by a nonseparable, block-banded, Toeplitz matrix, and a sine-transform technique is used to implement this algorithm efficiently. The reconstruction algorithms for both the most-likely-position array and the confidence-weighted array are described by similar equations, hence similar linear systems can be used to described the reconstruction algorithm for a discrete, confidence-weighted array, when the matrix and the entries in the data array are properly identified. It is found that the mean square-error depends on the ratio of the full width at half the maximum of time-of-flight measurement over the size of a pixel. When other parameters are fixed, the larger the pixel size, the smaller is the mean square-error. In the study of resolution, parameters that affect the impulse response of time-of-flight reconstruction algorithms are identified. It is found that the larger the pixel size, the larger is the standard deviation of the impulse response. This shows that small mean square-error and fine resolution are two contradictory requirements
Fast algorithm of track reconstruction for the Delphy TPC
International Nuclear Information System (INIS)
Maillard, J.
1984-01-01
We describe a simple geometrical method (polar inversion) to reconstruct tracks. When the magnetic field is constant in magnitude and direction. This method uses geometrical properties of the trajectories. In the case of the DELPHI apparatus, the track reconstruction is done using TPC informations. After explaining the algorithm, we give results on ''GEANT'' simulated events using the ''Lund'' generator. Today we get a computer time of the order of 1.2 milliseconds on a CDC 7600 and an efficiency of 98% [fr
Noise propagation in iterative reconstruction algorithms with line searches
International Nuclear Information System (INIS)
Qi, Jinyi
2003-01-01
In this paper we analyze the propagation of noise in iterative image reconstruction algorithms. We derive theoretical expressions for the general form of preconditioned gradient algorithms with line searches. The results are applicable to a wide range of iterative reconstruction problems, such as emission tomography, transmission tomography, and image restoration. A unique contribution of this paper comparing to our previous work [1] is that the line search is explicitly modeled and we do not use the approximation that the gradient of the objective function is zero. As a result, the error in the estimate of noise at early iterations is significantly reduced
ADART: an adaptive algebraic reconstruction algorithm for discrete tomography.
Maestre-Deusto, F Javier; Scavello, Giovanni; Pizarro, Joaquín; Galindo, Pedro L
2011-08-01
In this paper we suggest an algorithm based on the Discrete Algebraic Reconstruction Technique (DART) which is capable of computing high quality reconstructions from substantially fewer projections than required for conventional continuous tomography. Adaptive DART (ADART) goes a step further than DART on the reduction of the number of unknowns of the associated linear system achieving a significant reduction in the pixel error rate of reconstructed objects. The proposed methodology automatically adapts the border definition criterion at each iteration, resulting in a reduction of the number of pixels belonging to the border, and consequently of the number of unknowns in the general algebraic reconstruction linear system to be solved, being this reduction specially important at the final stage of the iterative process. Experimental results show that reconstruction errors are considerably reduced using ADART when compared to original DART, both in clean and noisy environments.
Optimization of reconstruction algorithms using Monte Carlo simulation
International Nuclear Information System (INIS)
Hanson, K.M.
1989-01-01
A method for optimizing reconstruction algorithms is presented that is based on how well a specified task can be performed using the reconstructed images. Task performance is numerically assessed by a Monte Carlo simulation of the complete imaging process including the generation of scenes appropriate to the desired application, subsequent data taking, reconstruction, and performance of the stated task based on the final image. The use of this method is demonstrated through the optimization of the Algebraic Reconstruction Technique (ART), which reconstructs images from their projections by an iterative procedure. The optimization is accomplished by varying the relaxation factor employed in the updating procedure. In some of the imaging situations studied, it is found that the optimization of constrained ART, in which a non-negativity constraint is invoked, can vastly increase the detectability of objects. There is little improvement attained for unconstrained ART. The general method presented may be applied to the problem of designing neutron-diffraction spectrometers. (author)
Convergence of iterative image reconstruction algorithms for Digital Breast Tomosynthesis
DEFF Research Database (Denmark)
Sidky, Emil; Jørgensen, Jakob Heide; Pan, Xiaochuan
2012-01-01
Most iterative image reconstruction algorithms are based on some form of optimization, such as minimization of a data-fidelity term plus an image regularizing penalty term. While achieving the solution of these optimization problems may not directly be clinically relevant, accurate optimization s...
A Practical Algorithm for Reconstructing Level-1 Phylogenetic Networks
K.T. Huber; L.J.J. van Iersel (Leo); S.M. Kelk (Steven); R. Suchecki
2010-01-01
htmlabstractRecently much attention has been devoted to the construction of phylogenetic networks which generalize phylogenetic trees in order to accommodate complex evolutionary processes. Here we present an efficient, practical algorithm for reconstructing level-1 phylogenetic networks - a type of
A practical algorithm for reconstructing level-1 phylogenetic networks
Huber, K.T.; Iersel, van L.J.J.; Kelk, S.M.; Suchecki, R.
2011-01-01
Recently, much attention has been devoted to the construction of phylogenetic networks which generalize phylogenetic trees in order to accommodate complex evolutionary processes. Here, we present an efficient, practical algorithm for reconstructing level-1 phylogenetic networks-a type of network
Inverse Monte Carlo: a unified reconstruction algorithm for SPECT
International Nuclear Information System (INIS)
Floyd, C.E.; Coleman, R.E.; Jaszczak, R.J.
1985-01-01
Inverse Monte Carlo (IMOC) is presented as a unified reconstruction algorithm for Emission Computed Tomography (ECT) providing simultaneous compensation for scatter, attenuation, and the variation of collimator resolution with depth. The technique of inverse Monte Carlo is used to find an inverse solution to the photon transport equation (an integral equation for photon flux from a specified source) for a parameterized source and specific boundary conditions. The system of linear equations so formed is solved to yield the source activity distribution for a set of acquired projections. For the studies presented here, the equations are solved using the EM (Maximum Likelihood) algorithm although other solution algorithms, such as Least Squares, could be employed. While the present results specifically consider the reconstruction of camera-based Single Photon Emission Computed Tomographic (SPECT) images, the technique is equally valid for Positron Emission Tomography (PET) if a Monte Carlo model of such a system is used. As a preliminary evaluation, experimentally acquired SPECT phantom studies for imaging Tc-99m (140 keV) are presented which demonstrate the quantitative compensation for scatter and attenuation for a two dimensional (single slice) reconstruction. The algorithm may be expanded in a straight forward manner to full three dimensional reconstruction including compensation for out of plane scatter
Iterative reconstruction of transcriptional regulatory networks: an algorithmic approach.
Directory of Open Access Journals (Sweden)
Christian L Barrett
2006-05-01
Full Text Available The number of complete, publicly available genome sequences is now greater than 200, and this number is expected to rapidly grow in the near future as metagenomic and environmental sequencing efforts escalate and the cost of sequencing drops. In order to make use of this data for understanding particular organisms and for discerning general principles about how organisms function, it will be necessary to reconstruct their various biochemical reaction networks. Principal among these will be transcriptional regulatory networks. Given the physical and logical complexity of these networks, the various sources of (often noisy data that can be utilized for their elucidation, the monetary costs involved, and the huge number of potential experiments approximately 10(12 that can be performed, experiment design algorithms will be necessary for synthesizing the various computational and experimental data to maximize the efficiency of regulatory network reconstruction. This paper presents an algorithm for experimental design to systematically and efficiently reconstruct transcriptional regulatory networks. It is meant to be applied iteratively in conjunction with an experimental laboratory component. The algorithm is presented here in the context of reconstructing transcriptional regulation for metabolism in Escherichia coli, and, through a retrospective analysis with previously performed experiments, we show that the produced experiment designs conform to how a human would design experiments. The algorithm is able to utilize probability estimates based on a wide range of computational and experimental sources to suggest experiments with the highest potential of discovering the greatest amount of new regulatory knowledge.
Preconditioned alternating projection algorithms for maximum a posteriori ECT reconstruction
International Nuclear Information System (INIS)
Krol, Andrzej; Li, Si; Shen, Lixin; Xu, Yuesheng
2012-01-01
We propose a preconditioned alternating projection algorithm (PAPA) for solving the maximum a posteriori (MAP) emission computed tomography (ECT) reconstruction problem. Specifically, we formulate the reconstruction problem as a constrained convex optimization problem with the total variation (TV) regularization. We then characterize the solution of the constrained convex optimization problem and show that it satisfies a system of fixed-point equations defined in terms of two proximity operators raised from the convex functions that define the TV-norm and the constraint involved in the problem. The characterization (of the solution) via the proximity operators that define two projection operators naturally leads to an alternating projection algorithm for finding the solution. For efficient numerical computation, we introduce to the alternating projection algorithm a preconditioning matrix (the EM-preconditioner) for the dense system matrix involved in the optimization problem. We prove theoretically convergence of the PAPA. In numerical experiments, performance of our algorithms, with an appropriately selected preconditioning matrix, is compared with performance of the conventional MAP expectation-maximization (MAP-EM) algorithm with TV regularizer (EM-TV) and that of the recently developed nested EM-TV algorithm for ECT reconstruction. Based on the numerical experiments performed in this work, we observe that the alternating projection algorithm with the EM-preconditioner outperforms significantly the EM-TV in all aspects including the convergence speed, the noise in the reconstructed images and the image quality. It also outperforms the nested EM-TV in the convergence speed while providing comparable image quality. (paper)
International Nuclear Information System (INIS)
Chaari, L.; Pesquet, J.Ch.; Chaari, L.; Ciuciu, Ph.; Benazza-Benyahia, A.
2011-01-01
To reduce scanning time and/or improve spatial/temporal resolution in some Magnetic Resonance Imaging (MRI) applications, parallel MRI acquisition techniques with multiple coils acquisition have emerged since the early 1990's as powerful imaging methods that allow a faster acquisition process. In these techniques, the full FOV image has to be reconstructed from the resulting acquired under sampled k-space data. To this end, several reconstruction techniques have been proposed such as the widely-used Sensitivity Encoding (SENSE) method. However, the reconstructed image generally presents artifacts when perturbations occur in both the measured data and the estimated coil sensitivity profiles. In this paper, we aim at achieving accurate image reconstruction under degraded experimental conditions (low magnetic field and high reduction factor), in which neither the SENSE method nor the Tikhonov regularization in the image domain give convincing results. To this end, we present a novel method for SENSE-based reconstruction which proceeds with regularization in the complex wavelet domain by promoting sparsity. The proposed approach relies on a fast algorithm that enables the minimization of regularized non-differentiable criteria including more general penalties than a classical l 1 term. To further enhance the reconstructed image quality, local convex constraints are added to the regularization process. In vivo human brain experiments carried out on Gradient-Echo (GRE) anatomical and Echo Planar Imaging (EPI) functional MRI data at 1.5 T indicate that our algorithm provides reconstructed images with reduced artifacts for high reduction factors. (authors)
Honey bee-inspired algorithms for SNP haplotype reconstruction problem
PourkamaliAnaraki, Maryam; Sadeghi, Mehdi
2016-03-01
Reconstructing haplotypes from SNP fragments is an important problem in computational biology. There have been a lot of interests in this field because haplotypes have been shown to contain promising data for disease association research. It is proved that haplotype reconstruction in Minimum Error Correction model is an NP-hard problem. Therefore, several methods such as clustering techniques, evolutionary algorithms, neural networks and swarm intelligence approaches have been proposed in order to solve this problem in appropriate time. In this paper, we have focused on various evolutionary clustering techniques and try to find an efficient technique for solving haplotype reconstruction problem. It can be referred from our experiments that the clustering methods relying on the behaviour of honey bee colony in nature, specifically bees algorithm and artificial bee colony methods, are expected to result in more efficient solutions. An application program of the methods is available at the following link. http://www.bioinf.cs.ipm.ir/software/haprs/
A reconstruction algorithms for helical cone-beam SPECT
International Nuclear Information System (INIS)
Weng, Y.; Zeng, G.L.; Gullberg, G.T.
1993-01-01
Cone-beam SPECT provides improved sensitivity for imaging small organs like the brain and heart. However, current cone-beam tomography with the focal point traversing a planar orbit does not acquire sufficient data to give an accurate reconstruction. In this paper, the authors employ a data-acquisition method which obtains complete data for cone-beam SPECT by simultaneously rotating the gamma camera and translating the patient bed, so that cone-beam projections can be obtained with the focal point traversing a helix surrounding the patient. An implementation of Grangeat's algorithm for helical cone-beam projections is developed. The algorithm requires a rebinning step to convert cone-beam data to parallel-beam data which are then reconstructed using the 3D Radon inversion. A fast new rebinning scheme is developed which uses all of the detected data to reconstruct the image and properly normalizes any multiply scanned data. This algorithm is shown to produce less artifacts than the commonly used Feldkamp algorithm when applied to either a circular planar orbit or a helical orbit acquisition. The algorithm can easily be extended to any arbitrary orbit
Event Reconstruction Algorithms for the ATLAS Trigger
Energy Technology Data Exchange (ETDEWEB)
Fonseca-Martin, T.; /CERN; Abolins, M.; /Michigan State U.; Adragna, P.; /Queen Mary, U. of London; Aleksandrov, E.; /Dubna, JINR; Aleksandrov, I.; /Dubna, JINR; Amorim, A.; /Lisbon, LIFEP; Anderson, K.; /Chicago U., EFI; Anduaga, X.; /La Plata U.; Aracena, I.; /SLAC; Asquith, L.; /University Coll. London; Avolio, G.; /CERN; Backlund, S.; /CERN; Badescu, E.; /Bucharest, IFIN-HH; Baines, J.; /Rutherford; Barria, P.; /Rome U. /INFN, Rome; Bartoldus, R.; /SLAC; Batreanu, S.; /Bucharest, IFIN-HH /CERN; Beck, H.P.; /Bern U.; Bee, C.; /Marseille, CPPM; Bell, P.; /Manchester U.; Bell, W.H.; /Glasgow U. /Pavia U. /INFN, Pavia /Regina U. /CERN /Annecy, LAPP /Paris, IN2P3 /Royal Holloway, U. of London /Napoli Seconda U. /INFN, Naples /Argonne /CERN /UC, Irvine /Barcelona, IFAE /Barcelona, Autonoma U. /CERN /Montreal U. /CERN /Glasgow U. /Michigan State U. /Bucharest, IFIN-HH /Napoli Seconda U. /INFN, Naples /New York U. /Barcelona, IFAE /Barcelona, Autonoma U. /Salento U. /INFN, Lecce /Pisa U. /INFN, Pisa /Bucharest, IFIN-HH /UC, Irvine /CERN /Glasgow U. /INFN, Genoa /Genoa U. /Lisbon, LIFEP /Napoli Seconda U. /INFN, Naples /UC, Irvine /Valencia U. /Rio de Janeiro Federal U. /University Coll. London /New York U.; /more authors..
2011-11-09
The ATLAS experiment under construction at CERN is due to begin operation at the end of 2007. The detector will record the results of proton-proton collisions at a center-of-mass energy of 14 TeV. The trigger is a three-tier system designed to identify in real-time potentially interesting events that are then saved for detailed offline analysis. The trigger system will select approximately 200 Hz of potentially interesting events out of the 40 MHz bunch-crossing rate (with 10{sup 9} interactions per second at the nominal luminosity). Algorithms used in the trigger system to identify different event features of interest will be described, as well as their expected performance in terms of selection efficiency, background rejection and computation time per event. The talk will concentrate on recent improvements and on performance studies, using a very detailed simulation of the ATLAS detector and electronics chain that emulates the raw data as it will appear at the input to the trigger system.
Event reconstruction algorithms for the ATLAS trigger
Energy Technology Data Exchange (ETDEWEB)
F-Martin, T; Avolio, G; Backlund, S [European Laboratory for Particle Physics (CERN), Geneva (Switzerland); Abolins, M [Michigan State University, Department of Physics and Astronomy, East Lansing, Michigan (United States); Adragna, P [Department of Physics, Queen Mary and Westfield College, University of London, London (United Kingdom); Aleksandrov, E; Aleksandrov, I [Joint Institute for Nuclear Research, Dubna (Russian Federation); Amorim, A [Laboratorio de Instrumentacao e Fisica Experimental, Lisboa (Portugal); Anderson, K [University of Chicago, Enrico Fermi Institute, Chicago, Illinois (United States); Anduaga, X [National University of La Plata, La Plata (United States); Aracena, I; Bartoldus, R [Stanford Linear Accelerator Center (SLAC), Stanford (United States); Asquith, L [Department of Physics and Astronomy, University College London, London (United Kingdom); Badescu, E [National Institute for Physics and Nuclear Engineering, Institute of Atomic Physics, Bucharest (Romania); Baines, J [Rutherford Appleton Laboratory, Chilton, Didcot (United Kingdom); Beck, H P [Laboratory for High Energy Physics, University of Bern, Bern (Switzerland); Bee, C [Centre de Physique des Particules de Marseille, IN2P3-CNRS, Marseille (France); Bell, P [Department of Physics and Astronomy, University of Manchester, Manchester (United Kingdom); Barria, P; Batreanu, S [and others
2008-07-01
The ATLAS experiment under construction at CERN is due to begin operation at the end of 2007. The detector will record the results of proton-proton collisions at a center-of-mass energy of 14 TeV. The trigger is a three-tier system designed to identify in real-time potentially interesting events that are then saved for detailed offline analysis. The trigger system will select approximately 200 Hz of potentially interesting events out of the 40 MHz bunch-crossing rate (with 10{sup 9} interactions per second at the nominal luminosity). Algorithms used in the trigger system to identify different event features of interest will be described, as well as their expected performance in terms of selection efficiency, background rejection and computation time per event. The talk will concentrate on recent improvements and on performance studies, using a very detailed simulation of the ATLAS detector and electronics chain that emulates the raw data as it will appear at the input to the trigger system.
Event reconstruction algorithms for the ATLAS trigger
International Nuclear Information System (INIS)
F-Martin, T; Avolio, G; Backlund, S; Abolins, M; Adragna, P; Aleksandrov, E; Aleksandrov, I; Amorim, A; Anderson, K; Anduaga, X; Aracena, I; Bartoldus, R; Asquith, L; Badescu, E; Baines, J; Beck, H P; Bee, C; Bell, P; Barria, P; Batreanu, S
2008-01-01
The ATLAS experiment under construction at CERN is due to begin operation at the end of 2007. The detector will record the results of proton-proton collisions at a center-of-mass energy of 14 TeV. The trigger is a three-tier system designed to identify in real-time potentially interesting events that are then saved for detailed offline analysis. The trigger system will select approximately 200 Hz of potentially interesting events out of the 40 MHz bunch-crossing rate (with 10 9 interactions per second at the nominal luminosity). Algorithms used in the trigger system to identify different event features of interest will be described, as well as their expected performance in terms of selection efficiency, background rejection and computation time per event. The talk will concentrate on recent improvements and on performance studies, using a very detailed simulation of the ATLAS detector and electronics chain that emulates the raw data as it will appear at the input to the trigger system
Guo, J.; Bücherl, T.; Zou, Y.; Guo, Z.
2011-09-01
Investigations on the fast neutron beam geometry for the NECTAR facility are presented. The results of MCNP simulations and experimental measurements of the beam distributions at NECTAR are compared. Boltzmann functions are used to describe the beam profile in the detection plane assuming the area source to be set up of large number of single neutron point sources. An iterative algebraic reconstruction algorithm is developed, realized and verified by both simulated and measured projection data. The feasibility for improved reconstruction in fast neutron computerized tomography at the NECTAR facility is demonstrated.
International Nuclear Information System (INIS)
Guo, J.; Buecherl, T.; Zou, Y.; Guo, Z.
2011-01-01
Investigations on the fast neutron beam geometry for the NECTAR facility are presented. The results of MCNP simulations and experimental measurements of the beam distributions at NECTAR are compared. Boltzmann functions are used to describe the beam profile in the detection plane assuming the area source to be set up of large number of single neutron point sources. An iterative algebraic reconstruction algorithm is developed, realized and verified by both simulated and measured projection data. The feasibility for improved reconstruction in fast neutron computerized tomography at the NECTAR facility is demonstrated.
Energy Technology Data Exchange (ETDEWEB)
Guo, J. [State Key Laboratory of Nuclear Physics and Technology and School of Physics, Peking University, 5 Yiheyuan Lu, Beijing 100871 (China); Lehrstuhl fuer Radiochemie, Technische Universitaet Muenchen, Garching 80748 (Germany); Buecherl, T. [Lehrstuhl fuer Radiochemie, Technische Universitaet Muenchen, Garching 80748 (Germany); Zou, Y., E-mail: zouyubin@pku.edu.cn [State Key Laboratory of Nuclear Physics and Technology and School of Physics, Peking University, 5 Yiheyuan Lu, Beijing 100871 (China); Guo, Z. [State Key Laboratory of Nuclear Physics and Technology and School of Physics, Peking University, 5 Yiheyuan Lu, Beijing 100871 (China)
2011-09-21
Investigations on the fast neutron beam geometry for the NECTAR facility are presented. The results of MCNP simulations and experimental measurements of the beam distributions at NECTAR are compared. Boltzmann functions are used to describe the beam profile in the detection plane assuming the area source to be set up of large number of single neutron point sources. An iterative algebraic reconstruction algorithm is developed, realized and verified by both simulated and measured projection data. The feasibility for improved reconstruction in fast neutron computerized tomography at the NECTAR facility is demonstrated.
Evaluation of reconstruction algorithms in SPECT neuroimaging: Pt. 1
International Nuclear Information System (INIS)
Heejoung Kim; Zeeberg, B.R.; Reba, R.C.
1993-01-01
In the presence of statistical noise, an iterative reconstruction algorithm (IRA) for the quantitative reconstruction of single-photon-emission computed tomographic (SPECT) brain images overcomes major limitations of applying the standard filtered back projection (FBP) reconstruction algorithm to projection data which have been degraded by convolution of the true radioactivity distribution with a finite-resolution distance-dependent detector response: (a) the non-uniformity within the grey (or white) matter voxels which results even though the true model is uniform within these voxels; (b) a significantly lower ratio of grey/white matter voxel values than in the true model; and (c) an inability to detect an altered radioactivity value within the grey (or white) matter voxels. It is normally expected that an algorithm which improves spatial resolution and quantitative accuracy might also increase the magnitude of the statistical noise in the reconstructed image. However, the noise properties in the IRA images are very similar to those in the FBP images. (Author)
Blind spectrum reconstruction algorithm with L0-sparse representation
International Nuclear Information System (INIS)
Liu, Hai; Zhang, Zhaoli; Liu, Sanyan; Shu, Jiangbo; Liu, Tingting; Zhang, Tianxu
2015-01-01
Raman spectrum often suffers from band overlap and Poisson noise. This paper presents a new blind Poissonian Raman spectrum reconstruction method, which incorporates the L 0 -sparse prior together with the total variation constraint into the maximum a posteriori framework. Furthermore, the greedy analysis pursuit algorithm is adopted to solve the L 0 -based minimization problem. Simulated and real spectrum experimental results show that the proposed method can effectively preserve spectral structure and suppress noise. The reconstructed Raman spectra are easily used for interpreting unknown chemical mixtures. (paper)
Development of Image Reconstruction Algorithms in electrical Capacitance Tomography
International Nuclear Information System (INIS)
Fernandez Marron, J. L.; Alberdi Primicia, J.; Barcala Riveira, J. M.
2007-01-01
The Electrical Capacitance Tomography (ECT) has not obtained a good development in order to be used at industrial level. That is due first to difficulties in the measurement of very little capacitances (in the range of femto farads) and second to the problem of reconstruction on- line of the images. This problem is due also to the small numbers of electrodes (maximum 16), that made the usual algorithms of reconstruction has many errors. In this work it is described a new purely geometrical method that could be used for this purpose. (Author) 4 refs
Parametric boundary reconstruction algorithm for industrial CT metrology application.
Yin, Zhye; Khare, Kedar; De Man, Bruno
2009-01-01
High-energy X-ray computed tomography (CT) systems have been recently used to produce high-resolution images in various nondestructive testing and evaluation (NDT/NDE) applications. The accuracy of the dimensional information extracted from CT images is rapidly approaching the accuracy achieved with a coordinate measuring machine (CMM), the conventional approach to acquire the metrology information directly. On the other hand, CT systems generate the sinogram which is transformed mathematically to the pixel-based images. The dimensional information of the scanned object is extracted later by performing edge detection on reconstructed CT images. The dimensional accuracy of this approach is limited by the grid size of the pixel-based representation of CT images since the edge detection is performed on the pixel grid. Moreover, reconstructed CT images usually display various artifacts due to the underlying physical process and resulting object boundaries from the edge detection fail to represent the true boundaries of the scanned object. In this paper, a novel algorithm to reconstruct the boundaries of an object with uniform material composition and uniform density is presented. There are three major benefits in the proposed approach. First, since the boundary parameters are reconstructed instead of image pixels, the complexity of the reconstruction algorithm is significantly reduced. The iterative approach, which can be computationally intensive, will be practical with the parametric boundary reconstruction. Second, the object of interest in metrology can be represented more directly and accurately by the boundary parameters instead of the image pixels. By eliminating the extra edge detection step, the overall dimensional accuracy and process time can be improved. Third, since the parametric reconstruction approach shares the boundary representation with other conventional metrology modalities such as CMM, boundary information from other modalities can be directly
Cone-beam and fan-beam image reconstruction algorithms based on spherical and circular harmonics
International Nuclear Information System (INIS)
Zeng, Gengsheng L; Gullberg, Grant T
2004-01-01
A cone-beam image reconstruction algorithm using spherical harmonic expansions is proposed. The reconstruction algorithm is in the form of a summation of inner products of two discrete arrays of spherical harmonic expansion coefficients at each cone-beam point of acquisition. This form is different from the common filtered backprojection algorithm and the direct Fourier reconstruction algorithm. There is no re-sampling of the data, and spherical harmonic expansions are used instead of Fourier expansions. As a special case, a new fan-beam image reconstruction algorithm is also derived in terms of a circular harmonic expansion. Computer simulation results for both cone-beam and fan-beam algorithms are presented for circular planar orbit acquisitions. The algorithms give accurate reconstructions; however, the implementation of the cone-beam reconstruction algorithm is computationally intensive. A relatively efficient algorithm is proposed for reconstructing the central slice of the image when a circular scanning orbit is used
Image Reconstruction Algorithm For Electrical Capacitance Tomography (ECT)
International Nuclear Information System (INIS)
Arko
2001-01-01
). Most image reconstruction algorithms for electrical capacitance tomography (ECT) use sensitivity maps as weighting factors. The computation is fast, involving a simple multiply-and- accumulate (MAC) operation, but the resulting image suffers from blurring due to the soft-field effect of the sensor. This paper presents a low cost iterative method employing proportional thresholding, which improves image quality dramatically. The strategy for implementation, computational cost, and achievable speed is examined when using a personal computer (PC) and Digital Signal Processor (DSP). For PC implementation, Watcom C++ 10.6 and Visual C++ 5.0 compilers were used. The experimental results are compared to the images reconstructed by commercially available software. The new algorithm improves the image quality significantly at a cost of a few iterations. This technique can be readily exploited for online applications
Energy reconstruction and calibration algorithms for the ATLAS electromagnetic calorimeter
Delmastro, M
2003-01-01
The work of this thesis is devoted to the study, development and optimization of the algorithms of energy reconstruction and calibration for the electromagnetic calorimeter (EMC) of the ATLAS experiment, presently under installation and commissioning at the CERN Large Hadron Collider in Geneva (Switzerland). A deep study of the electrical characteristics of the detector and of the signals formation and propagation is conduced: an electrical model of the detector is developed and analyzed through simulations; a hardware model (mock-up) of a group of the EMC readout cells has been built, allowing the direct collection and properties study of the signals emerging from the EMC cells. We analyze the existing multiple-sampled signal reconstruction strategy, showing the need of an improvement in order to reach the advertised performances of the detector. The optimal filtering reconstruction technique is studied and implemented, taking into account the differences between the ionization and calibration waveforms as e...
Tomographic reconstruction of transverse phase space from turn-by-turn profile data
Hancock, S; Lindroos, M
1999-01-01
Tomographic methods have the potential for useful application in beam diagnostics. The tomographic reconstruction of transverse phase space density from turn-by-turn profile data has been studied with particular attention to the effects of dispersion and chromaticity. It is shown that the modified Algebraic Reconstruction Technique (ART) that deals successfully with the problem of non-linear motion in the longitudinal plane cannot, in general, be extended to cover the transverse case. Instead, an approach is proposed in which the effect of dispersion is deconvoluted from the measured profiles before the phase space picture is reconstructed using either the modified ART algorithm or the inverse Radon Transform. This requires an accurate knowledge of the momentum distribution of the beam and the modified ART reconstruction of longitudinal phase space density yields just such information. The method has been tested extensively with simulated data.
A novel standalone track reconstruction algorithm for the LHCb upgrade
Quagliani, Renato
2018-01-01
During the LHC Run III, starting in 2020, the instantaneous luminosity of LHCb will be increased up to 2×1033 cm−2 s−1, five times larger than in Run II. The LHCb detector will then have to be upgraded in 2019. In fact, a full software event reconstruction will be performed at the full bunch crossing rate by the trigger, in order to profit of the higher instantaneous luminosity provided by the accelerator. In addition, all the tracking devices will be replaced and, in particular, a scintillating fiber tracker (SciFi) will be installed after the magnet, allowing to cope with the higher occupancy. The new running conditions, and the tighter timing constraints in the software trigger, represent a big challenge for the track reconstruction. This talk presents the design and performance of a novel algorithm that has been developed to reconstruct track segments using solely hits from the SciFi. This algorithm is crucial for the reconstruction of tracks originating from long-lived particles such as KS and Λ. ...
Optimization of reconstruction algorithms using Monte Carlo simulation
International Nuclear Information System (INIS)
Hanson, K.M.
1989-01-01
A method for optimizing reconstruction algorithms is presented that is based on how well a specified task can be performed using the reconstructed images. Task performance is numerically assessed by a Monte Carlo simulation of the complete imaging process including the generation of scenes appropriate to the desired application, subsequent data taking, reconstruction, and performance of the stated task based on the final image. The use of this method is demonstrated through the optimization of the Algebraic Reconstruction Technique (ART), which reconstructs images from their projections by a iterative procedure. The optimization is accomplished by varying the relaxation factor employed in the updating procedure. In some of the imaging situations studied, it is found that the optimization of constrained ART, in which a nonnegativity constraint is invoked, can vastly increase the detectability of objects. There is little improvement attained for unconstrained ART. The general method presented may be applied to the problem of designing neutron-diffraction spectrometers. 11 refs., 6 figs., 2 tabs
First results of genetic algorithm application in ML image reconstruction in emission tomography
International Nuclear Information System (INIS)
Smolik, W.
1999-01-01
This paper concerns application of genetic algorithm in maximum likelihood image reconstruction in emission tomography. The example of genetic algorithm for image reconstruction is presented. The genetic algorithm was based on the typical genetic scheme modified due to the nature of solved problem. The convergence of algorithm was examined. The different adaption functions, selection and crossover methods were verified. The algorithm was tested on simulated SPECT data. The obtained results of image reconstruction are discussed. (author)
New reconstruction algorithm in helical-volume CT
International Nuclear Information System (INIS)
Toki, Y.; Rifu, T.; Aradate, H.; Hirao, Y.; Ohyama, N.
1990-01-01
This paper reports on helical scanning that is an application of continuous scanning CT to acquire volume data in a short time for three-dimensional study. In a helical scan, the patient couch sustains movement during continuous-rotation scanning and then the acquired data is processed to synthesize a projection data set of vertical section by interpolation. But the synthesized section is not thin enough; also, the image may have artifacts caused by couch movement. A new reconstruction algorithm that helps resolve such problems has been developed and compared with the ordinary algorithm. The authors constructed a helical scan system based on TCT-900S, which can perform 1-second rotation continuously for 30 seconds. The authors measured section thickness using both algorithms on an AAPM phantom, and we also compared degree of artifacts on clinical data
Kim, Hyungjin; Park, Chang Min; Lee, Myunghee; Park, Sang Joon; Song, Yong Sub; Lee, Jong Hyuk; Hwang, Eui Jin; Goo, Jin Mo
2016-01-01
To identify the impact of reconstruction algorithms on CT radiomic features of pulmonary tumors and to reveal and compare the intra- and inter-reader and inter-reconstruction algorithm variability of each feature. Forty-two patients (M:F = 19:23; mean age, 60.43±10.56 years) with 42 pulmonary tumors (22.56±8.51mm) underwent contrast-enhanced CT scans, which were reconstructed with filtered back projection and commercial iterative reconstruction algorithm (level 3 and 5). Two readers independently segmented the whole tumor volume. Fifteen radiomic features were extracted and compared among reconstruction algorithms. Intra- and inter-reader variability and inter-reconstruction algorithm variability were calculated using coefficients of variation (CVs) and then compared. Among the 15 features, 5 first-order tumor intensity features and 4 gray level co-occurrence matrix (GLCM)-based features showed significant differences (palgorithms. As for the variability, effective diameter, sphericity, entropy, and GLCM entropy were the most robust features (CV≤5%). Inter-reader variability was larger than intra-reader or inter-reconstruction algorithm variability in 9 features. However, for entropy, homogeneity, and 4 GLCM-based features, inter-reconstruction algorithm variability was significantly greater than inter-reader variability (palgorithms. Inter-reconstruction algorithm variability was greater than inter-reader variability for entropy, homogeneity, and GLCM-based features.
Development of computed tomography system and image reconstruction algorithm
International Nuclear Information System (INIS)
Khairiah Yazid; Mohd Ashhar Khalid; Azaman Ahmad; Khairul Anuar Mohd Salleh; Ab Razak Hamzah
2006-01-01
Computed tomography is one of the most advanced and powerful nondestructive inspection techniques, which is currently used in many different industries. In several CT systems, detection has been by combination of an X-ray image intensifier and charge -coupled device (CCD) camera or by using line array detector. The recent development of X-ray flat panel detector has made fast CT imaging feasible and practical. Therefore this paper explained the arrangement of a new detection system which is using the existing high resolution (127 μm pixel size) flat panel detector in MINT and the image reconstruction technique developed. The aim of the project is to develop a prototype flat panel detector based CT imaging system for NDE. The prototype consisted of an X-ray tube, a flat panel detector system, a rotation table and a computer system to control the sample motion and image acquisition. Hence this project is divided to two major tasks, firstly to develop image reconstruction algorithm and secondly to integrate X-ray imaging components into one CT system. The image reconstruction algorithm using filtered back-projection method is developed and compared to other techniques. The MATLAB program is the tools used for the simulations and computations for this project. (Author)
Fast half-sibling population reconstruction: theory and algorithms.
Dexter, Daniel; Brown, Daniel G
2013-07-12
Kinship inference is the task of identifying genealogically related individuals. Kinship information is important for determining mating structures, notably in endangered populations. Although many solutions exist for reconstructing full sibling relationships, few exist for half-siblings. We consider the problem of determining whether a proposed half-sibling population reconstruction is valid under Mendelian inheritance assumptions. We show that this problem is NP-complete and provide a 0/1 integer program that identifies the minimum number of individuals that must be removed from a population in order for the reconstruction to become valid. We also present SibJoin, a heuristic-based clustering approach based on Mendelian genetics, which is strikingly fast. The software is available at http://github.com/ddexter/SibJoin.git+. Our SibJoin algorithm is reasonably accurate and thousands of times faster than existing algorithms. The heuristic is used to infer a half-sibling structure for a population which was, until recently, too large to evaluate.
Angle Statistics Reconstruction: a robust reconstruction algorithm for Muon Scattering Tomography
Stapleton, M.; Burns, J.; Quillin, S.; Steer, C.
2014-11-01
Muon Scattering Tomography (MST) is a technique for using the scattering of cosmic ray muons to probe the contents of enclosed volumes. As a muon passes through material it undergoes multiple Coulomb scattering, where the amount of scattering is dependent on the density and atomic number of the material as well as the path length. Hence, MST has been proposed as a means of imaging dense materials, for instance to detect special nuclear material in cargo containers. Algorithms are required to generate an accurate reconstruction of the material density inside the volume from the muon scattering information and some have already been proposed, most notably the Point of Closest Approach (PoCA) and Maximum Likelihood/Expectation Maximisation (MLEM) algorithms. However, whilst PoCA-based algorithms are easy to implement, they perform rather poorly in practice. Conversely, MLEM is a complicated algorithm to implement and computationally intensive and there is currently no published, fast and easily-implementable algorithm that performs well in practice. In this paper, we first provide a detailed analysis of the source of inaccuracy in PoCA-based algorithms. We then motivate an alternative method, based on ideas first laid out by Morris et al, presenting and fully specifying an algorithm that performs well against simulations of realistic scenarios. We argue this new algorithm should be adopted by developers of Muon Scattering Tomography as an alternative to PoCA.
An improved muon reconstruction algorithm for INO-ICAL experiment
International Nuclear Information System (INIS)
Bhattacharya, Kolahal; MandaI, Naba K.
2013-01-01
The charge current interaction of neutrino in INO-ICAL detector will be identified with a muon (μ ± ) in the detector whose kinematics is related with the kinematics of the neutrino. So, muon reconstruction is a very important step in achieving INO physics goals. The existing muon reconstruction package for INO-ICAL has poor performance in specific regimes of experimental interest: (a) for larger zenith angle (θ > 50°), (b) for lower energies (E < 1 GeV); mainly due to poor error propagation scheme insensitive to energy E, angle (θ, φ) and inhomogeneous magnetic field along the muon track. Since, a significant fraction of muons from atmospheric neutrino interactions will have initial energy < 1 GeV and almost uniform distribution in cosθ a robust package for muon reconstruction is essential. We have implemented higher order correction terms in the propagation of the state and error covariance matrices of the Kalman Iter. The algorithm ensures track element merging in most cases and also increases reconstruction efficiency. The performance of this package will be presented in comparison with the previous one. (author)
Yu, Nengjie; Li Qing Feng; Tang, Chuan-Xiang; Zheng, Shuxin
2005-01-01
A new method for low energy electron beam profile measurement is advanced, which presents a full 2-D beam profile distribution other than the traditional 2-D beam profile distribution given by 1-D vertical and horizontal beam profiles. The method is based on the CT (Computer Tomography) algorithm. Multi-sets of data about the 1-D beam profile projections are attained by rotating the multi-wire scanner. Then a 2-D beam profile is reconstructed from these projections with CT algorithm. The principle of this method is presented. The simulation and the experiment results are compared and analyzed in detail.
Research on compressive sensing reconstruction algorithm based on total variation model
Gao, Yu-xuan; Sun, Huayan; Zhang, Tinghua; Du, Lin
2017-12-01
Compressed sensing for breakthrough Nyquist sampling theorem provides a strong theoretical , making compressive sampling for image signals be carried out simultaneously. In traditional imaging procedures using compressed sensing theory, not only can it reduces the storage space, but also can reduce the demand for detector resolution greatly. Using the sparsity of image signal, by solving the mathematical model of inverse reconfiguration, realize the super-resolution imaging. Reconstruction algorithm is the most critical part of compression perception, to a large extent determine the accuracy of the reconstruction of the image.The reconstruction algorithm based on the total variation (TV) model is more suitable for the compression reconstruction of the two-dimensional image, and the better edge information can be obtained. In order to verify the performance of the algorithm, Simulation Analysis the reconstruction result in different coding mode of the reconstruction algorithm based on the TV reconstruction algorithm. The reconstruction effect of the reconfigurable algorithm based on TV based on the different coding methods is analyzed to verify the stability of the algorithm. This paper compares and analyzes the typical reconstruction algorithm in the same coding mode. On the basis of the minimum total variation algorithm, the Augmented Lagrangian function term is added and the optimal value is solved by the alternating direction method.Experimental results show that the reconstruction algorithm is compared with the traditional classical algorithm based on TV has great advantages, under the low measurement rate can be quickly and accurately recovers target image.
Optimization of the muon reconstruction algorithms for LHCb Run 2
Aaij, Roel; Dettori, Francesco; Dungs, Kevin; Lopes, Helder; Martinez Santos, Diego; Prisciandaro, Jessica; Sciascia, Barbara; Syropoulos, Vasileios; Stahl, Sascha; Vazquez Gomez, Ricardo
2017-01-01
The muon identiﬁcation algorithm in the LHCb HLT software trigger and offline reconstruction has been revisited in view of the LHC Run 2. This software has undergone a signiﬁcant refactorisation, resulting in a modularized common code base between the HLT and offline event processing. Because of the latter, the muon identiﬁcation is now identical in HLT and offline. The HLT1 algorithm sequence has been updated given the new rate and timing constraints. Also, information from the TT subdetector is used in order to reduce ghost tracks and optimize for low $p_T$ muons. The current software is presented here together with performance studies showing improved efficiencies and reduced timing.
Shape reconstruction from apparent contours theory and algorithms
Bellettini, Giovanni; Paolini, Maurizio
2015-01-01
Motivated by a variational model concerning the depth of the objects in a picture and the problem of hidden and illusory contours, this book investigates one of the central problems of computer vision: the topological and algorithmic reconstruction of a smooth three dimensional scene starting from the visible part of an apparent contour. The authors focus their attention on the manipulation of apparent contours using a finite set of elementary moves, which correspond to diffeomorphic deformations of three dimensional scenes. A large part of the book is devoted to the algorithmic part, with implementations, experiments, and computed examples. The book is intended also as a user's guide to the software code appcontour, written for the manipulation of apparent contours and their invariants. This book is addressed to theoretical and applied scientists working in the field of mathematical models of image segmentation.
The SRT reconstruction algorithm for semiquantification in PET imaging
Energy Technology Data Exchange (ETDEWEB)
Kastis, George A., E-mail: gkastis@academyofathens.gr [Research Center of Mathematics, Academy of Athens, Athens 11527 (Greece); Gaitanis, Anastasios [Biomedical Research Foundation of the Academy of Athens (BRFAA), Athens 11527 (Greece); Samartzis, Alexandros P. [Nuclear Medicine Department, Evangelismos General Hospital, Athens 10676 (Greece); Fokas, Athanasios S. [Department of Applied Mathematics and Theoretical Physics, University of Cambridge, Cambridge CB30WA, United Kingdom and Research Center of Mathematics, Academy of Athens, Athens 11527 (Greece)
2015-10-15
Purpose: The spline reconstruction technique (SRT) is a new, fast algorithm based on a novel numerical implementation of an analytic representation of the inverse Radon transform. The mathematical details of this algorithm and comparisons with filtered backprojection were presented earlier in the literature. In this study, the authors present a comparison between SRT and the ordered-subsets expectation–maximization (OSEM) algorithm for determining contrast and semiquantitative indices of {sup 18}F-FDG uptake. Methods: The authors implemented SRT in the software for tomographic image reconstruction (STIR) open-source platform and evaluated this technique using simulated and real sinograms obtained from the GE Discovery ST positron emission tomography/computer tomography scanner. All simulations and reconstructions were performed in STIR. For OSEM, the authors used the clinical protocol of their scanner, namely, 21 subsets and two iterations. The authors also examined images at one, four, six, and ten iterations. For the simulation studies, the authors analyzed an image-quality phantom with cold and hot lesions. Two different versions of the phantom were employed at two different hot-sphere lesion-to-background ratios (LBRs), namely, 2:1 and 4:1. For each noiseless sinogram, 20 Poisson realizations were created at five different noise levels. In addition to making visual comparisons of the reconstructed images, the authors determined contrast and bias as a function of the background image roughness (IR). For the real-data studies, sinograms of an image-quality phantom simulating the human torso were employed. The authors determined contrast and LBR as a function of the background IR. Finally, the authors present plots of contrast as a function of IR after smoothing each reconstructed image with Gaussian filters of six different sizes. Statistical significance was determined by employing the Wilcoxon rank-sum test. Results: In both simulated and real studies, SRT
The SRT reconstruction algorithm for semiquantification in PET imaging
International Nuclear Information System (INIS)
Kastis, George A.; Gaitanis, Anastasios; Samartzis, Alexandros P.; Fokas, Athanasios S.
2015-01-01
Purpose: The spline reconstruction technique (SRT) is a new, fast algorithm based on a novel numerical implementation of an analytic representation of the inverse Radon transform. The mathematical details of this algorithm and comparisons with filtered backprojection were presented earlier in the literature. In this study, the authors present a comparison between SRT and the ordered-subsets expectation–maximization (OSEM) algorithm for determining contrast and semiquantitative indices of 18 F-FDG uptake. Methods: The authors implemented SRT in the software for tomographic image reconstruction (STIR) open-source platform and evaluated this technique using simulated and real sinograms obtained from the GE Discovery ST positron emission tomography/computer tomography scanner. All simulations and reconstructions were performed in STIR. For OSEM, the authors used the clinical protocol of their scanner, namely, 21 subsets and two iterations. The authors also examined images at one, four, six, and ten iterations. For the simulation studies, the authors analyzed an image-quality phantom with cold and hot lesions. Two different versions of the phantom were employed at two different hot-sphere lesion-to-background ratios (LBRs), namely, 2:1 and 4:1. For each noiseless sinogram, 20 Poisson realizations were created at five different noise levels. In addition to making visual comparisons of the reconstructed images, the authors determined contrast and bias as a function of the background image roughness (IR). For the real-data studies, sinograms of an image-quality phantom simulating the human torso were employed. The authors determined contrast and LBR as a function of the background IR. Finally, the authors present plots of contrast as a function of IR after smoothing each reconstructed image with Gaussian filters of six different sizes. Statistical significance was determined by employing the Wilcoxon rank-sum test. Results: In both simulated and real studies, SRT
Track reconstruction algorithms for the CBM experiment at FAIR
International Nuclear Information System (INIS)
Lebedev, Andrey; Hoehne, Claudia; Kisel, Ivan; Ososkov, Gennady
2010-01-01
The Compressed Baryonic Matter (CBM) experiment at the future FAIR accelerator complex at Darmstadt is being designed for a comprehensive measurement of hadron and lepton production in heavy-ion collisions from 8-45 AGeV beam energy, producing events with large track multiplicity and high hit density. The setup consists of several detectors including as tracking detectors the silicon tracking system (STS), the muon detector (MUCH) or alternatively a set of Transition Radiation Detectors (TRD). In this contribution, the status of the track reconstruction software including track finding, fitting and propagation is presented for the MUCH and TRD detectors. The track propagation algorithm takes into account an inhomogeneous magnetic field and includes accurate calculation of multiple scattering and energy losses in the detector material. Track parameters and covariance matrices are estimated using the Kalman filter method and a Kalman filter modification by assigning weights to hits and using simulated annealing. Three different track finder algorithms based on track following have been developed which either allow for track branches, just select nearest hits or use the mentioned weighting method. The track reconstruction efficiency for central Au+Au collisions at 25 AGeV beam energy using events from the UrQMD model is at the level of 93-95% for both detectors.
International Nuclear Information System (INIS)
Koyama, Hisanobu; Ohno, Yoshiharu; Yamazaki, Youichi; Nogami, Munenobu; Kusaka, Akiko; Murase, Kenya; Sugimura, Kazuro
2010-01-01
This study aimed the influences of reconstruction algorithm for quantitative assessments in interstitial pneumonia patients. A total of 25 collagen vascular disease patients (nine male patients and 16 female patients; mean age, 57.2 years; age range 32-77 years) underwent thin-section MDCT examinations, and MDCT data were reconstructed with three kinds of reconstruction algorithm (two high-frequencies [A and B] and one standard [C]). In reconstruction algorithm B, the effect of low- and middle-frequency space was suppressed compared with reconstruction algorithm A. As quantitative CT parameters, kurtosis, skewness, and mean lung density (MLD) were acquired from a frequency histogram of the whole lung parenchyma in each reconstruction algorithm. To determine the difference of quantitative CT parameters affected by reconstruction algorithms, these parameters were compared statistically. To determine the relationships with the disease severity, these parameters were correlated with PFTs. In the results, all the histogram parameters values had significant differences each other (p < 0.0001) and those of reconstruction algorithm C were the highest. All MLDs had fair or moderate correlation with all parameters of PFT (-0.64 < r < -0.45, p < 0.05). Though kurtosis and skewness in high-frequency reconstruction algorithm A had significant correlations with all parameters of PFT (-0.61 < r < -0.45, p < 0.05), there were significant correlations only with diffusing capacity of carbon monoxide (DLco) and total lung capacity (TLC) in reconstruction algorithm C and with forced expiratory volume in 1 s (FEV1), DLco and TLC in reconstruction algorithm B. In conclusion, reconstruction algorithm has influence to quantitative assessments on chest thin-section MDCT examination in interstitial pneumonia patients.
Energy Technology Data Exchange (ETDEWEB)
Koyama, Hisanobu [Department of Radiology, Hyogo Kaibara Hospital, 5208-1 Kaibara, Kaibara-cho, Tanba 669-3395 (Japan)], E-mail: hisanobu19760104@yahoo.co.jp; Ohno, Yoshiharu [Department of Radiology, Kobe University Graduate School of Medicine, 7-5-2 Kusunoki-cho, Chuo-ku, Kobe 650-0017 (Japan)], E-mail: yosirad@kobe-u.ac.jp; Yamazaki, Youichi [Department of Medical Physics and Engineering, Faculty of Health Sciences, Graduate School of Medicine, Osaka University, 1-7 Yamadaoka, Suita 565-0871 (Japan)], E-mail: y.yamazk@sahs.med.osaka-u.ac.jp; Nogami, Munenobu [Division of PET, Institute of Biomedical Research and Innovation, 2-2 MInamimachi, Minatojima, Chu0-ku, Kobe 650-0047 (Japan)], E-mail: aznogami@fbri.org; Kusaka, Akiko [Division of Radiology, Kobe University Hospital, 7-5-2 Kusunoki-cho, Chuo-ku, Kobe 650-0017 (Japan)], E-mail: a.kusaka@hosp.kobe-u.ac.jp; Murase, Kenya [Department of Medical Physics and Engineering, Faculty of Health Sciences, Graduate School of Medicine, Osaka University, 1-7 Yamadaoka, Suita 565-0871 (Japan)], E-mail: murase@sahs.med.osaka-u.ac.jp; Sugimura, Kazuro [Department of Radiology, Kobe University Graduate School of Medicine, 7-5-2 Kusunoki-cho, Chuo-ku, Kobe 650-0017 (Japan)], E-mail: sugimura@med.kobe-u.ac.jp
2010-04-15
This study aimed the influences of reconstruction algorithm for quantitative assessments in interstitial pneumonia patients. A total of 25 collagen vascular disease patients (nine male patients and 16 female patients; mean age, 57.2 years; age range 32-77 years) underwent thin-section MDCT examinations, and MDCT data were reconstructed with three kinds of reconstruction algorithm (two high-frequencies [A and B] and one standard [C]). In reconstruction algorithm B, the effect of low- and middle-frequency space was suppressed compared with reconstruction algorithm A. As quantitative CT parameters, kurtosis, skewness, and mean lung density (MLD) were acquired from a frequency histogram of the whole lung parenchyma in each reconstruction algorithm. To determine the difference of quantitative CT parameters affected by reconstruction algorithms, these parameters were compared statistically. To determine the relationships with the disease severity, these parameters were correlated with PFTs. In the results, all the histogram parameters values had significant differences each other (p < 0.0001) and those of reconstruction algorithm C were the highest. All MLDs had fair or moderate correlation with all parameters of PFT (-0.64 < r < -0.45, p < 0.05). Though kurtosis and skewness in high-frequency reconstruction algorithm A had significant correlations with all parameters of PFT (-0.61 < r < -0.45, p < 0.05), there were significant correlations only with diffusing capacity of carbon monoxide (DLco) and total lung capacity (TLC) in reconstruction algorithm C and with forced expiratory volume in 1 s (FEV1), DLco and TLC in reconstruction algorithm B. In conclusion, reconstruction algorithm has influence to quantitative assessments on chest thin-section MDCT examination in interstitial pneumonia patients.
A parallel stereo reconstruction algorithm with applications in entomology (APSRA)
Bhasin, Rajesh; Jang, Won Jun; Hart, John C.
2012-03-01
We propose a fast parallel algorithm for the reconstruction of 3-Dimensional point clouds of insects from binocular stereo image pairs using a hierarchical approach for disparity estimation. Entomologists study various features of insects to classify them, build their distribution maps, and discover genetic links between specimens among various other essential tasks. This information is important to the pesticide and the pharmaceutical industries among others. When considering the large collections of insects entomologists analyze, it becomes difficult to physically handle the entire collection and share the data with researchers across the world. With the method presented in our work, Entomologists can create an image database for their collections and use the 3D models for studying the shape and structure of the insects thus making it easier to maintain and share. Initial feedback shows that the reconstructed 3D models preserve the shape and size of the specimen. We further optimize our results to incorporate multiview stereo which produces better overall structure of the insects. Our main contribution is applying stereoscopic vision techniques to entomology to solve the problems faced by entomologists.
Principle and Reconstruction Algorithm for Atomic-Resolution Holography
Matsushita, Tomohiro; Muro, Takayuki; Matsui, Fumihiko; Happo, Naohisa; Hosokawa, Shinya; Ohoyama, Kenji; Sato-Tomita, Ayana; Sasaki, Yuji C.; Hayashi, Kouichi
2018-06-01
Atomic-resolution holography makes it possible to obtain the three-dimensional (3D) structure around a target atomic site. Translational symmetry of the atomic arrangement of the sample is not necessary, and the 3D atomic image can be measured when the local structure of the target atomic site is oriented. Therefore, 3D local atomic structures such as dopants and adsorbates are observable. Here, the atomic-resolution holography comprising photoelectron holography, X-ray fluorescence holography, neutron holography, and their inverse modes are treated. Although the measurement methods are different, they can be handled with a unified theory. The algorithm for reconstructing 3D atomic images from holograms plays an important role. Although Fourier transform-based methods have been proposed, they require the multiple-energy holograms. In addition, they cannot be directly applied to photoelectron holography because of the phase shift problem. We have developed methods based on the fitting method for reconstructing from single-energy and photoelectron holograms. The developed methods are applicable to all types of atomic-resolution holography.
A fast iterative soft-thresholding algorithm for few-view CT reconstruction
Energy Technology Data Exchange (ETDEWEB)
Wu, Junfeng; Mou, Xuanqin; Zhang, Yanbo [Jiaotong Univ., Xi' an (China). Inst. of Image Processing and Pattern Recognition
2011-07-01
Iterative soft-thresholding algorithms with total variation regularization can produce high-quality reconstructions from few views and even in the presence of noise. However, these algorithms are known to converge quite slowly, with a proven theoretically global convergence rate O(1/k), where k is iteration number. In this paper, we present a fast iterative soft-thresholding algorithm for few-view fan beam CT reconstruction with a global convergence rate O(1/k{sup 2}), which is significantly faster than the iterative soft-thresholding algorithm. Simulation results demonstrate the superior performance of the proposed algorithm in terms of convergence speed and reconstruction quality. (orig.)
Novel image reconstruction algorithm for multi-phase flow tomography system using γ ray method
International Nuclear Information System (INIS)
Hao Kuihong; Wang Huaxiang; Gao Mei
2007-01-01
After analyzing the reason of image reconstructed algorithm by using the conventional back projection (IBP) is prone to produce spurious line, and considering the characteristic of multi-phase flow tomography, a novel image reconstruction algorithm is proposed, which carries out the intersection calculation using back projection data. This algorithm can obtain a perfect system point spread function, and can eliminate spurious line better. Simulating results show that the algorithm is effective for identifying multi-phase flow pattern. (authors)
A Super-resolution Reconstruction Algorithm for Surveillance Video
Directory of Open Access Journals (Sweden)
Jian Shao
2017-01-01
Full Text Available Recent technological developments have resulted in surveillance video becoming a primary method of preserving public security. Many city crimes are observed in surveillance video. The most abundant evidence collected by the police is also acquired through surveillance video sources. Surveillance video footage offers very strong support for solving criminal cases, therefore, creating an effective policy, and applying useful methods to the retrieval of additional evidence is becoming increasingly important. However, surveillance video has had its failings, namely, video footage being captured in low resolution (LR and bad visual quality. In this paper, we discuss the characteristics of surveillance video and describe the manual feature registration – maximum a posteriori – projection onto convex sets to develop a super-resolution reconstruction method, which improves the quality of surveillance video. From this method, we can make optimal use of information contained in the LR video image, but we can also control the image edge clearly as well as the convergence of the algorithm. Finally, we make a suggestion on how to adjust the algorithm adaptability by analyzing the prior information of target image.
Motion tolerant iterative reconstruction algorithm for cone-beam helical CT imaging
Energy Technology Data Exchange (ETDEWEB)
Takahashi, Hisashi; Goto, Taiga; Hirokawa, Koichi; Miyazaki, Osamu [Hitachi Medical Corporation, Chiba-ken (Japan). CT System Div.
2011-07-01
We have developed a new advanced iterative reconstruction algorithm for cone-beam helical CT. The features of this algorithm are: (a) it uses separable paraboloidal surrogate (SPS) technique as a foundation for reconstruction to reduce noise and cone-beam artifact, (b) it uses a view weight in the back-projection process to reduce motion artifact. To confirm the improvement of our proposed algorithm over other existing algorithm, such as Feldkamp-Davis-Kress (FDK) or SPS algorithm, we compared the motion artifact reduction, image noise reduction (standard deviation of CT number), and cone-beam artifact reduction on simulated and clinical data set. Our results demonstrate that the proposed algorithm dramatically reduces motion artifacts compared with the SPS algorithm, and decreases image noise compared with the FDK algorithm. In addition, the proposed algorithm potentially improves time resolution of iterative reconstruction. (orig.)
Bae, Kyung-hoon; Park, Changhan; Kim, Eun-soo
2008-03-01
In this paper, intermediate view reconstruction (IVR) using adaptive disparity search algorithm (ASDA) is for realtime 3-dimensional (3D) processing proposed. The proposed algorithm can reduce processing time of disparity estimation by selecting adaptive disparity search range. Also, the proposed algorithm can increase the quality of the 3D imaging. That is, by adaptively predicting the mutual correlation between stereo images pair using the proposed algorithm, the bandwidth of stereo input images pair can be compressed to the level of a conventional 2D image and a predicted image also can be effectively reconstructed using a reference image and disparity vectors. From some experiments, stereo sequences of 'Pot Plant' and 'IVO', it is shown that the proposed algorithm improves the PSNRs of a reconstructed image to about 4.8 dB by comparing with that of conventional algorithms, and reduces the Synthesizing time of a reconstructed image to about 7.02 sec by comparing with that of conventional algorithms.
Li, Yuanbo; Cui, Xiaoqian; Wang, Hongbei; Zhao, Mengge; Ding, Hongbin
2017-10-01
Digital speckle pattern interferometry (DSPI) can diagnose the topography evolution in real-time, continuous and non-destructive, and has been considered as a most promising technique for Plasma-Facing Components (PFCs) topography diagnostic under the complicated environment of tokamak. It is important for the study of digital speckle pattern interferometry to enhance speckle patterns and obtain the real topography of the ablated crater. In this paper, two kinds of numerical model based on flood-fill algorithm has been developed to obtain the real profile by unwrapping from the wrapped phase in speckle interference pattern, which can be calculated through four intensity images by means of 4-step phase-shifting technique. During the process of phase unwrapping by means of flood-fill algorithm, since the existence of noise pollution, and other inevitable factors will lead to poor quality of the reconstruction results, this will have an impact on the authenticity of the restored topography. The calculation of the quality parameters was introduced to obtain the quality-map from the wrapped phase map, this work presents two different methods to calculate the quality parameters. Then quality parameters are used to guide the path of flood-fill algorithm, and the pixels with good quality parameters are given priority calculation, so that the quality of speckle interference pattern reconstruction results are improved. According to the comparison between the flood-fill algorithm which is suitable for speckle pattern interferometry and the quality-guided flood-fill algorithm (with two different calculation approaches), the errors which caused by noise pollution and the discontinuous of the strips were successfully reduced.
Algorithm of hadron energy reconstruction for combined calorimeters in the DELPHI detector
International Nuclear Information System (INIS)
Gotra, Yu.N.; Tsyganov, E.N.; Zimin, N.I.; Zinchenko, A.I.
1989-01-01
The algorithm of hadron energy reconstruction from responses of electromagnetic and hadron calorimeters is described. The investigations have been carried out using the full-scale prototype of the hadron calorimeter cylindrical part modules. The supposed algorithm allows one to improve energy resolution by 5-7% with conserving the linearly of reconstructed hadron energy. 5 refs.; 4 figs.; 1 tab
The influence of image reconstruction algorithms on linear thorax EIT image analysis of ventilation
International Nuclear Information System (INIS)
Zhao, Zhanqi; Möller, Knut; Frerichs, Inéz; Pulletz, Sven; Müller-Lisse, Ullrich
2014-01-01
Analysis methods of electrical impedance tomography (EIT) images based on different reconstruction algorithms were examined. EIT measurements were performed on eight mechanically ventilated patients with acute respiratory distress syndrome. A maneuver with step increase of airway pressure was performed. EIT raw data were reconstructed offline with (1) filtered back-projection (BP); (2) the Dräger algorithm based on linearized Newton–Raphson (DR); (3) the GREIT (Graz consensus reconstruction algorithm for EIT) reconstruction algorithm with a circular forward model (GR C ) and (4) GREIT with individual thorax geometry (GR T ). Individual thorax contours were automatically determined from the routine computed tomography images. Five indices were calculated on the resulting EIT images respectively: (a) the ratio between tidal and deep inflation impedance changes; (b) tidal impedance changes in the right and left lungs; (c) center of gravity; (d) the global inhomogeneity index and (e) ventilation delay at mid-dorsal regions. No significant differences were found in all examined indices among the four reconstruction algorithms (p > 0.2, Kruskal–Wallis test). The examined algorithms used for EIT image reconstruction do not influence the selected indices derived from the EIT image analysis. Indices that validated for images with one reconstruction algorithm are also valid for other reconstruction algorithms. (paper)
The influence of image reconstruction algorithms on linear thorax EIT image analysis of ventilation.
Zhao, Zhanqi; Frerichs, Inéz; Pulletz, Sven; Müller-Lisse, Ullrich; Möller, Knut
2014-06-01
Analysis methods of electrical impedance tomography (EIT) images based on different reconstruction algorithms were examined. EIT measurements were performed on eight mechanically ventilated patients with acute respiratory distress syndrome. A maneuver with step increase of airway pressure was performed. EIT raw data were reconstructed offline with (1) filtered back-projection (BP); (2) the Dräger algorithm based on linearized Newton-Raphson (DR); (3) the GREIT (Graz consensus reconstruction algorithm for EIT) reconstruction algorithm with a circular forward model (GR(C)) and (4) GREIT with individual thorax geometry (GR(T)). Individual thorax contours were automatically determined from the routine computed tomography images. Five indices were calculated on the resulting EIT images respectively: (a) the ratio between tidal and deep inflation impedance changes; (b) tidal impedance changes in the right and left lungs; (c) center of gravity; (d) the global inhomogeneity index and (e) ventilation delay at mid-dorsal regions. No significant differences were found in all examined indices among the four reconstruction algorithms (p > 0.2, Kruskal-Wallis test). The examined algorithms used for EIT image reconstruction do not influence the selected indices derived from the EIT image analysis. Indices that validated for images with one reconstruction algorithm are also valid for other reconstruction algorithms.
A general algorithm for the reconstruction of jet events in e+e- annihilation
International Nuclear Information System (INIS)
Goddard, M.C.
1981-01-01
A general method is described to reconstruct a predetermined number of jets. It can reconstruct the jet axes as accurately as any existing algorithm and is up to one hundred times faster. Results are shown from the reconstruction of 2-jet, 3-jet and 4-jet Monte Carlo events. (author)
International Nuclear Information System (INIS)
Tang Jie; Nett, Brian E; Chen Guanghong
2009-01-01
Of all available reconstruction methods, statistical iterative reconstruction algorithms appear particularly promising since they enable accurate physical noise modeling. The newly developed compressive sampling/compressed sensing (CS) algorithm has shown the potential to accurately reconstruct images from highly undersampled data. The CS algorithm can be implemented in the statistical reconstruction framework as well. In this study, we compared the performance of two standard statistical reconstruction algorithms (penalized weighted least squares and q-GGMRF) to the CS algorithm. In assessing the image quality using these iterative reconstructions, it is critical to utilize realistic background anatomy as the reconstruction results are object dependent. A cadaver head was scanned on a Varian Trilogy system at different dose levels. Several figures of merit including the relative root mean square error and a quality factor which accounts for the noise performance and the spatial resolution were introduced to objectively evaluate reconstruction performance. A comparison is presented between the three algorithms for a constant undersampling factor comparing different algorithms at several dose levels. To facilitate this comparison, the original CS method was formulated in the framework of the statistical image reconstruction algorithms. Important conclusions of the measurements from our studies are that (1) for realistic neuro-anatomy, over 100 projections are required to avoid streak artifacts in the reconstructed images even with CS reconstruction, (2) regardless of the algorithm employed, it is beneficial to distribute the total dose to more views as long as each view remains quantum noise limited and (3) the total variation-based CS method is not appropriate for very low dose levels because while it can mitigate streaking artifacts, the images exhibit patchy behavior, which is potentially harmful for medical diagnosis.
Reconstruction algorithm in compressed sensing based on maximum a posteriori estimation
International Nuclear Information System (INIS)
Takeda, Koujin; Kabashima, Yoshiyuki
2013-01-01
We propose a systematic method for constructing a sparse data reconstruction algorithm in compressed sensing at a relatively low computational cost for general observation matrix. It is known that the cost of ℓ 1 -norm minimization using a standard linear programming algorithm is O(N 3 ). We show that this cost can be reduced to O(N 2 ) by applying the approach of posterior maximization. Furthermore, in principle, the algorithm from our approach is expected to achieve the widest successful reconstruction region, which is evaluated from theoretical argument. We also discuss the relation between the belief propagation-based reconstruction algorithm introduced in preceding works and our approach
Directory of Open Access Journals (Sweden)
Mingjian Sun
2015-01-01
Full Text Available Photoacoustic imaging is an innovative imaging technique to image biomedical tissues. The time reversal reconstruction algorithm in which a numerical model of the acoustic forward problem is run backwards in time is widely used. In the paper, a time reversal reconstruction algorithm based on particle swarm optimization (PSO optimized support vector machine (SVM interpolation method is proposed for photoacoustics imaging. Numerical results show that the reconstructed images of the proposed algorithm are more accurate than those of the nearest neighbor interpolation, linear interpolation, and cubic convolution interpolation based time reversal algorithm, which can provide higher imaging quality by using significantly fewer measurement positions or scanning times.
Maximum entropy algorithm and its implementation for the neutral beam profile measurement
Energy Technology Data Exchange (ETDEWEB)
Lee, Seung Wook; Cho, Gyu Seong [Korea Advanced Institute of Science and Technology, Taejon (Korea, Republic of); Cho, Yong Sub [Korea Atomic Energy Research Institute, Taejon (Korea, Republic of)
1997-12-31
A tomography algorithm to maximize the entropy of image using Lagrangian multiplier technique and conjugate gradient method has been designed for the measurement of 2D spatial distribution of intense neutral beams of KSTAR NBI (Korea Superconducting Tokamak Advanced Research Neutral Beam Injector), which is now being designed. A possible detection system was assumed and a numerical simulation has been implemented to test the reconstruction quality of given beam profiles. This algorithm has the good applicability for sparse projection data and thus, can be used for the neutral beam tomography. 8 refs., 3 figs. (Author)
Maximum entropy algorithm and its implementation for the neutral beam profile measurement
Energy Technology Data Exchange (ETDEWEB)
Lee, Seung Wook; Cho, Gyu Seong [Korea Advanced Institute of Science and Technology, Taejon (Korea, Republic of); Cho, Yong Sub [Korea Atomic Energy Research Institute, Taejon (Korea, Republic of)
1998-12-31
A tomography algorithm to maximize the entropy of image using Lagrangian multiplier technique and conjugate gradient method has been designed for the measurement of 2D spatial distribution of intense neutral beams of KSTAR NBI (Korea Superconducting Tokamak Advanced Research Neutral Beam Injector), which is now being designed. A possible detection system was assumed and a numerical simulation has been implemented to test the reconstruction quality of given beam profiles. This algorithm has the good applicability for sparse projection data and thus, can be used for the neutral beam tomography. 8 refs., 3 figs. (Author)
Directory of Open Access Journals (Sweden)
Chen Deyun
2013-01-01
Full Text Available According to the image reconstruction accuracy influenced by the “soft field” nature and ill-conditioned problems in electrical capacitance tomography, a superresolution image reconstruction algorithm based on Landweber is proposed in the paper, which is based on the working principle of the electrical capacitance tomography system. The method uses the algorithm which is derived by regularization of solutions derived and derives closed solution by fast Fourier transform of the convolution kernel. So, it ensures the certainty of the solution and improves the stability and quality of image reconstruction results. Simulation results show that the imaging precision and real-time imaging of the algorithm are better than Landweber algorithm, and this algorithm proposes a new method for the electrical capacitance tomography image reconstruction algorithm.
Parallelization of the model-based iterative reconstruction algorithm DIRA
International Nuclear Information System (INIS)
Oertenberg, A.; Sandborg, M.; Alm Carlsson, G.; Malusek, A.; Magnusson, M.
2016-01-01
New paradigms for parallel programming have been devised to simplify software development on multi-core processors and many-core graphical processing units (GPU). Despite their obvious benefits, the parallelization of existing computer programs is not an easy task. In this work, the use of the Open Multiprocessing (OpenMP) and Open Computing Language (OpenCL) frameworks is considered for the parallelization of the model-based iterative reconstruction algorithm DIRA with the aim to significantly shorten the code's execution time. Selected routines were parallelized using OpenMP and OpenCL libraries; some routines were converted from MATLAB to C and optimised. Parallelization of the code with the OpenMP was easy and resulted in an overall speedup of 15 on a 16-core computer. Parallelization with OpenCL was more difficult owing to differences between the central processing unit and GPU architectures. The resulting speedup was substantially lower than the theoretical peak performance of the GPU; the cause was explained. (authors)
FPGA Hardware Acceleration of a Phylogenetic Tree Reconstruction with Maximum Parsimony Algorithm
BLOCK, Henry; MARUYAMA, Tsutomu
2017-01-01
In this paper, we present an FPGA hardware implementation for a phylogenetic tree reconstruction with a maximum parsimony algorithm. We base our approach on a particular stochastic local search algorithm that uses the Progressive Neighborhood and the Indirect Calculation of Tree Lengths method. This method is widely used for the acceleration of the phylogenetic tree reconstruction algorithm in software. In our implementation, we define a tree structure and accelerate the search by parallel an...
Neural network algorithm for image reconstruction using the grid friendly projections
International Nuclear Information System (INIS)
Cierniak, R.
2011-01-01
Full text: The presented paper describes a development of original approach to the reconstruction problem using a recurrent neural network. Particularly, the 'grid-friendly' angles of performed projections are selected according to the discrete Radon transform (DRT) concept to decrease the number of projections required. The methodology of our approach is consistent with analytical reconstruction algorithms. Reconstruction problem is reformulated in our approach to optimization problem. This problem is solved in present concept using method based on the maximum likelihood methodology. The reconstruction algorithm proposed in this work is consequently adapted for more practical discrete fan beam projections. Computer simulation results show that the neural network reconstruction algorithm designed to work in this way improves obtained results and outperforms conventional methods in reconstructed image quality. (author)
A maximum-likelihood reconstruction algorithm for tomographic gamma-ray nondestructive assay
International Nuclear Information System (INIS)
Prettyman, T.H.; Estep, R.J.; Cole, R.A.; Sheppard, G.A.
1994-01-01
A new tomographic reconstruction algorithm for nondestructive assay with high resolution gamma-ray spectroscopy (HRGS) is presented. The reconstruction problem is formulated using a maximum-likelihood approach in which the statistical structure of both the gross and continuum measurements used to determine the full-energy response in HRGS is precisely modeled. An accelerated expectation-maximization algorithm is used to determine the optimal solution. The algorithm is applied to safeguards and environmental assays of large samples (for example, 55-gal. drums) in which high continuum levels caused by Compton scattering are routinely encountered. Details of the implementation of the algorithm and a comparative study of the algorithm's performance are presented
DEFF Research Database (Denmark)
Sidky, Emil Y.; Jørgensen, Jakob Heide; Pan, Xiaochuan
2012-01-01
The primal–dual optimization algorithm developed in Chambolle and Pock (CP) (2011 J. Math. Imag. Vis. 40 1–26) is applied to various convex optimization problems of interest in computed tomography (CT) image reconstruction. This algorithm allows for rapid prototyping of optimization problems...... for the purpose of designing iterative image reconstruction algorithms for CT. The primal–dual algorithm is briefly summarized in this paper, and its potential for prototyping is demonstrated by explicitly deriving CP algorithm instances for many optimization problems relevant to CT. An example application...
A fast 4D cone beam CT reconstruction method based on the OSC-TV algorithm.
Mascolo-Fortin, Julia; Matenine, Dmitri; Archambault, Louis; Després, Philippe
2018-01-01
Four-dimensional cone beam computed tomography allows for temporally resolved imaging with useful applications in radiotherapy, but raises particular challenges in terms of image quality and computation time. The purpose of this work is to develop a fast and accurate 4D algorithm by adapting a GPU-accelerated ordered subsets convex algorithm (OSC), combined with the total variation minimization regularization technique (TV). Different initialization schemes were studied to adapt the OSC-TV algorithm to 4D reconstruction: each respiratory phase was initialized either with a 3D reconstruction or a blank image. Reconstruction algorithms were tested on a dynamic numerical phantom and on a clinical dataset. 4D iterations were implemented for a cluster of 8 GPUs. All developed methods allowed for an adequate visualization of the respiratory movement and compared favorably to the McKinnon-Bates and adaptive steepest descent projection onto convex sets algorithms, while the 4D reconstructions initialized from a prior 3D reconstruction led to better overall image quality. The most suitable adaptation of OSC-TV to 4D CBCT was found to be a combination of a prior FDK reconstruction and a 4D OSC-TV reconstruction with a reconstruction time of 4.5 minutes. This relatively short reconstruction time could facilitate a clinical use.
Current profile reconstruction using electron temperature imaging diagnostics
International Nuclear Information System (INIS)
Tritz, K.; Stutman, D.; Delgado-Aparicio, L.F.; Finkenthal, M.; Pacella, D.; Kaita, R.; Stratton, B.; Sabbagh, S.
2004-01-01
Flux surface shape information can be used to constrain the current profile for reconstruction of the plasma equilibrium. One method of inferring flux surface shape relies on plasma x-ray emission; however, deviations from the flux surfaces due to impurity and density asymmetries complicate the interpretation. Electron isotherm surfaces should correspond well to the plasma flux surfaces, and equilibrium constraint modeling using this isotherm information constrains the current profile. The KFIT code is used to assess the profile uncertainty and to optimize the number, location and SNR required for the Te detectors. As Te imaging detectors we consider tangentially viewing, vertically spaced, linear gas electron multiplier arrays operated in pulse height analysis (PHA) mode and multifoil soft x-ray arrays. Isoflux coordinate sets provided by T e measurements offer a strong constraint on the equilibrium reconstruction in both a stacked horizontal array configuration and a crossed horizontal and vertical beam system, with q 0 determined to within ±4%. The required SNR can be provided with either PHA or multicolor diagnostic techniques, though the multicolor system requires ∼x4 better statistics for comparable final errors
Research on Image Reconstruction Algorithms for Tuber Electrical Resistance Tomography System
Directory of Open Access Journals (Sweden)
Jiang Zili
2016-01-01
Full Text Available The application of electrical resistance tomography (ERT technology has been expanded to the field of agriculture, and the concept of TERT (Tuber Electrical Resistance Tomography is proposed. On the basis of the research on the forward and the inverse problems of the TERT system, a hybrid algorithm based on genetic algorithm is proposed, which can be used in TERT system to monitor the growth status of the plant tubers. The image reconstruction of TERT system is different from the conventional ERT system for two phase-flow measurement. Imaging of TERT needs more precision measurement and the conventional ERT cares more about the image reconstruction speed. A variety of algorithms are analyzed and optimized for the purpose of making them suitable for TERT system. For example: linear back projection, modified Newton-Raphson and genetic algorithm. Experimental results showed that the novel hybrid algorithm is superior to other algorithm and it can effectively improve the image reconstruction quality.
Joint-2D-SL0 Algorithm for Joint Sparse Matrix Reconstruction
Directory of Open Access Journals (Sweden)
Dong Zhang
2017-01-01
Full Text Available Sparse matrix reconstruction has a wide application such as DOA estimation and STAP. However, its performance is usually restricted by the grid mismatch problem. In this paper, we revise the sparse matrix reconstruction model and propose the joint sparse matrix reconstruction model based on one-order Taylor expansion. And it can overcome the grid mismatch problem. Then, we put forward the Joint-2D-SL0 algorithm which can solve the joint sparse matrix reconstruction problem efficiently. Compared with the Kronecker compressive sensing method, our proposed method has a higher computational efficiency and acceptable reconstruction accuracy. Finally, simulation results validate the superiority of the proposed method.
Level-set-based reconstruction algorithm for EIT lung images: first clinical results.
Rahmati, Peyman; Soleimani, Manuchehr; Pulletz, Sven; Frerichs, Inéz; Adler, Andy
2012-05-01
We show the first clinical results using the level-set-based reconstruction algorithm for electrical impedance tomography (EIT) data. The level-set-based reconstruction method (LSRM) allows the reconstruction of non-smooth interfaces between image regions, which are typically smoothed by traditional voxel-based reconstruction methods (VBRMs). We develop a time difference formulation of the LSRM for 2D images. The proposed reconstruction method is applied to reconstruct clinical EIT data of a slow flow inflation pressure-volume manoeuvre in lung-healthy and adult lung-injury patients. Images from the LSRM and the VBRM are compared. The results show comparable reconstructed images, but with an improved ability to reconstruct sharp conductivity changes in the distribution of lung ventilation using the LSRM.
Level-set-based reconstruction algorithm for EIT lung images: first clinical results
International Nuclear Information System (INIS)
Rahmati, Peyman; Adler, Andy; Soleimani, Manuchehr; Pulletz, Sven; Frerichs, Inéz
2012-01-01
We show the first clinical results using the level-set-based reconstruction algorithm for electrical impedance tomography (EIT) data. The level-set-based reconstruction method (LSRM) allows the reconstruction of non-smooth interfaces between image regions, which are typically smoothed by traditional voxel-based reconstruction methods (VBRMs). We develop a time difference formulation of the LSRM for 2D images. The proposed reconstruction method is applied to reconstruct clinical EIT data of a slow flow inflation pressure–volume manoeuvre in lung-healthy and adult lung-injury patients. Images from the LSRM and the VBRM are compared. The results show comparable reconstructed images, but with an improved ability to reconstruct sharp conductivity changes in the distribution of lung ventilation using the LSRM. (paper)
Detector independent cellular automaton algorithm for track reconstruction
Energy Technology Data Exchange (ETDEWEB)
Kisel, Ivan; Kulakov, Igor; Zyzak, Maksym [Goethe Univ. Frankfurt am Main (Germany); Frankfurt Institute for Advanced Studies, Frankfurt am Main (Germany); GSI Helmholtzzentrum fuer Schwerionenforschung GmbH (Germany); Collaboration: CBM-Collaboration
2013-07-01
Track reconstruction is one of the most challenging problems of data analysis in modern high energy physics (HEP) experiments, which have to process per second of the order of 10{sup 7} events with high track multiplicity and density, registered by detectors of different types and, in many cases, located in non-homogeneous magnetic field. Creation of reconstruction package common for all experiments is considered to be important in order to consolidate efforts. The cellular automaton (CA) track reconstruction approach has been used successfully in many HEP experiments. It is very simple, efficient, local and parallel. Meanwhile it is intrinsically independent of detector geometry and good candidate for common track reconstruction. The CA implementation for the CBM experiment has been generalized and applied to the ALICE ITS and STAR HFT detectors. Tests with simulated collisions have been performed. The track reconstruction efficiencies are at the level of 95% for majority of the signal tracks for all detectors.
International Nuclear Information System (INIS)
Zeeberg, B.R.; Bacharach, S.; Carson, R.; Green, M.V.; Larson, S.M.; Soucaille, J.F.
1985-01-01
An algorithm is presented which permits the reconstruction of SPECT images in the presence of spatially varying attenuation. The algorithm considers the spatially variant attenuation as a perturbation of the constant attenuation case and computes a reconstructed image and a correction image to estimate the effects of this perturbation. The corrected image will be computed from these two images and is of comparable quality both visually and quantitatively to those simulated for zero or constant attenuation taken as standard reference images. In addition, the algorithm is time efficient, in that the time required is approximately 2.5 times that for a standard convolution-back projection algorithm
An Approximate Cone Beam Reconstruction Algorithm for Gantry-Tilted CT Using Tangential Filtering
Directory of Open Access Journals (Sweden)
Ming Yan
2006-01-01
Full Text Available FDK algorithm is a well-known 3D (three-dimensional approximate algorithm for CT (computed tomography image reconstruction and is also known to suffer from considerable artifacts when the scanning cone angle is large. Recently, it has been improved by performing the ramp filtering along the tangential direction of the X-ray source helix for dealing with the large cone angle problem. In this paper, we present an FDK-type approximate reconstruction algorithm for gantry-tilted CT imaging. The proposed method improves the image reconstruction by filtering the projection data along a proper direction which is determined by CT parameters and gantry-tilted angle. As a result, the proposed algorithm for gantry-tilted CT reconstruction can provide more scanning flexibilities in clinical CT scanning and is efficient in computation. The performance of the proposed algorithm is evaluated with turbell clock phantom and thorax phantom and compared with FDK algorithm and a popular 2D (two-dimensional approximate algorithm. The results show that the proposed algorithm can achieve better image quality for gantry-tilted CT image reconstruction.
A multiresolution approach to iterative reconstruction algorithms in X-ray computed tomography.
De Witte, Yoni; Vlassenbroeck, Jelle; Van Hoorebeke, Luc
2010-09-01
In computed tomography, the application of iterative reconstruction methods in practical situations is impeded by their high computational demands. Especially in high resolution X-ray computed tomography, where reconstruction volumes contain a high number of volume elements (several giga voxels), this computational burden prevents their actual breakthrough. Besides the large amount of calculations, iterative algorithms require the entire volume to be kept in memory during reconstruction, which quickly becomes cumbersome for large data sets. To overcome this obstacle, we present a novel multiresolution reconstruction, which greatly reduces the required amount of memory without significantly affecting the reconstructed image quality. It is shown that, combined with an efficient implementation on a graphical processing unit, the multiresolution approach enables the application of iterative algorithms in the reconstruction of large volumes at an acceptable speed using only limited resources.
Miller, D; Lippert, C; Vollmer, F; Bozinov, O; Benes, L; Schulte, D M; Sure, U
2012-09-01
Freehand three-dimensional ultrasound imaging (3D-US) is increasingly used in image-guided surgery. During image acquisition, a set of B-scans is acquired that is distributed in a non-parallel manner over the area of interest. Reconstructing these images into a regular array allows 3D visualization. However, the reconstruction process may introduce artefacts and may therefore reduce image quality. The aim of the study is to compare different algorithms with respect to image quality and diagnostic value for image guidance in neurosurgery. 3D-US data sets were acquired during surgery of various intracerebral lesions using an integrated ultrasound-navigation device. They were stored for post-hoc evaluation. Five different reconstruction algorithms, a standard multiplanar reconstruction with interpolation (MPR), a pixel nearest neighbour method (PNN), a voxel nearest neighbour method (VNN) and two voxel based distance-weighted algorithms (VNN2 and DW) were tested with respect to image quality and artefact formation. The capability of the algorithm to fill gaps within the sample volume was investigated and a clinical evaluation with respect to the diagnostic value of the reconstructed images was performed. MPR was significantly worse than the other algorithms in filling gaps. In an image subtraction test, VNN2 and DW reliably reconstructed images even if large amounts of data were missing. However, the quality of the reconstruction improved, if data acquisition was performed in a structured manner. When evaluating the diagnostic value of reconstructed axial, sagittal and coronal views, VNN2 and DW were judged to be significantly better than MPR and VNN. VNN2 and DW could be identified as robust algorithms that generate reconstructed US images with a high diagnostic value. These algorithms improve the utility and reliability of 3D-US imaging during intraoperative navigation. Copyright © 2012 John Wiley & Sons, Ltd.
International Nuclear Information System (INIS)
Bastiens, K.; Lemahieu, I.
1994-01-01
The application of a maximum entropy reconstruction algorithm to PET images requires a lot of computing resources. A parallel implementation could seriously reduce the execution time. However, programming a parallel application is still a non trivial task, needing specialized people. In this paper a programming environment based on a visual programming language is used for a parallel implementation of the reconstruction algorithm. This programming environment allows less experienced programmers to use the performance of multiprocessor systems. (authors)
Energy Technology Data Exchange (ETDEWEB)
Bastiens, K; Lemahieu, I [University of Ghent - ELIS Department, St. Pietersnieuwstraat 41, B-9000 Ghent (Belgium)
1994-12-31
The application of a maximum entropy reconstruction algorithm to PET images requires a lot of computing resources. A parallel implementation could seriously reduce the execution time. However, programming a parallel application is still a non trivial task, needing specialized people. In this paper a programming environment based on a visual programming language is used for a parallel implementation of the reconstruction algorithm. This programming environment allows less experienced programmers to use the performance of multiprocessor systems. (authors). 8 refs, 3 figs, 1 tab.
A practical exact maximum compatibility algorithm for reconstruction of recent evolutionary history
Cherry, Joshua L.
2017-01-01
Background Maximum compatibility is a method of phylogenetic reconstruction that is seldom applied to molecular sequences. It may be ideal for certain applications, such as reconstructing phylogenies of closely-related bacteria on the basis of whole-genome sequencing. Results Here I present an algorithm that rapidly computes phylogenies according to a compatibility criterion. Although based on solutions to the maximum clique problem, this algorithm deals properly with ambiguities in the data....
DEFF Research Database (Denmark)
Cappellin, C.; Pivnenko, Sergey; Jørgensen, E.
2013-01-01
This paper focuses on three important features of the 3D reconstruction algorithm of DIATOOL: the identification of array elements improper functioning and failure, the obtainable spatial resolution of the reconstructed fields and currents, and the filtering of undesired radiation and scattering...
An efficient algorithm for MR image reconstruction and compression
International Nuclear Information System (INIS)
Wang, Hang; Rosenfeld, D.; Braun, M.; Yan, Hong
1992-01-01
In magnetic resonance imaging (MRI), the original data are sampled in the spatial frequency domain. The sampled data thus constitute a set of discrete Fourier transform (DFT) coefficients. The image is usually reconstructed by taking inverse DFT. The image data may then be efficiently compressed using the discrete cosine transform (DCT). A method of using DCT to treat the sampled data is presented which combines two procedures, image reconstruction and data compression. This method may be particularly useful in medical picture archiving and communication systems where both image reconstruction and compression are important issues. 11 refs., 3 figs
Accurate 3D reconstruction by a new PDS-OSEM algorithm for HRRT
International Nuclear Information System (INIS)
Chen, Tai-Been; Horng-Shing Lu, Henry; Kim, Hang-Keun; Son, Young-Don; Cho, Zang- Hee
2014-01-01
State-of-the-art high resolution research tomography (HRRT) provides high resolution PET images with full 3D human brain scanning. But, a short time frame in dynamic study causes many problems related to the low counts in the acquired data. The PDS-OSEM algorithm was proposed to reconstruct the HRRT image with a high signal-to-noise ratio that provides accurate information for dynamic data. The new algorithm was evaluated by simulated image, empirical phantoms, and real human brain data. Meanwhile, the time activity curve was adopted to validate a reconstructed performance of dynamic data between PDS-OSEM and OP-OSEM algorithms. According to simulated and empirical studies, the PDS-OSEM algorithm reconstructs images with higher quality, higher accuracy, less noise, and less average sum of square error than those of OP-OSEM. The presented algorithm is useful to provide quality images under the condition of low count rates in dynamic studies with a short scan time. - Highlights: • The PDS-OSEM reconstructs PET images with iteratively compensating random and scatter corrections from prompt sinogram. • The PDS-OSEM can reconstruct PET images with low count data and data contaminations. • The PDS-OSEM provides less noise and higher quality of reconstructed images than those of OP-OSEM algorithm in statistical sense
A Total Variation Regularization Based Super-Resolution Reconstruction Algorithm for Digital Video
Directory of Open Access Journals (Sweden)
Zhang Liangpei
2007-01-01
Full Text Available Super-resolution (SR reconstruction technique is capable of producing a high-resolution image from a sequence of low-resolution images. In this paper, we study an efficient SR algorithm for digital video. To effectively deal with the intractable problems in SR video reconstruction, such as inevitable motion estimation errors, noise, blurring, missing regions, and compression artifacts, the total variation (TV regularization is employed in the reconstruction model. We use the fixed-point iteration method and preconditioning techniques to efficiently solve the associated nonlinear Euler-Lagrange equations of the corresponding variational problem in SR. The proposed algorithm has been tested in several cases of motion and degradation. It is also compared with the Laplacian regularization-based SR algorithm and other TV-based SR algorithms. Experimental results are presented to illustrate the effectiveness of the proposed algorithm.
International Nuclear Information System (INIS)
Mancini, R. C.
2007-01-01
We discuss several methods for the extraction of temperature and density spatial profiles in inertial confinement fusion implosion cores based on the analysis of the x-ray emission from spectroscopic tracers added to the deuterium fuel. The ideas rely on (1) detailed spectral models that take into account collisional-radiative atomic kinetics, Stark broadened line shapes, and radiation transport calculations, (2) the availability of narrow-band, gated pinhole and slit x-ray images, and space-resolved line spectra of the core, and (3) several data analysis and reconstruction methods that include a multi-objective search and optimization technique based on a novel application of Pareto genetic algorithms to plasma spectroscopy. The spectroscopic analysis yields the spatial profiles of temperature and density in the core at the collapse of the implosion, and also the extent of shell material mixing into the core. Results are illustrated with data recorded in implosion experiments driven by the OMEGA and Z facilities
Quasi Gradient Projection Algorithm for Sparse Reconstruction in Compressed Sensing
Directory of Open Access Journals (Sweden)
Xin Meng
2014-02-01
Full Text Available Compressed sensing is a novel signal sampling theory under the condition that the signal is sparse or compressible. The existing recovery algorithms based on the gradient projection can either need prior knowledge or recovery the signal poorly. In this paper, a new algorithm based on gradient projection is proposed, which is referred as Quasi Gradient Projection. The algorithm presented quasi gradient direction and two step sizes schemes along this direction. The algorithm doesn’t need any prior knowledge of the original signal. Simulation results demonstrate that the presented algorithm cans recovery the signal more correctly than GPSR which also don’t need prior knowledge. Meanwhile, the algorithm has a lower computation complexity.
International Nuclear Information System (INIS)
Nascimento, F.M.; Sergeenkov, S.; Araujo-Moreira, F.M.
2012-01-01
By using a specially designed algorithm (based on utilizing the so-called Hierarchical Data Format), we report on successful reconstruction of 3D profiles of local flux distribution within artificially prepared arrays of unshunted Nb-AlO x -Nb Josephson junctions from 2D surface images obtained via the scanning SQUID microscope. The analysis of the obtained results suggest that for large sweep areas, the local flux distribution significantly deviates from the conventional picture and exhibits a more complicated avalanche-type behavior with a prominent dendritic structure. -- Highlights: ► The penetration of external magnetic field into an array of Nb-AlO x -Nb Josephson junctions is studied. ► Using Scanning SQUID Microscope, 2D images of local flux distribution within array are obtained. ► Using specially designed pattern recognition algorithm, 3D flux profiles are reconstructed from 2D images.
International Nuclear Information System (INIS)
Bliznakova, K.; Kolitsi, Z.; Speller, R. D.; Horrocks, J. A.; Tromba, G.; Pallikarakis, N.
2010-01-01
Purpose: In this article, the image quality of reconstructed volumes by four algorithms for digital tomosynthesis, applied in the case of breast, is investigated using synchrotron radiation. Methods: An angular data set of 21 images of a complex phantom with heterogeneous tissue-mimicking background was obtained using the SYRMEP beamline at ELETTRA Synchrotron Light Laboratory, Trieste, Italy. The irradiated part was reconstructed using the multiple projection algorithm (MPA) and the filtered backprojection with ramp followed by hamming windows (FBR-RH) and filtered backprojection with ramp (FBP-R). Additionally, an algorithm for reducing the noise in reconstructed planes based on noise mask subtraction from the planes of the originally reconstructed volume using MPA (MPA-NM) has been further developed. The reconstruction techniques were evaluated in terms of calculations and comparison of the contrast-to-noise ratio (CNR) and artifact spread function. Results: It was found that the MPA-NM resulted in higher CNR, comparable with the CNR of FBP-RH for high contrast details. Low contrast objects are well visualized and characterized by high CNR using the simple MPA and the MPA-NM. In addition, the image quality of the reconstructed features in terms of CNR and visual appearance as a function of the initial number of projection images and the reconstruction arc was carried out. Slices reconstructed with more input projection images result in less reconstruction artifacts and higher detail CNR, while those reconstructed from projection images acquired in reduced angular range causes pronounced streak artifacts. Conclusions: Of the reconstruction algorithms implemented, the MPA-NM and MPA are a good choice for detecting low contrast objects, while the FBP-RH, FBP-R, and MPA-NM provide high CNR and well outlined edges in case of microcalcifications.
TV-constrained incremental algorithms for low-intensity CT image reconstruction
DEFF Research Database (Denmark)
Rose, Sean D.; Andersen, Martin S.; Sidky, Emil Y.
2015-01-01
constraint can be guided by an image reconstructed by filtered backprojection (FBP). We apply our algorithm to low-dose synchrotron X-ray CT data from the Advanced Photon Source (APS) at Argonne National Labs (ANL) to demonstrate its potential utility. We find that the algorithm provides a means of edge-preserving...
A First-Order Primal-Dual Reconstruction Algorithm for Few-View SPECT
DEFF Research Database (Denmark)
Wolf, Paul; Jørgensen, Jakob Heide; Gilat-Schmidt, Taly
2012-01-01
A sparsity-exploiting algorithm intended for few-view Single Photon Emission Computed Tomography (SPECT) reconstruction is proposed and characterized. The algorithm models the object as piecewise constant subject to a blurring operation. Monte Carlo simulations were performed to provide more proj...
International Nuclear Information System (INIS)
Van Gompel, Gert; Batenburg, K Joost; Defrise, Michel
2009-01-01
In this paper we consider the problem of reconstructing a two-dimensional star-shaped object of uniform density from truncated projections of the object. In particular, we prove that such an object is uniquely determined by its parallel projections sampled over a full π angular range with a detector that only covers an interior field-of-view, even if the density of the object is not known a priori. We analyze the stability of this reconstruction problem and propose a reconstruction algorithm. Simulation experiments demonstrate that the algorithm is capable of reconstructing a star-shaped object from interior data, even if the interior region is much smaller than the size of the object. In addition, we present results for a heuristic reconstruction algorithm called DART, that was recently proposed. The heuristic method is shown to yield accurate reconstructions if the density is known in advance, and to have a very good stability in the presence of noisy projection data. Finally, the performance of the DBP and DART algorithms is illustrated for the reconstruction of real micro-CT data of a diamond
Reconstruction of core axial power shapes using the alternating conditional expectation algorithm
International Nuclear Information System (INIS)
Lee, Eun Ki; Kim, Yong Hee; Cha, Kune Ho; Park, Moon Ghu
1999-01-01
We have introduced the alternating conditional expectation (ACE) algorithm in reconstructing 20-node axial core power shapes from five-level in-core detector powers. The core design code, Reactor Operation and Control Simulation (ROCS), calculates 3-dimensional power distributions for various core states, and the reference core-averaged axial power shapes and corresponding simulated detector powers are utilized to synthesize the axial power shape. By using the ACE algorithm, the optimal relationship between a dependent variable, the plane power, and independent variables, five detector powers, is determined without any preprocessing. A total of ∼3490 data sets per each cycle of YongGwang Nuclear (YGN) power plant units 3 and 4 is used for the regression. Continuous analytic function corresponding to each optimal transformation is calculated by simple regression model. The reconstructed axial power shapes of ∼21,200 cases are compared to the original ROCS axial power shapes. Also, to test the validity and accuracy of the new method, its performance is compared with that of the Fourier fitting method (FFM), a typical method of the deterministic approach. For a total of 21,204 data cases, the averages of root mean square (rms) error, axial peak error (ΔF z ), and axial shape index error (ΔASI) of new method are calculated as 0.81%, 0.51% and 0.00204, while those of FFM are 2.29%, 2.37% and 0.00264, respectively. We also evaluated the wide range of axial power profiles from the xenon-oscillation. The results show that the newly developed method is far superior to FFM; average rms and axial peak error are just ∼35 and ∼20% of those of FFM, respectively
Acceleration of the direct reconstruction of linear parametric images using nested algorithms
International Nuclear Information System (INIS)
Wang Guobao; Qi Jinyi
2010-01-01
Parametric imaging using dynamic positron emission tomography (PET) provides important information for biological research and clinical diagnosis. Indirect and direct methods have been developed for reconstructing linear parametric images from dynamic PET data. Indirect methods are relatively simple and easy to implement because the image reconstruction and kinetic modeling are performed in two separate steps. Direct methods estimate parametric images directly from raw PET data and are statistically more efficient. However, the convergence rate of direct algorithms can be slow due to the coupling between the reconstruction and kinetic modeling. Here we present two fast gradient-type algorithms for direct reconstruction of linear parametric images. The new algorithms decouple the reconstruction and linear parametric modeling at each iteration by employing the principle of optimization transfer. Convergence speed is accelerated by running more sub-iterations of linear parametric estimation because the computation cost of the linear parametric modeling is much less than that of the image reconstruction. Computer simulation studies demonstrated that the new algorithms converge much faster than the traditional expectation maximization (EM) and the preconditioned conjugate gradient algorithms for dynamic PET.
Solar Backscatter UV (SBUV total ozone and profile algorithm
Directory of Open Access Journals (Sweden)
P. K. Bhartia
2013-10-01
Full Text Available We describe the algorithm that has been applied to develop a 42 yr record of total ozone and ozone profiles from eight Solar Backscatter UV (SBUV instruments launched on NASA and NOAA satellites since April 1970. The Version 8 (V8 algorithm was released more than a decade ago and has been in use since then at NOAA to produce their operational ozone products. The current algorithm (V8.6 is basically the same as V8, except for updates to instrument calibration, incorporation of new ozone absorption cross-sections, and new ozone and cloud height climatologies. Since the V8 algorithm has been optimized for deriving monthly zonal mean (MZM anomalies for ozone assessment and model comparisons, our emphasis in this paper is primarily on characterizing the sources of errors that are relevant for such studies. When data are analyzed this way the effect of some errors, such as vertical smoothing of short-term variability, and noise due to clouds and aerosols diminish in importance, while the importance of others, such as errors due to vertical smoothing of the quasi-biennial oscillation (QBO and other periodic and aperiodic variations, become more important. With V8.6 zonal mean data we now provide smoothing kernels that can be used to compare anomalies in SBUV profile and partial ozone columns with models. In this paper we show how to use these kernels to compare SBUV data with Microwave Limb Sounder (MLS ozone profiles. These kernels are particularly useful for comparisons in the lower stratosphere where SBUV profiles have poor vertical resolution but partial column ozone values have high accuracy. We also provide our best estimate of the smoothing errors associated with SBUV MZM profiles. Since smoothing errors are the largest source of uncertainty in these profiles, they can be treated as error bars in deriving interannual variability and trends using SBUV data and for comparing with other measurements. In the V8 and V8.6 algorithms we derive total
A tracking algorithm for the reconstruction of the daughters of long-lived particles in LHCb
Dendek, Adam Mateusz
2018-01-01
A tracking algorithm for the reconstruction of the daughters of long-lived particles in LHCb 5 Jun 2018, 16:00 1h 30m Library, Centro San Domenico () LHC experiments Posters session Speaker Katharina Mueller (Universitaet Zuerich (CH)) Description The LHCb experiment at CERN operates a high precision and robust tracking system to reach its physics goals, including precise measurements of CP-violation phenomena in the heavy flavour quark sector and searches for New Physics beyond the Standard Model. The track reconstruction procedure is performed by a number of algorithms. One of these, PatLongLivedTracking, is optimised to reconstruct "downstream tracks", which are tracks originating from decays outside the LHCb vertex detector of long-lived particles, such as Ks or Λ0. After an overview of the LHCb tracking system, we provide a detailed description of the LHCb downstream track reconstruction algorithm. Its computational intelligence part is described in details, including the adaptation of the employed...
International Nuclear Information System (INIS)
Stevendaal, U. van; Schlomka, J.-P.; Harding, A.; Grass, M.
2003-01-01
Coherent scatter computed tomography (CSCT) is a reconstructive x-ray imaging technique that yields the spatially resolved coherent-scatter form factor of the investigated object. Reconstruction from coherently scattered x-rays is commonly done using algebraic reconstruction techniques (ART). In this paper, we propose an alternative approach based on filtered back-projection. For the first time, a three-dimensional (3D) filtered back-projection technique using curved 3D back-projection lines is applied to two-dimensional coherent scatter projection data. The proposed algorithm is tested with simulated projection data as well as with projection data acquired with a demonstrator setup similar to a multi-line CT scanner geometry. While yielding comparable image quality as ART reconstruction, the modified 3D filtered back-projection algorithm is about two orders of magnitude faster. In contrast to iterative reconstruction schemes, it has the advantage that subfield-of-view reconstruction becomes feasible. This allows a selective reconstruction of the coherent-scatter form factor for a region of interest. The proposed modified 3D filtered back-projection algorithm is a powerful reconstruction technique to be implemented in a CSCT scanning system. This method gives coherent scatter CT the potential of becoming a competitive modality for medical imaging or nondestructive testing
Reconstruction algorithm medical imaging DRR; Algoritmo de construccion de imagenes medicas DRR
Energy Technology Data Exchange (ETDEWEB)
Estrada Espinosa, J. C.
2013-07-01
The method of reconstruction for digital radiographic Imaging (DRR), is based on two orthogonal images, on the dorsal and lateral decubitus position of the simulation. DRR images are reconstructed with an algorithm that simulates running a conventional X-ray, a single rendition team, beam emitted is not divergent, in this case, the rays are considered to be parallel in the image reconstruction DRR, for this purpose, it is necessary to use all the values of the units (HU) hounsfield of each voxel in all axial cuts that form the study TC, finally obtaining the reconstructed image DRR performing a transformation from 3D to 2D. (Author)
Practical algorithms for simulation and reconstruction of digital in-line holograms.
Latychevskaia, Tatiana; Fink, Hans-Werner
2015-03-20
Here we present practical methods for simulation and reconstruction of in-line digital holograms recorded with plane and spherical waves. The algorithms described here are applicable to holographic imaging of an object exhibiting absorption as well as phase-shifting properties. Optimal parameters, related to distances, sampling rate, and other factors for successful simulation and reconstruction of holograms are evaluated and criteria for the achievable resolution are worked out. Moreover, we show that the numerical procedures for the reconstruction of holograms recorded with plane and spherical waves are identical under certain conditions. Experimental examples of holograms and their reconstructions are also discussed.
TauFinder: A Reconstruction Algorithm for τ Leptons at Linear Colliders
Muennich, A
2010-01-01
An algorithm to find and reconstruct τ leptons was developed, which targets τs that produce high energetic, low multiplicity jets as can be observed at multi TeV e+e− collisions. However, it makes no assumption about the decay of the τ candidate thus finding hadronic as well as leptonic decays. The algorithm delivers a reconstructed τ as seen by the detector. This note provides an overview of the algorithm, the cuts used and gives some evaluation of the performance. A first implementation is available within the ILC software framework as a MAR- LIN processor . Appendix A is intended as a short user manual.
A BPF-FBP tandem algorithm for image reconstruction in reverse helical cone-beam CT
International Nuclear Information System (INIS)
Cho, Seungryong; Xia, Dan; Pellizzari, Charles A.; Pan Xiaochuan
2010-01-01
Purpose: Reverse helical cone-beam computed tomography (CBCT) is a scanning configuration for potential applications in image-guided radiation therapy in which an accurate anatomic image of the patient is needed for image-guidance procedures. The authors previously developed an algorithm for image reconstruction from nontruncated data of an object that is completely within the reverse helix. The purpose of this work is to develop an image reconstruction approach for reverse helical CBCT of a long object that extends out of the reverse helix and therefore constitutes data truncation. Methods: The proposed approach comprises of two reconstruction steps. In the first step, a chord-based backprojection-filtration (BPF) algorithm reconstructs a volumetric image of an object from the original cone-beam data. Because there exists a chordless region in the middle of the reverse helix, the image obtained in the first step contains an unreconstructed central-gap region. In the second step, the gap region is reconstructed by use of a Pack-Noo-formula-based filteredbackprojection (FBP) algorithm from the modified cone-beam data obtained by subtracting from the original cone-beam data the reprojection of the image reconstructed in the first step. Results: The authors have performed numerical studies to validate the proposed approach in image reconstruction from reverse helical cone-beam data. The results confirm that the proposed approach can reconstruct accurate images of a long object without suffering from data-truncation artifacts or cone-angle artifacts. Conclusions: They developed and validated a BPF-FBP tandem algorithm to reconstruct images of a long object from reverse helical cone-beam data. The chord-based BPF algorithm was utilized for converting the long-object problem into a short-object problem. The proposed approach is applicable to other scanning configurations such as reduced circular sinusoidal trajectories.
Reconstruction of a digital core containing clay minerals based on a clustering algorithm
He, Yanlong; Pu, Chunsheng; Jing, Cheng; Gu, Xiaoyu; Chen, Qingdong; Liu, Hongzhi; Khan, Nasir; Dong, Qiaoling
2017-10-01
It is difficult to obtain a core sample and information for digital core reconstruction of mature sandstone reservoirs around the world, especially for an unconsolidated sandstone reservoir. Meanwhile, reconstruction and division of clay minerals play a vital role in the reconstruction of the digital cores, although the two-dimensional data-based reconstruction methods are specifically applicable as the microstructure reservoir simulation methods for the sandstone reservoir. However, reconstruction of clay minerals is still challenging from a research viewpoint for the better reconstruction of various clay minerals in the digital cores. In the present work, the content of clay minerals was considered on the basis of two-dimensional information about the reservoir. After application of the hybrid method, and compared with the model reconstructed by the process-based method, the digital core containing clay clusters without the labels of the clusters' number, size, and texture were the output. The statistics and geometry of the reconstruction model were similar to the reference model. In addition, the Hoshen-Kopelman algorithm was used to label various connected unclassified clay clusters in the initial model and then the number and size of clay clusters were recorded. At the same time, the K -means clustering algorithm was applied to divide the labeled, large connecting clusters into smaller clusters on the basis of difference in the clusters' characteristics. According to the clay minerals' characteristics, such as types, textures, and distributions, the digital core containing clay minerals was reconstructed by means of the clustering algorithm and the clay clusters' structure judgment. The distributions and textures of the clay minerals of the digital core were reasonable. The clustering algorithm improved the digital core reconstruction and provided an alternative method for the simulation of different clay minerals in the digital cores.
Reconstruction of a digital core containing clay minerals based on a clustering algorithm.
He, Yanlong; Pu, Chunsheng; Jing, Cheng; Gu, Xiaoyu; Chen, Qingdong; Liu, Hongzhi; Khan, Nasir; Dong, Qiaoling
2017-10-01
It is difficult to obtain a core sample and information for digital core reconstruction of mature sandstone reservoirs around the world, especially for an unconsolidated sandstone reservoir. Meanwhile, reconstruction and division of clay minerals play a vital role in the reconstruction of the digital cores, although the two-dimensional data-based reconstruction methods are specifically applicable as the microstructure reservoir simulation methods for the sandstone reservoir. However, reconstruction of clay minerals is still challenging from a research viewpoint for the better reconstruction of various clay minerals in the digital cores. In the present work, the content of clay minerals was considered on the basis of two-dimensional information about the reservoir. After application of the hybrid method, and compared with the model reconstructed by the process-based method, the digital core containing clay clusters without the labels of the clusters' number, size, and texture were the output. The statistics and geometry of the reconstruction model were similar to the reference model. In addition, the Hoshen-Kopelman algorithm was used to label various connected unclassified clay clusters in the initial model and then the number and size of clay clusters were recorded. At the same time, the K-means clustering algorithm was applied to divide the labeled, large connecting clusters into smaller clusters on the basis of difference in the clusters' characteristics. According to the clay minerals' characteristics, such as types, textures, and distributions, the digital core containing clay minerals was reconstructed by means of the clustering algorithm and the clay clusters' structure judgment. The distributions and textures of the clay minerals of the digital core were reasonable. The clustering algorithm improved the digital core reconstruction and provided an alternative method for the simulation of different clay minerals in the digital cores.
A three-dimensional reconstruction algorithm for an inverse-geometry volumetric CT system
International Nuclear Information System (INIS)
Schmidt, Taly Gilat; Fahrig, Rebecca; Pelc, Norbert J.
2005-01-01
An inverse-geometry volumetric computed tomography (IGCT) system has been proposed capable of rapidly acquiring sufficient data to reconstruct a thick volume in one circular scan. The system uses a large-area scanned source opposite a smaller detector. The source and detector have the same extent in the axial, or slice, direction, thus providing sufficient volumetric sampling and avoiding cone-beam artifacts. This paper describes a reconstruction algorithm for the IGCT system. The algorithm first rebins the acquired data into two-dimensional (2D) parallel-ray projections at multiple tilt and azimuthal angles, followed by a 3D filtered backprojection. The rebinning step is performed by gridding the data onto a Cartesian grid in a 4D projection space. We present a new method for correcting the gridding error caused by the finite and asymmetric sampling in the neighborhood of each output grid point in the projection space. The reconstruction algorithm was implemented and tested on simulated IGCT data. Results show that the gridding correction reduces the gridding errors to below one Hounsfield unit. With this correction, the reconstruction algorithm does not introduce significant artifacts or blurring when compared to images reconstructed from simulated 2D parallel-ray projections. We also present an investigation of the noise behavior of the method which verifies that the proposed reconstruction algorithm utilizes cross-plane rays as efficiently as in-plane rays and can provide noise comparable to an in-plane parallel-ray geometry for the same number of photons. Simulations of a resolution test pattern and the modulation transfer function demonstrate that the IGCT system, using the proposed algorithm, is capable of 0.4 mm isotropic resolution. The successful implementation of the reconstruction algorithm is an important step in establishing feasibility of the IGCT system
Mikhaylova, E; Kolstein, M; De Lorenzo, G; Chmeissani, M
2014-07-01
A novel positron emission tomography (PET) scanner design based on a room-temperature pixelated CdTe solid-state detector is being developed within the framework of the Voxel Imaging PET (VIP) Pathfinder project [1]. The simulation results show a great potential of the VIP to produce high-resolution images even in extremely challenging conditions such as the screening of a human head [2]. With unprecedented high channel density (450 channels/cm 3 ) image reconstruction is a challenge. Therefore optimization is needed to find the best algorithm in order to exploit correctly the promising detector potential. The following reconstruction algorithms are evaluated: 2-D Filtered Backprojection (FBP), Ordered Subset Expectation Maximization (OSEM), List-Mode OSEM (LM-OSEM), and the Origin Ensemble (OE) algorithm. The evaluation is based on the comparison of a true image phantom with a set of reconstructed images obtained by each algorithm. This is achieved by calculation of image quality merit parameters such as the bias, the variance and the mean square error (MSE). A systematic optimization of each algorithm is performed by varying the reconstruction parameters, such as the cutoff frequency of the noise filters and the number of iterations. The region of interest (ROI) analysis of the reconstructed phantom is also performed for each algorithm and the results are compared. Additionally, the performance of the image reconstruction methods is compared by calculating the modulation transfer function (MTF). The reconstruction time is also taken into account to choose the optimal algorithm. The analysis is based on GAMOS [3] simulation including the expected CdTe and electronic specifics.
International Nuclear Information System (INIS)
Tang Qiulin; Zeng, Gengsheng L; Gullberg, Grant T
2007-01-01
In this paper, we develop an approximate analytical reconstruction algorithm that compensates for uniform attenuation in 2D parallel-beam SPECT with a 180 0 acquisition. This new algorithm is in the form of a direct Fourier reconstruction. The complex variable central slice theorem is used to derive this algorithm. The image is reconstructed with the following steps: first, the attenuated projection data acquired over 180 deg. are extended to 360 deg. and the value for the uniform attenuator is changed to a negative value. The Fourier transform (FT) of the image in polar coordinates is obtained from the Fourier transform of an analytic function interpolated from an extension of the projection data according to the complex central slice theorem. Finally, the image is obtained by performing a 2D inverse Fourier transform. Computer simulations and comparison studies with a 360 deg. full-scan algorithm are provided
International Nuclear Information System (INIS)
Zeng, G.L.; Gullberg, G.T.
1990-01-01
Reconstruction artifacts in cone beam tomography are studied for filtered backprojection (Feldkamp) and iterative EM algorithms. The filtered backprojection algorithm uses a voxel-driven, interpolated backprojection to reconstruct the cone beam data; whereas, the iterative EM algorithm performs ray-driven projection and backprojection operations for each iteration. Two weight in schemes for the projection and backprojection operations in the EM algorithm are studied. One weights each voxel by the length of the ray through the voxel and the other equates the value of a voxel to the functional value of the midpoint of the line intersecting the voxel, which is obtained by interpolating between eight neighboring voxels. Cone beam reconstruction artifacts such as rings, bright vertical extremities, and slice-to slice cross talk are not found with parallel beam and fan beam geometries
Tau Reconstruction, Identification Algorithms and Performance in ATLAS
DEFF Research Database (Denmark)
Simonyan, M.
2013-01-01
identification of hadronically decaying tau leptons is achieved by using detailed information from tracking and calorimeter detector components. Variables describing the properties of calorimeter energy deposits and track reconstruction within tau candidates are combined in multi-variate discriminants...... by investigating single hadron calorimeter response, as well as kinematic distributions in Z¿ tt events....
Algorithms For Phylogeny Reconstruction In a New Mathematical Model
Lenzini, Gabriele; Marianelli, Silvia
1997-01-01
The evolutionary history of a set of species is represented by a tree called phylogenetic tree or phylogeny. Its structure depends on precise biological assumptions about the evolution of species. Problems related to phylogeny reconstruction (i.e., finding a tree representation of information
Miéville, Frédéric A.; Bolard, Gregory; Benkreira, Mohamed; Ayestaran, Paul; Gudinchet, François; Bochud, François; Verdun, Francis R.
2011-03-01
The noise power spectrum (NPS) is the reference metric for understanding the noise content in computed tomography (CT) images. To evaluate the noise properties of clinical multidetector (MDCT) scanners, local 2D and 3D NPSs were computed for different acquisition reconstruction parameters. A 64- and a 128-MDCT scanners were employed. Measurements were performed on a water phantom in axial and helical acquisition modes. CT dose index was identical for both installations. Influence of parameters such as the pitch, the reconstruction filter (soft, standard and bone) and the reconstruction algorithm (filtered-back projection (FBP), adaptive statistical iterative reconstruction (ASIR)) were investigated. Images were also reconstructed in the coronal plane using a reformat process. Then 2D and 3D NPS methods were computed. In axial acquisition mode, the 2D axial NPS showed an important magnitude variation as a function of the z-direction when measured at the phantom center. In helical mode, a directional dependency with lobular shape was observed while the magnitude of the NPS was kept constant. Important effects of the reconstruction filter, pitch and reconstruction algorithm were observed on 3D NPS results for both MDCTs. With ASIR, a reduction of the NPS magnitude and a shift of the NPS peak to the low frequency range were visible. 2D coronal NPS obtained from the reformat images was impacted by the interpolation when compared to 2D coronal NPS obtained from 3D measurements. The noise properties of volume measured in last generation MDCTs was studied using local 3D NPS metric. However, impact of the non-stationarity noise effect may need further investigations.
Directory of Open Access Journals (Sweden)
Schöning André
2016-01-01
Full Text Available Track reconstruction in high track multiplicity environments at current and future high rate particle physics experiments is a big challenge and very time consuming. The search for track seeds and the fitting of track candidates are usually the most time consuming steps in the track reconstruction. Here, a new and fast track reconstruction method based on hit triplets is proposed which exploits a three-dimensional fit model including multiple scattering and hit uncertainties from the very start, including the search for track seeds. The hit triplet based reconstruction method assumes a homogeneous magnetic field which allows to give an analytical solutions for the triplet fit result. This method is highly parallelizable, needs fewer operations than other standard track reconstruction methods and is therefore ideal for the implementation on parallel computing architectures. The proposed track reconstruction algorithm has been studied in the context of the Mu3e-experiment and a typical LHC experiment.
Volume reconstruction optimization for tomo-PIV algorithms applied to experimental data
Martins, Fabio J. W. A.; Foucaut, Jean-Marc; Thomas, Lionel; Azevedo, Luis F. A.; Stanislas, Michel
2015-08-01
Tomographic PIV is a three-component volumetric velocity measurement technique based on the tomographic reconstruction of a particle distribution imaged by multiple camera views. In essence, the performance and accuracy of this technique is highly dependent on the parametric adjustment and the reconstruction algorithm used. Although synthetic data have been widely employed to optimize experiments, the resulting reconstructed volumes might not have optimal quality. The purpose of the present study is to offer quality indicators that can be applied to data samples in order to improve the quality of velocity results obtained by the tomo-PIV technique. The methodology proposed can potentially lead to significantly reduction in the time required to optimize a tomo-PIV reconstruction, also leading to better quality velocity results. Tomo-PIV data provided by a six-camera turbulent boundary-layer experiment were used to optimize the reconstruction algorithms according to this methodology. Velocity statistics measurements obtained by optimized BIMART, SMART and MART algorithms were compared with hot-wire anemometer data and velocity measurement uncertainties were computed. Results indicated that BIMART and SMART algorithms produced reconstructed volumes with equivalent quality as the standard MART with the benefit of reduced computational time.
Volume reconstruction optimization for tomo-PIV algorithms applied to experimental data
International Nuclear Information System (INIS)
Martins, Fabio J W A; Foucaut, Jean-Marc; Stanislas, Michel; Thomas, Lionel; Azevedo, Luis F A
2015-01-01
Tomographic PIV is a three-component volumetric velocity measurement technique based on the tomographic reconstruction of a particle distribution imaged by multiple camera views. In essence, the performance and accuracy of this technique is highly dependent on the parametric adjustment and the reconstruction algorithm used. Although synthetic data have been widely employed to optimize experiments, the resulting reconstructed volumes might not have optimal quality. The purpose of the present study is to offer quality indicators that can be applied to data samples in order to improve the quality of velocity results obtained by the tomo-PIV technique. The methodology proposed can potentially lead to significantly reduction in the time required to optimize a tomo-PIV reconstruction, also leading to better quality velocity results. Tomo-PIV data provided by a six-camera turbulent boundary-layer experiment were used to optimize the reconstruction algorithms according to this methodology. Velocity statistics measurements obtained by optimized BIMART, SMART and MART algorithms were compared with hot-wire anemometer data and velocity measurement uncertainties were computed. Results indicated that BIMART and SMART algorithms produced reconstructed volumes with equivalent quality as the standard MART with the benefit of reduced computational time. (paper)
Directory of Open Access Journals (Sweden)
Berrak Akşam
2016-06-01
Full Text Available Objective: Most of the malignant cutaneous carcinomas are seen in the nasal region. Reconstruction of nasal defects is challenging because of the unique anatomic properties and complex structure of this region. In this study, we present our algorithm for the nasal skin defects that occurred after malignant skin tumor excisions. Material and Methods: Patients whose nasal skin was reconstructed after malignant skin tumor excision were included in the study. These patients were evaluated by their age, gender, comorbities, tumor location, tumor size, reconstruction type, histopathological diagnosis, and tumor recurrence. Results: A total of 130 patients (70 female, 60 male were evaluated. The average age of the patients was 67.8 years. Tumors were located mostly at the dorsum, alar region, and tip of the nose. When reconstruction methods were evaluated, primary closure was preferred in 14.6% patients, full thickness skin grafts were used in 25.3% patients, and reconstruction with flaps were the choice in 60% patients. Different flaps were used according to the subunits. Mostly, dorsal nasal flaps, bilobed flaps, nasolabial flaps, and forehead flaps were used. Conclusion: The defect-only reconstruction principle was accepted in this study. Previously described subunits, such as the dorsum, tip, alar region, lateral wall, columella, and soft triangles, of the nose were further divided into subregions by their anatomical relations. An algorithm was planned with these sub regions. In nasal skin reconstruction, this algorithm helps in selection the methods for the best results and minimize the complications.
Spectral CT metal artifact reduction with an optimization-based reconstruction algorithm
Gilat Schmidt, Taly; Barber, Rina F.; Sidky, Emil Y.
2017-03-01
Metal objects cause artifacts in computed tomography (CT) images. This work investigated the feasibility of a spectral CT method to reduce metal artifacts. Spectral CT acquisition combined with optimization-based reconstruction is proposed to reduce artifacts by modeling the physical effects that cause metal artifacts and by providing the flexibility to selectively remove corrupted spectral measurements in the spectral-sinogram space. The proposed Constrained `One-Step' Spectral CT Image Reconstruction (cOSSCIR) algorithm directly estimates the basis material maps while enforcing convex constraints. The incorporation of constraints on the reconstructed basis material maps is expected to mitigate undersampling effects that occur when corrupted data is excluded from reconstruction. The feasibility of the cOSSCIR algorithm to reduce metal artifacts was investigated through simulations of a pelvis phantom. The cOSSCIR algorithm was investigated with and without the use of a third basis material representing metal. The effects of excluding data corrupted by metal were also investigated. The results demonstrated that the proposed cOSSCIR algorithm reduced metal artifacts and improved CT number accuracy. For example, CT number error in a bright shading artifact region was reduced from 403 HU in the reference filtered backprojection reconstruction to 33 HU using the proposed algorithm in simulation. In the dark shading regions, the error was reduced from 1141 HU to 25 HU. Of the investigated approaches, decomposing the data into three basis material maps and excluding the corrupted data demonstrated the greatest reduction in metal artifacts.
Papaconstadopoulos, P; Levesque, I R; Maglieri, R; Seuntjens, J
2016-02-07
Direct determination of the source intensity distribution of clinical linear accelerators is still a challenging problem for small field beam modeling. Current techniques most often involve special equipment and are difficult to implement in the clinic. In this work we present a maximum-likelihood expectation-maximization (MLEM) approach to the source reconstruction problem utilizing small fields and a simple experimental set-up. The MLEM algorithm iteratively ray-traces photons from the source plane to the exit plane and extracts corrections based on photon fluence profile measurements. The photon fluence profiles were determined by dose profile film measurements in air using a high density thin foil as build-up material and an appropriate point spread function (PSF). The effect of other beam parameters and scatter sources was minimized by using the smallest field size ([Formula: see text] cm(2)). The source occlusion effect was reproduced by estimating the position of the collimating jaws during this process. The method was first benchmarked against simulations for a range of typical accelerator source sizes. The sources were reconstructed with an accuracy better than 0.12 mm in the full width at half maximum (FWHM) to the respective electron sources incident on the target. The estimated jaw positions agreed within 0.2 mm with the expected values. The reconstruction technique was also tested against measurements on a Varian Novalis Tx linear accelerator and compared to a previously commissioned Monte Carlo model. The reconstructed FWHM of the source agreed within 0.03 mm and 0.11 mm to the commissioned electron source in the crossplane and inplane orientations respectively. The impact of the jaw positioning, experimental and PSF uncertainties on the reconstructed source distribution was evaluated with the former presenting the dominant effect.
Goodenberger, Martin H; Wagner-Bartak, Nicolaus A; Gupta, Shiva; Liu, Xinming; Yap, Ramon Q; Sun, Jia; Tamm, Eric P; Jensen, Corey T
The purpose of this study was to compare abdominopelvic computed tomography images reconstructed with adaptive statistical iterative reconstruction-V (ASIR-V) with model-based iterative reconstruction (Veo 3.0), ASIR, and filtered back projection (FBP). Abdominopelvic computed tomography scans for 36 patients (26 males and 10 females) were reconstructed using FBP, ASIR (80%), Veo 3.0, and ASIR-V (30%, 60%, 90%). Mean ± SD patient age was 32 ± 10 years with mean ± SD body mass index of 26.9 ± 4.4 kg/m. Images were reviewed by 2 independent readers in a blinded, randomized fashion. Hounsfield unit, noise, and contrast-to-noise ratio (CNR) values were calculated for each reconstruction algorithm for further comparison. Phantom evaluation of low-contrast detectability (LCD) and high-contrast resolution was performed. Adaptive statistical iterative reconstruction-V 30%, ASIR-V 60%, and ASIR 80% were generally superior qualitatively compared with ASIR-V 90%, Veo 3.0, and FBP (P ASIR-V 60% with respective CNR values of 5.54 ± 2.39, 8.78 ± 3.15, and 3.49 ± 1.77 (P ASIR 80% had the best and worst spatial resolution, respectively. Adaptive statistical iterative reconstruction-V 30% and ASIR-V 60% provided the best combination of qualitative and quantitative performance. Adaptive statistical iterative reconstruction 80% was equivalent qualitatively, but demonstrated inferior spatial resolution and LCD.
Linear array implementation of the EM algorithm for PET image reconstruction
International Nuclear Information System (INIS)
Rajan, K.; Patnaik, L.M.; Ramakrishna, J.
1995-01-01
The PET image reconstruction based on the EM algorithm has several attractive advantages over the conventional convolution back projection algorithms. However, the PET image reconstruction based on the EM algorithm is computationally burdensome for today's single processor systems. In addition, a large memory is required for the storage of the image, projection data, and the probability matrix. Since the computations are easily divided into tasks executable in parallel, multiprocessor configurations are the ideal choice for fast execution of the EM algorithms. In tis study, the authors attempt to overcome these two problems by parallelizing the EM algorithm on a multiprocessor systems. The parallel EM algorithm on a linear array topology using the commercially available fast floating point digital signal processor (DSP) chips as the processing elements (PE's) has been implemented. The performance of the EM algorithm on a 386/387 machine, IBM 6000 RISC workstation, and on the linear array system is discussed and compared. The results show that the computational speed performance of a linear array using 8 DSP chips as PE's executing the EM image reconstruction algorithm is about 15.5 times better than that of the IBM 6000 RISC workstation. The novelty of the scheme is its simplicity. The linear array topology is expandable with a larger number of PE's. The architecture is not dependant on the DSP chip chosen, and the substitution of the latest DSP chip is straightforward and could yield better speed performance
International Nuclear Information System (INIS)
Choo, Ji Yung; Goo, Jin Mo; Park, Chang Min; Park, Sang Joon; Lee, Chang Hyun; Shim, Mi-Suk
2014-01-01
To evaluate filtered back projection (FBP) and two iterative reconstruction (IR) algorithms and their effects on the quantitative analysis of lung parenchyma and airway measurements on computed tomography (CT) images. Low-dose chest CT obtained in 281 adult patients were reconstructed using three algorithms: FBP, adaptive statistical IR (ASIR) and model-based IR (MBIR). Measurements of each dataset were compared: total lung volume, emphysema index (EI), airway measurements of the lumen and wall area as well as average wall thickness. Accuracy of airway measurements of each algorithm was also evaluated using an airway phantom. EI using a threshold of -950 HU was significantly different among the three algorithms in decreasing order of FBP (2.30 %), ASIR (1.49 %) and MBIR (1.20 %) (P < 0.01). Wall thickness was also significantly different among the three algorithms with FBP (2.09 mm) demonstrating thicker walls than ASIR (2.00 mm) and MBIR (1.88 mm) (P < 0.01). Airway phantom analysis revealed that MBIR showed the most accurate value for airway measurements. The three algorithms presented different EIs and wall thicknesses, decreasing in the order of FBP, ASIR and MBIR. Thus, care should be taken in selecting the appropriate IR algorithm on quantitative analysis of the lung. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Choo, Ji Yung [Seoul National University Medical Research Center, Department of Radiology, Seoul National University College of Medicine, and Institute of Radiation Medicine, Seoul (Korea, Republic of); Korea University Ansan Hospital, Ansan-si, Department of Radiology, Gyeonggi-do (Korea, Republic of); Goo, Jin Mo; Park, Chang Min; Park, Sang Joon [Seoul National University Medical Research Center, Department of Radiology, Seoul National University College of Medicine, and Institute of Radiation Medicine, Seoul (Korea, Republic of); Seoul National University, Cancer Research Institute, Seoul (Korea, Republic of); Lee, Chang Hyun; Shim, Mi-Suk [Seoul National University Medical Research Center, Department of Radiology, Seoul National University College of Medicine, and Institute of Radiation Medicine, Seoul (Korea, Republic of)
2014-04-15
To evaluate filtered back projection (FBP) and two iterative reconstruction (IR) algorithms and their effects on the quantitative analysis of lung parenchyma and airway measurements on computed tomography (CT) images. Low-dose chest CT obtained in 281 adult patients were reconstructed using three algorithms: FBP, adaptive statistical IR (ASIR) and model-based IR (MBIR). Measurements of each dataset were compared: total lung volume, emphysema index (EI), airway measurements of the lumen and wall area as well as average wall thickness. Accuracy of airway measurements of each algorithm was also evaluated using an airway phantom. EI using a threshold of -950 HU was significantly different among the three algorithms in decreasing order of FBP (2.30 %), ASIR (1.49 %) and MBIR (1.20 %) (P < 0.01). Wall thickness was also significantly different among the three algorithms with FBP (2.09 mm) demonstrating thicker walls than ASIR (2.00 mm) and MBIR (1.88 mm) (P < 0.01). Airway phantom analysis revealed that MBIR showed the most accurate value for airway measurements. The three algorithms presented different EIs and wall thicknesses, decreasing in the order of FBP, ASIR and MBIR. Thus, care should be taken in selecting the appropriate IR algorithm on quantitative analysis of the lung. (orig.)
International Nuclear Information System (INIS)
Tang Qiulin; Zeng, Gengsheng L; Gullberg, Grant T
2005-01-01
In this paper, we developed an analytical fan-beam reconstruction algorithm that compensates for uniform attenuation in SPECT. The new fan-beam algorithm is in the form of backprojection first, then filtering, and is mathematically exact. The algorithm is based on three components. The first one is the established generalized central-slice theorem, which relates the 1D Fourier transform of a set of arbitrary data and the 2D Fourier transform of the backprojected image. The second one is the fact that the backprojection of the fan-beam measurements is identical to the backprojection of the parallel measurements of the same object with the same attenuator. The third one is the stable analytical reconstruction algorithm for uniformly attenuated Radon data, developed by Metz and Pan. The fan-beam algorithm is then extended into a cone-beam reconstruction algorithm, where the orbit of the focal point of the cone-beam imaging geometry is a circle. This orbit geometry does not satisfy Tuy's condition and the obtained cone-beam algorithm is an approximation. In the cone-beam algorithm, the cone-beam data are first backprojected into the 3D image volume; then a slice-by-slice filtering is performed. This slice-by-slice filtering procedure is identical to that of the fan-beam algorithm. Both the fan-beam and cone-beam algorithms are efficient, and computer simulations are presented. The new cone-beam algorithm is compared with Bronnikov's cone-beam algorithm, and it is shown to have better performance with noisy projections
Zhang, Shang; Dong, Yuhan; Fu, Hongyan; Huang, Shao-Lun; Zhang, Lin
2018-02-22
The miniaturization of spectrometer can broaden the application area of spectrometry, which has huge academic and industrial value. Among various miniaturization approaches, filter-based miniaturization is a promising implementation by utilizing broadband filters with distinct transmission functions. Mathematically, filter-based spectral reconstruction can be modeled as solving a system of linear equations. In this paper, we propose an algorithm of spectral reconstruction based on sparse optimization and dictionary learning. To verify the feasibility of the reconstruction algorithm, we design and implement a simple prototype of a filter-based miniature spectrometer. The experimental results demonstrate that sparse optimization is well applicable to spectral reconstruction whether the spectra are directly sparse or not. As for the non-directly sparse spectra, their sparsity can be enhanced by dictionary learning. In conclusion, the proposed approach has a bright application prospect in fabricating a practical miniature spectrometer.
International Nuclear Information System (INIS)
Shen Le; Xing Yuxiang
2010-01-01
The derivative back-projection filtered algorithm for a helical cone-beam CT is a newly developed exact reconstruction method. Due to its large computational complexity, the reconstruction is rather slow for practical use. General purpose graphic processing unit (GPGPU) is an SIMD paralleled hardware architecture with powerful float-point operation capacity. In this paper,we propose a new method for PI-line choice and sampling grid, and a paralleled PI-line reconstruction algorithm implemented on NVIDIA's Compute Unified Device Architecture (CUDA). Numerical simulation studies are carried out to validate our method. Compared with conventional CPU implementation, the CUDA accelerated method provides images of the same quality with a speedup factor of 318. Optimization strategies for the GPU acceleration are presented. Finally, influence of the parameters of the PI-line samples on the reconstruction speed and image quality is discussed. (authors)
Zhang, Shang; Fu, Hongyan; Huang, Shao-Lun; Zhang, Lin
2018-01-01
The miniaturization of spectrometer can broaden the application area of spectrometry, which has huge academic and industrial value. Among various miniaturization approaches, filter-based miniaturization is a promising implementation by utilizing broadband filters with distinct transmission functions. Mathematically, filter-based spectral reconstruction can be modeled as solving a system of linear equations. In this paper, we propose an algorithm of spectral reconstruction based on sparse optimization and dictionary learning. To verify the feasibility of the reconstruction algorithm, we design and implement a simple prototype of a filter-based miniature spectrometer. The experimental results demonstrate that sparse optimization is well applicable to spectral reconstruction whether the spectra are directly sparse or not. As for the non-directly sparse spectra, their sparsity can be enhanced by dictionary learning. In conclusion, the proposed approach has a bright application prospect in fabricating a practical miniature spectrometer. PMID:29470406
An ART iterative reconstruction algorithm for computed tomography of diffraction enhanced imaging
International Nuclear Information System (INIS)
Wang Zhentian; Zhang Li; Huang Zhifeng; Kang Kejun; Chen Zhiqiang; Fang Qiaoguang; Zhu Peiping
2009-01-01
X-ray diffraction enhanced imaging (DEI) has extremely high sensitivity for weakly absorbing low-Z samples in medical and biological fields. In this paper, we propose an Algebra Reconstruction Technique (ART) iterative reconstruction algorithm for computed tomography of diffraction enhanced imaging (DEI-CT). An Ordered Subsets (OS) technique is used to accelerate the ART reconstruction. Few-view reconstruction is also studied, and a partial differential equation (PDE) type filter which has the ability of edge-preserving and denoising is used to improve the image quality and eliminate the artifacts. The proposed algorithm is validated with both the numerical simulations and the experiment at the Beijing synchrotron radiation facility (BSRF). (authors)
A stand-alone track reconstruction algorithm for the scintillating fibre tracker at the LHCb upgrade
Quagliani, Renato
2017-01-01
The LHCb upgrade detector project foresees the presence of a scintillating fiber tracker (SciFi) to be used during the LHC Run III, starting in 2020. The instantaneous luminosity will be increased up to $2\\times10^{33}$, five times larger than in Run II and a full software event reconstruction will be performed at the full bunch crossing rate by the trigger. The new running conditions, and the tighter timing constraints in the software trigger, represent a big challenge for track reconstruction. This poster presents the design and performance of a novel algorithm that has been developed to reconstruct track segments using solely hits from the SciFi. This algorithm is crucial for the reconstruction of tracks originating from long-lived particles such as $K_{S}^{0}$ and $\\Lambda$ and allows to greatly enhance the physics potential and capabilities of the LHCb upgrade when compared to its previous implementation.
The OMPS Limb Profiler Instrument: Two-Dimensional Retrieval Algorithm
Rault, Didier F.
2010-01-01
The upcoming Ozone Mapper and Profiler Suite (OMPS), which will be launched on the NPOESS Preparatory Project (NPP) platform in early 2011, will continue monitoring the global distribution of the Earth's middle atmosphere ozone and aerosol. OMPS is composed of three instruments, namely the Total Column Mapper (heritage: TOMS, OMI), the Nadir Profiler (heritage: SBUV) and the Limb Profiler (heritage: SOLSE/LORE, OSIRIS, SCIAMACHY, SAGE III). The ultimate goal of the mission is to better understand and quantify the rate of stratospheric ozone recovery. The focus of the paper will be on the Limb Profiler (LP) instrument. The LP instrument will measure the Earth's limb radiance (which is due to the scattering of solar photons by air molecules, aerosol and Earth surface) in the ultra-violet (UV), visible and near infrared, from 285 to 1000 nm. The LP simultaneously images the whole vertical extent of the Earth's limb through three vertical slits, each covering a vertical tangent height range of 100 km and each horizontally spaced by 250 km in the cross-track direction. Measurements are made every 19 seconds along the orbit track, which corresponds to a distance of about 150km. Several data analysis tools are presently being constructed and tested to retrieve ozone and aerosol vertical distribution from limb radiance measurements. The primary NASA algorithm is based on earlier algorithms developed for the SOLSE/LORE and SAGE III limb scatter missions. All the existing retrieval algorithms rely on a spherical symmetry assumption for the atmosphere structure. While this assumption is reasonable in most of the stratosphere, it is no longer valid in regions of prime scientific interest, such as polar vortex and UTLS regions. The paper will describe a two-dimensional retrieval algorithm whereby the ozone distribution is simultaneously retrieved vertically and horizontally for a whole orbit. The retrieval code relies on (1) a forward 2D Radiative Transfer code (to model limb
Goolsby, Eric W
2017-04-01
Ancestral state reconstruction is a method used to study the evolutionary trajectories of quantitative characters on phylogenies. Although efficient methods for univariate ancestral state reconstruction under a Brownian motion model have been described for at least 25 years, to date no generalization has been described to allow more complex evolutionary models, such as multivariate trait evolution, non-Brownian models, missing data, and within-species variation. Furthermore, even for simple univariate Brownian motion models, most phylogenetic comparative R packages compute ancestral states via inefficient tree rerooting and full tree traversals at each tree node, making ancestral state reconstruction extremely time-consuming for large phylogenies. Here, a computationally efficient method for fast maximum likelihood ancestral state reconstruction of continuous characters is described. The algorithm has linear complexity relative to the number of species and outperforms the fastest existing R implementations by several orders of magnitude. The described algorithm is capable of performing ancestral state reconstruction on a 1,000,000-species phylogeny in fewer than 2 s using a standard laptop, whereas the next fastest R implementation would take several days to complete. The method is generalizable to more complex evolutionary models, such as phylogenetic regression, within-species variation, non-Brownian evolutionary models, and multivariate trait evolution. Because this method enables fast repeated computations on phylogenies of virtually any size, implementation of the described algorithm can drastically alleviate the computational burden of many otherwise prohibitively time-consuming tasks requiring reconstruction of ancestral states, such as phylogenetic imputation of missing data, bootstrapping procedures, Expectation-Maximization algorithms, and Bayesian estimation. The described ancestral state reconstruction algorithm is implemented in the Rphylopars
Application aspects of advanced antenna diagnostics with the 3D reconstruction algorithm
DEFF Research Database (Denmark)
Cappellin, Cecilia; Pivnenko, Sergey
2015-01-01
This paper focuses on two important applications of the 3D reconstruction algorithm of the commercial software DIATOOL for antenna diagnostics. The first one is the accurate and detailed identification of array malfunctioning, thanks to the available enhanced spatial resolution of the reconstruct...... fields and currents. The second one is the filtering of the scattering from support structures and feed network leakage. Representative experimental results are presented and guidelines on the recommended measurement parameters for obtaining the best diagnostics results are provided....
Mariano-Goulart, D; Fourcade, M; Bernon, J L; Rossi, M; Zanca, M
2003-01-01
Thanks to an experimental study based on simulated and physical phantoms, the propagation of the stochastic noise in slices reconstructed using the conjugate gradient algorithm has been analysed versus iterations. After a first increase corresponding to the reconstruction of the signal, the noise stabilises before increasing linearly with iterations. The level of the plateau as well as the slope of the subsequent linear increase depends on the noise in the projection data.
International Nuclear Information System (INIS)
Jeon, Min-Gyu; Deguchi, Yoshihiro; Kamimoto, Takahiro; Doh, Deog-Hee; Cho, Gyeong-Rae
2017-01-01
Highlights: • The measured data were successfully used for generating absorption spectra. • Four different reconstruction algorithms, ART, MART, SART and SMART were evaluated. • The calculation speed of convergence by the SMART algorithm was the fastest. • SMART was the most reliable algorithm for reconstructing the multiple signals. - Abstract: Recent advent of the tunable lasers made to measure simultaneous temperature and concentration fields of the gases. CT-TDLAS (computed tomography-tunable diode laser absorption spectroscopy) is one the leading techniques for the measurements of temperature and concentration fields of the gases. In CT-TDLAS, the accuracies of the measurement results are strongly dependent upon the reconstruction algorithms. In this study, four different reconstruction algorithms have been tested numerically using experimental data sets measured by thermocouples for combustion fields. Three reconstruction algorithms, MART (multiplicative algebraic reconstruction technique) algorithm, SART (simultaneous algebraic reconstruction technique) algorithm and SMART (simultaneous multiplicative algebraic reconstruction technique) algorithm, are newly proposed for CT-TDLAS in this study. The calculation results obtained by the three algorithms have been compared with previous algorithm, ART (algebraic reconstruction technique) algorithm. Phantom data sets have been generated by the use of thermocouples data obtained in an actual experiment. The data of the Harvard HITRAN table in which the thermo-dynamical properties and the light spectrum of the H_2O are listed were used for the numerical test. The reconstructed temperature and concentration fields were compared with the original HITRAN data, through which the constructed methods are validated. The performances of the four reconstruction algorithms were demonstrated. This method is expected to enhance the practicality of CT-TDLAS.
A combination-weighted Feldkamp-based reconstruction algorithm for cone-beam CT
International Nuclear Information System (INIS)
Mori, Shinichiro; Endo, Masahiro; Komatsu, Shuhei; Kandatsu, Susumu; Yashiro, Tomoyasu; Baba, Masayuki
2006-01-01
The combination-weighted Feldkamp algorithm (CW-FDK) was developed and tested in a phantom in order to reduce cone-beam artefacts and enhance cranio-caudal reconstruction coverage in an attempt to improve image quality when utilizing cone-beam computed tomography (CBCT). Using a 256-slice cone-beam CT (256CBCT), image quality (CT-number uniformity and geometrical accuracy) was quantitatively evaluated in phantom and clinical studies, and the results were compared to those obtained with the original Feldkamp algorithm. A clinical study was done in lung cancer patients under breath holding and free breathing. Image quality for the original Feldkamp algorithm is degraded at the edge of the scan region due to the missing volume, commensurate with the cranio-caudal distance between the reconstruction and central planes. The CW-FDK extended the reconstruction coverage to equal the scan coverage and improved reconstruction accuracy, unaffected by the cranio-caudal distance. The extended reconstruction coverage with good image quality provided by the CW-FDK will be clinically investigated for improving diagnostic and radiotherapy applications. In addition, this algorithm can also be adapted for use in relatively wide cone-angle CBCT such as with a flat-panel detector CBCT
International Nuclear Information System (INIS)
Vaegler, Sven; Sauer, Otto; Stsepankou, Dzmitry; Hesser, Juergen
2015-01-01
The reduction of dose in cone beam computer tomography (CBCT) arises from the decrease of the tube current for each projection as well as from the reduction of the number of projections. In order to maintain good image quality, sophisticated image reconstruction techniques are required. The Prior Image Constrained Compressed Sensing (PICCS) incorporates prior images into the reconstruction algorithm and outperforms the widespread used Feldkamp-Davis-Kress-algorithm (FDK) when the number of projections is reduced. However, prior images that contain major variations are not appropriately considered so far in PICCS. We therefore propose the partial-PICCS (pPICCS) algorithm. This framework is a problem-specific extension of PICCS and enables the incorporation of the reliability of the prior images additionally. We assumed that the prior images are composed of areas with large and small deviations. Accordingly, a weighting matrix considered the assigned areas in the objective function. We applied our algorithm to the problem of image reconstruction from few views by simulations with a computer phantom as well as on clinical CBCT projections from a head-and-neck case. All prior images contained large local variations. The reconstructed images were compared to the reconstruction results by the FDK-algorithm, by Compressed Sensing (CS) and by PICCS. To show the gain of image quality we compared image details with the reference image and used quantitative metrics (root-mean-square error (RMSE), contrast-to-noise-ratio (CNR)). The pPICCS reconstruction framework yield images with substantially improved quality even when the number of projections was very small. The images contained less streaking, blurring and inaccurately reconstructed structures compared to the images reconstructed by FDK, CS and conventional PICCS. The increased image quality is also reflected in large RMSE differences. We proposed a modification of the original PICCS algorithm. The pPICCS algorithm
Toadere, Florin
2017-12-01
A spectral image processing algorithm that allows the illumination of the scene with different illuminants together with the reconstruction of the scene's reflectance is presented. Color checker spectral image and CIE A (warm light 2700 K), D65 (cold light 6500 K) and Cree TW Series LED T8 (4000 K) are employed for scene illumination. Illuminants used in the simulations have different spectra and, as a result of their illumination, the colors of the scene change. The influence of the illuminants on the reconstruction of the scene's reflectance is estimated. Demonstrative images and reflectance showing the operation of the algorithm are illustrated.
International Nuclear Information System (INIS)
Zhang Jin; Shi Daxin; Anastasio, Mark A; Sillanpaa, Jussi; Chang Jenghwa
2005-01-01
We propose and investigate weighted expectation maximization (EM) algorithms for image reconstruction in x-ray tomography. The development of the algorithms is motivated by the respiratory-gated megavoltage tomography problem, in which the acquired asymmetric cone-beam projections are limited in number and unevenly sampled over view angle. In these cases, images reconstructed by use of the conventional EM algorithm can contain ring- and streak-like artefacts that are attributable to a combination of data inconsistencies and truncation of the projection data. By use of computer-simulated and clinical gated fan-beam megavoltage projection data, we demonstrate that the proposed weighted EM algorithms effectively mitigate such image artefacts. (note)
Ogawa, Takahiro; Haseyama, Miki
2013-03-01
A missing texture reconstruction method based on an error reduction (ER) algorithm, including a novel estimation scheme of Fourier transform magnitudes is presented in this brief. In our method, Fourier transform magnitude is estimated for a target patch including missing areas, and the missing intensities are estimated by retrieving its phase based on the ER algorithm. Specifically, by monitoring errors converged in the ER algorithm, known patches whose Fourier transform magnitudes are similar to that of the target patch are selected from the target image. In the second approach, the Fourier transform magnitude of the target patch is estimated from those of the selected known patches and their corresponding errors. Consequently, by using the ER algorithm, we can estimate both the Fourier transform magnitudes and phases to reconstruct the missing areas.
Energy Technology Data Exchange (ETDEWEB)
Narasimhadhan, A.V.; Rajgopal, Kasi
2011-07-01
This paper presents a new hybrid filtered backprojection (FBP) algorithm for fan-beam and cone-beam scan. The hybrid reconstruction kernel is the sum of the ramp and Hilbert filters. We modify the redundancy weighting function to reduce the inverse square distance weighting in the backprojection to inverse distance weight. The modified weight also eliminates the derivative associated with the Hilbert filter kernel. Thus, the proposed reconstruction algorithm has the advantages of the inverse distance weight in the backprojection. We evaluate the performance of the new algorithm in terms of the magnitude level and uniformity in noise for the fan-beam geometry. The computer simulations show that the spatial resolution is nearly identical to the standard fan-beam ramp filtered algorithm while the noise is spatially uniform and the noise variance is reduced. (orig.)
Optimizing SRF Gun Cavity Profiles in a Genetic Algorithm Framework
International Nuclear Information System (INIS)
Hofler, Alicia; Evtushenko, Pavel; Marhauser, Frank
2009-01-01
Automation of DC photoinjector designs using a genetic algorithm (GA) based optimization is an accepted practice in accelerator physics. Allowing the gun cavity field profile shape to be varied can extend the utility of this optimization methodology to superconducting and normal conducting radio frequency (SRF/RF) gun based injectors. Finding optimal field and cavity geometry configurations can provide guidance for cavity design choices and verify existing designs. We have considered two approaches for varying the electric field profile. The first is to determine the optimal field profile shape that should be used independent of the cavity geometry, and the other is to vary the geometry of the gun cavity structure to produce an optimal field profile. The first method can provide a theoretical optimal and can illuminate where possible gains can be made in field shaping. The second method can produce more realistically achievable designs that can be compared to existing designs. In this paper, we discuss the design and implementation for these two methods for generating field profiles for SRF/RF guns in a GA based injector optimization scheme and provide preliminary results.
Energy Technology Data Exchange (ETDEWEB)
Jimenez, Edward S. [Sandia National Lab. (SNL-NM), Albuquerque, NM (United States); Orr, Laurel J. [Sandia National Lab. (SNL-NM), Albuquerque, NM (United States); Thompson, Kyle R. [Sandia National Lab. (SNL-NM), Albuquerque, NM (United States)
2013-09-01
The goal of this work is to develop a fast computed tomography (CT) reconstruction algorithm based on graphics processing units (GPU) that achieves significant improvement over traditional central processing unit (CPU) based implementations. The main challenge in developing a CT algorithm that is capable of handling very large datasets is parallelizing the algorithm in such a way that data transfer does not hinder performance of the reconstruction algorithm. General Purpose Graphics Processing (GPGPU) is a new technology that the Science and Technology (S&T) community is starting to adopt in many fields where CPU-based computing is the norm. GPGPU programming requires a new approach to algorithm development that utilizes massively multi-threaded environments. Multi-threaded algorithms in general are difficult to optimize since performance bottlenecks occur that are non-existent in single-threaded algorithms such as memory latencies. If an efficient GPU-based CT reconstruction algorithm can be developed; computational times could be improved by a factor of 20. Additionally, cost benefits will be realized as commodity graphics hardware could potentially replace expensive supercomputers and high-end workstations. This project will take advantage of the CUDA programming environment and attempt to parallelize the task in such a way that multiple slices of the reconstruction volume are computed simultaneously. This work will also take advantage of the GPU memory by utilizing asynchronous memory transfers, GPU texture memory, and (when possible) pinned host memory so that the memory transfer bottleneck inherent to GPGPU is amortized. Additionally, this work will take advantage of GPU-specific hardware (i.e. fast texture memory, pixel-pipelines, hardware interpolators, and varying memory hierarchy) that will allow for additional performance improvements.
On-line reconstruction algorithms for the CBM and ALICE experiments
International Nuclear Information System (INIS)
Gorbunov, Sergey
2013-01-01
This thesis presents various algorithms which have been developed for on-line event reconstruction in the CBM experiment at GSI, Darmstadt and the ALICE experiment at CERN, Geneve. Despite the fact that the experiments are different - CBM is a fixed target experiment with forward geometry, while ALICE has a typical collider geometry - they share common aspects when reconstruction is concerned. The thesis describes: - general modifications to the Kalman filter method, which allows one to accelerate, to improve, and to simplify existing fit algorithms; - developed algorithms for track fit in CBM and ALICE experiment, including a new method for track extrapolation in non-homogeneous magnetic field. - developed algorithms for primary and secondary vertex fit in the both experiments. In particular, a new method of reconstruction of decayed particles is presented. - developed parallel algorithm for the on-line tracking in the CBM experiment. - developed parallel algorithm for the on-line tracking in High Level Trigger of the ALICE experiment. - the realisation of the track finders on modern hardware, such as SIMD CPU registers and GPU accelerators. All the presented methods have been developed by or with the direct participation of the author.
Directory of Open Access Journals (Sweden)
Chae Young Lee
Full Text Available The purposes of this study were to optimize a proton computed tomography system (pCT for proton range verification and to confirm the pCT image reconstruction algorithm based on projection images generated with optimized parameters. For this purpose, we developed a new pCT scanner using the Geometry and Tracking (GEANT 4.9.6 simulation toolkit. GEANT4 simulations were performed to optimize the geometric parameters representing the detector thickness and the distance between the detectors for pCT. The system consisted of four silicon strip detectors for particle tracking and a calorimeter to measure the residual energies of the individual protons. The optimized pCT system design was then adjusted to ensure that the solution to a CS-based convex optimization problem would converge to yield the desired pCT images after a reasonable number of iterative corrections. In particular, we used a total variation-based formulation that has been useful in exploiting prior knowledge about the minimal variations of proton attenuation characteristics in the human body. Examinations performed using our CS algorithm showed that high-quality pCT images could be reconstructed using sets of 72 projections within 20 iterations and without any streaks or noise, which can be caused by under-sampling and proton starvation. Moreover, the images yielded by this CS algorithm were found to be of higher quality than those obtained using other reconstruction algorithms. The optimized pCT scanner system demonstrated the potential to perform high-quality pCT during on-line image-guided proton therapy, without increasing the imaging dose, by applying our CS based proton CT reconstruction algorithm. Further, we make our optimized detector system and CS-based proton CT reconstruction algorithm potentially useful in on-line proton therapy.
International Nuclear Information System (INIS)
Nunes, F.; Varela, P.; Silva, A.; Manso, M.; Santos, J.; Nunes, I.; Serra, F.; Kurzan, B.; Suttrop, W.
1997-01-01
Broadband reflectometry is a current technique that uses the round-trip group delays of reflected frequency-swept waves to measure density profiles of fusion plasmas. The main factor that may limit the accuracy of the reconstructed profiles is the interference of the probing waves with the plasma density fluctuations: plasma turbulence leads to random phase variations and magneto hydrodynamic activity produces mainly strong amplitude and phase modulations. Both effects cause the decrease, and eventually loss, of signal at some frequencies. Several data processing techniques can be applied to filter and/or interpolate noisy group delay data obtained from turbulent plasmas with a single frequency sweep. Here, we propose a more powerful algorithm performing two-dimensional regularization (in space and time) of data provided by multiple consecutive frequency sweeps, which leads to density profiles with improved accuracy. The new method is described and its application to simulated data corrupted by noise and missing data is considered. It is shown that the algorithm improves the identification of slowly varying plasma density perturbations by attenuating the effect of fast fluctuations and noise contained in experimental data. First results obtained with this method in ASDEX Upgrade tokamak are presented. copyright 1997 American Institute of Physics
Performance Comparison of Reconstruction Algorithms in Discrete Blind Multi-Coset Sampling
DEFF Research Database (Denmark)
Grigoryan, Ruben; Arildsen, Thomas; Tandur, Deepaknath
2012-01-01
This paper investigates the performance of different reconstruction algorithms in discrete blind multi-coset sampling. Multi-coset scheme is a promising compressed sensing architecture that can replace traditional Nyquist-rate sampling in the applications with multi-band frequency sparse signals...
Directory of Open Access Journals (Sweden)
Changgan SHU
2014-09-01
Full Text Available In the standard root multiple signal classification algorithm, the performance of direction of arrival estimation will reduce and even lose effect in circumstances that a low signal noise ratio and a small signals interval. By reconstructing and weighting the covariance matrix of received signal, the modified algorithm can provide more accurate estimation results. The computer simulation and performance analysis are given next, which show that under the condition of lower signal noise ratio and stronger correlation between signals, the proposed modified algorithm could provide preferable azimuth estimating performance than the standard method.
Lartillot, Nicolas; Rodrigue, Nicolas; Stubbs, Daniel; Richer, Jacques
2013-07-01
Modeling across site variation of the substitution process is increasingly recognized as important for obtaining more accurate phylogenetic reconstructions. Both finite and infinite mixture models have been proposed and have been shown to significantly improve on classical single-matrix models. Compared with their finite counterparts, infinite mixtures have a greater expressivity. However, they are computationally more challenging. This has resulted in practical compromises in the design of infinite mixture models. In particular, a fast but simplified version of a Dirichlet process model over equilibrium frequency profiles implemented in PhyloBayes has often been used in recent phylogenomics studies, while more refined model structures, more realistic and empirically more fit, have been practically out of reach. We introduce a message passing interface version of PhyloBayes, implementing the Dirichlet process mixture models as well as more classical empirical matrices and finite mixtures. The parallelization is made efficient thanks to the combination of two algorithmic strategies: a partial Gibbs sampling update of the tree topology and the use of a truncated stick-breaking representation for the Dirichlet process prior. The implementation shows close to linear gains in computational speed for up to 64 cores, thus allowing faster phylogenetic reconstruction under complex mixture models. PhyloBayes MPI is freely available from our website www.phylobayes.org.
International Nuclear Information System (INIS)
Xu, Qiaofeng; Sawatzky, Alex; Anastasio, Mark A.; Yang, Deshan; Tan, Jun
2016-01-01
Purpose: The development of iterative image reconstruction algorithms for cone-beam computed tomography (CBCT) remains an active and important research area. Even with hardware acceleration, the overwhelming majority of the available 3D iterative algorithms that implement nonsmooth regularizers remain computationally burdensome and have not been translated for routine use in time-sensitive applications such as image-guided radiation therapy (IGRT). In this work, two variants of the fast iterative shrinkage thresholding algorithm (FISTA) are proposed and investigated for accelerated iterative image reconstruction in CBCT. Methods: Algorithm acceleration was achieved by replacing the original gradient-descent step in the FISTAs by a subproblem that is solved by use of the ordered subset simultaneous algebraic reconstruction technique (OS-SART). Due to the preconditioning matrix adopted in the OS-SART method, two new weighted proximal problems were introduced and corresponding fast gradient projection-type algorithms were developed for solving them. We also provided efficient numerical implementations of the proposed algorithms that exploit the massive data parallelism of multiple graphics processing units. Results: The improved rates of convergence of the proposed algorithms were quantified in computer-simulation studies and by use of clinical projection data corresponding to an IGRT study. The accelerated FISTAs were shown to possess dramatically improved convergence properties as compared to the standard FISTAs. For example, the number of iterations to achieve a specified reconstruction error could be reduced by an order of magnitude. Volumetric images reconstructed from clinical data were produced in under 4 min. Conclusions: The FISTA achieves a quadratic convergence rate and can therefore potentially reduce the number of iterations required to produce an image of a specified image quality as compared to first-order methods. We have proposed and investigated
Xu, Qiaofeng; Yang, Deshan; Tan, Jun; Sawatzky, Alex; Anastasio, Mark A.
2016-01-01
Purpose: The development of iterative image reconstruction algorithms for cone-beam computed tomography (CBCT) remains an active and important research area. Even with hardware acceleration, the overwhelming majority of the available 3D iterative algorithms that implement nonsmooth regularizers remain computationally burdensome and have not been translated for routine use in time-sensitive applications such as image-guided radiation therapy (IGRT). In this work, two variants of the fast iterative shrinkage thresholding algorithm (FISTA) are proposed and investigated for accelerated iterative image reconstruction in CBCT. Methods: Algorithm acceleration was achieved by replacing the original gradient-descent step in the FISTAs by a subproblem that is solved by use of the ordered subset simultaneous algebraic reconstruction technique (OS-SART). Due to the preconditioning matrix adopted in the OS-SART method, two new weighted proximal problems were introduced and corresponding fast gradient projection-type algorithms were developed for solving them. We also provided efficient numerical implementations of the proposed algorithms that exploit the massive data parallelism of multiple graphics processing units. Results: The improved rates of convergence of the proposed algorithms were quantified in computer-simulation studies and by use of clinical projection data corresponding to an IGRT study. The accelerated FISTAs were shown to possess dramatically improved convergence properties as compared to the standard FISTAs. For example, the number of iterations to achieve a specified reconstruction error could be reduced by an order of magnitude. Volumetric images reconstructed from clinical data were produced in under 4 min. Conclusions: The FISTA achieves a quadratic convergence rate and can therefore potentially reduce the number of iterations required to produce an image of a specified image quality as compared to first-order methods. We have proposed and investigated
High-speed computation of the EM algorithm for PET image reconstruction
International Nuclear Information System (INIS)
Rajan, K.; Patnaik, L.M.; Ramakrishna, J.
1994-01-01
The PET image reconstruction based on the EM algorithm has several attractive advantages over the conventional convolution backprojection algorithms. However, two major drawbacks have impeded the routine use of the EM algorithm, namely, the long computational time due to slow convergence and the large memory required for the storage of the image, projection data and the probability matrix. In this study, the authors attempts to solve these two problems by parallelizing the EM algorithm on a multiprocessor system. The authors have implemented an extended hypercube (EH) architecture for the high-speed computation of the EM algorithm using the commercially available fast floating point digital signal processor (DSP) chips as the processing elements (PEs). The authors discuss and compare the performance of the EM algorithm on a 386/387 machine, CD 4360 mainframe, and on the EH system. The results show that the computational speed performance of an EH using DSP chips as PEs executing the EM image reconstruction algorithm is about 130 times better than that of the CD 4360 mainframe. The EH topology is expandable with more number of PEs
Microstructure Reconstruction of Sheet Molding Composite Using a Random Chips Packing Algorithm
Energy Technology Data Exchange (ETDEWEB)
Huang, Tianyu; Xu, Hongyi; Chen, Wei
2017-04-06
Fiber-reinforced polymer composites are strong candidates for structural materials to replace steel and light alloys in lightweight vehicle design because of their low density and relatively high strength. In the integrated computational materials engineering (ICME) development of carbon fiber composites, microstructure reconstruction algorithms are needed to generate material microstructure representative volume element (RVE) based on the material processing information. The microstructure RVE reconstruction enables the material property prediction by finite element analysis (FEA)This paper presents an algorithm to reconstruct the microstructure of a chopped carbon fiber/epoxy laminate material system produced by compression molding, normally known as sheet molding compounds (SMC). The algorithm takes the result from material’s manufacturing process as inputs, such as the orientation tensor of fibers, the chopped fiber sheet geometry, and the fiber volume fraction. The chopped fiber sheets are treated as deformable rectangle chips and a random packing algorithm is developed to pack these chips into a square plate. The RVE is built in a layer-by-layer fashion until the desired number of lamina is reached, then a fine tuning process is applied to finalize the reconstruction. Compared to the previous methods, this new approach has the ability to model bended fibers by allowing limited amount of overlaps of rectangle chips. Furthermore, the method does not need SMC microstructure images, for which the image-based characterization techniques have not been mature enough, as inputs. Case studies are performed and the results show that the statistics of the reconstructed microstructures generated by the algorithm matches well with the target input parameters from processing.
Accurate 3D reconstruction by a new PDS-OSEM algorithm for HRRT
Chen, Tai-Been; Horng-Shing Lu, Henry; Kim, Hang-Keun; Son, Young-Don; Cho, Zang-Hee
2014-03-01
State-of-the-art high resolution research tomography (HRRT) provides high resolution PET images with full 3D human brain scanning. But, a short time frame in dynamic study causes many problems related to the low counts in the acquired data. The PDS-OSEM algorithm was proposed to reconstruct the HRRT image with a high signal-to-noise ratio that provides accurate information for dynamic data. The new algorithm was evaluated by simulated image, empirical phantoms, and real human brain data. Meanwhile, the time activity curve was adopted to validate a reconstructed performance of dynamic data between PDS-OSEM and OP-OSEM algorithms. According to simulated and empirical studies, the PDS-OSEM algorithm reconstructs images with higher quality, higher accuracy, less noise, and less average sum of square error than those of OP-OSEM. The presented algorithm is useful to provide quality images under the condition of low count rates in dynamic studies with a short scan time.
International Nuclear Information System (INIS)
Sidky, Emil Y.; Pan Xiaochuan; Reiser, Ingrid S.; Nishikawa, Robert M.; Moore, Richard H.; Kopans, Daniel B.
2009-01-01
Purpose: The authors develop a practical, iterative algorithm for image-reconstruction in undersampled tomographic systems, such as digital breast tomosynthesis (DBT). Methods: The algorithm controls image regularity by minimizing the image total p variation (TpV), a function that reduces to the total variation when p=1.0 or the image roughness when p=2.0. Constraints on the image, such as image positivity and estimated projection-data tolerance, are enforced by projection onto convex sets. The fact that the tomographic system is undersampled translates to the mathematical property that many widely varied resultant volumes may correspond to a given data tolerance. Thus the application of image regularity serves two purposes: (1) Reduction in the number of resultant volumes out of those allowed by fixing the data tolerance, finding the minimum image TpV for fixed data tolerance, and (2) traditional regularization, sacrificing data fidelity for higher image regularity. The present algorithm allows for this dual role of image regularity in undersampled tomography. Results: The proposed image-reconstruction algorithm is applied to three clinical DBT data sets. The DBT cases include one with microcalcifications and two with masses. Conclusions: Results indicate that there may be a substantial advantage in using the present image-reconstruction algorithm for microcalcification imaging.
International Nuclear Information System (INIS)
Li Jing; Sun Yi; Zhu Peiping
2013-01-01
Differential phase-contrast computed tomography (DPC-CT) reconstruction problems are usually solved by using parallel-, fan- or cone-beam algorithms. For rod-shaped objects, the x-ray beams cannot recover all the slices of the sample at the same time. Thus, if a rod-shaped sample is required to be reconstructed by the above algorithms, one should alternately perform translation and rotation on this sample, which leads to lower efficiency. The helical cone-beam CT may significantly improve scanning efficiency for rod-shaped objects over other algorithms. In this paper, we propose a theoretically exact filter-backprojection algorithm for helical cone-beam DPC-CT, which can be applied to reconstruct the refractive index decrement distribution of the samples directly from two-dimensional differential phase-contrast images. Numerical simulations are conducted to verify the proposed algorithm. Our work provides a potential solution for inspecting the rod-shaped samples using DPC-CT, which may be applicable with the evolution of DPC-CT equipments. (paper)
Statistical image reconstruction for transmission tomography using relaxed ordered subset algorithms
International Nuclear Information System (INIS)
Kole, J S
2005-01-01
Statistical reconstruction methods offer possibilities for improving image quality as compared to analytical methods, but current reconstruction times prohibit routine clinical applications in x-ray computed tomography (CT). To reduce reconstruction times, we have applied (under) relaxation to ordered subset algorithms. This enables us to use subsets consisting of only single projection angle, effectively increasing the number of image updates within an entire iteration. A second advantage of applying relaxation is that it can help improve convergence by removing the limit cycle behaviour of ordered subset algorithms, which normally do not converge to an optimal solution but rather a suboptimal limit cycle consisting of as many points as there are subsets. Relaxation suppresses the limit cycle behaviour by decreasing the stepsize for approaching the solution. A simulation study for a 2D mathematical phantom and three different ordered subset algorithms shows that all three algorithms benefit from relaxation: equal noise-to-resolution trade-off can be achieved using fewer iterations than the conventional algorithms, while a lower minimal normalized mean square error (NMSE) clearly indicates a better convergence. Two different schemes for setting the relaxation parameter are studied, and both schemes yield approximately the same minimal NMSE
Implementation of a virtual laryngoscope system using efficient reconstruction algorithms.
Luo, Shouhua; Yan, Yuling
2009-08-01
Conventional fiberoptic laryngoscope may cause discomfort to the patient and in some cases it can lead to side effects that include perforation, infection and hemorrhage. Virtual laryngoscopy (VL) can overcome this problem and further it may lower the risk of operation failures. Very few virtual endoscope (VE) based investigations of the larynx have been described in the literature. CT data sets from a healthy subject were used for the VL studies. An algorithm of preprocessing and region-growing for 3-D image segmentation is developed. An octree based approach is applied in our VL system which facilitates a rapid construction of iso-surfaces. Some locating techniques are used for fast rendering and navigation (fly-through). Our VL visualization system provides for real time and efficient 'fly-through' navigation. The virtual camera can be arranged so that it moves along the airway in either direction. Snap shots were taken during fly-throughs. The system can automatically adjust the direction of the virtual camera and prevent collisions of the camera and the wall of the airway. A virtual laryngoscope (VL) system using OpenGL (Open Graphics Library) platform for interactive rendering and 3D visualization of the laryngeal framework and upper airway is established. OpenGL is supported on major operating systems and works with every major windowing system. The VL system runs on regular PC workstations and was successfully tested and evaluated using CT data from a normal subject.
Xia, Y.; Tian, J.; d'Angelo, P.; Reinartz, P.
2018-05-01
3D reconstruction of plants is hard to implement, as the complex leaf distribution highly increases the difficulty level in dense matching. Semi-Global Matching has been successfully applied to recover the depth information of a scene, but may perform variably when different matching cost algorithms are used. In this paper two matching cost computation algorithms, Census transform and an algorithm using a convolutional neural network, are tested for plant reconstruction based on Semi-Global Matching. High resolution close-range photogrammetric images from a handheld camera are used for the experiment. The disparity maps generated based on the two selected matching cost methods are comparable with acceptable quality, which shows the good performance of Census and the potential of neural networks to improve the dense matching.
Directory of Open Access Journals (Sweden)
Y. Xia
2018-05-01
Full Text Available 3D reconstruction of plants is hard to implement, as the complex leaf distribution highly increases the difficulty level in dense matching. Semi-Global Matching has been successfully applied to recover the depth information of a scene, but may perform variably when different matching cost algorithms are used. In this paper two matching cost computation algorithms, Census transform and an algorithm using a convolutional neural network, are tested for plant reconstruction based on Semi-Global Matching. High resolution close-range photogrammetric images from a handheld camera are used for the experiment. The disparity maps generated based on the two selected matching cost methods are comparable with acceptable quality, which shows the good performance of Census and the potential of neural networks to improve the dense matching.
Signal process and profile reconstruction of stress corrosion crack by eddy current test
International Nuclear Information System (INIS)
Zhang Siquan; Chen Tiequn; Liu Guixiong
2008-01-01
The reconstruction of crack profiles is very important in the NDE (nondestructive evaluation) of critical structures, such as pressure vessel and tubes in heat exchangers. First a wavelet transform signal processing technique is used to reduce noise and other non-defect signals from the signals of crack, and then based on an artificial neural network method, the crack profiles are reconstructed. Although the results reveal that this method is with many advantages such as a short CPU time and precision for reconstruction,it does have some drawbacks, for example, the database generation and network training is a much time consuming work. Moreover, this approach does not expressly reconstruct the distribution of conductivity inside a crack, so the reliability of a reconstructed crack shape is unknown. But in practical application, if we do not consider the multiple cracks, this method can be used to reconstruct the natural crack. (authors)
A Compressed Sensing-based Image Reconstruction Algorithm for Solar Flare X-Ray Observations
Energy Technology Data Exchange (ETDEWEB)
Felix, Simon; Bolzern, Roman; Battaglia, Marina, E-mail: simon.felix@fhnw.ch, E-mail: roman.bolzern@fhnw.ch, E-mail: marina.battaglia@fhnw.ch [University of Applied Sciences and Arts Northwestern Switzerland FHNW, 5210 Windisch (Switzerland)
2017-11-01
One way of imaging X-ray emission from solar flares is to measure Fourier components of the spatial X-ray source distribution. We present a new compressed sensing-based algorithm named VIS-CS, which reconstructs the spatial distribution from such Fourier components. We demonstrate the application of the algorithm on synthetic and observed solar flare X-ray data from the Reuven Ramaty High Energy Solar Spectroscopic Imager satellite and compare its performance with existing algorithms. VIS-CS produces competitive results with accurate photometry and morphology, without requiring any algorithm- and X-ray-source-specific parameter tuning. Its robustness and performance make this algorithm ideally suited for the generation of quicklook images or large image cubes without user intervention, such as for imaging spectroscopy analysis.
A Compressed Sensing-based Image Reconstruction Algorithm for Solar Flare X-Ray Observations
Felix, Simon; Bolzern, Roman; Battaglia, Marina
2017-11-01
One way of imaging X-ray emission from solar flares is to measure Fourier components of the spatial X-ray source distribution. We present a new compressed sensing-based algorithm named VIS_CS, which reconstructs the spatial distribution from such Fourier components. We demonstrate the application of the algorithm on synthetic and observed solar flare X-ray data from the Reuven Ramaty High Energy Solar Spectroscopic Imager satellite and compare its performance with existing algorithms. VIS_CS produces competitive results with accurate photometry and morphology, without requiring any algorithm- and X-ray-source-specific parameter tuning. Its robustness and performance make this algorithm ideally suited for the generation of quicklook images or large image cubes without user intervention, such as for imaging spectroscopy analysis.
Lohvithee, Manasavee; Biguri, Ander; Soleimani, Manuchehr
2017-12-01
There are a number of powerful total variation (TV) regularization methods that have great promise in limited data cone-beam CT reconstruction with an enhancement of image quality. These promising TV methods require careful selection of the image reconstruction parameters, for which there are no well-established criteria. This paper presents a comprehensive evaluation of parameter selection in a number of major TV-based reconstruction algorithms. An appropriate way of selecting the values for each individual parameter has been suggested. Finally, a new adaptive-weighted projection-controlled steepest descent (AwPCSD) algorithm is presented, which implements the edge-preserving function for CBCT reconstruction with limited data. The proposed algorithm shows significant robustness compared to three other existing algorithms: ASD-POCS, AwASD-POCS and PCSD. The proposed AwPCSD algorithm is able to preserve the edges of the reconstructed images better with fewer sensitive parameters to tune.
High resolution reconstruction of PET images using the iterative OSEM algorithm
International Nuclear Information System (INIS)
Doll, J.; Bublitz, O.; Werling, A.; Haberkorn, U.; Semmler, W.; Adam, L.E.; Pennsylvania Univ., Philadelphia, PA; Brix, G.
2004-01-01
Aim: Improvement of the spatial resolution in positron emission tomography (PET) by incorporation of the image-forming characteristics of the scanner into the process of iterative image reconstruction. Methods: All measurements were performed at the whole-body PET system ECAT EXACT HR + in 3D mode. The acquired 3D sinograms were sorted into 2D sinograms by means of the Fourier rebinning (FORE) algorithm, which allows the usage of 2D algorithms for image reconstruction. The scanner characteristics were described by a spatially variant line-spread function (LSF), which was determined from activated copper-64 line sources. This information was used to model the physical degradation processes in PET measurements during the course of 2D image reconstruction with the iterative OSEM algorithm. To assess the performance of the high-resolution OSEM algorithm, phantom measurements performed at a cylinder phantom, the hotspot Jaszczack phantom, and the 3D Hoffmann brain phantom as well as different patient examinations were analyzed. Results: Scanner characteristics could be described by a Gaussian-shaped LSF with a full-width at half-maximum increasing from 4.8 mm at the center to 5.5 mm at a radial distance of 10.5 cm. Incorporation of the LSF into the iteration formula resulted in a markedly improved resolution of 3.0 and 3.5 mm, respectively. The evaluation of phantom and patient studies showed that the high-resolution OSEM algorithm not only lead to a better contrast resolution in the reconstructed activity distributions but also to an improved accuracy in the quantification of activity concentrations in small structures without leading to an amplification of image noise or even the occurrence of image artifacts. Conclusion: The spatial and contrast resolution of PET scans can markedly be improved by the presented image restauration algorithm, which is of special interest for the examination of both patients with brain disorders and small animals. (orig.)
Rodriguez-Ruiz, Alejandro; Teuwen, Jonas; Vreemann, Suzan; Bouwman, Ramona W; van Engen, Ruben E; Karssemeijer, Nico; Mann, Ritse M; Gubern-Merida, Albert; Sechopoulos, Ioannis
2017-01-01
Background The image quality of digital breast tomosynthesis (DBT) volumes depends greatly on the reconstruction algorithm. Purpose To compare two DBT reconstruction algorithms used by the Siemens Mammomat Inspiration system, filtered back projection (FBP), and FBP with iterative optimizations (EMPIRE), using qualitative analysis by human readers and detection performance of machine learning algorithms. Material and Methods Visual grading analysis was performed by four readers specialized in breast imaging who scored 100 cases reconstructed with both algorithms (70 lesions). Scoring (5-point scale: 1 = poor to 5 = excellent quality) was performed on presence of noise and artifacts, visualization of skin-line and Cooper's ligaments, contrast, and image quality, and, when present, lesion visibility. In parallel, a three-dimensional deep-learning convolutional neural network (3D-CNN) was trained (n = 259 patients, 51 positives with BI-RADS 3, 4, or 5 calcifications) and tested (n = 46 patients, nine positives), separately with FBP and EMPIRE volumes, to discriminate between samples with and without calcifications. The partial area under the receiver operating characteristic curve (pAUC) of each 3D-CNN was used for comparison. Results EMPIRE reconstructions showed better contrast (3.23 vs. 3.10, P = 0.010), image quality (3.22 vs. 3.03, P algorithm provides DBT volumes with better contrast and image quality, fewer artifacts, and improved visibility of calcifications for human observers, as well as improved detection performance with deep-learning algorithms.
International Nuclear Information System (INIS)
Grzetic, S; Weldon, M; Noa, K; Gupta, N
2016-01-01
Purpose: This study compares the newly released MaxFOV Revision 1 EFOV reconstruction algorithm for GE RT590 to the older WideView EFOV algorithm. Two radiotherapy overlays from Q-fix and Diacor, are included in our analysis. Hounsfield Units (HU) generated with the WideView algorithm varied in the extended field (beyond 50cm) and the scanned object’s border varied from slice to slice. A validation of HU consistency between the two reconstruction algorithms is performed. Methods: A CatPhan 504 and CIRS062 Electron Density Phantom were scanned on a GE RT590 CT-Simulator. The phantoms were positioned in multiple locations within the scan field of view so some of the density plugs were outside the 50cm reconstruction circle. Images were reconstructed using both the WideView and MaxFOV algorithms. The HU for each scan were characterized both in average over a volume and in profile. Results: HU values are consistent between the two algorithms. Low-density material will have a slight increase in HU value and high-density material will have a slight decrease in HU value as the distance from the sweet spot increases. Border inconsistencies and shading artifacts are still present with the MaxFOV reconstruction on the Q-fix overlay but not the Diacor overlay (It should be noted that the Q-fix overlay is not currently GE-certified). HU values for water outside the 50cm FOV are within 40HU of reconstructions at the sweet spot of the scanner. CatPhan HU profiles show improvement with the MaxFOV algorithm as it approaches the scanner edge. Conclusion: The new MaxFOV algorithm improves the contour border for objects outside of the standard FOV when using a GE-approved tabletop. Air cavities outside of the standard FOV create inconsistent object borders. HU consistency is within GE specifications and the accuracy of the phantom edge improves. Further adjustments to the algorithm are being investigated by GE.
A fast method to emulate an iterative POCS image reconstruction algorithm.
Zeng, Gengsheng L
2017-10-01
Iterative image reconstruction algorithms are commonly used to optimize an objective function, especially when the objective function is nonquadratic. Generally speaking, the iterative algorithms are computationally inefficient. This paper presents a fast algorithm that has one backprojection and no forward projection. This paper derives a new method to solve an optimization problem. The nonquadratic constraint, for example, an edge-preserving denoising constraint is implemented as a nonlinear filter. The algorithm is derived based on the POCS (projections onto projections onto convex sets) approach. A windowed FBP (filtered backprojection) algorithm enforces the data fidelity. An iterative procedure, divided into segments, enforces edge-enhancement denoising. Each segment performs nonlinear filtering. The derived iterative algorithm is computationally efficient. It contains only one backprojection and no forward projection. Low-dose CT data are used for algorithm feasibility studies. The nonlinearity is implemented as an edge-enhancing noise-smoothing filter. The patient studies results demonstrate its effectiveness in processing low-dose x ray CT data. This fast algorithm can be used to replace many iterative algorithms. © 2017 American Association of Physicists in Medicine.
Reconstruction of sparse-view X-ray computed tomography using adaptive iterative algorithms.
Liu, Li; Lin, Weikai; Jin, Mingwu
2015-01-01
In this paper, we propose two reconstruction algorithms for sparse-view X-ray computed tomography (CT). Treating the reconstruction problems as data fidelity constrained total variation (TV) minimization, both algorithms adapt the alternate two-stage strategy: projection onto convex sets (POCS) for data fidelity and non-negativity constraints and steepest descent for TV minimization. The novelty of this work is to determine iterative parameters automatically from data, thus avoiding tedious manual parameter tuning. In TV minimization, the step sizes of steepest descent are adaptively adjusted according to the difference from POCS update in either the projection domain or the image domain, while the step size of algebraic reconstruction technique (ART) in POCS is determined based on the data noise level. In addition, projection errors are used to compare with the error bound to decide whether to perform ART so as to reduce computational costs. The performance of the proposed methods is studied and evaluated using both simulated and physical phantom data. Our methods with automatic parameter tuning achieve similar, if not better, reconstruction performance compared to a representative two-stage algorithm. Copyright © 2014 Elsevier Ltd. All rights reserved.
Computed tomographic reconstruction of beam profiles with a multi-wire chamber
International Nuclear Information System (INIS)
Alonso, J.R.; Tobias, C.A.; Chu, W.T.
1979-03-01
MEDUSA (MEdical Dose Uniformity SAmpler), a 16 plane multi-wire proportional chamber, has been built to accurately measure beam profiles. The large number of planes allows for reconstruction of highly detailed beam intensity structures by means of Fourier convolution reconstruction techniques. This instrument is being used for verification and tuning of the Bevalac radiotherapy beams, but has potential applications in many beam profile monitoring situations
International Nuclear Information System (INIS)
Zhao-Jiang, Chen; Shu-Yi, Zhang
2010-01-01
A new hybrid inversion method for depth profiling reconstruction of thermal conductivities of inhomogeneous solids is proposed based on multilayer quadrupole formalism of thermal waves, particle swarm optimization and sequential quadratic programming. The reconstruction simulations for several thermal conductivity profiles are performed to evaluate the applicability of the method. The numerical simulations demonstrate that the precision and insensitivity to noise of the inversion method are very satisfactory. (condensed matter: structure, mechanical and thermal properties)
Optimization of CT image reconstruction algorithms for the lung tissue research consortium (LTRC)
McCollough, Cynthia; Zhang, Jie; Bruesewitz, Michael; Bartholmai, Brian
2006-03-01
To create a repository of clinical data, CT images and tissue samples and to more clearly understand the pathogenetic features of pulmonary fibrosis and emphysema, the National Heart, Lung, and Blood Institute (NHLBI) launched a cooperative effort known as the Lung Tissue Resource Consortium (LTRC). The CT images for the LTRC effort must contain accurate CT numbers in order to characterize tissues, and must have high-spatial resolution to show fine anatomic structures. This study was performed to optimize the CT image reconstruction algorithms to achieve these criteria. Quantitative analyses of phantom and clinical images were conducted. The ACR CT accreditation phantom containing five regions of distinct CT attenuations (CT numbers of approximately -1000 HU, -80 HU, 0 HU, 130 HU and 900 HU), and a high-contrast spatial resolution test pattern, was scanned using CT systems from two manufacturers (General Electric (GE) Healthcare and Siemens Medical Solutions). Phantom images were reconstructed using all relevant reconstruction algorithms. Mean CT numbers and image noise (standard deviation) were measured and compared for the five materials. Clinical high-resolution chest CT images acquired on a GE CT system for a patient with diffuse lung disease were reconstructed using BONE and STANDARD algorithms and evaluated by a thoracic radiologist in terms of image quality and disease extent. The clinical BONE images were processed with a 3 x 3 x 3 median filter to simulate a thicker slice reconstructed in smoother algorithms, which have traditionally been proven to provide an accurate estimation of emphysema extent in the lungs. Using a threshold technique, the volume of emphysema (defined as the percentage of lung voxels having a CT number lower than -950 HU) was computed for the STANDARD, BONE, and BONE filtered. The CT numbers measured in the ACR CT Phantom images were accurate for all reconstruction kernels for both manufacturers. As expected, visual evaluation of the
Directory of Open Access Journals (Sweden)
Peigang Ning
Full Text Available OBJECTIVE: This work aims to explore the effects of adaptive statistical iterative reconstruction (ASiR and model-based iterative reconstruction (MBIR algorithms in reducing computed tomography (CT radiation dosages in abdominal imaging. METHODS: CT scans on a standard male phantom were performed at different tube currents. Images at the different tube currents were reconstructed with the filtered back-projection (FBP, 50% ASiR and MBIR algorithms and compared. The CT value, image noise and contrast-to-noise ratios (CNRs of the reconstructed abdominal images were measured. Volumetric CT dose indexes (CTDIvol were recorded. RESULTS: At different tube currents, 50% ASiR and MBIR significantly reduced image noise and increased the CNR when compared with FBP. The minimal tube current values required by FBP, 50% ASiR, and MBIR to achieve acceptable image quality using this phantom were 200, 140, and 80 mA, respectively. At the identical image quality, 50% ASiR and MBIR reduced the radiation dose by 35.9% and 59.9% respectively when compared with FBP. CONCLUSIONS: Advanced iterative reconstruction techniques are able to reduce image noise and increase image CNRs. Compared with FBP, 50% ASiR and MBIR reduced radiation doses by 35.9% and 59.9%, respectively.
Li Voti, R.; Sibilia, C.; Bertolotti, M.
2003-01-01
Photothermal depth profiling has been the subject of many papers in the last years. Inverse problems on different kinds of materials have been identified, classified, and solved. A first classification has been done according to the type of depth profile: the physical quantity to be reconstructed is the optical absorption in the problems of type I, the thermal effusivity for type II, and both of them for type III. Another classification may be done depending on the time scale of the pump beam heating (frequency scan, time scan), or on its geometrical symmetry (one- or three-dimensional). In this work we want to discuss two different approaches, the genetic algorithms (GA) [R. Li Voti, C. Melchiorri, C. Sibilia, and M. Bertolotti, Anal. Sci. 17, 410 (2001); R. Li Voti, Proceedings, IV Int. Workshop on Advances in Signal Processing for Non-Destructive Evaluation of Materials, Quebec, August 2001] and the thermal wave backscattering (TWBS) [R. Li Voti, G. L. Liakhou, S. Paoloni, C. Sibilia, and M. Bertolotti, Anal. Sci. 17, 414 (2001); J. C. Krapez and R. Li Voti, Anal. Sci. 17, 417 (2001)], showing their performances and limits of validity for several kinds of photothermal depth profiling problems: The two approaches are based on different mechanisms and exhibit obviously different features. GA may be implemented on the exact heat diffusion equation as follows: one chromosome is associated to each profile. The genetic evolution of the chromosome allows one to find better and better profiles, eventually converging towards the solution of the inverse problem. The main advantage is that GA may be applied to any arbitrary profile, but several disadvantages exist; for example, the complexity of the algorithm, the slow convergence, and consequently the computer time consumed. On the contrary, TWBS uses a simplified theoretical model of heat diffusion in inhomogeneous materials. According to such a model, the photothermal signal depends linearly on the thermal effusivity
Reconstruction of sound speed profile through natural generalized inverse technique
Digital Repository Service at National Institute of Oceanography (India)
Murty, T.V.R.; Somayajulu, Y.K.; Murty, C.S.
An acoustic model has been developed for reconstruction of vertical sound speed in a near stable or stratified ocean. Generalized inverse method is utilised in the model development. Numerical experiments have been carried out to account...
Improved iterative image reconstruction algorithm for the exterior problem of computed tomography
International Nuclear Information System (INIS)
Guo, Yumeng; Zeng, Li
2017-01-01
In industrial applications that are limited by the angle of a fan-beam and the length of a detector, the exterior problem of computed tomography (CT) uses only the projection data that correspond to the external annulus of the objects to reconstruct an image. Because the reconstructions are not affected by the projection data that correspond to the interior of the objects, the exterior problem is widely applied to detect cracks in the outer wall of large-sized objects, such as in-service pipelines. However, image reconstruction in the exterior problem is still a challenging problem due to truncated projection data and beam-hardening, both of which can lead to distortions and artifacts. Thus, developing an effective algorithm and adopting a scanning trajectory suited for the exterior problem may be valuable. In this study, an improved iterative algorithm that combines total variation minimization (TVM) with a region scalable fitting (RSF) model was developed for a unilateral off-centered scanning trajectory and can be utilized to inspect large-sized objects for defects. Experiments involving simulated phantoms and real projection data were conducted to validate the practicality of our algorithm. Furthermore, comparative experiments show that our algorithm outperforms others in suppressing the artifacts caused by truncated projection data and beam-hardening.
Wu, Zhejun; Kudenov, Michael W.
2017-05-01
This paper presents a reconstruction algorithm for the Spatial-Spectral Multiplexing (SSM) optical system. The goal of this algorithm is to recover the three-dimensional spatial and spectral information of a scene, given that a one-dimensional spectrometer array is used to sample the pupil of the spatial-spectral modulator. The challenge of the reconstruction is that the non-parametric representation of the three-dimensional spatial and spectral object requires a large number of variables, thus leading to an underdetermined linear system that is hard to uniquely recover. We propose to reparameterize the spectrum using B-spline functions to reduce the number of unknown variables. Our reconstruction algorithm then solves the improved linear system via a least- square optimization of such B-spline coefficients with additional spatial smoothness regularization. The ground truth object and the optical model for the measurement matrix are simulated with both spatial and spectral assumptions according to a realistic field of view. In order to test the robustness of the algorithm, we add Poisson noise to the measurement and test on both two-dimensional and three-dimensional spatial and spectral scenes. Our analysis shows that the root mean square error of the recovered results can be achieved within 5.15%.
Improved iterative image reconstruction algorithm for the exterior problem of computed tomography
Energy Technology Data Exchange (ETDEWEB)
Guo, Yumeng [Chongqing University, College of Mathematics and Statistics, Chongqing 401331 (China); Chongqing University, ICT Research Center, Key Laboratory of Optoelectronic Technology and System of the Education Ministry of China, Chongqing 400044 (China); Zeng, Li, E-mail: drlizeng@cqu.edu.cn [Chongqing University, College of Mathematics and Statistics, Chongqing 401331 (China); Chongqing University, ICT Research Center, Key Laboratory of Optoelectronic Technology and System of the Education Ministry of China, Chongqing 400044 (China)
2017-01-11
In industrial applications that are limited by the angle of a fan-beam and the length of a detector, the exterior problem of computed tomography (CT) uses only the projection data that correspond to the external annulus of the objects to reconstruct an image. Because the reconstructions are not affected by the projection data that correspond to the interior of the objects, the exterior problem is widely applied to detect cracks in the outer wall of large-sized objects, such as in-service pipelines. However, image reconstruction in the exterior problem is still a challenging problem due to truncated projection data and beam-hardening, both of which can lead to distortions and artifacts. Thus, developing an effective algorithm and adopting a scanning trajectory suited for the exterior problem may be valuable. In this study, an improved iterative algorithm that combines total variation minimization (TVM) with a region scalable fitting (RSF) model was developed for a unilateral off-centered scanning trajectory and can be utilized to inspect large-sized objects for defects. Experiments involving simulated phantoms and real projection data were conducted to validate the practicality of our algorithm. Furthermore, comparative experiments show that our algorithm outperforms others in suppressing the artifacts caused by truncated projection data and beam-hardening.
Energy Technology Data Exchange (ETDEWEB)
Lin, Yuan, E-mail: yuan.lin@duke.edu; Samei, Ehsan [Carl E. Ravin Advanced Imaging Laboratories, Duke University Medical Center, 2424 Erwin Road, Suite 302, Durham, North Carolina 27705 (United States)
2014-02-15
Purpose: In quantitative myocardial CT perfusion imaging, beam hardening effect due to dense bone and high concentration iodinated contrast agent can result in visible artifacts and inaccurate CT numbers. In this paper, an efficient polyenergetic Simultaneous Algebraic Reconstruction Technique (pSART) was presented to eliminate the beam hardening artifacts and to improve the CT quantitative imaging ability. Methods: Our algorithm made threea priori assumptions: (1) the human body is composed of several base materials (e.g., fat, breast, soft tissue, bone, and iodine); (2) images can be coarsely segmented to two types of regions, i.e., nonbone regions and noniodine regions; and (3) each voxel can be decomposed into a mixture of two most suitable base materials according to its attenuation value and its corresponding region type information. Based on the above assumptions, energy-independent accumulated effective lengths of all base materials can be fast computed in the forward ray-tracing process and be used repeatedly to obtain accurate polyenergetic projections, with which a SART-based equation can correctly update each voxel in the backward projecting process to iteratively reconstruct artifact-free images. This approach effectively reduces the influence of polyenergetic x-ray sources and it further enables monoenergetic images to be reconstructed at any arbitrarily preselected target energies. A series of simulation tests were performed on a size-variable cylindrical phantom and a realistic anthropomorphic thorax phantom. In addition, a phantom experiment was also performed on a clinical CT scanner to further quantitatively validate the proposed algorithm. Results: The simulations with the cylindrical phantom and the anthropomorphic thorax phantom showed that the proposed algorithm completely eliminated beam hardening artifacts and enabled quantitative imaging across different materials, phantom sizes, and spectra, as the absolute relative errors were reduced
International Nuclear Information System (INIS)
Lin, Yuan; Samei, Ehsan
2014-01-01
Purpose: In quantitative myocardial CT perfusion imaging, beam hardening effect due to dense bone and high concentration iodinated contrast agent can result in visible artifacts and inaccurate CT numbers. In this paper, an efficient polyenergetic Simultaneous Algebraic Reconstruction Technique (pSART) was presented to eliminate the beam hardening artifacts and to improve the CT quantitative imaging ability. Methods: Our algorithm made threea priori assumptions: (1) the human body is composed of several base materials (e.g., fat, breast, soft tissue, bone, and iodine); (2) images can be coarsely segmented to two types of regions, i.e., nonbone regions and noniodine regions; and (3) each voxel can be decomposed into a mixture of two most suitable base materials according to its attenuation value and its corresponding region type information. Based on the above assumptions, energy-independent accumulated effective lengths of all base materials can be fast computed in the forward ray-tracing process and be used repeatedly to obtain accurate polyenergetic projections, with which a SART-based equation can correctly update each voxel in the backward projecting process to iteratively reconstruct artifact-free images. This approach effectively reduces the influence of polyenergetic x-ray sources and it further enables monoenergetic images to be reconstructed at any arbitrarily preselected target energies. A series of simulation tests were performed on a size-variable cylindrical phantom and a realistic anthropomorphic thorax phantom. In addition, a phantom experiment was also performed on a clinical CT scanner to further quantitatively validate the proposed algorithm. Results: The simulations with the cylindrical phantom and the anthropomorphic thorax phantom showed that the proposed algorithm completely eliminated beam hardening artifacts and enabled quantitative imaging across different materials, phantom sizes, and spectra, as the absolute relative errors were reduced
Ooi, Adrian Sh; Kanapathy, Muholan; Ong, Yee Siang; Tan, Kok Chai; Tan, Bien Keem
2015-11-01
Scalp soft tissue defects are common and result from a variety of causes. Reconstructive methods should maximise cosmetic outcomes by maintaining hair-bearing tissue and aesthetic hairlines. This article outlines an algorithm based on a diverse clinical case series to optimise scalp soft tissue coverage. A retrospective analysis of scalp soft tissue reconstruction cases performed at the Singapore General Hospital between January 2004 and December 2013 was conducted. Forty-one patients were included in this study. The majority of defects aesthetic outcome while minimising complications and repeat procedures.
Direct cone-beam cardiac reconstruction algorithm with cardiac banding artifact correction
International Nuclear Information System (INIS)
Taguchi, Katsuyuki; Chiang, Beshan S.; Hein, Ilmar A.
2006-01-01
Multislice helical computed tomography (CT) is a promising noninvasive technique for coronary artery imaging. Various factors can cause inconsistencies in cardiac CT data, which can result in degraded image quality. These inconsistencies may be the result of the patient physiology (e.g., heart rate variations), the nature of the data (e.g., cone-angle), or the reconstruction algorithm itself. An algorithm which provides the best temporal resolution for each slice, for example, often provides suboptimal image quality for the entire volume since the cardiac temporal resolution (TRc) changes from slice to slice. Such variations in TRc can generate strong banding artifacts in multi-planar reconstruction images or three-dimensional images. Discontinuous heart walls and coronary arteries may compromise the accuracy of the diagnosis. A β-blocker is often used to reduce and stabilize patients' heart rate but cannot eliminate the variation. In order to obtain robust and optimal image quality, a software solution that increases the temporal resolution and decreases the effect of heart rate is highly desirable. This paper proposes an ECG-correlated direct cone-beam reconstruction algorithm (TCOT-EGR) with cardiac banding artifact correction (CBC) and disconnected projections redundancy compensation technique (DIRECT). First the theory and analytical model of the cardiac temporal resolution is outlined. Next, the performance of the proposed algorithms is evaluated by using computer simulations as well as patient data. It will be shown that the proposed algorithms enhance the robustness of the image quality against inconsistencies by guaranteeing smooth transition of heart cycles used in reconstruction
Energy Technology Data Exchange (ETDEWEB)
Park, Yeonok; Cho, Heemoon; Je, Uikyu; Cho, Hyosung, E-mail: hscho1@yonsei.ac.kr; Park, Chulkyu; Lim, Hyunwoo; Kim, Kyuseok; Kim, Guna; Park, Soyoung; Woo, Taeho; Choi, Sungil
2015-12-21
In this work, we have developed a prototype digital breast tomosynthesis (DBT) system which mainly consists of an x-ray generator (28 kV{sub p}, 7 mA s), a CMOS-type flat-panel detector (70-μm pixel size, 230.5×339 mm{sup 2} active area), and a rotational arm to move the x-ray generator in an arc. We employed a compressed-sensing (CS)-based reconstruction algorithm, rather than a common filtered-backprojection (FBP) one, for more accurate DBT reconstruction. Here the CS is a state-of-the-art mathematical theory for solving the inverse problems, which exploits the sparsity of the image with substantially high accuracy. We evaluated the reconstruction quality in terms of the detectability, the contrast-to-noise ratio (CNR), and the slice-sensitive profile (SSP) by using the mammographic accreditation phantom (Model 015, CIRS Inc.) and compared it to the FBP-based quality. The CS-based algorithm yielded much better image quality, preserving superior image homogeneity, edge sharpening, and cross-plane resolution, compared to the FBP-based one. - Highlights: • A prototype digital breast tomosynthesis (DBT) system is developed. • Compressed-sensing (CS) based reconstruction framework is employed. • We reconstructed high-quality DBT images by using the proposed reconstruction framework.
An accelerated threshold-based back-projection algorithm for Compton camera image reconstruction
International Nuclear Information System (INIS)
Mundy, Daniel W.; Herman, Michael G.
2011-01-01
Purpose: Compton camera imaging (CCI) systems are currently under investigation for radiotherapy dose reconstruction and verification. The ability of such a system to provide real-time images during dose delivery will be limited by the computational speed of the image reconstruction algorithm. In this work, the authors present a fast and simple method by which to generate an initial back-projected image from acquired CCI data, suitable for use in a filtered back-projection algorithm or as a starting point for iterative reconstruction algorithms, and compare its performance to the current state of the art. Methods: Each detector event in a CCI system describes a conical surface that includes the true point of origin of the detected photon. Numerical image reconstruction algorithms require, as a first step, the back-projection of each of these conical surfaces into an image space. The algorithm presented here first generates a solution matrix for each slice of the image space by solving the intersection of the conical surface with the image plane. Each element of the solution matrix is proportional to the distance of the corresponding voxel from the true intersection curve. A threshold function was developed to extract those pixels sufficiently close to the true intersection to generate a binary intersection curve. This process is repeated for each image plane for each CCI detector event, resulting in a three-dimensional back-projection image. The performance of this algorithm was tested against a marching algorithm known for speed and accuracy. Results: The threshold-based algorithm was found to be approximately four times faster than the current state of the art with minimal deficit to image quality, arising from the fact that a generically applicable threshold function cannot provide perfect results in all situations. The algorithm fails to extract a complete intersection curve in image slices near the detector surface for detector event cones having axes nearly
Time-Domain Techniques for Computation and Reconstruction of One-Dimensional Profiles
Directory of Open Access Journals (Sweden)
M. Rahman
2005-01-01
Full Text Available This paper presents a time-domain technique to compute the electromagnetic fields and to reconstruct the permittivity profile within a one-dimensional medium of finite length. The medium is characterized by a permittivity as well as conductivity profile which vary only with depth. The discussed scattering problem is thus one-dimensional. The modeling tool is divided into two different schemes which are named as the forward solver and the inverse solver. The task of the forward solver is to compute the internal fields of the specimen which is performed by Green’s function approach. When a known electromagnetic wave is incident normally on the media, the resulting electromagnetic field within the media can be calculated by constructing a Green’s operator. This operator maps the incident field on either side of the medium to the field at an arbitrary observation point. It is nothing but a matrix of integral operators with kernels satisfying known partial differential equations. The reflection and transmission behavior of the medium is also determined from the boundary values of the Green's operator. The inverse solver is responsible for solving an inverse scattering problem by reconstructing the permittivity profile of the medium. Though it is possible to use several algorithms to solve this problem, the invariant embedding method, also known as the layer-stripping method, has been implemented here due to the advantage that it requires a finite time trace of reflection data. Here only one round trip of reflection data is used, where one round trip is defined by the time required by the pulse to propagate through the medium and back again. The inversion process begins by retrieving the reflection kernel from the reflected wave data by simply using a deconvolution technique. The rest of the task can easily be performed by applying a numerical approach to determine different profile parameters. Both the solvers have been found to have the
A parallel algorithm for 3D particle tracking and Lagrangian trajectory reconstruction
International Nuclear Information System (INIS)
Barker, Douglas; Zhang, Yuanhui; Lifflander, Jonathan; Arya, Anshu
2012-01-01
Particle-tracking methods are widely used in fluid mechanics and multi-target tracking research because of their unique ability to reconstruct long trajectories with high spatial and temporal resolution. Researchers have recently demonstrated 3D tracking of several objects in real time, but as the number of objects is increased, real-time tracking becomes impossible due to data transfer and processing bottlenecks. This problem may be solved by using parallel processing. In this paper, a parallel-processing framework has been developed based on frame decomposition and is programmed using the asynchronous object-oriented Charm++ paradigm. This framework can be a key step in achieving a scalable Lagrangian measurement system for particle-tracking velocimetry and may lead to real-time measurement capabilities. The parallel tracking algorithm was evaluated with three data sets including the particle image velocimetry standard 3D images data set #352, a uniform data set for optimal parallel performance and a computational-fluid-dynamics-generated non-uniform data set to test trajectory reconstruction accuracy, consistency with the sequential version and scalability to more than 500 processors. The algorithm showed strong scaling up to 512 processors and no inherent limits of scalability were seen. Ultimately, up to a 200-fold speedup is observed compared to the serial algorithm when 256 processors were used. The parallel algorithm is adaptable and could be easily modified to use any sequential tracking algorithm, which inputs frames of 3D particle location data and outputs particle trajectories
Muckley, Matthew J; Noll, Douglas C; Fessler, Jeffrey A
2015-02-01
Sparsity-promoting regularization is useful for combining compressed sensing assumptions with parallel MRI for reducing scan time while preserving image quality. Variable splitting algorithms are the current state-of-the-art algorithms for SENSE-type MR image reconstruction with sparsity-promoting regularization. These methods are very general and have been observed to work with almost any regularizer; however, the tuning of associated convergence parameters is a commonly-cited hindrance in their adoption. Conversely, majorize-minimize algorithms based on a single Lipschitz constant have been observed to be slow in shift-variant applications such as SENSE-type MR image reconstruction since the associated Lipschitz constants are loose bounds for the shift-variant behavior. This paper bridges the gap between the Lipschitz constant and the shift-variant aspects of SENSE-type MR imaging by introducing majorizing matrices in the range of the regularizer matrix. The proposed majorize-minimize methods (called BARISTA) converge faster than state-of-the-art variable splitting algorithms when combined with momentum acceleration and adaptive momentum restarting. Furthermore, the tuning parameters associated with the proposed methods are unitless convergence tolerances that are easier to choose than the constraint penalty parameters required by variable splitting algorithms.
Li, Haoting; Chen, Rongqing; Xu, Canhua; Liu, Benyuan; Tang, Mengxing; Yang, Lin; Dong, Xiuzhen; Fu, Feng
2017-08-21
Dynamic brain electrical impedance tomography (EIT) is a promising technique for continuously monitoring the development of cerebral injury. While there are many reconstruction algorithms available for brain EIT, there is still a lack of study to compare their performance in the context of dynamic brain monitoring. To address this problem, we develop a framework for evaluating different current algorithms with their ability to correctly identify small intracranial conductivity changes. Firstly, a simulation 3D head phantom with realistic layered structure and impedance distribution is developed. Next several reconstructing algorithms, such as back projection (BP), damped least-square (DLS), Bayesian, split Bregman (SB) and GREIT are introduced. We investigate their temporal response, noise performance, location and shape error with respect to different noise levels on the simulation phantom. The results show that the SB algorithm demonstrates superior performance in reducing image error. To further improve the location accuracy, we optimize SB by incorporating the brain structure-based conductivity distribution priors, in which differences of the conductivities between different brain tissues and the inhomogeneous conductivity distribution of the skull are considered. We compare this novel algorithm (called SB-IBCD) with SB and DLS using anatomically correct head shaped phantoms with spatial varying skull conductivity. Main results and Significance: The results showed that SB-IBCD is the most effective in unveiling small intracranial conductivity changes, where it can reduce the image error by an average of 30.0% compared to DLS.
Reconstruction algorithms in the Super-Kamiokande large water Cherenkov detector
International Nuclear Information System (INIS)
Shiozawa, M.
1999-01-01
The Super-Kamiokande experiment, using a large underground water Cherenkov detector, has started its operation since first April, 1996. One of the main physics goals of this experiment is to measure the atmospheric neutrinos. Proton decay search is also an important topic. For these analyses, all measurement of physical quantities of an event such as vertex position, the number of Cherenkov rings, momentum, particle type and the number of decay electrons, is automatically performed by reconstruction algorithms. We attain enough quality of the analyses using these algorithms and several impressive results have been addressed
Reconstruction algorithms in the Super-Kamiokande large water Cherenkov detector
Shiozawa, M
1999-01-01
The Super-Kamiokande experiment, using a large underground water Cherenkov detector, has started its operation since first April, 1996. One of the main physics goals of this experiment is to measure the atmospheric neutrinos. Proton decay search is also an important topic. For these analyses, all measurement of physical quantities of an event such as vertex position, the number of Cherenkov rings, momentum, particle type and the number of decay electrons, is automatically performed by reconstruction algorithms. We attain enough quality of the analyses using these algorithms and several impressive results have been addressed.
Energy Technology Data Exchange (ETDEWEB)
Jacobsson Svärd, Staffan, E-mail: staffan.jacobsson_svard@physics.uu.se; Holcombe, Scott; Grape, Sophie
2015-05-21
A fuel assembly operated in a nuclear power plant typically contains 100–300 fuel rods, depending on fuel type, which become strongly radioactive during irradiation in the reactor core. For operational and security reasons, it is of interest to experimentally deduce rod-wise information from the fuel, preferably by means of non-destructive measurements. The tomographic SPECT technique offers such possibilities through its two-step application; (1) recording the gamma-ray flux distribution around the fuel assembly, and (2) reconstructing the assembly's internal source distribution, based on the recorded radiation field. In this paper, algorithms for performing the latter step and extracting quantitative relative rod-by-rod data are accounted for. As compared to application of SPECT in nuclear medicine, nuclear fuel assemblies present a much more heterogeneous distribution of internal attenuation to gamma radiation than the human body, typically with rods containing pellets of heavy uranium dioxide surrounded by cladding of a zirconium alloy placed in water or air. This inhomogeneity severely complicates the tomographic quantification of the rod-wise relative source content, and the deduction of conclusive data requires detailed modelling of the attenuation to be introduced in the reconstructions. However, as shown in this paper, simplified models may still produce valuable information about the fuel. Here, a set of reconstruction algorithms for SPECT on nuclear fuel assemblies are described and discussed in terms of their quantitative performance for two applications; verification of fuel assemblies' completeness in nuclear safeguards, and rod-wise fuel characterization. It is argued that a request not to base the former assessment on any a priori information brings constraints to which reconstruction methods that may be used in that case, whereas the use of a priori information on geometry and material content enables highly accurate quantitative
Layer-oriented multigrid wavefront reconstruction algorithms for multi-conjugate adaptive optics
Gilles, Luc; Ellerbroek, Brent L.; Vogel, Curtis R.
2003-02-01
Multi-conjugate adaptive optics (MCAO) systems with 104-105 degrees of freedom have been proposed for future giant telescopes. Using standard matrix methods to compute, optimize, and implement wavefront control algorithms for these systems is impractical, since the number of calculations required to compute and apply the reconstruction matrix scales respectively with the cube and the square of the number of AO degrees of freedom. In this paper, we develop an iterative sparse matrix implementation of minimum variance wavefront reconstruction for telescope diameters up to 32m with more than 104 actuators. The basic approach is the preconditioned conjugate gradient method, using a multigrid preconditioner incorporating a layer-oriented (block) symmetric Gauss-Seidel iterative smoothing operator. We present open-loop numerical simulation results to illustrate algorithm convergence.
International Nuclear Information System (INIS)
Du, Yi; Yu, Gongyi; Xiang, Xincheng; Wang, Xiangang; De Deene, Yves
2017-01-01
Computational simulations are used to investigate the convergence of a hybrid iterative algorithm for optical CT reconstruction, i.e. the simultaneous algebraic reconstruction technique (SART) integrated with ordered subsets (OS) iteration and total variation (TV) minimization regularization, or SART+OS+TV for short. The influence of parameter selection to reach convergence, spatial dose gradient integrity, MTF and convergent speed are discussed. It’s shown that the results of SART+OS+TV algorithm converge to the true values without significant bias, and MTF and convergent speed are affected by different parameter sets used for iterative calculation. In conclusion, the performance of the SART+OS+TV depends on parameter selection, which also implies that careful parameter tuning work is required and necessary for proper spatial performance and fast convergence. (paper)
Energy Technology Data Exchange (ETDEWEB)
Solomon, Justin, E-mail: justin.solomon@duke.edu [Carl E. Ravin Advanced Imaging Laboratories, Department of Radiology, Duke University Medical Center, Durham, North Carolina 27705 (United States); Samei, Ehsan [Carl E. Ravin Advanced Imaging Laboratories, Department of Radiology, Duke University Medical Center, Durham, North Carolina 27705 and Departments of Biomedical Engineering and Electrical and Computer Engineering, Pratt School of Engineering, Duke University, Durham, North Carolina 27705 (United States)
2014-09-15
Purpose: Quantum noise properties of CT images are generally assessed using simple geometric phantoms with uniform backgrounds. Such phantoms may be inadequate when assessing nonlinear reconstruction or postprocessing algorithms. The purpose of this study was to design anatomically informed textured phantoms and use the phantoms to assess quantum noise properties across two clinically available reconstruction algorithms, filtered back projection (FBP) and sinogram affirmed iterative reconstruction (SAFIRE). Methods: Two phantoms were designed to represent lung and soft-tissue textures. The lung phantom included intricate vessel-like structures along with embedded nodules (spherical, lobulated, and spiculated). The soft tissue phantom was designed based on a three-dimensional clustered lumpy background with included low-contrast lesions (spherical and anthropomorphic). The phantoms were built using rapid prototyping (3D printing) technology and, along with a uniform phantom of similar size, were imaged on a Siemens SOMATOM Definition Flash CT scanner and reconstructed with FBP and SAFIRE. Fifty repeated acquisitions were acquired for each background type and noise was assessed by estimating pixel-value statistics, such as standard deviation (i.e., noise magnitude), autocorrelation, and noise power spectrum. Noise stationarity was also assessed by examining the spatial distribution of noise magnitude. The noise properties were compared across background types and between the two reconstruction algorithms. Results: In FBP and SAFIRE images, noise was globally nonstationary for all phantoms. In FBP images of all phantoms, and in SAFIRE images of the uniform phantom, noise appeared to be locally stationary (within a reasonably small region of interest). Noise was locally nonstationary in SAFIRE images of the textured phantoms with edge pixels showing higher noise magnitude compared to pixels in more homogenous regions. For pixels in uniform regions, noise magnitude was
International Nuclear Information System (INIS)
Solomon, Justin; Samei, Ehsan
2014-01-01
Purpose: Quantum noise properties of CT images are generally assessed using simple geometric phantoms with uniform backgrounds. Such phantoms may be inadequate when assessing nonlinear reconstruction or postprocessing algorithms. The purpose of this study was to design anatomically informed textured phantoms and use the phantoms to assess quantum noise properties across two clinically available reconstruction algorithms, filtered back projection (FBP) and sinogram affirmed iterative reconstruction (SAFIRE). Methods: Two phantoms were designed to represent lung and soft-tissue textures. The lung phantom included intricate vessel-like structures along with embedded nodules (spherical, lobulated, and spiculated). The soft tissue phantom was designed based on a three-dimensional clustered lumpy background with included low-contrast lesions (spherical and anthropomorphic). The phantoms were built using rapid prototyping (3D printing) technology and, along with a uniform phantom of similar size, were imaged on a Siemens SOMATOM Definition Flash CT scanner and reconstructed with FBP and SAFIRE. Fifty repeated acquisitions were acquired for each background type and noise was assessed by estimating pixel-value statistics, such as standard deviation (i.e., noise magnitude), autocorrelation, and noise power spectrum. Noise stationarity was also assessed by examining the spatial distribution of noise magnitude. The noise properties were compared across background types and between the two reconstruction algorithms. Results: In FBP and SAFIRE images, noise was globally nonstationary for all phantoms. In FBP images of all phantoms, and in SAFIRE images of the uniform phantom, noise appeared to be locally stationary (within a reasonably small region of interest). Noise was locally nonstationary in SAFIRE images of the textured phantoms with edge pixels showing higher noise magnitude compared to pixels in more homogenous regions. For pixels in uniform regions, noise magnitude was
Petersen, T. C.; Ringer, S. P.
2010-03-01
Upon discerning the mere shape of an imaged object, as portrayed by projected perimeters, the full three-dimensional scattering density may not be of particular interest. In this situation considerable simplifications to the reconstruction problem are possible, allowing calculations based upon geometric principles. Here we describe and provide an algorithm which reconstructs the three-dimensional morphology of specimens from tilt series of images for application to electron tomography. Our algorithm uses a differential approach to infer the intersection of projected tangent lines with surfaces which define boundaries between regions of different scattering densities within and around the perimeters of specimens. Details of the algorithm implementation are given and explained using reconstruction calculations from simulations, which are built into the code. An experimental application of the algorithm to a nano-sized Aluminium tip is also presented to demonstrate practical analysis for a real specimen. Program summaryProgram title: STOMO version 1.0 Catalogue identifier: AEFS_v1_0 Program summary URL:http://cpc.cs.qub.ac.uk/summaries/AEFS_v1_0.html Program obtainable from: CPC Program Library, Queen's University, Belfast, N. Ireland Licensing provisions: Standard CPC licence, http://cpc.cs.qub.ac.uk/licence/licence.html No. of lines in distributed program, including test data, etc.: 2988 No. of bytes in distributed program, including test data, etc.: 191 605 Distribution format: tar.gz Programming language: C/C++ Computer: PC Operating system: Windows XP RAM: Depends upon the size of experimental data as input, ranging from 200 Mb to 1.5 Gb Supplementary material: Sample output files, for the test run provided, are available. Classification: 7.4, 14 External routines: Dev-C++ ( http://www.bloodshed.net/devcpp.html) Nature of problem: Electron tomography of specimens for which conventional back projection may fail and/or data for which there is a limited angular
Directory of Open Access Journals (Sweden)
Hamidreza Mousavi
2017-01-01
Full Text Available Recently different approaches have been developed in the field of sensor fault diagnostics based on Auto-Associative Neural Network (AANN. In this paper we present a novel algorithm called Self reconstructing Auto-Associative Neural Network (S-AANN which is able to detect and isolate single faulty sensor via reconstruction. We have also extended the algorithm to be applicable in multiple fault conditions. The algorithm uses a calibration model based on AANN. AANN can reconstruct the faulty sensor using non-faulty sensors due to correlation between the process variables, and mean of the difference between reconstructed and original data determines which sensors are faulty. The algorithms are tested on a Dimerization process. The simulation results show that the S-AANN can isolate multiple faulty sensors with low computational time that make the algorithm appropriate candidate for online applications.
Yu, Liang; Antoni, Jerome; Leclere, Quentin; Jiang, Weikang
2017-11-01
Acoustical source reconstruction is a typical inverse problem, whose minimum frequency of reconstruction hinges on the size of the array and maximum frequency depends on the spacing distance between the microphones. For the sake of enlarging the frequency of reconstruction and reducing the cost of an acquisition system, Cyclic Projection (CP), a method of sequential measurements without reference, was recently investigated (JSV,2016,372:31-49). In this paper, the Propagation based Fast Iterative Shrinkage Thresholding Algorithm (Propagation-FISTA) is introduced, which improves CP in two aspects: (1) the number of acoustic sources is no longer needed and the only making assumption is that of a "weakly sparse" eigenvalue spectrum; (2) the construction of the spatial basis is much easier and adaptive to practical scenarios of acoustical measurements benefiting from the introduction of propagation based spatial basis. The proposed Propagation-FISTA is first investigated with different simulations and experimental setups and is next illustrated with an industrial case.
International Nuclear Information System (INIS)
Ley-Zaporozhan, Julia; Ley, Sebastian; Weinheimer, Oliver; Iliyushenko, Svitlana; Erdugan, Serap; Eberhardt, Ralf; Fuxa, Adelheid; Mews, Juergen; Kauczor, Hans-Ulrich
2008-01-01
Purpose: The aim of the study was to compare the influence of different reconstruction algorithms on quantitative emphysema analysis in patients with severe emphysema. Material and methods: Twenty-five patients suffering from severe emphysema were included in the study. All patients underwent inspiratory MDCT (Aquilion-16, slice thickness 1/0.8 mm). The raw data were reconstructed using six different algorithms: bone kernel with beam hardening correction (BHC), soft tissue kernel with BHC; standard soft tissue kernel, smooth soft tissue kernel (internal reference standard), standard lung kernel, and high-convolution kernel. The only difference between image data sets was the algorithm employed to reconstruct the raw data, no additional radiation was required. CT data were analysed using self-written emphysema detection and quantification software providing lung volume, emphysema volume (EV), emphysema index (EI) and mean lung density (MLD). Results: The use of kernels with BHC led to a significant decrease in MLD (5%) and EI (61-79%) in comparison with kernels without BHC. The absolute difference (from smooth soft tissue kernel) in MLD ranged from -0.6 to -6.1 HU and were significant different for all kernels. The EV showed absolute differences between -0.05 and -0.4 L and was significantly different for all kernels. The EI showed absolute differences between -0.8 and -5.1 and was significantly different for all kernels. Conclusion: The use of kernels with BHC led to a significant decrease in MLD and EI. The absolute differences between different kernels without BHC were small but they were larger than the known interscan variation in patients. Thus, for follow-up examinations the same reconstruction algorithm has to be used and use of BHC has to be avoided
Evaluation of the OSC-TV iterative reconstruction algorithm for cone-beam optical CT.
Matenine, Dmitri; Mascolo-Fortin, Julia; Goussard, Yves; Després, Philippe
2015-11-01
The present work evaluates an iterative reconstruction approach, namely, the ordered subsets convex (OSC) algorithm with regularization via total variation (TV) minimization in the field of cone-beam optical computed tomography (optical CT). One of the uses of optical CT is gel-based 3D dosimetry for radiation therapy, where it is employed to map dose distributions in radiosensitive gels. Model-based iterative reconstruction may improve optical CT image quality and contribute to a wider use of optical CT in clinical gel dosimetry. This algorithm was evaluated using experimental data acquired by a cone-beam optical CT system, as well as complementary numerical simulations. A fast GPU implementation of OSC-TV was used to achieve reconstruction times comparable to those of conventional filtered backprojection. Images obtained via OSC-TV were compared with the corresponding filtered backprojections. Spatial resolution and uniformity phantoms were scanned and respective reconstructions were subject to evaluation of the modulation transfer function, image uniformity, and accuracy. The artifacts due to refraction and total signal loss from opaque objects were also studied. The cone-beam optical CT data reconstructions showed that OSC-TV outperforms filtered backprojection in terms of image quality, thanks to a model-based simulation of the photon attenuation process. It was shown to significantly improve the image spatial resolution and reduce image noise. The accuracy of the estimation of linear attenuation coefficients remained similar to that obtained via filtered backprojection. Certain image artifacts due to opaque objects were reduced. Nevertheless, the common artifact due to the gel container walls could not be eliminated. The use of iterative reconstruction improves cone-beam optical CT image quality in many ways. The comparisons between OSC-TV and filtered backprojection presented in this paper demonstrate that OSC-TV can potentially improve the rendering of
Ren, Zhong; Liu, Guodong; Huang, Zhen
2012-11-01
The image reconstruction is a key step in medical imaging (MI) and its algorithm's performance determinates the quality and resolution of reconstructed image. Although some algorithms have been used, filter back-projection (FBP) algorithm is still the classical and commonly-used algorithm in clinical MI. In FBP algorithm, filtering of original projection data is a key step in order to overcome artifact of the reconstructed image. Since simple using of classical filters, such as Shepp-Logan (SL), Ram-Lak (RL) filter have some drawbacks and limitations in practice, especially for the projection data polluted by non-stationary random noises. So, an improved wavelet denoising combined with parallel-beam FBP algorithm is used to enhance the quality of reconstructed image in this paper. In the experiments, the reconstructed effects were compared between the improved wavelet denoising and others (directly FBP, mean filter combined FBP and median filter combined FBP method). To determine the optimum reconstruction effect, different algorithms, and different wavelet bases combined with three filters were respectively test. Experimental results show the reconstruction effect of improved FBP algorithm is better than that of others. Comparing the results of different algorithms based on two evaluation standards i.e. mean-square error (MSE), peak-to-peak signal-noise ratio (PSNR), it was found that the reconstructed effects of the improved FBP based on db2 and Hanning filter at decomposition scale 2 was best, its MSE value was less and the PSNR value was higher than others. Therefore, this improved FBP algorithm has potential value in the medical imaging.
Design of 4D x-ray tomography experiments for reconstruction using regularized iterative algorithms
Mohan, K. Aditya
2017-10-01
4D X-ray computed tomography (4D-XCT) is widely used to perform non-destructive characterization of time varying physical processes in various materials. The conventional approach to improving temporal resolution in 4D-XCT involves the development of expensive and complex instrumentation that acquire data faster with reduced noise. It is customary to acquire data with many tomographic views at a high signal to noise ratio. Instead, temporal resolution can be improved using regularized iterative algorithms that are less sensitive to noise and limited views. These algorithms benefit from optimization of other parameters such as the view sampling strategy while improving temporal resolution by reducing the total number of views or the detector exposure time. This paper presents the design principles of 4D-XCT experiments when using regularized iterative algorithms derived using the framework of model-based reconstruction. A strategy for performing 4D-XCT experiments is presented that allows for improving the temporal resolution by progressively reducing the number of views or the detector exposure time. Theoretical analysis of the effect of the data acquisition parameters on the detector signal to noise ratio, spatial reconstruction resolution, and temporal reconstruction resolution is also presented in this paper.
A framelet-based iterative maximum-likelihood reconstruction algorithm for spectral CT
Wang, Yingmei; Wang, Ge; Mao, Shuwei; Cong, Wenxiang; Ji, Zhilong; Cai, Jian-Feng; Ye, Yangbo
2016-11-01
Standard computed tomography (CT) cannot reproduce spectral information of an object. Hardware solutions include dual-energy CT which scans the object twice in different x-ray energy levels, and energy-discriminative detectors which can separate lower and higher energy levels from a single x-ray scan. In this paper, we propose a software solution and give an iterative algorithm that reconstructs an image with spectral information from just one scan with a standard energy-integrating detector. The spectral information obtained can be used to produce color CT images, spectral curves of the attenuation coefficient μ (r,E) at points inside the object, and photoelectric images, which are all valuable imaging tools in cancerous diagnosis. Our software solution requires no change on hardware of a CT machine. With the Shepp-Logan phantom, we have found that although the photoelectric and Compton components were not perfectly reconstructed, their composite effect was very accurately reconstructed as compared to the ground truth and the dual-energy CT counterpart. This means that our proposed method has an intrinsic benefit in beam hardening correction and metal artifact reduction. The algorithm is based on a nonlinear polychromatic acquisition model for x-ray CT. The key technique is a sparse representation of iterations in a framelet system. Convergence of the algorithm is studied. This is believed to be the first application of framelet imaging tools to a nonlinear inverse problem.
Directory of Open Access Journals (Sweden)
Chong Fan
2017-03-01
Full Text Available A sub-block algorithm is usually applied in the super-resolution (SR reconstruction of images because of limitations in computer memory. However, the sub-block SR images can hardly achieve a seamless image mosaicking because of the uneven distribution of brightness and contrast among these sub-blocks. An effectively improved weighted Wallis dodging algorithm is proposed, aiming at the characteristic that SR reconstructed images are gray images with the same size and overlapping region. This algorithm can achieve consistency of image brightness and contrast. Meanwhile, a weighted adjustment sequence is presented to avoid the spatial propagation and accumulation of errors and the loss of image information caused by excessive computation. A seam line elimination method can share the partial dislocation in the seam line to the entire overlapping region with a smooth transition effect. Subsequently, the improved method is employed to remove the uneven illumination for 900 SR reconstructed images of ZY-3. Then, the overlapping image mosaic method is adopted to accomplish a seamless image mosaic based on the optimal seam line.
Effect of filters and reconstruction algorithms on I-124 PET in Siemens Inveon PET scanner
Ram Yu, A.; Kim, Jin Su
2015-10-01
Purpose: To assess the effects of filtering and reconstruction on Siemens I-124 PET data. Methods: A Siemens Inveon PET was used. Spatial resolution of I-124 was measured to a transverse offset of 50 mm from the center FBP, 2D ordered subset expectation maximization (OSEM2D), 3D re-projection algorithm (3DRP), and maximum a posteriori (MAP) methods were tested. Non-uniformity (NU), recovery coefficient (RC), and spillover ratio (SOR) parameterized image quality. Mini deluxe phantom data of I-124 was also assessed. Results: Volumetric resolution was 7.3 mm3 from the transverse FOV center when FBP reconstruction algorithms with ramp filter was used. MAP yielded minimal NU with β =1.5. OSEM2D yielded maximal RC. SOR was below 4% for FBP with ramp, Hamming, Hanning, or Shepp-Logan filters. Based on the mini deluxe phantom results, an FBP with Hanning or Parzen filters, or a 3DRP with Hanning filter yielded feasible I-124 PET data.Conclusions: Reconstruction algorithms and filters were compared. FBP with Hanning or Parzen filters, or 3DRP with Hanning filter yielded feasible data for quantifying I-124 PET.
Groce, J. L.; Izumi, K. H.; Markham, C. H.; Schwab, R. W.; Thompson, J. L.
1986-01-01
The Local Flow Management/Profile Descent (LFM/PD) algorithm designed for the NASA Transport System Research Vehicle program is described. The algorithm provides fuel-efficient altitude and airspeed profiles consistent with ATC restrictions in a time-based metering environment over a fixed ground track. The model design constraints include accommodation of both published profile descent procedures and unpublished profile descents, incorporation of fuel efficiency as a flight profile criterion, operation within the performance capabilities of the Boeing 737-100 airplane with JT8D-7 engines, and conformity to standard air traffic navigation and control procedures. Holding and path stretching capabilities are included for long delay situations.
International Nuclear Information System (INIS)
Leng Shuai; Zhuang Tingliang; Nett, Brian E; Chen Guanghong
2005-01-01
In this paper, we present a new algorithm designed for a specific data truncation problem in fan-beam CT. We consider a scanning configuration in which the fan-beam projection data are acquired from an asymmetrically positioned half-sized detector. Namely, the asymmetric detector only covers one half of the scanning field of view. Thus, the acquired fan-beam projection data are truncated at every view angle. If an explicit data rebinning process is not invoked, this data acquisition configuration will reek havoc on many known fan-beam image reconstruction schemes including the standard filtered backprojection (FBP) algorithm and the super-short-scan FBP reconstruction algorithms. However, we demonstrate that a recently developed fan-beam image reconstruction algorithm which reconstructs an image via filtering a backprojection image of differentiated projection data (FBPD) survives the above fan-beam data truncation problem. Namely, we may exactly reconstruct the whole image object using the truncated data acquired in a full scan mode (2π angular range). We may also exactly reconstruct a small region of interest (ROI) using the truncated projection data acquired in a short-scan mode (less than 2π angular range). The most important characteristic of the proposed reconstruction scheme is that an explicit data rebinning process is not introduced. Numerical simulations were conducted to validate the new reconstruction algorithm
A GREIT-type linear reconstruction algorithm for EIT using eigenimages
International Nuclear Information System (INIS)
Antink, Christoph Hoog; Pikkemaat, Robert; Leonhardt, Steffen
2013-01-01
Reconstruction in electrical impedance tomography (EIT) is a nonlinear, ill-posed inverse problem. Based on point-shaped training and evaluation data, the 'Graz consensus Reconstruction algorithm for EIT' (GREIT) constitutes a universal, homogenous method. While this is a very reasonable approach to the general problem, we ask the question if an optimized reconstruction method for a specific application of EIT, i.e. thoracic imaging, can be found. Instead of point-shaped training data we propose to use spatially extended training data consisting of eigenimages. To evaluate the quality of reconstruction of the proposed approach, figures of merit (FOMs) derived from the ones used in GREIT are developed. For the application of thoracic imaging, lung-shapes were segmented from a publicly available CT-database (www.dir-lab.com) and used to calculate the novel FOMs. With those, the general feasibility of using eigenimages is demonstrated and compared to the standard approach. In addition, it is shown that by using different sets of training data, the creation of an individually optimized linear method of reconstruction is possible.
A low-count reconstruction algorithm for Compton-based prompt gamma imaging
Huang, Hsuan-Ming; Liu, Chih-Chieh; Jan, Meei-Ling; Lee, Ming-Wei
2018-04-01
The Compton camera is an imaging device which has been proposed to detect prompt gammas (PGs) produced by proton–nuclear interactions within tissue during proton beam irradiation. Compton-based PG imaging has been developed to verify proton ranges because PG rays, particularly characteristic ones, have strong correlations with the distribution of the proton dose. However, accurate image reconstruction from characteristic PGs is challenging because the detector efficiency and resolution are generally low. Our previous study showed that point spread functions can be incorporated into the reconstruction process to improve image resolution. In this study, we proposed a low-count reconstruction algorithm to improve the image quality of a characteristic PG emission by pooling information from other characteristic PG emissions. PGs were simulated from a proton beam irradiated on a water phantom, and a two-stage Compton camera was used for PG detection. The results show that the image quality of the reconstructed characteristic PG emission is improved with our proposed method in contrast to the standard reconstruction method using events from only one characteristic PG emission. For the 4.44 MeV PG rays, both methods can be used to predict the positions of the peak and the distal falloff with a mean accuracy of 2 mm. Moreover, only the proposed method can improve the estimated positions of the peak and the distal falloff of 5.25 MeV PG rays, and a mean accuracy of 2 mm can be reached.
Novel automated inversion algorithm for temperature reconstruction using gas isotopes from ice cores
Directory of Open Access Journals (Sweden)
M. Döring
2018-06-01
Full Text Available Greenland past temperature history can be reconstructed by forcing the output of a firn-densification and heat-diffusion model to fit multiple gas-isotope data (δ15N or δ40Ar or δ15Nexcess extracted from ancient air in Greenland ice cores using published accumulation-rate (Acc datasets. We present here a novel methodology to solve this inverse problem, by designing a fully automated algorithm. To demonstrate the performance of this novel approach, we begin by intentionally constructing synthetic temperature histories and associated δ15N datasets, mimicking real Holocene data that we use as true values (targets to be compared to the output of the algorithm. This allows us to quantify uncertainties originating from the algorithm itself. The presented approach is completely automated and therefore minimizes the subjective impact of manual parameter tuning, leading to reproducible temperature estimates. In contrast to many other ice-core-based temperature reconstruction methods, the presented approach is completely independent from ice-core stable-water isotopes, providing the opportunity to validate water-isotope-based reconstructions or reconstructions where water isotopes are used together with δ15N or δ40Ar. We solve the inverse problem T(δ15N, Acc by using a combination of a Monte Carlo based iterative approach and the analysis of remaining mismatches between modelled and target data, based on cubic-spline filtering of random numbers and the laboratory-determined temperature sensitivity for nitrogen isotopes. Additionally, the presented reconstruction approach was tested by fitting measured δ40Ar and δ15Nexcess data, which led as well to a robust agreement between modelled and measured data. The obtained final mismatches follow a symmetric standard-distribution function. For the study on synthetic data, 95 % of the mismatches compared to the synthetic target data are in an envelope between 3.0 to 6.3 permeg for δ15N and 0.23 to 0
A hybrid reconstruction algorithm for fast and accurate 4D cone-beam CT imaging.
Yan, Hao; Zhen, Xin; Folkerts, Michael; Li, Yongbao; Pan, Tinsu; Cervino, Laura; Jiang, Steve B; Jia, Xun
2014-07-01
4D cone beam CT (4D-CBCT) has been utilized in radiation therapy to provide 4D image guidance in lung and upper abdomen area. However, clinical application of 4D-CBCT is currently limited due to the long scan time and low image quality. The purpose of this paper is to develop a new 4D-CBCT reconstruction method that restores volumetric images based on the 1-min scan data acquired with a standard 3D-CBCT protocol. The model optimizes a deformation vector field that deforms a patient-specific planning CT (p-CT), so that the calculated 4D-CBCT projections match measurements. A forward-backward splitting (FBS) method is invented to solve the optimization problem. It splits the original problem into two well-studied subproblems, i.e., image reconstruction and deformable image registration. By iteratively solving the two subproblems, FBS gradually yields correct deformation information, while maintaining high image quality. The whole workflow is implemented on a graphic-processing-unit to improve efficiency. Comprehensive evaluations have been conducted on a moving phantom and three real patient cases regarding the accuracy and quality of the reconstructed images, as well as the algorithm robustness and efficiency. The proposed algorithm reconstructs 4D-CBCT images from highly under-sampled projection data acquired with 1-min scans. Regarding the anatomical structure location accuracy, 0.204 mm average differences and 0.484 mm maximum difference are found for the phantom case, and the maximum differences of 0.3-0.5 mm for patients 1-3 are observed. As for the image quality, intensity errors below 5 and 20 HU compared to the planning CT are achieved for the phantom and the patient cases, respectively. Signal-noise-ratio values are improved by 12.74 and 5.12 times compared to results from FDK algorithm using the 1-min data and 4-min data, respectively. The computation time of the algorithm on a NVIDIA GTX590 card is 1-1.5 min per phase. High-quality 4D-CBCT imaging based
A hybrid reconstruction algorithm for fast and accurate 4D cone-beam CT imaging
Energy Technology Data Exchange (ETDEWEB)
Yan, Hao; Folkerts, Michael; Jiang, Steve B., E-mail: xun.jia@utsouthwestern.edu, E-mail: steve.jiang@UTSouthwestern.edu; Jia, Xun, E-mail: xun.jia@utsouthwestern.edu, E-mail: steve.jiang@UTSouthwestern.edu [Department of Radiation Oncology, The University of Texas, Southwestern Medical Center, Dallas, Texas 75390 (United States); Zhen, Xin [Department of Biomedical Engineering, Southern Medical University, Guangzhou, Guangdong 510515 (China); Li, Yongbao [Department of Radiation Oncology, The University of Texas, Southwestern Medical Center, Dallas, Texas 75390 and Department of Engineering Physics, Tsinghua University, Beijing 100084 (China); Pan, Tinsu [Department of Imaging Physics, The University of Texas, MD Anderson Cancer Center, Houston, Texas 77030 (United States); Cervino, Laura [Department of Radiation Medicine and Applied Sciences, University of California San Diego, La Jolla, California 92093 (United States)
2014-07-15
Purpose: 4D cone beam CT (4D-CBCT) has been utilized in radiation therapy to provide 4D image guidance in lung and upper abdomen area. However, clinical application of 4D-CBCT is currently limited due to the long scan time and low image quality. The purpose of this paper is to develop a new 4D-CBCT reconstruction method that restores volumetric images based on the 1-min scan data acquired with a standard 3D-CBCT protocol. Methods: The model optimizes a deformation vector field that deforms a patient-specific planning CT (p-CT), so that the calculated 4D-CBCT projections match measurements. A forward-backward splitting (FBS) method is invented to solve the optimization problem. It splits the original problem into two well-studied subproblems, i.e., image reconstruction and deformable image registration. By iteratively solving the two subproblems, FBS gradually yields correct deformation information, while maintaining high image quality. The whole workflow is implemented on a graphic-processing-unit to improve efficiency. Comprehensive evaluations have been conducted on a moving phantom and three real patient cases regarding the accuracy and quality of the reconstructed images, as well as the algorithm robustness and efficiency. Results: The proposed algorithm reconstructs 4D-CBCT images from highly under-sampled projection data acquired with 1-min scans. Regarding the anatomical structure location accuracy, 0.204 mm average differences and 0.484 mm maximum difference are found for the phantom case, and the maximum differences of 0.3–0.5 mm for patients 1–3 are observed. As for the image quality, intensity errors below 5 and 20 HU compared to the planning CT are achieved for the phantom and the patient cases, respectively. Signal-noise-ratio values are improved by 12.74 and 5.12 times compared to results from FDK algorithm using the 1-min data and 4-min data, respectively. The computation time of the algorithm on a NVIDIA GTX590 card is 1–1.5 min per phase
A practical exact maximum compatibility algorithm for reconstruction of recent evolutionary history.
Cherry, Joshua L
2017-02-23
Maximum compatibility is a method of phylogenetic reconstruction that is seldom applied to molecular sequences. It may be ideal for certain applications, such as reconstructing phylogenies of closely-related bacteria on the basis of whole-genome sequencing. Here I present an algorithm that rapidly computes phylogenies according to a compatibility criterion. Although based on solutions to the maximum clique problem, this algorithm deals properly with ambiguities in the data. The algorithm is applied to bacterial data sets containing up to nearly 2000 genomes with several thousand variable nucleotide sites. Run times are several seconds or less. Computational experiments show that maximum compatibility is less sensitive than maximum parsimony to the inclusion of nucleotide data that, though derived from actual sequence reads, has been identified as likely to be misleading. Maximum compatibility is a useful tool for certain phylogenetic problems, such as inferring the relationships among closely-related bacteria from whole-genome sequence data. The algorithm presented here rapidly solves fairly large problems of this type, and provides robustness against misleading characters than can pollute large-scale sequencing data.
International Nuclear Information System (INIS)
Zhang, Leihong; Liang, Dong
2016-01-01
In order to solve the problem that reconstruction efficiency and precision is not high, in this paper different samples are selected to reconstruct spectral reflectance, and a new kind of spectral reflectance reconstruction method based on the algorithm of compressive sensing is provided. Four different color numbers of matte color cards such as the ColorChecker Color Rendition Chart and Color Checker SG, the copperplate paper spot color card of Panton, and the Munsell colors card are chosen as training samples, the spectral image is reconstructed respectively by the algorithm of compressive sensing and pseudo-inverse and Wiener, and the results are compared. These methods of spectral reconstruction are evaluated by root mean square error and color difference accuracy. The experiments show that the cumulative contribution rate and color difference of the Munsell colors card are better than those of the other three numbers of color cards in the same conditions of reconstruction, and the accuracy of the spectral reconstruction will be affected by the training sample of different numbers of color cards. The key technology of reconstruction means that the uniformity and representation of the training sample selection has important significance upon reconstruction. In this paper, the influence of the sample selection on the spectral image reconstruction is studied. The precision of the spectral reconstruction based on the algorithm of compressive sensing is higher than that of the traditional algorithm of spectral reconstruction. By the MATLAB simulation results, it can be seen that the spectral reconstruction precision and efficiency are affected by the different color numbers of the training sample. (paper)
Performance of 3DOSEM and MAP algorithms for reconstructing low count SPECT acquisitions
Energy Technology Data Exchange (ETDEWEB)
Grootjans, Willem [Radboud Univ. Medical Center, Nijmegen (Netherlands). Dept. of Radiology and Nuclear Medicine; Leiden Univ. Medical Center (Netherlands). Dept. of Radiology; Meeuwis, Antoi P.W.; Gotthardt, Martin; Visser, Eric P. [Radboud Univ. Medical Center, Nijmegen (Netherlands). Dept. of Radiology and Nuclear Medicine; Slump, Cornelis H. [Univ. Twente, Enschede (Netherlands). MIRA Inst. for Biomedical Technology and Technical Medicine; Geus-Oei, Lioe-Fee de [Radboud Univ. Medical Center, Nijmegen (Netherlands). Dept. of Radiology and Nuclear Medicine; Univ. Twente, Enschede (Netherlands). MIRA Inst. for Biomedical Technology and Technical Medicine; Leiden Univ. Medical Center (Netherlands). Dept. of Radiology
2016-07-01
Low count single photon emission computed tomography (SPECT) is becoming more important in view of whole body SPECT and reduction of radiation dose. In this study, we investigated the performance of several 3D ordered subset expectation maximization (3DOSEM) and maximum a posteriori (MAP) algorithms for reconstructing low count SPECT images. Phantom experiments were conducted using the National Electrical Manufacturers Association (NEMA) NU2 image quality (IQ) phantom. The background compartment of the phantom was filled with varying concentrations of pertechnetate and indiumchloride, simulating various clinical imaging conditions. Images were acquired using a hybrid SPECT/CT scanner and reconstructed with 3DOSEM and MAP reconstruction algorithms implemented in Siemens Syngo MI.SPECT (Flash3D) and Hermes Hybrid Recon Oncology (Hyrid Recon 3DOSEM and MAP). Image analysis was performed by calculating the contrast recovery coefficient (CRC),percentage background variability (N%), and contrast-to-noise ratio (CNR), defined as the ratio between CRC and N%. Furthermore, image distortion is characterized by calculating the aspect ratio (AR) of ellipses fitted to the hot spheres. Additionally, the performance of these algorithms to reconstruct clinical images was investigated. Images reconstructed with 3DOSEM algorithms demonstrated superior image quality in terms of contrast and resolution recovery when compared to images reconstructed with filtered-back-projection (FBP), OSEM and 2DOSEM. However, occurrence of correlated noise patterns and image distortions significantly deteriorated the quality of 3DOSEM reconstructed images. The mean AR for the 37, 28, 22, and 17 mm spheres was 1.3, 1.3, 1.6, and 1.7 respectively. The mean N% increase in high and low count Flash3D and Hybrid Recon 3DOSEM from 5.9% and 4.0% to 11.1% and 9.0%, respectively. Similarly, the mean CNR decreased in high and low count Flash3D and Hybrid Recon 3DOSEM from 8.7 and 8.8 to 3.6 and 4
A reconstruction algorithm for electrical impedance tomography based on sparsity regularization
Jin, Bangti
2011-08-24
This paper develops a novel sparse reconstruction algorithm for the electrical impedance tomography problem of determining a conductivity parameter from boundary measurements. The sparsity of the \\'inhomogeneity\\' with respect to a certain basis is a priori assumed. The proposed approach is motivated by a Tikhonov functional incorporating a sparsity-promoting ℓ 1-penalty term, and it allows us to obtain quantitative results when the assumption is valid. A novel iterative algorithm of soft shrinkage type was proposed. Numerical results for several two-dimensional problems with both single and multiple convex and nonconvex inclusions were presented to illustrate the features of the proposed algorithm and were compared with one conventional approach based on smoothness regularization. © 2011 John Wiley & Sons, Ltd.
A cone-beam reconstruction algorithm using shift-variant filtering and cone-beam backprojection
International Nuclear Information System (INIS)
Defrise, M.; Clack, R.
1994-01-01
An exact inversion formula written in the form of shift-variant filtered-backprojection (FBP) is given for reconstruction from cone-beam data taken from any orbit satisfying Tuy's sufficiency conditions. The method is based on a result of Grangeat, involving the derivative of the three-dimensional (3-D) Radon transform, but unlike Grangeat's algorithm, no 3D rebinning step is required. Data redundancy, which occurs when several cone-beam projections supply the same values in the Radon domain, is handled using an elegant weighting function and without discarding data. The algorithm is expressed in a convenient cone-beam detector reference frame, and a specific example for the case of a dual orthogonal circular orbit is presented. When the method is applied to a single circular orbit, it is shown to be equivalent to the well-known algorithm of Feldkamp et al
Fully 3D PET image reconstruction using a fourier preconditioned conjugate-gradient algorithm
International Nuclear Information System (INIS)
Fessler, J.A.; Ficaro, E.P.
1996-01-01
Since the data sizes in fully 3D PET imaging are very large, iterative image reconstruction algorithms must converge in very few iterations to be useful. One can improve the convergence rate of the conjugate-gradient (CG) algorithm by incorporating preconditioning operators that approximate the inverse of the Hessian of the objective function. If the 3D cylindrical PET geometry were not truncated at the ends, then the Hessian of the penalized least-squares objective function would be approximately shift-invariant, i.e. G'G would be nearly block-circulant, where G is the system matrix. We propose a Fourier preconditioner based on this shift-invariant approximation to the Hessian. Results show that this preconditioner significantly accelerates the convergence of the CG algorithm with only a small increase in computation
Wu, Wei; Zhao, Dewei; Zhang, Huan
2015-12-01
Super-resolution image reconstruction is an effective method to improve the image quality. It has important research significance in the field of image processing. However, the choice of the dictionary directly affects the efficiency of image reconstruction. A sparse representation theory is introduced into the problem of the nearest neighbor selection. Based on the sparse representation of super-resolution image reconstruction method, a super-resolution image reconstruction algorithm based on multi-class dictionary is analyzed. This method avoids the redundancy problem of only training a hyper complete dictionary, and makes the sub-dictionary more representatives, and then replaces the traditional Euclidean distance computing method to improve the quality of the whole image reconstruction. In addition, the ill-posed problem is introduced into non-local self-similarity regularization. Experimental results show that the algorithm is much better results than state-of-the-art algorithm in terms of both PSNR and visual perception.
Identifying Septal Support Reconstructions for Saddle Nose Deformity: The Cakmak Algorithm.
Cakmak, Ozcan; Emre, Ismet Emrah; Ozkurt, Fazil Emre
2015-01-01
The saddle nose deformity is one of the most challenging problems in nasal surgery with a less predictable and reproducible result than other nasal procedures. The main feature of this deformity is loss of septal support with both functional and aesthetic implications. Most reports on saddle nose have focused on aesthetic improvement and neglected the reestablishment of septal support to improve airway. To explain how the Cakmak algorithm, an algorithm that describes various fixation techniques and grafts in different types of saddle nose deformities, aids in identifying saddle nose reconstructions that restore supportive nasal framework and provide the aesthetic improvements typically associated with procedures to correct saddle nose deformities. This algorithm presents septal support reconstruction of patients with saddle nose deformity based on the experience of the senior author in 206 patients with saddle nose deformity. Preoperative examination, intraoperative assessment, reconstruction techniques, graft materials, and patient evaluation of aesthetic success were documented, and 4 different types of saddle nose deformities were defined. The Cakmak algorithm classifies varying degrees of saddle nose deformity from type 0 to type 4 and helps identify the most appropriate surgical procedure to restore the supportive nasal framework and aesthetic dorsum. Among the 206 patients, 110 women and 96 men, mean (range) age was 39.7 years (15-68 years), and mean (range) of follow-up was 32 months (6-148 months). All but 12 patients had a history of previous nasal surgeries. Application of the Cakmak algorithm resulted in 36 patients categorized with type 0 saddle nose deformities; 79, type 1; 50, type 2; 20, type 3a; 7, type 3b; and 14, type 4. Postoperative photographs showed improvement of deformities, and patient surveys revealed aesthetic improvement in 201 patients and improvement in nasal breathing in 195 patients. Three patients developed postoperative infection
An evolutionary algorithm for tomographic reconstructions in limited data sets problems
International Nuclear Information System (INIS)
Turcanu, Catrinel; Craciunescu, Teddy
2000-01-01
The paper proposes a new method for tomographic reconstructions. Unlike nuclear medicine applications, in physical science problems we are often confronted with limited data sets: constraints in the number of projections or limited angle views. The problem of image reconstruction from projections may be considered as a problem of finding an image (solution) having projections that match the experimental ones. In our approach, we choose a statistical correlation coefficient to evaluate the fitness of any potential solution. The optimization process is carried out by an evolutionary algorithm. Our algorithm has some problem-oriented characteristics. One of them is that a chromosome, representing a potential solution, is not linear but coded as a matrix of pixels corresponding to a two-dimensional image. This kind of internal representation reflects the genuine manifestation and slight differences between two points situated in the original problem space give rise to similar differences once they become coded. Another particular feature is a newly built crossover operator: the grid-based crossover, suitable for high dimension two-dimensional chromosomes. Except for the population size and the dimension of the cutting grid for the grid-based crossover, all the other parameters of the algorithm are independent of the geometry of the tomographic reconstruction. The performances of the method are evaluated in comparison with a traditional tomographic method, based on the maximization of the entropy of the image, that proved to work well with limited data sets. The test phantom is typical for an application with limited data sets: the determination of the neutron energy spectra with time resolution in case of short-pulsed neutron emission. The qualitative judgement and also the quantitative one, based on some figures of merit, point out that the proposed method ensures an improved reconstruction of shapes, sizes and resolution in the image, even in the presence of noise
Derivation and implementation of a cone-beam reconstruction algorithm for nonplanar orbits
International Nuclear Information System (INIS)
Kudo, Hiroyuki; Saito, Tsuneo
1994-01-01
Smith and Grangeat derived a cone-beam inversion formula that can be applied when a nonplanar orbit satisfying the completeness condition is used. Although Grangeat's inversion formula is mathematically different from Smith's, they have similar overall structures to each other. The contribution of this paper is two-fold. First, based on the derivation of Smith, the authors point out that Grangeat's inversion formula and Smith's can be conveniently described using a single formula (the Smith-Grangeat inversion formula) that is in the form of space-variant filtering followed by cone-beam backprojection. Furthermore, the resulting formula is reformulated for data acquisition systems with a planar detector to obtain a new reconstruction algorithm. Second, the authors make two significant modifications to the new algorithm to reduce artifacts and numerical errors encountered in direct implementation of the new algorithm. As for exactness of the new algorithm, the following fact can be stated. The algorithm based on Grangeat's intermediate function is exact for any complete orbit, whereas that based on Smith's intermediate function should be considered as an approximate inverse excepting the special case where almost every plane in 3-D space meets the orbit. The validity of the new algorithm is demonstrated by simulation studies
Bieberle, M; Hampel, U
2015-06-13
Tomographic image reconstruction is based on recovering an object distribution from its projections, which have been acquired from all angular views around the object. If the angular range is limited to less than 180° of parallel projections, typical reconstruction artefacts arise when using standard algorithms. To compensate for this, specialized algorithms using a priori information about the object need to be applied. The application behind this work is ultrafast limited-angle X-ray computed tomography of two-phase flows. Here, only a binary distribution of the two phases needs to be reconstructed, which reduces the complexity of the inverse problem. To solve it, a new reconstruction algorithm (LSR) based on the level-set method is proposed. It includes one force function term accounting for matching the projection data and one incorporating a curvature-dependent smoothing of the phase boundary. The algorithm has been validated using simulated as well as measured projections of known structures, and its performance has been compared to the algebraic reconstruction technique and a binary derivative of it. The validation as well as the application of the level-set reconstruction on a dynamic two-phase flow demonstrated its applicability and its advantages over other reconstruction algorithms. © 2015 The Author(s) Published by the Royal Society. All rights reserved.
International Nuclear Information System (INIS)
Chen Jian-Lin; Li Lei; Wang Lin-Yuan; Cai Ai-Long; Xi Xiao-Qi; Zhang Han-Ming; Li Jian-Xin; Yan Bin
2015-01-01
The projection matrix model is used to describe the physical relationship between reconstructed object and projection. Such a model has a strong influence on projection and backprojection, two vital operations in iterative computed tomographic reconstruction. The distance-driven model (DDM) is a state-of-the-art technology that simulates forward and back projections. This model has a low computational complexity and a relatively high spatial resolution; however, it includes only a few methods in a parallel operation with a matched model scheme. This study introduces a fast and parallelizable algorithm to improve the traditional DDM for computing the parallel projection and backprojection operations. Our proposed model has been implemented on a GPU (graphic processing unit) platform and has achieved satisfactory computational efficiency with no approximation. The runtime for the projection and backprojection operations with our model is approximately 4.5 s and 10.5 s per loop, respectively, with an image size of 256×256×256 and 360 projections with a size of 512×512. We compare several general algorithms that have been proposed for maximizing GPU efficiency by using the unmatched projection/backprojection models in a parallel computation. The imaging resolution is not sacrificed and remains accurate during computed tomographic reconstruction. (paper)
Evaluation of imaging protocol for ECT based on CS image reconstruction algorithm
International Nuclear Information System (INIS)
Zhou Xiaolin; Yun Mingkai; Cao Xuexiang; Liu Shuangquan; Wang Lu; Huang Xianchao; Wei Long
2014-01-01
Single-photon emission computerized tomography and positron emission tomography are essential medical imaging tools, for which the sampling angle number and scan time should be carefully chosen to give a good compromise between image quality and radiopharmaceutical dose. In this study, the image quality of different acquisition protocols was evaluated via varied angle number and count number per angle with Monte Carlo simulation data. It was shown that, when similar imaging counts were used, the factor of acquisition counts was more important than that of the sampling number in emission computerized tomography. To further reduce the activity requirement and the scan duration, an iterative image reconstruction algorithm for limited-view and low-dose tomography based on compressed sensing theory has been developed. The total variation regulation was added to the reconstruction process to improve the signal to noise Ratio and reduce artifacts caused by the limited angle sampling. Maximization of the maximum likelihood of the estimated image and the measured data and minimization of the total variation of the image are alternatively implemented. By using this advanced algorithm, the reconstruction process is able to achieve image quality matching or exceed that of normal scans with only half of the injection radiopharmaceutical dose. (authors)
Analytical algorithm for the generation of polygonal projection data for tomographic reconstruction
International Nuclear Information System (INIS)
Davis, G.R.
1996-01-01
Tomographic reconstruction algorithms and filters can be tested using a mathematical phantom, that is, a computer program which takes numerical data as its input and outputs derived projection data. The input data is usually in the form of pixel ''densities'' over a regular grid, or position and dimensions of simple, geometrical objects. The former technique allows a greater variety of objects to be simulated, but is less suitable in the case when very small (relative to the ray-spacing) features are to be simulated. The second technique is normally used to simulate biological specimens, typically a human skull, modelled as a number of ellipses. This is not suitable for simulating non-biological specimens with features such as straight edges and fine cracks. We have therefore devised an algorithm for simulating objects described as a series of polygons. These polygons, or parts of them, may be smaller than the ray-spacing and there is no limit, except that imposed by computing resources, on the complexity, number or superposition of polygons. A simple test of such a phantom, reconstructed using the filtered back-projection method, revealed reconstruction artefacts not normally seen with ''biological'' phantoms. (orig.)
3D Fractal reconstruction of terrain profile data based on digital elevation model
International Nuclear Information System (INIS)
Huang, Y.M.; Chen, C.-J.
2009-01-01
Digital Elevation Model (DEM) often makes it difficult for terrain reconstruction and data storage due to the failure in acquisition of details with higher resolution. If original terrain of DEM can be simulated, resulting in geographical details can be represented precisely while reducing the data size, then an effective reconstruction scheme is essential. This paper adopts two sets of real-world 3D terrain profile data to proceed data reducing, i.e. data sampling randomly, then reconstruct them through 3D fractal reconstruction. Meanwhile, the quantitative and qualitative difference generated from different reduction rates were evaluated statistically. The research results show that, if 3D fractal interpolation method is applied to DEM reconstruction, the higher reduction rate can be obtained for DEM of larger data size with respect to that of smaller data size under the assumption that the entire terrain structure is still maintained.
Automated comparison of Bayesian reconstructions of experimental profiles with physical models
International Nuclear Information System (INIS)
Irishkin, Maxim
2014-01-01
In this work we developed an expert system that carries out in an integrated and fully automated way i) a reconstruction of plasma profiles from the measurements, using Bayesian analysis ii) a prediction of the reconstructed quantities, according to some models and iii) an intelligent comparison of the first two steps. This system includes systematic checking of the internal consistency of the reconstructed quantities, enables automated model validation and, if a well-validated model is used, can be applied to help detecting interesting new physics in an experiment. The work shows three applications of this quite general system. The expert system can successfully detect failures in the automated plasma reconstruction and provide (on successful reconstruction cases) statistics of agreement of the models with the experimental data, i.e. information on the model validity. (author) [fr
International Nuclear Information System (INIS)
Bueno, Josiane M.; Traina, Agma Juci M.; Cruvinel, Paulo E.
1995-01-01
This work presents an algorithm for three-dimensional digital image reconstruction. Such algorithms based on the combination of both a Fast Fourier Transform method with Hamming Window and the use of a tri-linear interpolation function. The algorithm allows not only the generation of three-dimensional spatial spin distribution maps for Magnetic Resonance Tomography data but also X and Y-rays linear attenuation coefficient maps for CT scanners. Results demonstrates the usefulness of the algorithm in three-dimensional image reconstruction by doing first two-dimensional reconstruction and rather after interpolation. The algorithm was developed in C++ language, and there are two available versions: one under the DOS environment, and the other under the UNIX/Sun environment. (author)
Precht, Helle; Thygesen, Jesper; Gerke, Oke; Egstrup, Kenneth; Waaler, Dag; Lambrechtsen, Jess
2016-12-01
Coronary computed tomography angiography (CCTA) requires high spatial and temporal resolution, increased low contrast resolution for the assessment of coronary artery stenosis, plaque detection, and/or non-coronary pathology. Therefore, new reconstruction algorithms, particularly iterative reconstruction (IR) techniques, have been developed in an attempt to improve image quality with no cost in radiation exposure. To evaluate whether adaptive statistical iterative reconstruction (ASIR) enhances perceived image quality in CCTA compared to filtered back projection (FBP). Thirty patients underwent CCTA due to suspected coronary artery disease. Images were reconstructed using FBP, 30% ASIR, and 60% ASIR. Ninety image sets were evaluated by five observers using the subjective visual grading analysis (VGA) and assessed by proportional odds modeling. Objective quality assessment (contrast, noise, and the contrast-to-noise ratio [CNR]) was analyzed with linear mixed effects modeling on log-transformed data. The need for ethical approval was waived by the local ethics committee as the study only involved anonymously collected clinical data. VGA showed significant improvements in sharpness by comparing FBP with ASIR, resulting in odds ratios of 1.54 for 30% ASIR and 1.89 for 60% ASIR ( P = 0.004). The objective measures showed significant differences between FBP and 60% ASIR ( P < 0.0001) for noise, with an estimated ratio of 0.82, and for CNR, with an estimated ratio of 1.26. ASIR improved the subjective image quality of parameter sharpness and, objectively, reduced noise and increased CNR.
An Algorithmic Approach to Total Breast Reconstruction with Free Tissue Transfer
Directory of Open Access Journals (Sweden)
Seong Cheol Yu
2013-05-01
Full Text Available As microvascular techniques continue to improve, perforator flap free tissue transfer is now the gold standard for autologous breast reconstruction. Various options are available for breast reconstruction with autologous tissue. These include the free transverse rectus abdominis myocutaneous (TRAM flap, deep inferior epigastric perforator flap, superficial inferior epigastric artery flap, superior gluteal artery perforator flap, and transverse/vertical upper gracilis flap. In addition, pedicled flaps can be very successful in the right hands and the right patient, such as the pedicled TRAM flap, latissimus dorsi flap, and thoracodorsal artery perforator. Each flap comes with its own advantages and disadvantages related to tissue properties and donor-site morbidity. Currently, the problem is how to determine the most appropriate flap for a particular patient among those potential candidates. Based on a thorough review of the literature and accumulated experiences in the author’s institution, this article provides a logical approach to autologous breast reconstruction. The algorithms presented here can be helpful to customize breast reconstruction to individual patient needs.
Development of an image reconstruction algorithm for a few number of projection data
International Nuclear Information System (INIS)
Vieira, Wilson S.; Brandao, Luiz E.; Braz, Delson
2007-01-01
An image reconstruction algorithm was developed for specific cases of radiotracer applications in industry (rotating cylindrical mixers), involving a very few number of projection data. The algorithm was planned for imaging radioactive isotope distributions around the center of circular planes. The method consists of adapting the original expectation maximization algorithm (EM) to solve the ill-posed emission tomography inverse problem in order to reconstruct transversal 2D images of an object with only four projections. To achieve this aim, counts of photons emitted by selected radioactive sources in the plane, after they had been simulated using the commercial software MICROSHIELD 5.05, constitutes the projections and a computational code (SPECTEM) was developed to generate activity vectors or images related to those sources. SPECTEM is flexible to support simultaneous changes of the detectors's geometry, the medium under investigation and the properties of the gamma radiation. As a consequence of the code had been followed correctly the proposed method, good results were obtained and they encouraged us to continue the next step of the research: the validation of SPECTEM utilizing experimental data to check its real performance. We aim this code will improve considerably radiotracer methodology, making easier the diagnosis of fails in industrial processes. (author)
Development of an image reconstruction algorithm for a few number of projection data
Energy Technology Data Exchange (ETDEWEB)
Vieira, Wilson S.; Brandao, Luiz E. [Instituto de Engenharia Nuclear (IEN-CNEN/RJ), Rio de Janeiro , RJ (Brazil)]. E-mails: wilson@ien.gov.br; brandao@ien.gov.br; Braz, Delson [Universidade Federal do Rio de Janeiro (UFRJ), RJ (Brazil). Coordenacao dos Programa de Pos-graduacao de Engenharia (COPPE). Lab. de Instrumentacao Nuclear]. E-mail: delson@mailhost.lin.ufrj.br
2007-07-01
An image reconstruction algorithm was developed for specific cases of radiotracer applications in industry (rotating cylindrical mixers), involving a very few number of projection data. The algorithm was planned for imaging radioactive isotope distributions around the center of circular planes. The method consists of adapting the original expectation maximization algorithm (EM) to solve the ill-posed emission tomography inverse problem in order to reconstruct transversal 2D images of an object with only four projections. To achieve this aim, counts of photons emitted by selected radioactive sources in the plane, after they had been simulated using the commercial software MICROSHIELD 5.05, constitutes the projections and a computational code (SPECTEM) was developed to generate activity vectors or images related to those sources. SPECTEM is flexible to support simultaneous changes of the detectors's geometry, the medium under investigation and the properties of the gamma radiation. As a consequence of the code had been followed correctly the proposed method, good results were obtained and they encouraged us to continue the next step of the research: the validation of SPECTEM utilizing experimental data to check its real performance. We aim this code will improve considerably radiotracer methodology, making easier the diagnosis of fails in industrial processes. (author)
International Nuclear Information System (INIS)
Ogino, Takashi; Egawa, Sunao
1991-01-01
New algorithms of CT value correction for reconstructing a radiotherapy simulation image through axial CT images were developed. One, designated plane weighting method, is to correct CT value in proportion to the position of the beam element passing through the voxel. The other, designated solid weighting method, is to correct CT value in proportion to the length of the beam element passing through the voxel and the volume of voxel. Phantom experiments showed fair spatial resolution in the transverse direction. In the longitudinal direction, however, spatial resolution of under slice thickness could not be obtained. Contrast resolution was equivalent for both methods. In patient studies, the reconstructed radiotherapy simulation image was almost similar in visual perception of the density resolution to a simulation film taken by X-ray simulator. (author)
International Nuclear Information System (INIS)
Roux, S.; Desbat, L.; Koenig, A.; Grangeat, P.
2005-01-01
In this paper we study a property of the filtering step of multi-cycle reconstruction algorithm used in the field of cardiac CT. We show that the common filtering step procedure is not optimal in the case of divergent geometry and decrease slightly the temporal resolution. We propose to use the filtering procedure related to the work of Noo at al ( F.Noo, M. Defrise, R. Clakdoyle, and H. Kudo. Image reconstruction from fan-beam projections on less than a short-scan. Phys. Med.Biol., 47:2525-2546, July 2002)and show that this alternative allows to reach the optimal temporal resolution with the same computational effort. (N.C.)
Suzuki, Shigeru; Machida, Haruhiko; Tanaka, Isao; Ueno, Eiko
2013-03-01
The purpose of this study was to evaluate the performance of model-based iterative reconstruction (MBIR) in measurement of the inner diameter of models of blood vessels and compare performance between MBIR and a standard filtered back projection (FBP) algorithm. Vascular models with wall thicknesses of 0.5, 1.0, and 1.5 mm were scanned with a 64-MDCT unit and densities of contrast material yielding 275, 396, and 542 HU. Images were reconstructed images by MBIR and FBP, and the mean diameter of each model vessel was measured by software automation. Twenty separate measurements were repeated for each vessel, and variance among the repeated measures was analyzed for determination of measurement error. For all nine model vessels, CT attenuation profiles were compared along a line passing through the luminal center on axial images reconstructed with FBP and MBIR, and the 10-90% edge rise distances at the boundary between the vascular wall and the lumen were evaluated. For images reconstructed with FBP, measurement errors were smallest for models with 1.5-mm wall thickness, except those filled with 275-HU contrast material, and errors grew as the density of the contrast material decreased. Measurement errors with MBIR were comparable to or less than those with FBP. In CT attenuation profiles of images reconstructed with MBIR, the 10-90% edge rise distances at the boundary between the lumen and vascular wall were relatively short for each vascular model compared with those of the profile curves of FBP images. MBIR is better than standard FBP for reducing reconstruction blur and improving the accuracy of diameter measurement at CT angiography.
Structural algorithm to reservoir reconstruction using passive seismic data (synthetic example)
Energy Technology Data Exchange (ETDEWEB)
Smaglichenko, Tatyana A.; Volodin, Igor A.; Lukyanitsa, Andrei A.; Smaglichenko, Alexander V.; Sayankina, Maria K. [Oil and Gas Research Institute, Russian Academy of Science, Gubkina str.3, 119333, Moscow (Russian Federation); Faculty of Computational Mathematics and Cybernetics, M.V. Lomonosov Moscow State University, Leninskie gory, 1, str.52,Second Teaching Building.119991 Moscow (Russian Federation); Shmidt' s Institute of Physics of the Earth, Russian Academy of Science, Bolshaya Gruzinskaya str. 10, str.1, 123995 Moscow (Russian Federation); Oil and Gas Research Institute, Russian Academy of Science, Gubkina str.3, 119333, Moscow (Russian Federation)
2012-09-26
Using of passive seismic observations to detect a reservoir is a new direction of prospecting and exploration of hydrocarbons. In order to identify thin reservoir model we applied the modification of Gaussian elimination method in conditions of incomplete synthetic data. Because of the singularity of a matrix conventional method does not work. Therefore structural algorithm has been developed by analyzing the given model as a complex model. Numerical results demonstrate of its advantage compared with usual way of solution. We conclude that the gas reservoir is reconstructed by retrieving of the image of encasing shale beneath it.
International Nuclear Information System (INIS)
Bufacchi, A.; Tripi, A.
1995-09-01
The aim of this paper is the presentation of a series of methodologies to recognize and to obtain a three-dimensional reconstruction of an inner architectural scene, using a gray level image obtained using a TV camera. In the first part of the work, a series of methods used to find the edges in an effective way are critically compared, obtaining a binary image, and then the application of the Hough transform to such binary image to find the straight lines in the original image are discussed. In the second part, an algorithm is shown in order to find the vanishing points in such image
Energy Technology Data Exchange (ETDEWEB)
Apfaltrer, Paul, E-mail: paul.apfaltrer@medma.uni-heidelberg.de [Department of Radiology and Radiological Science, Medical University of South Carolina, Charleston, PO Box 250322, 169 Ashley Avenue, Charleston, SC 29425 (United States); Institute of Clinical Radiology and Nuclear Medicine, Medical Faculty Mannheim, Heidelberg University, Theodor-Kutzer-Ufer 1-3, D-68167 Mannheim (Germany); Schoendube, Harald, E-mail: harald.schoendube@siemens.com [Siemens Healthcare, CT Division, Forchheim Siemens, Siemensstr. 1, 91301 Forchheim (Germany); Schoepf, U. Joseph, E-mail: schoepf@musc.edu [Department of Radiology and Radiological Science, Medical University of South Carolina, Charleston, PO Box 250322, 169 Ashley Avenue, Charleston, SC 29425 (United States); Allmendinger, Thomas, E-mail: thomas.allmendinger@siemens.com [Siemens Healthcare, CT Division, Forchheim Siemens, Siemensstr. 1, 91301 Forchheim (Germany); Tricarico, Francesco, E-mail: francescotricarico82@gmail.com [Department of Radiology and Radiological Science, Medical University of South Carolina, Charleston, PO Box 250322, 169 Ashley Avenue, Charleston, SC 29425 (United States); Department of Bioimaging and Radiological Sciences, Catholic University of the Sacred Heart, “A. Gemelli” Hospital, Largo A. Gemelli 8, Rome (Italy); Schindler, Andreas, E-mail: andreas.schindler@campus.lmu.de [Department of Radiology and Radiological Science, Medical University of South Carolina, Charleston, PO Box 250322, 169 Ashley Avenue, Charleston, SC 29425 (United States); Vogt, Sebastian, E-mail: sebastian.vogt@siemens.com [Siemens Healthcare, CT Division, Forchheim Siemens, Siemensstr. 1, 91301 Forchheim (Germany); Sunnegårdh, Johan, E-mail: johan.sunnegardh@siemens.com [Siemens Healthcare, CT Division, Forchheim Siemens, Siemensstr. 1, 91301 Forchheim (Germany); and others
2013-02-15
Objective: To evaluate the effect of a temporal resolution improvement method (TRIM) for cardiac CT on diagnostic image quality for coronary artery assessment. Materials and methods: The TRIM-algorithm employs an iterative approach to reconstruct images from less than 180° of projections and uses a histogram constraint to prevent the occurrence of limited-angle artifacts. This algorithm was applied in 11 obese patients (7 men, 67.2 ± 9.8 years) who had undergone second generation dual-source cardiac CT with 120 kV, 175–426 mAs, and 500 ms gantry rotation. All data were reconstructed with a temporal resolution of 250 ms using traditional filtered-back projection (FBP) and of 200 ms using the TRIM-algorithm. Contrast attenuation and contrast-to-noise-ratio (CNR) were measured in the ascending aorta. The presence and severity of coronary motion artifacts was rated on a 4-point Likert scale. Results: All scans were considered of diagnostic quality. Mean BMI was 36 ± 3.6 kg/m{sup 2}. Average heart rate was 60 ± 9 bpm. Mean effective dose was 13.5 ± 4.6 mSv. When comparing FBP- and TRIM reconstructed series, the attenuation within the ascending aorta (392 ± 70.7 vs. 396.8 ± 70.1 HU, p > 0.05) and CNR (13.2 ± 3.2 vs. 11.7 ± 3.1, p > 0.05) were not significantly different. A total of 110 coronary segments were evaluated. All studies were deemed diagnostic; however, there was a significant (p < 0.05) difference in the severity score distribution of coronary motion artifacts between FBP (median = 2.5) and TRIM (median = 2.0) reconstructions. Conclusion: The algorithm evaluated here delivers diagnostic imaging quality of the coronary arteries despite 500 ms gantry rotation. Possible applications include improvement of cardiac imaging on slower gantry rotation systems or mitigation of the trade-off between temporal resolution and CNR in obese patients.
International Nuclear Information System (INIS)
Majaron, B; Milanic, M
2010-01-01
Pulsed photothermal profiling involves reconstruction of temperature depth profile induced in a layered sample by single-pulse laser exposure, based on transient change in mid-infrared (IR) emission from its surface. Earlier studies have indicated that in watery tissues, featuring a pronounced spectral variation of mid-IR absorption coefficient, analysis of broadband radiometric signals within the customary monochromatic approximation adversely affects profiling accuracy. We present here an experimental comparison of pulsed photothermal profiling in layered agar gel samples utilizing a spectrally composite kernel matrix vs. the customary approach. By utilizing a custom reconstruction code, the augmented approach reduces broadening of individual temperature peaks to 14% of the absorber depth, in contrast to 21% obtained with the customary approach.
International Nuclear Information System (INIS)
Sun, Y.; Hou, Y.; Yan, Y.
2004-01-01
With the extensive application of industrial computed tomography in the field of non-destructive testing, how to improve the quality of the reconstructed image is receiving more and more concern. It is well known that in the existing cone-beam filtered backprojection reconstruction algorithms the cone angle is controlled within a narrow range. The reason of this limitation is the incompleteness of projection data when the cone angle increases. Thus the size of the tested workpiece is limited. Considering the characteristic of X-ray cone angle, an improved cone-beam filtered back-projection reconstruction algorithm taking account of angular correction is proposed in this paper. The aim of our algorithm is to correct the cone-angle effect resulted from the incompleteness of projection data in the conventional algorithm. The basis of the correction is the angular relationship among X-ray source, tested workpiece and the detector. Thus the cone angle is not strictly limited and this algorithm may be used to detect larger workpiece. Further more, adaptive wavelet filter is used to make multiresolution analysis, which can modify the wavelet decomposition series adaptively according to the demand for resolution of local reconstructed area. Therefore the computation and the time of reconstruction can be reduced, and the quality of the reconstructed image can also be improved. (author)
Simulation of 4-turn algorithms for reconstructing lattice optic functions from orbit measurements
International Nuclear Information System (INIS)
Koscielniak, S.; Iliev, A.
1994-06-01
We describe algorithms for reconstructing tune, closed-orbit, beta-function and phase advance from four individual turns of beam orbit acquisition data, under the assumption of coherent, almost linear and uncoupled betatron oscillations. To estimate the beta-function at, and phase advance between, position monitors, we require at least one anchor location consisting of two monitors separated by a drift. The algorithms were submitted to a Monte Carlo analysis to find the likely measurement accuracy of the optics functions in the KAON Factory Booster ring racetrack lattice, assuming beam position monitors with surveying and reading errors, and assuming an imperfect lattice with gradient and surveying errors. Some of the results of this study are reported. (author)
A new approximate algorithm for image reconstruction in cone-beam spiral CT at small cone-angles
International Nuclear Information System (INIS)
Schaller, S.; Flohr, T.; Steffen, P.
1996-01-01
This paper presents a new approximate algorithm for image reconstruction with cone-beam spiral CT data at relatively small cone-angles. Based on the algorithm of Wang et al., our method combines a special complementary interpolation with filtered backprojection. The presented algorithm has three main advantages over Wang's algorithm: (1) It overcomes the pitch limitation of Wang's algorithm. (2) It significantly improves z-resolution when suitable sampling schemes are applied. (3) It avoids the waste of applied radiation dose inherent to Wang's algorithm. Usage of the total applied dose is an important requirement in medical imaging. Our method has been implemented on a standard workstation. Reconstructions of computer-simulated data of different phantoms, assuming sampling conditions and image quality requirements typical to medical CT, show encouraging results
Study of a reconstruction algorithm for electrons in the ATLAS experiment in LHC
International Nuclear Information System (INIS)
Kerschen, N.
2006-09-01
The ATLAS experiment is a general purpose particle physics experiment mainly aimed at the discovery of the origin of mass through the research of the Higgs boson. In order to achieve this, the Large Hadron Collider at CERN will accelerate two proton beams and make them collide at the centre of the experiment. ATLAS will discover new particles through the measurement of their decay products. Electrons are such decay products: they produce an electromagnetic shower in the calorimeter by which they lose all their energy. The calorimeter is divided into cells and the deposited energy is reconstructed using an algorithm to assemble the cells into clusters. The purpose of this thesis is to study a new kind of algorithm adapting the cluster to the shower topology. In order to reconstruct the energy of the initially created electron, the cluster has to be calibrated by taking into account the energy lost in the dead material in front of the calorimeter. Therefore. a Monte-Carlo simulation of the ATLAS detector has been used to correct for effects of response modulation in position and in energy and to optimise the energy resolution as well as the linearity. An analysis of test beam data has been performed to study the behaviour of the algorithm in a more realistic environment. We show that the requirements of the experiment can be met for the linearity and resolution. The improvement of this new algorithm, compared to a fixed sized cluster. is the better recovery of Bremsstrahlung photons emitted by the electron in the material in front of the calorimeter. A Monte-Carlo analysis of the Higgs boson decay in four electrons confirms this result. (author)
Humphries, T.; Winn, J.; Faridani, A.
2017-08-01
Recent work in CT image reconstruction has seen increasing interest in the use of total variation (TV) and related penalties to regularize problems involving reconstruction from undersampled or incomplete data. Superiorization is a recently proposed heuristic which provides an automatic procedure to ‘superiorize’ an iterative image reconstruction algorithm with respect to a chosen objective function, such as TV. Under certain conditions, the superiorized algorithm is guaranteed to find a solution that is as satisfactory as any found by the original algorithm with respect to satisfying the constraints of the problem; this solution is also expected to be superior with respect to the chosen objective. Most work on superiorization has used reconstruction algorithms which assume a linear measurement model, which in the case of CT corresponds to data generated from a monoenergetic x-ray beam. Many CT systems generate x-rays from a polyenergetic spectrum, however, in which the measured data represent an integral of object attenuation over all energies in the spectrum. This inconsistency with the linear model produces the well-known beam hardening artifacts, which impair analysis of CT images. In this work we superiorize an iterative algorithm for reconstruction from polyenergetic data, using both TV and an anisotropic TV (ATV) penalty. We apply the superiorized algorithm in numerical phantom experiments modeling both sparse-view and limited-angle scenarios. In our experiments, the superiorized algorithm successfully finds solutions which are as constraints-compatible as those found by the original algorithm, with significantly reduced TV and ATV values. The superiorized algorithm thus produces images with greatly reduced sparse-view and limited angle artifacts, which are also largely free of the beam hardening artifacts that would be present if a superiorized version of a monoenergetic algorithm were used.
International Nuclear Information System (INIS)
Brady, S. L.; Yee, B. S.; Kaufman, R. A.
2012-01-01
Purpose: This study demonstrates a means of implementing an adaptive statistical iterative reconstruction (ASiR™) technique for dose reduction in computed tomography (CT) while maintaining similar noise levels in the reconstructed image. The effects of image quality and noise texture were assessed at all implementation levels of ASiR™. Empirically derived dose reduction limits were established for ASiR™ for imaging of the trunk for a pediatric oncology population ranging from 1 yr old through adolescence/adulthood. Methods: Image quality was assessed using metrics established by the American College of Radiology (ACR) CT accreditation program. Each image quality metric was tested using the ACR CT phantom with 0%–100% ASiR™ blended with filtered back projection (FBP) reconstructed images. Additionally, the noise power spectrum (NPS) was calculated for three common reconstruction filters of the trunk. The empirically derived limitations on ASiR™ implementation for dose reduction were assessed using (1, 5, 10) yr old and adolescent/adult anthropomorphic phantoms. To assess dose reduction limits, the phantoms were scanned in increments of increased noise index (decrementing mA using automatic tube current modulation) balanced with ASiR™ reconstruction to maintain noise equivalence of the 0% ASiR™ image. Results: The ASiR™ algorithm did not produce any unfavorable effects on image quality as assessed by ACR criteria. Conversely, low-contrast resolution was found to improve due to the reduction of noise in the reconstructed images. NPS calculations demonstrated that images with lower frequency noise had lower noise variance and coarser graininess at progressively higher percentages of ASiR™ reconstruction; and in spite of the similar magnitudes of noise, the image reconstructed with 50% or more ASiR™ presented a more smoothed appearance than the pre-ASiR™ 100% FBP image. Finally, relative to non-ASiR™ images with 100% of standard dose across the
Energy Technology Data Exchange (ETDEWEB)
Brady, S. L.; Yee, B. S.; Kaufman, R. A. [Department of Radiological Sciences, St. Jude Children' s Research Hospital, Memphis, Tennessee 38105 (United States)
2012-09-15
Purpose: This study demonstrates a means of implementing an adaptive statistical iterative reconstruction (ASiR Trade-Mark-Sign ) technique for dose reduction in computed tomography (CT) while maintaining similar noise levels in the reconstructed image. The effects of image quality and noise texture were assessed at all implementation levels of ASiR Trade-Mark-Sign . Empirically derived dose reduction limits were established for ASiR Trade-Mark-Sign for imaging of the trunk for a pediatric oncology population ranging from 1 yr old through adolescence/adulthood. Methods: Image quality was assessed using metrics established by the American College of Radiology (ACR) CT accreditation program. Each image quality metric was tested using the ACR CT phantom with 0%-100% ASiR Trade-Mark-Sign blended with filtered back projection (FBP) reconstructed images. Additionally, the noise power spectrum (NPS) was calculated for three common reconstruction filters of the trunk. The empirically derived limitations on ASiR Trade-Mark-Sign implementation for dose reduction were assessed using (1, 5, 10) yr old and adolescent/adult anthropomorphic phantoms. To assess dose reduction limits, the phantoms were scanned in increments of increased noise index (decrementing mA using automatic tube current modulation) balanced with ASiR Trade-Mark-Sign reconstruction to maintain noise equivalence of the 0% ASiR Trade-Mark-Sign image. Results: The ASiR Trade-Mark-Sign algorithm did not produce any unfavorable effects on image quality as assessed by ACR criteria. Conversely, low-contrast resolution was found to improve due to the reduction of noise in the reconstructed images. NPS calculations demonstrated that images with lower frequency noise had lower noise variance and coarser graininess at progressively higher percentages of ASiR Trade-Mark-Sign reconstruction; and in spite of the similar magnitudes of noise, the image reconstructed with 50% or more ASiR Trade-Mark-Sign presented a more
CSIR Research Space (South Africa)
Ngwangwa, HM
2008-07-01
Full Text Available on the road and driver to assess the integrity of road and vehicle infrastructure. In this paper, vehicle vibration data are applied to an artificial neural network to reconstruct the corresponding road surface profiles. The results show that the technique...
Design algorithm for generatrix profile of cylindrical crowned rollers
Directory of Open Access Journals (Sweden)
Creţu Spiridon
2017-01-01
Full Text Available The cross-section of roller profile controls the pressure distribution in the contact area and radically affects the roller bearings basic dynamic load rating and rating lives. Today the most used roller profiles are the logarithmic profile and cylindrical-crowned (ZB profile. The logarithmic profile has a continuous evolution with no discontinuities till the intersection with the end fillet while ZB profile has two more discontinuities at the intersections points between the crowning circle and straight line generatrix. Using a semianalytical method, a numerical study has been carried out to find the optimum ZB profile for rollers incorporated in cylindrical rollers bearings. The basic reference rating life (L10_r has been used as optimization criterion.
Directory of Open Access Journals (Sweden)
Song Chunsheng
2017-01-01
Full Text Available The adhesive-bonded joint of carbon fiber–reinforced plastic is one of the core components in aircraft structure design. It is an effective guarantee for the safety and reliability of the aerospace aircraft structure to use effective methods for monitoring and early warning of internal failure. In this article, the mapping relation model between the strain profiles of the adherend of the carbon fiber–reinforced plastic single-lap adhesive joint and the stiffness degradation evolution of adhesive layer was achieved by finite element software ABAQUS. The fiber Bragg grating was embedded in the adherend between the first and second layers at the end of the adhesive layer to calculate the reflection spectrum of fiber Bragg grating sensor region with improved T-matrix method for reconstruction of the adherend strain profile of fiber Bragg grating sensing area with the help of genetic algorithm. According to the reconstruction results, the maximum error between the ideal and reconstructed strain profile under different tension loads did not exceed 7.43%, showing a good coincidence degree. The monitoring method of the stiffness degradation evolution of adhesive layer of the carbon fiber–reinforced plastic single-lap joint based on the reconstruction of the adherend strain profile of fiber Bragg grating sensing area thus was figured out.
Evaluation of 3D reconstruction algorithms for a small animal PET camera
International Nuclear Information System (INIS)
Johnson, C.A.; Gandler, W.R.; Seidel, J.
1996-01-01
The use of paired, opposing position-sensitive phototube scintillation cameras (SCs) operating in coincidence for small animal imaging with positron emitters is currently under study. Because of the low sensitivity of the system even in 3D mode and the need to produce images with high resolution, it was postulated that a 3D expectation maximization (EM) reconstruction algorithm might be well suited for this application. We investigated four reconstruction algorithms for the 3D SC PET camera: 2D filtered back-projection (FBP), 2D ordered subset EM (OSEM), 3D reprojection (3DRP), and 3D OSEM. Noise was assessed for all slices by the coefficient of variation in a simulated uniform cylinder. Resolution was assessed from a simulation of 15 point sources in the warm background of the uniform cylinder. At comparable noise levels, the resolution achieved with OSEM (0.9-mm to 1.2-mm) is significantly better than that obtained with FBP or 3DRP (1.5-mm to 2.0-mm.) Images of a rat skull labeled with 18 F-fluoride suggest that 3D OSEM can improve image quality of a small animal PET camera
Algorithmic procedures for Bayesian MEG/EEG source reconstruction in SPM.
López, J D; Litvak, V; Espinosa, J J; Friston, K; Barnes, G R
2014-01-01
The MEG/EEG inverse problem is ill-posed, giving different source reconstructions depending on the initial assumption sets. Parametric Empirical Bayes allows one to implement most popular MEG/EEG inversion schemes (Minimum Norm, LORETA, etc.) within the same generic Bayesian framework. It also provides a cost-function in terms of the variational Free energy-an approximation to the marginal likelihood or evidence of the solution. In this manuscript, we revisit the algorithm for MEG/EEG source reconstruction with a view to providing a didactic and practical guide. The aim is to promote and help standardise the development and consolidation of other schemes within the same framework. We describe the implementation in the Statistical Parametric Mapping (SPM) software package, carefully explaining each of its stages with the help of a simple simulated data example. We focus on the Multiple Sparse Priors (MSP) model, which we compare with the well-known Minimum Norm and LORETA models, using the negative variational Free energy for model comparison. The manuscript is accompanied by Matlab scripts to allow the reader to test and explore the underlying algorithm. © 2013. Published by Elsevier Inc. All rights reserved.
Algorithmic procedures for Bayesian MEG/EEG source reconstruction in SPM☆
López, J.D.; Litvak, V.; Espinosa, J.J.; Friston, K.; Barnes, G.R.
2014-01-01
The MEG/EEG inverse problem is ill-posed, giving different source reconstructions depending on the initial assumption sets. Parametric Empirical Bayes allows one to implement most popular MEG/EEG inversion schemes (Minimum Norm, LORETA, etc.) within the same generic Bayesian framework. It also provides a cost-function in terms of the variational Free energy—an approximation to the marginal likelihood or evidence of the solution. In this manuscript, we revisit the algorithm for MEG/EEG source reconstruction with a view to providing a didactic and practical guide. The aim is to promote and help standardise the development and consolidation of other schemes within the same framework. We describe the implementation in the Statistical Parametric Mapping (SPM) software package, carefully explaining each of its stages with the help of a simple simulated data example. We focus on the Multiple Sparse Priors (MSP) model, which we compare with the well-known Minimum Norm and LORETA models, using the negative variational Free energy for model comparison. The manuscript is accompanied by Matlab scripts to allow the reader to test and explore the underlying algorithm. PMID:24041874
Performance of the reconstruction algorithms of the FIRST experiment pixel sensors vertex detector
Energy Technology Data Exchange (ETDEWEB)
Rescigno, R., E-mail: regina.rescigno@iphc.cnrs.fr [Institut Pluridisciplinaire Hubert Curien, 23 rue du Loess, 67037 Strasbourg Cedex 2 (France); Finck, Ch.; Juliani, D. [Institut Pluridisciplinaire Hubert Curien, 23 rue du Loess, 67037 Strasbourg Cedex 2 (France); Spiriti, E. [Istituto Nazionale di Fisica Nucleare - Laboratori Nazionali di Frascati (Italy); Istituto Nazionale di Fisica Nucleare - Sezione di Roma 3 (Italy); Baudot, J. [Institut Pluridisciplinaire Hubert Curien, 23 rue du Loess, 67037 Strasbourg Cedex 2 (France); Abou-Haidar, Z. [CNA, Sevilla (Spain); Agodi, C. [Istituto Nazionale di Fisica Nucleare - Laboratori Nazionali del Sud (Italy); Alvarez, M.A.G. [CNA, Sevilla (Spain); Aumann, T. [GSI Helmholtzzentrum für Schwerionenforschung, Darmstadt (Germany); Battistoni, G. [Istituto Nazionale di Fisica Nucleare - Sezione di Milano (Italy); Bocci, A. [CNA, Sevilla (Spain); Böhlen, T.T. [European Organization for Nuclear Research CERN, Geneva (Switzerland); Medical Radiation Physics, Karolinska Institutet and Stockholm University, Stockholm (Sweden); Boudard, A. [CEA-Saclay, IRFU/SPhN, Gif sur Yvette Cedex (France); Brunetti, A.; Carpinelli, M. [Istituto Nazionale di Fisica Nucleare - Sezione di Cagliari (Italy); Università di Sassari (Italy); Cirrone, G.A.P. [Istituto Nazionale di Fisica Nucleare - Laboratori Nazionali del Sud (Italy); Cortes-Giraldo, M.A. [Departamento de Fisica Atomica, Molecular y Nuclear, University of Sevilla, 41080-Sevilla (Spain); Cuttone, G.; De Napoli, M. [Istituto Nazionale di Fisica Nucleare - Laboratori Nazionali del Sud (Italy); Durante, M. [GSI Helmholtzzentrum für Schwerionenforschung, Darmstadt (Germany); and others
2014-12-11
Hadrontherapy treatments use charged particles (e.g. protons and carbon ions) to treat tumors. During a therapeutic treatment with carbon ions, the beam undergoes nuclear fragmentation processes giving rise to significant yields of secondary charged particles. An accurate prediction of these production rates is necessary to estimate precisely the dose deposited into the tumours and the surrounding healthy tissues. Nowadays, a limited set of double differential carbon fragmentation cross-section is available. Experimental data are necessary to benchmark Monte Carlo simulations for their use in hadrontherapy. The purpose of the FIRST experiment is to study nuclear fragmentation processes of ions with kinetic energy in the range from 100 to 1000 MeV/u. Tracks are reconstructed using information from a pixel silicon detector based on the CMOS technology. The performances achieved using this device for hadrontherapy purpose are discussed. For each reconstruction step (clustering, tracking and vertexing), different methods are implemented. The algorithm performances and the accuracy on reconstructed observables are evaluated on the basis of simulated and experimental data.
Blanken, T.C.; Felici, F.; Rapson, C.J.; de Baar, M.R.; Heemels, W.P.M.H.
2018-01-01
A model-based approach to real-time reconstruction of the particle density profile in tokamak plasmas is presented, based on a dynamic state estimator. Traditionally, the density profile is reconstructed in real-time by solving an ill-conditioned inversion problem using a measurement at a single
Barca, Patrizio; Giannelli, Marco; Fantacci, Maria Evelina; Caramella, Davide
2018-06-01
Computed tomography (CT) is a useful and widely employed imaging technique, which represents the largest source of population exposure to ionizing radiation in industrialized countries. Adaptive Statistical Iterative Reconstruction (ASIR) is an iterative reconstruction algorithm with the potential to allow reduction of radiation exposure while preserving diagnostic information. The aim of this phantom study was to assess the performance of ASIR, in terms of a number of image quality indices, when different reconstruction blending levels are employed. CT images of the Catphan-504 phantom were reconstructed using conventional filtered back-projection (FBP) and ASIR with reconstruction blending levels of 20, 40, 60, 80, and 100%. Noise, noise power spectrum (NPS), contrast-to-noise ratio (CNR) and modulation transfer function (MTF) were estimated for different scanning parameters and contrast objects. Noise decreased and CNR increased non-linearly up to 50 and 100%, respectively, with increasing blending level of reconstruction. Also, ASIR has proven to modify the NPS curve shape. The MTF of ASIR reconstructed images depended on tube load/contrast and decreased with increasing blending level of reconstruction. In particular, for low radiation exposure and low contrast acquisitions, ASIR showed lower performance than FBP, in terms of spatial resolution for all blending levels of reconstruction. CT image quality varies substantially with the blending level of reconstruction. ASIR has the potential to reduce noise whilst maintaining diagnostic information in low radiation exposure CT imaging. Given the opposite variation of CNR and spatial resolution with the blending level of reconstruction, it is recommended to use an optimal value of this parameter for each specific clinical application.
Median prior constrained TV algorithm for sparse view low-dose CT reconstruction.
Liu, Yi; Shangguan, Hong; Zhang, Quan; Zhu, Hongqing; Shu, Huazhong; Gui, Zhiguo
2015-05-01
It is known that lowering the X-ray tube current (mAs) or tube voltage (kVp) and simultaneously reducing the total number of X-ray views (sparse view) is an effective means to achieve low-dose in computed tomography (CT) scan. However, the associated image quality by the conventional filtered back-projection (FBP) usually degrades due to the excessive quantum noise. Although sparse-view CT reconstruction algorithm via total variation (TV), in the scanning protocol of reducing X-ray tube current, has been demonstrated to be able to result in significant radiation dose reduction while maintain image quality, noticeable patchy artifacts still exist in reconstructed images. In this study, to address the problem of patchy artifacts, we proposed a median prior constrained TV regularization to retain the image quality by introducing an auxiliary vector m in register with the object. Specifically, the approximate action of m is to draw, in each iteration, an object voxel toward its own local median, aiming to improve low-dose image quality with sparse-view projection measurements. Subsequently, an alternating optimization algorithm is adopted to optimize the associative objective function. We refer to the median prior constrained TV regularization as "TV_MP" for simplicity. Experimental results on digital phantoms and clinical phantom demonstrated that the proposed TV_MP with appropriate control parameters can not only ensure a higher signal to noise ratio (SNR) of the reconstructed image, but also its resolution compared with the original TV method. Copyright © 2015 Elsevier Ltd. All rights reserved.
International Nuclear Information System (INIS)
Kim, Hyungjin; Park, Chang Min; Song, Yong Sub; Lee, Sang Min; Goo, Jin Mo
2014-01-01
Purpose: To evaluate the influence of radiation dose settings and reconstruction algorithms on the measurement accuracy and reproducibility of semi-automated pulmonary nodule volumetry. Materials and methods: CT scans were performed on a chest phantom containing various nodules (10 and 12 mm; +100, −630 and −800 HU) at 120 kVp with tube current–time settings of 10, 20, 50, and 100 mAs. Each CT was reconstructed using filtered back projection (FBP), iDose 4 and iterative model reconstruction (IMR). Semi-automated volumetry was performed by two radiologists using commercial volumetry software for nodules at each CT dataset. Noise, contrast-to-noise ratio and signal-to-noise ratio of CT images were also obtained. The absolute percentage measurement errors and differences were then calculated for volume and mass. The influence of radiation dose and reconstruction algorithm on measurement accuracy, reproducibility and objective image quality metrics was analyzed using generalized estimating equations. Results: Measurement accuracy and reproducibility of nodule volume and mass were not significantly associated with CT radiation dose settings or reconstruction algorithms (p > 0.05). Objective image quality metrics of CT images were superior in IMR than in FBP or iDose 4 at all radiation dose settings (p < 0.05). Conclusion: Semi-automated nodule volumetry can be applied to low- or ultralow-dose chest CT with usage of a novel iterative reconstruction algorithm without losing measurement accuracy and reproducibility
Energy Technology Data Exchange (ETDEWEB)
Kim, Hyungjin, E-mail: khj.snuh@gmail.com [Department of Radiology, Seoul National University College of Medicine, Institute of Radiation Medicine, Seoul National University Medical Research Center, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Park, Chang Min, E-mail: cmpark@radiol.snu.ac.kr [Department of Radiology, Seoul National University College of Medicine, Institute of Radiation Medicine, Seoul National University Medical Research Center, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Cancer Research Institute, Seoul National University, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Song, Yong Sub, E-mail: terasong@gmail.com [Department of Radiology, Seoul National University College of Medicine, Institute of Radiation Medicine, Seoul National University Medical Research Center, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Lee, Sang Min, E-mail: sangmin.lee.md@gmail.com [Department of Radiology, Seoul National University College of Medicine, Institute of Radiation Medicine, Seoul National University Medical Research Center, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Goo, Jin Mo, E-mail: jmgoo@plaza.snu.ac.kr [Department of Radiology, Seoul National University College of Medicine, Institute of Radiation Medicine, Seoul National University Medical Research Center, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of); Cancer Research Institute, Seoul National University, 101, Daehangno, Jongno-gu, Seoul 110-744 (Korea, Republic of)
2014-05-15
Purpose: To evaluate the influence of radiation dose settings and reconstruction algorithms on the measurement accuracy and reproducibility of semi-automated pulmonary nodule volumetry. Materials and methods: CT scans were performed on a chest phantom containing various nodules (10 and 12 mm; +100, −630 and −800 HU) at 120 kVp with tube current–time settings of 10, 20, 50, and 100 mAs. Each CT was reconstructed using filtered back projection (FBP), iDose{sup 4} and iterative model reconstruction (IMR). Semi-automated volumetry was performed by two radiologists using commercial volumetry software for nodules at each CT dataset. Noise, contrast-to-noise ratio and signal-to-noise ratio of CT images were also obtained. The absolute percentage measurement errors and differences were then calculated for volume and mass. The influence of radiation dose and reconstruction algorithm on measurement accuracy, reproducibility and objective image quality metrics was analyzed using generalized estimating equations. Results: Measurement accuracy and reproducibility of nodule volume and mass were not significantly associated with CT radiation dose settings or reconstruction algorithms (p > 0.05). Objective image quality metrics of CT images were superior in IMR than in FBP or iDose{sup 4} at all radiation dose settings (p < 0.05). Conclusion: Semi-automated nodule volumetry can be applied to low- or ultralow-dose chest CT with usage of a novel iterative reconstruction algorithm without losing measurement accuracy and reproducibility.
Kim, Hyungjin; Park, Chang Min; Song, Yong Sub; Lee, Sang Min; Goo, Jin Mo
2014-05-01
To evaluate the influence of radiation dose settings and reconstruction algorithms on the measurement accuracy and reproducibility of semi-automated pulmonary nodule volumetry. CT scans were performed on a chest phantom containing various nodules (10 and 12mm; +100, -630 and -800HU) at 120kVp with tube current-time settings of 10, 20, 50, and 100mAs. Each CT was reconstructed using filtered back projection (FBP), iDose(4) and iterative model reconstruction (IMR). Semi-automated volumetry was performed by two radiologists using commercial volumetry software for nodules at each CT dataset. Noise, contrast-to-noise ratio and signal-to-noise ratio of CT images were also obtained. The absolute percentage measurement errors and differences were then calculated for volume and mass. The influence of radiation dose and reconstruction algorithm on measurement accuracy, reproducibility and objective image quality metrics was analyzed using generalized estimating equations. Measurement accuracy and reproducibility of nodule volume and mass were not significantly associated with CT radiation dose settings or reconstruction algorithms (p>0.05). Objective image quality metrics of CT images were superior in IMR than in FBP or iDose(4) at all radiation dose settings (pvolumetry can be applied to low- or ultralow-dose chest CT with usage of a novel iterative reconstruction algorithm without losing measurement accuracy and reproducibility. Copyright © 2014 Elsevier Ireland Ltd. All rights reserved.
Wu, Junfeng; Dai, Fang; Hu, Gang; Mou, Xuanqin
2018-04-18
Excessive radiation exposure in computed tomography (CT) scans increases the chance of developing cancer and has become a major clinical concern. Recently, statistical iterative reconstruction (SIR) with l0-norm dictionary learning regularization has been developed to reconstruct CT images from the low dose and few-view dataset in order to reduce radiation dose. Nonetheless, the sparse regularization term adopted in this approach is l0-norm, which cannot guarantee the global convergence of the proposed algorithm. To address this problem, in this study we introduced the l1-norm dictionary learning penalty into SIR framework for low dose CT image reconstruction, and developed an alternating minimization algorithm to minimize the associated objective function, which transforms CT image reconstruction problem into a sparse coding subproblem and an image updating subproblem. During the image updating process, an efficient model function approach based on balancing principle is applied to choose the regularization parameters. The proposed alternating minimization algorithm was evaluated first using real projection data of a sheep lung CT perfusion and then using numerical simulation based on sheep lung CT image and chest image. Both visual assessment and quantitative comparison using terms of root mean square error (RMSE) and structural similarity (SSIM) index demonstrated that the new image reconstruction algorithm yielded similar performance with l0-norm dictionary learning penalty and outperformed the conventional filtered backprojection (FBP) and total variation (TV) minimization algorithms.
Energy Technology Data Exchange (ETDEWEB)
Melli, Seyed Ali, E-mail: sem649@mail.usask.ca [Department of Electrical and Computer Engineering, University of Saskatchewan, Saskatoon, SK (Canada); Wahid, Khan A. [Department of Electrical and Computer Engineering, University of Saskatchewan, Saskatoon, SK (Canada); Babyn, Paul [Department of Medical Imaging, University of Saskatchewan, Saskatoon, SK (Canada); Montgomery, James [College of Medicine, University of Saskatchewan, Saskatoon, SK (Canada); Snead, Elisabeth [Western College of Veterinary Medicine, University of Saskatchewan, Saskatoon, SK (Canada); El-Gayed, Ali [College of Medicine, University of Saskatchewan, Saskatoon, SK (Canada); Pettitt, Murray; Wolkowski, Bailey [College of Agriculture and Bioresources, University of Saskatchewan, Saskatoon, SK (Canada); Wesolowski, Michal [Department of Medical Imaging, University of Saskatchewan, Saskatoon, SK (Canada)
2016-01-11
Synchrotron source propagation-based X-ray phase contrast computed tomography is increasingly used in pre-clinical imaging. However, it typically requires a large number of projections, and subsequently a large radiation dose, to produce high quality images. To improve the applicability of this imaging technique, reconstruction algorithms that can reduce the radiation dose and acquisition time without degrading image quality are needed. The proposed research focused on using a novel combination of Douglas–Rachford splitting and randomized Kaczmarz algorithms to solve large-scale total variation based optimization in a compressed sensing framework to reconstruct 2D images from a reduced number of projections. Visual assessment and quantitative performance evaluations of a synthetic abdomen phantom and real reconstructed image of an ex-vivo slice of canine prostate tissue demonstrate that the proposed algorithm is competitive in reconstruction process compared with other well-known algorithms. An additional potential benefit of reducing the number of projections would be reduction of time for motion artifact to occur if the sample moves during image acquisition. Use of this reconstruction algorithm to reduce the required number of projections in synchrotron source propagation-based X-ray phase contrast computed tomography is an effective form of dose reduction that may pave the way for imaging of in-vivo samples.
International Nuclear Information System (INIS)
Cao, Z.J.; Tsui, B.M.
1993-01-01
Conventional single-orbit cone beam tomography presents special problems. They include incomplete sampling and inadequate three-dimensional (3D) reconstruction algorithm. The commonly used Feldkamp reconstruction algorithm simply extends the two-dimensional (2D) fan beam algorithm to 3D cone beam geometry. A truly 3D reconstruction formulation has been derived for the single-orbit cone beam SPECT based on the 3D Fourier slice theorem. In the formulation, a nonstationary filter which depends on the distance from the central plane of the cone beam was derived. The filter is applied to the 2D projection data in directions along and normal to the axis-of-rotation. The 3D reconstruction algorithm with the nonstationary filter was evaluated using both computer simulation and experimental measurements. Significant improvement in image quality was demonstrated in terms of decreased artifacts and distortions in cone beam reconstructed images. However, compared with the Feldkamp algorithm, a five-fold increase in processing time is required. Further improvement in image quality needs complete sampling in frequency space
Algorithms for Reconstruction of Undersampled Atomic Force Microscopy Images Supplementary Material
DEFF Research Database (Denmark)
2017-01-01
Two Jupyter Notebooks showcasing reconstructions of undersampled atomic force microscopy images. The reconstructions were obtained using a variety of interpolation and reconstruction methods.......Two Jupyter Notebooks showcasing reconstructions of undersampled atomic force microscopy images. The reconstructions were obtained using a variety of interpolation and reconstruction methods....
Directory of Open Access Journals (Sweden)
Helle Precht
2016-12-01
Full Text Available Background Coronary computed tomography angiography (CCTA requires high spatial and temporal resolution, increased low contrast resolution for the assessment of coronary artery stenosis, plaque detection, and/or non-coronary pathology. Therefore, new reconstruction algorithms, particularly iterative reconstruction (IR techniques, have been developed in an attempt to improve image quality with no cost in radiation exposure. Purpose To evaluate whether adaptive statistical iterative reconstruction (ASIR enhances perceived image quality in CCTA compared to filtered back projection (FBP. Material and Methods Thirty patients underwent CCTA due to suspected coronary artery disease. Images were reconstructed using FBP, 30% ASIR, and 60% ASIR. Ninety image sets were evaluated by five observers using the subjective visual grading analysis (VGA and assessed by proportional odds modeling. Objective quality assessment (contrast, noise, and the contrast-to-noise ratio [CNR] was analyzed with linear mixed effects modeling on log-transformed data. The need for ethical approval was waived by the local ethics committee as the study only involved anonymously collected clinical data. Results VGA showed significant improvements in sharpness by comparing FBP with ASIR, resulting in odds ratios of 1.54 for 30% ASIR and 1.89 for 60% ASIR (P = 0.004. The objective measures showed significant differences between FBP and 60% ASIR (P < 0.0001 for noise, with an estimated ratio of 0.82, and for CNR, with an estimated ratio of 1.26. Conclusion ASIR improved the subjective image quality of parameter sharpness and, objectively, reduced noise and increased CNR.
Electron bunch profile reconstruction in the few fs regime using coherent Smith-Purcell radiation
International Nuclear Information System (INIS)
Bartolini, R; Delerue, N; Doucas, G; Reichold, A; Clarke, C
2012-01-01
Advanced accelerators for fourth generation light sources based on high brightness linacs or laser-driven wakefield accelerators will operate with intense, highly relativistic electron bunches that are only a few fs long. Diagnostic techniques for the determination of temporal profile of such bunches are required to be non invasive, single shot, economic and with the required resolution in the fs regime. The use of a radiative process such as coherent Smith-Purcell radiation (SPR), is particularly promising with this respect. In this technique the beam is made to radiate a small amount of electromagnetic radiation and the temporal profile is reconstructed from the measured spectral distribution of the radiation. We summarise the advantages of SPR and present the design parameters and preliminary results of the experiments at the FACET facility at SLAC. We also discuss a new approach to the problem of the recovery of the 'missing phase', which is essential for the accurate reconstruction of the temporal bunch profile.
Electron Bunch Profile Reconstruction in the Few fs Regime using Coherent Smith-Purcell Radiation
International Nuclear Information System (INIS)
Bartolini, R.; Clarke, C.; Delerue, N.; Doucasa, G.; Reicholda, A.
2012-01-01
Advanced accelerators for fourth generation light sources based on high brightness linacs or laser-driven wakefield accelerators will operate with intense, highly relativistic electron bunches that are only a few fs long. Diagnostic techniques for the determination of temporal profile of such bunches are required to be non invasive, single shot, economic and with the required resolution in the fs regime. The use of a radiative process such as coherent Smith-Purcell radiation (SPR), is particularly promising with this respect. In this technique the beam is made to radiate a small amount of electromagnetic radiation and the temporal profile is reconstructed from the measured spectral distribution of the radiation. We summarise the advantages of SPR and present the design parameters and preliminary results of the experiments at the FACET facility at SLAC. We also discuss a new approach to the problem of the recovery of the 'missing phase', which is essential for the accurate reconstruction of the temporal bunch profile.
Huang, Weilin; Wang, Runqiu; Chen, Yangkang
2018-05-01
Microseismic signal is typically weak compared with the strong background noise. In order to effectively detect the weak signal in microseismic data, we propose a mathematical morphology based approach. We decompose the initial data into several morphological multiscale components. For detection of weak signal, a non-stationary weighting operator is proposed and introduced into the process of reconstruction of data by morphological multiscale components. The non-stationary weighting operator can be obtained by solving an inversion problem. The regularized non-stationary method can be understood as a non-stationary matching filtering method, where the matching filter has the same size as the data to be filtered. In this paper, we provide detailed algorithmic descriptions and analysis. The detailed algorithm framework, parameter selection and computational issue for the regularized non-stationary morphological reconstruction (RNMR) method are presented. We validate the presented method through a comprehensive analysis through different data examples. We first test the proposed technique using a synthetic data set. Then the proposed technique is applied to a field project, where the signals induced from hydraulic fracturing are recorded by 12 three-component geophones in a monitoring well. The result demonstrates that the RNMR can improve the detectability of the weak microseismic signals. Using the processed data, the short-term-average over long-term average picking algorithm and Geiger's method are applied to obtain new locations of microseismic events. In addition, we show that the proposed RNMR method can be used not only in microseismic data but also in reflection seismic data to detect the weak signal. We also discussed the extension of RNMR from 1-D to 2-D or a higher dimensional version.
Reconstructing Dynamic Promoter Activity Profiles from Reporter Gene Data.
Kannan, Soumya; Sams, Thomas; Maury, Jérôme; Workman, Christopher T
2018-03-16
Accurate characterization of promoter activity is important when designing expression systems for systems biology and metabolic engineering applications. Promoters that respond to changes in the environment enable the dynamic control of gene expression without the necessity of inducer compounds, for example. However, the dynamic nature of these processes poses challenges for estimating promoter activity. Most experimental approaches utilize reporter gene expression to estimate promoter activity. Typically the reporter gene encodes a fluorescent protein that is used to infer a constant promoter activity despite the fact that the observed output may be dynamic and is a number of steps away from the transcription process. In fact, some promoters that are often thought of as constitutive can show changes in activity when growth conditions change. For these reasons, we have developed a system of ordinary differential equations for estimating dynamic promoter activity for promoters that change their activity in response to the environment that is robust to noise and changes in growth rate. Our approach, inference of dynamic promoter activity (PromAct), improves on existing methods by more accurately inferring known promoter activity profiles. This method is also capable of estimating the correct scale of promoter activity and can be applied to quantitative data sets to estimate quantitative rates.
Energy Technology Data Exchange (ETDEWEB)
Krohn, Thomas [RWTH University Hospital Aachen, Department of Nuclear Medicine, Aachen (Germany); Ulm University, Department of Nuclear Medicine, Ulm (Germany); Birmes, Anita; Winz, Oliver H.; Drude, Natascha I. [RWTH University Hospital Aachen, Department of Nuclear Medicine, Aachen (Germany); Mottaghy, Felix M. [RWTH University Hospital Aachen, Department of Nuclear Medicine, Aachen (Germany); Maastricht UMC+, Department of Nuclear Medicine, Maastricht (Netherlands); Behrendt, Florian F. [RWTH University Hospital Aachen, Department of Nuclear Medicine, Aachen (Germany); Radiology Institute ' ' Aachen Land' ' , Wuerselen (Germany); Verburg, Frederik A. [RWTH University Hospital Aachen, Department of Nuclear Medicine, Aachen (Germany); University Hospital Giessen and Marburg, Department of Nuclear Medicine, Marburg (Germany)
2017-04-15
To investigate whether the numbers of lymph node metastases and coeliac ganglia delineated on [{sup 68}Ga]PSMA-HBED-CC PET/CT scans differ among datasets generated using different reconstruction algorithms. Data were constructed using the BLOB-OS-TF, BLOB-OS and 3D-RAMLA algorithms. All reconstructions were assessed by two nuclear medicine physicians for the number of pelvic/paraaortal lymph node metastases as well the number of coeliac ganglia. Standardized uptake values (SUV) were also calculated in different regions. At least one [{sup 68}Ga]PSMA-HBED-CC PET/CT-positive pelvic or paraaortal lymph node metastasis was found in 49 and 35 patients using the BLOB-OS-TF algorithm, in 42 and 33 patients using the BLOB-OS algorithm, and in 41 and 31 patients using the 3D-RAMLA algorithm, respectively, and a positive ganglion was found in 92, 59 and 24 of 100 patients using the three algorithms, respectively. Quantitatively, the SUVmean and SUVmax were significantly higher with the BLOB-OS algorithm than with either the BLOB-OS-TF or the 3D-RAMLA algorithm in all measured regions (p < 0.001 for all comparisons). The differences between the SUVs with the BLOB-OS-TF- and 3D-RAMLA algorithms were not significant in the aorta (SUVmean, p = 0.93; SUVmax, p = 0.97) but were significant in all other regions (p < 0.001 in all cases). The SUVmean ganglion/gluteus ratio was significantly higher with the BLOB-OS-TF algorithm than with either the BLOB-OS or the 3D-RAMLA algorithm and was significantly higher with the BLOB-OS than with the 3D-RAMLA algorithm (p < 0.001 in all cases). The results of [{sup 68}Ga]PSMA-HBED-CC PET/CT are affected by the reconstruction algorithm used. The highest number of lesions and physiological structures will be visualized using a modern algorithm employing time-of-flight information. (orig.)
International Nuclear Information System (INIS)
Viana, R.S.; Yoriyaz, H.; Santos, A.
2011-01-01
The Expectation-Maximization (E-M) algorithm is an iterative computational method for maximum likelihood (M-L) estimates, useful in a variety of incomplete-data problems. Due to its stochastic nature, one of the most relevant applications of E-M algorithm is the reconstruction of emission tomography images. In this paper, the statistical formulation of the E-M algorithm was applied to the in vivo spectrographic imaging of stable isotopes called Neutron Stimulated Emission Computed Tomography (NSECT). In the process of E-M algorithm iteration, the conditional probability distribution plays a very important role to achieve high quality image. This present work proposes an alternative methodology for the generation of the conditional probability distribution associated to the E-M reconstruction algorithm, using the Monte Carlo code MCNP5 and with the application of the reciprocity theorem. (author)
Energy Technology Data Exchange (ETDEWEB)
Viana, R.S.; Yoriyaz, H.; Santos, A., E-mail: rodrigossviana@gmail.com, E-mail: hyoriyaz@ipen.br, E-mail: asantos@ipen.br [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2011-07-01
The Expectation-Maximization (E-M) algorithm is an iterative computational method for maximum likelihood (M-L) estimates, useful in a variety of incomplete-data problems. Due to its stochastic nature, one of the most relevant applications of E-M algorithm is the reconstruction of emission tomography images. In this paper, the statistical formulation of the E-M algorithm was applied to the in vivo spectrographic imaging of stable isotopes called Neutron Stimulated Emission Computed Tomography (NSECT). In the process of E-M algorithm iteration, the conditional probability distribution plays a very important role to achieve high quality image. This present work proposes an alternative methodology for the generation of the conditional probability distribution associated to the E-M reconstruction algorithm, using the Monte Carlo code MCNP5 and with the application of the reciprocity theorem. (author)
Directory of Open Access Journals (Sweden)
Logsdon Benjamin A
2012-04-01
Full Text Available Abstract Background We propose a novel variational Bayes network reconstruction algorithm to extract the most relevant disease factors from high-throughput genomic data-sets. Our algorithm is the only scalable method for regularized network recovery that employs Bayesian model averaging and that can internally estimate an appropriate level of sparsity to ensure few false positives enter the model without the need for cross-validation or a model selection criterion. We use our algorithm to characterize the effect of genetic markers and liver gene expression traits on mouse obesity related phenotypes, including weight, cholesterol, glucose, and free fatty acid levels, in an experiment previously used for discovery and validation of network connections: an F2 intercross between the C57BL/6 J and C3H/HeJ mouse strains, where apolipoprotein E is null on the background. Results We identified eleven genes, Gch1, Zfp69, Dlgap1, Gna14, Yy1, Gabarapl1, Folr2, Fdft1, Cnr2, Slc24a3, and Ccl19, and a quantitative trait locus directly connected to weight, glucose, cholesterol, or free fatty acid levels in our network. None of these genes were identified by other network analyses of this mouse intercross data-set, but all have been previously associated with obesity or related pathologies in independent studies. In addition, through both simulations and data analysis we demonstrate that our algorithm achieves superior performance in terms of power and type I error control than other network recovery algorithms that use the lasso and have bounds on type I error control. Conclusions Our final network contains 118 previously associated and novel genes affecting weight, cholesterol, glucose, and free fatty acid levels that are excellent obesity risk candidates.
Assist feature printability prediction by 3-D resist profile reconstruction
Zheng, Xin; Huang, Jensheng; Chin, Fook; Kazarian, Aram; Kuo, Chun-Chieh
2012-06-01
properties may then be used to optimize the printability vs. efficacy of an SRAF either prior to or during an Optical Proximity Correction (OPC) run. The process models that are used during OPC have never been able to reliably predict which SRAFs will print. This appears to be due to the fact that OPC process models are generally created using data that does not include printed subresolution patterns. An enhancement to compact modeling capability to predict Assist Features (AF) printability is developed and discussed. A hypsometric map representing 3-D resist profile was built by applying a first principle approximation to estimate the "energy loss" from the resist top to bottom. Such a 3-D resist profile is an extrapolation of a well calibrated traditional OPC model without any additional information. Assist features are detected at either top of resist (dark field) or bottom of resist (bright field). Such detection can be done by just extracting top or bottom resist models from our 3-D resist model. There is no measurement of assist features needed when we build AF but it can be included if interested but focusing on resist calibration to account for both exposure dosage and focus change sensitivities. This approach significantly increases resist model's capability for predicting printed SRAF accuracy. And we don't need to calibrate an SRAF model in addition to the OPC model. Without increase in computation time, this compact model can draw assist feature contour with real placement and size at any vertical plane. The result is compared and validated with 3-D rigorous modeling as well as SEM images. Since this method does not change any form of compact modeling, it can be integrated into current MBAF solutions without any additional work.
A comparison of semiglobal and local dense matching algorithms for surface reconstruction
Directory of Open Access Journals (Sweden)
E. Dall'Asta
2014-06-01
Full Text Available Encouraged by the growing interest in automatic 3D image-based reconstruction, the development and improvement of robust stereo matching techniques is one of the most investigated research topic of the last years in photogrammetry and computer vision. The paper is focused on the comparison of some stereo matching algorithms (local and global which are very popular both in photogrammetry and computer vision. In particular, the Semi-Global Matching (SGM, which realizes a pixel-wise matching and relies on the application of consistency constraints during the matching cost aggregation, will be discussed. The results of some tests performed on real and simulated stereo image datasets, evaluating in particular the accuracy of the obtained digital surface models, will be presented. Several algorithms and different implementation are considered in the comparison, using freeware software codes like MICMAC and OpenCV, commercial software (e.g. Agisoft PhotoScan and proprietary codes implementing Least Square e Semi-Global Matching algorithms. The comparisons will also consider the completeness and the level of detail within fine structures, and the reliability and repeatability of the obtainable data.
A comparison of semiglobal and local dense matching algorithms for surface reconstruction
Dall'Asta, E.; Roncella, R.
2014-06-01
Encouraged by the growing interest in automatic 3D image-based reconstruction, the development and improvement of robust stereo matching techniques is one of the most investigated research topic of the last years in photogrammetry and computer vision. The paper is focused on the comparison of some stereo matching algorithms (local and global) which are very popular both in photogrammetry and computer vision. In particular, the Semi-Global Matching (SGM), which realizes a pixel-wise matching and relies on the application of consistency constraints during the matching cost aggregation, will be discussed. The results of some tests performed on real and simulated stereo image datasets, evaluating in particular the accuracy of the obtained digital surface models, will be presented. Several algorithms and different implementation are considered in the comparison, using freeware software codes like MICMAC and OpenCV, commercial software (e.g. Agisoft PhotoScan) and proprietary codes implementing Least Square e Semi-Global Matching algorithms. The comparisons will also consider the completeness and the level of detail within fine structures, and the reliability and repeatability of the obtainable data.
Chen, Yihang; Xiao, Chijie; Yang, Xiaoyi; Wang, Tianbo; Xu, Tianchao; Yu, Yi; Xu, Min; Wang, Long; Lin, Chen; Wang, Xiaogang
2017-10-01
The Laser-driven Ion beam trace probe (LITP) is a new diagnostic method for measuring poloidal magnetic field (Bp) and radial electric field (Er) in tokamaks. LITP injects a laser-driven ion beam into the tokamak, and Bp and Er profiles can be reconstructed using tomography methods. A reconstruction code has been developed to validate the LITP theory, and both 2D reconstruction of Bp and simultaneous reconstruction of Bp and Er have been attained. To reconstruct from experimental data with noise, Maximum Entropy and Gaussian-Bayesian tomography methods were applied and improved according to the characteristics of the LITP problem. With these improved methods, a reconstruction error level below 15% has been attained with a data noise level of 10%. These methods will be further tested and applied in the following LITP experiments. Supported by the ITER-CHINA program 2015GB120001, CHINA MOST under 2012YQ030142 and National Natural Science Foundation Abstract of China under 11575014 and 11375053.
Banjak, Hussein; Grenier, Thomas; Epicier, Thierry; Koneti, Siddardha; Roiban, Lucian; Gay, Anne-Sophie; Magnin, Isabelle; Peyrin, Françoise; Maxim, Voichita
2018-06-01
Fast tomography in Environmental Transmission Electron Microscopy (ETEM) is of a great interest for in situ experiments where it allows to observe 3D real-time evolution of nanomaterials under operating conditions. In this context, we are working on speeding up the acquisition step to a few seconds mainly with applications on nanocatalysts. In order to accomplish such rapid acquisitions of the required tilt series of projections, a modern 4K high-speed camera is used, that can capture up to 100 images per second in a 2K binning mode. However, due to the fast rotation of the sample during the tilt procedure, noise and blur effects may occur in many projections which in turn would lead to poor quality reconstructions. Blurred projections make classical reconstruction algorithms inappropriate and require the use of prior information. In this work, a regularized algebraic reconstruction algorithm named SIRT-FISTA-TV is proposed. The performance of this algorithm using blurred data is studied by means of a numerical blur introduced into simulated images series to mimic possible mechanical instabilities/drifts during fast acquisitions. We also present reconstruction results from noisy data to show the robustness of the algorithm to noise. Finally, we show reconstructions with experimental datasets and we demonstrate the interest of fast tomography with an ultra-fast acquisition performed under environmental conditions, i.e. gas and temperature, in the ETEM. Compared to classically used SIRT and SART approaches, our proposed SIRT-FISTA-TV reconstruction algorithm provides higher quality tomograms allowing easier segmentation of the reconstructed volume for a better final processing and analysis. Copyright © 2018 Elsevier B.V. All rights reserved.
MO-FG-204-04: How Iterative Reconstruction Algorithms Affect the NPS of CT Images
International Nuclear Information System (INIS)
Li, G; Liu, X; Dodge, C; Jensen, C; Rong, J
2015-01-01
Purpose: To evaluate how the third generation model based iterative reconstruction (MBIR) compares with filtered back-projection (FBP), adaptive statistical iterative reconstruction (ASiR), and the second generation MBIR based on noise power spectrum (NPS) analysis over a wide range of clinically applicable dose levels. Methods: The Catphan 600 CTP515 module, surrounded by an oval, fat-equivalent ring to mimic patient size/shape, was scanned on a GE HD750 CT scanner at 1, 2, 3, 6, 12 and 19mGy CTDIvol levels with typical patient scan parameters: 120kVp, 0.8s, 40mm beam width, large SFOV, 0.984 pitch and reconstructed thickness 2.5mm (VEO3.0: Abd/Pelvis with Texture and NR05). At each CTDIvol level, 10 repeated scans were acquired for achieving sufficient data sampling. The images were reconstructed using Standard kernel with FBP; 20%, 40% and 70% ASiR; and two versions of MBIR (VEO2.0 and 3.0). For evaluating the effect of the ROI spatial location to the Result of NPS, 4 ROI groups were categorized based on their distances from the center of the phantom. Results: VEO3.0 performed inferiorly comparing to VEO2.0 over all dose levels. On the other hand, at low dose levels (less than 3 mGy), it clearly outperformed ASiR and FBP, in NPS values. Therefore, the lower the dose level, the relative performance of MBIR improves. However, the shapes of the NPS show substantial differences in horizontal and vertical sampling dimensions. These differences may determine the characteristics of the noise/texture features in images, and hence, play an important role in image interpretation. Conclusion: The third generation MBIR did not improve over the second generation MBIR in term of NPS analysis. The overall performance of both versions of MBIR improved as compared to other reconstruction algorithms when dose was reduced. The shapes of the NPS curves provided additional value for future characterization of the image noise/texture features
MO-FG-204-04: How Iterative Reconstruction Algorithms Affect the NPS of CT Images
Energy Technology Data Exchange (ETDEWEB)
Li, G; Liu, X; Dodge, C; Jensen, C; Rong, J [UT MD Anderson Cancer Center, Houston, TX (United States)
2015-06-15
Purpose: To evaluate how the third generation model based iterative reconstruction (MBIR) compares with filtered back-projection (FBP), adaptive statistical iterative reconstruction (ASiR), and the second generation MBIR based on noise power spectrum (NPS) analysis over a wide range of clinically applicable dose levels. Methods: The Catphan 600 CTP515 module, surrounded by an oval, fat-equivalent ring to mimic patient size/shape, was scanned on a GE HD750 CT scanner at 1, 2, 3, 6, 12 and 19mGy CTDIvol levels with typical patient scan parameters: 120kVp, 0.8s, 40mm beam width, large SFOV, 0.984 pitch and reconstructed thickness 2.5mm (VEO3.0: Abd/Pelvis with Texture and NR05). At each CTDIvol level, 10 repeated scans were acquired for achieving sufficient data sampling. The images were reconstructed using Standard kernel with FBP; 20%, 40% and 70% ASiR; and two versions of MBIR (VEO2.0 and 3.0). For evaluating the effect of the ROI spatial location to the Result of NPS, 4 ROI groups were categorized based on their distances from the center of the phantom. Results: VEO3.0 performed inferiorly comparing to VEO2.0 over all dose levels. On the other hand, at low dose levels (less than 3 mGy), it clearly outperformed ASiR and FBP, in NPS values. Therefore, the lower the dose level, the relative performance of MBIR improves. However, the shapes of the NPS show substantial differences in horizontal and vertical sampling dimensions. These differences may determine the characteristics of the noise/texture features in images, and hence, play an important role in image interpretation. Conclusion: The third generation MBIR did not improve over the second generation MBIR in term of NPS analysis. The overall performance of both versions of MBIR improved as compared to other reconstruction algorithms when dose was reduced. The shapes of the NPS curves provided additional value for future characterization of the image noise/texture features.
Cui, Xiaoming; Li, Tao; Li, Xin; Zhou, Weihua
2015-05-01
The aim of this study was to evaluate the in vivo performance of four image reconstruction algorithms in a high-definition CT (HDCT) scanner with improved spatial resolution for the evaluation of coronary artery stents and intrastent lumina. Thirty-nine consecutive patients with a total of 71 implanted coronary stents underwent coronary CT angiography (CCTA) on a HDCT (Discovery CT 750 HD; GE Healthcare) with the high-resolution scanning mode. Four different reconstruction algorithms (HD-stand, HD-detail; HD-stand-plus; HD-detail-plus) were applied to reconstruct the stented coronary arteries. Image quality for stent characterization was assessed. Image noise and intrastent luminal diameter were measured. The relationship between the measurement of inner stent diameter (ISD) and the true stent diameter (TSD) and stent type were analysed. The stent-dedicated kernel (HD-detail) offered the highest percentage (53.5%) of good image quality for stent characterization and the highest ratio (68.0±8.4%) of visible stent lumen/true stent lumen for luminal diameter measurement at the expense of an increased overall image noise. The Pearson correlation coefficient between the ISD and TSD measurement and spearman correlation coefficient between the ISD measurement and stent type were 0.83 and 0.48, respectively. Compared with standard reconstruction algorithms, high-definition CT imaging technique with dedicated high-resolution reconstruction algorithm provides more accurate stent characterization and intrastent luminal diameter measurement. Copyright © 2015 Elsevier Ireland Ltd. All rights reserved.
International Nuclear Information System (INIS)
Sunaguchi, Naoki; Yuasa, Tetsuya; Gupta, Rajiv; Ando, Masami
2015-01-01
The main focus of this paper is reconstruction of tomographic phase-contrast image from a set of projections. We propose an efficient reconstruction algorithm for differential phase-contrast computed tomography that can considerably reduce the number of projections required for reconstruction. The key result underlying this research is a projection theorem that states that the second derivative of the projection set is linearly related to the Laplacian of the tomographic image. The proposed algorithm first reconstructs the Laplacian image of the phase-shift distribution from the second-derivative of the projections using total variation regularization. The second step is to obtain the phase-shift distribution by solving a Poisson equation whose source is the Laplacian image previously reconstructed under the Dirichlet condition. We demonstrate the efficacy of this algorithm using both synthetically generated simulation data and projection data acquired experimentally at a synchrotron. The experimental phase data were acquired from a human coronary artery specimen using dark-field-imaging optics pioneered by our group. Our results demonstrate that the proposed algorithm can reduce the number of projections to approximately 33% as compared with the conventional filtered backprojection method, without any detrimental effect on the image quality
Triggerless Readout with Time and Amplitude Reconstruction of Event Based on Deconvolution Algorithm
International Nuclear Information System (INIS)
Kulis, S.; Idzik, M.
2011-01-01
In future linear colliders like CLIC, where the period between the bunch crossings is in a sub-nanoseconds range ( 500 ps), an appropriate detection technique with triggerless signal processing is needed. In this work we discuss a technique, based on deconvolution algorithm, suitable for time and amplitude reconstruction of an event. In the implemented method the output of a relatively slow shaper (many bunch crossing periods) is sampled and digitalised in an ADC and then the deconvolution procedure is applied to digital data. The time of an event can be found with a precision of few percent of sampling time. The signal to noise ratio is only slightly decreased after passing through the deconvolution filter. The performed theoretical and Monte Carlo studies are confirmed by the results of preliminary measurements obtained with the dedicated system comprising of radiation source, silicon sensor, front-end electronics, ADC and further digital processing implemented on a PC computer. (author)
Complex amplitude reconstruction by iterative amplitude-phase retrieval algorithm with reference
Shen, Cheng; Guo, Cheng; Tan, Jiubin; Liu, Shutian; Liu, Zhengjun
2018-06-01
Multi-image iterative phase retrieval methods have been successfully applied in plenty of research fields due to their simple but efficient implementation. However, there is a mismatch between the measurement of the first long imaging distance and the sequential interval. In this paper, an amplitude-phase retrieval algorithm with reference is put forward without additional measurements or priori knowledge. It gets rid of measuring the first imaging distance. With a designed update formula, it significantly raises the convergence speed and the reconstruction fidelity, especially in phase retrieval. Its superiority over the original amplitude-phase retrieval (APR) method is validated by numerical analysis and experiments. Furthermore, it provides a conceptual design of a compact holographic image sensor, which can achieve numerical refocusing easily.
International Nuclear Information System (INIS)
Qin, M; Chen, D Y; Wang, L L; Yu, X Y
2006-01-01
The subject investigated in this paper is the ECT system of 8-electrode oil-water two-phase flow, and the measuring principle is analysed. In ART image-reconstruction algorithm, an adaptive threshold image reconstruction is presented to improve quality of image reconstruction and calculating accuracy of concentration, and generally the measurement error is about 1%. Such method can well solve many defects that other measurement methods may have, such as slow speed, high cost, and poor security and so on. Therefore, it offers a new method for the concentration measurement of oil-water two-phase flow
International Nuclear Information System (INIS)
Kollár, László E; Lucas, Gary P; Zhang, Zhichao
2014-01-01
An analytical method is developed for the reconstruction of velocity profiles using measured potential distributions obtained around the boundary of a multi-electrode electromagnetic flow meter (EMFM). The method is based on the discrete Fourier transform (DFT), and is implemented in Matlab. The method assumes the velocity profile in a section of a pipe as a superposition of polynomials up to sixth order. Each polynomial component is defined along a specific direction in the plane of the pipe section. For a potential distribution obtained in a uniform magnetic field, this direction is not unique for quadratic and higher-order components; thus, multiple possible solutions exist for the reconstructed velocity profile. A procedure for choosing the optimum velocity profile is proposed. It is applicable for single-phase or two-phase flows, and requires measurement of the potential distribution in a non-uniform magnetic field. The potential distribution in this non-uniform magnetic field is also calculated for the possible solutions using weight values. Then, the velocity profile with the calculated potential distribution which is closest to the measured one provides the optimum solution. The reliability of the method is first demonstrated by reconstructing an artificial velocity profile defined by polynomial functions. Next, velocity profiles in different two-phase flows, based on results from the literature, are used to define the input velocity fields. In all cases, COMSOL Multiphysics is used to model the physical specifications of the EMFM and to simulate the measurements; thus, COMSOL simulations produce the potential distributions on the internal circumference of the flow pipe. These potential distributions serve as inputs for the analytical method. The reconstructed velocity profiles show satisfactory agreement with the input velocity profiles. The method described in this paper is most suitable for stratified flows and is not applicable to axisymmetric flows in
SU-F-T-441: Dose Calculation Accuracy in CT Images Reconstructed with Artifact Reduction Algorithm
Energy Technology Data Exchange (ETDEWEB)
Ng, C; Chan, S; Lee, F; Ngan, R [Queen Elizabeth Hospital (Hong Kong); Lee, V [University of Hong Kong, Hong Kong, HK (Hong Kong)
2016-06-15
Purpose: Accuracy of radiotherapy dose calculation in patients with surgical implants is complicated by two factors. First is the accuracy of CT number, second is the dose calculation accuracy. We compared measured dose with dose calculated on CT images reconstructed with FBP and an artifact reduction algorithm (OMAR, Philips) for a phantom with high density inserts. Dose calculation were done with Varian AAA and AcurosXB. Methods: A phantom was constructed with solid water in which 2 titanium or stainless steel rods could be inserted. The phantom was scanned with the Philips Brillance Big Bore CT. Image reconstruction was done with FBP and OMAR. Two 6 MV single field photon plans were constructed for each phantom. Radiochromic films were placed at different locations to measure the dose deposited. One plan has normal incidence on the titanium/steel rods. In the second plan, the beam is at almost glancing incidence on the metal rods. Measurements were then compared with dose calculated with AAA and AcurosXB. Results: The use of OMAR images slightly improved the dose calculation accuracy. The agreement between measured and calculated dose was best with AXB and image reconstructed with OMAR. Dose calculated on titanium phantom has better agreement with measurement. Large discrepancies were seen at points directly above and below the high density inserts. Both AAA and AXB underestimated the dose directly above the metal surface, while overestimated the dose below the metal surface. Doses measured downstream of metal were all within 3% of calculated values. Conclusion: When doing treatment planning for patients with metal implants, care must be taken to acquire correct CT images to improve dose calculation accuracy. Moreover, great discrepancies in measured and calculated dose were observed at metal/tissue interface. Care must be taken in estimating the dose in critical structures that come into contact with metals.
Fast cross-projection algorithm for reconstruction of seeds in prostate brachytherapy
International Nuclear Information System (INIS)
Narayanan, Sreeram; Cho, Paul S.; Marks, Robert J. II
2002-01-01
A fast method of seed matching and reconstruction in prostrate brachytherapy is proposed. Previous approaches have required all seeds to be matched with all other seeds in other projections. The fast cross-projection algorithm for the reconstruction of seeds (Fast-CARS) allows for matching of a given seed with a subset of seeds in other projections. This subset lies in a proximal region centered about the projection of a line, connecting the seed to its source, onto other projection planes. The proposed technique permits a significant reduction in computational overhead, as measured by the required number of matching tests. The number of multiplications and additions is also vastly reduced at no trade-off in accuracy. Because of its speed, Fast-CARS can be used in applications requiring real-time performance such as intraoperative dosimetry of prostate brachytherapy. Furthermore, the proposed method makes practical the use of a larger number of views as opposed to previous techniques limited to a maximum use of three views
Superficial temporal artery flap for reconstruction of complex facial defects: A new algorithm
Directory of Open Access Journals (Sweden)
Tarek M. Elbanoby
2018-03-01
Full Text Available Background A variety of island flaps can be based on the superficial temporal artery with variable tissue composition. They can be used for defect reconstruction, cavity resurfacing, facial hair restoration, or contracture release. Methods Seventy-two patients underwent facial reconstruction using a superficial temporal artery island flap from October 2010 to October 2014. The defects had various etiologies, including trauma, burns, tumors, exposed hardware, and congenital causes. We classified the patients by indication into 5 groups: cavity resurfacing, contracture release, facial hair restoration, skin coverage, and combined. The demographic data of the patients, defect characteristics, operative procedures, postoperative results, and complications were retrospectively documented. The follow-up period ranged from 24 to 54 months. Results A total of 24 females and 48 males were included in this study. The mean age of the patients was 33.7±15.6 years. The flaps were used for contracture release in 13 cases, cavity resurfacing in 10 cases, skin coverage in 17 cases, facial hair restoration in 19 cases, and combined defects in 13 cases. No major complications were reported. Conclusion: Based on our experiences with the use of superficial temporal artery island flaps, we have developed a detailed approach for the optimal management of patients with composite facial defects. The aim of this article is to provide the reader with a systematic algorithm to use for such patients.
Compositional-prior-guided image reconstruction algorithm for multi-modality imaging
Fang, Qianqian; Moore, Richard H.; Kopans, Daniel B.; Boas, David A.
2010-01-01
The development of effective multi-modality imaging methods typically requires an efficient information fusion model, particularly when combining structural images with a complementary imaging modality that provides functional information. We propose a composition-based image segmentation method for X-ray digital breast tomosynthesis (DBT) and a structural-prior-guided image reconstruction for a combined DBT and diffuse optical tomography (DOT) breast imaging system. Using the 3D DBT images from 31 clinically measured healthy breasts, we create an empirical relationship between the X-ray intensities for adipose and fibroglandular tissue. We use this relationship to then segment another 58 healthy breast DBT images from 29 subjects into compositional maps of different tissue types. For each breast, we build a weighted-graph in the compositional space and construct a regularization matrix to incorporate the structural priors into a finite-element-based DOT image reconstruction. Use of the compositional priors enables us to fuse tissue anatomy into optical images with less restriction than when using a binary segmentation. This allows us to recover the image contrast captured by DOT but not by DBT. We show that it is possible to fine-tune the strength of the structural priors by changing a single regularization parameter. By estimating the optical properties for adipose and fibroglandular tissue using the proposed algorithm, we found the results are comparable or superior to those estimated with expert-segmentations, but does not involve the time-consuming manual selection of regions-of-interest. PMID:21258460
Development of tomographic reconstruction algorithms for the PIXE analysis of biological samples
International Nuclear Information System (INIS)
Nguyen, D.T.
2008-05-01
The development of 3-dimensional microscopy techniques offering a spatial resolution of 1 μm or less has opened a large field of investigation in Cell Biology. Amongst them, an interesting advantage of ion beam micro-tomography is its ability to give quantitative results in terms of local concentrations in a direct way, using Particle Induced X-ray Emission (PIXET) combined to Scanning Transmission Ion Microscopy (STIMT) Tomography. After a brief introduction of existing reconstruction techniques, we present the principle of the DISRA code, the most complete written so far, which is the basis of the present work. We have modified and extended the DISRA algorithm by considering the specific aspects of biologic specimens. Moreover, correction procedures were added in the code to reduce noise in the tomograms. For portability purpose, a Windows graphic interface was designed to easily enter and modify experimental parameters used in the reconstruction, and control the several steps of data reduction. Results of STIMT and PIXET experiments on reference specimens and on human cancer cells will be also presented. (author)
Super resolution reconstruction of μ-CT image of rock sample using neighbour embedding algorithm
Wang, Yuzhu; Rahman, Sheik S.; Arns, Christoph H.
2018-03-01
X-ray computed tomography (μ-CT) is considered to be the most effective way to obtain the inner structure of rock sample without destructions. However, its limited resolution hampers its ability to probe sub-micro structures which is critical for flow transportation of rock sample. In this study, we propose an innovative methodology to improve the resolution of μ-CT image using neighbour embedding algorithm where low frequency information is provided by μ-CT image itself while high frequency information is supplemented by high resolution scanning electron microscopy (SEM) image. In order to obtain prior for reconstruction, a large number of image patch pairs contain high- and low- image patches are extracted from the Gaussian image pyramid generated by SEM image. These image patch pairs contain abundant information about tomographic evolution of local porous structures under different resolution spaces. Relying on the assumption of self-similarity of porous structure, this prior information can be used to supervise the reconstruction of high resolution μ-CT image effectively. The experimental results show that the proposed method is able to achieve the state-of-the-art performance.
Fast and Easy 3D Reconstruction with the Help of Geometric Constraints and Genetic Algorithms
Annich, Afafe; El Abderrahmani, Abdellatif; Satori, Khalid
2017-09-01
The purpose of the work presented in this paper is to describe new method of 3D reconstruction from one or more uncalibrated images. This method is based on two important concepts: geometric constraints and genetic algorithms (GAs). At first, we are going to discuss the combination between bundle adjustment and GAs that we have proposed in order to improve 3D reconstruction efficiency and success. We used GAs in order to improve fitness quality of initial values that are used in the optimization problem. It will increase surely convergence rate. Extracted geometric constraints are used first to obtain an estimated value of focal length that helps us in the initialization step. Matching homologous points and constraints is used to estimate the 3D model. In fact, our new method gives us a lot of advantages: reducing the estimated parameter number in optimization step, decreasing used image number, winning time and stabilizing good quality of 3D results. At the end, without any prior information about our 3D scene, we obtain an accurate calibration of the cameras, and a realistic 3D model that strictly respects the geometric constraints defined before in an easy way. Various data and examples will be used to highlight the efficiency and competitiveness of our present approach.
Accelerating statistical image reconstruction algorithms for fan-beam x-ray CT using cloud computing
Srivastava, Somesh; Rao, A. Ravishankar; Sheinin, Vadim
2011-03-01
Statistical image reconstruction algorithms potentially offer many advantages to x-ray computed tomography (CT), e.g. lower radiation dose. But, their adoption in practical CT scanners requires extra computation power, which is traditionally provided by incorporating additional computing hardware (e.g. CPU-clusters, GPUs, FPGAs etc.) into a scanner. An alternative solution is to access the required computation power over the internet from a cloud computing service, which is orders-of-magnitude more cost-effective. This is because users only pay a small pay-as-you-go fee for the computation resources used (i.e. CPU time, storage etc.), and completely avoid purchase, maintenance and upgrade costs. In this paper, we investigate the benefits and shortcomings of using cloud computing for statistical image reconstruction. We parallelized the most time-consuming parts of our application, the forward and back projectors, using MapReduce, the standard parallelization library on clouds. From preliminary investigations, we found that a large speedup is possible at a very low cost. But, communication overheads inside MapReduce can limit the maximum speedup, and a better MapReduce implementation might become necessary in the future. All the experiments for this paper, including development and testing, were completed on the Amazon Elastic Compute Cloud (EC2) for less than $20.
A retrieval algorithm of hydrometer profile for submillimeter-wave radiometer
Liu, Yuli; Buehler, Stefan; Liu, Heguang
2017-04-01
Vertical profiles of particle microphysics perform vital functions for the estimation of climatic feedback. This paper proposes a new algorithm to retrieve the profile of the parameters of the hydrometeor(i.e., ice, snow, rain, liquid cloud, graupel) based on passive submillimeter-wave measurements. These parameters include water content and particle size. The first part of the algorithm builds the database and retrieves the integrated quantities. Database is built up by Atmospheric Radiative Transfer Simulator(ARTS), which uses atmosphere data to simulate the corresponding brightness temperature. Neural network, trained by the precalculated database, is developed to retrieve the water path for each type of particles. The second part of the algorithm analyses the statistical relationship between water path and vertical parameters profiles. Based on the strong dependence existing between vertical layers in the profiles, Principal Component Analysis(PCA) technique is applied. The third part of the algorithm uses the forward model explicitly to retrieve the hydrometeor profiles. Cost function is calculated in each iteration, and Differential Evolution(DE) algorithm is used to adjust the parameter values during the evolutionary process. The performance of this algorithm is planning to be verified for both simulation database and measurement data, by retrieving profiles in comparison with the initial one. Results show that this algorithm has the ability to retrieve the hydrometeor profiles efficiently. The combination of ARTS and optimization algorithm can get much better results than the commonly used database approach. Meanwhile, the concept that ARTS can be used explicitly in the retrieval process shows great potential in providing solution to other retrieval problems.
Maximum entropy based reconstruction of soft X ray emissivity profiles in W7-AS
International Nuclear Information System (INIS)
Ertl, K.; Linden, W. von der; Dose, V.; Weller, A.
1996-01-01
The reconstruction of 2-D emissivity profiles from soft X ray tomography measurements constitutes a highly underdetermined and ill-posed inversion problem, because of the restricted viewing access, the number of chords and the increased noise level in most plasma devices. An unbiased and consistent probabilistic approach within the framework of Bayesian inference is provided by the maximum entropy method, which is independent of model assumptions, but allows any prior knowledge available to be incorporated. The formalism is applied to the reconstruction of emissivity profiles in an NBI heated plasma discharge to determine the dependence of the Shafranov shift on β, the reduction of which was a particular objective in designing the advanced W7-AS stellarator. (author). 40 refs, 7 figs
Energy Technology Data Exchange (ETDEWEB)
Mokhatab, Saied [OIEC - Oil Industries' Engineering and Construction Group, Tehran (Iran, Islamic Republic of); Vatani, Ali [University of Tehran (Iran, Islamic Republic of)
2003-07-01
The main objective of the present study has been the development of a relatively simple analytical algorithm for predicting flow temperature and pressure profiles along the two-phase, gas/gas-condensate transmission pipelines. Results demonstrate the ability of the method to predict reasonably accurate pressure gradient and temperature gradient profiles under operating conditions. (author)
Energy Technology Data Exchange (ETDEWEB)
Cui, Xiaoming, E-mail: mmayzy2008@126.com; Li, Tao, E-mail: litaofeivip@163.com; Li, Xin, E-mail: lx0803@sina.com.cn; Zhou, Weihua, E-mail: wangxue0606@gmail.com
2015-05-15
Highlights: • High-resolution scan mode is appropriate for imaging coronary stent. • HD-detail reconstruction algorithm is stent-dedicated kernel. • The intrastent lumen visibility also depends on stent diameter and material. - Abstract: Objective: The aim of this study was to evaluate the in vivo performance of four image reconstruction algorithms in a high-definition CT (HDCT) scanner with improved spatial resolution for the evaluation of coronary artery stents and intrastent lumina. Materials and methods: Thirty-nine consecutive patients with a total of 71 implanted coronary stents underwent coronary CT angiography (CCTA) on a HDCT (Discovery CT 750 HD; GE Healthcare) with the high-resolution scanning mode. Four different reconstruction algorithms (HD-stand, HD-detail; HD-stand-plus; HD-detail-plus) were applied to reconstruct the stented coronary arteries. Image quality for stent characterization was assessed. Image noise and intrastent luminal diameter were measured. The relationship between the measurement of inner stent diameter (ISD) and the true stent diameter (TSD) and stent type were analysed. Results: The stent-dedicated kernel (HD-detail) offered the highest percentage (53.5%) of good image quality for stent characterization and the highest ratio (68.0 ± 8.4%) of visible stent lumen/true stent lumen for luminal diameter measurement at the expense of an increased overall image noise. The Pearson correlation coefficient between the ISD and TSD measurement and spearman correlation coefficient between the ISD measurement and stent type were 0.83 and 0.48, respectively. Conclusions: Compared with standard reconstruction algorithms, high-definition CT imaging technique with dedicated high-resolution reconstruction algorithm provides more accurate stent characterization and intrastent luminal diameter measurement.
Matsen, Frederick A
2010-06-01
This article introduces constNJ (constrained neighbor-joining), an algorithm for phylogenetic reconstruction of sets of trees with constrained pairwise rooted subtree-prune-regraft (rSPR) distance. We are motivated by the problem of constructing sets of trees that must fit into a recombination, hybridization, or similar network. Rather than first finding a set of trees that are optimal according to a phylogenetic criterion (e.g., likelihood or parsimony) and then attempting to fit them into a network, constNJ estimates the trees while enforcing specified rSPR distance constraints. The primary input for constNJ is a collection of distance matrices derived from sequence blocks which are assumed to have evolved in a tree-like manner, such as blocks of an alignment which do not contain any recombination breakpoints. The other input is a set of rSPR constraint inequalities for any set of pairs of trees. constNJ is consistent and a strict generalization of the neighbor-joining algorithm; it uses the new notion of maximum agreement partitions (MAPs) to assure that the resulting trees satisfy the given rSPR distance constraints.
Yan, Mingfei; Hu, Huasi; Otake, Yoshie; Taketani, Atsushi; Wakabayashi, Yasuo; Yanagimachi, Shinzo; Wang, Sheng; Pan, Ziheng; Hu, Guang
2018-05-01
Thermal neutron computer tomography (CT) is a useful tool for visualizing two-phase flow due to its high imaging contrast and strong penetrability of neutrons for tube walls constructed with metallic material. A novel approach for two-phase flow CT reconstruction based on an improved adaptive genetic algorithm with sparsity constraint (IAGA-SC) is proposed in this paper. In the algorithm, the neighborhood mutation operator is used to ensure the continuity of the reconstructed object. The adaptive crossover probability P c and mutation probability P m are improved to help the adaptive genetic algorithm (AGA) achieve the global optimum. The reconstructed results for projection data, obtained from Monte Carlo simulation, indicate that the comprehensive performance of the IAGA-SC algorithm exceeds the adaptive steepest descent-projection onto convex sets (ASD-POCS) algorithm in restoring typical and complex flow regimes. It especially shows great advantages in restoring the simply connected flow regimes and the shape of object. In addition, the CT experiment for two-phase flow phantoms was conducted on the accelerator-driven neutron source to verify the performance of the developed IAGA-SC algorithm.
International Nuclear Information System (INIS)
Pedron, Antoine
2013-01-01
This thesis work is placed between the scientific domain of ultrasound non-destructive testing and algorithm-architecture adequation. Ultrasound non-destructive testing includes a group of analysis techniques used in science and industry to evaluate the properties of a material, component, or system without causing damage. In order to characterise possible defects, determining their position, size and shape, imaging and reconstruction tools have been developed at CEA-LIST, within the CIVA software platform. Evolution of acquisition sensors implies a continuous growth of datasets and consequently more and more computing power is needed to maintain interactive reconstructions. General purpose processors (GPP) evolving towards parallelism and emerging architectures such as GPU allow large acceleration possibilities than can be applied to these algorithms. The main goal of the thesis is to evaluate the acceleration than can be obtained for two reconstruction algorithms on these architectures. These two algorithms differ in their parallelization scheme. The first one can be properly parallelized on GPP whereas on GPU, an intensive use of atomic instructions is required. Within the second algorithm, parallelism is easier to express, but loop ordering on GPP, as well as thread scheduling and a good use of shared memory on GPU are necessary in order to obtain efficient results. Different API or libraries, such as OpenMP, CUDA and OpenCL are evaluated through chosen benchmarks. An integration of both algorithms in the CIVA software platform is proposed and different issues related to code maintenance and durability are discussed. (author) [fr
International Nuclear Information System (INIS)
Quan, E; Lalush, D S
2009-01-01
We present a faster iterative reconstruction algorithm based on the ordered-subset convex (OSC) algorithm for transmission CT. The OSC algorithm was modified such that it calculates the normalization term before the iterative process in order to save computational cost. The modified version requires only one backprojection per iteration as compared to two required for the original OSC. We applied the modified OSC (MOSC) algorithm to a rotation-free micro-CT system that we proposed previously, observed its performance, and compared with the OSC algorithm for 3D cone-beam reconstruction. Measurements on the reconstructed images as well as the point spread functions show that MOSC is quite similar to OSC; in noise-resolution trade-off, MOSC is comparable with OSC in a regular-noise situation and it is slightly worse than OSC in an extremely high-noise situation. The timing record shows that MOSC saves 25-30% CPU time, depending on the number of iterations used. We conclude that the MOSC algorithm is more efficient than OSC and provides comparable images.
Energy Technology Data Exchange (ETDEWEB)
Niemkiewicz, J; Palmiotti, A; Miner, M; Stunja, L; Bergene, J [Lehigh Valley Health Network, Allentown, PA (United States)
2014-06-01
Purpose: Metal in patients creates streak artifacts in CT images. When used for radiation treatment planning, these artifacts make it difficult to identify internal structures and affects radiation dose calculations, which depend on HU numbers for inhomogeneity correction. This work quantitatively evaluates a new metal artifact reduction (MAR) CT image reconstruction algorithm (GE Healthcare CT-0521-04.13-EN-US DOC1381483) when metal is present. Methods: A Gammex Model 467 Tissue Characterization phantom was used. CT images were taken of this phantom on a GE Optima580RT CT scanner with and without steel and titanium plugs using both the standard and MAR reconstruction algorithms. HU values were compared pixel by pixel to determine if the MAR algorithm altered the HUs of normal tissues when no metal is present, and to evaluate the effect of using the MAR algorithm when metal is present. Also, CT images of patients with internal metal objects using standard and MAR reconstruction algorithms were compared. Results: Comparing the standard and MAR reconstructed images of the phantom without metal, 95.0% of pixels were within ±35 HU and 98.0% of pixels were within ±85 HU. Also, the MAR reconstruction algorithm showed significant improvement in maintaining HUs of non-metallic regions in the images taken of the phantom with metal. HU Gamma analysis (2%, 2mm) of metal vs. non-metal phantom imaging using standard reconstruction resulted in an 84.8% pass rate compared to 96.6% for the MAR reconstructed images. CT images of patients with metal show significant artifact reduction when reconstructed with the MAR algorithm. Conclusion: CT imaging using the MAR reconstruction algorithm provides improved visualization of internal anatomy and more accurate HUs when metal is present compared to the standard reconstruction algorithm. MAR reconstructed CT images provide qualitative and quantitative improvements over current reconstruction algorithms, thus improving radiation
International Nuclear Information System (INIS)
Niemkiewicz, J; Palmiotti, A; Miner, M; Stunja, L; Bergene, J
2014-01-01
Purpose: Metal in patients creates streak artifacts in CT images. When used for radiation treatment planning, these artifacts make it difficult to identify internal structures and affects radiation dose calculations, which depend on HU numbers for inhomogeneity correction. This work quantitatively evaluates a new metal artifact reduction (MAR) CT image reconstruction algorithm (GE Healthcare CT-0521-04.13-EN-US DOC1381483) when metal is present. Methods: A Gammex Model 467 Tissue Characterization phantom was used. CT images were taken of this phantom on a GE Optima580RT CT scanner with and without steel and titanium plugs using both the standard and MAR reconstruction algorithms. HU values were compared pixel by pixel to determine if the MAR algorithm altered the HUs of normal tissues when no metal is present, and to evaluate the effect of using the MAR algorithm when metal is present. Also, CT images of patients with internal metal objects using standard and MAR reconstruction algorithms were compared. Results: Comparing the standard and MAR reconstructed images of the phantom without metal, 95.0% of pixels were within ±35 HU and 98.0% of pixels were within ±85 HU. Also, the MAR reconstruction algorithm showed significant improvement in maintaining HUs of non-metallic regions in the images taken of the phantom with metal. HU Gamma analysis (2%, 2mm) of metal vs. non-metal phantom imaging using standard reconstruction resulted in an 84.8% pass rate compared to 96.6% for the MAR reconstructed images. CT images of patients with metal show significant artifact reduction when reconstructed with the MAR algorithm. Conclusion: CT imaging using the MAR reconstruction algorithm provides improved visualization of internal anatomy and more accurate HUs when metal is present compared to the standard reconstruction algorithm. MAR reconstructed CT images provide qualitative and quantitative improvements over current reconstruction algorithms, thus improving radiation
International Nuclear Information System (INIS)
1976-01-01
A tomograph which is capable of gathering divergent radiations and reconstruct them in signal profiles or images each corresponding with a beam of parallel rays is discussed which may eliminate the interfering point dispersion function which normally occurs
Behavioral Profiling of Scada Network Traffic Using Machine Learning Algorithms
2014-03-27
Acquisition ( SCADA ) System Overview SCADA systems control and monitor processes for water distribution, oil and natural gas pipelines , electrical...the desire for remote control and monitoring of industrial processes. The ability to identify SCADA devices on a mixed traffic network with zero...optimal attribute subset, while maintaining the desired TPR of .99 for SCADA network traffic. The attributes and ML algorithms chosen for
Resilient algorithms for reconstructing and simulating gappy flow fields in CFD
Energy Technology Data Exchange (ETDEWEB)
Lee, Seungjoon; Karniadakis, George Em [Division of Applied Mathematics, Brown University, Providence, RI 02912 (United States); Kevrekidis, Ioannis G, E-mail: george_karniadakis@brown.edu [Department of Chemical and Biological Engineering, Brown University, Providence, RI 02912 (United States); PACM, Princeton University, Princeton, NJ 08544 (United States)
2015-10-15
It is anticipated that in future generations of massively parallel computer systems a significant portion of processors may suffer from hardware or software faults rendering large-scale computations useless. In this work we address this problem from the algorithmic side, proposing resilient algorithms that can recover from such faults irrespective of their fault origin. In particular, we set the foundations of a new class of algorithms that will combine numerical approximations with machine learning methods. To this end, we consider three types of fault scenarios: (1) a gappy region but with no previous gaps and no contamination of surrounding simulation data, (2) a space–time gappy region but with full spatiotemporal information and no contamination, and (3) previous gaps with contamination of surrounding data. To recover from such faults we employ different reconstruction and simulation methods, namely the projective integration, the co-Kriging interpolation, and the resimulation method. In order to compare the effectiveness of these methods for the different processor faults and to quantify the error propagation in each case, we perform simulations of two benchmark flows, flow in a cavity and flow past a circular cylinder. In general, the projective integration seems to be the most effective method when the time gaps are small, and the resimulation method is the best when the time gaps are big while the co-Kriging method is independent of time gaps. Furthermore, the projective integration method and the co-Kriging method are found to be good estimation methods for the initial and boundary conditions of the resimulation method in scenario (3). (paper)
Treatment Algorithms Based on Tumor Molecular Profiling: The Essence of Precision Medicine Trials.
Le Tourneau, Christophe; Kamal, Maud; Tsimberidou, Apostolia-Maria; Bedard, Philippe; Pierron, Gaëlle; Callens, Céline; Rouleau, Etienne; Vincent-Salomon, Anne; Servant, Nicolas; Alt, Marie; Rouzier, Roman; Paoletti, Xavier; Delattre, Olivier; Bièche, Ivan
2016-04-01
With the advent of high-throughput molecular technologies, several precision medicine (PM) studies are currently ongoing that include molecular screening programs and PM clinical trials. Molecular profiling programs establish the molecular profile of patients' tumors with the aim to guide therapy based on identified molecular alterations. The aim of prospective PM clinical trials is to assess the clinical utility of tumor molecular profiling and to determine whether treatment selection based on molecular alterations produces superior outcomes compared with unselected treatment. These trials use treatment algorithms to assign patients to specific targeted therapies based on tumor molecular alterations. These algorithms should be governed by fixed rules to ensure standardization and reproducibility. Here, we summarize key molecular, biological, and technical criteria that, in our view, should be addressed when establishing treatment algorithms based on tumor molecular profiling for PM trials. © The Author 2015. Published by Oxford University Press.
Energy Technology Data Exchange (ETDEWEB)
Chun, Se Young [School of Electrical and Computer Engineering, Ulsan National Institute of Science and Technology (UNIST), Ulsan (Korea, Republic of)
2016-03-15
PET and SPECT are important tools for providing valuable molecular information about patients to clinicians. Advances in nuclear medicine hardware technologies and statistical image reconstruction algorithms enabled significantly improved image quality. Sequentially or simultaneously acquired anatomical images such as CT and MRI from hybrid scanners are also important ingredients for improving the image quality of PET or SPECT further. High-quality anatomical information has been used and investigated for attenuation and scatter corrections, motion compensation, and noise reduction via post-reconstruction filtering and regularization in inverse problems. In this article, we will review works using anatomical information for molecular image reconstruction algorithms for better image quality by describing mathematical models, discussing sources of anatomical information for different cases, and showing some examples.
Directory of Open Access Journals (Sweden)
GRANDIN, P. H.
2014-06-01
Full Text Available Recommendation systems based on collaborative filtering are open by nature, what makes them vulnerable to profile injection attacks that insert biased evaluations in the system database in order to manipulate recommendations. In this paper we evaluate the stability and robustness of collaborative filtering algorithms applied to semantic web services recommendation when submitted to random and segment profile injection attacks. We evaluated four algorithms: (1 IMEAN, that makes predictions using the average of the evaluations received by the target item; (2 UMEAN, that makes predictions using the average of the evaluation made by the target user; (3 an algorithm based on the k-nearest neighbor (k-NN method and (4, an algorithm based on the k-means clustering method.The experiments showed that the UMEAN algorithm is not affected by the attacks and that IMEAN is the most vulnerable of all algorithms tested. Nevertheless, both UMEAN and IMEAN have little practical application due to the low precision of their predictions. Among the algorithms with intermediate tolerance to attacks but with good prediction performance, the algorithm based on k-nn proved to be more robust and stable than the algorithm based on k-means.
GPS tomography: validation of reconstructed 3-D humidity fields with radiosonde profiles
Directory of Open Access Journals (Sweden)
M. Shangguan
2013-09-01
Full Text Available Water vapor plays an important role in meteorological applications; GeoForschungsZentrum (GFZ therefore developed a tomographic system to derive 3-D distributions of the tropospheric water vapor above Germany using GPS data from about 300 ground stations. Input data for the tomographic reconstructions are generated by the Earth Parameter and Orbit determination System (EPOS software of the GFZ, which provides zenith total delay (ZTD, integrated water vapor (IWV and slant total delay (STD data operationally with a temporal resolution of 2.5 min (STD and 15 min (ZTD, IWV. The water vapor distribution in the atmosphere is derived by tomographic reconstruction techniques. The quality of the solution is dependent on many factors such as the spatial coverage of the atmosphere with slant paths, the spatial distribution of their intersections and the accuracy of the input observations. Independent observations are required to validate the tomographic reconstructions and to get precise information on the accuracy of the derived 3-D water vapor fields. To determine the quality of the GPS tomography, more than 8000 vertical water vapor profiles at 13 German radiosonde stations were used for the comparison. The radiosondes were launched twice a day (at 00:00 UTC and 12:00 UTC in 2007. In this paper, parameters of the entire profiles such as the wet refractivity, and the zenith wet delay have been compared. Before the validation the temporal and spatial distribution of the slant paths, serving as a basis for tomographic reconstruction, as well as their angular distribution were studied. The mean wet refractivity differences between tomography and radiosonde data for all points vary from −1.3 to 0.3, and the root mean square is within the range of 6.5–9. About 32% of 6803 profiles match well, 23% match badly and 45% are difficult to classify as they match only in parts.
GPS tomography. Validation of reconstructed 3-D humidity fields with radiosonde profiles
Energy Technology Data Exchange (ETDEWEB)
Shangguan, M.; Bender, M.; Ramatschi, M.; Dick, G.; Wickert, J. [Helmholtz Centre Potsdam, German Research Centre for Geosciences (GFZ), Potsdam (Germany); Raabe, A. [Leipzig Institute for Meteorology (LIM), Leipzig (Germany); Galas, R. [Technische Univ. Berlin (Germany). Dept. for Geodesy and Geoinformation Sciences
2013-11-01
Water vapor plays an important role in meteorological applications; GeoForschungsZentrum (GFZ) therefore developed a tomographic system to derive 3-D distributions of the tropospheric water vapor above Germany using GPS data from about 300 ground stations. Input data for the tomographic reconstructions are generated by the Earth Parameter and Orbit determination System (EPOS) software of the GFZ, which provides zenith total delay (ZTD), integrated water vapor (IWV) and slant total delay (STD) data operationally with a temporal resolution of 2.5 min (STD) and 15 min (ZTD, IWV). The water vapor distribution in the atmosphere is derived by tomographic reconstruction techniques. The quality of the solution is dependent on many factors such as the spatial coverage of the atmosphere with slant paths, the spatial distribution of their intersections and the accuracy of the input observations. Independent observations are required to validate the tomographic reconstructions and to get precise information on the accuracy of the derived 3-D water vapor fields. To determine the quality of the GPS tomography, more than 8000 vertical water vapor profiles at 13 German radiosonde stations were used for the comparison. The radiosondes were launched twice a day (at 00:00 UTC and 12:00 UTC) in 2007. In this paper, parameters of the entire profiles such as the wet refractivity, and the zenith wet delay have been compared. Before the validation the temporal and spatial distribution of the slant paths, serving as a basis for tomographic reconstruction, as well as their angular distribution were studied. The mean wet refractivity differences between tomography and radiosonde data for all points vary from -1.3 to 0.3, and the root mean square is within the range of 6.5-9. About 32% of 6803 profiles match well, 23% match badly and 45% are difficult to classify as they match only in parts.
Energy Technology Data Exchange (ETDEWEB)
Llacer Martos, S.; Herraiz Lablanca, M. D.; Puchal Ane, R.
2011-07-01
This paper compares the image quality obtained with each of the algorithms is evaluated and its running time, to optimize the choice of algorithm to use taking into account both the quality of the reconstructed image as the time spent on the reconstruction.
k-Nearest Neighbors Algorithm in Profiling Power Analysis Attacks
Directory of Open Access Journals (Sweden)
Z. Martinasek
2016-06-01
Full Text Available Power analysis presents the typical example of successful attacks against trusted cryptographic devices such as RFID (Radio-Frequency IDentifications and contact smart cards. In recent years, the cryptographic community has explored new approaches in power analysis based on machine learning models such as Support Vector Machine (SVM, RF (Random Forest and Multi-Layer Perceptron (MLP. In this paper, we made an extensive comparison of machine learning algorithms in the power analysis. For this purpose, we implemented a verification program that always chooses the optimal settings of individual machine learning models in order to obtain the best classification accuracy. In our research, we used three datasets, the first containing the power traces of an unprotected AES (Advanced Encryption Standard implementation. The second and third datasets are created independently from public available power traces corresponding to a masked AES implementation (DPA Contest v4. The obtained results revealed some interesting facts, namely, an elementary k-NN (k-Nearest Neighbors algorithm, which has not been commonly used in power analysis yet, shows great application potential in practice.
Patra, Rusha; Dutta, Pranab K.
2015-07-01
Reconstruction of the absorption coefficient of tissue with good contrast is of key importance in functional diffuse optical imaging. A hybrid approach using model-based iterative image reconstruction and a genetic algorithm is proposed to enhance the contrast of the reconstructed image. The proposed method yields an observed contrast of 98.4%, mean square error of 0.638×10-3, and object centroid error of (0.001 to 0.22) mm. Experimental validation of the proposed method has also been provided with tissue-like phantoms which shows a significant improvement in image quality and thus establishes the potential of the method for functional diffuse optical tomography reconstruction with continuous wave setup. A case study of finger joint imaging is illustrated as well to show the prospect of the proposed method in clinical diagnosis. The method can also be applied to the concentration measurement of a region of interest in a turbid medium.
Energy Technology Data Exchange (ETDEWEB)
Sakai, Fumikazu; Tsuuchi, Yasuhiko; Suzuki, Keiko; Ueno, Keiko; Yamada, Takayuki; Okawa, Tomohiko [Tokyo Women`s Medical Coll. (Japan); Yun, Shen; Horiuchi, Tetsuya; Kimura, Fumiko
1998-05-01
We describe our trial to reduce cardiac motion artifacts on HR-CT images caused by cardiac pulsation by combining use of subsecond CT (scan time 0.8 s) and a special cine reconstruction algorithm (cine reconstruction algorithm with 180-degree helical interpolation). Eleven to 51 HR-CT images were reconstructed with the special cine reconstruction algorithm at the pitch of 0.1 (0.08 s) from the data obtained by two to six contigious rotation scans at the same level. Images with the fewest cardiac motion artifacts were selected for evaluation. These images were compared with those reconstructed with a conventional cine reconstruction algorithm and step-by-step scan. In spite of its increased radiation exposure, technical complexity and slight degradation of spatial resolution, our method was useful in reducing cardiac motion artifacts on HR-CT images in regions adjacent to the heart. (author)
Adjustable internal structure for reconstructing gradient index profile of crystalline lens.
Bahrami, Mehdi; Goncharov, Alexander V; Pierscionek, Barbara K
2014-03-01
Employing advanced technologies in studying the crystalline lens of the eye has improved our understanding of the refractive index gradient of the lens. Reconstructing and studying such a complex structure requires models with adaptable internal geometry that can be altered to simulate geometrical and optical changes of the lens with aging. In this Letter, we introduce an optically well-defined, geometrical structure for modeling the gradient refractive index profile of the crystalline lens with the advantage of an adjustable internal structure that is not available with existing models. The refractive index profile assigned to this rotationally symmetric geometry is calculated numerically, yet it is shown that this does not limit the model. The study provides a basis for developing lens models with sophisticated external and internal structures without the need for analytical solutions to calculate refractive index profiles.
International Nuclear Information System (INIS)
Matsumoto, Keiichi; Ohnishi, Hideo; Niida, Hideharu; Nishimura, Yoshihiro; Wada, Yasuhiro; Kida, Tetsuo
2003-01-01
The maximum likelihood expectation maximization (ML-EM) algorithm has become available as an alternative to filtered back projection in SPECT. The actual physical performance may be different depending on the manufacturer and model, because of differences in computational details. The purpose of this study was to investigate the characteristics of seven different types of ML-EM algorithms using simple simulation data. Seven ML-EM algorithm programs were used: Genie (GE), esoft (Siemens), HARP-III (Hitachi), GMS-5500UI (Toshiba), Pegasys (ADAC), ODYSSEY-FX (Marconi), and Windows-PC (original software). Projection data of a 2-pixel-wide line source in the center of the field of view were simulated without attenuation or scatter. Images were reconstructed with ML-EM by changing the number of iterations from 1 to 45 for each algorithm. Image quality was evaluated after a reconstruction using full width at half maximum (FWHM), full width at tenth maximum (FWTM), and the total counts of the reconstructed images. In the maximum number of iterations, the difference in the FWHM value was up to 1.5 pixels, and that of FWTM, no less than 2.0 pixels. The total counts of the reconstructed images in the initial few iterations were larger or smaller than the converged value depending on the initial values. Our results for the simplest simulation data suggest that each ML-EM algorithm itself provides a simulation image. We should keep in mind which algorithm is being used and its computational details, when physical and clinical usefulness are compared. (author)
Probability- and curve-based fractal reconstruction on 2D DEM terrain profile
International Nuclear Information System (INIS)
Lai, F.-J.; Huang, Y.M.
2009-01-01
Data compression and reconstruction has been playing important roles in information science and engineering. As part of them, image compression and reconstruction that mainly deal with image data set reduction for storage or transmission and data set restoration with least loss is still a topic deserved a great deal of works to focus on. In this paper we propose a new scheme in comparison with the well-known Improved Douglas-Peucker (IDP) method to extract characteristic or feature points of two-dimensional digital elevation model (2D DEM) terrain profile to compress data set. As for reconstruction in use of fractal interpolation, we propose a probability-based method to speed up the fractal interpolation execution to a rate as high as triple or even ninefold of the regular. In addition, a curve-based method is proposed in the study to determine the vertical scaling factor that much affects the generation of the interpolated data points to significantly improve the reconstruction performance. Finally, an evaluation is made to show the advantage of employing the proposed new method to extract characteristic points associated with our novel fractal interpolation scheme.
Lin, Qingyang; Andrew, Matthew; Thompson, William; Blunt, Martin J.; Bijeljic, Branko
2018-05-01
Non-invasive laboratory-based X-ray microtomography has been widely applied in many industrial and research disciplines. However, the main barrier to the use of laboratory systems compared to a synchrotron beamline is its much longer image acquisition time (hours per scan compared to seconds to minutes at a synchrotron), which results in limited application for dynamic in situ processes. Therefore, the majority of existing laboratory X-ray microtomography is limited to static imaging; relatively fast imaging (tens of minutes per scan) can only be achieved by sacrificing imaging quality, e.g. reducing exposure time or number of projections. To alleviate this barrier, we introduce an optimized implementation of a well-known iterative reconstruction algorithm that allows users to reconstruct tomographic images with reasonable image quality, but requires lower X-ray signal counts and fewer projections than conventional methods. Quantitative analysis and comparison between the iterative and the conventional filtered back-projection reconstruction algorithm was performed using a sandstone rock sample with and without liquid phases in the pore space. Overall, by implementing the iterative reconstruction algorithm, the required image acquisition time for samples such as this, with sparse object structure, can be reduced by a factor of up to 4 without measurable loss of sharpness or signal to noise ratio.
The Nonlocal Sparse Reconstruction Algorithm by Similarity Measurement with Shearlet Feature Vector
Directory of Open Access Journals (Sweden)
Wu Qidi
2014-01-01
Full Text Available Due to the limited accuracy of conventional methods with image restoration, the paper supplied a nonlocal sparsity reconstruction algorithm with similarity measurement. To improve the performance of restoration results, we proposed two schemes to dictionary learning and sparse coding, respectively. In the part of the dictionary learning, we measured the similarity between patches from degraded image by constructing the Shearlet feature vector. Besides, we classified the patches into different classes with similarity and trained the cluster dictionary for each class, by cascading which we could gain the universal dictionary. In the part of sparse coding, we proposed a novel optimal objective function with the coding residual item, which can suppress the residual between the estimate coding and true sparse coding. Additionally, we show the derivation of self-adaptive regularization parameter in optimization under the Bayesian framework, which can make the performance better. It can be indicated from the experimental results that by taking full advantage of similar local geometric structure feature existing in the nonlocal patches and the coding residual suppression, the proposed method shows advantage both on visual perception and PSNR compared to the conventional methods.
Clapuyt, Francois; Vanacker, Veerle; Van Oost, Kristof
2016-05-01
Combination of UAV-based aerial pictures and Structure-from-Motion (SfM) algorithm provides an efficient, low-cost and rapid framework for remote sensing and monitoring of dynamic natural environments. This methodology is particularly suitable for repeated topographic surveys in remote or poorly accessible areas. However, temporal analysis of landform topography requires high accuracy of measurements and reproducibility of the methodology as differencing of digital surface models leads to error propagation. In order to assess the repeatability of the SfM technique, we surveyed a study area characterized by gentle topography with an UAV platform equipped with a standard reflex camera, and varied the focal length of the camera and location of georeferencing targets between flights. Comparison of different SfM-derived topography datasets shows that precision of measurements is in the order of centimetres for identical replications which highlights the excellent performance of the SfM workflow, all parameters being equal. The precision is one order of magnitude higher for 3D topographic reconstructions involving independent sets of ground control points, which results from the fact that the accuracy of the localisation of ground control points strongly propagates into final results.
Rapisarda, E; Bettinardi, V; Thielemans, K; Gilardi, M C
2010-07-21
The interest in positron emission tomography (PET) and particularly in hybrid integrated PET/CT systems has significantly increased in the last few years due to the improved quality of the obtained images. Nevertheless, one of the most important limits of the PET imaging technique is still its poor spatial resolution due to several physical factors originating both at the emission (e.g. positron range, photon non-collinearity) and at detection levels (e.g. scatter inside the scintillating crystals, finite dimensions of the crystals and depth of interaction). To improve the spatial resolution of the images, a possible way consists of measuring the point spread function (PSF) of the system and then accounting for it inside the reconstruction algorithm. In this work, the system response of the GE Discovery STE operating in 3D mode has been characterized by acquiring (22)Na point sources in different positions of the scanner field of view. An image-based model of the PSF was then obtained by fitting asymmetric two-dimensional Gaussians on the (22)Na images reconstructed with small pixel sizes. The PSF was then incorporated, at the image level, in a three-dimensional ordered subset maximum likelihood expectation maximization (OS-MLEM) reconstruction algorithm. A qualitative and quantitative validation of the algorithm accounting for the PSF has been performed on phantom and clinical data, showing improved spatial resolution, higher contrast and lower noise compared with the corresponding images obtained using the standard OS-MLEM algorithm.
International Nuclear Information System (INIS)
Laurent, C.; Chassery, J.M.; Peyrin, F.; Girerd, C.
1996-01-01
This paper deals with the parallel implementations of reconstruction methods in 3D tomography. 3D tomography requires voluminous data and long computation times. Parallel computing, on MIMD computers, seems to be a good approach to manage this problem. In this study, we present the different steps of the parallelization on an abstract parallel computer. Depending on the method, we use two main approaches to parallelize the algorithms: the local approach and the global approach. Experimental results on MIMD computers are presented. Two 3D images reconstructed from realistic data are showed
Genetic Algorithm for Opto-thermal Skin Hydration Depth Profiling Measurements
Cui, Y.; Xiao, Perry; Imhof, R. E.
2013-09-01
Stratum corneum is the outermost skin layer, and the water content in stratum corneum plays a key role in skin cosmetic properties as well as skin barrier functions. However, to measure the water content, especially the water concentration depth profile, within stratum corneum is very difficult. Opto-thermal emission radiometry, or OTTER, is a promising technique that can be used for such measurements. In this paper, a study on stratum corneum hydration depth profiling by using a genetic algorithm (GA) is presented. The pros and cons of a GA compared against other inverse algorithms such as neural networks, maximum entropy, conjugate gradient, and singular value decomposition will be discussed first. Then, it will be shown how to use existing knowledge to optimize a GA for analyzing the opto-thermal signals. Finally, these latest GA results on hydration depth profiling of stratum corneum under different conditions, as well as on the penetration profiles of externally applied solvents, will be shown.
International Nuclear Information System (INIS)
Vaegler, Sven
2016-01-01
The treatment of cancer in radiation therapy is achievable today by techniques that enable highly conformal dose distributions and steep dose gradients. In order to avoid mistreatment, these irradiation techniques have necessitated enhanced patient localization techniques. With an integrated x-ray tube at modern linear accelerators kV-projections can be acquired over a sufficiently large angular space and can be reconstructed to a volumetric image data set from the current situation of the patient prior to irradiation. The so-called Cone-Beam-CT (CBCT) allows a precise verification of patient positioning as well as adaptive radiotherapy. The benefits of an improved patient positioning due to a daily performed CBCT's is contrary to an increased and not negligible radiation exposure of the patient. In order to decrease the radiation exposure, substantial research effort is focused on various dose reduction strategies. Prominent strategies are the decrease of the charge per projection, the reduction of the number of projections as well as the reduction of the acquisition space. Unfortunately, these acquisition schemes lead to images with degraded quality with the widely used Feldkamp-Davis-Kress image reconstruction algorithm. More sophisticated image reconstruction techniques can deal with these dose-reduction strategies without degrading the image quality. A frequently investigated method is the image reconstruction by minimizing the total variation (TV), which is also known as Compressed Sensing (CS). A Compressed Sensing-based reconstruction framework that includes prior images into the reconstruction algorithm is the Prior-Image-Constrained- Compressed-Sensing algorithm (PICCS). The images reconstructed by PICCS outperform the reconstruction results of the conventional Feldkamp-Davis-Kress algorithm (FDK) based method if only a small number of projections are available. However, a drawback of PICCS is that major deviations between prior image data sets and the
Group-decoupled multi-group pin power reconstruction utilizing nodal solution 1D flux profiles
International Nuclear Information System (INIS)
Yu, Lulin; Lu, Dong; Zhang, Shaohong; Wang, Dezhong
2014-01-01
Highlights: • A direct fitting multi-group pin power reconstruction method is developed. • The 1D nodal solution flux profiles are used as the condition. • The least square fit problem is analytically solved. • A slowing down source improvement method is applied. • The method shows good accuracy for even challenging problems. - Abstract: A group-decoupled direct fitting method is developed for multi-group pin power reconstruction, which avoids both the complication of obtaining 2D analytic multi-group flux solution and any group-coupled iteration. A unique feature of the method is that in addition to nodal volume and surface average fluxes and corner fluxes, transversely-integrated 1D nodal solution flux profiles are also used as the condition to determine the 2D intra-nodal flux distribution. For each energy group, a two-dimensional expansion with a nine-term polynomial and eight hyperbolic functions is used to perform a constrained least square fit to the 1D intra-nodal flux solution profiles. The constraints are on the conservation of nodal volume and surface average fluxes and corner fluxes. Instead of solving the constrained least square fit problem numerically, we solve it analytically by fully utilizing the symmetry property of the expansion functions. Each of the 17 unknown expansion coefficients is expressed in terms of nodal volume and surface average fluxes, corner fluxes and transversely-integrated flux values. To determine the unknown corner fluxes, a set of linear algebraic equations involving corner fluxes is established via using the current conservation condition on all corners. Moreover, an optional slowing down source improvement method is also developed to further enhance the accuracy of the reconstructed flux distribution if needed. Two test examples are shown with very good results. One is a four-group BWR mini-core problem with all control blades inserted and the other is the seven-group OECD NEA MOX benchmark, C5G7
International Nuclear Information System (INIS)
Yusa, Noritaka; Chen, Zhenmao; Miya, Kenzo; Cheng, Weiying
2002-01-01
This paper proposes a reliability parameter to enhance an version scheme developed by authors. The scheme is based upon an artificial neural network that simulates mapping between eddy current signals and crack profiles. One of the biggest advantages of the scheme is that it can deal with conductive cracks, which is necessary to reconstruct natural cracks. However, it has one significant disadvantage: the reliability of reconstructed profiles was unknown. The parameter provides an index for assessment of the crack profile and overcomes this disadvantage. After the parameter is validated by reconstruction of simulated cracks, it is applied to reconstruction of natural cracks that occurred in steam generator tubes of a pressurized water reactor. It is revealed that the parameter is applicable to not only simulated cracks but also natural ones. (author)
Directory of Open Access Journals (Sweden)
Karla Vittori
2008-12-01
Full Text Available We propose a new distance algorithm for phylogenetic estimation based on Ant Colony Optimization (ACO, named Ant-Based Phylogenetic Reconstruction (ABPR. ABPR joins two taxa iteratively based on evolutionary distance among sequences, while also accounting for the quality of the phylogenetic tree built according to the total length of the tree. Similar to optimization algorithms for phylogenetic estimation, the algorithm allows exploration of a larger set of nearly optimal solutions. We applied the algorithm to four empirical data sets of mitochondrial DNA ranging from 12 to 186 sequences, and from 898 to 16,608 base pairs, and covering taxonomic levels from populations to orders. We show that ABPR performs better than the commonly used Neighbor-Joining algorithm, except when sequences are too closely related (e.g., population-level sequences. The phylogenetic relationships recovered at and above species level by ABPR agree with conventional views. However, like other algorithms of phylogenetic estimation, the proposed algorithm failed to recover expected relationships when distances are too similar or when rates of evolution are very variable, leading to the problem of long-branch attraction. ABPR, as well as other ACO-based algorithms, is emerging as a fast and accurate alternative method of phylogenetic estimation for large data sets.
International Nuclear Information System (INIS)
Kubo, S.; Narihara, K.; Tomita, Y.; Hasegawa, M.; Tsuzuki, T.; Mohri, A.
1988-01-01
A multichannel HCN laser interferometer system has been developed to investigate the plasma electron confinement properties in SPAC VII device. Maximum entropy method is applied to reconstruct the electron density profile from measured line integrated data. Particle diffusion coefficient in the peripheral region of the REB ring core spherator was obtained from the evolution of the density profile. (author)
International Nuclear Information System (INIS)
Huang, Q; Zeng, G L; You, J; Gullberg, G T
2005-01-01
In this paper, Novikov's inversion formula of the attenuated two-dimensional (2D) Radon transform is applied to the reconstruction of attenuated fan-beam projections acquired with equal detector spacing and of attenuated cone-beam projections acquired with a flat planar detector and circular trajectory. The derivation of the fan-beam algorithm is obtained by transformation from parallel-beam coordinates to fan-beam coordinates. The cone-beam reconstruction algorithm is an extension of the fan-beam reconstruction algorithm using Feldkamp-Davis-Kress's (FDK) method. Computer simulations indicate that the algorithm is efficient and is accurate in reconstructing slices close to the central slice of the cone-beam orbit plane. When the attenuation map is set to zero the implementation is equivalent to the FDK method. Reconstructed images are also shown for noise corrupted projections
Dietrich, Felix
2017-01-01
Reconstructing the invariant mass in a Higgs boson decay event containing tau leptons turns out to be a challenging endeavour. The aim of this summer student project is to implement a new algorithm for this task, using deep neural networks and machine learning. The results are compared to SVFit, an existing algorithm that uses dynamical likelihood techniques. A neural network is found that reaches the accuracy of SVFit at low masses and even surpasses it at higher masses, while at the same time providing results a thousand times faster.
International Nuclear Information System (INIS)
Gomez-Cardona, Daniel; Nagle, Scott K.; Li, Ke; Chen, Guang-Hong; Robinson, Terry E.
2015-01-01
Purpose: Wall thickness (WT) is an airway feature of great interest for the assessment of morphological changes in the lung parenchyma. Multidetector computed tomography (MDCT) has recently been used to evaluate airway WT, but the potential risk of radiation-induced carcinogenesis—particularly in younger patients—might limit a wider use of this imaging method in clinical practice. The recent commercial implementation of the statistical model-based iterative reconstruction (MBIR) algorithm, instead of the conventional filtered back projection (FBP) algorithm, has enabled considerable radiation dose reduction in many other clinical applications of MDCT. The purpose of this work was to study the impact of radiation dose and MBIR in the MDCT assessment of airway WT. Methods: An airway phantom was scanned using a clinical MDCT system (Discovery CT750 HD, GE Healthcare) at 4 kV levels and 5 mAs levels. Both FBP and a commercial implementation of MBIR (Veo TM , GE Healthcare) were used to reconstruct CT images of the airways. For each kV–mAs combination and each reconstruction algorithm, the contrast-to-noise ratio (CNR) of the airways was measured, and the WT of each airway was measured and compared with the nominal value; the relative bias and the angular standard deviation in the measured WT were calculated. For each airway and reconstruction algorithm, the overall performance of WT quantification across all of the 20 kV–mAs combinations was quantified by the sum of squares (SSQs) of the difference between the measured and nominal WT values. Finally, the particular kV–mAs combination and reconstruction algorithm that minimized radiation dose while still achieving a reference WT quantification accuracy level was chosen as the optimal acquisition and reconstruction settings. Results: The wall thicknesses of seven airways of different sizes were analyzed in the study. Compared with FBP, MBIR improved the CNR of the airways, particularly at low radiation dose
International Nuclear Information System (INIS)
Cao, Xiaoqing; Xie, Qingguo; Xiao, Peng
2015-01-01
List mode format is commonly used in modern positron emission tomography (PET) for image reconstruction due to certain special advantages. In this work, we proposed a list mode based regularized relaxed ordered subset (LMROS) algorithm for static PET imaging. LMROS is able to work with regularization terms which can be formulated as twice differentiable convex functions. Such a versatility would make LMROS a convenient and general framework for fulfilling different regularized list mode reconstruction methods. LMROS was applied to two simulated undersampling PET imaging scenarios to verify its effectiveness. Convex quadratic function, total variation constraint, non-local means and dictionary learning based regularization methods were successfully realized for different cases. The results showed that the LMROS algorithm was effective and some regularization methods greatly reduced the distortions and artifacts caused by undersampling. (paper)
Energy Technology Data Exchange (ETDEWEB)
Gomez-Cardona, Daniel [Department of Medical Physics, University of Wisconsin-Madison School of Medicine and Public Health, 1111 Highland Avenue, Madison, Wisconsin 53705 (United States); Nagle, Scott K. [Department of Medical Physics, University of Wisconsin-Madison School of Medicine and Public Health, 1111 Highland Avenue, Madison, Wisconsin 53705 (United States); Department of Radiology, University of Wisconsin-Madison School of Medicine and Public Health, 600 Highland Avenue, Madison, Wisconsin 53792 (United States); Department of Pediatrics, University of Wisconsin-Madison School of Medicine and Public Health, 600 Highland Avenue, Madison, Wisconsin 53792 (United States); Li, Ke; Chen, Guang-Hong, E-mail: gchen7@wisc.edu [Department of Medical Physics, University of Wisconsin-Madison School of Medicine and Public Health, 1111 Highland Avenue, Madison, Wisconsin 53705 (United States); Department of Radiology, University of Wisconsin-Madison School of Medicine and Public Health, 600 Highland Avenue, Madison, Wisconsin 53792 (United States); Robinson, Terry E. [Department of Pediatrics, Stanford School of Medicine, 770 Welch Road, Palo Alto, California 94304 (United States)
2015-10-15
Purpose: Wall thickness (WT) is an airway feature of great interest for the assessment of morphological changes in the lung parenchyma. Multidetector computed tomography (MDCT) has recently been used to evaluate airway WT, but the potential risk of radiation-induced carcinogenesis—particularly in younger patients—might limit a wider use of this imaging method in clinical practice. The recent commercial implementation of the statistical model-based iterative reconstruction (MBIR) algorithm, instead of the conventional filtered back projection (FBP) algorithm, has enabled considerable radiation dose reduction in many other clinical applications of MDCT. The purpose of this work was to study the impact of radiation dose and MBIR in the MDCT assessment of airway WT. Methods: An airway phantom was scanned using a clinical MDCT system (Discovery CT750 HD, GE Healthcare) at 4 kV levels and 5 mAs levels. Both FBP and a commercial implementation of MBIR (Veo{sup TM}, GE Healthcare) were used to reconstruct CT images of the airways. For each kV–mAs combination and each reconstruction algorithm, the contrast-to-noise ratio (CNR) of the airways was measured, and the WT of each airway was measured and compared with the nominal value; the relative bias and the angular standard deviation in the measured WT were calculated. For each airway and reconstruction algorithm, the overall performance of WT quantification across all of the 20 kV–mAs combinations was quantified by the sum of squares (SSQs) of the difference between the measured and nominal WT values. Finally, the particular kV–mAs combination and reconstruction algorithm that minimized radiation dose while still achieving a reference WT quantification accuracy level was chosen as the optimal acquisition and reconstruction settings. Results: The wall thicknesses of seven airways of different sizes were analyzed in the study. Compared with FBP, MBIR improved the CNR of the airways, particularly at low radiation dose
DEFF Research Database (Denmark)
Enevoldsen, Lotte H; Menashi, Changez A K; Andersen, Ulrik B
2013-01-01
time (HT) protocols and Evolution for Cardiac Software. METHODS: We studied 45 consecutive, non-selected patients referred for a clinically indicated routine 2-day stress/rest (99m)Tc-Sestamibi myocardial perfusion SPECT. All patients underwent an FT and an HT scan. Both FT and HT scans were processed......-RR) and for quantitative analysis (FT-FBP, HT-FBP, and HT-RR). The datasets were analyzed using commercially available QGS/QPS software and read by two observers evaluating image quality and clinical interpretation. Image quality was assessed on a 10-cm visual analog scale score. RESULTS: HT imaging was associated......: Use of RR reconstruction algorithms compensates for loss of image quality associated with reduced scan time. Both HT acquisition and RR reconstruction algorithm had significant effects on motion and perfusion parameters obtained with standard software, but these effects were relatively small...
International Nuclear Information System (INIS)
Ströhl, Florian; Kaminski, Clemens F
2015-01-01
We demonstrate the reconstruction of images obtained by multifocal structured illumination microscopy, MSIM, using a joint Richardson–Lucy, jRL-MSIM, deconvolution algorithm, which is based on an underlying widefield image-formation model. The method is efficient in the suppression of out-of-focus light and greatly improves image contrast and resolution. Furthermore, it is particularly well suited for the processing of noise corrupted data. The principle is verified on simulated as well as experimental data and a comparison of the jRL-MSIM approach with the standard reconstruction procedure, which is based on image scanning microscopy, ISM, is made. Our algorithm is efficient and freely available in a user friendly software package. (paper)
Energy Technology Data Exchange (ETDEWEB)
Morozov, A; Fraga, F A F; Fraga, M M F R; Margato, L M S; Pereira, L [LIP-Coimbra and Departamento de Física, Universidade de Coimbra, Rua Larga, Coimbra (Portugal); Defendi, I; Jurkovic, M [Forschungs-Neutronenquelle Heinz Maier-Leibnitz (FRM II), TUM, Lichtenbergstr. 1, Garching (Germany); Engels, R; Kemmerling, G [Zentralinstitut für Elektronik, Forschungszentrum Jülich GmbH, Wilhelm-Johnen-Straße, Jülich (Germany); Gongadze, A; Guerard, B; Manzin, G; Niko, H; Peyaud, A; Piscitelli, F [Institut Laue Langevin, 6 Rue Jules Horowitz, Grenoble (France); Petrillo, C; Sacchetti, F [Istituto Nazionale per la Fisica della Materia, Unità di Perugia, Via A. Pascoli, Perugia (Italy); Raspino, D; Rhodes, N J; Schooneveld, E M, E-mail: andrei@coimbra.lip.pt [Science and Technology Facilities Council, Rutherford Appleton Laboratory, Harwell Oxford, Didcot (United Kingdom); others, and
2013-05-01
The software package ANTS (Anger-camera type Neutron detector: Toolkit for Simulations), developed for simulation of Anger-type gaseous detectors for thermal neutron imaging was extended to include a module for experimental data processing. Data recorded with a sensor array containing up to 100 photomultiplier tubes (PMT) or silicon photomultipliers (SiPM) in a custom configuration can be loaded and the positions and energies of the events can be reconstructed using the Center-of-Gravity, Maximum Likelihood or Least Squares algorithm. A particular strength of the new module is the ability to reconstruct the light response functions and relative gains of the photomultipliers from flood field illumination data using adaptive algorithms. The performance of the module is demonstrated with simulated data generated in ANTS and experimental data recorded with a 19 PMT neutron detector. The package executables are publicly available at http://coimbra.lip.pt/∼andrei/.
Ltaief, Hatem; Luszczek, Piotr R.; Dongarra, Jack
2011-01-01
This paper presents the power profile of two high performance dense linear algebra libraries i.e., LAPACK and PLASMA. The former is based on block algorithms that use the fork-join paradigm to achieve parallel performance. The latter uses fine
Reconstructing matter profiles of spherically compensated cosmic regions in ΛCDM cosmology
de Fromont, Paul; Alimi, Jean-Michel
2018-02-01
The absence of a physically motivated model for large-scale profiles of cosmic voids limits our ability to extract valuable cosmological information from their study. In this paper, we address this problem by introducing the spherically compensated cosmic regions, named CoSpheres. Such cosmic regions are identified around local extrema in the density field and admit a unique compensation radius R1 where the internal spherical mass is exactly compensated. Their origin is studied by extending the standard peak model and implementing the compensation condition. Since the compensation radius evolves as the Universe itself, R1(t) ∝ a(t), CoSpheres behave as bubble Universes with fixed comoving volume. Using the spherical collapse model, we reconstruct their profiles with a very high accuracy until z = 0 in N-body simulations. CoSpheres are symmetrically defined and reconstructed for both central maximum (seeding haloes and galaxies) and minimum (identified with cosmic voids). We show that the full non-linear dynamics can be solved analytically around this particular compensation radius, providing useful predictions for cosmology. This formalism highlights original correlations between local extremum and their large-scale cosmic environment. The statistical properties of these spherically compensated cosmic regions and the possibilities to constrain efficiently both cosmology and gravity will be investigated in companion papers.
Mary, D.; Ferrari, A.; Ferrari, C.; Deguignet, J.; Vannier, M.
2016-12-01
With millions of receivers leading to TerraByte data cubes, the story of the giant SKA telescope is also that of collaborative efforts from radioastronomy, signal processing, optimization and computer sciences. Reconstructing SKA cubes poses two challenges. First, the majority of existing algorithms work in 2D and cannot be directly translated into 3D. Second, the reconstruction implies solving an inverse problem and it is not clear what ultimate limit we can expect on the error of this solution. This study addresses (of course partially) both challenges. We consider an extremely simple data acquisition model, and we focus on strategies making it possible to implement 3D reconstruction algorithms that use state-of-the-art image/spectral regularization. The proposed approach has two main features: (i) reduced memory storage with respect to a previous approach; (ii) efficient parallelization and ventilation of the computational load over the spectral bands. This work will allow to implement and compare various 3D reconstruction approaches in a large scale framework.
Linear: A Novel Algorithm for Reconstructing Slitless Spectroscopy from HST/WFC3
Ryan, R. E., Jr.; Casertano, S.; Pirzkal, N.
2018-03-01
We present a grism extraction package (LINEAR) designed to reconstruct 1D spectra from a collection of slitless spectroscopic images, ideally taken at a variety of orientations, dispersion directions, and/or dither positions. Our approach is to enumerate every transformation between all direct image positions (i.e., a potential source) and the collection of grism images at all relevant wavelengths. This leads to solving a large, sparse system of linear equations, which we invert using the standard LSQR algorithm. We implement a number of color and geometric corrections (such as flat field, pixel-area map, source morphology, and spectral bandwidth), but assume many effects have been calibrated out (such as basic reductions, background subtraction, and astrometric refinement). We demonstrate the power of our approach with several Monte Carlo simulations and the analysis of archival data. The simulations include astrometric and photometric uncertainties, sky-background estimation, and signal-to-noise calculations. The data are G141 observations obtained with the Wide-Field Camera 3 of the Hubble Ultra-Deep Field, and show the power of our formalism by improving the spectral resolution without sacrificing the signal-to-noise (a tradeoff that is often made by current approaches). Additionally, our approach naturally accounts for source contamination, which is only handled heuristically by present softwares. We conclude with a discussion of various observations where our approach will provide much improved spectral 1D spectra, such as crowded fields (star or galaxy clusters), spatially resolved spectroscopy, or surveys with strict completeness requirements. At present our software is heavily geared for Wide-Field Camera 3 IR, however we plan extend the codebase for additional instruments.
Mizas, Ch; Sirakoulis, G Ch; Mardiris, V; Karafyllidis, I; Glykos, N; Sandaltzopoulos, R
2008-04-01
Change of DNA sequence that fuels evolution is, to a certain extent, a deterministic process because mutagenesis does not occur in an absolutely random manner. So far, it has not been possible to decipher the rules that govern DNA sequence evolution due to the extreme complexity of the entire process. In our attempt to approach this issue we focus solely on the mechanisms of mutagenesis and deliberately disregard the role of natural selection. Hence, in this analysis, evolution refers to the accumulation of genetic alterations that originate from mutations and are transmitted through generations without being subjected to natural selection. We have developed a software tool that allows modelling of a DNA sequence as a one-dimensional cellular automaton (CA) with four states per cell which correspond to the four DNA bases, i.e. A, C, T and G. The four states are represented by numbers of the quaternary number system. Moreover, we have developed genetic algorithms (GAs) in order to determine the rules of CA evolution that simulate the DNA evolution process. Linear evolution rules were considered and square matrices were used to represent them. If DNA sequences of different evolution steps are available, our approach allows the determination of the underlying evolution rule(s). Conversely, once the evolution rules are deciphered, our tool may reconstruct the DNA sequence in any previous evolution step for which the exact sequence information was unknown. The developed tool may be used to test various parameters that could influence evolution. We describe a paradigm relying on the assumption that mutagenesis is governed by a near-neighbour-dependent mechanism. Based on the satisfactory performance of our system in the deliberately simplified example, we propose that our approach could offer a starting point for future attempts to understand the mechanisms that govern evolution. The developed software is open-source and has a user-friendly graphical input interface.
A system for the 3D reconstruction of retracted-septa PET data using the EM algorithm
International Nuclear Information System (INIS)
Johnson, C.A.; Yan, Y.; Carson, R.E.; Martino, R.L.; Daube-Witherspoon, M.E.
1995-01-01
The authors have implemented the EM reconstruction algorithm for volume acquisition from current generation retracted-septa PET scanners. Although the software was designed for a GE Advance scanner, it is easily adaptable to other 3D scanners. The reconstruction software was written for an Intel iPSC/860 parallel computer with 128 compute nodes. Running on 32 processors, the algorithm requires approximately 55 minutes per iteration to reconstruct a 128 x 128 x 35 image. No projection data compression schemes or other approximations were used in the implementation. Extensive use of EM system matrix (C ij ) symmetries (including the 8-fold in-plane symmetries, 2-fold axial symmetries, and axial parallel line redundancies) reduces the storage cost by a factor of 188. The parallel algorithm operates on distributed projection data which are decomposed by base-symmetry angles. Symmetry operators copy and index the C ij chord to the form required for the particular symmetry. The use of asynchronous reads, lookup tables, and optimized image indexing improves computational performance
International Nuclear Information System (INIS)
Puchner, Stefan B.; Ferencik, Maros; Maurovich-Horvat, Pal; Nakano, Masataka; Otsuka, Fumiyuki; Virmani, Renu; Kauczor, Hans-Ulrich; Hoffmann, Udo; Schlett, Christopher L.
2015-01-01
To evaluate whether iterative reconstruction algorithms improve the diagnostic accuracy of coronary CT angiography (CCTA) for detection of lipid-core plaque (LCP) compared to histology. CCTA and histological data were acquired from three ex vivo hearts. CCTA images were reconstructed using filtered back projection (FBP), adaptive-statistical (ASIR) and model-based (MBIR) iterative algorithms. Vessel cross-sections were co-registered between FBP/ASIR/MBIR and histology. Plaque area 2 : 5.78 ± 2.29 vs. 3.39 ± 1.68 FBP; 5.92 ± 1.87 vs. 3.43 ± 1.62 ASIR; 6.40 ± 1.55 vs. 3.49 ± 1.50 MBIR; all p < 0.0001). AUC for detecting LCP was 0.803/0.850/0.903 for FBP/ASIR/MBIR and was significantly higher for MBIR compared to FBP (p = 0.01). MBIR increased sensitivity for detection of LCP by CCTA. Plaque area <60 HU in CCTA was associated with LCP in histology regardless of the reconstruction algorithm. However, MBIR demonstrated higher accuracy for detecting LCP, which may improve vulnerable plaque detection by CCTA. (orig.)
International Nuclear Information System (INIS)
Bosak, K.; Blum, J.; Joffrin, E.
2004-01-01
Recent development of real-time equilibrium code Equinox using a fixed-point algorithm allow major plasma magnetic parameters to be identified in real-time, using rigorous analytical method. The code relies on the boundary flux code providing magnetic flux values on the first wall of vacuum vessel. By means of least-square minimization of differences between magnetic field obtained from previous solution and the next measurements the code identifies the source term of the non-linear Grad-Shafranov equation. The strict use of analytical equations together with a flexible algorithm offers an opportunity to include new measurements into stable magnetic equilibrium code and compare the results directly between several tokamaks while maintaining the same physical model (i.e. no iron model is necessary inside the equilibrium code). The successful implementation of this equilibrium code for JET and Tore Supra has already been published. In this paper, we show the preliminary results of predictive runs of the Equinox code using the ITER geometry. Because the real-time control experiments of plasma profile at JET using the code has been shown unstable when using magnetic and polarimetric measurements (that could be indirectly translated into accuracy vs robustness tradeoff), we plan an outline of the algorithm that will allow us to further constrain the plasma current profile using the central value of pressure of the plasma in real-time in order to better define the poloidal beta (this constraint is not necessary with purely magnetic equilibrium). (authors)
Energy Technology Data Exchange (ETDEWEB)
Bosak, K.; Blum, J. [Universite de Nice-Sophia-Antipolis, Lab. J. A. Dieudonne, 06 - Nice (France); Joffrin, E. [Association Euratom-CEA Cadarache, 13 - Saint-Paul-lez-Durance (France). Dept. de Recherches sur la Fusion Controlee
2004-07-01
Recent development of real-time equilibrium code Equinox using a fixed-point algorithm allow major plasma magnetic parameters to be identified in real-time, using rigorous analytical method. The code relies on the boundary flux code providing magnetic flux values on the first wall of vacuum vessel. By means of least-square minimization of differences between magnetic field obtained from previous solution and the next measurements the code identifies the source term of the non-linear Grad-Shafranov equation. The strict use of analytical equations together with a flexible algorithm offers an opportunity to include new measurements into stable magnetic equilibrium code and compare the results directly between several tokamaks while maintaining the same physical model (i.e. no iron model is necessary inside the equilibrium code). The successful implementation of this equilibrium code for JET and Tore Supra has already been published. In this paper, we show the preliminary results of predictive runs of the Equinox code using the ITER geometry. Because the real-time control experiments of plasma profile at JET using the code has been shown unstable when using magnetic and polarimetric measurements (that could be indirectly translated into accuracy vs robustness tradeoff), we plan an outline of the algorithm that will allow us to further constrain the plasma current profile using the central value of pressure of the plasma in real-time in order to better define the poloidal beta (this constraint is not necessary with purely magnetic equilibrium). (authors)
Melli, S. Ali; Wahid, Khan A.; Babyn, Paul; Cooper, David M. L.; Gopi, Varun P.
2016-12-01
Synchrotron X-ray Micro Computed Tomography (Micro-CT) is an imaging technique which is increasingly used for non-invasive in vivo preclinical imaging. However, it often requires a large number of projections from many different angles to reconstruct high-quality images leading to significantly high radiation doses and long scan times. To utilize this imaging technique further for in vivo imaging, we need to design reconstruction algorithms that reduce the radiation dose and scan time without reduction of reconstructed image quality. This research is focused on using a combination of gradient-based Douglas-Rachford splitting and discrete wavelet packet shrinkage image denoising methods to design an algorithm for reconstruction of large-scale reduced-view synchrotron Micro-CT images with acceptable quality metrics. These quality metrics are computed by comparing the reconstructed images with a high-dose reference image reconstructed from 1800 equally spaced projections spanning 180°. Visual and quantitative-based performance assessment of a synthetic head phantom and a femoral cortical bone sample imaged in the biomedical imaging and therapy bending magnet beamline at the Canadian Light Source demonstrates that the proposed algorithm is superior to the existing reconstruction algorithms. Using the proposed reconstruction algorithm to reduce the number of projections in synchrotron Micro-CT is an effective way to reduce the overall radiation dose and scan time which improves in vivo imaging protocols.
Liu, Iching; Sun, Ying
1992-10-01
A system for reconstructing 3-D vascular structure from two orthogonally projected images is presented. The formidable problem of matching segments between two views is solved using knowledge of the epipolar constraint and the similarity of segment geometry and connectivity. The knowledge is represented in a rule-based system, which also controls the operation of several computational algorithms for tracking segments in each image, representing 2-D segments with directed graphs, and reconstructing 3-D segments from matching 2-D segment pairs. Uncertain reasoning governs the interaction between segmentation and matching; it also provides a framework for resolving the matching ambiguities in an iterative way. The system was implemented in the C language and the C Language Integrated Production System (CLIPS) expert system shell. Using video images of a tree model, the standard deviation of reconstructed centerlines was estimated to be 0.8 mm (1.7 mm) when the view direction was parallel (perpendicular) to the epipolar plane. Feasibility of clinical use was shown using x-ray angiograms of a human chest phantom. The correspondence of vessel segments between two views was accurate. Computational time for the entire reconstruction process was under 30 s on a workstation. A fully automated system for two-view reconstruction that does not require the a priori knowledge of vascular anatomy is demonstrated.
Directory of Open Access Journals (Sweden)
Jesse Robert Zaneveld
2014-08-01
Full Text Available Complex symbioses between animal or plant hosts and their associated microbiotas can involve thousands of species and millions of genes. Because of the number of interacting partners, it is often impractical to study all organisms or genes in these host-microbe symbioses individually. Yet new phylogenetic predictive methods can use the wealth of accumulated data on diverse model organisms to make inferences into the properties of less well-studied species and gene families. Predictive functional profiling methods use evolutionary models based on the properties of studied relatives to put bounds on the likely characteristics of an organism or gene that has not yet been studied in detail. These techniques have been applied to predict diverse features of host-associated microbial communities ranging from the enzymatic function of uncharacterized genes to the gene content of uncultured microorganisms. We consider these phylogenetically-informed predictive techniques from disparate fields as examples of a general class of algorithms for Hidden State Prediction (HSP, and argue that HSP methods have broad value in predicting organismal traits in a variety of contexts, including the study of complex host-microbe symbioses.
Zaneveld, Jesse R R; Thurber, Rebecca L V
2014-01-01
Complex symbioses between animal or plant hosts and their associated microbiotas can involve thousands of species and millions of genes. Because of the number of interacting partners, it is often impractical to study all organisms or genes in these host-microbe symbioses individually. Yet new phylogenetic predictive methods can use the wealth of accumulated data on diverse model organisms to make inferences into the properties of less well-studied species and gene families. Predictive functional profiling methods use evolutionary models based on the properties of studied relatives to put bounds on the likely characteristics of an organism or gene that has not yet been studied in detail. These techniques have been applied to predict diverse features of host-associated microbial communities ranging from the enzymatic function of uncharacterized genes to the gene content of uncultured microorganisms. We consider these phylogenetically informed predictive techniques from disparate fields as examples of a general class of algorithms for Hidden State Prediction (HSP), and argue that HSP methods have broad value in predicting organismal traits in a variety of contexts, including the study of complex host-microbe symbioses.
Directory of Open Access Journals (Sweden)
Ioannis P. Panapakidis
2018-02-01
Full Text Available Due to high implementation rates of smart meter systems, considerable amount of research is placed in machine learning tools for data handling and information retrieval. A key tool in load data processing is clustering. In recent years, a number of researches have proposed different clustering algorithms in the load profiling field. The present paper provides a methodology for addressing the aforementioned problem through Multi-Criteria Decision Analysis (MCDA and namely, using the Technique for Order of Preference by Similarity to Ideal Solution (TOPSIS. A comparison of the algorithms is employed. Next, a single test case on the selection of an algorithm is examined. User specific weights are applied and based on these weight values, the optimal algorithm is drawn.
International Nuclear Information System (INIS)
Smarda, M; Alexopoulou, E; Mazioti, A; Kordolaimi, S; Ploussi, A; Efstathopoulos, E; Priftis, K
2015-01-01
Purpose of the study is to determine the appropriate iterative reconstruction (IR) algorithm level that combines image quality and diagnostic confidence, for pediatric patients undergoing high-resolution computed tomography (HRCT). During the last 2 years, a total number of 20 children up to 10 years old with a clinical presentation of chronic bronchitis underwent HRCT in our department's 64-detector row CT scanner using the iDose IR algorithm, with almost similar image settings (80kVp, 40-50 mAs). CT images were reconstructed with all iDose levels (level 1 to 7) as well as with filtered-back projection (FBP) algorithm. Subjective image quality was evaluated by 2 experienced radiologists in terms of image noise, sharpness, contrast and diagnostic acceptability using a 5-point scale (1=excellent image, 5=non-acceptable image). Artifacts existance was also pointed out. All mean scores from both radiologists corresponded to satisfactory image quality (score ≤3), even with the FBP algorithm use. Almost excellent (score <2) overall image quality was achieved with iDose levels 5 to 7, but oversmoothing artifacts appearing with iDose levels 6 and 7 affected the diagnostic confidence. In conclusion, the use of iDose level 5 enables almost excellent image quality without considerable artifacts affecting the diagnosis. Further evaluation is needed in order to draw more precise conclusions. (paper)
Duan, Jizhong; Liu, Yu; Jing, Peiguang
2018-02-01
Self-consistent parallel imaging (SPIRiT) is an auto-calibrating model for the reconstruction of parallel magnetic resonance imaging, which can be formulated as a regularized SPIRiT problem. The Projection Over Convex Sets (POCS) method was used to solve the formulated regularized SPIRiT problem. However, the quality of the reconstructed image still needs to be improved. Though methods such as NonLinear Conjugate Gradients (NLCG) can achieve higher spatial resolution, these methods always demand very complex computation and converge slowly. In this paper, we propose a new algorithm to solve the formulated Cartesian SPIRiT problem with the JTV and JL1 regularization terms. The proposed algorithm uses the operator splitting (OS) technique to decompose the problem into a gradient problem and a denoising problem with two regularization terms, which is solved by our proposed split Bregman based denoising algorithm, and adopts the Barzilai and Borwein method to update step size. Simulation experiments on two in vivo data sets demonstrate that the proposed algorithm is 1.3 times faster than ADMM for datasets with 8 channels. Especially, our proposal is 2 times faster than ADMM for the dataset with 32 channels. Copyright © 2017 Elsevier Inc. All rights reserved.
International Nuclear Information System (INIS)
Egger, M.L.; Scheurer, A.H.; Joseph, C.
1996-01-01
The issue of long reconstruction times in PET has been addressed from several points of view, resulting in an affordable dedicated system capable of handling routine 3D reconstruction in a few minutes per frame: on the hardware side using fast processors and a parallel architecture, and on the software side, using efficient implementations of computationally less intensive algorithms. Execution times obtained for the PRT-1 data set on a parallel system of five hybrid nodes, each combining an Alpha processor for computation and a transputer for communication, are the following (256 sinograms of 96 views by 128 radial samples): Ramp algorithm 56 s, Favor 81 s and reprojection algorithm of Kinahan and Rogers 187 s. The implementation of fast rebinning algorithms has shown our hardware platform to become communications-limited; they execute faster on a conventional single-processor Alpha workstation: single-slice rebinning 7 s, Fourier rebinning 22 s, 2D filtered backprojection 5 s. The scalability of the system has been demonstrated, and a saturation effect at network sizes above ten nodes has become visible; new T9000-based products lifting most of the constraints on network topology and link throughput are expected to result in improved parallel efficiency and scalability properties
Development of site profiles for dose reconstruction used in worker compensation claims.
Kenoyer, Judson L; Scalsky, Edward D; Taulbee, Timothy D
2008-07-01
For the purpose of dose reconstruction, personal dosimeter data and measured intakes through bioassay analysis (i.e., in-vivo and in-vitro measurements) should be used whenever possible and given precedence over area monitoring data, survey data, or source term data. However, this is not always possible. A worker's exposure record may be incomplete or missing, or, based on directives and guidelines at the time, a worker may not have been monitored during his or her time of employment. In an effort to recognize, analyze, and incorporate all possible considerations of potential exposures, the National Institute for Occupational Safety and Health Radiation Dose Reconstruction Program developed "site profiles" for all of the major U.S. Department of Energy sites and Atomic Weapons Employer sites. Site profiles are technical documents that (1) provide a brief, general overview of the site; (2) identify the facilities on site with a brief description of the processes and radionuclides used in these processes; (3) contain detailed information on the historical detection limits for film, thermoluminescent dosimeter, and bioassay measurements that are used by the dose reconstructor to interpret a worker's available monitoring records; and (4) provide important supporting information for the dose reconstructor to use if the monitoring data are inadequate or unavailable. When a complete set of monitoring data for an individual is unavailable, it is the parameters in the site profile that are of the most use to the dose reconstructor. These parameters include facility monitoring data (by radionuclide, mechanism of intake, year of exposure, location within a facility); occupational medical x rays and techniques used; environmental measurements (by area on site, radiation type, energy range); minimum detectable activities of the types and kinds of instruments used to detect the different radionuclides; specific source terms (quantities of material and their molecular form
International Nuclear Information System (INIS)
Bettinardi, V.; Gilardi, M.C.; Fazio, F.; Alenius, S.; Ruotsalainen, U.; Numminen, P.; Teraes, M.
2003-01-01
An ordered subsets (OS) reconstruction algorithm based on the median root prior (MRP) and inter-update median filtering was implemented for the reconstruction of low count statistics transmission (TR) scans. The OS-MRP-TR algorithm was evaluated using an experimental phantom, simulating positron emission tomography (PET) whole-body (WB) studies, as well as patient data. Various experimental conditions, in terms of TR scan time (from 1 h to 1 min), covering a wide range of TR count statistics were evaluated. The performance of the algorithm was assessed by comparing the mean value of the attenuation coefficient (MVAC) of known tissue types and the coefficient of variation (CV) for low-count TR images, reconstructed with the OS-MRP-TR algorithm, with reference values obtained from high-count TR images reconstructed with a filtered back-projection (FBP) algorithm. The reconstructed OS-MRP-TR images were then used for attenuation correction of the corresponding emission (EM) data. EM images reconstructed with attenuation correction generated by OS-MRP-TR images, of low count statistics, were compared with the EM images corrected for attenuation using reference (high statistics) TR data. In all the experimental situations considered, the OS-MRP-TR algorithm showed: (1) a tendency towards a stable solution in terms of MVAC; (2) a difference in the MVAC of within 5% for a TR scan of 1 min reconstructed with the OS-MRP-TR and a TR scan of 1 h reconstructed with the FBP algorithm; (3) effectiveness in noise reduction, particularly for low count statistics data [using a specific parameter configuration the TR images reconstructed with OS-MRP-TR(1 min) had a lower CV than the corresponding TR images of a 1-h scan reconstructed with the FBP algorithm]; (4) a difference of within 3% between the mean counts in the EM images attenuation corrected using the OS-MRP-TR images of 1 min and the mean counts in the EM images attenuation corrected using the OS-MRP-TR images of 1 h; (5
Energy Technology Data Exchange (ETDEWEB)
Lee, Eun Sun [Department of Radiology, Seoul National University Hospital (Korea, Republic of); Institute of Radiation Medicine, Seoul National University Hospital (Korea, Republic of); Kim, Se Hyung, E-mail: shkim7071@gmail.com [Department of Radiology, Seoul National University Hospital (Korea, Republic of); Institute of Radiation Medicine, Seoul National University Hospital (Korea, Republic of); Im, Jong Pil; Kim, Sang Gyun [Department of Internal Medicine, Seoul National University Hospital (Korea, Republic of); Shin, Cheong-il; Han, Joon Koo; Choi, Byung Ihn [Department of Radiology, Seoul National University Hospital (Korea, Republic of); Institute of Radiation Medicine, Seoul National University Hospital (Korea, Republic of)
2015-04-15
Highlights: •We assessed the effect of reconstruction algorithms on CAD in ultra-low dose CTC. •30 patients underwent ultra-low dose CTC using 120 and 100 kVp with 10 mAs. •CT was reconstructed with FBP, ASiR and Veo and then, we applied a CAD system. •Per-polyp sensitivity of CAD in ULD CT can be improved with the IR algorithms. •Despite of an increase in the number of FPs with IR, it was still acceptable. -- Abstract: Purpose: To assess the effect of different reconstruction algorithms on computer-aided diagnosis (CAD) performance in ultra-low-dose CT colonography (ULD CTC). Materials and methods: IRB approval and informed consents were obtained. Thirty prospectively enrolled patients underwent non-contrast CTC at 120 kVp/10 mAs in supine and 100 kVp/10 mAs in prone positions, followed by same-day colonoscopy. Images were reconstructed with filtered back projection (FBP), 80% adaptive statistical iterative reconstruction (ASIR80), and model-based iterative reconstruction (MBIR). A commercial CAD system was applied and per-polyp sensitivities and numbers of false-positives (FPs) were compared among algorithms. Results: Mean effective radiation dose of CTC was 1.02 mSv. Of 101 polyps detected and removed by colonoscopy, 61 polyps were detected on supine and on prone CTC datasets on consensus unblinded review, resulting in 122 visible polyps (32 polyps <6 mm, 52 6–9.9 mm, and 38 ≥ 10 mm). Per-polyp sensitivity of CAD for all polyps was highest with MBIR (56/122, 45.9%), followed by ASIR80 (54/122, 44.3%) and FBP (43/122, 35.2%), with significant differences between FBP and IR algorithms (P < 0.017). Per-polyp sensitivity for polyps ≥ 10 mm was also higher with MBIR (25/38, 65.8%) and ASIR80 (24/38, 63.2%) than with FBP (20/38, 58.8%), albeit without statistical significance (P > 0.017). Mean number of FPs was significantly different among algorithms (FBP, 1.4; ASIR, 2.1; MBIR, 2.4) (P = 0.011). Conclusion: Although the performance of stand-alone CAD
International Nuclear Information System (INIS)
Lee, Eun Sun; Kim, Se Hyung; Im, Jong Pil; Kim, Sang Gyun; Shin, Cheong-il; Han, Joon Koo; Choi, Byung Ihn
2015-01-01
Highlights: •We assessed the effect of reconstruction algorithms on CAD in ultra-low dose CTC. •30 patients underwent ultra-low dose CTC using 120 and 100 kVp with 10 mAs. •CT was reconstructed with FBP, ASiR and Veo and then, we applied a CAD system. •Per-polyp sensitivity of CAD in ULD CT can be improved with the IR algorithms. •Despite of an increase in the number of FPs with IR, it was still acceptable. -- Abstract: Purpose: To assess the effect of different reconstruction algorithms on computer-aided diagnosis (CAD) performance in ultra-low-dose CT colonography (ULD CTC). Materials and methods: IRB approval and informed consents were obtained. Thirty prospectively enrolled patients underwent non-contrast CTC at 120 kVp/10 mAs in supine and 100 kVp/10 mAs in prone positions, followed by same-day colonoscopy. Images were reconstructed with filtered back projection (FBP), 80% adaptive statistical iterative reconstruction (ASIR80), and model-based iterative reconstruction (MBIR). A commercial CAD system was applied and per-polyp sensitivities and numbers of false-positives (FPs) were compared among algorithms. Results: Mean effective radiation dose of CTC was 1.02 mSv. Of 101 polyps detected and removed by colonoscopy, 61 polyps were detected on supine and on prone CTC datasets on consensus unblinded review, resulting in 122 visible polyps (32 polyps <6 mm, 52 6–9.9 mm, and 38 ≥ 10 mm). Per-polyp sensitivity of CAD for all polyps was highest with MBIR (56/122, 45.9%), followed by ASIR80 (54/122, 44.3%) and FBP (43/122, 35.2%), with significant differences between FBP and IR algorithms (P < 0.017). Per-polyp sensitivity for polyps ≥ 10 mm was also higher with MBIR (25/38, 65.8%) and ASIR80 (24/38, 63.2%) than with FBP (20/38, 58.8%), albeit without statistical significance (P > 0.017). Mean number of FPs was significantly different among algorithms (FBP, 1.4; ASIR, 2.1; MBIR, 2.4) (P = 0.011). Conclusion: Although the performance of stand-alone CAD
Directory of Open Access Journals (Sweden)
P. Sibanda
2011-02-01
Full Text Available Successful empirical modeling of the topside ionosphere relies on the availability of good quality measured data. The Alouette, ISIS and Intercosmos-19 satellite missions provided large amounts of topside sounder data, but with limited coverage of relevant geophysical conditions (e.g., geographic location, diurnal, seasonal and solar activity by each individual mission. Recently, methods for inferring the electron density distribution in the topside ionosphere from Global Positioning System (GPS-based total electron content (TEC measurements have been developed. This study is focused on the modeling efforts in South Africa and presents the implementation of a technique for reconstructing the topside ionospheric electron density (Ne using a combination of GPS-TEC and ionosonde measurements and empirically obtained Upper Transition Height (UTH. The technique produces reasonable profiles as determined by the global models already in operation. With the added advantage that the constructed profiles are tied to reliable measured GPS-TEC and the empirically determined upper transition height, the technique offers a higher level of confidence in the resulting Ne profiles.
Energy Technology Data Exchange (ETDEWEB)
Velo, Alexandre F.; Carvalho, Diego V.; Alvarez, Alexandre G.; Hamada, Margarida M.; Mesquita, Carlos H., E-mail: afvelo@usp.br [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2017-07-01
The greatest impact of the tomography technology currently occurs in medicine. The success is due to the human body presents standardized dimensions with well-established composition. These conditions are not found in industrial objects. In industry, there is much interest in using the tomography in order to know the inner of (1) the manufactured industrial objects or (2) the machines and their means of production. In these cases, the purpose of the tomography is to (a) control the quality of the final product and (b) to optimize production, contributing to the pilot phase of the projects and analyzing the quality of the means of production. This scan system is a non-destructive, efficient and fast method for providing sectional images of industrial objects and is able to show the dynamic processes and the dispersion of the materials structures within these objects. In this context, it is important that the reconstructed image presents a great spatial resolution with a satisfactory temporal resolution. Thus the algorithm to reconstruct the images has to meet these requirements. This work consists in the analysis of three different iterative algorithm methods, such Maximum Likelihood Estimation Method (MLEM), Maximum Likelihood Transmitted Method (MLTR) and Simultaneous Iterative Reconstruction Method (SIRT. The analysis consists on measurement of the contrast to noise ratio (CNR), the root mean square error (RMSE) and the Modulation Transfer Function (MTF), to know which algorithm fits better the conditions in order to optimize system. The algorithms and the image quality analysis were performed by the Matlab® 2013b. (author)
Lalush, D. S.; Tsui, B. M. W.
1998-06-01
We study the statistical convergence properties of two fast iterative reconstruction algorithms, the rescaled block-iterative (RBI) and ordered subset (OS) EM algorithms, in the context of cardiac SPECT with 3D detector response modeling. The Monte Carlo method was used to generate nearly noise-free projection data modeling the effects of attenuation, detector response, and scatter from the MCAT phantom. One thousand noise realizations were generated with an average count level approximating a typical T1-201 cardiac study. Each noise realization was reconstructed using the RBI and OS algorithms for cases with and without detector response modeling. For each iteration up to twenty, we generated mean and variance images, as well as covariance images for six specific locations. Both OS and RBI converged in the mean to results that were close to the noise-free ML-EM result using the same projection model. When detector response was not modeled in the reconstruction, RBI exhibited considerably lower noise variance than OS for the same resolution. When 3D detector response was modeled, the RBI-EM provided a small improvement in the tradeoff between noise level and resolution recovery, primarily in the axial direction, while OS required about half the number of iterations of RBI to reach the same resolution. We conclude that OS is faster than RBI, but may be sensitive to errors in the projection model. Both OS-EM and RBI-EM are effective alternatives to the EVIL-EM algorithm, but noise level and speed of convergence depend on the projection model used.
International Nuclear Information System (INIS)
Velo, Alexandre F.; Carvalho, Diego V.; Alvarez, Alexandre G.; Hamada, Margarida M.; Mesquita, Carlos H.
2017-01-01
The greatest impact of the tomography technology currently occurs in medicine. The success is due to the human body presents standardized dimensions with well-established composition. These conditions are not found in industrial objects. In industry, there is much interest in using the tomography in order to know the inner of (1) the manufactured industrial objects or (2) the machines and their means of production. In these cases, the purpose of the tomography is to (a) control the quality of the final product and (b) to optimize production, contributing to the pilot phase of the projects and analyzing the quality of the means of production. This scan system is a non-destructive, efficient and fast method for providing sectional images of industrial objects and is able to show the dynamic processes and the dispersion of the materials structures within these objects. In this context, it is important that the reconstructed image presents a great spatial resolution with a satisfactory temporal resolution. Thus the algorithm to reconstruct the images has to meet these requirements. This work consists in the analysis of three different iterative algorithm methods, such Maximum Likelihood Estimation Method (MLEM), Maximum Likelihood Transmitted Method (MLTR) and Simultaneous Iterative Reconstruction Method (SIRT. The analysis consists on measurement of the contrast to noise ratio (CNR), the root mean square error (RMSE) and the Modulation Transfer Function (MTF), to know which algorithm fits better the conditions in order to optimize system. The algorithms and the image quality analysis were performed by the Matlab® 2013b. (author)
Application of Improved APO Algorithm in Vulnerability Assessment and Reconstruction of Microgrid
Xie, Jili; Ma, Hailing
2018-01-01
Artificial Physics Optimization (APO) has good global search ability and can avoid the premature convergence phenomenon in PSO algorithm, which has good stability of fast convergence and robustness. On the basis of APO of the vector model, a reactive power optimization algorithm based on improved APO algorithm is proposed for the static structure and dynamic operation characteristics of microgrid. The simulation test is carried out through the IEEE 30-bus system and the result shows that the algorithm has better efficiency and accuracy compared with other optimization algorithms.
An Algorithm For Climate-Quality Atmospheric Profiling Continuity From EOS Aqua To Suomi-NPP
Moncet, J. L.
2015-12-01
We will present results from an algorithm that is being developed to produce climate-quality atmospheric profiling earth system data records (ESDRs) for application to hyperspectral sounding instrument data from Suomi-NPP, EOS Aqua, and other spacecraft. The current focus is on data from the S-NPP Cross-track Infrared Sounder (CrIS) and Advanced Technology Microwave Sounder (ATMS) instruments as well as the Atmospheric InfraRed Sounder (AIRS) on EOS Aqua. The algorithm development at Atmospheric and Environmental Research (AER) has common heritage with the optimal estimation (OE) algorithm operationally processing S-NPP data in the Interface Data Processing Segment (IDPS), but the ESDR algorithm has a flexible, modular software structure to support experimentation and collaboration and has several features adapted to the climate orientation of ESDRs. Data record continuity benefits from the fact that the same algorithm can be applied to different sensors, simply by providing suitable configuration and data files. The radiative transfer component uses an enhanced version of optimal spectral sampling (OSS) with updated spectroscopy, treatment of emission that is not in local thermodynamic equilibrium (non-LTE), efficiency gains with "global" optimal sampling over all channels, and support for channel selection. The algorithm is designed for adaptive treatment of clouds, with capability to apply "cloud clearing" or simultaneous cloud parameter retrieval, depending on conditions. We will present retrieval results demonstrating the impact of a new capability to perform the retrievals on sigma or hybrid vertical grid (as opposed to a fixed pressure grid), which particularly affects profile accuracy over land with variable terrain height and with sharp vertical structure near the surface. In addition, we will show impacts of alternative treatments of regularization of the inversion. While OE algorithms typically implement regularization by using background estimates from
Energy Technology Data Exchange (ETDEWEB)
Price, Ryan G. [Department of Radiation Oncology, Henry Ford Health Systems, Detroit, Michigan 48202 and Wayne State University School of Medicine, Detroit, Michigan 48201 (United States); Vance, Sean; Cattaneo, Richard; Elshaikh, Mohamed A.; Chetty, Indrin J.; Glide-Hurst, Carri K., E-mail: churst2@hfhs.org [Department of Radiation Oncology, Henry Ford Health Systems, Detroit, Michigan 48202 (United States); Schultz, Lonni [Department of Public Health Sciences, Henry Ford Health Systems, Detroit, Michigan 48202 (United States)
2014-08-15
Purpose: Iterative reconstruction (IR) reduces noise, thereby allowing dose reduction in computed tomography (CT) while maintaining comparable image quality to filtered back-projection (FBP). This study sought to characterize image quality metrics, delineation, dosimetric assessment, and other aspects necessary to integrate IR into treatment planning. Methods: CT images (Brilliance Big Bore v3.6, Philips Healthcare) were acquired of several phantoms using 120 kVp and 25–800 mAs. IR was applied at levels corresponding to noise reduction of 0.89–0.55 with respect to FBP. Noise power spectrum (NPS) analysis was used to characterize noise magnitude and texture. CT to electron density (CT-ED) curves were generated over all IR levels. Uniformity as well as spatial and low contrast resolution were quantified using a CATPHAN phantom. Task specific modulation transfer functions (MTF{sub task}) were developed to characterize spatial frequency across objects of varied contrast. A prospective dose reduction study was conducted for 14 patients undergoing interfraction CT scans for high-dose rate brachytherapy. Three physicians performed image quality assessment using a six-point grading scale between the normal-dose FBP (reference), low-dose FBP, and low-dose IR scans for the following metrics: image noise, detectability of the vaginal cuff/bladder interface, spatial resolution, texture, segmentation confidence, and overall image quality. Contouring differences between FBP and IR were quantified for the bladder and rectum via overlap indices (OI) and Dice similarity coefficients (DSC). Line profile and region of interest analyses quantified noise and boundary changes. For two subjects, the impact of IR on external beam dose calculation was assessed via gamma analysis and changes in digitally reconstructed radiographs (DRRs) were quantified. Results: NPS showed large reduction in noise magnitude (50%), and a slight spatial frequency shift (∼0.1 mm{sup −1}) with
Nguyen, Nam-Ninh; Srihari, Sriganesh; Leong, Hon Wai; Chong, Ket-Fah
2015-10-01
Determining the entire complement of enzymes and their enzymatic functions is a fundamental step for reconstructing the metabolic network of cells. High quality enzyme annotation helps in enhancing metabolic networks reconstructed from the genome, especially by reducing gaps and increasing the enzyme coverage. Currently, structure-based and network-based approaches can only cover a limited number of enzyme families, and the accuracy of homology-based approaches can be further improved. Bottom-up homology-based approach improves the coverage by rebuilding Hidden Markov Model (HMM) profiles for all known enzymes. However, its clustering procedure relies firmly on BLAST similarity score, ignoring protein domains/patterns, and is sensitive to changes in cut-off thresholds. Here, we use functional domain architecture to score the association between domain families and enzyme families (Domain-Enzyme Association Scoring, DEAS). The DEAS score is used to calculate the similarity between proteins, which is then used in clustering procedure, instead of using sequence similarity score. We improve the enzyme annotation protocol using a stringent classification procedure, and by choosing optimal threshold settings and checking for active sites. Our analysis shows that our stringent protocol EnzDP can cover up to 90% of enzyme families available in Swiss-Prot. It achieves a high accuracy of 94.5% based on five-fold cross-validation. EnzDP outperforms existing methods across several testing scenarios. Thus, EnzDP serves as a reliable automated tool for enzyme annotation and metabolic network reconstruction. Available at: www.comp.nus.edu.sg/~nguyennn/EnzDP .
International Nuclear Information System (INIS)
Son, H.H.; Luong, P.T.; Loan, N.T.
1990-04-01
The problems of Remote Sensing (passive or active) are investigated on the base of main principle which consists in interpretation of radiometric electromagnetic measurements in such spectral interval where the radiation is sensitive to interested physical property of medium. Those problems such as an analysis of composition and structure of atmosphere using the records of scattered radiation, cloud identification, investigation of thermodynamic state and composition of system, reconstructing the atmospheric temperature profile on the base of data processing of infrared radiation emitted by system Earth-Atmosphere... belong to class of inverse problems of mathematical physics which are often incorrect. Int his paper a new class of regularized solution corresponding to general formulated RATP-problem is considered. (author). 14 refs, 3 figs, 3 tabs
International Nuclear Information System (INIS)
Irishkin, M.; Imbeaux, F.; Aniel, T.; Artaud, J.F.
2015-01-01
Highlights: • We developed a method for automated comparison of experimental data with models. • A unique platform implements Bayesian analysis and integrated modelling tools. • The method is tokamak-generic and is applied to Tore Supra and JET pulses. • Validation of a heat transport model is carried out. • We quantified the uncertainties due to Te profiles in current diffusion simulations. - Abstract: In the context of present and future long pulse tokamak experiments yielding a growing size of measured data per pulse, automating data consistency analysis and comparisons of measurements with models is a critical matter. To address these issues, the present work describes an expert system that carries out in an integrated and fully automated way (i) a reconstruction of plasma profiles from the measurements, using Bayesian analysis (ii) a prediction of the reconstructed quantities, according to some models and (iii) a comparison of the first two steps. The first application shown is devoted to the development of an automated comparison method between the experimental plasma profiles reconstructed using Bayesian methods and time dependent solutions of the transport equations. The method was applied to model validation of a simple heat transport model with three radial shape options. It has been tested on a database of 21 Tore Supra and 14 JET shots. The second application aims at quantifying uncertainties due to the electron temperature profile in current diffusion simulations. A systematic reconstruction of the Ne, Te, Ti profiles was first carried out for all time slices of the pulse. The Bayesian 95% highest probability intervals on the Te profile reconstruction were then used for (i) data consistency check of the flux consumption and (ii) defining a confidence interval for the current profile simulation. The method has been applied to one Tore Supra pulse and one JET pulse.
Direct cone beam SPECT reconstruction with camera tilt
International Nuclear Information System (INIS)
Jianying Li; Jaszczak, R.J.; Greer, K.L.; Coleman, R.E.; Zongjian Cao; Tsui, B.M.W.
1993-01-01
A filtered backprojection (FBP) algorithm is derived to perform cone beam (CB) single-photon emission computed tomography (SPECT) reconstruction with camera tilt using circular orbits. This algorithm reconstructs the tilted angle CB projection data directly by incorporating the tilt angle into it. When the tilt angle becomes zero, this algorithm reduces to that of Feldkamp. Experimentally acquired phantom studies using both a two-point source and the three-dimensional Hoffman brain phantom have been performed. The transaxial tilted cone beam brain images and profiles obtained using the new algorithm are compared with those without camera tilt. For those slices which have approximately the same distance from the detector in both tilt and non-tilt set-ups, the two transaxial reconstructions have similar profiles. The two-point source images reconstructed from this new algorithm and the tilted cone beam brain images are also compared with those reconstructed from the existing tilted cone beam algorithm. (author)
Chen, Chi-Kan
2017-07-26
The identification of genetic regulatory networks (GRNs) provides insights into complex cellular processes. A class of recurrent neural networks (RNNs) captures the dynamics of GRN. Algorithms combining the RNN and machine learning schemes were proposed to reconstruct small-scale GRNs using gene expression time series. We present new GRN reconstruction methods with neural networks. The RNN is extended to a class of recurrent multilayer perceptrons (RMLPs) with latent nodes. Our methods contain two steps: the edge rank assignment step and the network construction step. The former assigns ranks to all possible edges by a recursive procedure based on the estimated weights of wires of RNN/RMLP (RE RNN /RE RMLP ), and the latter constructs a network consisting of top-ranked edges under which the optimized RNN simulates the gene expression time series. The particle swarm optimization (PSO) is applied to optimize the parameters of RNNs and RMLPs in a two-step algorithm. The proposed RE RNN -RNN and RE RMLP -RNN algorithms are tested on synthetic and experimental gene expression time series of small GRNs of about 10 genes. The experimental time series are from the studies of yeast cell cycle regulated genes and E. coli DNA repair genes. The unstable estimation of RNN using experimental time series having limited data points can lead to fairly arbitrary predicted GRNs. Our methods incorporate RNN and RMLP into a two-step structure learning procedure. Results show that the RE RMLP using the RMLP with a suitable number of latent nodes to reduce the parameter dimension often result in more accurate edge ranks than the RE RNN using the regularized RNN on short simulated time series. Combining by a weighted majority voting rule the networks derived by the RE RMLP -RNN using different numbers of latent nodes in step one to infer the GRN, the method performs consistently and outperforms published algorithms for GRN reconstruction on most benchmark time series. The framework of two
International Nuclear Information System (INIS)
Tseng, Hsin-Wu; Kupinski, Matthew A.; Fan, Jiahua; Sainath, Paavana; Hsieh, Jiang
2014-01-01
Purpose: A number of different techniques have been developed to reduce radiation dose in x-ray computed tomography (CT) imaging. In this paper, the authors will compare task-based measures of image quality of CT images reconstructed by two algorithms: conventional filtered back projection (FBP), and a new iterative reconstruction algorithm (IR). Methods: To assess image quality, the authors used the performance of a channelized Hotelling observer acting on reconstructed image slices. The selected channels are dense difference Gaussian channels (DDOG).A body phantom and a head phantom were imaged 50 times at different dose levels to obtain the data needed to assess image quality. The phantoms consisted of uniform backgrounds with low contrast signals embedded at various locations. The tasks the observer model performed included (1) detection of a signal of known location and shape, and (2) detection and localization of a signal of known shape. The employed DDOG channels are based on the response of the human visual system. Performance was assessed using the areas under ROC curves and areas under localization ROC curves. Results: For signal known exactly (SKE) and location unknown/signal shape known tasks with circular signals of different sizes and contrasts, the authors’ task-based measures showed that a FBP equivalent image quality can be achieved at lower dose levels using the IR algorithm. For the SKE case, the range of dose reduction is 50%–67% (head phantom) and 68%–82% (body phantom). For the study of location unknown/signal shape known, the dose reduction range can be reached at 67%–75% for head phantom and 67%–77% for body phantom case. These results suggest that the IR images at lower dose settings can reach the same image quality when compared to full dose conventional FBP images. Conclusions: The work presented provides an objective way to quantitatively assess the image quality of a newly introduced CT IR algorithm. The performance of the
Impact of the algorithm of iterative reconstruction ASIR in the CTDI of studies in TCHMC
International Nuclear Information System (INIS)
Ambroa Rey, E. M.; Vazquez Vazquez, R.; Gimenez Insua, M.; Sanchez Garcia, M.; Otero Martinez, C.; Luna Vega, V.; Mosquera Sueiro, J.; Lobato Busto, R.; Pombar Camean, M.
2013-01-01
The objective of this work is to make a comparison of the doses in the 10 protocols most commonly used in our Center, before and after the commissioning of the software ASIR (Adaptive statistical iterative reconstruction). (Author)
Ltaief, Hatem
2011-08-31
This paper presents the power profile of two high performance dense linear algebra libraries i.e., LAPACK and PLASMA. The former is based on block algorithms that use the fork-join paradigm to achieve parallel performance. The latter uses fine-grained task parallelism that recasts the computation to operate on submatrices called tiles. In this way tile algorithms are formed. We show results from the power profiling of the most common routines, which permits us to clearly identify the different phases of the computations. This allows us to isolate the bottlenecks in terms of energy efficiency. Our results show that PLASMA surpasses LAPACK not only in terms of performance but also in terms of energy efficiency. © 2011 Springer-Verlag.
International Nuclear Information System (INIS)
Barras, Heloise; Dunet, Vincent; Hachulla, Anne-Lise; Grimm, Jochen; Beigelman-Aubry, Catherine
2016-01-01
Model-based iterative reconstruction (MBIR) reduces image noise and improves image quality (IQ) but its influence on post-processing tools including maximal intensity projection (MIP) and minimal intensity projection (mIP) remains unknown. To evaluate the influence on IQ of MBIR on native, mIP, MIP axial and coronal reformats of reduced dose computed tomography (RD-CT) chest acquisition. Raw data of 50 patients, who underwent a standard dose CT (SD-CT) and a follow-up RD-CT with a CT dose index (CTDI) of 2–3 mGy, were reconstructed by MBIR and FBP. Native slices, 4-mm-thick MIP, and 3-mm-thick mIP axial and coronal reformats were generated. The relative IQ, subjective IQ, image noise, and number of artifacts were determined in order to compare different reconstructions of RD-CT with reference SD-CT. The lowest noise was observed with MBIR. RD-CT reconstructed by MBIR exhibited the best relative and subjective IQ on coronal view regardless of the post-processing tool. MBIR generated the lowest rate of artefacts on coronal mIP/MIP reformats and the highest one on axial reformats, mainly represented by distortions and stairsteps artifacts. The MBIR algorithm reduces image noise but generates more artifacts than FBP on axial mIP and MIP reformats of RD-CT. Conversely, it significantly improves IQ on coronal views, without increasing artifacts, regardless of the post-processing technique
International Nuclear Information System (INIS)
Birguel, Oezlem; Eyueboglu, B Murat; Ider, Y Ziya
2003-01-01
Conventional injected-current electrical impedance tomography (EIT) and magnetic resonance imaging (MRI) techniques can be combined to reconstruct high resolution true conductivity images. The magnetic flux density distribution generated by the internal current density distribution is extracted from MR phase images. This information is used to form a fine detailed conductivity image using an Ohm's law based update equation. The reconstructed conductivity image is assumed to differ from the true image by a scale factor. EIT surface potential measurements are then used to scale the reconstructed image in order to find the true conductivity values. This process is iterated until a stopping criterion is met. Several simulations are carried out for opposite and cosine current injection patterns to select the best current injection pattern for a 2D thorax model. The contrast resolution and accuracy of the proposed algorithm are also studied. In all simulation studies, realistic noise models for voltage and magnetic flux density measurements are used. It is shown that, in contrast to the conventional EIT techniques, the proposed method has the capability of reconstructing conductivity images with uniform and high spatial resolution. The spatial resolution is limited by the larger element size of the finite element mesh and twice the magnetic resonance image pixel size
A new approach for quantitative evaluation of reconstruction algorithms in SPECT
International Nuclear Information System (INIS)
Raeisi, E.; Rajabi, H.; Aghamiri, S. M. R.
2006-01-01
In nuclear medicine, phantoms are mainly used to evaluate the overall performance of the imaging systems, and practically there is no phantom exclusively designed for the evaluation of the software performance. In this study the Hoffman brain phantom was used for quantitative evaluation of reconstruction techniques. The phantom is modified to acquire tomographic and planar image of the same structure. The planar image may be used as the reference image to evaluate the quality of reconstructed slices, using the companion software developed in MATLAB. Materials and Methods: The designed phantom was composed of 4 independent 2D slices that could have been placed juxtapose to the 3D phantom. Each slice was composed of objects of different size and shape (for example: circle, triangle, and rectangle). Each 2D slice was imaged at distances ranging from 0 to 15 cm from the collimator surface. The phantom in 3D configuration was imaged acquiring 128 views of 128*128 matrix size. Reconstruction was performed using different filtering condition and the reconstructed images were compared to the corresponding planar images. The modulation transfer function, scatter fraction and attenuation map were calculated for each reconstructed image. Results: Since all the parameters of the acquisition were identical for the 2D and the 3D imaging, it was assumed that the difference in the quality of the images has exclusively been due to the reconstruction condition. The planar images were assumed to be the most perfect images which could be obtained with the system. The comparison of the reconstructed slices with the corresponding planar images yielded the optimum reconstruction condition. The results clearly showed that Wiener filter yields superior quality image among the entire tested filters. The extent of the improvement has been quantified in terms of universal image quality index. Conclusion : The phantom and the accompanying software were evaluated and found to be quite useful in
Directory of Open Access Journals (Sweden)
Maryam Shokouhi
2017-06-01
Full Text Available Distribution networks are designed as a ring and operated as a radial form. Therefore, the reconfiguration is a simple and cost-effective way to use existing facilities without the need for any new equipment in distribution networks to achieve various objectives such as: power loss reduction, feeder overload reduction, load balancing, voltage profile improvement, reducing the number of switching considering constraints that ultimately result in the power loss reduction. In this paper, a new method based on the Ant Lion algorithm (a modern meta-heuristic algorithm is provided for the reconfiguration of distribution networks. Considering the extension of the distribution networks and complexity of their communications networks, and the various parameters, using smart techniques is inevitable. The proposed approach is tested on the IEEE 33 & 69-bus radial standard distribution networks. The Evaluation of results in MATLAB software shows the effectiveness of the Ant Lion algorithm in the distribution network reconfiguration.
Maldonado Puente, Bryan Patricio
2014-01-01
The inner detector of the ATLAS experiment has two types of silicon detectors used for tracking: Pixel Detector and SCT (semiconductor tracker). Once a proton-proton collision occurs, the result- ing particles pass through these detectors and these are recorded as hits on the detector surfaces. A medium to high energy particle passes through seven different surfaces of the two detectors, leaving seven hits, while lower energy particles can leave many more hits as they circle through the detector. For a typical event during the expected operational conditions, there are 30 000 hits in average recorded by the sensors. Only high energy particles are of interest for physics analysis and are taken into account for the path reconstruction; thus, a filtering process helps to discard the low energy particles produced in the collision. The following report presents a solution for increasing the speed of the filtering process in the path reconstruction algorithm.
Luck, Margaux; Schmitt, Caroline; Talbi, Neila; Gouya, Laurent; Caradeuc, Cédric; Puy, Hervé; Bertho, Gildas; Pallet, Nicolas
2018-01-01
Metabolomic profiling combines Nuclear Magnetic Resonance spectroscopy with supervised statistical analysis that might allow to better understanding the mechanisms of a disease. In this study, the urinary metabolic profiling of individuals with porphyrias was performed to predict different types of disease, and to propose new pathophysiological hypotheses. Urine 1 H-NMR spectra of 73 patients with asymptomatic acute intermittent porphyria (aAIP) and familial or sporadic porphyria cutanea tarda (f/sPCT) were compared using a supervised rule-mining algorithm. NMR spectrum buckets bins, corresponding to rules, were extracted and a logistic regression was trained. Our rule-mining algorithm generated results were consistent with those obtained using partial least square discriminant analysis (PLS-DA) and the predictive performance of the model was significant. Buckets that were identified by the algorithm corresponded to metabolites involved in glycolysis and energy-conversion pathways, notably acetate, citrate, and pyruvate, which were found in higher concentrations in the urines of aAIP compared with PCT patients. Metabolic profiling did not discriminate sPCT from fPCT patients. These results suggest that metabolic reprogramming occurs in aAIP individuals, even in the absence of overt symptoms, and supports the relationship that occur between heme synthesis and mitochondrial energetic metabolism.
Hardware-Efficient Design of Real-Time Profile Shape Matching Stereo Vision Algorithm on FPGA
Directory of Open Access Journals (Sweden)
Beau Tippetts
2014-01-01
Full Text Available A variety of platforms, such as micro-unmanned vehicles, are limited in the amount of computational hardware they can support due to weight and power constraints. An efficient stereo vision algorithm implemented on an FPGA would be able to minimize payload and power consumption in microunmanned vehicles, while providing 3D information and still leaving computational resources available for other processing tasks. This work presents a hardware design of the efficient profile shape matching stereo vision algorithm. Hardware resource usage is presented for the targeted micro-UV platform, Helio-copter, that uses the Xilinx Virtex 4 FX60 FPGA. Less than a fifth of the resources on this FGPA were used to produce dense disparity maps for image sizes up to 450 × 375, with the ability to scale up easily by increasing BRAM usage. A comparison is given of accuracy, speed performance, and resource usage of a census transform-based stereo vision FPGA implementation by Jin et al. Results show that the profile shape matching algorithm is an efficient real-time stereo vision algorithm for hardware implementation for resource limited systems such as microunmanned vehicles.
Directory of Open Access Journals (Sweden)
Charles M Malata
2015-10-01
Full Text Available Introduction: The deep inferior epigastric artery perforator (DIEP free flap is the gold standard for autologous breast reconstruction. However, using a single vascular pedicle may not yield sufficient tissue in patients with midline scars or insufficient lower abdominal pannus. Double-pedicled free flaps overcome this problem using different vascular arrangements to harvest the entire lower abdominal flap. The literature is, however, sparse regarding technique selection. We therefore reviewed our experience in order to formulate an algorithm and comprehensive classification for this purpose. Methods: All patients undergoing unilateral double-pedicled abdominal perforator free flap breast reconstruction (AFFBR by a single surgeon (CMM over 40 months were reviewed from a prospectively collected database. Results: Of the 112 consecutive breast free flaps performed, 25 (22% utilised two vascular pedicles. The mean patient age was 45 years (range=27-54. All flaps but one (which used the thoracodorsal system were anastomosed to the internal mammary vessels using the rib-preservation technique. The surgical duration was 656 minutes (range=468-690 mins. The median flap weight was 618g (range=432-1275g and the mastectomy weight was 445g (range=220-896g. All flaps were successful and only three patients requested minor liposuction to reduce and reshape their reconstructed breasts.Conclusion: Bipedicled free abdominal perforator flaps, employed in a fifth of all our AFFBRs, are a reliable and safe option for unilateral breast reconstruction. They, however, necessitate clear indications to justify the additional technical complexity and surgical duration. Our algorithm and comprehensive classification facilitate technique selection for the anastomotic permutations and successful execution of these operations.
Indian Academy of Sciences (India)
polynomial) division have been found in Vedic Mathematics which are dated much before Euclid's algorithm. A programming language Is used to describe an algorithm for execution on a computer. An algorithm expressed using a programming.
Energy Technology Data Exchange (ETDEWEB)
Puchner, Stefan B. [Massachusetts General Hospital, Harvard Medical School, Cardiac MR PET CT Program, Department of Radiology, Boston, MA (United States); Medical University of Vienna, Department of Biomedical Imaging and Image-Guided Therapy, Vienna (Austria); Ferencik, Maros [Massachusetts General Hospital, Harvard Medical School, Cardiac MR PET CT Program, Department of Radiology, Boston, MA (United States); Harvard Medical School, Division of Cardiology, Massachusetts General Hospital, Boston, MA (United States); Maurovich-Horvat, Pal [Massachusetts General Hospital, Harvard Medical School, Cardiac MR PET CT Program, Department of Radiology, Boston, MA (United States); Semmelweis University, MTA-SE Lenduelet Cardiovascular Imaging Research Group, Heart and Vascular Center, Budapest (Hungary); Nakano, Masataka; Otsuka, Fumiyuki; Virmani, Renu [CV Path Institute Inc., Gaithersburg, MD (United States); Kauczor, Hans-Ulrich [University Hospital Heidelberg, Ruprecht-Karls-University of Heidelberg, Department of Diagnostic and Interventional Radiology, Heidelberg (Germany); Hoffmann, Udo [Massachusetts General Hospital, Harvard Medical School, Cardiac MR PET CT Program, Department of Radiology, Boston, MA (United States); Schlett, Christopher L. [Massachusetts General Hospital, Harvard Medical School, Cardiac MR PET CT Program, Department of Radiology, Boston, MA (United States); University Hospital Heidelberg, Ruprecht-Karls-University of Heidelberg, Department of Diagnostic and Interventional Radiology, Heidelberg (Germany)
2015-01-15
To evaluate whether iterative reconstruction algorithms improve the diagnostic accuracy of coronary CT angiography (CCTA) for detection of lipid-core plaque (LCP) compared to histology. CCTA and histological data were acquired from three ex vivo hearts. CCTA images were reconstructed using filtered back projection (FBP), adaptive-statistical (ASIR) and model-based (MBIR) iterative algorithms. Vessel cross-sections were co-registered between FBP/ASIR/MBIR and histology. Plaque area <60 HU was semiautomatically quantified in CCTA. LCP was defined by histology as fibroatheroma with a large lipid/necrotic core. Area under the curve (AUC) was derived from logistic regression analysis as a measure of diagnostic accuracy. Overall, 173 CCTA triplets (FBP/ASIR/MBIR) were co-registered with histology. LCP was present in 26 cross-sections. Average measured plaque area <60 HU was significantly larger in LCP compared to non-LCP cross-sections (mm{sup 2}: 5.78 ± 2.29 vs. 3.39 ± 1.68 FBP; 5.92 ± 1.87 vs. 3.43 ± 1.62 ASIR; 6.40 ± 1.55 vs. 3.49 ± 1.50 MBIR; all p < 0.0001). AUC for detecting LCP was 0.803/0.850/0.903 for FBP/ASIR/MBIR and was significantly higher for MBIR compared to FBP (p = 0.01). MBIR increased sensitivity for detection of LCP by CCTA. Plaque area <60 HU in CCTA was associated with LCP in histology regardless of the reconstruction algorithm. However, MBIR demonstrated higher accuracy for detecting LCP, which may improve vulnerable plaque detection by CCTA. (orig.)
Albert, A.; André, M.; Anghinolfi, M.; Anton, G.; Ardid, M.; Aubert, J.-J.; Avgitas, T.; Baret, B.; Barrios-Martí, J.; Basa, S.; Bertin, V.; Biagi, S.; Bormuth, R.; Bourret, S.; Bouwhuis, M.C.; Bruijn, R.; Brunner, J.; Busto, J.; Capone, A.; Caramete, L.; Carr, J.; Celli, S.; Chiarusi, T.; Circella, M.; Coelho, C.O.A.; Coleiro, A.; Coniglione, R.; Costantini, H.; Coyle, P.; Creusot, A.; Deschamps, A.; De Bonis, G.; Distefano, C.; Di Palma, I.; Domi, A.; Donzaud, C.; Dornic, D.; Drouhin, D.; Eberl, T.; El Bojaddaini, I.; Elsässer, D.; Enzenhofer, A.; Felis, I.; Folger, F.; Fusco, L.A.; Galata, S.; Gay, P.; Giordano, V.; Glotin, H.; Grégoire, T.; Gracia-Ruiz, R.; Graf, K.; Hallmann, S.; van Haren, H.; Heijboer, A.J.; Hello, Y.; Hernandez-Rey, J.J.; Hößl, J.; Hofestädt, J.; Hugon, C.; Illuminati, G.; James, C.W.; de Jong, M.; Jongen, M.; Kadler, M.; Kalekin, O.; Katz, U.; Kießling, D.; Kouchner, A.; Kreter, M.; Kreykenbohm, I.; Kulikovskiy, V.; Lachaud, C.; Lahmann, R.; Lefevre, D.; Leonora, E.; Lotze, M.; Loucatos, S.; Marcelin, M.; Margiotta, A.; Marinelli, A.; Martinez-Mora, J.A.; Mele, R.; Melis, K.; Michael, T.; Migliozzi, P.; Moussa, A.; Nezri, E.; Organokov, M.; Pavalas, G.E.; Pellegrino, C.; Perrina, C.; Piattelli, P.; Popa, V.; Pradier, T.; Quinn, L.; Racca, C.; Riccobene, G.; Sanchez-Losa, A.; Saldaña, M.; Salvadori, I.; Samtleben, D.F.E.; Sanguineti, M.; Sapienza, P.; Schussler, F.; Sieger, C.; Spurio, M.; Stolarczyk, T.; Taiuti, M.; Tayalati, Y.; Trovato, A.; Turpin, D.; Tönnis, C.; Vallage, B.; Van Elewyck, V.; Versari, F.; Vivolo, D.; Vizzoca, A.; Wilms, J.; Zornoza, J.D.; Zuniga, J.
2017-01-01
A novel algorithm to reconstruct neutrino-induced particle showers within the ANTARES neutrino telescope is presented. The method achieves a median angular resolution of 6∘ for shower energies below 100 TeV. Applying this algorithm to 6 years of data taken with the ANTARES detector, 8 events with
International Nuclear Information System (INIS)
Arpinar, V E; Muftuler, L T; Hamamura, M J; Degirmenci, E
2012-01-01
Magnetic resonance electrical impedance tomography (MREIT) is a technique that produces images of conductivity in tissues and phantoms. In this technique, electrical currents are applied to an object and the resulting magnetic flux density is measured using magnetic resonance imaging (MRI) and the conductivity distribution is reconstructed using these MRI data. Currently, the technique is used in research environments, primarily studying phantoms and animals. In order to translate MREIT to clinical applications, strict safety standards need to be established, especially for safe current limits. However, there are currently no standards for safe current limits specific to MREIT. Until such standards are established, human MREIT applications need to conform to existing electrical safety standards in medical instrumentation, such as IEC601. This protocol limits patient auxiliary currents to 100 µA for low frequencies. However, published MREIT studies have utilized currents 10–400 times larger than this limit, bringing into question whether the clinical applications of MREIT are attainable under current standards. In this study, we investigated the feasibility of MREIT to accurately reconstruct the relative conductivity of a simple agarose phantom using 200 µA total injected current and tested the performance of two MREIT reconstruction algorithms. These reconstruction algorithms used are the iterative sensitivity matrix method (SMM) by Ider and Birgul (1998 Elektrik 6 215–25) with Tikhonov regularization and the harmonic B Z proposed by Oh et al (2003 Magn. Reason. Med. 50 875–8). The reconstruction techniques were tested at both 200 µA and 5 mA injected currents to investigate their noise sensitivity at low and high current conditions. It should be noted that 200 µA total injected current into a cylindrical phantom generates only 14.7 µA current in imaging slice. Similarly, 5 mA total injected current results in 367 µA in imaging slice. Total acquisition
International Nuclear Information System (INIS)
Kacarska, Marija; Loskovska, Suzana
2002-01-01
In this paper comparative analysis between different EIT algorithms is presented. Analysis is made for spatial and temporal resolution of obtained images by several different algorithms. Discussions consider spatial resolution dependent on data acquisition method, too. Obtained results show that conventional applied-current EIT is more powerful compared to induced-current EIT. (Author)
International Nuclear Information System (INIS)
Tang Xiangyang; Hsieh Jiang; Nilsen, Roy A; Dutta, Sandeep; Samsonov, Dmitry; Hagiwara, Akira
2006-01-01
Based on the structure of the original helical FDK algorithm, a three-dimensional (3D)-weighted cone beam filtered backprojection (CB-FBP) algorithm is proposed for image reconstruction in volumetric CT under helical source trajectory. In addition to its dependence on view and fan angles, the 3D weighting utilizes the cone angle dependency of a ray to improve reconstruction accuracy. The 3D weighting is ray-dependent and the underlying mechanism is to give a favourable weight to the ray with the smaller cone angle out of a pair of conjugate rays but an unfavourable weight to the ray with the larger cone angle out of the conjugate ray pair. The proposed 3D-weighted helical CB-FBP reconstruction algorithm is implemented in the cone-parallel geometry that can improve noise uniformity and image generation speed significantly. Under the cone-parallel geometry, the filtering is naturally carried out along the tangential direction of the helical source trajectory. By exploring the 3D weighting's dependence on cone angle, the proposed helical 3D-weighted CB-FBP reconstruction algorithm can provide significantly improved reconstruction accuracy at moderate cone angle and high helical pitches. The 3D-weighted CB-FBP algorithm is experimentally evaluated by computer-simulated phantoms and phantoms scanned by a diagnostic volumetric CT system with a detector dimension of 64 x 0.625 mm over various helical pitches. The computer simulation study shows that the 3D weighting enables the proposed algorithm to reach reconstruction accuracy comparable to that of exact CB reconstruction algorithms, such as the Katsevich algorithm, under a moderate cone angle (4 deg.) and various helical pitches. Meanwhile, the experimental evaluation using the phantoms scanned by a volumetric CT system shows that the spatial resolution along the z-direction and noise characteristics of the proposed 3D-weighted helical CB-FBP reconstruction algorithm are maintained very well in comparison to the FDK
Cordeiro, Peter G; Chen, Constance M
2012-01-01
Reconstruction of complex midfacial defects is best approached with a clear algorithm. The goals of reconstruction are functional and aesthetic. Over a 15-year period (1992 to 2006), a single surgeon (P.G.C.) performed 100 flaps to reconstruct the following midfacial defects: type I, limited maxillectomy (n = 20); type IIA, subtotal maxillectomy with resection of less than 50 percent of the palate (n = 8); type IIB, subtotal maxillectomy with resection of greater than 50 percent of the palate (n = 8); type IIIA, total maxillectomy with preservation of the orbital contents (n = 22); type IIIB, total maxillectomy with orbital exenteration (n = 23); and type IV, orbitomaxillectomy (n = 19). Free flaps were used in 94 cases (94 percent), and pedicled flaps were used in six (6 percent). One hundred flaps were performed in 96 patients (69 males, 72 percent; 27 females, 28 percent); four patients underwent a second flap reconstruction due to recurrent disease (n = 4, 4 percent). Average patient age was 49.2 years (range, 13 to 81 years). Free-flap survival was 100 percent, with one partial flap loss (1 percent). Five patients suffered systemic complications (5.2 percent), and four died within 30 days of hospitalization (4.2 percent). Over 50 percent of patients returned to normal diet and speech. Almost 60 percent were judged to have an excellent aesthetic result. Free-tissue transfer offers the most effective and reliable form of reconstruction for complex maxillectomy defects. Rectus abdominis and radial forearm free flaps in combination with immediate bone grafting or as osteocutaneous flaps consistently provide the best functional and aesthetic results. Therapeutic, IV.
Kim, Hyun Keol; Montejo, Ludguier D; Jia, Jingfei; Hielscher, Andreas H
2017-06-01
We introduce here the finite volume formulation of the frequency-domain simplified spherical harmonics model with n -th order absorption coefficients (FD-SP N ) that approximates the frequency-domain equation of radiative transfer (FD-ERT). We then present the FD-SP N based reconstruction algorithm that recovers absorption and scattering coefficients in biological tissue. The FD-SP N model with 3 rd order absorption coefficient (i.e., FD-SP 3 ) is used as a forward model to solve the inverse problem. The FD-SP 3 is discretized with a node-centered finite volume scheme and solved with a restarted generalized minimum residual (GMRES) algorithm. The absorption and scattering coefficients are retrieved using a limited-memory Broyden-Fletcher-Goldfarb-Shanno (L-BFGS) algorithm. Finally, the forward and inverse algorithms are evaluated using numerical phantoms with optical properties and size that mimic small-volume tissue such as finger joints and small animals. The forward results show that the FD-SP 3 model approximates the FD-ERT (S 12 ) solution within relatively high accuracy; the average error in the phase (<3.7%) and the amplitude (<7.1%) of the partial current at the boundary are reported. From the inverse results we find that the absorption and scattering coefficient maps are more accurately reconstructed with the SP 3 model than those with the SP 1 model. Therefore, this work shows that the FD-SP 3 is an efficient model for optical tomographic imaging of small-volume media with non-diffuse properties both in terms of computational time and accuracy as it requires significantly lower CPU time than the FD-ERT (S 12 ) and also it is more accurate than the FD-SP 1 .
DART: a robust algorithm for fast reconstruction of three-dimensional grain maps
DEFF Research Database (Denmark)
Batenburg, K.J.; Sijbers, J.; Poulsen, Henning Friis
2010-01-01
and moderate noise levels, DART is shown to generate essentially perfect two-dimensional grain maps for as few as three projections per grain with running times on a PC in the range of less than a second. This is seen as opening up the possibility for fast reconstructions in connection with in situ studies....
An algorithm for the reconstruction of neutrino-induced showers in the ANTARES neutrino telescope
Albert, A.; André, M.; Anghinolfi, M.; Anton, G.; Ardid, M.; Aubert, J.-J.; Avgitas, T.; Baret, B.; Barrios-Martí, J.; Basa, S.; Belhorma, B.; Bertin, V.; Biagi, S.; Bormuth, R.; Bourret, S.; Bouwhuis, M.C.; Brânzas, H.; Bruijn, R.; Brunner, J.; Busto, J.; Capone, A.; Caramete, L.; Carr, J.; Celli, S.; El Moursli, R.C.; Chiarusi, T.; Circella, M.; Coelho, J.A.B.; Coleiro, A.; Coniglione, R.; Costantini, H.; Coyle, P.; Creusot, A.; Díaz, A.F.; Deschamps, A.; de Boris, G.; Distefano, C.; Di Palma, I.; Domi, A.; Donzaud, C.; Dornic, D.; Drouhin, D.; Eberl, T.; El Bojaddaini, I.; El Khayati, N.; Elsässer, D.; Enzenhöfer, A.; Ettahiri, A.; Fassi, F.; Felis, I.; Fusco, L.A.; Gay, P.; Giordano, V.; Glotin, H.; Grégoire, T.; Garcia Ruiz, R.; Graf, K.; Hallmann, S.; van Haren, H.; Heijboer, A.J.; Hello, Y.; Hernandez-Rey, J.J.; Hößl, J.; Hofestädt, J.; Hugon, C.; Illuminati, G.; James, J.W.; de Jong, M.; Jongen, M.; Kadler, M.; Kalekin, O.; Katz, U.; Kießling, D.; Kouchner, A.; Kreter, M.; Kreykenbohm, I.; Kulikovskiy, V.; Lachaud, C.; Lahmann, R.; Lefevre, D.; Leonora, E.; Lotze, M.; Loucatos, S.; Marcelin, M.; Margiotta, A.; Marinelli, A.; Martinez-Mora, J.A.; Mele, R.; Melis, K.; Michael, T.; Migliozzi, P.; Moussa, A.; Navas, S.; Nezri, E.; Organokov, M.; Pavalas, G.E.; Pellegrino, C.; Perrina, C.; Piattelli, P.; Popa, V.; Pradier, T.; Quinn, L.; Racca, C.; Riccobene, G.; Sánchez-Losa, A.; Saldaña, M.; Salvadori, I.; Samtleben, D.F.E.; Sanguineti, M.; Sapienza, P.; Schüssler, F.; Sieger, C.; Spurio, M.; Stolarczyk, Th.; Taiuti, M.; Tayalati, Y.; Trovato, A.; Turpin, D.; Tönnis, C.; Vallage, B.; Van Elewyck, V.; Versari, F.; Vivolo, D.; Vizzoca, A.; Wilms, J.; Zornoza, J.D.; Zúñiga, J.
2017-01-01
Muons created by nm charged current (CC) interactions in the water surrounding the ANTARES neutrino telescopehave been almost exclusively used so far in searches for cosmic neutrino sources. Due to their long range, highlyenergetic muons inducing Cherenkov radiation in the water are reconstructed
A reconstruction algorithm for electrical impedance tomography based on sparsity regularization
Jin, Bangti; Khan, Taufiquar; Maass, Peter
2011-01-01
type was proposed. Numerical results for several two-dimensional problems with both single and multiple convex and nonconvex inclusions were presented to illustrate the features of the proposed algorithm and were compared with one conventional approach
Duan, Yuping; Bouslimi, Dalel; Yang, Guanyu; Shu, Huazhong; Coatrieux, Gouenou
2017-07-01
In this paper, we focus on the "blind" identification of the computed tomography (CT) scanner that has produced a CT image. To do so, we propose a set of noise features derived from the image chain acquisition and which can be used as CT-scanner footprint. Basically, we propose two approaches. The first one aims at identifying a CT scanner based on an original sensor pattern noise (OSPN) that is intrinsic to the X-ray detectors. The second one identifies an acquisition system based on the way this noise is modified by its three-dimensional (3-D) image reconstruction algorithm. As these reconstruction algorithms are manufacturer dependent and kept secret, our features are used as input to train a support vector machine (SVM) based classifier to discriminate acquisition systems. Experiments conducted on images issued from 15 different CT-scanner models of 4 distinct manufacturers demonstrate that our system identifies the origin of one CT image with a detection rate of at least 94% and that it achieves better performance than sensor pattern noise (SPN) based strategy proposed for general public camera devices.
Nash evolutionary algorithms : Testing problem size in reconstruction problems in frame structures
Greiner, D.; Periaux, Jacques; Emperador, J.M.; Galván, B.; Winter, G.
2016-01-01
The use of evolutionary algorithms has been enhanced in recent years for solving real engineering problems, where the requirements of intense computational calculations are needed, especially when computational engineering simulations are involved (use of finite element method, boundary element method, etc). The coupling of game-theory concepts in evolutionary algorithms has been a recent line of research which could enhance the efficiency of the optimum design procedure and th...
Maximum likelihood reconstruction in fully 3D PET via the SAGE algorithm
International Nuclear Information System (INIS)
Ollinger, J.M.; Goggin, A.S.
1996-01-01
The SAGE and ordered subsets algorithms have been proposed as fast methods to compute penalized maximum likelihood estimates in PET. We have implemented both for use in fully 3D PET and completed a preliminary evaluation. The technique used to compute the transition matrix is fully described. The evaluation suggests that the ordered subsets algorithm converges much faster than SAGE, but that it stops short of the optimal solution
Directory of Open Access Journals (Sweden)
Zacharias Kamarianakis
2014-07-01
Full Text Available This paper presents the architecture of a software platform implemented in C++, for the purpose of testing and evaluation of reconstruction algorithms in X-ray imaging. The fundamental elements of the platform are classes, tightened together in a logical hierarchy. Real world objects as an X-ray source or a flat detector can be defined and implemented as instances of corresponding classes. Various operations (e.g. 3D transformations, loading, saving, filtering of images, creation of planar or curved objects of various dimensions have been incorporated in the software tool as class methods, as well. The user can easily set up any arrangement of the imaging chain objects in 3D space and experiment with many different trajectories and configurations. Selected 3D volume reconstructions using simulated data acquired in specific scanning trajectories are used as a demonstration of the tool. The platform is considered as a basic tool for future investigations of new reconstruction methods in combination with various scanning configurations.
Atoche, Alejandro Castillo; Castillo, Javier Vázquez
2012-01-01
A high-speed dual super-systolic core for reconstructive signal processing (SP) operations consists of a double parallel systolic array (SA) machine in which each processing element of the array is also conceptualized as another SA in a bit-level fashion. In this study, we addressed the design of a high-speed dual super-systolic array (SSA) core for the enhancement/reconstruction of remote sensing (RS) imaging of radar/synthetic aperture radar (SAR) sensor systems. The selected reconstructive SP algorithms are efficiently transformed in their parallel representation and then, they are mapped into an efficient high performance embedded computing (HPEC) architecture in reconfigurable Xilinx field programmable gate array (FPGA) platforms. As an implementation test case, the proposed approach was aggregated in a HW/SW co-design scheme in order to solve the nonlinear ill-posed inverse problem of nonparametric estimation of the power spatial spectrum pattern (SSP) from a remotely sensed scene. We show how such dual SSA core, drastically reduces the computational load of complex RS regularization techniques achieving the required real-time operational mode. PMID:22736964
Directory of Open Access Journals (Sweden)
Dong Zhang
2014-02-01
Full Text Available This paper presents a new profile shape matching stereovision algorithm that is designed to extract 3D information in real time. This algorithm obtains 3D information by matching profile intensity shapes of each corresponding row of the stereo image pair. It detects the corresponding matching patterns of the intensity profile rather than the intensity values of individual pixels or pixels in a small neighbourhood. This approach reduces the effect of the intensity and colour variations caused by lighting differences. As with all real-time vision algorithms, there is always a trade-off between accuracy and processing speed. This algorithm achieves a balance between the two to produce accurate results for real-time applications. To demonstrate its performance, the proposed algorithm is tested for human pose and hand gesture recognition to control a smart phone and an entertainment system.
Solomon, Justin; Ba, Alexandre; Bochud, François; Samei, Ehsan
2016-12-01
To use novel voxel-based 3D printed textured phantoms in order to compare low-contrast detectability between two reconstruction algorithms, FBP (filtered-backprojection) and SAFIRE (sinogram affirmed iterative reconstruction) and determine what impact background texture (i.e., anatomical noise) has on estimating the dose reduction potential of SAFIRE. Liver volumes were segmented from 23 abdominal CT cases. The volumes were characterized in terms of texture features from gray-level co-occurrence and run-length matrices. Using a 3D clustered lumpy background (CLB) model, a fitting technique based on a genetic optimization algorithm was used to find CLB textures that were reflective of the liver textures, accounting for CT system factors of spatial blurring and noise. With the modeled background texture as a guide, four cylindrical phantoms (Textures A-C and uniform, 165 mm in diameter, and 30 mm height) were designed, each containing 20 low-contrast spherical signals (6 mm diameter at nominal contrast levels of ∼3.2, 5.2, 7.2, 10, and 14 HU with four repeats per signal). The phantoms were voxelized and input into a commercial multimaterial 3D printer (Object Connex 350), with custom software for voxel-based printing (using principles of digital dithering). Images of the textured phantoms and a corresponding uniform phantom were acquired at six radiation dose levels (SOMATOM Flash, Siemens Healthcare) and observer model detection performance (detectability index of a multislice channelized Hotelling observer) was estimated for each condition (5 contrasts × 6 doses × 2 reconstructions × 4 backgrounds = 240 total conditions). A multivariate generalized regression analysis was performed (linear terms, no interactions, random error term, log link function) to assess whether dose, reconstruction algorithm, signal contrast, and background type have statistically significant effects on detectability. Also, fitted curves of detectability (averaged across contrast levels
Barbre, Robert E., Jr.
2012-01-01
This paper presents the process used by the Marshall Space Flight Center Natural Environments Branch (EV44) to quality control (QC) data from the Kennedy Space Center's 50-MHz Doppler Radar Wind Profiler for use in vehicle wind loads and steering commands. The database has been built to mitigate limitations of using the currently archived databases from weather balloons. The DRWP database contains wind measurements from approximately 2.7-18.6 km altitude at roughly five minute intervals for the August 1997 to December 2009 period of record, and the extensive QC process was designed to remove spurious data from various forms of atmospheric and non-atmospheric artifacts. The QC process is largely based on DRWP literature, but two new algorithms have been developed to remove data contaminated by convection and excessive first guess propagations from the Median Filter First Guess Algorithm. In addition to describing the automated and manual QC process in detail, this paper describes the extent of the data retained. Roughly 58% of all possible wind observations exist in the database, with approximately 100 times as many complete profile sets existing relative to the EV44 balloon databases. This increased sample of near-continuous wind profile measurements may help increase launch availability by reducing the uncertainty of wind changes during launch countdown
International Nuclear Information System (INIS)
Liu, Patrick T.; Pavlicek, William P.; Peter, Mary B.; Roberts, Catherine C.; Paden, Robert G.; Spangehl, Mark J.
2009-01-01
Despite recent advances in CT technology, metal orthopedic implants continue to cause significant artifacts on many CT exams, often obscuring diagnostic information. We performed this prospective study to evaluate the effectiveness of an experimental metal artifact reduction (MAR) image reconstruction program for CT. We examined image quality on CT exams performed in patients with hip arthroplasties as well as other types of implanted metal orthopedic devices. The exam raw data were reconstructed using two different methods, the standard filtered backprojection (FBP) program and the MAR program. Images were evaluated for quality of the metal-cement-bone interfaces, trabeculae ≤1 cm from the metal, trabeculae 5 cm apart from the metal, streak artifact, and overall soft tissue detail. The Wilcoxon Rank Sum test was used to compare the image scores from the large and small prostheses. Interobserver agreement was calculated. When all patients were grouped together, the MAR images showed mild to moderate improvement over the FBP images. However, when the cases were divided by implant size, the MAR images consistently received higher image quality scores than the FBP images for large metal implants (total hip prostheses). For small metal implants (screws, plates, staples), conversely, the MAR images received lower image quality scores than the FBP images due to blurring artifact. The difference of image scores for the large and small implants was significant (p=0.002). Interobserver agreement was found to be high for all measures of image quality (k>0.9). The experimental MAR reconstruction algorithm significantly improved CT image quality for patients with large metal implants. However, the MAR algorithm introduced blurring artifact that reduced image quality with small metal implants. (orig.)
Tomše, Petra; Jensterle, Luka; Rep, Sebastijan; Grmek, Marko; Zaletel, Katja; Eidelberg, David; Dhawan, Vijay; Ma, Yilong; Trošt, Maja
2017-09-01
To evaluate the reproducibility of the expression of Parkinson's Disease Related Pattern (PDRP) across multiple sets of 18F-FDG-PET brain images reconstructed with different reconstruction algorithms. 18F-FDG-PET brain imaging was performed in two independent cohorts of Parkinson's disease (PD) patients and normal controls (NC). Slovenian cohort (20 PD patients, 20 NC) was scanned with Siemens Biograph mCT camera and reconstructed using FBP, FBP+TOF, OSEM, OSEM+TOF, OSEM+PSF and OSEM+PSF+TOF. American Cohort (20 PD patients, 7 NC) was scanned with GE Advance camera and reconstructed using 3DRP, FORE-FBP and FORE-Iterative. Expressions of two previously-validated PDRP patterns (PDRP-Slovenia and PDRP-USA) were calculated. We compared the ability of PDRP to discriminate PD patients from NC, differences and correlation between the corresponding subject scores and ROC analysis results across the different reconstruction algorithms. The expression of PDRP-Slovenia and PDRP-USA networks was significantly elevated in PD patients compared to NC (palgorithms. PDRP expression strongly correlated between all studied algorithms and the reference algorithm (r⩾0.993, palgorithms varied within 0.73 and 0.08 of the reference value for PDRP-Slovenia and PDRP-USA, respectively. ROC analysis confirmed high similarity in sensitivity, specificity and AUC among all studied reconstruction algorithms. These results show that the expression of PDRP is reproducible across a variety of reconstruction algorithms of 18F-FDG-PET brain images. PDRP is capable of providing a robust metabolic biomarker of PD for multicenter 18F-FDG-PET images acquired in the context of differential diagnosis or clinical trials. Copyright © 2017 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.
Dong, Jian; Hayakawa, Yoshihiko; Kannenberg, Sven; Kober, Cornelia
2013-02-01
The objective of this study was to reduce metal-induced streak artifact on oral and maxillofacial x-ray computed tomography (CT) images by developing the fast statistical image reconstruction system using iterative reconstruction algorithms. Adjacent CT images often depict similar anatomical structures in thin slices. So, first, images were reconstructed using the same projection data of an artifact-free image. Second, images were processed by the successive iterative restoration method where projection data were generated from reconstructed image in sequence. Besides the maximum likelihood-expectation maximization algorithm, the ordered subset-expectation maximization algorithm (OS-EM) was examined. Also, small region of interest (ROI) setting and reverse processing were applied for improving performance. Both algorithms reduced artifacts instead of slightly decreasing gray levels. The OS-EM and small ROI reduced the processing duration without apparent detriments. Sequential and reverse processing did not show apparent effects. Two alternatives in iterative reconstruction methods were effective for artifact reduction. The OS-EM algorithm and small ROI setting improved the performance. Copyright © 2012 Elsevier Inc. All rights reserved.
Directory of Open Access Journals (Sweden)
Ramon Barber
2014-02-01
Full Text Available Nowadays the automobile industry is becoming more and more demanding as far as quality is concerned. Within the wide variety of processes in which this quality must be ensured, those regarding the squeezing of the auto bodywork are especially important due to the fact that the quality of the resulting product is tested manually by experts, leading to inaccuracies of all types. In this paper, an algorithm is proposed for the automated evaluation of the imperfections in the sheets of the bodywork after the squeezing process. The algorithm processes the profile signals from a retroreflective image and characterizes an imperfection. It is based on a convergence criterion that follows the line of the maximum gradient of the imperfection and gives its geometrical characteristics as a result: maximum gradient, length, width, and area.
Performance of the reconstruction algorithms of the FIRST experiment pixel sensors vertex detector
Rescigno, R; Juliani, D; Spiriti, E; Baudot, J; Abou-Haidar, Z; Agodi, C; Alvarez, M A G; Aumann, T; Battistoni, G; Bocci, A; Böhlen, T T; Boudard, A; Brunetti, A; Carpinelli, M; Cirrone, G A P; Cortes-Giraldo, M A; Cuttone, G; De Napoli, M; Durante, M; Gallardo, M I; Golosio, B; Iarocci, E; Iazzi, F; Ickert, G; Introzzi, R; Krimmer, J; Kurz, N; Labalme, M; Leifels, Y; Le Fevre, A; Leray, S; Marchetto, F; Monaco, V; Morone, M C; Oliva, P; Paoloni, A; Patera, V; Piersanti, L; Pleskac, R; Quesada, J M; Randazzo, N; Romano, F; Rossi, D; Rousseau, M; Sacchi, R; Sala, P; Sarti, A; Scheidenberger, C; Schuy, C; Sciubba, A; Sfienti, C; Simon, H; Sipala, V; Tropea, S; Vanstalle, M; Younis, H
2014-01-01
Hadrontherapy treatments use charged particles (e.g. protons and carbon ions) to treat tumors. During a therapeutic treatment with carbon ions, the beam undergoes nuclear fragmentation processes giving rise to significant yields of secondary charged particles. An accurate prediction of these production rates is necessary to estimate precisely the dose deposited into the tumours and the surrounding healthy tissues. Nowadays, a limited set of double differential carbon fragmentation cross-section is available. Experimental data are necessary to benchmark Monte Carlo simulations for their use in hadrontherapy. The purpose of the FIRST experiment is to study nuclear fragmentation processes of ions with kinetic energy in the range from 100 to 1000 MeV/u. Tracks are reconstructed using information from a pixel silicon detector based on the CMOS technology. The performances achieved using this device for hadrontherapy purpose are discussed. For each reconstruction step (clustering, tracking and vertexing), different...
Trusiak, Maciej; Patorski, Krzysztof; Wielgus, Maciej
2012-10-08
Presented method for fringe pattern enhancement has been designed for processing and analyzing low quality fringe patterns. It uses a modified fast and adaptive bidimensional empirical mode decomposition (FABEMD) for the extraction of bidimensional intrinsic mode functions (BIMFs) from an interferogram. Fringe pattern is then selectively reconstructed (SR) taking the regions of selected BIMFs with high modulation values only. Amplitude demodulation and normalization of the reconstructed image is conducted using the spiral phase Hilbert transform (HS). It has been tested using computer generated interferograms and real data. The performance of the presented SR-FABEMD-HS method is compared with other normalization techniques. Its superiority, potential and robustness to high fringe density variations and the presence of noise, modulation and background illumination defects in analyzed fringe patterns has been corroborated.
Armstrong, Ian S; Hoffmann, Sandra A
2016-11-01
The interest in quantitative single photon emission computer tomography (SPECT) shows potential in a number of clinical applications and now several vendors are providing software and hardware solutions to allow 'SUV-SPECT' to mirror metrics used in PET imaging. This brief technical report assesses the accuracy of activity concentration measurements using a new algorithm 'xSPECT' from Siemens Healthcare. SPECT/CT data were acquired from a uniform cylinder with 5, 10, 15 and 20 s/projection and NEMA image quality phantom with 25 s/projection. The NEMA phantom had hot spheres filled with an 8 : 1 activity concentration relative to the background compartment. Reconstructions were performed using parameters defined by manufacturer presets available with the algorithm. The accuracy of activity concentration measurements was assessed. A dose calibrator-camera cross-calibration factor (CCF) was derived from the uniform phantom data. In uniform phantom images, a positive bias was observed, ranging from ∼6% in the lower count images to ∼4% in the higher-count images. On the basis of the higher-count data, a CCF of 0.96 was derived. As expected, considerable negative bias was measured in the NEMA spheres using region mean values whereas positive bias was measured in the four largest NEMA spheres. Nonmonotonically increasing recovery curves for the hot spheres suggested the presence of Gibbs edge enhancement from resolution modelling. Sufficiently accurate activity concentration measurements can easily be measured on images reconstructed with the xSPECT algorithm without a CCF. However, the use of a CCF is likely to improve accuracy further. A manual conversion of voxel values into SUV should be possible, provided that the patient weight, injected activity and time between injection and imaging are all known accurately.
Pontone, Gianluca; Muscogiuri, Giuseppe; Andreini, Daniele; Guaricci, Andrea I; Guglielmo, Marco; Baggiano, Andrea; Fazzari, Fabio; Mushtaq, Saima; Conte, Edoardo; Annoni, Andrea; Formenti, Alberto; Mancini, Elisabetta; Verdecchia, Massimo; Campari, Alessandro; Martini, Chiara; Gatti, Marco; Fusini, Laura; Bonfanti, Lorenzo; Consiglio, Elisa; Rabbat, Mark G; Bartorelli, Antonio L; Pepi, Mauro
2018-03-27
A new postprocessing algorithm named adaptive statistical iterative reconstruction (ASIR)-V has been recently introduced. The aim of this article was to analyze the impact of ASIR-V algorithm on signal, noise, and image quality of coronary computed tomography angiography. Fifty consecutive patients underwent clinically indicated coronary computed tomography angiography (Revolution CT; GE Healthcare, Milwaukee, WI). Images were reconstructed using filtered back projection and ASIR-V 0%, and a combination of filtered back projection and ASIR-V 20%-80% and ASIR-V 100%. Image noise, signal-to-noise ratio (SNR), and contrast-to-noise ratio (CNR) were calculated for left main coronary artery (LM), left anterior descending artery (LAD), left circumflex artery (LCX), and right coronary artery (RCA) and were compared between the different postprocessing algorithms used. Similarly a four-point Likert image quality score of coronary segments was graded for each dataset and compared. A cutoff value of P ASIR-V 0%, ASIR-V 100% demonstrated a significant reduction of image noise in all coronaries (P ASIR-V 0%, SNR was significantly higher with ASIR-V 60% in LM (P ASIR-V 0%, CNR for ASIR-V ≥60% was significantly improved in LM (P ASIR-V ≥80%. ASIR-V 60% had significantly better Likert image quality scores compared to ASIR-V 0% in segment-, vessel-, and patient-based analyses (P ASIR-V 60% provides the optimal balance between image noise, SNR, CNR, and image quality. Copyright © 2018 The Association of University Radiologists. Published by Elsevier Inc. All rights reserved.
Analytic reconstruction algorithms for triple-source CT with horizontal data truncation
International Nuclear Information System (INIS)
Chen, Ming; Yu, Hengyong
2015-01-01
Purpose: This paper explores a triple-source imaging method with horizontal data truncation to enlarge the field of view (FOV) for big objects. Methods: The study is conducted by using theoretical analysis, mathematical deduction, and numerical simulations. The proposed algorithms are implemented in c + + and MATLAB. While the basic platform is constructed in MATLAB, the computationally intensive segments are coded in c + +, which are linked via a MEX interface. Results: A triple-source circular scanning configuration with horizontal data truncation is developed, where three pairs of x-ray sources and detectors are unevenly distributed on the same circle to cover the whole imaging object. For this triple-source configuration, a fan-beam filtered backprojection-type algorithm is derived for truncated full-scan projections without data rebinning. The algorithm is also extended for horizontally truncated half-scan projections and cone-beam projections in a Feldkamp-type framework. Using their method, the FOV is enlarged twofold to threefold to scan bigger objects with high speed and quality. The numerical simulation results confirm the correctness and effectiveness of the developed algorithms. Conclusions: The triple-source scanning configuration with horizontal data truncation cannot only keep most of the advantages of a traditional multisource system but also cover a larger FOV for big imaging objects. In addition, because the filtering is shift-invariant, the proposed algorithms are very fast and easily parallelized on graphic processing units
Super-resolution reconstruction of MR image with a novel residual learning network algorithm
Shi, Jun; Liu, Qingping; Wang, Chaofeng; Zhang, Qi; Ying, Shihui; Xu, Haoyu
2018-04-01
Spatial resolution is one of the key parameters of magnetic resonance imaging (MRI). The image super-resolution (SR) technique offers an alternative approach to improve the spatial resolution of MRI due to its simplicity. Convolutional neural networks (CNN)-based SR algorithms have achieved state-of-the-art performance, in which the global residual learning (GRL) strategy is now commonly used due to its effectiveness for learning image details for SR. However, the partial loss of image details usually happens in a very deep network due to the degradation problem. In this work, we propose a novel residual learning-based SR algorithm for MRI, which combines both multi-scale GRL and shallow network block-based local residual learning (LRL). The proposed LRL module works effectively in capturing high-frequency details by learning local residuals. One simulated MRI dataset and two real MRI datasets have been used to evaluate our algorithm. The experimental results show that the proposed SR algorithm achieves superior performance to all of the other compared CNN-based SR algorithms in this work.
Performance of the ATLAS track reconstruction algorithms in dense environments in LHC Run 2
Energy Technology Data Exchange (ETDEWEB)
Aaboud, M. [Univ. Mohamed Premier et LPTPM, Oujda (Morocco). Faculte des Sciences; Aad, G. [CPPM, Aix-Marseille Univ. et CNRS/IN2P3, Marseille (France); Abbott, B. [Oklahoma Univ., Norman, OK (United States). Homer L. Dodge Dept. of Physics and Astronomy; Collaboration: ATLAS Collaboration; and others
2017-10-15
With the increase in energy of the Large Hadron Collider to a centre-of-mass energy of 13 TeV for Run 2, events with dense environments, such as in the cores of high-energy jets, became a focus for new physics searches as well as measurements of the Standard Model. These environments are characterized by charged-particle separations of the order of the tracking detectors sensor granularity. Basic track quantities are compared between 3.2 fb{sup -1} of data collected by the ATLAS experiment and simulation of proton-proton collisions producing high-transverse-momentum jets at a centre-of-mass energy of 13 TeV. The impact of charged-particle separations and multiplicities on the track reconstruction performance is discussed. The track reconstruction efficiency in the cores of jets with transverse momenta between 200 and 1600 GeV is quantified using a novel, data-driven, method. The method uses the energy loss, dE/dx, to identify pixel clusters originating from two charged particles. Of the charged particles creating these clusters, the measured fraction that fail to be reconstructed is 0.061 ± 0.006(stat.) ± 0.014(syst.) and 0.093 ± 0.017(stat.) ± 0.021(syst.) for jet transverse momenta of 200-400 GeV and 1400-1600 GeV, respectively. (orig.)
Performance of the ATLAS track reconstruction algorithms in dense environments in LHC Run 2.
Aaboud, M; Aad, G; Abbott, B; Abdallah, J; Abdinov, O; Abeloos, B; Abidi, S H; AbouZeid, O S; Abraham, N L; Abramowicz, H; Abreu, H; Abreu, R; Abulaiti, Y; Acharya, B S; Adachi, S; Adamczyk, L; Adelman, J; Adersberger, M; Adye, T; Affolder, A A; Agatonovic-Jovin, T; Agheorghiesei, C; Aguilar-Saavedra, J A; Ahlen, S P; Ahmadov, F; Aielli, G; Akatsuka, S; Akerstedt, H; Åkesson, T P A; Akimov, A V; Alberghi, G L; Albert, J; Albicocco, P; Alconada Verzini, M J; Aleksa, M; Aleksandrov, I N; Alexa, C; Alexander, G; Alexopoulos, T; Alhroob, M; Ali, B; Aliev, M; Alimonti, G; Alison, J; Alkire, S P; Allbrooke, B M M; Allen, B W; Allport, P P; Aloisio, A; Alonso, A; Alonso, F; Alpigiani, C; Alshehri, A A; Alstaty, M; Alvarez Gonzalez, B; Álvarez Piqueras, D; Alviggi, M G; Amadio, B T; Amaral Coutinho, Y; Amelung, C; Amidei, D; Santos, S P Amor Dos; Amorim, A; Amoroso, S; Amundsen, G; Anastopoulos, C; Ancu, L S; Andari, N; Andeen, T; Anders, C F; Anders, J K; Anderson, K J; Andreazza, A; Andrei, V; Angelidakis, S; Angelozzi, I; Angerami, A; Anisenkov, A V; Anjos, N; Annovi, A; Antel, C; Antonelli, M; Antonov, A; Antrim, D J; Anulli, F; Aoki, M; Aperio Bella, L; Arabidze, G; Arai, Y; Araque, J P; Araujo Ferraz, V; Arce, A T H; Ardell, R E; Arduh, F A; Arguin, J-F; Argyropoulos, S; Arik, M; Armbruster, A J; Armitage, L J; Arnaez, O; Arnold, H; Arratia, M; Arslan, O; Artamonov, A; Artoni, G; Artz, S; Asai, S; Asbah, N; Ashkenazi, A; Asquith, L; Assamagan, K; Astalos, R; Atkinson, M; Atlay, N B; Augsten, K; Avolio, G; Axen, B; Ayoub, M K; Azuelos, G; Baas, A E; Baca, M J; Bachacou, H; Bachas, K; Backes, M; Backhaus, M; Bagnaia, P; Bahrasemani, H; Baines, J T; Bajic, M; Baker, O K; Baldin, E M; Balek, P; Balli, F; Balunas, W K; Banas, E; Banerjee, Sw; Bannoura, A A E; Barak, L; Barberio, E L; Barberis, D; Barbero, M; Barillari, T; Barisits, M-S; Barklow, T; Barlow, N; Barnes, S L; Barnett, B M; Barnett, R M; Barnovska-Blenessy, Z; Baroncelli, A; Barone, G; Barr, A J; Barranco Navarro, L; Barreiro, F; Barreiro Guimarães da Costa, J; Bartoldus, R; Barton, A E; Bartos, P; Basalaev, A; Bassalat, A; Bates, R L; Batista, S J; Batley, J R; Battaglia, M; Bauce, M; Bauer, F; Bawa, H S; Beacham, J B; Beattie, M D; Beau, T; Beauchemin, P H; Bechtle, P; Beck, H P; Becker, K; Becker, M; Beckingham, M; Becot, C; Beddall, A J; Beddall, A; Bednyakov, V A; Bedognetti, M; Bee, C P; Beermann, T A; Begalli, M; Begel, M; Behr, J K; Bell, A S; Bella, G; Bellagamba, L; Bellerive, A; Bellomo, M; Belotskiy, K; Beltramello, O; Belyaev, N L; Benary, O; Benchekroun, D; Bender, M; Bendtz, K; Benekos, N; Benhammou, Y; Benhar Noccioli, E; Benitez, J; Benjamin, D P; Benoit, M; Bensinger, J R; Bentvelsen, S; Beresford, L; Beretta, M; Berge, D; Bergeaas Kuutmann, E; Berger, N; Beringer, J; Berlendis, S; Bernard, N R; Bernardi, G; Bernius, C; Bernlochner, F U; Berry, T; Berta, P; Bertella, C; Bertoli, G; Bertolucci, F; Bertram, I A; Bertsche, C; Bertsche, D; Besjes, G J; Bessidskaia Bylund, O; Bessner, M; Besson, N; Betancourt, C; Bethani, A; Bethke, S; Bevan, A J; Beyer, J; Bianchi, R M; Biebel, O; Biedermann, D; Bielski, R; Biesuz, N V; Biglietti, M; Billoud, T R V; Bilokon, H; Bindi, M; Bingul, A; Bini, C; Biondi, S; Bisanz, T; Bittrich, C; Bjergaard, D M; Black, C W; Black, J E; Black, K M; Blair, R E; Blazek, T; Bloch, I; Blocker, C; Blue, A; Blum, W; Blumenschein, U; Blunier, S; Bobbink, G J; Bobrovnikov, V S; Bocchetta, S S; Bocci, A; Bock, C; Boehler, M; Boerner, D; Bogavac, D; Bogdanchikov, A G; Bohm, C; Boisvert, V; Bokan, P; Bold, T; Boldyrev, A S; Bolz, A E; Bomben, M; Bona, M; Boonekamp, M; Borisov, A; Borissov, G; Bortfeldt, J; Bortoletto, D; Bortolotto, V; Boscherini, D; Bosman, M; Bossio Sola, J D; Boudreau, J; Bouffard, J; Bouhova-Thacker, E V; Boumediene, D; Bourdarios, C; Boutle, S K; Boveia, A; Boyd, J; Boyko, I R; Bracinik, J; Brandt, A; Brandt, G; Brandt, O; Bratzler, U; Brau, B; Brau, J E; Breaden Madden, W D; Brendlinger, K; Brennan, A J; Brenner, L; Brenner, R; Bressler, S; Briglin, D L; Bristow, T M; Britton, D; Britzger, D; Brochu, F M; Brock, I; Brock, R; Brooijmans, G; Brooks, T; Brooks, W K; Brosamer, J; Brost, E; Broughton, J H; de Renstrom, P A Bruckman; Bruncko, D; Bruni, A; Bruni, G; Bruni, L S; Brunt, B H; Bruschi, M; Bruscino, N; Bryant, P; Bryngemark, L; Buanes, T; Buat, Q; Buchholz, P; Buckley, A G; Budagov, I A; Buehrer, F; Bugge, M K; Bulekov, O; Bullock, D; Burch, T J; Burckhart, H; Burdin, S; Burgard, C D; Burger, A M; Burghgrave, B; Burka, K; Burke, S; Burmeister, I; Burr, J T P; Busato, E; Büscher, D; Büscher, V; Bussey, P; Butler, J M; Buttar, C M; Butterworth, J M; Butti, P; Buttinger, W; Buzatu, A; Buzykaev, A R; Cabrera Urbán, S; Caforio, D; Cairo, V M; Cakir, O; Calace, N; Calafiura, P; Calandri, A; Calderini, G; Calfayan, P; Callea, G; Caloba, L P; Calvente Lopez, S; Calvet, D; Calvet, S; Calvet, T P; Camacho Toro, R; Camarda, S; Camarri, P; Cameron, D; Caminal Armadans, R; Camincher, C; Campana, S; Campanelli, M; Camplani, A; Campoverde, A; Canale, V; Cano Bret, M; Cantero, J; Cao, T; Capeans Garrido, M D M; Caprini, I; Caprini, M; Capua, M; Carbone, R M; Cardarelli, R; Cardillo, F; Carli, I; Carli, T; Carlino, G; Carlson, B T; Carminati, L; Carney, R M D; Caron, S; Carquin, E; Carrá, S; Carrillo-Montoya, G D; Carvalho, J; Casadei, D; Casado, M P; Casolino, M; Casper, D W; Castelijn, R; Castillo Gimenez, V; Castro, N F; Catinaccio, A; Catmore, J R; Cattai, A; Caudron, J; Cavaliere, V; Cavallaro, E; Cavalli, D; Cavalli-Sforza, M; Cavasinni, V; Celebi, E; Ceradini, F; Cerda Alberich, L; Cerqueira, A S; Cerri, A; Cerrito, L; Cerutti, F; Cervelli, A; Cetin, S A; Chafaq, A; Chakraborty, D; Chan, S K; Chan, W S; Chan, Y L; Chang, P; Chapman, J D; Charlton, D G; Chau, C C; Chavez Barajas, C A; Che, S; Cheatham, S; Chegwidden, A; Chekanov, S; Chekulaev, S V; Chelkov, G A; Chelstowska, M A; Chen, C; Chen, H; Chen, S; Chen, S; Chen, X; Chen, Y; Cheng, H C; Cheng, H J; Cheplakov, A; Cheremushkina, E; Cherkaoui El Moursli, R; Chernyatin, V; Cheu, E; Chevalier, L; Chiarella, V; Chiarelli, G; Chiodini, G; Chisholm, A S; Chitan, A; Chiu, Y H; Chizhov, M V; Choi, K; Chomont, A R; Chouridou, S; Christodoulou, V; Chromek-Burckhart, D; Chu, M C; Chudoba, J; Chuinard, A J; Chwastowski, J J; Chytka, L; Ciftci, A K; Cinca, D; Cindro, V; Cioara, I A; Ciocca, C; Ciocio, A; Cirotto, F; Citron, Z H; Citterio, M; Ciubancan, M; Clark, A; Clark, B L; Clark, M R; Clark, P J; Clarke, R N; Clement, C; Coadou, Y; Cobal, M; Coccaro, A; Cochran, J; Colasurdo, L; Cole, B; Colijn, A P; Collot, J; Colombo, T; Conde Muiño, P; Coniavitis, E; Connell, S H; Connelly, I A; Constantinescu, S; Conti, G; Conventi, F; Cooke, M; Cooper-Sarkar, A M; Cormier, F; Cormier, K J R; Corradi, M; Corriveau, F; Cortes-Gonzalez, A; Cortiana, G; Costa, G; Costa, M J; Costanzo, D; Cottin, G; Cowan, G; Cox, B E; Cranmer, K; Crawley, S J; Creager, R A; Cree, G; Crépé-Renaudin, S; Crescioli, F; Cribbs, W A; Cristinziani, M; Croft, V; Crosetti, G; Cueto, A; Cuhadar Donszelmann, T; Cukierman, A R; Cummings, J; Curatolo, M; Cúth, J; Czirr, H; Czodrowski, P; D'amen, G; D'Auria, S; D'eramo, L; D'Onofrio, M; Da Cunha Sargedas De Sousa, M J; Via, C Da; Dabrowski, W; Dado, T; Dai, T; Dale, O; Dallaire, F; Dallapiccola, C; Dam, M; Dandoy, J R; Daneri, M F; Dang, N P; Daniells, A C; Dann, N S; Danninger, M; Hoffmann, M Dano; Dao, V; Darbo, G; Darmora, S; Dassoulas, J; Dattagupta, A; Daubney, T; Davey, W; David, C; Davidek, T; Davies, M; Davis, D R; Davison, P; Dawe, E; Dawson, I; De, K; de Asmundis, R; De Benedetti, A; De Castro, S; De Cecco, S; De Groot, N; de Jong, P; De la Torre, H; De Lorenzi, F; De Maria, A; De Pedis, D; De Salvo, A; De Sanctis, U; De Santo, A; De Vasconcelos Corga, K; De Vivie De Regie, J B; Dearnaley, W J; Debbe, R; Debenedetti, C; Dedovich, D V; Dehghanian, N; Deigaard, I; Del Gaudio, M; Del Peso, J; Prete, T Del; Delgove, D; Deliot, F; Delitzsch, C M; Dell'Acqua, A; Dell'Asta, L; Dell'Orso, M; Della Pietra, M; Della Volpe, D; Delmastro, M; Delporte, C; Delsart, P A; DeMarco, D A; Demers, S; Demichev, M; Demilly, A; Denisov, S P; Denysiuk, D; Derendarz, D; Derkaoui, J E; Derue, F; Dervan, P; Desch, K; Deterre, C; Dette, K; Devesa, M R; Deviveiros, P O; Dewhurst, A; Dhaliwal, S; Di Bello, F A; Di Ciaccio, A; Di Ciaccio, L; Di Clemente, W K; Di Donato, C; Di Girolamo, A; Di Girolamo, B; Di Micco, B; Di Nardo, R; Di Petrillo, K F; Di Simone, A; Di Sipio, R; Di Valentino, D; Diaconu, C; Diamond, M; Dias, F A; Diaz, M A; Diehl, E B; Dietrich, J; Díez Cornell, S; Dimitrievska, A; Dingfelder, J; Dita, P; Dita, S; Dittus, F; Djama, F; Djobava, T; Djuvsland, J I; do Vale, M A B; Dobos, D; Dobre, M; Doglioni, C; Dolejsi, J; Dolezal, Z; Donadelli, M; Donati, S; Dondero, P; Donini, J; Dopke, J; Doria, A; Dova, M T; Doyle, A T; Drechsler, E; Dris, M; Du, Y; Duarte-Campderros, J; Dubreuil, A; Duchovni, E; Duckeck, G; Ducourthial, A; Ducu, O A; Duda, D; Dudarev, A; Dudder, A Chr; Duffield, E M; Duflot, L; Dührssen, M; Dumancic, M; Dumitriu, A E; Duncan, A K; Dunford, M; Duran Yildiz, H; Düren, M; Durglishvili, A; Duschinger, D; Dutta, B; Dyndal, M; Eckardt, C; Ecker, K M; Edgar, R C; Eifert, T; Eigen, G; Einsweiler, K; Ekelof, T; Kacimi, M El; Kosseifi, R El; Ellajosyula, V; Ellert, M; Elles, S; Ellinghaus, F; Elliot, A A; Ellis, N; Elmsheuser, J; Elsing, M; Emeliyanov, D; Enari, Y; Endner, O C; Ennis, J S; Erdmann, J; Ereditato, A; Ernis, G; Ernst, M; Errede, S; Escalier, M; Escobar, C; Esposito, B; Estrada Pastor, O; Etienvre, A I; Etzion, E; Evans, H; Ezhilov, A; Ezzi, M; Fabbri, F; Fabbri, L; Facini, G; Fakhrutdinov, R M; Falciano, S; Falla, R J; Faltova, J; Fang, Y; Fanti, M; Farbin, A; Farilla, A; Farina, C; Farina, E M; Farooque, T; Farrell, S; Farrington, S M; Farthouat, P; Fassi, F; Fassnacht, P; Fass