International Nuclear Information System (INIS)
Oda, Naotaka.
1993-01-01
The device of the present invention greatly saves an analog processing section such as an analog filter and an analog processing circuit. That is, the device of the present invention comprises (1) a neutron flux detection means for detecting neutron fluxed in the reactor, (2) a digital filter means for dividing signals corresponding to the detected neutron fluxes into predetermined frequency band regions, (3) a calculation processing means for applying a calculation processing corresponding to the frequency band regions to the neutron flux detection signals divided by the digital filter means. With such a constitution, since the neutron detection signals are processed by the digital filter means, the accuracy is improved and the change for the property of the filter is facilitated. Further, when a neutron flux level is obtained, a calculation processing corresponding to the frequency band region can be conducted without the analog processing circuit. Accordingly, maintenance and accuracy are improved by greatly decreasing the number of parts. Further, since problems inherent to the analog circuit are solved, neutron fluxes are monitored at high reliability. (I.S.)
International Nuclear Information System (INIS)
Madhavi, V.; Phatak, P.R.; Bahadur, C.; Bayala, A.K.; Jakati, R.K.; Sathian, V.
2003-01-01
Full text: A compact size neutron flux monitor has been developed incorporating standard boards developed for smart radiation monitors. The sensitivity of the monitors is 0.4cps/nV. It has been tested up to 2075 nV flux with standard neutron sources. It shows convincing results even in high flux areas like 6m away from the accelerator in RMC (Parel) for 106/107 nV. These monitors have a focal and remote display, alarm function with potential free contacts for centralized control and additional provision of connectivity via RS485/Ethernet. This paper describes the construction, working and results of the above flux monitor
Neutron flux monitoring device
International Nuclear Information System (INIS)
Shimazu, Yoichiro.
1995-01-01
In a neutron flux monitoring device, there are disposed a neutron flux measuring means for outputting signals in accordance with the intensity of neutron fluxes, a calculation means for calculating a self power density spectrum at a frequency band suitable to an object to be measured based on the output of the neutron flux measuring means, an alarm set value generation means for outputting an alarm set value as a comparative reference, and an alarm judging means for comparing the alarm set value with the outputted value of the calculation means to judge requirement of generating an alarm and generate an alarm in accordance with the result of the judgement. Namely, the time-series of neutron flux signals is put to fourier transformation for a predetermined period of time by the calculation means, and from each of square sums for real number component and imaginary number component for each of the frequencies, a self power density spectrum in the frequency band suitable to the object to be measured is calculated. Then, when the set reference value is exceeded, an alarm is generated. This can reliably prevent generation of erroneous alarm due to neutron flux noises and can accurately generate an alarm at an appropriate time. (N.H.)
Neutron flux monitoring device
International Nuclear Information System (INIS)
Goto, Yasushi; Mitsubori, Minehisa; Ohashi, Kazunori.
1997-01-01
The present invention provides a neutron flux monitoring device for preventing occurrence of erroneous reactor scram caused by the elevation of the indication of a start region monitor (SRM) due to a factor different from actual increase of neutron fluxes. Namely, judgement based on measured values obtained by a pulse counting method and a judgment based on measured values obtained by a Cambel method are combined. A logic of switching neutron flux measuring method to be used for monitoring, namely, switching to an intermediate region when both of the judgements are valid is adopted. Then, even if the indication value is elevated based on the Cambel method with no increase of the counter rate in a neutron source region, the switching to the intermediate region is not conducted. As a result, erroneous reactor scram such as 'shorter reactor period' can be avoided. (I.S.)
International Nuclear Information System (INIS)
Sahni, D.C.; Sharma, A.
2000-01-01
The integral form of one-speed, spherically symmetric neutron transport equation with isotropic scattering is considered. Two standard problems are solved using normal mode expansion technique. The expansion coefficients are obtained by solving their singular integral equations. It is shown that these expansion coefficients provide a representation of all spherical harmonics moments of the angular flux as a superposition of Bessel functions. It is seen that large errors occur in the computation of higher moments unless we take certain precautions. The reasons for this phenomenon are explained. They throw some light on the failure of spherical harmonics method in treating spherical geometry problems as observed by Aronsson
International Nuclear Information System (INIS)
Seki, Eiji; Tai, Ichiro.
1984-01-01
Purpose: To maintain the measuring accuracy and the reponse time within an allowable range in accordance with the change of neutron fluxes in a nuclear reactor pressure vessel. Constitution: Neutron fluxes within a nuclear reactor pressure vessel are detected by detectors, converted into pulse signals and amplified in a range switching amplifier. The amplified signals are further converted through an A/D converter and digital signals from the converter are subjected to a square operation in an square operation circuit. The output from the circuit is inputted into an integration circuit to selectively accumulate the constant of 1/2n, 1 - 1/2n (n is a positive integer) respectively for two continuing signals to perform weighing. Then, the addition is carried out to calculate the integrated value and the addition number is changed by the chane in the number n to vary the integrating time. The integrated value is inputted into a control circuit to control the value of n so that the fluctuation and the calculation time for the integrated value are within a predetermined range and, at the same time, the gain of the range switching amplifier is controlled. (Seki, T.)
Notes on neutron flux measurement
International Nuclear Information System (INIS)
Alcala Ruiz, F.
1984-01-01
The main purpose of this work is to get an useful guide to carry out topical neutron flux measurements. Although the foil activation technique is used in the majority of the cases, other techniques, such as those based on fission chambers and self-powered neutron detectors, are also shown. Special interest is given to the description and application of corrections on the measurement of relative and absolute induced activities by several types of detectors (scintillators, G-M and gas proportional counters). The thermal arid epithermal neutron fluxes, as determined in this work, are conventional or effective (West cots fluxes), which are extensively used by the reactor experimentalists; however, we also give some expressions where they are related to the integrated neutron fluxes, which are used in neutron calculations. (Author) 16 refs
Neutron flux stabilization in the NG-150 neutron generators
International Nuclear Information System (INIS)
Kuz'min, L.E.; Makarov, S.A.; Pronman, I.M.
1986-01-01
Problem of metal tritium target lifetime increase and neutron flux stabilization in the NG-150 neutron generators is studied. Possibility on neutron flux stabilization using the mass analyzer for low-angle (4 deg and 41 deg) mass separation of a beam in thre components, which fall on a target simultaneously, is confirmed experimentally. Basic generator parameters are: accelerating voltage of 150 kV, total beam current on a target of 1.5 mA, beam current density of 0.3-1.6 mA/cm 2 , beam diameter of 8 mm. The initial neutron flux on the targets of 0.73 mg/cm 2 thick constituted 1.1x10 11 ssup(-1). The neutron flux monitoring was accomplished from recoil proton recording by a plastic scintillator. Flux decrease by more than 5% served as a signel for measuring mass analyzer magnetic field providing beam displacement on a target and restoration of the given flux. The NG-150 generator neutron flux stabilization was attained during 2h
CARNAC, Neutron Flux and Neutron Spectra in Criticality Accident
International Nuclear Information System (INIS)
Bessis, J.
1976-01-01
Nature of physical problem solved: Calculation of flux and neutron spectra in the case of a criticality accident. The method is unsophisticated but fast. The program is divided into two parts: (1) The code CRITIC is based on the Fermi age equation and evaluates the neutron number per fission emitted from a moderate critical system and its energy spectrum. (2) The code NARCISSE uses concrete current albedo, evaluates the product of neutron reflection on walls of the source containment and calculates the resulting flux at any point, and its energy distribution into 21 groups. The results obtained seem satisfactory, if compared with a Monte Carlo program
Wide range neutron flux monitor
International Nuclear Information System (INIS)
Endo, Yorimasa; Fukushima, Toshiki.
1983-01-01
Purpose: To provide a wide range neutron-flux monitor adapted such that the flux monitoring function and alarming function can automatically by shifted from pulse counting system to cambel method system. Constitution: A wide range neutron-flux monitor comprises (la) pulse counting system and (lb) cambel-method system for inputting detection signals from neutron detectors and separating them into signals for the pulse measuring system and the cambel measuring system, (2) overlap detection and calculation circuit for detecting the existence of the overlap of two output signals from the (la) and (lb) systems, and (3) trip circuit for judging the abnormal state of neutron detectors upon input of the detection signals. (Seki, T.)
The secondary neutron sources for generation of particular neutron fluxes
International Nuclear Information System (INIS)
Tracz, G.
2007-07-01
The foregoing paper presents the doctor's thesis entitled '' The secondary neutron sources for generation of particular neutron fluxes ''. Two secondary neutron sources have been designed, which exploit already existing primary sources emitting neutrons of energies different from the desired ones. The first source is devoted to boron-neutron capture therapy (BNCT). The research reactor MARIA at the Institute of Atomic Energy in Swierk (Poland) is the primary source of the reactor thermal neutrons, while the secondary source should supply epithermal neutrons. The other secondary source is the pulsed source of thermal neutrons that uses fast 14 MeV neutrons from a pulsed generator at the Institute of Nuclear Physics PAN in Krakow (Poland). The physical problems to be solved in the two mentioned cases are different. Namely, in order to devise the BNCT source the initial energy of particles ought to be increased, whilst in the other case the fast neutrons have to be moderated. Slowing down of neutrons is relatively easy since these particles lose energy when they scatter in media; the most effective moderators are the materials which contain light elements (mostly hydrogen). In order to increase the energy of neutrons from thermal to epithermal (the BNCT case) the so-called neutron converter should be exploited. It contains a fissile material, 235 U. The thermal neutrons from the reactor cause fission of uranium and fast neutrons are emitted from the converter. Then fissile neutrons of energy of a few MeV are slowed down to the required epithermal energy range. The design of both secondary sources have been conducted by means of Monte Carlo simulations, which have been carried out using the MCNP code. In the case of the secondary pulsed thermal neutron source, some of the calculated results have been verified experimentally. (author)
maximum neutron flux at thermal nuclear reactors
International Nuclear Information System (INIS)
Strugar, P.
1968-10-01
Since actual research reactors are technically complicated and expensive facilities it is important to achieve savings by appropriate reactor lattice configurations. There is a number of papers, and practical examples of reactors with central reflector, dealing with spatial distribution of fuel elements which would result in higher neutron flux. Common disadvantage of all the solutions is that the choice of best solution is done starting from the anticipated spatial distributions of fuel elements. The weakness of these approaches is lack of defined optimization criteria. Direct approach is defined as follows: determine the spatial distribution of fuel concentration starting from the condition of maximum neutron flux by fulfilling the thermal constraints. Thus the problem of determining the maximum neutron flux is solving a variational problem which is beyond the possibilities of classical variational calculation. This variational problem has been successfully solved by applying the maximum principle of Pontrjagin. Optimum distribution of fuel concentration was obtained in explicit analytical form. Thus, spatial distribution of the neutron flux and critical dimensions of quite complex reactor system are calculated in a relatively simple way. In addition to the fact that the results are innovative this approach is interesting because of the optimization procedure itself [sr
Energy Technology Data Exchange (ETDEWEB)
Ceolin, Celina; Schramm, Marcelo; Bodmann, Bardo Ernst Josef; Vilhena, Marco Tullio Mena Barreto de [Universidade Federal do Rio Grande do Sul, Porto Alegre (Brazil). Programa de Pos-Graduacao em Engenharia Mecanica; Bogado Leite, Sergio de Queiroz [Comissao Nacional de Energia Nuclear, Rio de Janeiro (Brazil)
2014-11-15
In this work the authors solved the steady state neutron diffusion equation for a multi-layer slab assuming the multi-group energy model. The method to solve the equation system is based on an expansion in Taylor Series resulting in an analytical expression. The results obtained can be used as initial condition for neutron space kinetics problems. The neutron scalar flux was expanded in a power series, and the coefficients were found by using the ordinary differential equation and the boundary and interface conditions. The effective multiplication factor k was evaluated using the power method. We divided the domain into several slabs to guarantee the convergence with a low truncation order. We present the formalism together with some numerical simulations.
Neutron flux enhancement at LASREF
International Nuclear Information System (INIS)
Sommer, W.F.; Ferguson, P.D.; Wechsler, M.S.
1992-01-01
The accelerator at the Los Alamos Meson Physiscs Facility produces a 1 mA beam of protons at an energy of 800 MeV. Since 1985, the Los Alamos Spallation Radiation Effects Facility (LASREF) has made use of the neutron flux that is generated as the incident protons interact with the targets and a copper beam stop. A variety of basic and applied experiments in radiation damage and radiation effects have been completed. Recent studies indicate that the flux at LASREF can be increased by at least a factor of 10 from the present level of about 5 E + 17 m -2 s -1 . This requires changing the beam stop material from Cu to W and optimizing the geometry of the beam-target interaction region. These studies are motivated by the need for a large volume, high energy, and high intensity neutron source in the development of materials for advanced energy concepts such as fusion reactors. (orig.)
Neutron flux control systems validation
International Nuclear Information System (INIS)
Hascik, R.
2003-01-01
In nuclear installations main requirement is to obtain corresponding nuclear safety in all operation conditions. From the nuclear safety point of view is commissioning and start-up after reactor refuelling appropriate period for safety systems verification. In this paper, methodology, performance and results of neutron flux measurements systems validation is presented. Standard neutron flux measuring chains incorporated into the reactor protection and control system are used. Standard neutron flux measuring chain contains detector, preamplifier, wiring to data acquisition unit, data acquisition unit, wiring to display at control room and display at control room. During reactor outage only data acquisition unit and wiring and displaying at reactor control room is verified. It is impossible to verify detector, preamplifier and wiring to data acquisition recording unit during reactor refuelling according to low power. Adjustment and accurate functionality of these chains is confirmed by start-up rate (SUR) measurement during start-up tests after refuelling of the reactors. This measurement has direct impact to nuclear safety and increase operational nuclear safety level. Briefly description of each measuring system is given. Results are illustrated on measurements performed at Bohunice NPP during reactor start-up tests. Main failures and their elimination are described (Authors)
Neutron flux enhancement at LASREF
International Nuclear Information System (INIS)
Sommer, W.F.; Ferguson, P.D.; Wechsler, M.S.
1991-01-01
The accelerator at the Los Alamos Meson Physics Facility produces a 1-mA beam of protons at an energy of 800 MeV. Since 1985, the Los Alamos Spallation Radiation Effects Facility (LASREF) has made use of the neutron flux that is generated as the incident protons interact with the nuclei in targets and a copper beam stop. A variety of basic and applied experiments in radiation damage and radiation effects have been completed. Recent studies indicate that the flux at LASREF can be increased by at least a factor of ten from the present level of about 5 E+17 m -2 s -1 . This requires changing the beam-stop material from Cu to W and optimizing the geometry of the beam-target interaction region. These studies are motivated by the need for a large volume, high energy, and high intensity neutron source in the development of materials for advanced energy concepts such as fusion reactors. 18 refs., 7 figs., 2 tabs
Neutron flux enhancement at LASREF
Energy Technology Data Exchange (ETDEWEB)
Sommer, W.F. (Los Alamos National Lab., Los Alamos, NM (United States)); Ferguson, P.D. (Univ. of Missouri, Rolla, MO (United States)); Wechsler, M.S. (Iowa State Univ., Ames, IA (United States))
1992-09-01
The accelerator at the Los Alamos Meson Physiscs Facility produces a 1 mA beam of protons at an energy of 800 MeV. Since 1985, the Los Alamos Spallation Radiation Effects Facility (LASREF) has made use of the neutron flux that is generated as the incident protons interact with the targets and a copper beam stop. A variety of basic and applied experiments in radiation damage and radiation effects have been completed. Recent studies indicate that the flux at LASREF can be increased by at least a factor of 10 from the present level of about 5 E + 17 m[sup -2] s[sup -1]. This requires changing the beam stop material from Cu to W and optimizing the geometry of the beam-target interaction region. These studies are motivated by the need for a large volume, high energy, and high intensity neutron source in the development of materials for advanced energy concepts such as fusion reactors. (orig.).
Maximum neutron flux in thermal reactors
International Nuclear Information System (INIS)
Strugar, P.V.
1968-12-01
Direct approach to the problem is to calculate spatial distribution of fuel concentration if the reactor core directly using the condition of maximum neutron flux and comply with thermal limitations. This paper proved that the problem can be solved by applying the variational calculus, i.e. by using the maximum principle of Pontryagin. Mathematical model of reactor core is based on the two-group neutron diffusion theory with some simplifications which make it appropriate from maximum principle point of view. Here applied theory of maximum principle are suitable for application. The solution of optimum distribution of fuel concentration in the reactor core is obtained in explicit analytical form. The reactor critical dimensions are roots of a system of nonlinear equations and verification of optimum conditions can be done only for specific examples
Linker, J. A.; Caplan, R. M.; Downs, C.; Riley, P.; Mikic, Z.; Lionello, R.; Henney, C. J.; Arge, C. N.; Liu, Y.; Derosa, M. L.; Yeates, A.; Owens, M. J.
2017-10-01
The heliospheric magnetic field is of pivotal importance in solar and space physics. The field is rooted in the Sun’s photosphere, where it has been observed for many years. Global maps of the solar magnetic field based on full-disk magnetograms are commonly used as boundary conditions for coronal and solar wind models. Two primary observational constraints on the models are (1) the open field regions in the model should approximately correspond to coronal holes (CHs) observed in emission and (2) the magnitude of the open magnetic flux in the model should match that inferred from in situ spacecraft measurements. In this study, we calculate both magnetohydrodynamic and potential field source surface solutions using 14 different magnetic maps produced from five different types of observatory magnetograms, for the time period surrounding 2010 July. We have found that for all of the model/map combinations, models that have CH areas close to observations underestimate the interplanetary magnetic flux, or, conversely, for models to match the interplanetary flux, the modeled open field regions are larger than CHs observed in EUV emission. In an alternative approach, we estimate the open magnetic flux entirely from solar observations by combining automatically detected CHs for Carrington rotation 2098 with observatory synoptic magnetic maps. This approach also underestimates the interplanetary magnetic flux. Our results imply that either typical observatory maps underestimate the Sun’s magnetic flux, or a significant portion of the open magnetic flux is not rooted in regions that are obviously dark in EUV and X-ray emission.
Energy Technology Data Exchange (ETDEWEB)
Linker, J. A.; Caplan, R. M.; Downs, C.; Riley, P.; Mikic, Z.; Lionello, R. [Predictive Science Inc., 9990 Mesa Rim Road, Suite 170, San Diego, CA 92121 (United States); Henney, C. J. [Air Force Research Lab/Space Vehicles Directorate, 3550 Aberdeen Avenue SE, Kirtland AFB, NM (United States); Arge, C. N. [Science and Exploration Directorate, NASA/GSFC, Greenbelt, MD 20771 (United States); Liu, Y. [W. W. Hansen Experimental Physics Laboratory, Stanford University, Stanford, CA 94305 (United States); Derosa, M. L. [Lockheed Martin Solar and Astrophysics Laboratory, 3251 Hanover Street B/252, Palo Alto, CA 94304 (United States); Yeates, A. [Department of Mathematical Sciences, Durham University, Durham, DH1 3LE (United Kingdom); Owens, M. J., E-mail: linkerj@predsci.com [Space and Atmospheric Electricity Group, Department of Meteorology, University of Reading, Earley Gate, P.O. Box 243, Reading RG6 6BB (United Kingdom)
2017-10-10
The heliospheric magnetic field is of pivotal importance in solar and space physics. The field is rooted in the Sun’s photosphere, where it has been observed for many years. Global maps of the solar magnetic field based on full-disk magnetograms are commonly used as boundary conditions for coronal and solar wind models. Two primary observational constraints on the models are (1) the open field regions in the model should approximately correspond to coronal holes (CHs) observed in emission and (2) the magnitude of the open magnetic flux in the model should match that inferred from in situ spacecraft measurements. In this study, we calculate both magnetohydrodynamic and potential field source surface solutions using 14 different magnetic maps produced from five different types of observatory magnetograms, for the time period surrounding 2010 July. We have found that for all of the model/map combinations, models that have CH areas close to observations underestimate the interplanetary magnetic flux, or, conversely, for models to match the interplanetary flux, the modeled open field regions are larger than CHs observed in EUV emission. In an alternative approach, we estimate the open magnetic flux entirely from solar observations by combining automatically detected CHs for Carrington rotation 2098 with observatory synoptic magnetic maps. This approach also underestimates the interplanetary magnetic flux. Our results imply that either typical observatory maps underestimate the Sun’s magnetic flux, or a significant portion of the open magnetic flux is not rooted in regions that are obviously dark in EUV and X-ray emission.
EL-2 reactor: Thermal neutron flux distribution
International Nuclear Information System (INIS)
Rousseau, A.; Genthon, J.P.
1958-01-01
The flux distribution of thermal neutrons in EL-2 reactor is studied. The reactor core and lattices are described as well as the experimental reactor facilities, in particular, the experimental channels and special facilities. The measurement shows that the thermal neutron flux increases in the central channel when enriched uranium is used in place of natural uranium. However the thermal neutron flux is not perturbed in the other reactor channels by the fuel modification. The macroscopic flux distribution is measured according the radial positioning of fuel rods. The longitudinal neutron flux distribution in a fuel rod is also measured and shows no difference between enriched and natural uranium fuel rods. In addition, measurements of the flux distribution have been effectuated for rods containing other material as steel or aluminium. The neutron flux distribution is also studied in all the experimental channels as well as in the thermal column. The determination of the distribution of the thermal neutron flux in all experimental facilities, the thermal column and the fuel channels has been made with a heavy water level of 1825 mm and is given for an operating power of 1000 kW. (M.P.)
Neutron flux measurement utilizing Campbell technique
International Nuclear Information System (INIS)
Kropik, M.
2000-01-01
Application of the Campbell technique for the neutron flux measurement is described in the contribution. This technique utilizes the AC component (noise) of a neutron chamber signal rather than a usually used DC component. The Campbell theorem, originally discovered to describe noise behaviour of valves, explains that the root mean square of the AC component of the chamber signal is proportional to the neutron flux (reactor power). The quadratic dependence of the reactor power on the root mean square value usually permits to accomplish the whole current power range of the neutron flux measurement by only one channel. Further advantage of the Campbell technique is that large pulses of the response to neutrons are favoured over small pulses of the response to gamma rays in the ratio of their mean square charge transfer and thus, the Campbell technique provides an excellent gamma rays discrimination in the current operational range of a neutron chamber. The neutron flux measurement channel using state of the art components was designed and put into operation. Its linearity, accuracy, dynamic range, time response and gamma discrimination were tested on the VR-1 nuclear reactor in Prague, and behaviour under high neutron flux (accident conditions) was tested on the TRIGA nuclear reactor in Vienna. (author)
Neutron flux measurements in PUSPATI Triga Reactor
International Nuclear Information System (INIS)
Gui Ah Auu; Mohamad Amin Sharifuldin Salleh; Mohamad Ali Sufi.
1983-01-01
Neutron flux measurement in the PUSPATI TRIGA Reactor (PTR) was initiated after its commissioning on 28 June 1982. Initial measured thermal neutron flux at the bottom of the rotary specimen rack (rotating) and in-core pneumatic terminus were 3.81E+11 n/cm 2 sec and 1.10E+12n/cm 2 sec respectively at 100KW. Work to complete the neutron flux data are still going on. The cadmium ratio, thermal and epithermal neutron flux are measured in the reactor core, rotary specimen rack, in-core pneumatic terminus and thermal column. Bare and Cadmium covered gold foils and wires are used for the above measurement. The activities of the irradiated gold foils and wires are determined using Ge(Li) and hyperpure germinium detectors. (author)
Self-powered neutron flux detector
International Nuclear Information System (INIS)
Kroon, J.
1979-01-01
A self-powered neutron flux detector having an emitter electrode, at least a major portion of which is, 95 Mo encased in a tubular collector electrode and separated therefrom by dielectric material. The 95 Mo emitter electrode has experimentally shown a 98% prompt response, is primarily sensitive to neutron flux, has adequate sensitivity and has low burn up. Preferably the emitter electrode is molybdenum which has been enriched 75% to 99% by weight with 95 Mo
Neutron flux distribution forecasting device of reactor
International Nuclear Information System (INIS)
Uematsu, Hitoshi
1991-01-01
A neutron flux distribution is forecast by using current data obtained from a reactor. That is, the device of the present invention comprises (1) a neutron flux monitor disposed in various positions in the reactor, (2) a forecasting means for calculating and forecasting a one-dimensional neutron flux distribution relative to imaginable events by using data obtained from the neutron flux monitor and physical models, and (3) a display means for displaying the results forecast in the forecasting means to a reactor operation console. Since the forecast values for the one-dimensional neutron flux distribution relative to the imaginable events are calculated in the device of the present invention by using data obtained from the neutron flux monitor and the physical models, the data as a base of the calculation are new and the period for calculating the forecast values can be shortened. Accordingly, although there is a worry of providing some errors in the forecast values, they can be utilized sufficiently as reference data. As a result, the reactor can be operated more appropriately. (I.N.)
A theoretical model for predicting neutron fluxes for cyclic Neutron ...
African Journals Online (AJOL)
A theoretical model has been developed for prediction of thermal neutron fluxes required for cyclic irradiations of a sample to obtain the same activity previously used for the detection of any radionuclide of interest. The model is suitable for radiotracer production or for long-lived neutron activation products where the ...
Neutron flux measurement by mobile detectors
International Nuclear Information System (INIS)
Verchain, M.
1987-01-01
Various incore instrumentation systems and their technological evolution are first reviewed. Then, for 1300 MWe PWR nuclear power plant, temperature and neutron flux measurement are described. Mobile fission chambers, with their large measuring range and accurate location allow a good knowledge of the core. Other incore measures are possible because of flux detector thimble tubes inserted in the reactor core [fr
CFRMF neutron flux gradient and spectral determinations
International Nuclear Information System (INIS)
Rogers, J.W.; Turk, E.H.; Hogg, C.H.
1976-01-01
Recently more accurate and complete measurements of the flux gradient have been measured by the activation of 235 U and Au samples. Neutron spectrum characteristics were studied by making activation measurements with and without the ends of the CFRMF test region plugged with 10 B. These measurements define the flux gradient to +-1 to 2% and indicate there is no detectable streaming of thermal or resonance neutrons from the ends in the central 30 cm of the CFRMF test region. Measurements of the Cd ratio of Au foil activations were conducted and these results also indicate there is no streaming of thermal and resonance neutrons into the CFRMF test region
Energy Technology Data Exchange (ETDEWEB)
Le Meur, R; Lecomte, P [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1968-07-01
The operation of neutron-electron converters designed for measuring thermal neutron fluxes is examined. The principle is to produce short lived isotopes emitting beta particles, by activation, and to measure their activity not by extracting them from the reactor, but directly in the reactor using the emitted electrons to deflect the needle of a galvanometer placed outside the flux. After a theoretical study, the results of the measurements are presented; particular attention is paid to a new type of converter characterized by a layer structure. The converters are very useful for obtaining flux distributions with more than 10{sup 7} neutrons cm{sup -2}*sec{sup -1}. They work satisfactorily in pressurized carbon dioxide at 400 Celsius degrees. Some points still have to be cleared up however concerning interfering currents in the detectors and the behaviour of the dielectrics under irradiation. (authors) [French] On examine le fonctionnement de convertisseurs neutrons electrons destines a des mesures de flux de neutrons thermiques. Le principe est de former par activation des isotopes a periodes courtes et a emission beta et de mesurer leur activite non pas en les sortant du reacteur, mais directement en pile, utilisant les electrons emis pour faire devier l'aiguille d'un galvanometre place hors flux. Apres une etude theorique, on indique des resultats de mesures obtenus, en insistant particulierement sur un nouveau type de convertisseur, caracterise par sa structure stratifiee. Les convertisseurs sont tres interessants pour tracer, des cartes de flux a partir de 10{sup 7} neutrons cm{sup -2}*s{sup -1}. Ils sont utilisables pour des flux de 10{sup 14} neutrons cm{sup -2}*s{sup -1}. Ils fonctionnent correctement dans du gaz carbonique sous pression a 400 C. Des points restent cependant a eclaircir concernant les courants parasites dans les detecteurs et le comportement des dielectriques pendant leur irradiation. (auteur)
Measurements of neutron flux in the RA reactor
International Nuclear Information System (INIS)
Raisic, N.
1961-12-01
This report includes the following separate parts: Thermal neutron flux in the experimental channels od RA reactor; Epithermal neutron flux in the experimental channels od RA reactor; Fast neutron flux in the experimental channels od RA reactor; Thermal neutron flux in the thermal column and biological experimental channel; Neutronic measurements in the RA reactor cell; Temperature reactivity coefficient of the RA reactor; design of the device for measuring the activity of wire [sr
Review of unfolding methods for neutron flux dosimetry
International Nuclear Information System (INIS)
Stallmann, F.W.; Kam, F.B.K.
1975-01-01
The primary method in reactor dosimetry is the foil activation technique. To translate the activation measurements into neutron fluxes, a special data processing technique called unfolding is needed. Some general observations about the problems and the reliability of this approach to reactor dosimetry are presented. Current unfolding methods are reviewed. 12 references. (auth)
Solving the uncommon reactor core neutronics problems
International Nuclear Information System (INIS)
Vondy, D.R.; Fowler, T.B.
1980-01-01
The common reactor core neutronics problems have fundamental neutron space, energy spectrum solutions. Typically the most positive eigenvalue is associated with an all-positive flux for the pseudo-steady-state condition (k/sub eff/), or the critical state is to be effected by selective adjustment of some variable such as the fuel concentration. With sophistication in reactor analysis has come the demand for solutions of other, uncommon neutronics problems. Importance functionss are needed for sensitivity and uncertainty analyses, as for ratios of intergral reaction rates such as the fuel conversion (breeding) ratio. The dominant higher harmonic solution is needed in stability analysis. Typically the desired neutronics solution must contain negative values to qualify as a higher harmonic or to satisfy a fixed source containing negative values. Both regular and adjoint solutions are of interest as are special integrals of the solutions to support analysis
EL-2 reactor: Thermal neutron flux distribution; EL-2: Repartition du flux de neutrons thermiques
Energy Technology Data Exchange (ETDEWEB)
Rousseau, A; Genthon, J P [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1958-07-01
The flux distribution of thermal neutrons in EL-2 reactor is studied. The reactor core and lattices are described as well as the experimental reactor facilities, in particular, the experimental channels and special facilities. The measurement shows that the thermal neutron flux increases in the central channel when enriched uranium is used in place of natural uranium. However the thermal neutron flux is not perturbed in the other reactor channels by the fuel modification. The macroscopic flux distribution is measured according the radial positioning of fuel rods. The longitudinal neutron flux distribution in a fuel rod is also measured and shows no difference between enriched and natural uranium fuel rods. In addition, measurements of the flux distribution have been effectuated for rods containing other material as steel or aluminium. The neutron flux distribution is also studied in all the experimental channels as well as in the thermal column. The determination of the distribution of the thermal neutron flux in all experimental facilities, the thermal column and the fuel channels has been made with a heavy water level of 1825 mm and is given for an operating power of 1000 kW. (M.P.)
Device for controlling neutron flux
International Nuclear Information System (INIS)
Hirukawa, Koji.
1979-01-01
Purpose: To separately provide a reflux-type control tube and a scramming control rod, and arrange respective control tube groups concentrically from the core to the outside, and add a monitoring material thereto, thereby simplifying control system, flattening and controlling the core power. Constitution: At the central part of four fuel assemblies there are arranged reflux type control tubes filled with a solution of a neutron control substance, and four of the fundamental units are assembled and scramming control rods and provided at the center of these fundamental units in a freely insertable and removable manner, thereby forming a unit. The units of reflux type control tubes filled with the solutions of the neutron control substances having different concentrations are arranged concentrically from the core to the outside. Further, a monitoring substance such as phosphoric acid or the like, displaying similar behaviors as the solution is added in the solution, and the concentration of the solution is continuously measured. (Sekiya, K.)
Self-powered neutron flux detector assembly
International Nuclear Information System (INIS)
Allan, C.J.; McIntyre, I.L.
1980-01-01
A self-powered neutron flux detector has both the central emitter electrode and its surrounding collector electrode made of inconel 600. The lead cables may also be made of inconel. Other nickel alloys, or iron, nickel, titamium, chromium, zirconium or their alloys may also be used for the electrodes
Neutron flux enhancement in the NRAD reactor
International Nuclear Information System (INIS)
Weeks, A.A.; Heidel, C.C.; Imel, G.R.
1988-01-01
In 1987 a series of experiments were conducted at the NRAD reactor facility at Argonne National Laboratory - West (ANL-W) to investigate the possibility of increasing the thermal neutron content at the end of the reactor's east beam tube through the use of hydrogenous flux traps. It was desired to increase the thermal flux for a series of experiments to be performed in the east radiography cell, in which the enhanced flux was required in a relatively small volume. Hence, it was feasible to attempt to focus the cross section of the beam to a smaller area. Two flux traps were constructed from unborated polypropylene and tested to determine their effectiveness. Both traps were open to the entire cross-sectional area of the neutron beam (as it emerges from the wall and enters the beam room). The sides then converged such that at the end of the trap the beam would be 'focused' to a greater intensity. The differences in the two flux traps were primarily in length, and hence angle to the beam as the inlet and outlet cross-sectional areas were held constant. The experiments have contributed to the design of a flux trap in which a thermal flux of nearly 10 9 was obtained, with an enhancement of 6.61
Neutron flux enhancement in the NRAD reactor
International Nuclear Information System (INIS)
Weeks, A.A.; Heidel, C.C.; Imel, G.R.
1988-01-01
In 1987 a series of experiments were conducted at the NRAD reactor facility at Argonne National Laboratory - West (ANL-W) to investigate the possibility of increasing the thermal neutron content at the end of the reactor's east beam tube through the use of hydrogenous flux traps. It was desired to increase the thermal flux for a series of experiments to be performed in the east radiography cell, in which the enhanced flux was required in a relatively small volume. Hence, it was feasible to attempt to focus the cross section of the beam to a smaller area. Two flux traps were constructed from unborated polypropylene and tested to determine their effectiveness. Both traps were open to the entire cross-sectional area of the neutron beam (as it emerges from the wall and enters the beam room). The sides then converged such that at the end of the trap the beam would be 'focused' to a greater intensity. The differences in the two flux traps were primarily in length, and hence angle to the beam as the inlet and outlet cross-sectional areas were held constant. It should be noted that merely placing a slab of polypropylene in the beam will not yield significant multiplication as neutrons are primarily scattered away
An alternative method for the measurement of neutron flux
Indian Academy of Sciences (India)
A simple and easy method for measuring the neutron flux is presented. This paper deals with the experimental verification of neutron dose rate–flux relationship for a non-dissipative medium. Though the neutron flux cannot be obtained from the dose rate in a dissipative medium, experimental result shows that for ...
Energy Technology Data Exchange (ETDEWEB)
Raisic, N [Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Serbia and Montenegro)
1961-12-15
This report includes results of the following measurements performed at the RA reactor: thermal neutron flux in the experimental channels, epithermal and fast neutron flux, neutron flux in the biological shield, neutron flux distribution in the reactor cell.
Generating energy dependent neutron flux maps for effective ...
African Journals Online (AJOL)
For activation analysis and irradiation scheme of miniature neutron source reactor, designers or engineers usually require information on thermal neutron flux levels and other energy group flux levels (such as fast, resonance and epithermal). A methodology for readily generating such flux maps and flux profiles for any ...
SHREDI, Neutron Flux and Neutron Activation in 2-D Shields by Removal Diffusion
International Nuclear Information System (INIS)
Daneri, A.; Toselli, G.
1976-01-01
1 - Nature of physical problem solved: SHREDI is a removal - diffusion neutron shielding code. The program computes neutron fluxes and activations in bidimensional sections (x,y or r,z) of the shield. It is also possible to consider shielding points with the same y or z coordinate (mono-dimensional problems). 2 - Method of solution: The integrals which define the removal fluxes are computed in some shield points by means of a particular algorithm based on the Simpson's and trapezoidal rules. For the diffusion calculation the finite difference method is used. The removal sources are interpolated in all diffusion points by Chebyshev polynomials. 3 - Restrictions on the complexity of the problem: Maxima: number of removal energy groups NGR = 40; number of diffusion energy groups NGD = 40; number of the reactor core and shield materials NCMP = 50; number of core mesh points in r (or x) direction for integral calculation = 75; number of core mesh points in z (or y) direction for integral calculation = 75; number of core mesh points in theta (or z) direction for integral calculation = 75; number of shield mesh points for the neutron flux calculation in r (or x) direction NPX = 200; number of shield mesh points for the neutron flux calculation in z (or y) direction NPY = 200; n.b. (NPX * NPY) le 12000
SLAROM, Neutron Flux Distribution and Spectra in Lattice Cell
International Nuclear Information System (INIS)
Nakagawa, M.; Tsuchihashi, K.
2002-01-01
1 - Description of program or function: SLAROM solves the neutron integral transport equations to determine flux distribution and spectra in a lattice and calculates cell averaged effective cross sections. 2 - Method of solution: Collision probability method for cell calculation and 1D diffusion for core calculation. 3 - Restrictions on the complexity of the problem: Variable dimensions are used throughout the program so that computer core requirements depend on a variety of program parameters
Linden, P; Dahl, B; Pázsit, I; Por, G
1999-01-01
We have performed laboratory measurements of the neutron flux and its gradient in a static model experiment, similar to a model problem proposed in Pazsit (Ann. Nucl. Energy 24 (1997) 1257). The experimental system consists of a radioactive neutron source located in a water tank. The measurements are performed using a recently developed very small optical fibre detector. The measured values of the flux and its gradient are then used to test the possibility of localising the source. The results show that it is possible to measure the flux on the circumference of a circle and from this calculate the flux gradient vector. Then, by comparison of the measured quantities with corresponding MCNP calculations, both the direction and the distance to the source are found and thus the position of the source can be determined.
The measurements of thermal neutron flux distribution in a paraffin
Indian Academy of Sciences (India)
The term `thermal flux' implies a Maxwellian distribution of velocity and energy corresponding to the most probable velocity of 2200 ms-1 at 293.4 K. In order to measure the thermal neutron flux density, the foil activation method was used. Thermal neutron flux determination in paraffin phantom by counting the emitted rays of ...
The isotope density inverse problem in multigroup neutron transport
International Nuclear Information System (INIS)
Zazula, J.M.
1981-01-01
The inverse problem for stationary multigroup anisotropic neutron transport is discussed in order to search for isotope densities in multielement medium. The spatial- and angular-integrated form of neutron transport equation, in terms of the flux in a group - density of an element spatial correlation, leads to a set of integral functionals for the densities weighted by the group fluxes. Some methods of approximation to make the problem uniquently solvable are proposed. Particularly P 0 angular flux information and the spherically-symetrical geometry of an infinite medium are considered. The numerical calculation using this method related to sooner evaluated direct problem data gives promising agreement with primary densities. This approach would be the basis for further application in an elemental analysis of a medium, using an isotopic neutron source and a moving, energy-dependent neutron detector. (author)
First flux measurement in a SINQ supermirror neutron guide
Energy Technology Data Exchange (ETDEWEB)
Janssen, S.; Schlumpf, N.; Bauer, G. [Paul Scherrer Inst. (PSI), Villigen (Switzerland)
1997-09-01
On Dec. 3, 1996, the Swiss spallation neutron source SINQ was taken into operation and produced its first neutrons successfully. The neutron spectrum within one of the supermirror guides was estimated by a chopper Time-of-Flight method. The result shows a 30% higher neutron intensity at the flux maximum than expected from previous Monte-Carlo simulations. (author) 1 fig., 4 refs.
Albedo analytical method for multi-scattered neutron flux calculation in cavity
International Nuclear Information System (INIS)
Shin, Kazuo; Selvi, S.; Hyodo, Tomonori
1986-01-01
A simple formula which describes multi-scattered neutron flux in a spherical cavity was derived based on the albedo concept. The formura treats a neutron source which has an arbitrary energy-angle distribution and is placed at any point in the cavity. The derived formula was applied to the estimation of neutron fluxes in two cavities, i.e. a spherical concrete cell with a 14-MeV neutron source at the center and the ''YAYOI'' reactor cavity with a pencil beam of reactor neutrons. The results of the analytical formula agreed very well with the reference data in the both problems. It was concluded that the formula is applicable to estimate the neutron fluxes in a spherical cell except for special cases that tangential source neutrons are incident to the cavity wall. (author)
Benchmarking time-dependent neutron problems with Monte Carlo codes
International Nuclear Information System (INIS)
Couet, B.; Loomis, W.A.
1990-01-01
Many nuclear logging tools measure the time dependence of a neutron flux in a geological formation to infer important properties of the formation. The complex geometry of the tool and the borehole within the formation does not permit an exact deterministic modelling of the neutron flux behaviour. While this exact simulation is possible with Monte Carlo methods the computation time does not facilitate quick turnaround of results useful for design and diagnostic purposes. Nonetheless a simple model based on the diffusion-decay equation for the flux of neutrons of a single energy group can be useful in this situation. A combination approach where a Monte Carlo calculation benchmarks a deterministic model in terms of the diffusion constants of the neutrons propagating in the media and their flux depletion rates thus offers the possibility of quick calculation with assurance as to accuracy. We exemplify this approach with the Monte Carlo benchmarking of a logging tool problem, showing standoff and bedding response. (author)
A Novel Detector for High Neutron Flux Measurements
International Nuclear Information System (INIS)
Singo, T. D.; Wyngaardt, S. M.; Papka, P.; Dobson, R. T.
2010-01-01
Measuring alpha particles from a neutron induced break-up reaction with a mass spectrometer can be an excellent tool for detecting neutrons in a high neutron flux environment. Break-up reactions of 6 Li and 12 C can be used in the detection of slow and fast neutrons, respectively. A high neutron flux detection system that integrates the neutron energy sensitive material and helium mass spectrometer has been developed. The description of the detector configuration is given and it is soon to be tested at iThemba LABS, South Africa.
The development of ex-core neutron flux monitoring system for integral reactor
Energy Technology Data Exchange (ETDEWEB)
Lee, J. K.; Kwon, H. J.; Park, H. Y.; Koo, I. S
2004-12-01
Due to the arrangement of major components within the reactor vessel, the integral reactor has relatively long distance between the core support barrel and the reactor vessel when compared with the currently operating plants. So, a neutron flux leakage at the ex-vessel represents a relatively low flux level which may generate some difficulties in obtaining a wide range of neutron flux information including the source range one. This fact may have an impact upon the design and fabrication of an ex-core neutron flux detector. Therefore, it is required to study neutron flux detectors that are suitable for the installation location and characteristics of an integral reactor. The physical constraints of an integral reactor should be considered when one designs and develops the ex-core neutron flux monitoring detectors and their systems. As a possible installation location of the integral reactor ex-core neutron flux detector assembly, two candidate locations are considered, that is, one is between the core support barrel and the reactor vessel and the other is within the Internal Shielding Tank(IST). And, for these locations, some factors such as the environmental requirements and geometrical restrictions are investigated In the case of considering the inside of the IST as a ex-core neutron flux detector installation position, an electrical insulation problem and a low neutron flux measurement problem arose and when considering the inside of the reactor vessel, a detector's sensitivity variation problem, an electrical insulation problem, a detector's insertion and withdrawal problem, and a high neutron flux measurement problem were encountered. Through a survey of the detector installation of the currently operating plants and detector manufacturer's products, the proposed structure and specifications of an ex-core neutron flux detector are suggested. And, the joint ownership strategy for a proposed detector model is also depicted. At the end, by studying
The development of ex-core neutron flux monitoring system for integral reactor
International Nuclear Information System (INIS)
Lee, J. K.; Kwon, H. J.; Park, H. Y.; Koo, I. S.
2004-12-01
Due to the arrangement of major components within the reactor vessel, the integral reactor has relatively long distance between the core support barrel and the reactor vessel when compared with the currently operating plants. So, a neutron flux leakage at the ex-vessel represents a relatively low flux level which may generate some difficulties in obtaining a wide range of neutron flux information including the source range one. This fact may have an impact upon the design and fabrication of an ex-core neutron flux detector. Therefore, it is required to study neutron flux detectors that are suitable for the installation location and characteristics of an integral reactor. The physical constraints of an integral reactor should be considered when one designs and develops the ex-core neutron flux monitoring detectors and their systems. As a possible installation location of the integral reactor ex-core neutron flux detector assembly, two candidate locations are considered, that is, one is between the core support barrel and the reactor vessel and the other is within the Internal Shielding Tank(IST). And, for these locations, some factors such as the environmental requirements and geometrical restrictions are investigated In the case of considering the inside of the IST as a ex-core neutron flux detector installation position, an electrical insulation problem and a low neutron flux measurement problem arose and when considering the inside of the reactor vessel, a detector's sensitivity variation problem, an electrical insulation problem, a detector's insertion and withdrawal problem, and a high neutron flux measurement problem were encountered. Through a survey of the detector installation of the currently operating plants and detector manufacturer's products, the proposed structure and specifications of an ex-core neutron flux detector are suggested. And, the joint ownership strategy for a proposed detector model is also depicted. At the end, by studying the ex
Problems and prospects of neutron imaging
International Nuclear Information System (INIS)
Kobayashi, Hisao
2008-01-01
Technical problems and future prospects of neutron imaging and neutron radiography are reviewed and discussed for further development. For technical problems, neutron sources together with cold neutron, ultra-cold neutron, epithermal and fast-neutron beams, energy converters, and the intensity of neutron beam, dynamic range associated with imaging procedure, etc, are reviewed. As standardization, such indicators as beam purity, sensitivity, image quality, and beam quality are discussed and limitation of neutron radiography is also presented. As neutron imaging has developed as a nondestructive testing technique in industrial applications, further problems and prospects of quality control and qualification to perform neutron radiography, standardization and international cooperation of neutron imaging are discussed. (S. Ohno)
MURALB - a programme for calculating neutron fluxes in many groups
International Nuclear Information System (INIS)
MacDougall, J.
1977-09-01
The program MURALB solves the multi-group transport equation (with no upscatter) in many equal lethargy groups to produce neutron fluxes in these groups. The code has been made very flexible by confining the spatial flux solution to a single subroutine which takes as input the cross section data and source for a single group and calculates the flux for that group. In this way by supplying different versions of this routine different geometries and methods of solution of the transport equation may be treated. At present plane, cylindrical and spherical diffusion theory and collision probability solutions are available, together with a two region collision probability solution for a rod in a square cell. There is no basic restriction to one dimension but the practical size of problem tends to be limited to about 30 spatial regions by core storage requirements. In addition to the flux solution, the code calculates neutron balance, reaction rates and few groups cross sections for each mesh region, together with the values averaged over the system (cell or reactor). The program is available both as a stand-alone code and integrated into the COSMOS system. (author)
Investigating The Neutron Flux Distribution Of The Miniature Neutron Source Reactor MNSR Type
International Nuclear Information System (INIS)
Nguyen Hoang Hai; Do Quang Binh
2011-01-01
Neutron flux distribution is the important characteristic of nuclear reactor. In this article, four energy group neutron flux distributions of the miniature neutron source reactor MNSR type versus radial and axial directions are investigated in case the control rod is fully withdrawn. In addition, the effect of control rod positions on the thermal neutron flux distribution is also studied. The group constants for all reactor components are generated by the WIMSD code, and the neutron flux distributions are calculated by the CITATION code. The results show that the control rod positions only affect in the planning area for distribution in the region around the control rod. (author)
Analyzer of neutron flux in real time
International Nuclear Information System (INIS)
Rojas S, A.S.; Carrillo M, R.A.; Balderas, E.G.
1999-01-01
With base in the study of the real signals of neutron flux of instability events occurred in the Laguna Verde nuclear power plant where the nucleus oscillation phenomena of the reactor are in the 0 to 2.5 Hz range, it has been seen the possibility about the development a surveillance and diagnostic equipment capable to analyze in real time the behavior of nucleus in this frequencies range. An important method for surveillance the stability of the reactor nucleus is the use of the Power spectral density which allows to determine the frequencies and amplitudes contained in the signals. It is used an instrument carried out by LabVIEW graphic programming with a data acquisition card of 16 channels which works at Windows 95/98 environment. (Author)
Device for measuring neutron-flux distribution density
International Nuclear Information System (INIS)
Rozenbljum, N.D.; Mitelman, M.G.; Kononovich, A.A.; Kirsanov, V.S.; Zagadkin, V.A.
1977-01-01
An arrangement is described for measuring the distribution of neutron flux density over the height of a nuclear reactor core and which may be used for monitoring energy release or for detecting deviations of neutron flux from an optimal level so that subsequent balance can be achieved. It avoids mutual interference of detectors. Full constructional details are given. (UK)
APPLE, Plot of 1-D Multigroup Neutron Flux and Gamma Flux and Reaction Rates from ANISN
International Nuclear Information System (INIS)
Kawasaki, Hiromitsu; Seki, Yasushi
1983-01-01
A - Description of problem or function: The APPLE-2 code has the following functions: (1) It plots multi-group energy spectra of neutron and/or gamma ray fluxes calculated by ANISN, DOT-3.5, and MORSE. (2) It gives an overview plot of multi-group neutron fluxes calculated by ANISN and DOT-3.5. The scalar neutron flux phi(r,E) is plotted with the spatial parameter r linear along the Y-axis, logE along the X-axis and log phi(r,E) in the Z direction. (3) It calculates the spatial distribution and region volume integrated values of reaction rates using the scalar flux calculated with ANISN and DOT-3.5. (4) Reaction rate distribution along the R or Z direction may be plotted. (5) An overview plot of reaction rates or scalar fluxes summed over specified groups may be plotted. R(ri,zi) or phi(ri,zi) is plotted with spatial parameters r and z along the X- and Y-axes in an orthogonal coordinate system. (6) Angular flux calculated by ANISN is rearranged and a shell source at any specified spatial mesh point may be punched out in FIDO format. The shell source obtained may be employed in solving deep penetration problems with ANISN, when the entire reactor system is divided into two or more parts and the neutron fluxes in two adjoining parts are connected by using the shell source. B - Method of solution: (a) The input data specification is made as simple as possible by making use of the input data required in the radiation transport code. For example, geometry related data in ANISN and DOT are transmitted to APPLE-2 along with scalar flux data so as to reduce duplicity and errors in reproducing these data. (b) Most the input data follow the free form FIDO format developed at Oak Ridge National Laboratory and used in the ANISN code. Furthermore, the mixture specifying method used in ANISN is also employed by APPLE-2. (c) Libraries for some standard response functions required in fusion reactor design have been prepared and are made available to users of the 42-group neutron
Dimensional reduction of a generalized flux problem
International Nuclear Information System (INIS)
Moroz, A.
1992-01-01
In this paper, a generalized flux problem with Abelian and non-Abelian fluxes is considered. In the Abelian case we shall show that the generalized flux problem for tight-binding models of noninteracting electrons on either 2n- or (2n + 1)-dimensional lattice can always be reduced to an n-dimensional hopping problem. A residual freedom in this reduction enables one to identify equivalence classes of hopping Hamiltonians which have the same spectrum. In the non-Abelian case, the reduction is not possible in general unless the flux tensor factorizes into an Abelian one times are element of the corresponding algebra
Neutron flux calculation by means of Monte Carlo methods
International Nuclear Information System (INIS)
Barz, H.U.; Eichhorn, M.
1988-01-01
In this report a survey of modern neutron flux calculation procedures by means of Monte Carlo methods is given. Due to the progress in the development of variance reduction techniques and the improvements of computational techniques this method is of increasing importance. The basic ideas in application of Monte Carlo methods are briefly outlined. In more detail various possibilities of non-analog games and estimation procedures are presented, problems in the field of optimizing the variance reduction techniques are discussed. In the last part some important international Monte Carlo codes and own codes of the authors are listed and special applications are described. (author)
TORT/MCNP coupling method for the calculation of neutron flux around a core of BWR
International Nuclear Information System (INIS)
Kurosawa, M.
2005-01-01
For the analysis of BWR neutronics performance, accurate data are required for neutron flux distribution over the In-Reactor Pressure Vessel equipments taking into account the detailed geometrical arrangement. The TORT code can calculate neutron flux around a core of BWR in a three-dimensional geometry model, but has difficulties in fine geometrical modelling and lacks huge computer resource. On the other hand, the MCNP code enables the calculation of the neutron flux with a detailed geometry model, but requires very long sampling time to give enough number of particles. Therefore, a TORT/MCNP coupling method has been developed to eliminate the two problems mentioned above in each code. In this method, the TORT code calculates angular flux distribution on the core surface and the MCNP code calculates neutron spectrum at the points of interest using the flux distribution. The coupling method will be used as the DOT-DOMINO-MORSE code system. This TORT/MCNP coupling method was applied to calculate the neutron flux at points where induced radioactivity data were measured for 54 Mn and 60 Co and the radioactivity calculations based on the neutron flux obtained from the above method were compared with the measured data. (authors)
TORT/MCNP coupling method for the calculation of neutron flux around a core of BWR.
Kurosawa, Masahiko
2005-01-01
For the analysis of BWR neutronics performance, accurate data are required for neutron flux distribution over the In-Reactor Pressure Vessel equipments taking into account the detailed geometrical arrangement. The TORT code can calculate neutron flux around a core of BWR in a three-dimensional geometry model, but has difficulties in fine geometrical modelling and lacks huge computer resource. On the other hand, the MCNP code enables the calculation of the neutron flux with a detailed geometry model, but requires very long sampling time to give enough number of particles. Therefore, a TORT/MCNP coupling method has been developed to eliminate the two problems mentioned above in each code. In this method, the TORT code calculates angular flux distribution on the core surface and the MCNP code calculates neutron spectrum at the points of interest using the flux distribution. The coupling method will be used as the DOT-DOMINO-MORSE code system. This TORT/MCNP coupling method was applied to calculate the neutron flux at points where induced radioactivity data were measured for 54Mn and 60Co and the radioactivity calculations based on the neutron flux obtained from the above method were compared with the measured data.
Neutron flux measurements in C-9 capsule pressure tube
International Nuclear Information System (INIS)
Barbos, D.; Roth, C. S.; Gugiu, D.; Preda, M.
2001-01-01
C-9 capsule is a fuel testing facility in which the testing consists of a daily cycle ranging between the limits 100% power to 50% power. C-9 in-pile section with sample holder an instrumentation are introduced in G-9 and G-10 experimental channels. The experimental fuel channel has a maximum value when the in-pile section (pressure tube) is in G-9 channel and minimum value in G-10 channel. In this paper the main goals are determination or measurements of: - axial thermal neutron flux distribution in C-9 pressure tube both in G-9 and G-10 channel; - ratio of maximum neutron flux value in G-9 and the same value in G-9 channel and the same value in G-10 channel; - neutron flux-spectrum. On the basis of axial neutron flux distribution measurements, the experimental fuel element in sample holder position in set. Both axial neutron flux distribution of thermal neutrons and neutron flux-spectrum were performed using multi- foil activation technique. Activation rates were obtained by absolute measurements of the induced activity using gamma spectroscopy methods. To determine the axial thermal neutron flux distribution in G-9 and G-10, Cu 100% wire was irradiated at the reactor power of 2 MW. Ratio between the two maximum values, in G-9 and G-10 channels, is 2.55. Multi-foil activation method was used for neutron flux spectrum measurements. The neutron spectra and flux were obtained from reaction rate measurements by means of SAND 2 code. To obtain gamma-ray spectra, a HPGe detector connected to a multichannel analyzer was used. The spectrometer is absolute efficiency calibrated. The foils were irradiated at 2 MW reactor power in previously determined maximum flux position resulted from wire measurements. This reaction rates were normalized for 10 MW reactor power. Neutron self shielding corrections for the activation foils were applied. The self-shielding corrections are computed using Monte Carlo simulation methods. The measured integral flux is 1.1·10 14 n/cm 2 s
Compilation of neutron flux density spectra and reaction rates in different neutron fields. V.3
International Nuclear Information System (INIS)
Ertek, C.
1980-04-01
Upon the recommendation of the International Working Group of Reactor Radiation Measurements (IWGRRM) a compilation of documents containing neutron flux density spectra and the reaction rates obtained by activiation and fission foils in different neutron fields is presented
Energy Technology Data Exchange (ETDEWEB)
Brisbois, J; Lott, M; Manent, G [Commissariat a l' Energie Atomique, Fontenay-aux-Roses (France). Centre d' Etudes Nucleaires
1964-07-01
The problem of neutron flux measurements using activation detectors is studied in the particular case of protection research. It is shown how it possible, it is possible, using a known thermal flux, to organise a coherent calibration system for all the detectors. The rapid neutron detectors are calibrated with respect to a reference detector (phosphorus) in a natural uranium converter; the intermediate neutron detectors with respect to gold in the axial channel of ZOE. This method makes it possible to minimise the errors due to the activation cross-sections. A brief description is given of the counting room of the Pile Safety Study Service, as well of the practical utilisation characteristics of the counters employed. (authors) [French] Le probleme de la mesure des flux de neutrons au moyen de detecteurs par activation est etudie dans le cas particulier des etudes de protections. On montre comment, a partir d'un flux thermique connu, on peut organiser un systeme coherent d'etalonnage de tous les detecteurs. Les detecteurs de neutrons rapides sont etalonnes par rapport a un detecteur de reference (phosphore) dans un convertisseur en uranium naturel; les detecteurs de neutrons intermediaires, par rapport a l'or dans le canal axial de ZOE, Cette methode permet de minimiser les erreurs dues aux sections efficaces d'activation. On decrit sommairement la salle de comptage du Service d'Etudes de Protections de Piles et on indique les caracteristiques d'emploi pratique des detecteurs utilises. (auteurs)
Using thermalizers in measuring 'Ukryttia' object's FCM neutron fluxes
Krasnyanskaya, O G; Odinokin, G I; Pavlovich, V N
2003-01-01
The results of research of a thermalizer (heater) width influence on neutron thermalization efficiency during FCM neutron flux measuring in the 'Ukryttia' are described. The calculations of neutron flux densities were performed by the Monte-Carlo method with the help of computer code MCNP-4C for FCM different models.Three possible installations of detectors were considered: on FCM surface,inside the FCM, and inside the concrete under the FCM layer. It was shown,that in order to increase the sensitivity of neutron detectors in intermediate and fast neutrons field,and consequently, to decrease the dependence of the readings of spectral distribution of neutron flux,it is necessary to position the detector inside the so-called thermalizer or heater. The most reasonable application of thick 'heaters' is the situation, when the detector is placed on FCM surface.
International Nuclear Information System (INIS)
Li, X. Z.; Wei, Q. M.; Liu, B.; Zhu, X. G.; Ren, S. L.
2007-01-01
Pursuing nuclear energy with no nuclear contamination has been a long endeavor since the first fission reactor in 1942. Four major concepts have been the key issues: i.e. resonance, negative feed back, self-sustaining, nuclear radiation. When nuclear energy was just discovered in laboratory, the key issue was to enlarge it from the micro-scale to the macro-scale. Slowing-down the neutrons was the key issue to enhance the fission cross-section in order to build-up the neutron flux through the chain-reactions using resonance between neutron and fissile materials. Once the chain-reaction was realized, the negative feed-back was the key issue to keep the neutron flux at the allowable level. The negative reaction coefficient was introduced by the thermal expansion, and the resonant absorption in cadmium or boron was used to have a self-sustaining fission reactor with neutron flux. Then the strong neutron flux became the origin of all nuclear contamination, and a heavy shielding limits the application of the nuclear energy. The fusion approach to nuclear energy was much longer; nevertheless, it evolved with the similar issues. The resonance between deuteron and triton was resorted to enlarge the fusion cross section in order to keep a self-sustaining hot plasma. However, the 14 MeV neutron emission became the origin of all nuclear contamination again. Deuteron plus helium-3 fusion reaction was proposed to avoid neutron emission although there are two more difficulties: the helium-3 is supposed to be carried back from the moon; and much more higher temperature plasma has to be confined while 50 years needed to realized the deuteron-triton plasma already. Even if deuteron plus helium-3 fusion plasma might be realized in a much higher temperature plasma, we still have the neutron emission from the deuteron-deuteron fusion reaction in the deuteron plus helium-3 fusion plasma. Polarized deuteron-deuteron fusion reaction was proposed early in 1980's to select the neutron
Fast neutron fluxes distribution in Egyptian ilmenite concrete
International Nuclear Information System (INIS)
Megahed, R.M.; Abou El-Nasr, T.Z.; Bashter, I.I.
1978-01-01
This work is concerned with the study of the distribution of fast neutron fluxes in a new type of heavy concrete made from Egyptian ilmenite ores. The neutron source used was a collimated beam of reactor neutrons emitted from one of the horizontal channels of the ET-RR-1 reactor. Measurements were carried-out using phosphorous activation detectors. Iso-flux curves were represented which give directly the shape and thickness required to attenuate the emitted fast neutron flux to a certain value. The relaxation lengths were also evaluated from the measured data for both disc monodirectional source and infinite plane monodirectional source. The obtained values were compared with that calculated using the derived values of relative number densities and microscopic removal cross-sections of the different constituents. The obtained data show that ilmenite concrete attenuates fast neutron flux more strongly than ordinary concrete. A semiemperical formula was derived to calculate the fast neutron flux at different thicknesses along the beam axis. Another semiemperical formula was also derived to calculate the fast neutron flux in ordinary concrete along the beam axis using the corresponding value in ilmenite concrete
Monitoring the fast neutrons in a high flux: The case for 242Pu fission chambers
International Nuclear Information System (INIS)
Filliatre, P.; Jammes, C.; Oriol, L.; Geslot, B.; Vermeeren, L.
2009-01-01
Fission chambers are widely used for on-line monitoring of neutron fluxes in irradiation reactors. A selective measurement of a component of interest of the neutron flux is possible in principle thanks to a careful choice of the deposit material. However, measuring the fast component is challenging when the flux is high (up to 10 15 n/cm 2 /s) with a significant thermal component. The main problem is that the isotopic content of a material selected for its good response to fast neutrons evolves with irradiation, so that the material is more and more sensitive to thermal neutrons. Within the framework of the FNDS (Fast Neutron Detector System) project, we design tools that simulate the evolution of the isotopic composition and fission rate for several deposits under any given flux. In the case of a high flux with a significant thermal component, 242 Pu is shown after a comprehensive study of all possibilities to be the best choice for measuring the fast component, as long as its purity is sufficient. If an estimate of the thermal flux is independently available, one can correct the signal for that component. This suggests a system of two detectors, one of which being used for such a correction. It is of very high interest when the detectors must be operated up to a high neutron fluence. (authors)
Monitoring the fast neutrons in a high flux: The case for {sup 242}Pu fission chambers
Energy Technology Data Exchange (ETDEWEB)
Filliatre, P.; Jammes, C.; Oriol, L.; Geslot, B. [Commissariat a l' Energie Atomique, DEN/SPEX/LDCI, Centre de Cadarache, F-13108 Saint-Paul-lez-Durance (France); Vermeeren, L. [SCK-CEN, Boeretang 200, B-2400 Mol (Belgium)
2009-07-01
Fission chambers are widely used for on-line monitoring of neutron fluxes in irradiation reactors. A selective measurement of a component of interest of the neutron flux is possible in principle thanks to a careful choice of the deposit material. However, measuring the fast component is challenging when the flux is high (up to 10{sup 15} n/cm{sup 2}/s) with a significant thermal component. The main problem is that the isotopic content of a material selected for its good response to fast neutrons evolves with irradiation, so that the material is more and more sensitive to thermal neutrons. Within the framework of the FNDS (Fast Neutron Detector System) project, we design tools that simulate the evolution of the isotopic composition and fission rate for several deposits under any given flux. In the case of a high flux with a significant thermal component, {sup 242}Pu is shown after a comprehensive study of all possibilities to be the best choice for measuring the fast component, as long as its purity is sufficient. If an estimate of the thermal flux is independently available, one can correct the signal for that component. This suggests a system of two detectors, one of which being used for such a correction. It is of very high interest when the detectors must be operated up to a high neutron fluence. (authors)
Some neutronic studies on flux-trap type moderator
International Nuclear Information System (INIS)
Kiyanagi, Y.; Watanabe, N.
1991-01-01
The neutronic performance of a flux-trap type moderator was studied by computer simulation in connection with the KENS-II target-moderator system. It was confirmed that this system can provide 1.3-1.4 times higher neutron intensity than a traditional wing-geometry moderator system. (author)
Determination of average activating thermal neutron flux in bulk samples
International Nuclear Information System (INIS)
Doczi, R.; Csikai, J.; Doczi, R.; Csikai, J.; Hassan, F. M.; Ali, M.A.
2004-01-01
A previous method used for the determination of the average neutron flux within bulky samples has been applied for the measurements of hydrogen contents of different samples. An analytical function is given for the description of the correlation between the activity of Dy foils and the hydrogen concentrations. Results obtained by the activation and the thermal neutron reflection methods are compared
HAV-1-A multipurpose multimonitor for reactor neutron flux characterization
International Nuclear Information System (INIS)
Diaz Rizo, O.; Alvarez, I.; Herrera, E.; Lima, L.; Tores, J.; Lopez, M.C.; Ixquiac, M.
1996-01-01
A simple method non-solid multi monitor HAV-1 for the systematic evaluation of reactor neutron flux parameters for K o neutron activation analysis is presented. Solution of Au, Zr, Co, Zn, Sn, U and Th (deposited in filter paper) are used to study the parameters alpha and f. Dissolved Lu is used to neutron temperature (Tn) determination, according to the Wescott's formalism. A multipurpose multi monitor HAV-1 preparation, certification and evaluations presented
HAV-1-A multipurpose multimonitor for reactor neutron flux characterization
Energy Technology Data Exchange (ETDEWEB)
Diaz Rizo, O; Alvarez, I; Herrera, E; Lima, L; Tores, J [Secretaria Ejecutiva para Asuntos Nucleares, Holguin (Cuba). Delegacion Territorial; Manso, M V [Centro de Isotopos, La Habana (Cuba); Lopez, M C [Instituto Nacional de Investigaciones Nucleares, Mexico City (Mexico); Ixquiac, M [Universidad de San Carlos de Guatemala, Guatemala City (Guatemala)
1997-12-31
A simple method non-solid multi monitor HAV-1 for the systematic evaluation of reactor neutron flux parameters for K{sub o} neutron activation analysis is presented. Solution of Au, Zr, Co, Zn, Sn, U and Th (deposited in filter paper) are used to study the parameters alpha and f. Dissolved Lu is used to neutron temperature (Tn) determination, according to the Wescott`s formalism. A multipurpose multi monitor HAV-1 preparation, certification and evaluations presented.
Bayesian statistics applied to neutron activation data for reactor flux spectrum analysis
International Nuclear Information System (INIS)
Chiesa, Davide; Previtali, Ezio; Sisti, Monica
2014-01-01
Highlights: • Bayesian statistics to analyze the neutron flux spectrum from activation data. • Rigorous statistical approach for accurate evaluation of the neutron flux groups. • Cross section and activation data uncertainties included for the problem solution. • Flexible methodology applied to analyze different nuclear reactor flux spectra. • The results are in good agreement with the MCNP simulations of neutron fluxes. - Abstract: In this paper, we present a statistical method, based on Bayesian statistics, to analyze the neutron flux spectrum from the activation data of different isotopes. The experimental data were acquired during a neutron activation experiment performed at the TRIGA Mark II reactor of Pavia University (Italy) in four irradiation positions characterized by different neutron spectra. In order to evaluate the neutron flux spectrum, subdivided in energy groups, a system of linear equations, containing the group effective cross sections and the activation rate data, has to be solved. However, since the system’s coefficients are experimental data affected by uncertainties, a rigorous statistical approach is fundamental for an accurate evaluation of the neutron flux groups. For this purpose, we applied the Bayesian statistical analysis, that allows to include the uncertainties of the coefficients and the a priori information about the neutron flux. A program for the analysis of Bayesian hierarchical models, based on Markov Chain Monte Carlo (MCMC) simulations, was used to define the problem statistical model and solve it. The first analysis involved the determination of the thermal, resonance-intermediate and fast flux components and the dependence of the results on the Prior distribution choice was investigated to confirm the reliability of the Bayesian analysis. After that, the main resonances of the activation cross sections were analyzed to implement multi-group models with finer energy subdivisions that would allow to determine the
Compilation of neutron flux density spectra and reaction rates in different neutron fields
International Nuclear Information System (INIS)
Ertek, C.
1979-07-01
Upon the recommendation of International Working Group of Reactor Radiation Measurements (IWGRRM), the compilation of neutron flux density spectra and the reaction rates obtained by activation and fission foils in different neutron fields is presented. The neutron fields considered are as follows: 1/E; iron block; LWR core and pressure vessel; LMFBR core and blanket; CTR first wall and blanket; fission spectrum
Development of high flux thermal neutron generator for neutron activation analysis
Energy Technology Data Exchange (ETDEWEB)
Vainionpaa, Jaakko H., E-mail: hannes@adelphitech.com [Adelphi Technology, 2003 E Bayshore Rd, Redwood City, CA 94063 (United States); Chen, Allan X.; Piestrup, Melvin A.; Gary, Charles K. [Adelphi Technology, 2003 E Bayshore Rd, Redwood City, CA 94063 (United States); Jones, Glenn [G& J Jones Enterprice, 7486 Brighton Ct, Dublin, CA 94568 (United States); Pantell, Richard H. [Department of Electrical Engineering, Stanford University, Stanford, CA (United States)
2015-05-01
The new model DD110MB neutron generator from Adelphi Technology produces thermal (<0.5 eV) neutron flux that is normally achieved in a nuclear reactor or larger accelerator based systems. Thermal neutron fluxes of 3–5 · 10{sup 7} n/cm{sup 2}/s are measured. This flux is achieved using four ion beams arranged concentrically around a target chamber containing a compact moderator with a central sample cylinder. Fast neutron yield of ∼2 · 10{sup 10} n/s is created at the titanium surface of the target chamber. The thickness and material of the moderator is selected to maximize the thermal neutron flux at the center. The 2.5 MeV neutrons are quickly thermalized to energies below 0.5 eV and concentrated at the sample cylinder. The maximum flux of thermal neutrons at the target is achieved when approximately half of the neutrons at the sample area are thermalized. In this paper we present simulation results used to characterize performance of the neutron generator. The neutron flux can be used for neutron activation analysis (NAA) prompt gamma neutron activation analysis (PGNAA) for determining the concentrations of elements in many materials. Another envisioned use of the generator is production of radioactive isotopes. DD110MB is small enough for modest-sized laboratories and universities. Compared to nuclear reactors the DD110MB produces comparable thermal flux but provides reduced administrative and safety requirements and it can be run in pulsed mode, which is beneficial in many neutron activation techniques.
Recent developments on micrometric fission chambers for high neutron fluxes
International Nuclear Information System (INIS)
Letourneau, A.; Bringer, O.; Dupont, E.; Marie, F.; Panebianco, S.; Toussaint, J. C.; Veyssiere, C.; Chabod, S.; Breaud, S.; Oriol, L.
2009-01-01
With the development of innovative nuclear systems and new generation neutron sources, the nuclear instrumentation should be adapted. Since several years, we developed microscopic fission chambers to study the transmutation of minor actinides in high thermal-neutron fluxes. The recent developments done to fulfill the drastic conditions of irradiations are described in this paper together with the feedback from the measurements. Two installations were used: the HFR of the ILL for its highest thermal neutron flux of the world and the MEGAPIE target which was the first 1 MW liquid Pb-Bi spallation target in the world. (authors)
Recent developments on micrometric fission chambers for high neutron fluxes
Energy Technology Data Exchange (ETDEWEB)
Letourneau, A. [Irfu, Service de Physique Nucleaire, CEA-Saclay, 91191 Gif-sur-Yvette (France); Bringer, O.; Dupont, E.; Marie, F.; Panebianco, S.; Toussaint, J. C.; Veyssiere, C. [Irfu, CEA-Saclay, 91191 Gif-sur-Yvette (France); Chabod, S. [LPSC, Universite Joseph Fourier Grenoble 1, CNRS/IN2P3 (France); Breaud, S.; Oriol, L. [DEN/DER/SPEX, CEA-Cadarache, Saint-Paul-lez-Durances (France)
2009-07-01
With the development of innovative nuclear systems and new generation neutron sources, the nuclear instrumentation should be adapted. Since several years, we developed microscopic fission chambers to study the transmutation of minor actinides in high thermal-neutron fluxes. The recent developments done to fulfill the drastic conditions of irradiations are described in this paper together with the feedback from the measurements. Two installations were used: the HFR of the ILL for its highest thermal neutron flux of the world and the MEGAPIE target which was the first 1 MW liquid Pb-Bi spallation target in the world. (authors)
The new high flux neutron source FRM-2 in Munich
International Nuclear Information System (INIS)
Roegler, H.J.; Wierheim, G.
2002-01-01
Quite some years ago in 1974 to be exact, the first consideration on a new neutron source started at the technical university of Munich (Germany). 27 years later the new high flux neutron source (FRM-2) was read for hot operation, now delayed by a refused approval for its third partial license by the federal government of Germany despite a wide support from the scientific community. FRM-2 is a tank-type research reactor cooled by water, moderated by heavy water and whose thermal power was limited to 20 MW maximum. The extreme compact core together with the applied inverse flux principle led to a neutron flux design value of 8.10 18 n/m 2 .s at the reflector peak. 10 beam tubes will allow an optimized use of the high neutron flux. A hot neutron source with graphite at about 2200 Celsius degrees and a cold neutron source with liquid D 2 at about 25 K will provide shifted energy spectra. The utilization of FRM-2 is many-fold: neutronography and tomography, medical irradiation, radio-nuclide production, doping of pure silicon, neutron activation analysis. (A.C.)
Absolute measurement of neutron fluxes inside the reactor core
International Nuclear Information System (INIS)
Ajdacic, S. V.
1964-10-01
The subject of this work is the development and study of two methods of neutron measurements in nuclear reactors, the new method of high neutron flux measurements and the Li 6 -semiconductor neutron spectrometer. This work is presented in four sections: Section I. The introduction explains the need for neutron measurements in reactors. A critical survey is given of the existing methods of high neutron flux measurement and methods of fast neutron spectrum determination. Section II. Theoretical basis of the work of semiconductor counters and their most important characteristics are given. Section III. The main point of this section is in presenting the basis of the new method which the author developed, i.e., the long-tube method, and the results obtained by it, with particular emphasis on absolute measurement of high neutron fluxes. Advantages and limitations of this method are discussed in details at the end of this section. Section IV. A comparison of the existing semiconductor neutron spectrometers is made and their advantages and shortcomings underlined. A critical analysis of the obtained results with the Li 6 -semiconductor spectrometer with plane geometry is given. A new type of Li 6 -semiconductor spectrometer is described, its characteristics experimentally determined, and a comparison of it with a classical Li 6 -spectrometer made (author)
Absolute measurement of neutron fluxes inside the reactor core
Energy Technology Data Exchange (ETDEWEB)
Ajdacic, S V [Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Serbia and Montenegro)
1964-10-15
The subject of this work is the development and study of two methods of neutron measurements in nuclear reactors, the new method of high neutron flux measurements and the Li{sup 6}-semiconductor neutron spectrometer. This work is presented in four sections: Section I. The introduction explains the need for neutron measurements in reactors. A critical survey is given of the existing methods of high neutron flux measurement and methods of fast neutron spectrum determination. Section II. Theoretical basis of the work of semiconductor counters and their most important characteristics are given. Section III. The main point of this section is in presenting the basis of the new method which the author developed, i.e., the long-tube method, and the results obtained by it, with particular emphasis on absolute measurement of high neutron fluxes. Advantages and limitations of this method are discussed in details at the end of this section. Section IV. A comparison of the existing semiconductor neutron spectrometers is made and their advantages and shortcomings underlined. A critical analysis of the obtained results with the Li{sup 6}-semiconductor spectrometer with plane geometry is given. A new type of Li{sup 6}-semiconductor spectrometer is described, its characteristics experimentally determined, and a comparison of it with a classical Li{sup 6}-spectrometer made (author)
Calculation of neutron fluxes in biological shield of the TRIGA Mark II reactor
International Nuclear Information System (INIS)
Bozic, M.; Zagar, T.; Ravnik, M.
2001-01-01
The complete calculation of neutron fluxes in biological shield and verification with experimental results is presented. Calculated results are obtained with TORT code (TORT-Three Dimensional Oak Ridge Discrete Ordinates Neutron/Photon Transport Code). Experimental results used for comparison are available from irradiation experiment with selected type of concrete and other materials in irradiation channel 4 in TRIGA Mark II reactor. These experimental results were used as a benchmark. Homogeneous type of problem (without inserted irradiation channel) and problem with asymmetry (inserted beam port 4, filled with different materials) were of interest for neutron flux calculation. Deviation from material data set up as original parameters is also considered (first of all presence of water in concrete and density of concrete) for type of concrete in biological shield and for selected type of concrete in irradiation channel. BUGLE-96 (47 neutron energy groups) library is used. Excellent agreement between calculated and experimental results for reaction rate is received.(author)
Energy Technology Data Exchange (ETDEWEB)
Didi, Abdessamad; Dadouch, Ahmed; Tajmouati, Jaouad; Bekkouri, Hassane [Advanced Technology and Integration System, Dept. of Physics, Faculty of Science Dhar Mehraz, University Sidi Mohamed Ben Abdellah, Fez (Morocco); Jai, Otman [Laboratory of Radiation and Nuclear Systems, Dept. of Physics, Faculty of Sciences, Tetouan (Morocco)
2017-06-15
Americium–beryllium (Am-Be; n, γ) is a neutron emitting source used in various research fields such as chemistry, physics, geology, archaeology, medicine, and environmental monitoring, as well as in the forensic sciences. It is a mobile source of neutron activity (20 Ci), yielding a small thermal neutron flux that is water moderated. The aim of this study is to develop a model to increase the neutron thermal flux of a source such as Am-Be. This study achieved multiple advantageous results: primarily, it will help us perform neutron activation analysis. Next, it will give us the opportunity to produce radio-elements with short half-lives. Am-Be single and multisource (5 sources) experiments were performed within an irradiation facility with a paraffin moderator. The resulting models mainly increase the thermal neutron flux compared to the traditional method with water moderator.
Neutron flux measuring system for nuclear reactor
International Nuclear Information System (INIS)
Aoki, Kazuo.
1977-01-01
Purpose: To avoid the generation of an undesired scram signal due to abrupt changes in the neutron level given to the detectors disposed near the boundary between the moderator and the atmosphere. Constitution: In a nuclear reactor adapted to conduct power control by the change of the level in the moderator such as heavy water, the outputs from the neutron detectors disposed vertically are averaged and the nuclear reactor is scramed corresponding to the averaged value. In this system, moderator level detectors are additionally provided to the nuclear reactor and their outputs, moderator level signal, are sent to a power averaging device where the output signals of the neutron detectors are judged if they are delivered from neutrons in the moderator or not depending on the magnitude of the level signal and the outputs of the detectors out of the moderator are substantially excluded. The reactor interlock signal from the device is utilized as a scram signal. (Seki, T.)
International Nuclear Information System (INIS)
Overwater, R.M.W.; Hoogenboom, J.E.
1994-01-01
At the Delft University of Technology Interfaculty Reactor Institute, a facility has been installed to irradiate cylindrical samples with diameters up to 15 cm and weights up to 50 kg for instrumental neutron activation analysis (INAA) purposes. To be able to do quantitative INAA on voluminous samples, it is necessary to correct for gamma-ray absorption, gamma-ray scattering, neutron absorption, and neutron scattering in the sample. The neutron absorption and the neutron scattering are discussed. An analytical solution is obtained for the diffusion equation in the geometry of the irradiation facility. For samples with known composition, the neutron flux--as a function of position in the sample--can be calculated directly. Those of unknown composition require additional flux measurements on which least-squares fitting must be done to obtain both the thermal neutron diffusion coefficient D s and the diffusion length L s of the sample. Experiments are performed to test the theory
International Nuclear Information System (INIS)
Noorddin Ibrahim; Rosnie Akang
2009-01-01
Full text: One of the major problems encountered during the irradiation of large inhomogeneous samples in performing activation analysis using neutron is the perturbation of the neutron field due to absorption and scattering of neutron within the sample as well as along the neutron guide in the case of prompt gamma activation analysis. The magnitude of this perturbation shown by self-shielding coefficient and flux depression depend on several factors including the average neutron energy, the size and shape of the sample, as well as the macroscopic absorption cross section of the sample. In this study, we use Monte Carlo N-Particle codes to simulate the variation of neutron self-shielding coefficient and thermal flux depression factor as a function of the macroscopic thermal absorption cross section. The simulation works was carried out using the high performance computing facility available at UTM while the experimental work was performed at the tangential beam port of Reactor TRIGA PUSPATI, Malaysia Nuclear Agency. The neutron flux measured along the beam port is found to be in good agreement with the simulated data. Our simulation results also reveal that total flux perturbation factor decreases as the value of absorption increases. This factor is close to unity for low absorbing sample and tends towards zero for strong absorber. In addition, sample with long mean chord length produces smaller flux perturbation than the shorter mean chord length. When comparing both the graphs of self-shielding factor and total disturbance, we can conclude that the total disturbance of the thermal neutron flux on the large samples is dominated by the self-shielding effect. (Author)
International Nuclear Information System (INIS)
Zareen Khan Abdul Jalil Khan; Mohd Idris Taib; Izhar Abu Husin; Nurfarhana Ayuni
2010-01-01
This paper presents the comparison study on In-Core neutron detector using for online flux mapping of Research and Power reactor. Technical description of in-core neutron also taken into consideration to identify the different characterization of neutron detector and describe on Self Power neutron detector (SPND) for online neutron flux mapping. Able to provide information on the neutron flux distribution and understand how in-core neutron detector are being used in nuclear power plant including to enable to state the principles of neutron detector. (author)
Device for detecting neutron flux in nuclear reactor. [BWR
Energy Technology Data Exchange (ETDEWEB)
Bessho, Y; Nishizawa, Y
1976-07-30
The object of the invention is to ensure accuracy in the operation of the nuclear reactor by reducing the difference that results between the readings of a Traversing Incore Probe (TIP) and a Local Power Range Monitor (LPRM) when the neutron flux distribution undergoes a change. In an apparatus for detecting neutrons in a nuclear reactor, an LPRM sensor comprising a layer containing a substance capable of nuclear fission, a section filled with argon gas and a collector is constructed so as to surround a TIP within a TIP guide tube at the height of the reactor axis. In this way, the LPRM detects the average value of neutron distribution in the region surrounding the TIP, so that no great difference between the readings of both the sensors is produced even if the neutron flux distribution is changed.
Neutron flux measurements at the Wendelstein VII-A stellarator
International Nuclear Information System (INIS)
Weller, A.; Maassberg, H.
1985-10-01
In addition to charge exchange analysis (CX) and charge exchange recombination spectroscopy (CXRS), the time evolution of the central ion temperature during neutral beam heated plasma discharges in the Wendelstein VII-A stellarator is derived from the neutron flux from thermal D-D reactions. In general, good quantitative agreement between the different methods is obtained. Neutron flux measurements also permit to investigate the slowing down of fast D + -ions from neutral beam injection (NBI). The results agree well with the predictions based on the assumption of a collisional slowing down mechanism. (orig.)
Neutron flux response to regulating rod random vibrations
International Nuclear Information System (INIS)
Dach, K.; Nemec, J.; Pecinka, L.
The relation is presented for the mean square value of the deflection of the rod for the n-th vibration shape on an arbitrary site. The relation may serve the obtaining of a variable which may be used both in a mechanical, i.e., stress analysis and in the determination of neutron flux fluctuations. It is demonstrated that the vibration frequency introduced in the reactor by the regulating rod has the same response in the neutron flux. This effect was used in the localization of an enormously vibrating regulating rod. (J.P.)
Determination of photon detector coefficient in neutron flux study
International Nuclear Information System (INIS)
Jedol Dayou; Azali Muhammad; Abd Aziz Mohamed; Abdul Razak Daud; Elias Saniman
1995-01-01
The efficiency of photon detector which is normally used in neutron flux measurement has been studied. The data obtain have been plotted using mathematical models in the form of reciprocal, exponential and semilog equation and subsequently efficiency coefficient of the detector has been determined. Beside that, energy quadratic equation model has also been used. It has been found that equation model selection is very important in the detector efficiency coefficient determination. In the case of energy quadratic equation, it has been found that the selection of energy set influenced the result. It can be concluded that energy quadratic equation is the best model in the neutron flux determination
Self-powered neutron and gamma-ray flux detector
International Nuclear Information System (INIS)
Allan, C.J.; Shields, R.B.; Lynch, G.F.; Cuttler, J.M.
1980-01-01
A new type of self-powered neutron detector was developed which is sensitive to both the neutron and gamma-ray fluxes. The emitter comprises two parts. The central emitter core is made of materials that generate high-energy electrons on exposure to neutrons. The outer layer acts as a gamma-ray/electron converter, and since it has a higher atomic number and higher back-scattering coefficient than the collector, increases the net outflow or emmission of electrons. The collector, which is around the emitter outer layer, is insulated from the outer layer electrically with dielectric insulation formed from compressed metal-oxide powder. The fraction of electrons given off by the emitter that is reflected back by the collector is less than the fraction of electrons emitted by the collector that is reflected back by the emitter. The thickness of the outer layer needed to achieve this result is very small. A detector of this design responds to external reactor gamma-rays as well as to neutron capture gamma-rays from the collector. The emitter core is either nickel, iron or titanium, or alloys based on these metals. The outer layer is made of platinum, tantalum, osmium, molybdenum or cerium. The detector is particularly useful for monitoring neutron and gamma ray flux intensities in nuclear reactor cores in which the neutron and gamma ray flux intensities are closely proportional, are unltimately related to the fission rate, and are used as measurements of nuclear reactor power. (DN)
Neutron Flux Distribution on Neutron Radiography Facility After Fixing the Collimator
International Nuclear Information System (INIS)
Supandi; Parikin; Mohtar; Sunardi; Roestam, S
1996-01-01
The Radiography Neutron Facility consists of an inner collimator, outer collimator, main shutter, second shutter and the sample chamber with 300 mm in diameter. Neutron beam quality depends on the neutron flux intensities distribution, L/D ratio Cd ratio, neutron/gamma ratio. The results show that the neutron flux intensity was 2.83 x 107 n cm-2.s-1, with deviation of + 7.8 % and it was distributed homogeneously at the sample position of 200 mm diameter. The beam characteristics were L/D ratio 98 and Rod 8, and neutron gamma ratio 3.08 x 105n.cm-2.mR-1 and Reactor Power was 20 MW. This technique can be used to examine sample with diameter of < 200 mm
Calculation of neutron flux in the presence of a source
International Nuclear Information System (INIS)
Planchard, J.
1993-09-01
Neutron sources are introduced into the reactors to initiate the chain reaction. For safety reasons, we have to know the distribution and evolution of the flux throughout the startup phase. The flux is calculated iteratively but convergence of the process can slow down arbitrarily as we approach criticality. A calculation method is presented, with a convergence speed which does not depend on the negative reactivity when it is small. (author). 7 refs
Excitation of neutron flux waves in reactor core transients
International Nuclear Information System (INIS)
Carew, J.F.; Neogy, P.
1983-01-01
An analysis of the excitation of neutron flux waves in reactor core transients has been performed. A perturbation theory solution has been developed for the time-dependent thermal diffusion equation in which the absorption cross section undergoes a rapid change, as in a PWR rod ejection accident (REA). In this analysis the unperturbed reactor flux states provide the basis for the spatial representation of the flux solution. Using a simplified space-time representation for the cross section change, the temporal integrations have been carried out and analytic expressions for the modal flux amplitudes determined. The first order modal excitation strength is determined by the spatial overlap between the initial and final flux states, and the cross section perturbation. The flux wave amplitudes are found to be largest for rapid transients involving large reactivity perturbations
Neutron energy spectrum flux profile of Ghana's miniature neutron source reactor core
International Nuclear Information System (INIS)
Sogbadji, R.B.M.; Abrefah, R.G.; Ampomah-Amoako, E.; Agbemava, S.E.; Nyarko, B.J.B.
2011-01-01
Highlights: → The total neutron flux spectrum of the compact core of Ghana's miniature neutron source reactor was studied. → Using 20,484 energy grids, the thermal, slowing down and fast neutron energy regions were studied. - Abstract: The total neutron flux spectrum of the compact core of Ghana's miniature neutron source reactor was understudied using the Monte Carlo method. To create small energy groups, 20,484 energy grids were used for the three neutron energy regions: thermal, slowing down and fast. The moderator, the inner irradiation channels, the annulus beryllium reflector and the outer irradiation channels were the region monitored. The thermal neutrons recorded their highest flux in the inner irradiation channel with a peak flux of (1.2068 ± 0.0008) x 10 12 n/cm 2 s, followed by the outer irradiation channel with a peak flux of (7.9166 ± 0.0055) x 10 11 n/cm 2 s. The beryllium reflector recorded the lowest flux in the thermal region with a peak flux of (2.3288 ± 0.0004) x 10 11 n/cm 2 s. The peak values of the thermal energy range occurred in the energy range (1.8939-3.7880) x 10 -08 MeV. The inner channel again recorded the highest flux of (1.8745 ± 0.0306) x 10 09 n/cm 2 s at the lower energy end of the slowing down region between 8.2491 x 10 -01 MeV and 8.2680 x 10 -01 MeV, but was over taken by the moderator as the neutron energies increased to 2.0465 MeV. The outer irradiation channel recorded the lowest flux in this region. In the fast region, the core, where the moderator is found, the highest flux was recorded as expected, at a peak flux of (2.9110 ± 0.0198) x 10 08 n/cm 2 s at 6.961 MeV. The inner channel recorded the second highest while the outer channel and annulus beryllium recorded very low flux in this region. The flux values in this region reduce asymptotically to 20 MeV.
Radiation damage in molybdenum and tungsten in high neutron fluxes
Energy Technology Data Exchange (ETDEWEB)
Veljkovic, S; Milasin, N [Institute of Nuclear Sciences Boris Kidric, Department of Reactor Materials, Vinca, Beograd (Serbia and Montenegro)
1964-04-15
The effects of radiation on molybdenum and tungsten in high neutron fluxes are presented. The changes induced, particularly defects with a high migration activation energy, are analyzed. The correlation of these changes with the basic concepts of radiation damage in solids is considered. An attempt is made to relate the defects studied with the changes in macroscopic properties (author)
TORT application in reactor pressure vessel neutron flux calculations
International Nuclear Information System (INIS)
Belousov, S.I.; Ilieva, K.D.; Antonov, S.Y.
1994-01-01
The neutron flux values onto reactor pressure vessel for WWER-1000 and WWER-440 reactors, at the places important for metal embrittlement surveillance have been calculated by 3 dimensional code TORT and synthesis method. The comparison of the results received by both methods confirms their good consistency. (authors). 13 refs., 4 tabs
Equipment for thermal neutron flux measurements in reactor R2
Energy Technology Data Exchange (ETDEWEB)
Johansson, E; Nilsson, T; Claeson, S
1960-04-15
For most of the thermal neutron flux measurements in reactor R2 cobalt wires will be used. The loading and removal of these wires from the reactor core will be performed by means of a long aluminium tube and electromagnets. After irradiation the wires will be scanned in a semi-automatic device.
Radiation damage in molybdenum and tungsten in high neutron fluxes
International Nuclear Information System (INIS)
Veljkovic, S.; Milasin, N.
1964-01-01
The effects of radiation on molybdenum and tungsten in high neutron fluxes are presented. The changes induced, particularly defects with a high migration activation energy, are analyzed. The correlation of these changes with the basic concepts of radiation damage in solids is considered. An attempt is made to relate the defects studied with the changes in macroscopic properties (author)
Neutron flux in a periodical slab geometry (1960)
International Nuclear Information System (INIS)
Lamare, J. de; Mathelot, P.; Cadhilac, M.
1960-01-01
In the present report, we explain an original method to perform exact calculations of neutron flux in either of two geometries: a slab surrounded by an infinite multiplying medium or a periodical, one dimensional array of two different media. (author) [fr
Transient neutrons flux behaviour in a spherical reactor core
International Nuclear Information System (INIS)
Souza, A.W.A. de.
1978-11-01
This work studies the transient neutron flux in a fast reactor of spherical geometry. The burning of U 235 nuclei is equated and two kinds of reflector were studied. The numeric solutions are then compared with the results for those reflectors. (author) [pt
Spatial distribution of neutron flux for the A-711 neutron generator
International Nuclear Information System (INIS)
Essiet, A. E.; Owolabi, S. A.; Adesanmi, C. A.; Balogun, F. A.
1996-01-01
The spatial distribution of neutron flux for the Kaman sciences A-711 neutron generator recently installed at the Centre for Energy Research and Development (CERD), Ile-Ife Nigeria has been determined. At an operational tube current of 2.0 mA and high voltage power supply (HVPS) of 158 kV, the neutron flux increases from 1.608 ± 0.021*10 8 n/cm 2 s at the top of the irradiated plastic vial to 2.640 ± 0.022*10 8 n/cm 2 s at the centre, and then decreases to 1.943 ± 0.02* 8 n/cm 2 s at the bottom. The flux density is strongly dependent on the diameter of deuteron at the tritium target, and within this range a source strength of 10 8 n/s has been measured for the A-711 neutron generator
Study of the neutron flux distribution in acylindrical reactor
Directory of Open Access Journals (Sweden)
A. Vidal-Ferràndiz
2017-08-01
Full Text Available In the Energy Engineering Degree of the Universitat Politècnica de València, the students attend to the Nuclear Technology course, in which the basic knowledge of this technology is presented. A main objective of this technology is to obtain neutron population distribution inside a reactor core, in order to maintain the fission reaction chain. As this activity cannot be experimentally developed, mathematical modelling is of great importance to achieve such objective. One of the computer laboratories proposed consists in the neutron flux determination analytically and numerically in a cylindrical geometry. The analytical solution makes use of the Bessel functions and is a good example of their applications. Alternatively, a numerical solution based on finite differences is used to obtain an approximate solution of the neutron flux. In this work, different discretizations of the cylindrical geometry are implemented and their results are compared.
Utilization of cold neutron beams at intermediate flux reactors
International Nuclear Information System (INIS)
Clark, D.D.
1992-01-01
With the advent of cold neutron beam (CNB) facilities at U.S. reactors [National Institute of Standards and Technology (NIST) in 1991; Cornell University and the University of Texas at Austin, anticipated in 1992], it is appropriate to reexamine the types of research for which they are likely to be best suited or uniquely suited. With the exception of a small-angle neutron scattering facility at Brookhaven National Laboratory, there has been no prior experience in the United States with such beams, but they have been extensively used at European reactors where cold neutron sources and neutron guides were developed some years age. This paper does not discuss specialized cases such as ultracold neutrons or very high flux facilities such as the Institute Laue-Langevin ractor and the proposed advanced neutron source. Instead, it concentrates on potential utilization of CNBs at intermediate-flux reactors such as at Cornell and Texas, i.e., in the 1-MW range and operated <24 h a day
International Nuclear Information System (INIS)
Ljubenov, V.; Milosevic, M.
2004-11-01
The procedures for the numerical and experimental determination of the neutron flux in the zones with the strong neutron absorption and leakage are described in this paper. Numerical procedure is based on the application of the SCALE-4.4a code system where the Dancoff factors are determined by the VEGA2DAN code. Two main parts of the experimental methodology are measurement of the activity of irradiated foils and determination of the averaged neutron absorption cross-section in the foils by the SCALE-4.4a calculation procedure. The proposed procedures have been applied for the determination of the neutron flux in the internal neutron converter used with the RB reactor core configuration number 114. (author)
Methods and applications in high flux neutron imaging
International Nuclear Information System (INIS)
Ballhausen, H.
2007-01-01
This treatise develops new methods for high flux neutron radiography and high flux neutron tomography and describes some of their applications in actual experiments. Instead of single images, time series can be acquired with short exposure times due to the available high intensity. To best use the increased amount of information, new estimators are proposed, which extract accurate results from the recorded ensembles, even if the individual piece of data is very noisy and in addition severely affected by systematic errors such as an influence of gamma background radiation. The spatial resolution of neutron radiographies, usually limited by beam divergence and inherent resolution of the scintillator, can be significantly increased by scanning the sample with a pinhole-micro-collimator. This technique circumvents any limitations in present detector design and, due to the available high intensity, could be successfully tested. Imaging with scattered neutrons as opposed to conventional total attenuation based imaging determines separately the absorption and scattering cross sections within the sample. For the first time even coherent angle dependent scattering could be visualized space-resolved. New applications of high flux neutron imaging are presented, such as materials engineering experiments on innovative metal joints, time-resolved tomography on multilayer stacks of fuel cells under operation, and others. A new implementation of an algorithm for the algebraic reconstruction of tomography data executes even in case of missing information, such as limited angle tomography, and returns quantitative reconstructions. The setup of the world-leading high flux radiography and tomography facility at the Institut Laue-Langevin is presented. A comprehensive appendix covers the physical and technical foundations of neutron imaging. (orig.)
Fast neutron flux analyzer with real-time digital pulse shape discrimination
Energy Technology Data Exchange (ETDEWEB)
Ivanova, A.A., E-mail: a.a.ivanova@inp.nsk.su [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Zubarev, P.V. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Novosibirsk State Technical University, 630092 Novosibirsk (Russian Federation); Ivanenko, S.V. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Khilchenko, A.D. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Novosibirsk State Technical University, 630092 Novosibirsk (Russian Federation); Kotelnikov, A.I. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Polosatkin, S.V. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Novosibirsk State Technical University, 630092 Novosibirsk (Russian Federation); Novosibirsk State University, 630090 Novosibirsk (Russian Federation); Puryga, E.A.; Shvyrev, V.G. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Novosibirsk State Technical University, 630092 Novosibirsk (Russian Federation); Sulyaev, Yu.S. [Budker Institute of Nuclear Physics SB RAS, 630090 Novosibirsk (Russian Federation); Novosibirsk State University, 630090 Novosibirsk (Russian Federation)
2016-08-11
Investigation of subthermonuclear plasma confinement and heating in magnetic fusion devices such as GOL–3 and GDT at the Budker Institute (Novosibirsk, Russia) requires sophisticated equipment for neutron-, gamma- diagnostics and upgrading data acquisition systems with online data processing. Measurement of fast neutron flux with stilbene scintillation detectors raised the problem of discrimination of the neutrons (n) from background cosmic particles (muons) and neutron-induced gamma rays (γ). This paper describes a fast neutron flux analyzer with real-time digital pulse-shape discrimination (DPSD) algorithm FPGA-implemented for the GOL–3 and GDT devices. This analyzer was tested and calibrated with the help of {sup 137}Cs and {sup 252}Cf radiation sources. The Figures of Merit (FOM) calculated for different energy cuts are presented. - Highlights: • Electronic equipment for measurement of fast neutron flux with stilbene scintillator is presented. • FPGA-implemented digital pulse-shape discrimination algorithm by charge comparison method is shown. • Calibration of analyzer was carried out with {sup 137}Cs and {sup 252}Cf. • Figures of Merit (FOM) values for energy cuts from 1/8 Cs to 2 Cs are from 1.264 to 2.34 respectively.
Energy Technology Data Exchange (ETDEWEB)
Obradovic, D; Bosevski, T [The Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Yugoslavia)
1965-07-01
Using semiconductor detectors for measuring the neutron flux distribution is considered suitable and faster than using activation foils. Results of radial neutron flux distribution obtained by semiconductor detectors are presented.
LANSCE steady state unperturbed thermal neutron fluxes at 100 μA
International Nuclear Information System (INIS)
Russell, G.J.
1989-01-01
The ''maximum'' unperturbed, steady state thermal neutron flux for LANSCE is calculated to be 2 /times/ 10 13 n/cm 2 -s for 100 μA of 800-MeV protons. This LANSCE neutron flux is a comparable entity to a steady state reactor thermal neutron flux. LANSCE perturbed steady state thermal neutron fluxes have also been calculated. Because LANSCE is a pulsed neutron source, much higher ''peak'' (in time) neutron fluxes can be generated than at a steady state reactor source. 5 refs., 5 figs
Green's function approach to neutron flux discontinuities
International Nuclear Information System (INIS)
Saad, E.A.; El-Wakil, S.A.
1980-01-01
The present work is devoted to the presentation of analytical method for the calculation of elastically and inelastically slowed down neutrons in an infinite non-absorbing medium. On the basis of the central limit theory (CLT) and the integral transform technique the slowing down equation including inelastic scattering, in terms of the Green function of elastic scattering, is solved. The Green function is decomposed according to the number of collisions. Placzec discontinuity associated with elastic scattering in addition to two discontinuities due to inelastic scattering are investigated. Numerical calculations for Fe 56 show that the elastic discontinuity produces about 41.8% change in the collision density whilst the ratio of the inelastic collision density discontinuity at qsub(o)sup(+) to the Placzec discontinuity at usub(o) + 1n 1/oc gives 55.7 percent change. (author)
Monte Carlo method for neutron transport problems
International Nuclear Information System (INIS)
Asaoka, Takumi
1977-01-01
Some methods for decreasing variances in Monte Carlo neutron transport calculations are presented together with the results of sample calculations. A general purpose neutron transport Monte Carlo code ''MORSE'' was used for the purpose. The first method discussed in this report is the method of statistical estimation. As an example of this method, the application of the coarse-mesh rebalance acceleration method to the criticality calculation of a cylindrical fast reactor is presented. Effective multiplication factor and its standard deviation are presented as a function of the number of histories and comparisons are made between the coarse-mesh rebalance method and the standard method. Five-group neutron fluxes at core center are also compared with the result of S4 calculation. The second method is the method of correlated sampling. This method was applied to the perturbation calculation of control rod worths in a fast critical assembly (FCA-V-3) Two methods of sampling (similar flight paths and identical flight paths) are tested and compared with experimental results. For every cases the experimental value lies within the standard deviation of the Monte Carlo calculations. The third method is the importance sampling. In this report a biased selection of particle flight directions discussed. This method was applied to the flux calculation in a spherical fast neutron system surrounded by a 10.16 cm iron reflector. Result-direction biasing, path-length stretching, and no biasing are compared with S8 calculation. (Aoki, K.)
Criticality experiments to provide benchmark data on neutron flux traps
International Nuclear Information System (INIS)
Bierman, S.R.
1988-06-01
The experimental measurements covered by this report were designed to provide benchmark type data on water moderated LWR type fuel arrays containing neutron flux traps. The experiments were performed at the US Department of Energy Hanford Critical Mass Laboratory, operated by Pacific Northwest Laboratory. The experimental assemblies consisted of 2 /times/ 2 arrays of 4.31 wt % 235 U enriched UO 2 fuel rods, uniformly arranged in water on a 1.891 cm square center-to-center spacing. Neutron flux traps were created between the fuel units using metal plates containing varying amounts of boron. Measurements were made to determine the effect that boron loading and distance between the fuel and flux trap had on the amount of fuel required for criticality. Also, measurements were made, using the pulse neutron source technique, to determine the effect of boron loading on the effective neutron multiplications constant. On two assemblies, reaction rate measurements were made using solid state track recorders to determine absolute fission rates in 235 U and 238 U. 14 refs., 12 figs., 7 tabs
Fast neutron flux and intracranial dose distribution at a neutron irradiation facility
International Nuclear Information System (INIS)
Matsumoto, Tetsuo; Aizawa, Otohiko; Nozaki, Tetsuya
1981-01-01
A head phantom filled with water was used to measure the fast neutron flux using 115 In(n, n')sup(115m)In and 103 Rh(n, n')sup(103m)Rh reactions. γ-ray from sup(115m)In and x-ray from sup(103m)Rh were detected by a Ge(Li) and a Na(Tl)I counter, respectively. TLD was used to investigate the γ-dose rate distribution inside the phantom. Flux of fast neutron inside the phantom was about 1 x 10 6 n/cm 2 sec, which was 3 order smaller than that of thermal neutron. The fast neutron flux decreased to 1/10 at 15 cm depth, and γ-dose rate was about 200 R/h at 100 kW inside the phantom. Total dose at the surface was 350 rad/h, to which, fast neutrons contributed more than γ-rays. The rate of fast neutron dose was about 10% of thermal neutron's in Kerma dose unit (rad), however, the rate was highly dependent on RBE value. (Nakanishi, T.)
Real time neutron flux monitoring using Rh self powered neutron detector
International Nuclear Information System (INIS)
Juna, Byung Jin; Lee, Byung Chul; Park, Sang Jun; Jung, Hoan Sung
2012-01-01
Rhodium (Rh) self powered neutron detectors (SPNDs) are widely used for on line monitoring of local neutron flux. Its signal is slower than the actual variation of neutron flux owing to a delayed β decay of the Rh activation product, but real time monitoring is possible by solving equations between the neutron reaction rate in the detector and its signal. While the measuring system is highly reliable, the accuracy depends on the method solving the equations and accuracy of the parameters in the equations. The uncertain parameters are the contribution of gamma rays to the signal, and the branching ratios of Rh 104 and Rh 104m after the neutron absorption of Rh 103. Real time neutron flux monitoring using Rh SPNDs has been quite successful for neutron transmutation doping (NTD) at HANARO. We revisited the initial data used for the verification of a real time monitoring system, to refine algorithm for a better solution and to check the parameters for correctness. As a result, we suggest an effective way to determine the prompt parameter
Real time neutron flux monitoring using Rh self powered neutron detector
Energy Technology Data Exchange (ETDEWEB)
Juna, Byung Jin; Lee, Byung Chul; Park, Sang Jun; Jung, Hoan Sung [KAERI, Daejeon (Korea, Republic of)
2012-10-15
Rhodium (Rh) self powered neutron detectors (SPNDs) are widely used for on line monitoring of local neutron flux. Its signal is slower than the actual variation of neutron flux owing to a delayed {beta} decay of the Rh activation product, but real time monitoring is possible by solving equations between the neutron reaction rate in the detector and its signal. While the measuring system is highly reliable, the accuracy depends on the method solving the equations and accuracy of the parameters in the equations. The uncertain parameters are the contribution of gamma rays to the signal, and the branching ratios of Rh 104 and Rh 104m after the neutron absorption of Rh 103. Real time neutron flux monitoring using Rh SPNDs has been quite successful for neutron transmutation doping (NTD) at HANARO. We revisited the initial data used for the verification of a real time monitoring system, to refine algorithm for a better solution and to check the parameters for correctness. As a result, we suggest an effective way to determine the prompt parameter.
Analysis of neutron flux measurement systems using statistical functions
International Nuclear Information System (INIS)
Pontes, Eduardo Winston
1997-01-01
This work develops an integrated analysis for neutron flux measurement systems using the concepts of cumulants and spectra. Its major contribution is the generalization of Campbell's theorem in the form of spectra in the frequency domain, and its application to the analysis of neutron flux measurement systems. Campbell's theorem, in its generalized form, constitutes an important tool, not only to find the nth-order frequency spectra of the radiation detector, but also in the system analysis. The radiation detector, an ionization chamber for neutrons, is modeled for cylindrical, plane and spherical geometries. The detector current pulses are characterized by a vector of random parameters, and the associated charges, statistical moments and frequency spectra of the resulting current are calculated. A computer program is developed for application of the proposed methodology. In order for the analysis to integrate the associated electronics, the signal processor is studied, considering analog and digital configurations. The analysis is unified by developing the concept of equivalent systems that can be used to describe the cumulants and spectra in analog or digital systems. The noise in the signal processor input stage is analysed in terms of second order spectrum. Mathematical expressions are presented for cumulants and spectra up to fourth order, for important cases of filter positioning relative to detector spectra. Unbiased conventional estimators for cumulants are used, and, to evaluate systems precision and response time, expressions are developed for their variances. Finally, some possibilities for obtaining neutron radiation flux as a function of cumulants are discussed. In summary, this work proposes some analysis tools which make possible important decisions in the design of better neutron flux measurement systems. (author)
A method for neutron dosimetry in ultrahigh flux environments
International Nuclear Information System (INIS)
Ougouag, A.M.; Wemple, C.A.; Rogers, J.W.
1996-01-01
A method for neutron dosimetry in ultrahigh flux environments is developed, and devices embodying it are proposed and simulated using a Monte Carlo code. The new approach no longer assumes a linear relationship between the fluence and the activity of the nuclides formed by irradiation. It accounts for depletion of the original ''foil'' material and for decay and depletion of the formed nuclides. In facilities where very high fluences are possible, the fluences inferred by activity measurements may be ambiguous. A method for resolving these ambiguities is also proposed and simulated. The new method and proposed devices should make possible the use of materials not traditionally considered desirable for neutron activation dosimetry
Gas loop - continuous measurement of thermal and fast neutron fluxes
International Nuclear Information System (INIS)
Droulers, Y.; Pleyber, G.; Sciers, P.; Maurin, G.
1964-01-01
The measurement method described in this report can be applied both to thermal and fast neutron fluxes. A description is given of two practical applications in each of these two domains. This method is particularly suitable for measurements carried out on 'loop' type equipment. The measurement of the relative flux variations are carried out with an accuracy of 5 per cent. The choice of the shape of the gas circuit leaves a considerable amount of liberty for the adaptation of the measurement circuit to the experimental conditions. (authors) [fr
International Nuclear Information System (INIS)
2007-01-01
1 - Description of program or function: VIRGIN calculates un-collided flux and reactions due to transmission of a mono-directional beam of neutrons through any thickness of material. In order to simulate an experimental measurement the results are given as integrals over energy tally groups (as opposed to point-wise in energy). IAEA0932/10: This version include the updates up to January 30, 2007. Changes in ENDF/B-VII Format and procedures, as well as the evaluations themselves, make it impossible for versions of the ENDF/B pre-processing codes earlier than PREPRO 2007 (2007 Version) to accurately process current ENDF/B-VII evaluations. The present code can handle all existing ENDF/B-VI evaluations through release 8, which will be the last release of ENDF/B-VI. Modifications from previous versions: Virgin VERS. 2007-1 (Jan. 2007): checked against all ENDF/B-VII; increased in-core page size from 60,000 to 240,000. 2 - Method of solution: By taking the ratio of reactions to flux in each group an equivalent spatially dependent group averaged cross section is calculated. 3 - Restrictions on the complexity of the problem: The evaluated data must be in the ENDF/B format. However it must be linear-linear interpolable in energy-cross section between tabulated points. Since only cross sections (file 3) are used, this program will work on any version of ENDF/B
Finite element based composite solution for neutron transport problems
International Nuclear Information System (INIS)
Mirza, A.N.; Mirza, N.M.
1995-01-01
A finite element treatment for solving neutron transport problems is presented. The employs region-wise discontinuous finite elements for the spatial representation of the neutron angular flux, while spherical harmonics are used for directional dependence. Composite solutions has been obtained by using different orders of angular approximations in different parts of a system. The method has been successfully implemented for one dimensional slab and two dimensional rectangular geometry problems. An overall reduction in the number of nodal coefficients (more than 60% in some cases as compared to conventional schemes) has been achieved without loss of accuracy with better utilization of computational resources. The method also provides an efficient way of handling physically difficult situations such as treatment of voids in duct problems and sharply changing angular flux. It is observed that a great wealth of information about the spatial and directional dependence of the angular flux is obtained much more quickly as compared to Monte Carlo method, where most of the information in restricted to the locality of immediate interest. (author)
In-core neutron flux measurements at PARR using self powered neutron detector
International Nuclear Information System (INIS)
Hussain, A.; Ansari, S.A.
1989-10-01
This report describes experimental reactor physics measure ments at PARR using the in-core neutron detectors. Rhodium self powered neutron detectors (SPND) were used in the PARR core and several measurements were made aimed at detector calibration, response time determination and neutron flux measurements. The detectors were calibrated at low power using gold foils and full power by the thermal channel. Based on this calibration it was observed that the detector response remains almost linear throughout the power range. The self powered detectors were used for on-line determination of absolute neutron flux in the core as well as the spatial distribution of neutron flux or reactor power. The experimental, axial and horizontal flux mapping results at certain locations in the core are presented. The total response time of rhodium detector was experimentally determined to be about 5 minutes, which agree well with the theoretical results. Because of longer response time of SPND of the detectors it is not possible to use them in the reactor protection system. (author). 10 figs
Modeling the Effects of Meteorological Conditions on the Neutron Flux
2017-05-22
about 2% between day and night on a given day [2]. In the 1960s, the launch of satellites allowed scientists to measure the sun’s cosmic rays outside...hour, a 20% variation, over five months of data collection with large variation between days . Meteorological data were collected with two commercially...contributes to the formation of the neutron flux. To account for the earth’s magnetic field, scientists have done extensive three-dimensional analysis
Optimization of neutron flux distribution in Isotope Production Reactor
International Nuclear Information System (INIS)
Valladares, G.L.
1988-01-01
In order to optimize the thermal neutrons flux distribution in a Radioisotope Production and Research Reactor, the influence of two reactor parameters was studied, namely the Vmod / Vcomb ratio and the core volume. The reactor core is built with uranium oxide pellets (UO 2 ) mounted in rod clusters, with an enrichment level of ∼3 %, similar to LIGHT WATER POWER REATOR (LWR) fuel elements. (author) [pt
Analog model for analysis of spatial instability of neutron flux
International Nuclear Information System (INIS)
Radanovic, Lj.
1964-12-01
The objective of this task was to develop a model for analysing spatial instability of the neutron flux and defining the optimum number and position of regulating rods. The developed model enables calculation of higher harmonics to be taken into account for each type of reactor, to define zones for regulation rods, position and number of points for detecting reactor state, and number and position of the regulating rods
FNR demonstration experiments Part II: Subcadmium neutron flux measurements
International Nuclear Information System (INIS)
Wehe, D.K.; King, J.S.
1983-01-01
The FNR HEU-LEU Demonstration Experiments include a comprehensive set of experiments to identify and quantify significant operational differences between two nuclear fuel enrichments. One aspect of these measurements, the subcadmium flux profiling, is the subject of this paper. The flux profiling effort has been accomplished through foil and wire activations, and by rhodium self-powered neutron detector (SPND) mappings. Within the experimental limitations discussed, the program to measure subcadmium flux profiles, lead to the following conclusions: (1) Replacement of a single fresh HEU element by a fresh LEU element at the center of an equilibrium HEU core produces a local flux depression. The ratio of HEU to LEU local flux is 1.19 ± .036, which is, well within experimental uncertainty, equal to the inverse of the U-235 masses for the two elements. (2) Whole core replacement of a large 38 element equilibrium HEU core by a fresh or nearly unburned LEU core reduces the core flux and raises the flux in both D 2 O and H 2 O reflectors. The reduction in the central core region is 40% to 10.0% for the small fresh 29 element LEU core, and 16% to 18% for a 31 element LEU core 482) with low average burnup 2 O reflector fluxes relative to core fluxes as measured by SPND with a fixed value of sensitivity, are in gross disagreement with the same flux ratios measured by Fe and Rh wire activations. Space dependent refinements of S are calculated to give some improvement in the discrepancy but the major part of the correction remains to be resolved
Simulation of neutron fluxes around the W7-X Stellarator
Energy Technology Data Exchange (ETDEWEB)
Andersson, Jenny
1999-12-01
A new fusion experiment, the WENDELSTEIN 7-X Stellarator (W7-X), will be undertaken in Greifswald in Germany. Measurements of the neutron flux will provide information on fusion reaction rates and possibly also on ion temperatures as function of time. For this purpose moderating neutron counters will be designed, tested, calibrated and eventually used at W7-X. Extensive Monte-Carlo simulations have been performed in order to select the most suitable detector and moderator combination with a flat response function and highest achievable efficiency. Different detector configurations with different moderating materials have been tried out, showing that a 32 cm thick graphite moderating BF{sub 3} -counter gives the desired flat response and sufficient efficiency. Neutron spectra calculations have been made for different torus models and the influence of floor, walls and ceiling (i.e. reactor hall) have been investigated. Presented results suggest that a more detailed torus model significantly reduces the number of neutron counts at the detector. Calculations including the reactor hall indicate a tendency of shifting the neutron spectra towards the thermal region. The main part of the scattered neutrons are back-scattered from the floor. Finally, calculations on the graphite moderating BF{sub 3} -counter in the detailed torus environment were performed in order to assess the absolute response function under the influence of the reactor hall. The results show that the detector count rate will increase by only 5-7 % when the reactor hall is taken into account. With a stellarator generating 10{sup 12} to 10{sup 16} neutrons per second the detector count rate will be 2x10{sup 5} to 2x10{sup 9} neutrons per second.
Simulation of neutron fluxes around the W7-X Stellarator
International Nuclear Information System (INIS)
Andersson, Jenny
1999-12-01
A new fusion experiment, the WENDELSTEIN 7-X Stellarator (W7-X), will be undertaken in Greifswald in Germany. Measurements of the neutron flux will provide information on fusion reaction rates and possibly also on ion temperatures as function of time. For this purpose moderating neutron counters will be designed, tested, calibrated and eventually used at W7-X. Extensive Monte-Carlo simulations have been performed in order to select the most suitable detector and moderator combination with a flat response function and highest achievable efficiency. Different detector configurations with different moderating materials have been tried out, showing that a 32 cm thick graphite moderating BF 3 -counter gives the desired flat response and sufficient efficiency. Neutron spectra calculations have been made for different torus models and the influence of floor, walls and ceiling (i.e. reactor hall) have been investigated. Presented results suggest that a more detailed torus model significantly reduces the number of neutron counts at the detector. Calculations including the reactor hall indicate a tendency of shifting the neutron spectra towards the thermal region. The main part of the scattered neutrons are back-scattered from the floor. Finally, calculations on the graphite moderating BF 3 -counter in the detailed torus environment were performed in order to assess the absolute response function under the influence of the reactor hall. The results show that the detector count rate will increase by only 5-7 % when the reactor hall is taken into account. With a stellarator generating 10 12 to 10 16 neutrons per second the detector count rate will be 2x10 5 to 2x10 9 neutrons per second
Neutron flux determination at the IPR-R1 Triga Mark I neutron beam extractor
International Nuclear Information System (INIS)
Zangirolami, Dante Marco; Maretti Junior, Fausto; Ferreira, Andrea Vidal
2009-01-01
The IPR-R1 Triga Mark I Reactor located at the CDTN/CNEN, Belo Horizonte, Brazil, has been operating since November of 1960. In this work, measurements of thermal and epithermal neutron flux along the IPR-R1 neutron beam extractor were performed by neutron activation of reference materials using the two foils method. The obtained results were compared with results from two previous works: an experimental measurement done in a previous reactor core configuration and a numerical work made by Monte Carlo simulation using the actual reactor core configuration. The main purpose of this work is to update the measured data to the actual reactor core configuration. (author)
Evaluation of thermal margin during BWR neutron flux oscillation
International Nuclear Information System (INIS)
Takeuchi, Yutaka; Takigawa, Yukio; Chuman, Kazuto; Ebata, Shigeo
1992-01-01
Fuel integrity is very important, from the view point of nuclear power plant safety. Recently, neutron flux oscillations were observed at several BWR plants. The present paper describes the evaluations of the thermal margin during BWR neutron flux oscillations, using a three-dimensional transient code. The thermal margin is evaluated as MCPR (minimum critical power ratio). The LaSalle-2 event was simulated and the MCPR during the event was evaluated. It was a core-wide oscillation, at which a large neutron flux oscillation amplitude was observed. The results indicate that the MCPR had a sufficient margin with regard to the design limit. A regional oscillation mode, which is different from a core-wide oscillation, was simulated and the MCPR response was compared with that for the LaSalle-2 event. The MCPR decrement is greater in the regional oscillation, than in the core wide -oscillation, because of the sensitivity difference in a flow-to-power gain. A study was carried out about regional oscillation detectability, from the MCPR response view point. Even in a hypothetically severe case, the regional oscillation is detectable by LPRM signals. (author)
International Nuclear Information System (INIS)
Schuerrer, F.
1980-01-01
For characterizing heterogene configurations of pebble-bed reactors the fine structure of the flux distribution as well as the determination of the macroscopic neutronphysical quantities are of interest. When calculating system parameters of Wigner-Seitz-cells the usual codes for neutron spectra calculation always neglect the modulation of the neutron flux by the influence of neighbouring spheres. To judge the error arising from that procedure it is necessary to determinate the flux distribution in the surrounding of a spherical fuel element. In the present paper an approximation method to calculate the flux distribution in the two-sphere model is developed. This method is based on the exactly solvable problem of the flux determination of a point source of neutrons in an infinite medium, which contains a spherical perturbation zone eccentric to the point source. An iteration method allows by superposing secondary fields and alternately satisfying the conditions of continuity on the surface of each of the two fuel elements to advance to continually improving approximations. (orig.) 891 RW/orig. 892 CKA [de
Fast neutron flux in heavy water reactors; Flux de neutrons rapides dans les piles a eau lourde
Energy Technology Data Exchange (ETDEWEB)
Brisbois, J; Katz, S [Commissariat a l' Energie Atomique, Centre d' Etudes Nucleaires de Fontenay-aux-Roses, 92 (France)
1966-07-01
The possibility of calculating the fast neutron flux in a natural uranium-heavy water lattice by superposition of the individual contributions of the different fuel elements was verified using a one-dimension Monte-Carlo code. The results obtained are in good agreement with experimental measurements done in the core and reflector of the reactor AQUILON. (author) [French] La possibilite de calculer le flux de neutrons rapides dans un reseau d'uranium naturel a eau lourde par superposition des apports des divers barreaux, a ete verifiee en utilisant un code Monte-Carlo monodimensionel. Les resultats obtenus concordent avec des mesures experimentales effectuees dans le coeur et reacteur de la pile Aquilon. (auteurs)
Fusion neutron yield and flux calculation on HT-7 superconducting tokamak
International Nuclear Information System (INIS)
Fu Yanzhang; Zhu Yubao; Chen Juequan
2006-01-01
Neutron yield and flux have been numerically estimated on HT-7 tokamak. The total fusion neutron yield and neutron flux distribution on different positions and azimuth angles of the device are presented. Analyses on the errors induced by ion temperature and density distribution factors are given in detail. The results of the calculations provide a useful database for neutron diagnostics and neutron radiation protection. (authors)
Neutron scattering at the high-flux isotope reactor
International Nuclear Information System (INIS)
Cable, J.W. Chakoumakos, B.C.; Dai, P.
1995-01-01
The title facilities offer the brightest source of neutrons in the national user program. Neutron scattering experiments probe the structure and dynamics of materials in unique and complementary ways as compared to x-ray scattering methods and provide fundamental data on materials of interest to solid state physicists, chemists, biologists, polymer scientists, colloid scientists, mineralogists, and metallurgists. Instrumentation at the High- Flux Isotope Reactor includes triple-axis spectrometers for inelastic scattering experiments, a single-crystal four diffractometer for crystal structural studies, a high-resolution powder diffractometer for nuclear and magnetic structure studies, a wide-angle diffractometer for dynamic powder studies and measurements of diffuse scattering in crystals, a small-angle neutron scattering (SANS) instrument used primarily to study structure-function relationships in polymers and biological macromolecules, a neutron reflectometer for studies of surface and thin-film structures, and residual stress instrumentation for determining macro- and micro-stresses in structural metals and ceramics. Research highlights of these areas will illustrate the current state of neutron science to study the physical properties of materials
High Flux Isotope Reactor cold neutron source reference design concept
International Nuclear Information System (INIS)
Selby, D.L.; Lucas, A.T.; Hyman, C.R.
1998-05-01
In February 1995, Oak Ridge National Laboratory's (ORNL's) deputy director formed a group to examine the need for upgrades to the High Flux Isotope Reactor (HFIR) system in light of the cancellation of the Advanced neutron Source Project. One of the major findings of this study was that there was an immediate need for the installation of a cold neutron source facility in the HFIR complex. In May 1995, a team was formed to examine the feasibility of retrofitting a liquid hydrogen (LH 2 ) cold source facility into an existing HFIR beam tube. The results of this feasibility study indicated that the most practical location for such a cold source was the HB-4 beam tube. This location provides a potential flux environment higher than the Institut Laue-Langevin (ILL) vertical cold source and maximizes the space available for a future cold neutron guide hall expansion. It was determined that this cold neutron beam would be comparable, in cold neutron brightness, to the best facilities in the world, and a decision was made to complete a preconceptual design study with the intention of proceeding with an activity to install a working LH 2 cold source in the HFIR HB-4 beam tube. During the development of the reference design the liquid hydrogen concept was changed to a supercritical hydrogen system for a number of reasons. This report documents the reference supercritical hydrogen design and its performance. The cold source project has been divided into four phases: (1) preconceptual, (2) conceptual design and testing, (3) detailed design and procurement, and (4) installation and operation. This report marks the conclusion of the conceptual design phase and establishes the baseline reference concept
High Flux Isotope Reactor cold neutron source reference design concept
Energy Technology Data Exchange (ETDEWEB)
Selby, D.L.; Lucas, A.T.; Hyman, C.R. [and others
1998-05-01
In February 1995, Oak Ridge National Laboratory`s (ORNL`s) deputy director formed a group to examine the need for upgrades to the High Flux Isotope Reactor (HFIR) system in light of the cancellation of the Advanced neutron Source Project. One of the major findings of this study was that there was an immediate need for the installation of a cold neutron source facility in the HFIR complex. In May 1995, a team was formed to examine the feasibility of retrofitting a liquid hydrogen (LH{sub 2}) cold source facility into an existing HFIR beam tube. The results of this feasibility study indicated that the most practical location for such a cold source was the HB-4 beam tube. This location provides a potential flux environment higher than the Institut Laue-Langevin (ILL) vertical cold source and maximizes the space available for a future cold neutron guide hall expansion. It was determined that this cold neutron beam would be comparable, in cold neutron brightness, to the best facilities in the world, and a decision was made to complete a preconceptual design study with the intention of proceeding with an activity to install a working LH{sub 2} cold source in the HFIR HB-4 beam tube. During the development of the reference design the liquid hydrogen concept was changed to a supercritical hydrogen system for a number of reasons. This report documents the reference supercritical hydrogen design and its performance. The cold source project has been divided into four phases: (1) preconceptual, (2) conceptual design and testing, (3) detailed design and procurement, and (4) installation and operation. This report marks the conclusion of the conceptual design phase and establishes the baseline reference concept.
Online In-Core Thermal Neutron Flux Measurement for the Validation of Computational Methods
International Nuclear Information System (INIS)
Mohamad Hairie Rabir; Muhammad Rawi Mohamed Zin; Yahya Ismail
2016-01-01
In order to verify and validate the computational methods for neutron flux calculation in RTP calculations, a series of thermal neutron flux measurement has been performed. The Self Powered Neutron Detector (SPND) was used to measure thermal neutron flux to verify the calculated neutron flux distribution in the TRIGA reactor. Measurements results obtained online for different power level of the reactor. The experimental results were compared to the calculations performed with Monte Carlo code MCNP using detailed geometrical model of the reactor. The calculated and measured thermal neutron flux in the core are in very good agreement indicating that the material and geometrical properties of the reactor core are modelled well. In conclusion one can state that our computational model describes very well the neutron flux distribution in the reactor core. Since the computational model properly describes the reactor core it can be used for calculations of reactor core parameters and for optimization of RTP utilization. (author)
On Line Neutron Flux Mapping in Fuel Coolant Channels of a Research Reactor
International Nuclear Information System (INIS)
Barbot, Loic; Domergue, Christophe; Villard, Jean-Francois; Destouches, Christophe; Braoudakis, George; Wassink, David; Sinclair, Bradley; Osborn, John-C.; Wu, Huayou; Blandin, C.; Thevenin, Mathieu; Corre, Gwenole; Normand, Stephane
2013-06-01
This work deals with the on-line neutron flux mapping of the OPAL research reactor. A specific irradiation device has been set up to investigate fuel coolant channels using subminiature fission chambers to get thermal neutron flux profiles. Experimental results are compared to first neutronic calculations and show good agreement (C/E ∼0.97). (authors)
Bowman, Charles D.
1992-01-01
Apparatus for nuclear transmutation and power production using an intense accelerator-generated thermal neutron flux. High thermal neutron fluxes generated from the action of a high power proton accelerator on a spallation target allows the efficient burn-up of higher actinide nuclear waste by a two-step process. Additionally, rapid burn-up of fission product waste for nuclides having small thermal neutron cross sections, and the practicality of small material inventories while achieving significant throughput derive from employment of such high fluxes. Several nuclear technology problems are addressed including 1. nuclear energy production without a waste stream requiring storage on a geological timescale, 2. the burn-up of defense and commercial nuclear waste, and 3. the production of defense nuclear material. The apparatus includes an accelerator, a target for neutron production surrounded by a blanket region for transmutation, a turbine for electric power production, and a chemical processing facility. In all applications, the accelerator power may be generated internally from fission and the waste produced thereby is transmuted internally so that waste management might not be required beyond the human lifespan.
Bowman, C.D.
1992-11-03
Apparatus for nuclear transmutation and power production using an intense accelerator-generated thermal neutron flux. High thermal neutron fluxes generated from the action of a high power proton accelerator on a spallation target allows the efficient burn-up of higher actinide nuclear waste by a two-step process. Additionally, rapid burn-up of fission product waste for nuclides having small thermal neutron cross sections, and the practicality of small material inventories while achieving significant throughput derive from employment of such high fluxes. Several nuclear technology problems are addressed including 1. nuclear energy production without a waste stream requiring storage on a geological timescale, 2. the burn-up of defense and commercial nuclear waste, and 3. the production of defense nuclear material. The apparatus includes an accelerator, a target for neutron production surrounded by a blanket region for transmutation, a turbine for electric power production, and a chemical processing facility. In all applications, the accelerator power may be generated internally from fission and the waste produced thereby is transmuted internally so that waste management might not be required beyond the human lifespan.
Upgrade of the compact neutron spectrometer for high flux environments
Osipenko, M.; Bellucci, A.; Ceriale, V.; Corsini, D.; Gariano, G.; Gatti, F.; Girolami, M.; Minutoli, S.; Panza, F.; Pillon, M.; Ripani, M.; Trucchi, D. M.
2018-03-01
In this paper new version of the 6Li-based neutron spectrometer for high flux environments is described. The new spectrometer was built with commercial single crystal Chemical Vapour Deposition diamonds of electronic grade. These crystals feature better charge collection as well as higher radiation hardness. New metal contacts approaching ohmic conditions were deposited on the diamonds suppressing build-up of space charge observed in the previous prototypes. New passive preamplification of the signal at detector side was implemented to improve its resolution. This preamplification is based on the RF transformer not sensitive to high neutron flux. The compact mechanical design allowed to reduce detector size to a tube of 1 cm diameter and 13 cm long. The spectrometer was tested in the thermal column of TRIGA reactor and at the DD neutron generator. The test results indicate an energy resolution of 300 keV (FWHM), reduced to 72 keV (RMS) excluding energy loss, and coincidence timing resolution of 160 ps (FWHM). The measured data are in agreement with Geant4 simulations except for larger energy loss tail presumably related to imperfections of metal contacts and glue expansion.
The Neutron-Gamma Pulse Shape Discrimination Method for Neutron Flux Detection in the ITER
International Nuclear Information System (INIS)
Xu Xiufeng; Li Shiping; Cao Hongrui; Yin Zejie; Yuan Guoliang; Yang Qingwei
2013-01-01
The neutron flux monitor (NFM), as a significant diagnostic system in the International Thermonuclear Experimental Reactor (ITER), will play an important role in the readings of a series of key parameters in the fusion reaction process. As the core of the main electronic system of the NFM, the neutron-gamma pulse shape discrimination (n-γ PSD) can distinguish the neutron pulse from the gamma pulse and other disturbing pulses according to the thresholds of the rising time and the amplitude pre-installed on the board, the double timing point CFD method is used to get the rising time of the pulse. The n-γ PSD can provide an accurate neutron count. (magnetically confined plasma)
Flux depression and the absolute measurement of the thermal neutron flux density
International Nuclear Information System (INIS)
Bensch, Friedrich.
1977-01-01
The thermal neutron flux depression in a diffusing medium by an absorbing foil has been treated in numerous papers. The results are re-examined in an attempt to find a uniform and physically meaningful representation of the 'activation correction'. This quantity can be split up into a combination of probabilities. Thus, it is possible to determine the activation correction for any moderator and foil material. Measurements confirm the utility of the concepts introduced
A numerical approach to the time dependent neutron flux using the Laplace transform technique
International Nuclear Information System (INIS)
El-Demerdash, A; Beynon, T.D.
1979-01-01
In this study a time dependent transport problem in which an isotopic neutron source emits a pulse of neutrons into a finite sphere has been solved by a numerical Laplace transform technique. The object has been to investigate the time behaviour of the neutron field in the moderators at times shortly after the neutron source initiation, that is in the nanosecond time period. The basis of the solution is a numercial evaluation of the Laplace transform of the flux in the linear Boltzmann equation with the use of a modified version of a steady state energy multi-group spatially dependent code. The explicit or direct inversion of the Laplace transformed flux is complicated to be solved numerically due to the ill-conditioned matrix obtained. The suggested method of solutions depends on choice of a function that satisfies the physical condition known from the neutron behaviour and that has a Laplace inversion which is analytically amenable. By employing a least square fitting procedure the function is modified in order to minimize the error in the Laplace transformed values and hence in the time dependent solution. This method has been applied satisfactorily in comparison to analytical and experimental results
International Nuclear Information System (INIS)
Bozic, M.; Zagar, T.; Ravnik, M.
2003-01-01
Neutron fluxes in different spatial locations in biological shield are obtained with TORT code (TORT-Three Dimensional Oak Ridge Discrete Ordinates Neutron/Photon Transport Code). Libraries used with TORT code were BUGLE-96 library (coupled library with 47 neutron groups and 20 gamma groups) and VITAMIN-B6 library (coupled library with 199 neutron groups and 42 gamma groups). BUGLE-96 library is derived from VITAMIN-B6 library. 2-D and 3-D models for homogeneous type of problem (without inserted beam port 4) and problem with asymmetry (non-homogeneous problem; inserted beam port 4, filled with different materials) were of interest for neutron flux calculation. The main purpose is to verify the possibility for using 2-D approximation model instead of large 3-D model in some calculations. Another purpose of this paper was to compare neutron spectral constants obtained from neutron fluxes (3-D model) determined with smaller BUGLE-96 library with new constants obtained from fluxes calculated with bigger VITAMIN-B6 library. These neutron spectral constants are used in isotopic calculation with SCALE code package (ORIGEN-S). In past only neutron spectral constants determined by neutron fluxes from BUGLE-96 library were used. Experimental results used for isotopic composition comparison are available from irradiation experiment with selected type of concrete and other materials in beam port 4 (irradiation channel 4) in TRIGA Mark II reactor. These experimental results were used as a benchmark in this paper. (author)
MCNP and visualization of neutron flux and power distributions
International Nuclear Information System (INIS)
Snoj, L.; Lengar, I.; Zerovnik, G.; Ravnik, M.
2009-01-01
The visualization of neutron flux and power distributions in two nuclear reactors (TRIG A type research reactor and typical PWR) and one thermonuclear reactor (tokamak type) are treated in the paper. The distributions are calculated with MCNP computer code and presented using Amira and Voxler software. The results in the form of figures are presented in the paper together with comments qualitatively explaining the figures. The remembrance of most of the people is better, if they visualize a process. Therefore a representation of the reactor and neutron transport parameters is a convenient modern educational tool for (nuclear power plant) operators, nuclear engineers, students and specialists involved in reactor operation and design. Visualization is applicable not only in education and training, but also as a tool for core and irradiation planning. (authors)
Evaluation of neutron flux in the Pool Critical Assembly
International Nuclear Information System (INIS)
Lippincott, E.P.; Ruddy, F.H.; Gold, R.; Kellogg, L.S.; Roberts, J.H.
1984-09-01
A recently completed series of experiments in the Pool Critical Assembly (PCA) at Oak Ridge National Laboratory (ORNL) provided extensive neutron flux characterization of a mockup pressure vessel configuration. Considerable effort has been made to understand the uncertainties of the various measurements made in the PCA and to resolve discrepancies in the data. Additional measurements are available for similar configurations in the Oak Ridge Reactor-Poolside Facility (ORR-PSF) at ORNL and in the NESDIP facility in the UK. Comparisons of these results, together with associated neutron field calculations, enable a better evaluation of the actual uncertainties and realistic limits of accuracy to be assessed. Such assessments are especially valuable when the accuracy improvements of benchmark referencing are to be included and extrapolations to new configurations are made
Transmutation studies of minor actinides in high intensity neutron fluxes
International Nuclear Information System (INIS)
Fioni, G.; Bolognese, T.; Cribier, M.; Marie, F.; Roettger, S.; Faust, H.; Leconte, Ph.
1999-01-01
Integral measurements of nuclear data and of the transmutation potential in specific neutron fluxes, constitute the fastest and essential way to overcome to the large uncertainties present in the nuclear data libraries. In the frame of the activities of the Directorate for Science of Matter (DSM) of the French Atomic Energy Authority (CEA), a new project is proposed so as to carry out integral measurements relevant for nuclear waste transmutation systems. A new beam tube will be installed to irradiate actinides and fission fragment samples at different distances from the fuel element of the ILL reactor. Variable neutron energy spectra could then be obtained by choosing the distance between the sample and the fuel element, opening the way to the determination of the ideal physical conditions to incinerate nuclear waste in hybrid transmutation systems. (author)
DEMONR, Monte-Carlo Shielding Calculation for Neutron Flux and Neutron Spectra, Teaching Program
International Nuclear Information System (INIS)
Courtney, J. C.
1987-01-01
1 - Description of problem or function: DEMONR treats the behavior of neutrons in a slab shield. It is frequently used as a teaching tool. 2 - Method of solution: An unbiased Monte Carlo code calculates the number, energy, and direction of neutrons that penetrate or are reflected from a shield. 3 - Restrictions on the complexity of the problem: Only one shield may be used in each problem. The shield material may be a single element or a homogeneous mixture of elements with a single effective atomic weight. Only elastic scattering and neutron capture processes are allowed. The source is a point located on one face of the slab. It provides a cosine distribution of current. Monoenergetic or fission spectrum neutrons may be selected
Measurement of neutron flux distribution by semiconductor detector
International Nuclear Information System (INIS)
Obradovic, D.; Bosevski, T.
1964-01-01
Application of semiconductor detectors for measuring neutron flux distribution is about 10 times faster than measurements by activation foils and demands significantly lower reactor power. Following corrections are avoided: mass of activation foils which influences the self shielding, nuclear decay during activity measurements; counter dead-time. It is possible to control the measured data during experiment and repeat measurements if needed. Precision of the measurement is higher since it is possible to choose the wanted statistics. The method described in this paper is applied for measurements at the RB reactor. It is concluded that the method is suitable for fast measurements but the activation analysis is still indispensable
Neutronic modelling of the Harwell MTR's: some recent problems
International Nuclear Information System (INIS)
Taylor, N.P.
1984-01-01
Use of the Harwell Materials Testing Reactors for the irradiation of experimental rigs gives rise to a number of requirements for calculations of neutron fluxes. In addition photon fluxes are required for estimates of nuclear heating rates. A range of calculational methods are employed, from simple cell to whole reactor models, and the latter have been extended for preliminary design studies for the next generation of MTR to replace DIDO and PLUTO. The technique used for these various models are described in this note, with emphasis on the areas in which modelling problems are encountered. The applications divide into three distinct areas: calculations concerning rigs irradiated within the reactor core, those for rigs positioned in the D 2 O reflector surrounding the core, and design studies for a replacement reactor. (Auth.)
Generalized diffusion theory for calculating the neutron transport scalar flux
International Nuclear Information System (INIS)
Alcouffe, R.E.
1975-01-01
A generalization of the neutron diffusion equation is introduced, the solution of which is an accurate approximation to the transport scalar flux. In this generalization the auxiliary transport calculations of the system of interest are utilized to compute an accurate, pointwise diffusion coefficient. A procedure is specified to generate and improve this auxiliary information in a systematic way, leading to improvement in the calculated diffusion scalar flux. This improvement is shown to be contingent upon satisfying the condition of positive calculated-diffusion coefficients, and an algorithm that ensures this positivity is presented. The generalized diffusion theory is also shown to be compatible with conventional diffusion theory in the sense that the same methods and codes can be used to calculate a solution for both. The accuracy of the method compared to reference S/sub N/ transport calculations is demonstrated for a wide variety of examples. (U.S.)
International Nuclear Information System (INIS)
Erben, O.
1980-01-01
The coefficients of thermal and epithermal neutron flux density depression and self-shielding for the SPN detectors with vanadium, rhodium, silver and cobalt emitters are presented, (for cobalt SPN detectors the functions describing the absorbtion of neutrons along the emitter cross-section are also shown). Using these coefficients and previously published beta particle escape efficiencies, sensitivities are determined for the principal types of detectors produced by Les Cables de Lyon and SODERN companies. The experiments and their results verifying the validity of the theoretical work are described. (author)
Energy Technology Data Exchange (ETDEWEB)
Guyonvarh, M; Lecomte, P; Le Meur, R [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1967-07-01
It is necessary to know, in irradiation loops, the thermal neutron flux after the irradiation device has been introduced and without being obliged to wait for the discharge of this device. In order to measure the flux and to control it continuously, one possible method is to place in the flux a coiled steel tube through which air passes. By measuring the activity of argon 41, and with a knowledge of the flow rate and the temperature of the air, it is possible to calculate the flux. An air-circulation flux controller is described and the relationship between the flux and the count rate is established The accuracy of an absolute measurement is about 14 per cent; that of a relative measurement is about 3 per cent. The measurement can be carried out equally well whether the reactor is operating at maximum or at low power. The measurement range goes from 10{sup 9} to lO{sup 15} n.cm{sup -2}.sec{sup -1}, and it would be possible after a few modifications to measure fluxes between 10{sup 5} and 10{sup 15} n.cm{sup -2}.sec{sup -1}. Finally, the method is very safe to operate: there is little risk of irradiation because of the low specific activity of the argon-41 formed, and no risk of contamination because the decay product of argon-41 is stable. This method, which is now being used in loops, is thus very practical. (authors) [French] Sur des boucles d'irradiation il est necessaire de connaitre le flux de neutrons thermiques apres mise en place du dispositif d'irradiation et sans etre oblige d'attendre le detournement de ce dispositif. Pour mesurer le flux et le controler en permanence, une methode consiste a placer sous flux un serpentin en acier dans lequel on fait circuler de l'air. La mesure d'activite d'argon 41 permet de calculer le flux, connaissant le debit et la temperature de l'air. Un controleur de flux par circulation d'air est decrit et la relation entre le flux et le taux de comptage est etablie. La precision d'une mesure absolue est de l'ordre de 14 pour
Design basis for creep of zirconium alloy components in a fast neutron flux
International Nuclear Information System (INIS)
Ross-Ross, P.A.; Fidleris, V.
1975-01-01
The chalk River Nuclear Laboratory's experience with the creep of zirconium alloys in a neutron flux is described. Fast neutron flux changes the creep behaviour of zirconium alloys and new design criteria for in-reactor applications are needed. From experimental results empirical relations describing the effects of neutron flux, stress, temperature, time and anisotropy on creep rate were established. The relations are applied to the design of pressure tubes. (author)
Design basis for creep of zirconium alloy components in a fast neutron flux
International Nuclear Information System (INIS)
Ross-Ross, P.A.; Fidleris, V.
1974-01-01
The Chalk River Nuclear Laboratory's experience with the creep of zirconium alloys in a neutron flux is described. Fast neutron flux changes the creep behavior of zirconium alloys and new design criteria for in-reactor applications are needed. From experimental results empirical relations describing the effects of neutron flux, stress, temperature, time, and anisotropy on creep rate were established. The relations are applied to the design of pressure tubes. (author)
International Nuclear Information System (INIS)
Fourmentel, D.; Filliatre, P.; Barbot, L.; Villard, J.-F.; Lyoussi, A.; Geslot, B.; Malo, J.-Y.; Carcreff, H.; Reynard-Carette, C.
2013-06-01
Thermal neutron flux is one of the most important nuclear parameter to be measured on-line in Material Testing Reactors (MTRs). In particular two types of sensors with different physical operating principles are commonly used: self-powered neutron detectors (SPND) and fission chambers with uranium 235 coating. This work aims to compare on one hand the thermal neutron flux evaluation given by these two types of sensors and on the other hand to compare these evaluations with activation dosimeter measurements, which are considered as the reference for absolute neutron flux assessment. This study was conducted in an irradiation experiment, called CARMEN-1, performed during 2012 in OSIRIS reactor (CEA Saclay - France). The CARMEN-1 experiment aims to improve the neutron and photon flux and nuclear heating measurements in MTRs. In this paper we focus on the thermal neutron flux measurements performed in CARMEN-1 experiment. The use of fission chambers to measure the absolute thermal neutron flux in MTRs is not very usual. An innovative calibration method for fission chambers operated in Campbell mode has been developed at the CEA Cadarache (France) and tested for the first time in the CARMEN-1 experiment. The results of these measurements are discussed, with the objective to measure with the best accuracy the thermal neutron flux in the future Jules Horowitz Reactor. (authors)
Measuring neutron flux density in near-vessel space of a commercial WWER-1000 reactor
International Nuclear Information System (INIS)
Borodkin, G.I.; Eremin, A.N.; Lomakin, S.S.; Morozov, A.G.
1987-01-01
Distribution of neutron flux density in two experimental channels on the reactor vessel external surface and in ionization chamber channel of a commercial WWER-1000 reactor, is measured by the activation detector technique. Azimuthal distributions of fast and thermal neutron fluxes and height distributions of fast neutron flux density within energy range >1.2 and 2.3 MeV are obtained. Conclusion is made, that reactor core state and its structural peculiarities in the measurement range essentially affect space and energy distribution of neutron field near the vessel. It should be taken into account when determining permissible neutron fluence for the reactor vessel
International Nuclear Information System (INIS)
Talamo, A.; Gohar, Y.; Aliberti, G.; Zhong, Z.; Bournos, V.; Fokov, Y.; Kiyavitskaya, H.; Routkovskaya, C.; Serafimovich, I.
2010-01-01
In 1997, Bretscher calculated the effective delayed neutron fraction by the k-ratio method. The Bretscher's approach is based on calculating the multiplication factor of a nuclear reactor core with and without the contribution of delayed neutrons. The multiplication factor set by the delayed neutrons (the delayed multiplication factor) is obtained as the difference between the total and the prompt multiplication factors. Bretscher evaluated the effective delayed neutron fraction as the ratio between the delayed and total multiplication factors (therefore the method is often referred to as k-ratio method). In the present work, the k-ratio method is applied by deterministic nuclear codes. The ENDF/B nuclear data library of the fuel isotopes ( 238 U and 238 U) have been processed by the NJOY code with and without the delayed neutron data to prepare multigroup WIMSD nuclear data libraries for the DRAGON code. The DRAGON code has been used for preparing the PARTISN macroscopic cross sections. This calculation methodology has been applied to the YALINA-Thermal assembly of Belarus. The assembly has been modeled and analyzed using PARTISN code with 69 energy groups and 60 different material zones. The deterministic and Monte Carlo results for the effective delayed neutron fraction obtained by the k-ratio method agree very well. The results also agree with the values obtained by using the adjoint flux. (authors)
International Nuclear Information System (INIS)
Strugar, P.
1964-12-01
Calculation and experimental results shown in this paper show that higher thermal neutron flux is obtained in the reactor core with central horizontal reflector at the same power level. The flux is increased when the moderation capability of the core is decreased. Apart from increase of the thermal component of the neutron flux in the experimental channels, the central reflector causes decrease of the epithermal neutron flux and gamma radiation intensity. This is very useful for studying (n, γ) reaction, neutron diffraction, etc. [sr
The epithermal neutron-flux distribution in the reactor RA - Vinca
International Nuclear Information System (INIS)
Marinkov, V.; Bikit, I.; Martinc, R.; Veskovic, M.; Slivka, J.; Vaderna, S.
1987-01-01
The distribution of the epithermal neutron flux in the reactor RA - Vinca has been measured by means of Zr - activation detectors. In the channel VK-8 non-homogeneous flux distribution was observed (author) [sr
Neutron point-flux calculation by Monte Carlo
International Nuclear Information System (INIS)
Eichhorn, M.
1986-04-01
A survey of the usual methods for estimating flux at a point is given. The associated variance-reducing techniques in direct Monte Carlo games are explained. The multigroup Monte Carlo codes MC for critical systems and PUNKT for point source-point detector-systems are represented, and problems in applying the codes to practical tasks are discussed. (author)
International Nuclear Information System (INIS)
Sadikov, I.I.; Zinov'ev, V.G.; Sadikova, Z.O.; Salimov, M.I.
2006-01-01
Full text: For effective work of a reactor, and correct planning of experiments related to the reactor irradiation of various materials it is required to control a neutron flux in the given irradiation point for a long irradiation period. For realization of research works on topazes ennobling under irradiation by reactor neutrons as well as by secondary neutrons produced in a cyclotron it is necessary to know the total neutron flux and spectra. To resolve the problem a technique for registration of neutrons with different energy and calculation of a neutrons spectrum in the given irradiation points in reactor channels and in cyclotron behind the nickel target has been developed. Neutron flux density and energy spectra were monitored by use of the following nuclear reactions: 59 Co(n,γ) 60 Co, 197 Au(n,γ) 198 Au, 58 Ni(n,p) 58 Co, 24 Mg(n,p) 24 Na, 48 Ti(n,p) 48 Sc, 46 Ti(n,p) 46 Sc, 54 Fe(n,p) 54 Mn, 89 Y(n,2n) 88 Y, 60 Ni(np) 60 Co. Gamma spectrometer composed of HPGe detector (Rel. Eff. - 15%) and Digital Spectra Analyzer DSA-1000 (Canberra Ind., USA) was used to measure gamma activity of irradiated samples. Acquired gamma spectra were processed by means of Genie 2000 standard software package. The σ(E) functions and neutron spectra were calculated by using the least squares method and approximating the tabular and experimental data with power polynomials. The developed technique was applied for the adjustment of the topazes irradiation regimes in the reactor core and under secondary neutrons flux from a nickel target in the cyclotron. The given technique allows to calculate a logarithmic spectrum of neutrons in a energy range from 0,025 eV up to 12 MeV with the uncertainty of about 10 %. (author)
A fission ionization detector for neutron flux measurements at a spallation source
Energy Technology Data Exchange (ETDEWEB)
Wender, S.A. (Los Alamos National Lab., Los Alamos, NM (United States)); Balestrini, S. (Los Alamos National Lab., Los Alamos, NM (United States)); Brown, A. (Los Alamos National Lab., Los Alamos, NM (United States)); Haight, R.C. (Los Alamos National Lab., Los Alamos, NM (United States)); Laymon, C.M. (Los Alamos National Lab., Los Alamos, NM (United States)); Lee, T.M. (Los Alamos National Lab., Los Alamos, NM (United States)); Lisowski, P.W. (Los Alamos National Lab., Los Alamos, NM (United States)); McCorkle, W. (Los Alamos National Lab., Los Alamos, NM (United States)); Nelson, R.O. (Los Alamos National Lab., Los Alamos, NM (United States)); Parker, W. (Los Alamos National Lab., Los Alamos, NM (United States)); Hill, N.W. (Oak Ridge National Lab., Oak Ridge, TN (United States))
1993-11-15
The construction of a neutron flux monitor that can measure absolute neutron intensities in the neutron energy range from below 1 MeV to over 500 MeV is described. The detector consists of an ionization chamber with several thin deposits of fissionable material. The ionization chamber is thin enough that it does not significantly affect the neutron beam and may be left in the neutron flight path during experimental measurements to continuously monitor the beam flux. The use of this monitor at the continuous-energy spallation neutron source at the WNR target area at LAMPF is described. (orig.)
A fission ionization detector for neutron flux measurements at a spallation source
International Nuclear Information System (INIS)
Wender, S.A.; Balestrini, S.; Brown, A.; Haight, R.C.; Laymon, C.M.; Lee, T.M.; Lisowski, P.W.; McCorkle, W.; Nelson, R.O.; Parker, W.; Hill, N.W.
1993-01-01
The construction of a neutron flux monitor that can measure absolute neutron intensities in the neutron energy range from below 1 MeV to over 500 MeV is described. The detector consists of an ionization chamber with several thin deposits of fissionable material. The ionization chamber is thin enough that it does not significantly affect the neutron beam and may be left in the neutron flight path during experimental measurements to continuously monitor the beam flux. The use of this monitor at the continuous-energy spallation neutron source at the WNR target area at LAMPF is described. (orig.)
International Nuclear Information System (INIS)
Cartier, J.; Casoli, P.; Chappert, F.
2013-01-01
In this paper, we present calibration methods in order to estimate reactor neutron flux spectrum and its uncertainties by using integral activation measurements. These techniques are performed using Bayesian and MCMC framework. These methods are applied to integral activation experiments in the cavity of the CALIBAN reactor. We estimate the neutron flux and its related uncertainties. The originality of this work is that these uncertainties take into account measurements uncertainties, cross-sections uncertainties and model error. In particular, our results give a very good approximation of the total flux and indicate that neutron flux from MCNP simulation for energies above about 5 MeV seems to overestimate the 'real flux'. (authors)
International Nuclear Information System (INIS)
Jammes, Christian; Filliatre, Philippe; Verma, Vasudha; Hellesen, Carl; Jacobsson Svard, Staffan
2015-01-01
Neutron flux monitoring system forms an integral part of the design of a Generation IV sodium cooled fast reactor system. Diverse possibilities of detector systems installation have to be investigated with respect to practicality and feasibility according to the detection parameters. In this paper, we demonstrate the feasibility of using self powered neutron detectors as in-core detectors in fast reactors for detecting local change in neutron flux distribution. We show that the gamma contribution from fission products decay in the fuel and activation of structural materials is very small compared to the fission gammas. Thus, it is possible for the in-core SPND signal to follow changes in local neutron flux as they are proportional to each other. This implies that the signal from an in-core SPND can provide dynamic information on the neutron flux perturbations occurring inside the reactor core. (authors)
Energy Technology Data Exchange (ETDEWEB)
Jammes, Christian; Filliatre, Philippe [CEA, DEN, DER, Instrumentation Sensors and Dosimetry Laboratory, Cadarache, F-13108 St Paul-Lez-Durance, (France); Verma, Vasudha; Hellesen, Carl; Jacobsson Svard, Staffan [Division of Applied Nuclear Physics, Uppsala University, SE-75120 Uppsala, (Sweden)
2015-07-01
Neutron flux monitoring system forms an integral part of the design of a Generation IV sodium cooled fast reactor system. Diverse possibilities of detector systems installation have to be investigated with respect to practicality and feasibility according to the detection parameters. In this paper, we demonstrate the feasibility of using self powered neutron detectors as in-core detectors in fast reactors for detecting local change in neutron flux distribution. We show that the gamma contribution from fission products decay in the fuel and activation of structural materials is very small compared to the fission gammas. Thus, it is possible for the in-core SPND signal to follow changes in local neutron flux as they are proportional to each other. This implies that the signal from an in-core SPND can provide dynamic information on the neutron flux perturbations occurring inside the reactor core. (authors)
Design features of HANARO Neutron Flux Monitoring System and its operating experiences
International Nuclear Information System (INIS)
Kim, Young-Ki; Ahn, Guk-Hoon
1999-01-01
The Neutron Flux Monitoring System for HANARO provides reliable neutron flux measurement from reactor shutdown to reactor full power level ranging 10 decades from 10 0 nv to 10 10 nv. The neutron flux monitoring system consists of a guarded fission chamber, amplifier and signal processor. The neutron flux as the measure of reactor power is continuously monitored by six(6) fission chambers mounted on the courtside wall of the reflector tank in the pool. Three(3) of the fission chambers are used for reactor power control, while the other three(3) are used for tripping the reactor in case of power excursion. Only the wide range fission chamber-based neutron monitoring system is employed for neutron power measurement thereby source range and intermediate range detectors are not necessary and the number of neutron monitoring channels are minimized at HANARO. (author)
Directory of Open Access Journals (Sweden)
Abdessamad Didi
2017-06-01
Full Text Available Americium–beryllium (Am-Be; n, γ is a neutron emitting source used in various research fields such as chemistry, physics, geology, archaeology, medicine, and environmental monitoring, as well as in the forensic sciences. It is a mobile source of neutron activity (20 Ci, yielding a small thermal neutron flux that is water moderated. The aim of this study is to develop a model to increase the neutron thermal flux of a source such as Am-Be. This study achieved multiple advantageous results: primarily, it will help us perform neutron activation analysis. Next, it will give us the opportunity to produce radio-elements with short half-lives. Am-Be single and multisource (5 sources experiments were performed within an irradiation facility with a paraffin moderator. The resulting models mainly increase the thermal neutron flux compared to the traditional method with water moderator.
International Nuclear Information System (INIS)
Khattab, K.; Omar, H.; Ghazi, N.
2009-01-01
A 3-D (R, θ , Z) neutronic model for the Miniature Neutron Source Reactor (MNSR) was developed earlier to conduct the reactor neutronic analysis. The group constants for all the reactor components were generated using the WIMSD4 code. The reactor excess reactivity and the four group neutron flux distributions were calculated using the CITATION code. This model is used in this paper to calculate the point wise four energy group neutron flux distributions in the MNSR versus the radius, angle and reactor axial directions. Good agreement is noticed between the measured and the calculated thermal neutron flux in the inner and the outer irradiation site with relative difference less than 7% and 5% respectively. (author)
Development of self-powered neutron detectors for neutron flux monitoring in HCLL and HCPB ITER-TBM
International Nuclear Information System (INIS)
Angelone, M.; Klix, A.; Pillon, M.; Batistoni, P.; Fischer, U.; Santagata, A.
2014-01-01
Highlights: •Self powered neutron detector (SPND) is attractive neutron monitor for TBM in ITER. •In hard neutron spectra (e.g. TBM) there is the need to optimize their response. •Three state-of-the-art SPNDs were tested using fast and 14 MeV neutrons. •The response of SPNDs is much lower than in thermal neutron flux. •FISPACT calculations performed to find out candidate materials in hard spectra. -- Abstract: Self powered neutron detectors (SPND) have a number of interesting properties (e.g. small dimensions, capability to operate in harsh environments, absence of external bias), so they are attractive neutron monitors for TBM in ITER. However, commercially available SPNDs are optimized for operation in a thermal nuclear reactor where the neutron spectrum is much softer than that expected in a TBM. This fact can limit the use of SPND in a TBM since the effective cross sections for the production of beta emitters are much lower in a fast neutron spectrum. This work represents the first attempt to study SPNDs as neutron flux monitors for TBM. Three state-of-the-art SPND available on the market were bought and tested using fast neutrons at TAPIRO fast neutron source of ENEA Casaccia and with 14 MeV neutrons at the Frascati neutron generator (FNG). The results clearly indicate that in fast neutron spectra, the response of SPNDs is much lower than in thermal neutron flux. Activation calculations were performed using the FISPACT code to find out possible material candidates for SPND suitable for operation in TBM neutron spectra
Development of self-powered neutron detectors for neutron flux monitoring in HCLL and HCPB ITER-TBM
Energy Technology Data Exchange (ETDEWEB)
Angelone, M., E-mail: maurizio.angelone@enea.it [Associazione ENEA-EURATOM sulla FusioneENEA C.R. Frascati, Via E. Fermi 45, 00044 Frascati (Italy); Klix, A. [Association KIT-EURATOM, Karlsruhe Institute of Technology (KIT), Hermann-von-Helmholtz-Platz 1, 76344 Eggenstein-Leopoldshafen (Germany); Pillon, M.; Batistoni, P. [Associazione ENEA-EURATOM sulla FusioneENEA C.R. Frascati, Via E. Fermi 45, 00044 Frascati (Italy); Fischer, U. [Association KIT-EURATOM, Karlsruhe Institute of Technology (KIT), Hermann-von-Helmholtz-Platz 1, 76344 Eggenstein-Leopoldshafen (Germany); Santagata, A. [ENEA C.R. Casaccia, via Anguillarese Km. 1,300, 00100 Roma (Italy)
2014-10-15
Highlights: •Self powered neutron detector (SPND) is attractive neutron monitor for TBM in ITER. •In hard neutron spectra (e.g. TBM) there is the need to optimize their response. •Three state-of-the-art SPNDs were tested using fast and 14 MeV neutrons. •The response of SPNDs is much lower than in thermal neutron flux. •FISPACT calculations performed to find out candidate materials in hard spectra. -- Abstract: Self powered neutron detectors (SPND) have a number of interesting properties (e.g. small dimensions, capability to operate in harsh environments, absence of external bias), so they are attractive neutron monitors for TBM in ITER. However, commercially available SPNDs are optimized for operation in a thermal nuclear reactor where the neutron spectrum is much softer than that expected in a TBM. This fact can limit the use of SPND in a TBM since the effective cross sections for the production of beta emitters are much lower in a fast neutron spectrum. This work represents the first attempt to study SPNDs as neutron flux monitors for TBM. Three state-of-the-art SPND available on the market were bought and tested using fast neutrons at TAPIRO fast neutron source of ENEA Casaccia and with 14 MeV neutrons at the Frascati neutron generator (FNG). The results clearly indicate that in fast neutron spectra, the response of SPNDs is much lower than in thermal neutron flux. Activation calculations were performed using the FISPACT code to find out possible material candidates for SPND suitable for operation in TBM neutron spectra.
International Nuclear Information System (INIS)
Mansur, Ralph S.; Barros, Ricardo C.
2011-01-01
We describe a method to determine the neutron scalar flux in a slab using monoenergetic diffusion model. To achieve this goal we used three ingredients in the computational code that we developed on the Scilab platform: a spectral nodal method that generates numerical solution for the one-speed slab-geometry fixed source diffusion problem with no spatial truncation errors; a spatial reconstruction scheme to yield detailed profile of the coarse-mesh solution; and an angular reconstruction scheme to yield approximately the neutron angular flux profile at a given location of the slab migrating in a given direction. Numerical results are given to illustrate the efficiency of the offered code. (author)
Maximum neutron flux in thermal reactors; Maksimum neutronskog fluksa kod termalnih reaktora
Energy Technology Data Exchange (ETDEWEB)
Strugar, P V [Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Yugoslavia)
1968-07-01
Direct approach to the problem is to calculate spatial distribution of fuel concentration if the reactor core directly using the condition of maximum neutron flux and comply with thermal limitations. This paper proved that the problem can be solved by applying the variational calculus, i.e. by using the maximum principle of Pontryagin. Mathematical model of reactor core is based on the two-group neutron diffusion theory with some simplifications which make it appropriate from maximum principle point of view. Here applied theory of maximum principle are suitable for application. The solution of optimum distribution of fuel concentration in the reactor core is obtained in explicit analytical form. The reactor critical dimensions are roots of a system of nonlinear equations and verification of optimum conditions can be done only for specific examples.
Advanced neutron source design: burnout heat flux correlation development
International Nuclear Information System (INIS)
Gambill, W.R.; Mochizuki, T.
1988-01-01
In the advanced neutron source reactor (ANSR) fuel element region, heat fluxes will be elevated. Early designs corresponded to average and estimated hot-spot fluxes of 11 to 12 and 21 to 22 MW/m 2 , respectively. Design changes under consideration may lower these values to ∼ 9 and 17 MW/m 1 . In either event, the development of a satisfactory burnout heat flux correlation is an important element among the many thermal-hydraulic design issues, since the critical power ratio will depend in part on its validity. Relatively little work in the area of subcooled-flow burnout has been published over the past 12 yr. The authors have compared seven burnout correlations and modifications therefore with several sets of experimental data, of which the most relevant to the ANS core are those referenced. The best overall agreement between the correlations tested and these data is currently provided by a modification of Thorgerson et al. correlation. The variable ranges of the experimental data are outlined and the results of the correlation comparisons are summarized
Wahid, Kareem; Sanchez, Patrick; Hannan, Mohammad
2014-03-01
In the field of nuclear science, neutron flux is an intrinsic property of nuclear reaction facilities that is the basis for experimental irradiation calculations and analysis. In the Rio Grande Valley (Texas), the UTPA Neutron Research Facility (NRF) is currently the only neutron facility available for experimental research purposes. The facility is comprised of a 20-microgram californium-252 neutron source surrounded by a shielding cascade containing different irradiation cavities. Thermal and fast neutron flux values for the UTPA NRF have yet to be fully investigated and may be of particular interest to biomedical studies in low neutron dose applications. Though a variety of techniques exist for the characterization of neutron flux, neutron activation analysis (NAA) of metal and nonmetal foils is a commonly utilized experimental method because of its detection sensitivity and availability. The aim of our current investigation is to employ foil activation in the determination of neutron flux values for the UTPA NSRF for further research purposes. Neutron spectrum unfolding of the acquired experimental data via specialized software and subsequent comparison for consistency with computational models lends confidence to the results.
Measurement and simulation of thermal neutron flux distribution in the RTP core
Rabir, Mohamad Hairie B.; Jalal Bayar, Abi Muttaqin B.; Hamzah, Na'im Syauqi B.; Mustafa, Muhammad Khairul Ariff B.; Karim, Julia Bt. Abdul; Zin, Muhammad Rawi B. Mohamed; Ismail, Yahya B.; Hussain, Mohd Huzair B.; Mat Husin, Mat Zin B.; Dan, Roslan B. Md; Ismail, Ahmad Razali B.; Husain, Nurfazila Bt.; Jalil Khan, Zareen Khan B. Abdul; Yakin, Shaiful Rizaide B. Mohd; Saad, Mohamad Fauzi B.; Masood, Zarina Bt.
2018-01-01
The in-core thermal neutron flux distribution was determined using measurement and simulation methods for the Malaysian’s PUSPATI TRIGA Reactor (RTP). In this work, online thermal neutron flux measurement using Self Powered Neutron Detector (SPND) has been performed to verify and validate the computational methods for neutron flux calculation in RTP calculations. The experimental results were used as a validation to the calculations performed with Monte Carlo code MCNP. The detail in-core neutron flux distributions were estimated using MCNP mesh tally method. The neutron flux mapping obtained revealed the heterogeneous configuration of the core. Based on the measurement and simulation, the thermal flux profile peaked at the centre of the core and gradually decreased towards the outer side of the core. The results show a good agreement (relatively) between calculation and measurement where both show the same radial thermal flux profile inside the core: MCNP model over estimation with maximum discrepancy around 20% higher compared to SPND measurement. As our model also predicts well the neutron flux distribution in the core it can be used for the characterization of the full core, that is neutron flux and spectra calculation, dose rate calculations, reaction rate calculations, etc.
Application of direct discrete method (DDM) to multigroup neutron transport problems
International Nuclear Information System (INIS)
Vosoughi, Naser; Salehi, Ali Akbar; Shahriari, Majid
2003-01-01
The Direct Discrete Method (DDM), which produced excellent results for one-group neutron transport problems, has been developed for multigroup energy. A multigroup neutron transport discrete equation has been produced for a cylindrical shape fuel element with and without associated coolant regions with two boundary conditions. The calculations are illustrated for two-group energy by graphs showing the fast and thermal fluxes. The validity of the results are tested against the results obtained by the ANISN code. (author)
Measured and Predicted Neutron Flux Distributions in a Material Surrounding a Cylindrical Duct
Energy Technology Data Exchange (ETDEWEB)
Nilsson, J; Sandlin, R
1966-03-15
The radial fast neutron flux attenuations in the material (iron) surrounding ducts of diameters 7, 9, and 15 cm and total duct length of about 1.5 m have been investigated with and without neutron scattering cans filled with D{sub 2}O in the duct. Experimentally the problem was solved by the use of foil activation techniques. Theoretically it was attacked by, in the first place, a Monte Carlo program specially written for this purpose and utilizing an importance sampling technique. In the second place non- and single-scattering removal flux codes were tried, and also simple hand calculations. The Monte Carlo results accounted well for the fast flux attenuation, while the non- and single-scattering methods overestimated the attenuation generally by a factor of 10 or less. Simple hand calculations using three empirical parameters could be fitted to the measured data within a factor of 1.2 - 1.3 at penetration depths greater than 3 - 4 cm. The distribution of the D{sub 2}O-scattered flux could well be described in terms of single scattering.
High-flux neutron source based on a liquid-lithium target
Halfon, S.; Feinberg, G.; Paul, M.; Arenshtam, A.; Berkovits, D.; Kijel, D.; Nagler, A.; Eliyahu, I.; Silverman, I.
2013-04-01
A prototype compact Liquid Lithium Target (LiLiT), able to constitute an accelerator-based intense neutron source, was built. The neutron source is intended for nuclear astrophysical research, boron neutron capture therapy (BNCT) in hospitals and material studies for fusion reactors. The LiLiT setup is presently being commissioned at Soreq Nuclear research Center (SNRC). The lithium target will produce neutrons through the 7Li(p,n)7Be reaction and it will overcome the major problem of removing the thermal power generated by a high-intensity proton beam, necessary for intense neutron flux for the above applications. The liquid-lithium loop of LiLiT is designed to generate a stable lithium jet at high velocity on a concave supporting wall with free surface toward the incident proton beam (up to 10 kW). During off-line tests, liquid lithium was flown through the loop and generated a stable jet at velocity higher than 5 m/s on the concave supporting wall. The target is now under extensive test program using a high-power electron-gun. Up to 2 kW electron beam was applied on the lithium flow at velocity of 4 m/s without any flow instabilities or excessive evaporation. High-intensity proton beam irradiation will take place at SARAF (Soreq Applied Research Accelerator Facility) superconducting linear accelerator currently in commissioning at SNRC.
Maximum neutron flux at thermal nuclear reactors; Maksimum neutronskog fluksa kod termalnih reaktora
Energy Technology Data Exchange (ETDEWEB)
Strugar, P [Institute of Nuclear Sciences Vinca, Beograd (Serbia and Montenegro)
1968-10-15
Since actual research reactors are technically complicated and expensive facilities it is important to achieve savings by appropriate reactor lattice configurations. There is a number of papers, and practical examples of reactors with central reflector, dealing with spatial distribution of fuel elements which would result in higher neutron flux. Common disadvantage of all the solutions is that the choice of best solution is done starting from the anticipated spatial distributions of fuel elements. The weakness of these approaches is lack of defined optimization criteria. Direct approach is defined as follows: determine the spatial distribution of fuel concentration starting from the condition of maximum neutron flux by fulfilling the thermal constraints. Thus the problem of determining the maximum neutron flux is solving a variational problem which is beyond the possibilities of classical variational calculation. This variational problem has been successfully solved by applying the maximum principle of Pontrjagin. Optimum distribution of fuel concentration was obtained in explicit analytical form. Thus, spatial distribution of the neutron flux and critical dimensions of quite complex reactor system are calculated in a relatively simple way. In addition to the fact that the results are innovative this approach is interesting because of the optimization procedure itself. [Serbo-Croat] Savremeni reaktori za fizicka i tehnoloska istrazivanja predstavljaju tehnicki komplikovanu i skupu masinu. Iz tog razloga su opravdana nastojanja da se podesnim rasporedom goriva u jezgru reaktora dodje do sto ekonomicnijeg rjesenja. U literaturi postoji vise radova, cak i konkretnih realizacija u vidu reaktora sa reflektorom u centru, koji se bave odredjivanjem takve prostorne zavisnosti koncentracije goriva koja pod odredjenim uslovima daje najveci neutronski fluks. Zajednicki nedostatak svih pomenutih rjesenja je u tome sto se polazi od pretpostavljenih prostornih distribucija
Kitis, G; Wiescher, M; Dahlfors, M; Soares, J
2002-01-01
We propose to measure the neutron capture cross sections of $^{139}$La, of $^{93}$Zr (t$_{1/2}$)=1.5 10$^{6}$ yr), and of all the stable Zr isotopes at n_TOF. The aim of these measurements is to improve the accuracy of existing results by at least a factor of three in order to meet the quality required for using the s-process nucleosynthesis as a diagnostic tool for neutron exposure and neutron flux during the He burning stages of stellar evolution. Combining these results with a wealth of recent information coming from high-resolution stellar spectroscopy and from the detailed analysis of presolar dust grains will shed new light on the chemical history of the universe. The investigated cross sections are also needed for technological applications, in particular since $^{93}$Zr is one of the major long-lived fission products.
Calculation of the thermal neutron flux depression in the loop VISA-1
International Nuclear Information System (INIS)
Martinc, R.
1961-01-01
Among other applications, the VISA-1 loop is to be used for thermal load testing of materials. For this type of testing one should know the maximum power generated in the loop. This power is determined from the maximum thermal neutron flux in the VK-5 channel and mean flux depression in the fissile component of the loop. Thermal neutron flux depression is caused by neutron absorption in the components of the loop, shape of the components and neutron leaking through gaps as well as properties of the surrounding medium of the core. All these parameters were taken into account for calculating the depression of thermal neutron flux in the VISA-1 loop. Two group diffusion theory was used. Fast neutron from the fission in the loop and slowed down were taken into account. Depression of the thermal neutron flux is expressed by depression factor which represents the ratio of the mean thermal neutron flux in the fissile loop component and the thermal neutron flux in the VK-5 without the loop. Calculation error was estimated and it was recommended to determine the depression factor experimentally as well [sr
Reliability analysis of neutron flux monitoring system for PFBR
International Nuclear Information System (INIS)
Rajesh, M.G.; Bhatnagar, P.V.; Das, D.; Pithawa, C.K.; Vinod, Gopika; Rao, V.V.S.S.
2010-01-01
The Neutron Flux Monitoring System (NFMS) measures reactor power, rate of change of power and reactivity changes in the core in all states of operation and shutdown. The system consists of instrument channels that are designed and built to have high reliability. All channels are required to have a Mean Time Between Failures (MTBF) of 150000 hours minimum. Failure Mode and Effects Analysis (FMEA) and failure rate estimation of NFMS channels has been carried out. FMEA is carried out in compliance with MIL-STD-338B. Reliability estimation of the channels is done according to MIL-HDBK-217FN2. Paper discusses the methodology followed for FMEA and failure rate estimation of two safety channels and results. (author)
Neutron flux and power in RTP core-15
Energy Technology Data Exchange (ETDEWEB)
Rabir, Mohamad Hairie, E-mail: m-hairie@nuclearmalaysia.gov.my; Zin, Muhammad Rawi Md; Usang, Mark Dennis; Bayar, Abi Muttaqin Jalal; Hamzah, Na’im Syauqi Bin [Nuclear and reactor Physics Section, Nuclear Technology Center, Technical Support Division, Malaysian Nuclear Agency, Bangi, 43000 Kajang, Selangor (Malaysia)
2016-01-22
PUSPATI TRIGA Reactor achieved initial criticality on June 28, 1982. The reactor is designed to effectively implement the various fields of basic nuclear research, manpower training, and production of radioisotopes. This paper describes the reactor parameters calculation for the PUSPATI TRIGA REACTOR (RTP); focusing on the application of the developed reactor 3D model for criticality calculation, analysis of power and neutron flux distribution of TRIGA core. The 3D continuous energy Monte Carlo code MCNP was used to develop a versatile and accurate full model of the TRIGA reactor. The model represents in detailed all important components of the core with literally no physical approximation. The consistency and accuracy of the developed RTP MCNP model was established by comparing calculations to the available experimental results and TRIGLAV code calculation.
The calculation of neutron flux using Monte Carlo method
Günay, Mehtap; Bardakçı, Hilal
2017-09-01
In this study, a hybrid reactor system was designed by using 99-95% Li20Sn80 + 1-5% RG-Pu, 99-95% Li20Sn80 + 1-5% RG-PuF4, and 99-95% Li20Sn80 + 1-5% RG-PuO2 fluids, ENDF/B-VII.0 evaluated nuclear data library and 9Cr2WVTa structural material. The fluids were used in the liquid first wall, liquid second wall (blanket) and shield zones of a fusion-fission hybrid reactor system. The neutron flux was calculated according to the mixture components, radial, energy spectrum in the designed hybrid reactor system for the selected fluids, library and structural material. Three-dimensional nucleonic calculations were performed using the most recent version MCNPX-2.7.0 the Monte Carlo code.
Measurement of absolute neutron flux in LWSCR based on the nuclear track method
International Nuclear Information System (INIS)
Sadeghzadeh, J.; Nassiri Mofakham, N.; Khajehmiri, Z.
2012-01-01
Highlights: ► Up to now the spectral parameters of thermal neutrons are measured with activation foils that are not always reliable in low flux systems. ► We applied a solid state nuclear track detector to measure the absolute neutron flux in the light water sub-critical reactor (LWSCR). ► Experiments concerning fission track detecting were performed and were investigated using the Monte Carlo code MCNP. ► The neutron fluxes obtained in experiment are in fairly good agreement with the results obtained by MCNP. - Abstract: In the present paper, a solid state nuclear track detector is applied to measure the absolute neutron flux in the light water sub-critical reactor (LWSCR) in Nuclear Science and Technology Research Institute (NSTRI). Up to now, the spectral parameters of thermal neutrons have been measured with activation foils that are not always reliable in low flux systems. The method investigated here is the irradiation method. Experiments concerning fission track detecting were performed. The experiment including neutron flux calculation method has also been investigated using the Monte Carlo code MCNP. The analysis shows that the values of neutron flux obtained by experiment are in fairly good agreement with the results obtained by MCNP. Thus, this method may be able to predict the absolute value of neutron flux at LWSCR and other similar reactors.
Transmutation and neutron flux studies with fission chambers in the MEGAPIE target
International Nuclear Information System (INIS)
Chabod, S.; Foucher, Y.; Letourneau, A.; Marie, F.; Toussaint, J.C.; Blandin, Ch.; Chartier, F.; Fioni, G.
2005-01-01
Eight fission micro chambers will be inserted inside the central rod of the 1 MW liquid Pb-Bi MEGAPIE target in order to study the transmutation of two major actinides and to measure the neutron flux at a level of 5%. These chambers were developed for high neutron fluxes and tested at Laue Langevin Institute. (authors)
One group neutron flux at a point in a cylindrical reactor cell calculated by Monte Carlo
Energy Technology Data Exchange (ETDEWEB)
Kocic, A [Institute of Nuclear Sciences Vinca, Beograd (Serbia and Montenegro)
1974-01-15
Mean values of the neutron flux over material regions and the neutron flux at space points in a cylindrical annular cell (one group model) have been calculated by Monte Carlo. The results are compared with those obtained by an improved collision probability method (author)
Digital module for neutron flux measurement by Campbell method
International Nuclear Information System (INIS)
Baratte, G.
1987-02-01
The study reported here concerns a wide range measurement channel for reactor control instrumentation but it may also be useful for specific measurements requiring the Campbell method. A wide range measurement channel allows the processing of the signal issued from a single fission chamber so it's possible to insure control of nuclear reactors in three different running modes: pulse processing, fluctuations and current. The study described in this note includes three parts: - the analogical wide range neutron measurement channel is presented in the first chapter; the fluctuation mode is thoroughly studied; the results of tests and proper limitations of analogical processing are summarized. A theoretical study of the neutron flux measurement by numerical calculation of the fluctuation signal variance is given in the second chapter. The digital module is described in the third chapter; the results of experiments are analysed. The validity of the digital method is proved by means of a practical realisation. The performances obtained with the digital fluctuation test model may be compared with those given by the analogical fluctuation channel which can be used for the control of lower fission rates. The digital module may also be used for any fluctuation measurement where very short response time and broad spectral band of analysis are not strictly necessary [fr
SINUPERM N: a new digital neutron flux density monitoring system
International Nuclear Information System (INIS)
Flick, H.A.
1993-01-01
The new SINUPERM N System is developed for Neutron Monitoring in nuclear power plants. The development was started in 1989 (with the design specification) and will be finished in 1993 (with the qualification). The first built will be the nuclear power plant in Borselle (Netherlands). The design is based on a microprocessor system with a digital signal processor for calculations and signal filtering. The separation between analogue-input signals and digital processing enables for each detector type special input modules and standard output interfaces e.g. field - bus. The wide range of the Neutron Flux Density from 10 -2 cm -2 s -1 up to 10 8 cm -2 s -1 for the out-of-pile instrumentation and up to 10 14 cm -2 s -1 for the in-core-instrumentation will be covered by the SINUPERM N system. The requirements to be met by the SINUPERM N system are the IEEE 323, IEC 987 and the German standard KTA-3503 for safety systems. Other standards for instrumentation and control systems like IEC 801, IEC 1131 and IEC 68 for EMV, climatic and seismic requirements are also included in the hardware type test. The software requirement depends on the IEC 880 standard. (author). 3 figs
The effect of an iron plug on the neutron flux distributions in water
International Nuclear Information System (INIS)
Lotfi, A.; Maayouf, R.M.A.; Megahid, R.
1978-01-01
This work is concerned with studying both fast and thermal neutron fluxes distribution in water and its perturbation due to the presence of a cylindrical iron plug. The measurements were carried out using a collimated neutron beam emitted from one of the horizontal channels of the ET-RR-1 reactor. The fast neutron fluxes were measured using phosphorus activation detectors, while the thermal neutron ones were measured using fission fragment track detectors from glass. The results show that the presence of an iron plug causes a remarkable change in the intensities of both the fast and thermal neutron fluxes distribution in the water medium surrounding the iron plug. The flux intensities at the peaks, formed beyond the iron plug in case of thermal neutrons, are also compared with values calculated using the available emperical formula
Determination of neutron flux distribution in an Am-Be irradiator using the MCNP.
Shtejer-Diaz, K; Zamboni, C B; Zahn, G S; Zevallos-Chávez, J Y
2003-10-01
A neutron irradiator has been assembled at IPEN facilities to perform qualitative-quantitative analysis of many materials using thermal and fast neutrons outside the nuclear reactor premises. To establish the prototype specifications, the neutron flux distribution and the absorbed dose rates were calculated using the MCNP computer code. These theoretical predictions then allow one to discuss the optimum irradiator design and its performance.
Development of SiC Neutron Detector Assembly to Measure the Neutron Flux of the Reactor Core
Energy Technology Data Exchange (ETDEWEB)
Park, Se Hwan; Park, June Sic; Shin, Hee Sung; Kim, Ho Dong [Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of); Kim, Yong Kyun [Hanyang University, Seoul (Korea, Republic of)
2012-05-15
At present, the conventional detector to measure the neutron at harsh environment is a Self Powered Neutron Detector (SPND). Rhodium(Rh)-103 is in the SPND. When neutron is incident on the Rhodium, the neutron capture reaction occurs, and the Rh-103 is converted to Rh-104. The Rh-104 is decayed to Pd-104 by {beta}-decay, and electrons are generated as the decay products. Because of the half life of Rh-104, approximately 5 minutes are required for the SPND output to reach the equilibrium condition. Therefore the on-line monitoring of the nuclear reactor state is limited if the neutron flux in the reactor core is monitored with the SPND. Silicon carbide (SiC) has the possibility to be developed as neutron detector at harsh environment, because the SiC can be operative at high temperature and high neutron flux conditions. Previously, the basic operation properties of the SiC detector were studied. Also, the radiation response of the SiC detector was studied at high neutron and gamma dose rate. The measurement results for an ex-core neutron flux monitor or a neutron flux monitor of the spent fuel were published. The SiC detector was also developed as neutron detector to measure the fissile material with active interrogation method. However, the studies about the development of SiC detector are still limited. In the present work, the radiation damage effect of the SiC detector was studied. The detector structure was determined based on the study, and a neutron detector assembly was made with the SiC detectors. The neutron and gamma-ray response of the detector assembly is presented in this paper. The detector assembly was positioned in the HANARO research reactor core, the performance test was done. The preliminary results are also included in this paper
International Nuclear Information System (INIS)
Yavar, A.R.; Sarmani, S.B.; Wood, A.K.; Fadzil, S.M.; Radir, M.H.; Khoo, K.S.
2011-01-01
Determination of thermal to fast neutron flux ratio (f fast ) and fast neutron flux (φ fast ) is required for fast neutron reactions, fast neutron activation analysis, and for correcting interference reactions. The f fast and subsequently φ fast were determined using the absolute method. The f fast ranged from 48 to 155, and the φ fast was found in the range 1.03x10 10 -4.89x10 10 n cm -2 s -1 . These values indicate an acceptable conformity and applicable for installation of the fast neutron facility at the MNA research reactor.
Research on measurement of neutron flux in irradiation channels of research reactor
International Nuclear Information System (INIS)
Yin Zhitao; Lv Zheng; Wang Yulin; Zheng Wuqin
2014-01-01
Relative distribution of thermal neutron flux in the irradiation channel is measured by classical activation foil method. After that, on a representative point in the irradiation channel, neutron temperature and absolute neutron flux are also measured. Cadmium ratio correction method is used to check the experiment result in the end. Comparative analysis shows that the results from two different methods are agreed pretty well, which adds the credibility of experiment results. (authors)
Connection factor calculation for isotopic neutron flux measurements with foil detectors
International Nuclear Information System (INIS)
Avila L, J.
1987-01-01
Thermal and resonance neutron self-shielding factors, neutron flux distortion and edge effects as well as a connection factor for neutron flux profile around a foil detector have been calculated. A general expression for resonance self shielding factor is presented in order to take into account the most important resonances for a given isotope. A computer program SPRESYTER.BAS was written and results for In-115 and Au-197 foils are given
Miller, Marcelo E; Sztejnberg, Manuel L; González, Sara J; Thorp, Silvia I; Longhino, Juan M; Estryk, Guillermo
2011-12-01
A rhodium self-powered neutron detector (Rh SPND) has been specifically developed by the Comisión Nacional de Energía Atómica (CNEA) of Argentina to measure locally and in real time thermal neutron fluxes in patients treated with boron neutron capture therapy (BNCT). In this work, the thermal and epithermal neutron response of the Rh SPND was evaluated by studying the detector response to two different reactor spectra. In addition, during clinical trials of the BNCT Project of the CNEA, on-line neutron flux measurements using the specially designed detector were assessed. The first calibration of the detector was done with the well-thermalized neutron spectrum of the CNEA RA-3 reactor thermal column. For this purpose, the reactor spectrum was approximated by a Maxwell-Boltzmann distribution in the thermal energy range. The second calibration was done at different positions along the central axis of a water-filled cylindrical phantom, placed in the mixed thermal-epithermal neutron beam of CNEA RA-6 reactor. In this latter case, the RA-6 neutron spectrum had been well characterized by both calculation and measurement, and it presented some marked differences with the ideal spectrum considered for SPND calibrations at RA-3. In addition, the RA-6 neutron spectrum varied with depth in the water phantom and thus the percentage of the epithermal contribution to the total neutron flux changed at each measurement location. Local (one point-position) and global (several points-positions) and thermal and mixed-field thermal neutron sensitivities were determined from these measurements. Thermal neutron flux was also measured during BNCT clinical trials within the irradiation fields incident on the patients. In order to achieve this, the detector was placed on patient's skin at dosimetric reference points for each one of the fields. System stability was adequate for this kind of measurement. Local mixed-field thermal neutron sensitivities and global thermal and mixed
Energy Technology Data Exchange (ETDEWEB)
Miller, Marcelo E.; Sztejnberg, Manuel L.; Gonzalez, Sara J.; Thorp, Silvia I.; Longhino, Juan M.; Estryk, Guillermo [Comision Nacional de Energia Atomica, Av. del Libertador 8250, Ciudad de Buenos Aires 1429 (Argentina); Comision Nacional de Energia Atomica, Av. del Libertador 8250, Ciudad de Buenos Aires 1429, Argentina and CONICET, Av. Rivadavia 1917, Ciudad de Buenos Aires 1033 (Argentina); Comision Nacional de Energia Atomica, Av. del Libertador 8250, Ciudad de Buenos Aires 1429 (Argentina)
2011-12-15
Purpose: A rhodium self-powered neutron detector (Rh SPND) has been specifically developed by the Comision Nacional de Energia Atomica (CNEA) of Argentina to measure locally and in real time thermal neutron fluxes in patients treated with boron neutron capture therapy (BNCT). In this work, the thermal and epithermal neutron response of the Rh SPND was evaluated by studying the detector response to two different reactor spectra. In addition, during clinical trials of the BNCT Project of the CNEA, on-line neutron flux measurements using the specially designed detector were assessed. Methods: The first calibration of the detector was done with the well-thermalized neutron spectrum of the CNEA RA-3 reactor thermal column. For this purpose, the reactor spectrum was approximated by a Maxwell-Boltzmann distribution in the thermal energy range. The second calibration was done at different positions along the central axis of a water-filled cylindrical phantom, placed in the mixed thermal-epithermal neutron beam of CNEA RA-6 reactor. In this latter case, the RA-6 neutron spectrum had been well characterized by both calculation and measurement, and it presented some marked differences with the ideal spectrum considered for SPND calibrations at RA-3. In addition, the RA-6 neutron spectrum varied with depth in the water phantom and thus the percentage of the epithermal contribution to the total neutron flux changed at each measurement location. Local (one point-position) and global (several points-positions) and thermal and mixed-field thermal neutron sensitivities were determined from these measurements. Thermal neutron flux was also measured during BNCT clinical trials within the irradiation fields incident on the patients. In order to achieve this, the detector was placed on patient's skin at dosimetric reference points for each one of the fields. System stability was adequate for this kind of measurement. Results: Local mixed-field thermal neutron sensitivities and
International Nuclear Information System (INIS)
Miller, Marcelo E.; Sztejnberg, Manuel L.; Gonzalez, Sara J.; Thorp, Silvia I.; Longhino, Juan M.; Estryk, Guillermo
2011-01-01
Purpose: A rhodium self-powered neutron detector (Rh SPND) has been specifically developed by the Comision Nacional de Energia Atomica (CNEA) of Argentina to measure locally and in real time thermal neutron fluxes in patients treated with boron neutron capture therapy (BNCT). In this work, the thermal and epithermal neutron response of the Rh SPND was evaluated by studying the detector response to two different reactor spectra. In addition, during clinical trials of the BNCT Project of the CNEA, on-line neutron flux measurements using the specially designed detector were assessed. Methods: The first calibration of the detector was done with the well-thermalized neutron spectrum of the CNEA RA-3 reactor thermal column. For this purpose, the reactor spectrum was approximated by a Maxwell-Boltzmann distribution in the thermal energy range. The second calibration was done at different positions along the central axis of a water-filled cylindrical phantom, placed in the mixed thermal-epithermal neutron beam of CNEA RA-6 reactor. In this latter case, the RA-6 neutron spectrum had been well characterized by both calculation and measurement, and it presented some marked differences with the ideal spectrum considered for SPND calibrations at RA-3. In addition, the RA-6 neutron spectrum varied with depth in the water phantom and thus the percentage of the epithermal contribution to the total neutron flux changed at each measurement location. Local (one point-position) and global (several points-positions) and thermal and mixed-field thermal neutron sensitivities were determined from these measurements. Thermal neutron flux was also measured during BNCT clinical trials within the irradiation fields incident on the patients. In order to achieve this, the detector was placed on patient's skin at dosimetric reference points for each one of the fields. System stability was adequate for this kind of measurement. Results: Local mixed-field thermal neutron sensitivities and global
Thermal neutron flux distribution in ET-RR-2 reactor thermal column
Directory of Open Access Journals (Sweden)
Imam Mahmoud M.
2002-01-01
Full Text Available The thermal column in the ET-RR-2 reactor is intended to promote a thermal neutron field of high intensity and purity to be used for following tasks: (a to provide a thermal neutron flux in the neutron transmutation silicon doping, (b to provide a thermal flux in the neutron activation analysis position, and (c to provide a thermal neutron flux of high intensity to the head of one of the beam tubes leading to the room specified for boron thermal neutron capture therapy. It was, therefore, necessary to determine the thermal neutron flux at above mentioned positions. In the present work, the neutron flux in the ET-RR-2 reactor system was calculated by applying the three dimensional diffusion depletion code TRITON. According to these calculations, the reactor system is composed of the core, surrounding external irradiation grid, beryllium block, thermal column and the water reflector in the reactor tank next to the tank wall. As a result of these calculations, the thermal neutron fluxes within the thermal column and at irradiation positions within the thermal column were obtained. Apart from this, the burn up results for the start up core calculated according to the TRITION code were compared with those given by the reactor designer.
International Nuclear Information System (INIS)
Alzergy, A.A.; Emara, N.M.; Abd El-Latif, A.A.; El-Saady, S.M.M.; Emara, N.M.; Abd El-Latif, A.A.
2010-01-01
This work was performed to investigate the biological effects of different thermal neutron fluxes (0.27x10 8 , 0.52X10 8 , 1.089X10 8 , 2.16X10 8 and 4.32X10 8 ) on liver and kidney of male mice using neutron irradiation cell with Ra-Be(α,n) 3 mCi neutron source Leybold (55930). Exposed to various fluxes of thermal neutron induced a dramatic alterations in hepatic and renal functions as indicated by biochemical estimation of several parameters (bilirubin, SGT, and alkaline phosphate .Urea , total protein, and albumin) and confirmed by histological examinations Thermal neutron exposure induces marked increase in the serum activities of total bilirubin, alanine amino transaminase (ALT or GPT), and alkaline phosphate, whereas, urea, total protein and albumin showed marked decline as compared to control group. The physiological changes induced in thermal neutron fluxes dependent manner. Histopathological results revealed mild to severe type of necrosis, and degenerative changes in liver and kidney of male mice exposed to thermal neutron fluxes. Also it was found that the histopathological alterations induced in thermal neutron fluxes dependent manner. It was found that exposed to thermal neutron fluxes irradiation plays prominent role in the development of the physiological alterations in male Swiss albino mice. The Former up normalities as a result of the sequence events followed interaction of radiation with the former biological mater (liver and kidney) of male Swiss albino mice, which are, physical, physicochemical, chemical, and biological stages.
Effect of time dependence of neutron flux on the plutonium 238 production
International Nuclear Information System (INIS)
Rudik, A.P.
1975-01-01
An analytical treatment is given of the dependence of a plutonium-238 yield when irradiating neptunium-237 on the time variation of the neutron flux. This dependence is governed by the two physical factors: the competition between the beta decay and the neptunium-238 depletion, and the differences in the depletion rate of neptunium-237, neptunium-238 and plutonium-238 with thermal and resonance neutrons. The role of the neptunium-238 nonprompt decay has been studied and the order of the effect determined, and it has been established that it is advantageous to irradiate neptunium-237 in the lowest central neutron flux possible for thermal and resonance neutron reactors. A perturbation theory has been developed to determine the role of the time variation of the thermal neutron flux and it has been established that, the integral neutron flux being the same, it is advantageous to irradiate the material in a drooping rather than a constant flux of neutrons. The simplest example of great time variations of a neutron flux has been considered and it has been established that in this particular case the conclusions made in the framework of the perturbation theory find a qualitative support. Convenient formulas are given for determining the number of neutrons used to produce plutonium-238
Energy Technology Data Exchange (ETDEWEB)
Hathout, A M [National Center for Nuclear Safety and Radiation Control, NC-NSRC, Atomic Energy Authority, Cairo (Egypt)
1996-03-01
The narrow resonance approximation is applicable for all low-energy resonances and the heaviest nuclides. It is of great importance in neutron calculations, hence, fertile isotopes do not undergo fission at resonance energies. The effect of overestimating the self shielded group averaged cross-section data for a given resonance nuclide can be fairly serious. In the present work, a detailed study, and derivation of the problem of self-shielding are carried-out through the information of Hansen-roach library which is used for criticality safety analysis. The intermediate neutron flux spectrum is analyzed, using the narrow resonance approximation. The resonance self-shielded values of various cross-sections are determined. 4 figs., 3 tabs.
International Nuclear Information System (INIS)
Hoogenboom, J.E.
1981-01-01
An adjoint Monte Carlo technique is described for the solution of neutron transport problems. The optimum biasing function for a zero-variance collision estimator is derived. The optimum treatment of an analog of a non-velocity thermal group has also been derived. The method is extended to multiplying systems, especially for eigenfunction problems to enable the estimate of averages over the unknown fundamental neutron flux distribution. A versatile computer code, FOCUS, has been written, based on the described theory. Numerical examples are given for a shielding problem and a critical assembly, illustrating the performance of the FOCUS code. 19 refs
Applications of neutrons for laboratory and industrial activation analysis problems
International Nuclear Information System (INIS)
Szabo, Elek; Bakos, Laszlo
1986-01-01
This chapter presents some particular applications and case studies of neutrons in activation analysis for research and industrial development purposes. The reactor neutrons have been applied in Hungarian laboratories for semiconductor research, for analysis of geological (lunar) samples, and for a special comparator measurement of samples. Some industrial applications of neutron generator and sealed sources for analytical problems are presented. Finally, prompt neutron activation analysis is outlined briefly. (R.P.)
Can a large neutron excess help solve the baryon loading problem in gamma-Ray burst fireballs?
Fuller; Pruet; Abazajian
2000-09-25
We point out that the baryon loading problem in gamma-ray burst (GRB) models can be ameliorated if a significant fraction of the baryons which inertially confine the fireball is converted to neutrons. A high neutron fraction can result in a reduced transfer of energy from relativistic light particles in the fireball to baryons. The energy needed to produce the required relativistic flow in the GRB is consequently reduced, in some cases by orders of magnitude. A high neutron-to-proton ratio has been calculated in neutron star-merger fireball environments. Significant neutron excess also could occur near compact objects with high neutrino fluxes.
International Nuclear Information System (INIS)
Khan, M.J.; Aslam; Ahmad, N.; Ahmed, R.; Ahmad, S.I.
2005-01-01
The neutron moderation level in a nuclear reactor has a strong influence on core multiplication, reactivity control, fuel burnup, neutron fluxes etc. In the study presented in this article, the effects of neutron moderation level on core reactivity and neutron fluxes in a typical heavy water moderated nuclear research reactor is explored and the results are discussed. (author)
Measuring planetary neutron albedo fluxes by remote gamma-ray sensing
International Nuclear Information System (INIS)
Haines, E.L.; Metzger, A.E.
1984-01-01
A remote-sensing γ-ray spectrometer (GRS) is capable of measuring planetary surface composition through the detection of characteristic gamma rays. In addition, the planetary neutron leakage flux may be detected by means of a thin neutron absorber surrounding the γ-ray detector which converts the neutron flux into a γ-ray flux having a unique energy signature. The γ rays representing the neutron flux are observed against interference consisting of cosmic γ rays, planetary continuum and line emission, and a variety of gamma rays arising from cosmic-ray particle interactions with the γ-ray spectrometer and spacecraft (SC). In this paper the amplitudes of planetary and non-planetary neutron fluxes are assessed and their impact on the sensitivity of measurement is calculated for a lunar orbiter mission and a comet nucleus rendezvous mission. For a 100 h observation period from an altitude of 100 km, a GRS on a lunar orbiter can detect a thermal neutron albedo flux as low as 0.002 cm -2 s -1 and measure the expected flux of approx.=0.6 cm -2 s -1 with an uncertainty of 0.001 cm -2 s -1 . A GRS rendezvousing with a comet at a distance equal to the radius of the comet's nucleus, again for a 100 h observation time, should detect a thermal neutron albedo flux at a level of 0.006 cm -2 s -1 and measure the expected flux of approx.=0.4 cm -2 s -1 with an uncertainty of 0.004 cm -2 s -1 . Mapping the planetary neutron flux jointly with the direct detection of H will not only provide a more accurate model for translating observed γ-ray fluxes into concentrations but will also extend the effective sampling depth and should provide a capability for simple stratigraphic modeling of hydrogen. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Zanette, Rodrigo [Universidade Federal do Rio Grande do Sul (UFRGS), Porto Alegre, RS (Brazil). Programa de Pós-Graduação em Matemática Aplicada; Petersen, Claudio Z.; Tavares, Matheus G., E-mail: rodrigozanette@hotmail.com, E-mail: claudiopetersen@yahoo.com.br, E-mail: matheus.gulartetavares@gmail.com [Universidade Federal de Pelotas (UFPEL), RS (Brazil). Programa de Pós-Graduação em Modelagem Matemática
2017-07-01
We describe in this work the application of the modified power method for solve the multigroup neutron diffusion eigenvalue problem in slab geometry considering two-dimensions for nuclear reactor global calculations. It is well known that criticality calculations can often be best approached by solving eigenvalue problems. The criticality in nuclear reactors physics plays a relevant role since establishes the ratio between the numbers of neutrons generated in successive fission reactions. In order to solve the eigenvalue problem, a modified power method is used to obtain the dominant eigenvalue (effective multiplication factor (K{sub eff})) and its corresponding eigenfunction (scalar neutron flux), which is non-negative in every domain, that is, physically relevant. The innovation of this work is solving the neutron diffusion equation in analytical form for each new iteration of the power method. For solve this problem we propose to apply the Finite Fourier Sine Transform on one of the spatial variables obtaining a transformed problem which is resolved by well-established methods for ordinary differential equations. The inverse Fourier transform is used to reconstruct the solution for the original problem. It is known that the power method is an iterative source method in which is updated by the neutron flux expression of previous iteration. Thus, for each new iteration, the neutron flux expression becomes larger and more complex due to analytical solution what makes propose that it be reconstructed through an polynomial interpolation. The methodology is implemented to solve a homogeneous problem and the results are compared with works presents in the literature. (author)
An alternative method for the measurement of neutron flux
Indian Academy of Sciences (India)
dose rate. We have used a 241AmBe neutron source for neutron irradiation, and the neutron dose rate and count rate were ... neutron sources, e.g., for the characterization of superheated droplet detectors (SDD). [1–6]. The SDD is a .... Grants Commission (UGC) for the financial assistance provided for this work. References.
Neutron Flux Characterization of Irradiation Holes for Irradiation Test at HANARO
Directory of Open Access Journals (Sweden)
Yang Seong Woo
2016-01-01
Full Text Available The High flux Advanced Neutron Application ReactOr (HANARO is a unique research reactor in the Republic of Korea, and has been used for irradiation testing since 1998. To conduct irradiation tests for nuclear materials, the irradiation holes of CT and OR5 have been used due to a high fast-neutron flux. Because the neutron flux must be accurately calculated to evaluate the neutron fluence of irradiated material, it was conducted using MCNP. The neutron flux was measured using fluence monitor wires to verify the calculated result. Some evaluations have been conducted, however, more than 20% errors have frequently occurred at the OR irradiation hole, while a good agreement between the calculated and measured data was shown at the CT irradiation hole.
Energy Technology Data Exchange (ETDEWEB)
Raisic, N [Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Serbia and Montenegro)
1961-12-15
This report includes the following separate parts: Thermal neutron flux in the experimental channels od RA reactor; Epithermal neutron flux in the experimental channels od RA reactor; Fast neutron flux in the experimental channels od RA reactor; Thermal neutron flux in the thermal column and biological experimental channel; Neutronic measurements in the RA reactor cell; Temperature reactivity coefficient of the RA reactor; design of the device for measuring the activity of wire. [Serbo-Croat] Ovaj izvestaj sadrzi sledece referate: Fluks termalnih neutrona u eksperimentalnim kanalima reaktora RA; Fluks epitermalnih neutrona u eksperimentalnim kanalima reaktora RA; Fluks brzih neutrona u eksperimentalnim kanalima reaktora RA; Fluks termalnih neurona u termalnoj koloni i bioloskom eksperimentalnom kanalu; Neutronska merenja u elementarnoj celiji reaktora RA; Temperaturni koeficijent reaktivnosti reaktora RA; Projekat uredjaja za merenje radioaktivnosti zice.
Determination of neutron flux densities in WWR-S reactor core
International Nuclear Information System (INIS)
Tomasek, F.
1989-04-01
The method is described of determining neutron flux densities and neutron fluences using activation detectors. The basic definitions and relations for determining reaction rates, fluence and neutron flux as well as the characteristics of some reactions and of sitable activation detectors are reported. The flux densities were determined of thermal and fast neutrons and of gamma quanta in the WWR-S reactor core. The data measured in the period 1984-1987 are tabulated. Cross sections for the individual reactions were determined from spectra measurements processed using program SAND-II and cross section library ENDF-B IV. Neutron flux densities were also measured for the WWR-S reactor vertical channels. (E.J.). 10 figs., 8 tabs., 111 refs
Notes on neutron flux measurement; Notas sobre medida de flujos neutronicos
Energy Technology Data Exchange (ETDEWEB)
Alcala Ruiz, F
1984-07-01
The main purpose of this work is to get an useful guide to carry out topical neutron flux measurements. Although the foil activation technique is used in the majority of the cases, other techniques, such as those based on fission chambers and self-powered neutron detectors, are also shown. Special interest is given to the description and application of corrections on the measurement of relative and absolute induced activities by several types of detectors (scintillators, G-M and gas proportional counters). The thermal arid epithermal neutron fluxes, as determined in this work, are conventional or effective (West cots fluxes), which are extensively used by the reactor experimentalists; however, we also give some expressions where they are related to the integrated neutron fluxes, which are used in neutron calculations. (Author) 16 refs.
Energy Technology Data Exchange (ETDEWEB)
Koch, L; Labeyrie, J; Tarassenko, S [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1958-07-01
The detector described is designed for the instantaneous measurement of thermal neutron fluxes, in the presence of high {gamma} ray activity; this detector can withstand temperatures as high as 500 deg. C. It is based on the following principle: radioactive atoms resulting from heavy-nucleus fission are carried by a gas flow to a detector recording their {beta} and {gamma} disintegration. Thermal neutron fluxes as low as few neutrons per cm{sup 2} per second can be measured. This detector may be used to control a nuclear reactor, to plot the thermal flux distribution with an excellent definition (1 mm{sup 2}) for fluxes higher than 10{sup 8} n/cm{sup 2}/s. The time response of the system to a sharp variation of flux is limited, in case of large fluxes, to the transit time of the gas flow between the fission product emitter and the detector; of the order of one tenth of a sec per meter of piping. The detector may also be applied for spectroscopy of fission products eider than 0,1 s. (author)Fren. [French] On decrit un appareil permettant la mesure instantanee des flux de neutrons thermiques accompagnes de flux intenses de rayons {gamma} et situes dans des enceintes pouvant etre portees a des temperatures superieures a 500 deg. C. On utilise la radioactivite des atomes resultant de la fission des noyaux lourds; ces atomes sont entraines par un courant gazeux vers un detecteur de radioactivite qui enregistre leurs desintegrations {beta} et {gamma}. On peut mesurer des flux partir de quelques neutrons thermiques par cm{sup 2} et par seconde. L'appareil permet de suivre la puissance d'un reacteur atomique, de tracer des cartes de densite de neutrons avec une tres bonne definition (1 mm{sup 2}) dans le cas de flux superieurs a 10{sup 8} cm{sup 2}/s. Le temps de reponse du systeme a une variation du flux de neutrons est limite, poes flux importants, par le temps de transit du gaz entre l'emetteur de produits de fission et le detecteur: soit quelques dizaines de
International Nuclear Information System (INIS)
Dale, Gregory E.; Gahl, John M.
2005-01-01
There are several potential uses for a high-flux thermal neutron source in both industrial and clinical applications. The viable commercial implementation of these applications requires a low-cost, high-flux thermal neutron generator suitable for installation in industrial and clinical environments. This paper describes the Monte Carlo for N-Particle modeling results of a high-flux thermal neutron source driven with an electron accelerator. An electron linear accelerator (linac), fitted with a standard X-ray converter, can produce high neutron yields in materials with low photonuclear threshold energies, such as D and 9 Be. Results indicate that a 10-MeV, 10-kW electron linac can produce on the order of 10 12 n/s in a heavy water photoneutron target. The thermal neutron flux in an unreflected heavy water target is calculated to be on the order of 10 10 n.cm -2 .s. The sensitivity of these answers to heavy water purity is also investigated, specifically the dilution of heavy water with light water. It is shown that the peak thermal neutron flux is not adversely effected by dilution up to a light water weight fraction of 35%
International Nuclear Information System (INIS)
Andrejsek, K.
1977-01-01
The calculation is made of the thermal neutron flux in the moderator and reflector by solving the neutron diffusion equation using the four-group theory. The correction for neutron absorption in the moderator was carried out using the perturbation theory. The calculation was carried out for four groups with the following energy ranges: the first group 2 MeV to 3 keV, the second group 3 keV to 5 eV, the third group 5 eV to 0.025 eV and the fourth group 0.025 eV. The values of the macroscopic cross section of capture and scattering, of the diffusion coefficient, the macroscopic cross section of the moderator, of the neutron age and the extrapolation length for the water-graphite moderator used in the calculations are given. The spatial distribution of the thermal neutron flux is graphically represented for graphite of a 30, 40, and 50 cm radius and for graphite of a 30 and 40 cm radius with a 10 cm water reflector; a graphic comparison is made of the distribution of the thermal neutron flux in water and in graphite, both 40 cm in radius. The system of graphite with reflector proved to be the best and most efficient system for raising the flux density of thermal neutrons. (J.P.)
Thermal problems on high flux beam lines
International Nuclear Information System (INIS)
Avery, R.T.
1983-09-01
Wiggler and undulator magnets can provide very intense photon flux densities to beam line components. This paper addresses some thermal/materials consequences due to such impingement. The LBL/Exxon/SSRL hybrid-wiggler Beam Line VI now nearing operation will be able to provide up to approx. 7 kW of total photon power at planned SPEAR operating conditions. The first masks are located at 6.5 meters from the source and may receive a peak power density (transverse to the beam) exceeding 20 kW/cm 2 . Significantly, this heat transfer rate exceeds that radiated from the sun's surface (7 kW/cm 2 ) and is comparable to that of welding torches. Clearing, cooling and configuration are of critical importance. Configurations for the first fixed mask, the movable mask, and the pivot mask on this beam line are presented together with considerations of thermal stress fatigue and of heat transfer by conduction to water-cooling circuits. Some preliminary information on heating of crystals and mirrors is also presented
Gamma-ray-spectroscopy following high-flux 14-MeV neutron activation
Energy Technology Data Exchange (ETDEWEB)
Williams, R.E.
1981-10-12
The Rotating Target Neutron Source (RTNS-I), a high-intensity source of 14-MeV neutrons at the Lawrence Livermore National Laboratory (LLNL), has been used for applications in activation analysis, inertial-confinement-fusion diagnostic development, and fission decay-heat studies. The fast-neutron flux from the RTNS-I is at least 50 times the maximum fluxes available from typical neutron generators, making these applications possible. Facilities and procedures necessary for gamma-ray spectroscopy of samples irradiated at the RTNS-I were developed.
Gamma-ray-spectroscopy following high-flux 14-MeV neutron activation
International Nuclear Information System (INIS)
Williams, R.E.
1981-01-01
The Rotating Target Neutron Source (RTNS-I), a high-intensity source of 14-MeV neutrons at the Lawrence Livermore National Laboratory (LLNL), has been used for applications in activation analysis, inertial-confinement-fusion diagnostic development, and fission decay-heat studies. The fast-neutron flux from the RTNS-I is at least 50 times the maximum fluxes available from typical neutron generators, making these applications possible. Facilities and procedures necessary for gamma-ray spectroscopy of samples irradiated at the RTNS-I were developed
Monte Carlo simulation study of the muon-induced neutron flux at LNGS
International Nuclear Information System (INIS)
Persiani, R.; Garbini, M.; Massoli, F.; Sartorelli, G; Selvi, M.
2011-01-01
Muon-induced neutrons are ultimate background for all the experiments searching for rare events in underground laboratories. Several measurements and simulations were performed concerning the neutron production and propagation but there are disagreements between experimental data and simulations. In this work we present our Monte-Carlo simulation study, based on Geant4, to estimate the muon-induced neutron flux at LNGS. The obtained integral flux of neutrons above 1 MeV is 2.31 x 10 -10 n/cm 2 /s.
International Nuclear Information System (INIS)
Shafii, Mohammad Ali; Meidianti, Rahma; Wildian,; Fitriyani, Dian; Tongkukut, Seni H. J.; Arkundato, Artoto
2014-01-01
Theoretical analysis of integral neutron transport equation using collision probability (CP) method with quadratic flux approach has been carried out. In general, the solution of the neutron transport using the CP method is performed with the flat flux approach. In this research, the CP method is implemented in the cylindrical nuclear fuel cell with the spatial of mesh being conducted into non flat flux approach. It means that the neutron flux at any point in the nuclear fuel cell are considered different each other followed the distribution pattern of quadratic flux. The result is presented here in the form of quadratic flux that is better understanding of the real condition in the cell calculation and as a starting point to be applied in computational calculation
Analysis of the neutron flux in an annular pulsed reactor by using finite volume method
Energy Technology Data Exchange (ETDEWEB)
Silva, Mário A.B. da; Narain, Rajendra; Bezerra, Jair de L., E-mail: mabs500@gmail.com, E-mail: narain@ufpe.br, E-mail: jairbezerra@gmail.com [Universidade Federal de Pernambuco (UFPE), Recife, PE (Brazil). Centro de Tecnologia e Geociências. Departamento de Energia Nuclear
2017-07-01
Production of very intense neutron sources is important for basic nuclear physics and for material testing and isotope production. Nuclear reactors have been used as sources of intense neutron fluxes, although the achievement of such levels is limited by the inability to remove fission heat. Periodic pulsed reactors provide very intense fluxes by a rotating modulator near a subcritical core. A concept for the production of very intense neutron fluxes that combines features of periodic pulsed reactors and steady state reactors was proposed by Narain (1997). Such a concept is known as Very Intense Continuous High Flux Pulsed Reactor (VICHFPR) and was analyzed by using diffusion equation with moving boundary conditions and Finite Difference Method with Crank-Nicolson formalism. This research aims to analyze the flux distribution in the Very Intense Continuous Flux High Pulsed Reactor (VICHFPR) by using the Finite Volume Method and compares its results with those obtained by the previous computational method. (author)
Energy Technology Data Exchange (ETDEWEB)
Shafii, Mohammad Ali, E-mail: mashafii@fmipa.unand.ac.id; Meidianti, Rahma, E-mail: mashafii@fmipa.unand.ac.id; Wildian,, E-mail: mashafii@fmipa.unand.ac.id; Fitriyani, Dian, E-mail: mashafii@fmipa.unand.ac.id [Department of Physics, Andalas University Padang West Sumatera Indonesia (Indonesia); Tongkukut, Seni H. J. [Department of Physics, Sam Ratulangi University Manado North Sulawesi Indonesia (Indonesia); Arkundato, Artoto [Department of Physics, Jember University Jember East Java Indonesia (Indonesia)
2014-09-30
Theoretical analysis of integral neutron transport equation using collision probability (CP) method with quadratic flux approach has been carried out. In general, the solution of the neutron transport using the CP method is performed with the flat flux approach. In this research, the CP method is implemented in the cylindrical nuclear fuel cell with the spatial of mesh being conducted into non flat flux approach. It means that the neutron flux at any point in the nuclear fuel cell are considered different each other followed the distribution pattern of quadratic flux. The result is presented here in the form of quadratic flux that is better understanding of the real condition in the cell calculation and as a starting point to be applied in computational calculation.
Analysis of the neutron flux in an annular pulsed reactor by using finite volume method
International Nuclear Information System (INIS)
Silva, Mário A.B. da; Narain, Rajendra; Bezerra, Jair de L.
2017-01-01
Production of very intense neutron sources is important for basic nuclear physics and for material testing and isotope production. Nuclear reactors have been used as sources of intense neutron fluxes, although the achievement of such levels is limited by the inability to remove fission heat. Periodic pulsed reactors provide very intense fluxes by a rotating modulator near a subcritical core. A concept for the production of very intense neutron fluxes that combines features of periodic pulsed reactors and steady state reactors was proposed by Narain (1997). Such a concept is known as Very Intense Continuous High Flux Pulsed Reactor (VICHFPR) and was analyzed by using diffusion equation with moving boundary conditions and Finite Difference Method with Crank-Nicolson formalism. This research aims to analyze the flux distribution in the Very Intense Continuous Flux High Pulsed Reactor (VICHFPR) by using the Finite Volume Method and compares its results with those obtained by the previous computational method. (author)
Energy Technology Data Exchange (ETDEWEB)
Patil, B.J., E-mail: bjp@physics.unipune.ernet.i [Department of Physics, University of Pune, Pune 411 007 (India); Chavan, S.T.; Pethe, S.N.; Krishnan, R. [SAMEER, IIT Powai Campus, Mumbai 400 076 (India); Dhole, S.D., E-mail: sanjay@physics.unipune.ernet.i [Department of Physics, University of Pune, Pune 411 007 (India)
2010-09-15
The 6 MeV race track microtron based pulsed neutron source has been designed specifically for the elemental analysis of short lived activation products, where the low neutron flux requirement is desirable. Electrons impinges on a e-{gamma} target to generate bremsstrahlung radiations, which further produces neutrons by photonuclear reaction in {gamma}-n target. The optimisation of these targets along with their spectra were estimated using FLUKA code. The measurement of neutron flux was carried out by activation of vanadium at different scattering angles. Angular distribution of neutron flux indicates that the flux decreases with increase in the angle and are in good agreement with the FLUKA simulation.
Study of variation of materials patients room's door related of neutron flux iradiation
Nirmalasari, Yuliana Dian; Suparmi, A.; Sardjono, Y.
2017-08-01
The treatment chamber of patients has been simulating with MCNPX Code. Optimation of simulation design of Irradiation chamber is corresponding to ISO standards for 30 MeV cyclotron generator. The simulation has used the variation of door's materials that was applied at treatment room's door. The variation of materials was Stainless Steel 202 and Pb, the thickness Pb and stainless steel 202 with the thickness were 2 cm, respectively. Neutron flux that was radiated to stainless steel 202 in the sequence was 3.34195 × 105 n . Cm-2 s-1 and 8.41568 × 104 n . Cm-2 s-1, while for Pb was 4.01349 × 105 n . Cm-2 s-1 and 2.58058 × 104 n . Cm-2 s-1. The further, neutron flux that was radiated to Pb and stainless steel 202 with the thickness were 4 cm in sequence was 4.00601 × 105 n . Cm-2 s-1 and 1.71713 × 104 n . Cm-2 s-1 for Pb, while for SS 202 was 3.09925 × 105 n . Cm-2 s-1. From this ratio we concluded that material Pb absorbed higher neutron flux than material Stainless Steel 202. On the other hand, the cost of Pb was more expensive than Stainless Steel 202. In addition, the material Stainless Steel 202 was obtaine more easily than the material Pb. There fore to overcome the economics problem, can try to build the door with stainless still 202 sheet and Pb sheet together. The further, the neutron dose with 2 cm of thickness was 7.69603 × 10-2 Gy and 2.10623 × 10-2 Gy for SS 202, while for Pb was 4.19444 × 10-2 Gy and 1.50581 × 10-2 Gy. While the neutron dose with 4 cm of thickness for SS 202 was 9.39602 × 10-2 Gy and for Pb was 4.46541 × 10-2 Gy and 1.50502 × 10-2 Gy. We recommend that this simulation should be further optimized.
Flux distribution in phantom for biomedical use of beam-type thermal neutrons
International Nuclear Information System (INIS)
Aoki, Kazuhiko; Kobayashi, Tooru; Kanda, Keiji; Kimura, Itsuro
1985-01-01
For boron neutron capture therapy, the thermal neutron beam is worth using as therapeutic neutron irradiation without useless and unfavorable exposure of normal tissues around tumor and for microanalysis system to measure ppm-order 10 B concentrations in tissue and to search for the location of the metastasis of tumor. In the present study, the thermal neutron flux distribution in a phantom, when beam-type thermal neutrons were incident on it, was measured at the KUR Neutron Guide Tube. The measurements were carried out by two different methods using indium foil. The one is an ordinary foil activation technique by using the 115 In(n, γ) 116m 1 In reactions, while the other is to detect γ-rays from the 115 In(n, γ) 116m 2 In reactions during neutron irradiations with a handy-type Ge detector. The calculations with DOT 3.5 were performed to examine thermal neutron flux in the phantom for various beam size and phantom size. The experimental and calculated results are in good agreement and it is shown that the second type measurement has a potential for practical application as a new monitoring system of the thermal neutron flux in a living body for boron neutron capture therapy. (author)
Energy Technology Data Exchange (ETDEWEB)
Marini, P., E-mail: marini@cenbg.in2p3.fr [CENBG, CNRS/IN2P3-Université de Bordeaux, Chemin du Solarium B.P. 120, 33175 Gradignan (France); Mathieu, L. [CENBG, CNRS/IN2P3-Université de Bordeaux, Chemin du Solarium B.P. 120, 33175 Gradignan (France); Acosta, L. [Instituto de Física, Universidad Nacional Autónoma de México, Apartado Postal 20-364, México D.F. 01000 (Mexico); Aïche, M.; Czajkowski, S.; Jurado, B.; Tsekhanovich, I. [CENBG, CNRS/IN2P3-Université de Bordeaux, Chemin du Solarium B.P. 120, 33175 Gradignan (France)
2017-01-01
Proton-recoil detectors (PRDs), based on the well known standard H(n,p) elastic scattering cross section, are the preferred instruments to perform precise quasi-absolute neutron flux measurements above 1 MeV. The limitations of using a single silicon detector as PRD at a continuous neutron beam facility are investigated, with the aim of extending such measurements to neutron energies below 1 MeV. This requires a systematic investigation of the background sources affecting the neutron flux measurement. Experiments have been carried out at the AIFIRA facility to identify these sources. A study on the role of the silicon detector thickness on the background is presented and an energy limit on the use of a single silicon detector to achieve a neutron flux precision better than 1% is given.
International Nuclear Information System (INIS)
Ljubenov, V.; Milosevic, M.
2003-01-01
A procedure for the neutron flux determination in a neutron field with an arbitrary energy spectrum, based on the using of standard methods for the measurement of irradiated foils activity and on the application of the SCALE-4.4a code system for averaged cross section calculation is described in this paper. Proposed procedure allows to include the energy spectrum of neutron flux reestablished in the location of irradiated foils and the resonance self-shielding effects in the foils also. Example application of this procedure is given for the neutron flux determination inside the neutron filter with boron placed in the centre of heavy water critical assembly RB at the Vinca Institute (author)
On line local measurement of thermal neutron flux on BNCT patient using SPND
International Nuclear Information System (INIS)
Miller, M.E.; Sztejnberg Goncalves-Carralves, M.L.; Gonzalez, S.J.
2006-01-01
The first on-line neutron flux measurement on a patient using a self-powered neutron detector (SPND) was assessed during the fourth clinical trial of the Boron Neutron Capture Therapy (BNCT) Project carried out at the National Atomic Energy Commission of Argentina (CNEA) and the medical center Angel H. Roffo. The SPND was specially developed and assembled for BNCT by CNEA. Its small size, 1 cm sensible length and 1.9 mm diameter, allowed performing a localized measurement. Since the treated tumors were cutaneous melanomas of nodular type, the SPND was located on the patient's skin. The patient was exposed to three different and consecutive fields and in each of them the SPND was used to measure local thermal neutron fluxes at selected dosimetric reference points. The values of the measured fluxes agreed with the ones estimated by calculation. This trial also demonstrated the usefulness of the SPND for assessing flux on-line. (author)
Design and fabrication of self-powered in-core neutron flux monitor assembly
International Nuclear Information System (INIS)
Chung, M.K.; Cho, S.W.; Kang, H.D.; Cho, K.K.; Cho, B.S.; Kang, S.S.
1980-01-01
This is the final report on the prototypical fabrication of an in-core neutron flux monitor detector assembly for a specific power reactor conducted by KAERI from July 1, 1978 to December 31, 1979. It is well known that power reactors require a large number of in-core neutron flux detector for reactor regulation and the structures of detector assemblies are different from reactor to reactor. Therefore, from the nature of this project, it should be noted here that the target model of the prototypical farbrication of an in-core neutron flux monitor detector assembly is a VFD-2 System for Wolsung CANDU. It is concluded that fabrication of in-core neutron flux monitor detector assembly for CANDU reactor is technically feasible and will bring economical benefit as much as 50 % of the unit price if they are fabricated in Korea by using partially materials which are available from local market. (author)
Proposal for a new method of reactor neutron flux distribution determination
Energy Technology Data Exchange (ETDEWEB)
Popic, V R [Institute of nuclear sciences Boris Kidric, Vinca, Beograd (Serbia and Montenegro)
1964-01-15
A method, based on the measurements of the activity produced in a medium flowing with variable velocity through a reactor, for the determination of the neutron flux distribution inside a reactor is considered theoretically (author)
Neutron-diffraction investigations of flux-lines in superconductors
Energy Technology Data Exchange (ETDEWEB)
Forgan, E M [Birmingham Univ. (United Kingdom); Lee, S L [Saint Andrews Univ. (United Kingdom); McKPaul, D [Warwick Univ., Coventry (United Kingdom); Mook, H A [Oak Ridge National Lab., TN (United States); Cubitt, R [Institut Max von Laue - Paul Langevin (ILL), 38 - Grenoble (France)
1997-04-01
SANS has proved an extremely useful tool for investigating flux-line structures within the bulk of superconductors. With high-T{sub c} materials, the scattered intensities are weak, but careful measurements are giving important new information about flux lattices, flux pinning and flux-lattice melting. (author). 10 refs.
Measurement and evaluation of fast neutron flux of CT and OR5 irradiation hole in HANARO
International Nuclear Information System (INIS)
Yang, Seong Woo; Choo, Kee Nam; Lee, Seung-Kyu; Kim, Yong Kyun
2012-01-01
The irradiation test has been conducted to evaluate the irradiation performance of many materials by a material capsule at HANARO. Since the fast neutron fluence above 1 MeV is important for the irradiation test of material, it must be measured and evaluated exactly at each irradiation hole. Therefore, a fast neutron flux was measured and evaluated by a 09M-02K capsule irradiated in an OR5 irradiation hole and a 10M-01K capsule irradiated in a CT irradiation hole. Fe, Ni, and Ti wires as the fluence monitor were used for the detection of fast neutron flux. Before the irradiation test, the neutron flux and spectrum was calculated for each irradiation hole using an MCNP code. After the irradiation test, the activity of the fluence monitor was measured by an HPGe detector and the reaction rate was calculated. For the OR5 irradiation hole, the radial difference of the fast neutron flux was observed from a calculated data due to the OR5 irradiation hole being located outside the core. Furthermore, a control absorber rod was withdrawn from the core as the increase of the irradiation time at the same irradiation cycle, so the distribution of neutron flux was changed from the beginning to the end of the cycle. These effects were considered to evaluate the fast neutron flux. Neutron spectrums of the CT and OR5 irradiation hole were adjusted by the measured data. The fluxes of a fast neutron above 1 MeV were compared with calculated and measured value. Although the maximum difference was shown at 18.48%, most of the results showed good agreement. (author)
Scattered Neutron Tomography Based on A Neutron Transport Inverse Problem
International Nuclear Information System (INIS)
William Charlton
2007-01-01
Neutron radiography and computed tomography are commonly used techniques to non-destructively examine materials. Tomography refers to the cross-sectional imaging of an object from either transmission or reflection data collected by illuminating the object from many different directions
Self-powered neutron and γ-ray flux detector
International Nuclear Information System (INIS)
Allan, C.J.
1983-01-01
According to the invention there is provided a self-powered neutron and γ-ray flux detector, comprising: a) an emitter core wire; b) an emitter outer layer around the core wire and of different metal thereto; c) a metal collector around the emitter core wire and the emitter outer layer; and d) dielectric insulation electrically insulating the emitter core wire and the emitter outer layer from the metal collector. The improvement comprises: a) the overall diameter of the emitter core wire and the emitter outer layer is at least of the order of 0.4 mm in diameter; b) the emitter outer layer covers only of the order of l0 percent of the order of 90 percent of the emitter core wire surface area and comprises at least one band around the emitter core wire and is of a thickness in the range of the order 0.02 mm to of the order of 0.07 mm; and c) the metal of the emitter core wire, the metal of the emitter outer layer, the metal of the metal collector, the overall diameter of the emitter core wire and the emitter outer layer and the surface area of the emitter core wire that is covered by the emitter outer layer are selected so that the detector has a prompt fraction in the range of the order of 90 percent to of the order of 96 percent and has a dynamic response which substantially matches the dynamic response of the power in the fuel of the nuclear reactor in which the detector is to be used
Energy Technology Data Exchange (ETDEWEB)
Cartier, J. [Commissariat a l' Energie Atomique et aux Energies Alternatives CEA, DAM, DIF, F-91297 Arpajon (France); Casoli, P. [Commissariat a l' Energie Atomique et aux Energies Alternatives CEA, DAM, Valduc, F-21120 Is sur Tille (France); Chappert, F. [Commissariat a l' Energie Atomique et aux Energies Alternatives CEA, DAM, DIF, F-91297 Arpajon (France)
2013-07-01
In this paper, we present calibration methods in order to estimate reactor neutron flux spectrum and its uncertainties by using integral activation measurements. These techniques are performed using Bayesian and MCMC framework. These methods are applied to integral activation experiments in the cavity of the CALIBAN reactor. We estimate the neutron flux and its related uncertainties. The originality of this work is that these uncertainties take into account measurements uncertainties, cross-sections uncertainties and model error. In particular, our results give a very good approximation of the total flux and indicate that neutron flux from MCNP simulation for energies above about 5 MeV seems to overestimate the 'real flux'. (authors)
Calculation of gamma-rays and fast neutrons fluxes with the program Mercure-4
International Nuclear Information System (INIS)
Baur, A.; Dupont, C.; Totth, B.
1978-01-01
The program MERCURE-4 evaluates gamma ray or fast neutron attenuation, through laminated or bulky three-dimensionnal shields. The method used is that of line of sight point attenuation kernel, the scattered rays being taken into account by means of build-up factors for γ and removal cross sections for fast neutrons. The integration of the point kernel over the range of sources distributed in space and energy, is performed by the Monte-Carlo method, with an automatic adjustment of the importance functions. Since it is operationnal the program MERCURE-4 has been intensively used for many various problems, for example: - the calculation of gamma heating in reactor cores, control rods and shielding screens, as well as in experimental devices and irradiation loops; - the evaluation of fast neutron fluxes and corresponding damage in structural materials of reactors (vessel steels...); - the estimation of gamma dose rates on nuclear instrumentation in the reactors, around the reactor circuits and around spent fuel shipping casks
Transport calculation of neutron flux distribution in reflector of PW reactor
International Nuclear Information System (INIS)
Remec, I.
1982-01-01
Two-dimensional transport calculation of the neutron flux and spectrum in the equatorial plain of PW reactor, using computer program DOT 3, is presented. Results show significant differences between neutron fields in which test samples and reactor vessel are exposed. (author)
Simulations of muon-induced neutron flux at large depths underground
International Nuclear Information System (INIS)
Kudryavtsev, V.A.; Spooner, N.J.C.; McMillan, J.E.
2003-01-01
The production of neutrons by cosmic-ray muons at large depths underground is discussed. The most recent versions of the muon propagation code MUSIC, and particle transport code FLUKA are used to evaluate muon and neutron fluxes. The results of simulations are compared with experimental data
Energy Technology Data Exchange (ETDEWEB)
Blaesius, Christoph; Herb, Joachim; Kuentzel, Matthias
2016-01-15
Aim of the project was to find possible explanations for the neutron flux fluctuations and their changes over the last decades in German PWR. Several models concerning thermal hydraulics, structural mechanics and neutron physics were evaluated. It was shown that up to now no models are available that could explain the observed phenomena. Future studies should focus on interdisciplinary coupling of different models.
Flux distribution by neutrons semi-conductors detectors during the startup of the EL4 reactor
International Nuclear Information System (INIS)
Fuster, S.; Tarabella, A.
1967-01-01
The Cea developed neutron semi-conductors detectors which allows a quasi-instantaneous monitoring of neutrons flux distribution, when placed in a reactor during the tests. These detectors have been experimented in the EL4 reactor. The experiment and the results are presented and compared with reference mappings. (A.L.B.)
Measurements of thermal and fast neutron fluxes at the TRIGA reactor
International Nuclear Information System (INIS)
Zerdin, F.; Grabovsek, Z.; Klinc, T.; Solinc, H.
1966-01-01
Gold foils were placed at different positions in the TRIGA reactor core and in the experimental devices. Absolute values of the thermal neutron flux at these positions were obtained by coincidence method. Preliminary fast neutron spectrum was measured by threshold detector and by 'Li 6 sandwich' detector. A short description of the applied method and obtained measurements results are included [sl
Fast solution of neutron diffusion problem by reduced basis finite element method
International Nuclear Information System (INIS)
Chunyu, Zhang; Gong, Chen
2018-01-01
Highlights: •An extremely efficient method is proposed to solve the neutron diffusion equation with varying the cross sections. •Three orders of speedup is achieved for IAEA benchmark problems. •The method may open a new possibility of efficient high-fidelity modeling of large scale problems in nuclear engineering. -- Abstract: For the important applications which need carry out many times of neutron diffusion calculations such as the fuel depletion analysis and the neutronics-thermohydraulics coupling analysis, fast and accurate solutions of the neutron diffusion equation are demanding but necessary. In the present work, the certified reduced basis finite element method is proposed and implemented to solve the generalized eigenvalue problems of neutron diffusion with variable cross sections. The order reduced model is built upon high-fidelity finite element approximations during the offline stage. During the online stage, both the k eff and the spatical distribution of neutron flux can be obtained very efficiently for any given set of cross sections. Numerical tests show that a speedup of around 1100 is achieved for the IAEA two-dimensional PWR benchmark problem and a speedup of around 3400 is achieved for the three-dimensional counterpart with the fission cross-sections, the absorption cross-sections and the scattering cross-sections treated as parameters.
International Nuclear Information System (INIS)
Chatani, Hiroshi
2001-05-01
The reactions of Au(n, γ) 198 Au and Ti(n, p) 47 or 48 Sc were used for the measurements of the thermal and epithermal (thermal + epithermal) and the fast neutron flux distributions, respectively. In the case of Hyd (Hydraulic conveyer), the thermal + epithermal and fast neutron flux distributions in the horizontal direction in the capsule are especially flat; the distortion of the fluxes are 0.6% and 5.4%, respectively. However, these neutron fluxes in the vertical direction are low at the top and high at the bottom of the capsule. These differences between the top and bottom are 14% for both distributions. On the other hand, in polyethylene capsules of Pn-1, 2, 3 (Pneumatic tubes Nos. 1, 2, 3), in contrast with Hyd, these neutron flux distributions in the horizontal direction have gradients of 8 - 18% per 2.5 cm diameter, and those on the vertical axis have a distortion of approximately 5%. The strength of the epithermal dE/E component relative to the neutron density including both thermal and epithermal neutrons, i.e., the epithermal index, for the hydraulic conveyer (Hyd) and pneumatic tube No.2 (Pn-2), in which the irradiation experiments can be achieved, are determined by the multiple foil activation method using the reactions of Au(n, γ) 198 Au and Co(n, γ) 60(m+g) Co. The epithermal index observed in an aluminum capsule of Hyd is 0.034-0.04, and the Westcott thermal neutron flux is 1.2x10 14 cm -2 sec -1 at approximately 1 cm above the bottom. The epithermal index in a Pn-2 polyethylene capsule was measured by not only the multiple foil activation method but also the Cd-ratio method in which the Au(n, γ) 198 Au reaction in a cadmium cover is also used. The epithermal index is 0.045 - 0.055, and the thermal neutron flux is 1.8x10 13 cm -2 sec -1 . (J.P.N.)
Solving the uncommon nuclear reactor core neutronics problems
International Nuclear Information System (INIS)
Vondy, D.R.; Fowler, T.B.
1983-01-01
Calculational procedures have been implemented for solving importance and higher harmonic neutronics problems. Solutions are obtained routinely to support analysis of reactor core performance, treating up to three space coordinates with the multigroup diffusion theory approximation to neutron transport. The techniques used and some of the calculational difficulties are discussed
Energy Technology Data Exchange (ETDEWEB)
Fourmentel, D.; Villard, J. F.; Lyoussi, A. [DEN Reactor Studies Dept., French Nuclear Energy and Alternative Energies Commission, CEA Cadarache, 13108 Saint Paul-Lez-Durance (France); Reynard-Carette, C. [Laboratoire Chimie Provence LCP UMR 6264, Univ. of Provence, Centre St. Jerome, 13397 Marseille Cedex 20 (France); Bignan, G.; Chauvin, J. P.; Gonnier, C.; Guimbal, P.; Malo, J. Y. [DEN Reactor Studies Dept., French Nuclear Energy and Alternative Energies Commission, CEA Cadarache, 13108 Saint Paul-Lez-Durance (France); Carette, M.; Janulyte, A.; Merroun, O.; Brun, J.; Zerega, Y.; Andre, J. [Laboratoire Chimie Provence LCP UMR 6264, Univ. of Provence, Centre St. Jerome, 13397 Marseille Cedex 20 (France)
2011-07-01
We study the combined analysis of nuclear measurements to improve the knowledge of the irradiation conditions in the experimental locations of the future Jules Horowitz Reactor (JHR). The goal of the present work is to measure more accurately neutron flux, photon flux and nuclear heating in the reactor. In a Material Testing Reactor (MTR), nuclear heating is a crucial parameter to design the experimental devices to be irradiated in harsh nuclear conditions. This parameter drives the temperature of the devices and of the samples. The numerical codes can predict this parameter but in-situ measurements are necessary to reach the expected accuracy. For this reason, one objective of the IN-CORE program [1] is to study the combined measurements of neutron and photon flux and their cross advanced interpretation. It should be reminded that both neutron and photon sensors are not totally selective as their signals are due to neutron and photon interactions. We intend to measure the neutron flux by three different kinds of sensors (Uranium Fission chamber, Plutonium Fission chamber and Self Powered Neutron Detector), the photon flux by two different sensors (Ionization chamber and Self Powered Gamma Detector) and the nuclear heating by two different ones (Differential calorimeter and Gamma Thermometer). For the same parameter, we expect that the use of different kinds of sensors will allow a better estimation of the aimed parameter by mixing different spectrum responses and different neutron and gamma contributions. An experimental test called CARMEN-1 is scheduled in OSIRIS reactor (CEA Saclay - France) at the end of 2011, with the goal to map irradiation locations in the reactor reflector to get a first validation of the analysis model. This article focuses on the sensor selection for CARMEN-1 experiment and to the way to link neutron and photon flux measurements in view to reduce their uncertainties but also to better assess the neutron and photon contributions to nuclear
International Nuclear Information System (INIS)
Fourmentel, D.; Villard, J. F.; Lyoussi, A.; Reynard-Carette, C.; Bignan, G.; Chauvin, J. P.; Gonnier, C.; Guimbal, P.; Malo, J. Y.; Carette, M.; Janulyte, A.; Merroun, O.; Brun, J.; Zerega, Y.; Andre, J.
2011-01-01
We study the combined analysis of nuclear measurements to improve the knowledge of the irradiation conditions in the experimental locations of the future Jules Horowitz Reactor (JHR). The goal of the present work is to measure more accurately neutron flux, photon flux and nuclear heating in the reactor. In a Material Testing Reactor (MTR), nuclear heating is a crucial parameter to design the experimental devices to be irradiated in harsh nuclear conditions. This parameter drives the temperature of the devices and of the samples. The numerical codes can predict this parameter but in-situ measurements are necessary to reach the expected accuracy. For this reason, one objective of the IN-CORE program [1] is to study the combined measurements of neutron and photon flux and their cross advanced interpretation. It should be reminded that both neutron and photon sensors are not totally selective as their signals are due to neutron and photon interactions. We intend to measure the neutron flux by three different kinds of sensors (Uranium Fission chamber, Plutonium Fission chamber and Self Powered Neutron Detector), the photon flux by two different sensors (Ionization chamber and Self Powered Gamma Detector) and the nuclear heating by two different ones (Differential calorimeter and Gamma Thermometer). For the same parameter, we expect that the use of different kinds of sensors will allow a better estimation of the aimed parameter by mixing different spectrum responses and different neutron and gamma contributions. An experimental test called CARMEN-1 is scheduled in OSIRIS reactor (CEA Saclay - France) at the end of 2011, with the goal to map irradiation locations in the reactor reflector to get a first validation of the analysis model. This article focuses on the sensor selection for CARMEN-1 experiment and to the way to link neutron and photon flux measurements in view to reduce their uncertainties but also to better assess the neutron and photon contributions to nuclear
Analysis of Neutron Flux Using Monte Carlo Methods
International Nuclear Information System (INIS)
Picha, Roppon
2007-08-01
Full text: The energy profile of neutrons from a fission reactor core and a neutron irradiation setup are simulated. The neutron doses deposited inside casings of aluminum, cadmium, and tantalum are studied via MCNP simulations to estimate the doses received by materials with different types of shielding. It is found that the difference in dose reduction between cadmium and tantalum is most pronounced at the thermal energy region
Mitigation of end flux peaking in CANDU fuel bundles using neutron absorbers
Energy Technology Data Exchange (ETDEWEB)
Pierce, D.; Chan, P.K., E-mail: dylan.pierce@rmc.ca [Royal Military College of Canada, Kingston ON, (Canada); Shen, W. [Canadian Nuclear Safety Commission, Ottawa ON, (Canada)
2015-07-01
End flux peaking (EFP) is a phenomenon where a region of elevated neutron flux occurs between two adjoining fuel bundles. These peaks lead to an increase in fission rate and therefore greater heat generation. It is known that addition of neutron absorbers into fuel bundles can help mitigate EFP, yet implementation in Canada Deuterium Uranium (CANDU) type reactors using natural uranium fuel has not been pursued. Monte Carlo N-Particle code (MCNP) 6.1 was used to simulate the addition of a small amount of neutron absorbers strategically within the fuel pellets. This paper will present some preliminary results collected thus far. (author)
Neutron flux measurement with 6Li and 7Li dual glass scintillators by γ compensation method
International Nuclear Information System (INIS)
Ji Changsong; Zhang Shulan; Zhang Shuheng
1996-01-01
Based on the characteristics of 6 Li glass scintillator which is sensitive to both neutron and gamma rays, and 7 Li glass scintillator which is sensitive to gamma rays only, a new method of detecting weak neutron flux under interference of strong gamma radiation has been investigated by means of 6 Li- 7 Li pair glass scintillator gamma compensation method. The result of neutron flux measurement by above-mentioned method with an error of about 1% when the gamma ray interference is up to 18.7% has been obtained
Neutron flux measurement with 6Li and 7Li dual glass scintillators by γ compensation method
International Nuclear Information System (INIS)
Ji Changsong; Zhang Shulan; Zhang Shuheng
1998-01-01
Based on the characteristics of 6 Li glass scintillator which is sensitive to both neutron and gamma rays, and 7 Li glass scintillator which is sensitive to gamma rays only, a new method of detecting weak neutron flux under interference of strong gamma radiation has been investigated by mans of 6 Li- 7 Li dual glass scintillator gamma compensation method. The result of neutron flux measurement by above-mentioned method with an error of about 1% when the gamma ray interference is up to 18.7% has been obtained
International Nuclear Information System (INIS)
Bertalot, L.
1987-01-01
A feasibility study has been carried out on the use of in-core fission chambers as neutron diagnostic tools to perform neutron flux measurements on the blanket component of NET. The high neutron and gamma fluxes and the severe thermal-mechanical and magnetic conditions of the blanket structure have been taken into account in this analysis. Preliminary design criteria and specifications of an in-core detector are presented for NET application. A research and development programme is outlined which aims to obtain more information on the tecnological constraints arising from the severe conditions of the NET blanket
A Method for the neutron flux determination during the activation process
International Nuclear Information System (INIS)
Maayouf, R.M.A.; Khalil, M.I.
2000-01-01
The present work deals with an accurate method for determining the neutron flux coming out from a neutron source during the experimental measurements. Accordingly, a suitable detector, followed by preamplifier and amplifier, is connected to a data acquisition system designed specially for this purpose; and the number of neutrons detected during every sampling period is stored in the PC. The historical file can be used to compute the average or the integral flux during any time period; considering the detector efficiency, geometrical arrangement and the amplification gain
Modelisation and distribution of neutron flux in radium-beryllium source (226Ra-Be)
Didi, Abdessamad; Dadouch, Ahmed; Jai, Otman
2017-09-01
Using the Monte Carlo N-Particle code (MCNP-6), to analyze the thermal, epithermal and fast neutron fluxes, of 3 millicuries of radium-beryllium, for determine the qualitative and quantitative of many materials, using method of neutron activation analysis. Radium-beryllium source of neutron is established to practical work and research in nuclear field. The main objective of this work was to enable us harness the profile flux of radium-beryllium irradiation, this theoretical study permits to discuss the design of the optimal irradiation and performance for increased the facility research and education of nuclear physics.
Measurement of spectra and neutron fluxes on artificial earth satellites from the Cosmos series
Dudkin, V. Y.; Kovalev, Y. Y.; Novikova, M. R.; Potapov, Y. V.; Skvortsov, S. S.; Smirennyy, L. N.
1975-01-01
In 1966-1967 measurements were carried out at the altitudes of 200 to 400 km to determine the spectra and fluxes of fast neutrons inside the hermetically sealed artificial earth satellites of the Cosmos series. The detectors used were nuclear emulsions of the B9 and BR types and an emulsion of the P9 type, filled with Li and P. Spectra and fluxes of neutrons in the range of energies from thermal energies to 10 MeV are presented. Neutron doses are also estimated.
Measurement of the fast neutron flux in the MNSR inner irradiation site
International Nuclear Information System (INIS)
Khattab, K.
2007-01-01
The WIMSD4 code was used to calculate the fast neutron flux spectrum and the fast neutron fission cross sections for 238 U, using six energy groups ranging from 0.5 to 10 MeV. These results, with the measured radioactivities of the 140 Ba, 131 I, 103 Ru, 95 Zr and 97 Zr fission products emerging from the fission of the 238 U foil covered with a cadmium filter, were used to measure the fast neutron flux in the Syrian MNSR inner irradiation site. (author)
International Nuclear Information System (INIS)
Diaz Rizo, O.; Herrera Peraza, E.
1996-01-01
The non ideality of the epithermal neutron flux distribution at a reactor site parameter (made, using Chloramine-T method. Radiochemical purity and stability of the labelled product were determined by radiochromatography. The labelled Melagenine-II showed two radioactive fractions thermal-to-epithermal neutron ratio (f) were determined in the 3 typical irradiations positions of the TRIGA Mark III reactor of the National Nuclear Research Institute, Salazar, Mexico, using the Cd-ratio for multi monitor and bare bi-isotopic monitor methods respectively. This characterization is of use in the K o - method of neutron activation analysis, recently introduced at the Institute
Energy Technology Data Exchange (ETDEWEB)
Weill, J; Duchene, J P [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1958-07-01
The ionisation chamber described is designed for the measurement of slow neutron fluxes of average or low intensity, in the presence, eventually, of very high gamma fluxes. The capture of a slow neutron by a fissile material, in this case {sup 235}U, gives rise to fission fragments, high-energy particles which ionise the gas contained in the chamber. The neutrons are detected by virtue of the potential pulses, on the collecting electrode of the chamber, deriving from the collection of the ions produced by the fission fragments. The pulses are counted by means of a measuring system consisting of a preamplifier, a 2 Mc amplifier, a discriminator and an electronic scale with numerator or integrator. The general characteristics are as follows: sensitivity to neutrons: 0.07 kicks/n/cm{sup 2}.s, sensitivity to {gamma} rays: zero up to 3.10{sup 4} R/H, a background noise at the normal discrimination voltage: 0.01 kicks/s, working H.T.: -500 V, capacity: 40 {mu}{mu}F, average height of pulse: 8 mV, limits of use: from several neutrons to 10{sup 6} n/cm{sup 2}.s. This chamber may be used in all cases where low fluxes of slow neutrons must be measured, especially in the presence of high gamma fluxes, for example in the checking of Pu concentrations in an extraction plant or for the starting up of reactors. (author) [French] La chambre d'ionisation decrite est destinee a la mesure des flux de neutrons lents d'intensite moyenne ou faible, en presence eventuelle de flux gamma tres importants. La capture d'un neutron lent par un materiau fissile, en l'occurrence {sup 235}U, donne naissance a des fragments de fission, particules de grande energie qui ionisent le gaz contenu dans la chambre. Les neutrons sont detectes grace aux impulsions de potentiel, sur l'electrode collectrice de la chambre, provenant de la collection des ions produits par les fragments de fission. Une chaine de mesure comprenant un preamplificateur, un amplificateur 2 Mc, un discriminateur, une echelle
Three new nondestructive evaluation tools based on high flux neutron sources
International Nuclear Information System (INIS)
Hubbard, C.R.; Raine, D.; Peascoe, R.; Wright, M.
1997-01-01
Nondestructive evaluation methods and systems based on specific attributes of neutron interactions with materials are being developed. The special attributes of neutrons are low attenuation in most engineering materials, strong interaction with low Z elements, and epithermal neutron absorption resonances. The three methods under development at ORNL include neutron based tomography and radiography; through thickness, nondestructive texture mapping; and internal, noninvasive temperature measurement. All three techniques require high flux sources such as the High Flux Isotope Reactor, a steady state source, or the Oak Ridge Electron Linear Accelerator, a pulsed neutron source. Neutrons are quite penetrating in most engineering materials and thus can be useful to detect internal flaws and features. Hydrogen atoms, such as in a hydrocarbon fuel, lubricant, or a metal hydride, are relatively opaque to neutron transmission and thus neutron based tomography/radiography is ideal to image their presence. Texture, the nonrandom orientation of crystalline grains within materials, can be mapped nondestructively using neutron diffraction methods. Epithermal neutron resonance absorption is being studied as a noncontacting temperature sensor. This paper highlights the underlying physics of the methods, progress in development, and the potential benefits for science and industry of the three facilities
Relative measurement of the fluxes of thermal, resonant and rapid neutrons in reactor G1
International Nuclear Information System (INIS)
Carle, R.; Mazancourt, T. de
1957-01-01
We sought to determine the behavior of the thermal, resonant and rapid neutron fluxes in the multiplier-reflector transition region, in the two principal directions of the system. We have also measured the variation of these different fluxes in the body of the multiplier medium in a canal filled with graphite and in an empty canal. The results are given in the form of curves representing: - the variation of the ratio of the thermal flux to the rapid flux in axial and radial transitions - the behavior of the thermal and resonant fluxes and the variation of their ratio in the same regions. (author) [fr
International Nuclear Information System (INIS)
Snoj, Luka; Ravnik, Matjaz; Lengar, Igor
2008-01-01
Modern Monte Carlo computer codes (e.g. MCNP) for neutron transport allow calculation of detailed neutron flux and power distribution in complex geometries with resolution of ∼1 mm. Moreover they enable the calculation of individual particle tracks, scattering and absorption events. With the use of advanced software for 3D visualization (e.g. Amira, Voxler, etc.) one can create and present neutron flux and power distribution in a 'user friendly' way convenient for educational purposes. One can view axial, radial or any other spatial distribution of the neutron flux and power distribution in a nuclear reactor from various perspectives and in various modalities of presentation. By visualizing the distribution of scattering and absorption events and individual particle tracks one can visualize neutron transport parameters (mean free path, diffusion length, macroscopic cross section, up-scattering, thermalization, etc.) from elementary point of view. Most of the people remember better, if they visualize the processes. Therefore the representation of the reactor and neutron transport parameters is a convenient modern educational tool for the (nuclear power plant) operators, nuclear engineers, students and specialists involved in reactor operation and design. The visualization of neutron flux and power distributions in Jozef Stefan Institute TRIGA Mark II research reactor is treated in the paper. The distributions are calculated with MCNP computer code and presented using Amira and Voxler software. The results in the form of figures are presented in the paper together with comments qualitatively explaining the figures. (authors)
Energy Technology Data Exchange (ETDEWEB)
Droulers, Y; Pleyber, G; Sciers, P; Maurin, G [Commissariat a l' Energie Atomique, Grenoble (France). Centre d' Etudes Nucleaires
1964-07-01
The measurement method described in this report can be applied both to thermal and fast neutron fluxes. A description is given of two practical applications in each of these two domains. This method is particularly suitable for measurements carried out on 'loop' type equipment. The measurement of the relative flux variations are carried out with an accuracy of 5 per cent. The choice of the shape of the gas circuit leaves a considerable amount of liberty for the adaptation of the measurement circuit to the experimental conditions. (authors) [French] La methode de mesure defrite dans ce rapport s1 applique aussi bien au flux de neutrons thermiques, qu'au flux de neutrons rapides. On donne la description de deux realisations pratiques dans chacun de ces domaines. Cette methode est particulierement adaptee a des mesures effectuees sur des dispositifs du type 'boucle'. La mesure des variations relatives de flux se fait avec une precision de 5 pour cent. Le choix de la configuration du circuit gazeux donne une grande souplesse dans l'adaptation du circuit de mesure aux conditions experimentales. (auteurs)
Intermediate neutron spectrum problems and the intermediate neutron spectrum experiment
International Nuclear Information System (INIS)
Jaegers, P.J.; Sanchez, R.G.
1996-01-01
Criticality benchmark data for intermediate energy spectrum systems does not exist. These systems are dominated by scattering and fission events induced by neutrons with energies between 1 eV and 1 MeV. Nuclear data uncertainties have been reported for such systems which can not be resolved without benchmark critical experiments. Intermediate energy spectrum systems have been proposed for the geological disposition of surplus fissile materials. Without the proper benchmarking of the nuclear data in the intermediate energy spectrum, adequate criticality safety margins can not be guaranteed. The Zeus critical experiment now under construction will provide this necessary benchmark data
High-accuracy determination of the neutron flux at n{sub T}OF
Energy Technology Data Exchange (ETDEWEB)
Barbagallo, M.; Colonna, N.; Mastromarco, M.; Meaze, M.; Tagliente, G.; Variale, V. [Sezione di Bari, INFN, Bari (Italy); Guerrero, C.; Andriamonje, S.; Boccone, V.; Brugger, M.; Calviani, M.; Cerutti, F.; Chin, M.; Ferrari, A.; Kadi, Y.; Losito, R.; Versaci, R.; Vlachoudis, V. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Tsinganis, A. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); National Technical University of Athens (NTUA), Athens (Greece); Tarrio, D.; Duran, I.; Leal-Cidoncha, E.; Paradela, C. [Universidade de Santiago de Compostela, Santiago (Spain); Altstadt, S.; Goebel, K.; Langer, C.; Reifarth, R.; Schmidt, S.; Weigand, M. [Johann-Wolfgang-Goethe Universitaet, Frankfurt (Germany); Andrzejewski, J.; Marganiec, J.; Perkowski, J. [Uniwersytet Lodzki, Lodz (Poland); Audouin, L.; Leong, L.S.; Tassan-Got, L. [Centre National de la Recherche Scientifique/IN2P3 - IPN, Orsay (France); Becares, V.; Cano-Ott, D.; Garcia, A.R.; Gonzalez-Romero, E.; Martinez, T.; Mendoza, E. [Centro de Investigaciones Energeticas Medioambientales y Tecnologicas (CIEMAT), Madrid (Spain); Becvar, F.; Krticka, M.; Kroll, J.; Valenta, S. [Charles University, Prague (Czech Republic); Belloni, F.; Fraval, K.; Gunsing, F.; Lampoudis, C.; Papaevangelou, T. [Commissariata l' Energie Atomique (CEA) Saclay - Irfu, Gif-sur-Yvette (France); Berthoumieux, E.; Chiaveri, E. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Commissariata l' Energie Atomique (CEA) Saclay - Irfu, Gif-sur-Yvette (France); Billowes, J.; Ware, T.; Wright, T. [University of Manchester, Manchester (United Kingdom); Bosnar, D.; Zugec, P. [University of Zagreb, Department of Physics, Faculty of Science, Zagreb (Croatia); Calvino, F.; Cortes, G.; Gomez-Hornillos, M.B.; Riego, A. [Universitat Politecnica de Catalunya, Barcelona (Spain); Carrapico, C.; Goncalves, I.F.; Sarmento, R.; Vaz, P. [Universidade Tecnica de Lisboa, Instituto Tecnologico e Nuclear, Instituto Superior Tecnico, Lisboa (Portugal); Cortes-Giraldo, M.A.; Praena, J.; Quesada, J.M.; Sabate-Gilarte, M. [Universidad de Sevilla, Sevilla (Spain); Diakaki, M.; Karadimos, D.; Kokkoris, M.; Vlastou, R. [National Technical University of Athens (NTUA), Athens (Greece); Domingo-Pardo, C.; Giubrone, G.; Tain, J.L. [CSIC-Universidad de Valencia, Instituto de Fisica Corpuscular, Valencia (Spain); Dressler, R.; Kivel, N.; Schumann, D.; Steinegger, P. [Paul Scherrer Institut, Villigen PSI (Switzerland); Dzysiuk, N.; Mastinu, P.F. [Laboratori Nazionali di Legnaro, INFN, Rome (Italy); Eleftheriadis, C.; Manousos, A. [Aristotle University of Thessaloniki, Thessaloniki (Greece); Ganesan, S.; Gurusamy, P.; Saxena, A. [Bhabha Atomic Research Centre (BARC), Mumbai (IN); Griesmayer, E.; Jericha, E.; Leeb, H. [Technische Universitaet Wien, Atominstitut, Wien (AT); Hernandez-Prieto, A. [European Organization for Nuclear Research (CERN), Geneva (CH); Universitat Politecnica de Catalunya, Barcelona (ES); Jenkins, D.G.; Vermeulen, M.J. [University of York, Heslington, York (GB); Kaeppeler, F. [Institut fuer Kernphysik, Karlsruhe Institute of Technology, Campus Nord, Karlsruhe (DE); Koehler, P. [Oak Ridge National Laboratory (ORNL), Oak Ridge (US); Lederer, C. [Johann-Wolfgang-Goethe Universitaet, Frankfurt (DE); University of Vienna, Faculty of Physics, Vienna (AT); Massimi, C.; Mingrone, F.; Vannini, G. [Universita di Bologna (IT); INFN, Sezione di Bologna, Dipartimento di Fisica, Bologna (IT); Mengoni, A.; Ventura, A. [Agenzia nazionale per le nuove tecnologie, l' energia e lo sviluppo economico sostenibile (ENEA), Bologna (IT); Milazzo, P.M. [Sezione di Trieste, INFN, Trieste (IT); Mirea, M. [Horia Hulubei National Institute of Physics and Nuclear Engineering - IFIN HH, Bucharest - Magurele (RO); Mondalaers, W.; Plompen, A.; Schillebeeckx, P. [Institute for Reference Materials and Measurements, European Commission JRC, Geel (BE); Pavlik, A.; Wallner, A. [University of Vienna, Faculty of Physics, Vienna (AT); Rauscher, T. [University of Basel, Department of Physics and Astronomy, Basel (CH); Roman, F. [European Organization for Nuclear Research (CERN), Geneva (CH); Horia Hulubei National Institute of Physics and Nuclear Engineering - IFIN HH, Bucharest - Magurele (RO); Rubbia, C. [European Organization for Nuclear Research (CERN), Geneva (CH); Laboratori Nazionali del Gran Sasso dell' INFN, Assergi (AQ) (IT); Weiss, C. [European Organization for Nuclear Research (CERN), Geneva (CH); Johann-Wolfgang-Goethe Universitaet, Frankfurt (DE)
2013-12-15
The neutron flux of the n{sub T}OF facility at CERN was measured, after installation of the new spallation target, with four different systems based on three neutron-converting reactions, which represent accepted cross sections standards in different energy regions. A careful comparison and combination of the different measurements allowed us to reach an unprecedented accuracy on the energy dependence of the neutron flux in the very wide range (thermal to 1 GeV) that characterizes the n{sub T}OF neutron beam. This is a pre-requisite for the high accuracy of cross section measurements at n{sub T}OF. An unexpected anomaly in the neutron-induced fission cross section of {sup 235}U is observed in the energy region between 10 and 30keV, hinting at a possible overestimation of this important cross section, well above currently assigned uncertainties. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Yavar, A.R. [School of Applied Physics, Faculty of Science and Technology, National University of Malaysia (UKM), 43600 Bangi, Selangor (Malaysia); Sarmani, S.B. [School of Chemical Sciences and Food Technology, Faculty of Science and Technology, National University of Malaysia (UKM), 43600 Bangi, Selangor (Malaysia); Wood, A.K. [Analytical Chemistry Application Group, Industrial Technology Division, Malaysian Nuclear Agency (MNA), Bangi, 43000 Kajang, Selangor (Malaysia); Fadzil, S.M. [School of Applied Physics, Faculty of Science and Technology, National University of Malaysia (UKM), 43600 Bangi, Selangor (Malaysia); Radir, M.H. [Analytical Chemistry Application Group, Industrial Technology Division, Malaysian Nuclear Agency (MNA), Bangi, 43000 Kajang, Selangor (Malaysia); Khoo, K.S., E-mail: khoo@ukm.m [School of Applied Physics, Faculty of Science and Technology, National University of Malaysia (UKM), 43600 Bangi, Selangor (Malaysia)
2011-05-15
Determination of thermal to fast neutron flux ratio (f{sub fast}) and fast neutron flux ({phi}{sub fast}) is required for fast neutron reactions, fast neutron activation analysis, and for correcting interference reactions. The f{sub fast} and subsequently {phi}{sub fast} were determined using the absolute method. The f{sub fast} ranged from 48 to 155, and the {phi}{sub fast} was found in the range 1.03x10{sup 10}-4.89x10{sup 10} n cm{sup -2} s{sup -1}. These values indicate an acceptable conformity and applicable for installation of the fast neutron facility at the MNA research reactor.
Flux form Semi-Lagrangian methods for parabolic problems
Directory of Open Access Journals (Sweden)
Bonaventura Luca
2016-09-01
Full Text Available A semi-Lagrangian method for parabolic problems is proposed, that extends previous work by the authors to achieve a fully conservative, flux-form discretization of linear and nonlinear diffusion equations. A basic consistency and stability analysis is proposed. Numerical examples validate the proposed method and display its potential for consistent semi-Lagrangian discretization of advection diffusion and nonlinear parabolic problems.
Design and construction of a 10B coated ion chamber for the measurement of a thermal neutron flux
International Nuclear Information System (INIS)
Vite T, M.
1979-01-01
A model of an ionization chamber cover with 10 B, for the measurement of thermal neutron flux is presented, the developed chamber is made of a cylindrical vessel in which interior exist 7 aluminium electrodes which by electrodeposition were covered with a coat of 10 B with a thickness of 0.68 mg/cm 2 . Once the chamber is filled with N 2 to a pressure of 1.33 atm., it was exposed to a flux of thermal neutrons at the order 10 4 n/cm 2 -seg obtaining an ionization current of 10 -11 amp. which can be easily distinguished from the leak current which order is of 10 -12 amp. The conventional electronics associated to the chamber allows to process the ionization current in pulse form. The relation of noise to signal is approximately of 8, for which the pulses can be differentiated from the electronic noise without any problem. (author)
International Nuclear Information System (INIS)
Klix, A.; Batistoni, P.; Boettger, R.; Lebrun-Grandie, D.; Fischer, U.; Henniger, J.; Leichtle, D.; Villari, R.
2010-01-01
Fast neutron and gamma-ray flux spectra and time-of-arrival spectra of slow neutrons have been measured in a neutronics mock-up of the European Helium-Cooled Lithium-Lead Test Blanket Module with the aim to validate nuclear cross-section data. The mock-up was irradiated with fusion peak neutrons from the DT neutron generator of the Technical University of Dresden. A well characterized cylindrical NE-213 scintillator was inserted into two positions in the LiPb/EUROFER assembly. Pulse height spectra from neutrons and gamma-rays were recorded from the NE-213 output. The spectra were then unfolded with experimentally obtained response matrices of the NE-213 detector. Time-of-arrival spectra of slow neutrons were measured with a 3 He counter placed in the mock-up, and the neutron generator was operated in pulsed mode. Monte Carlo calculations using the MCNP code and nuclear cross-section data from the JEFF-3.1.1 and FENDL-2.1 libraries were performed and the results are compared with the experimental results. A good agreement of measurement and calculation was found with some deviations in certain energy intervals.
The increase of the subthermal neutron flux by using a cold neutron source at the FRG-1
International Nuclear Information System (INIS)
Krueger, A.; Turgut, M.H.
1986-01-01
The increase of the subthermal neutron flux (wavelength range 4-6 A by a cold neutron source (CNS) at a radial beam tube of the FRG-1 reactor is investigated in combination with different reflectors (H 2 O, C, Be, D 2 O). Advantage factors on the basis of the directed neutron flux, resulting from the use of the CNS, are calculated for various configurations. In addition, the influence of different scattering models (gas, Koppel/Young) for the CNS, group structure, and structural materials are described. Finally, the CNS assembly which is going to be installed at the FRG-1 is treated in detail. For the calculations the transport code NEUTRA and the spectral code GGC-4 are used. (orig.) [de
Vermeeren, L.; Wéber, M.
2003-06-01
A set of ten Self-Powered Neutron Detectors with Co, Rh and Ag emitters has been irradiated in several channels of the BR2 research reactor at SCK•CEN aiming at a comparison of their performance as thermal neutron flux detectors under various conditions. To allow for a correct interpretation of their signals, all detector sensitivity contributions (prompt and delayed) were calculated using a dedicated Monte Carlo model. The various contributions were also measured separately; the agreement between calculated and experimental data, including data from activation dosimetry, was excellent. Detailed neutron flux profiles were obtained from the SPND data, after correction for the finite detector lengths and for the slow response of delayed SPNDs.
Energy Technology Data Exchange (ETDEWEB)
Nakane, Yoshihiro; Sakamoto, Yukio [Japan Atomic Energy Research Inst., Tokai, Ibaraki (Japan). Tokai Research Establishment; Hayashi, Katsumi [and others
1996-09-01
Benchmark problems were prepared for evaluating the calculation codes and the nuclear data for accelerator shielding design by the Accelerator Shielding Working Group of the Research Committee on Reactor Physics in JAERI. Four benchmark problems: transmission of quasi-monoenergetic neutrons generated by 43 MeV and 68 MeV protons through iron and concrete shields at TIARA of JAERI, neutron fluxes in and around an iron beam stop irradiated by 500 MeV protons at KEK, reaction rate distributions inside a thick concrete shield irradiated by 6.2 GeV protons at LBL, and neutron and hadron fluxes inside an iron beam stop irradiated by 24 GeV protons at CERN are compiled in this document. Calculational configurations and neutron reaction cross section data up to 500 MeV are provided. (author)
Ali, Nur Syazwani Mohd; Hamzah, Khaidzir; Mohamad Idris, Faridah; Hairie Rabir, Mohamad
2018-01-01
The thermal neutron flux measurement has been conducted at the out-core location using self-powered neutron detectors (SPNDs). This work represents the first attempt to study SPNDs as neutron flux sensor for developing the fault detection system (FDS) focusing on neutron flux parameters. The study was conducted to test the reliability of the SPND’s signal by measuring the neutron flux through the interaction between neutrons and emitter materials of the SPNDs. Three SPNDs were used to measure the flux at four different radial locations which located at the fission chamber cylinder, 10cm above graphite reflector, between graphite reflector and tank liner and fuel rack. The measurements were conducted at 750 kW reactor power. The outputs from SPNDs were collected through data acquisition system and were corrected to obtain the actual neutron flux due to delayed responses from SPNDs. The measurements showed that thermal neutron flux between fission chamber location near to the tank liner and fuel rack were between 5.18 × 1011 nv to 8.45 × 109 nv. The average thermal neutron flux showed a good agreement with those from previous studies that has been made using simulation at the same core configuration at the nearest irradiation facilities with detector locations.
Finite element method for neutron diffusion problems in hexagonal geometry
International Nuclear Information System (INIS)
Wei, T.Y.C.; Hansen, K.F.
1975-06-01
The use of the finite element method for solving two-dimensional static neutron diffusion problems in hexagonal reactor configurations is considered. It is investigated as a possible alternative to the low-order finite difference method. Various piecewise polynomial spaces are examined for their use in hexagonal problems. The central questions which arise in the design of these spaces are the degree of incompleteness permissible and the advantages of using a low-order space fine-mesh approach over that of a high-order space coarse-mesh one. There is also the question of the degree of smoothness required. Two schemes for the construction of spaces are described and a number of specific spaces, constructed with the questions outlined above in mind, are presented. They range from a complete non-Lagrangian, non-Hermite quadratic space to an incomplete ninth order space. Results are presented for two-dimensional problems typical of a small high temperature gas-cooled reactor. From the results it is concluded that the space used should at least include the complete linear one. Complete spaces are to be preferred to totally incomplete ones. Once function continuity is imposed any additional degree of smoothness is of secondary importance. For flux shapes typical of the small high temperature gas-cooled reactor the linear space fine-mesh alternative is to be preferred to the perturbation quadratic space coarse-mesh one and the low-order finite difference method is to be preferred over both finite element schemes
Progress made by Framatome in neutron flux mapping technology
International Nuclear Information System (INIS)
Mourlevat, J.L.
1999-01-01
A fixed in-core system based on self-powered neutron detectors (SPNDs) could improve the power distribution measurement accuracy and hence improve core operating margin knowledge. For that purpose, Framatome has tested rhodium neutron detectors as fixed in-core sensors. Two such in-core strings were installed during tie summer of 1997 at Golfech Unit 2, for a three-year experiment. (author)
Fission ionisation chamber for the measurement of low fluxes of slow neutrons
International Nuclear Information System (INIS)
Weill, J.; Duchene, J.P.
1958-01-01
The ionisation chamber described is designed for the measurement of slow neutron fluxes of average or low intensity, in the presence, eventually, of very high gamma fluxes. The capture of a slow neutron by a fissile material, in this case 235 U, gives rise to fission fragments, high-energy particles which ionise the gas contained in the chamber. The neutrons are detected by virtue of the potential pulses, on the collecting electrode of the chamber, deriving from the collection of the ions produced by the fission fragments. The pulses are counted by means of a measuring system consisting of a preamplifier, a 2 Mc amplifier, a discriminator and an electronic scale with numerator or integrator. The general characteristics are as follows: sensitivity to neutrons: 0.07 kicks/n/cm 2 .s, sensitivity to γ rays: zero up to 3.10 4 R/H, a background noise at the normal discrimination voltage: 0.01 kicks/s, working H.T.: -500 V, capacity: 40 μμF, average height of pulse: 8 mV, limits of use: from several neutrons to 10 6 n/cm 2 .s. This chamber may be used in all cases where low fluxes of slow neutrons must be measured, especially in the presence of high gamma fluxes, for example in the checking of Pu concentrations in an extraction plant or for the starting up of reactors. (author) [fr
Validation of neutron flux redistribution factors in JSI TRIGA reactor due to control rod movements
International Nuclear Information System (INIS)
Kaiba, Tanja; Žerovnik, Gašper; Jazbec, Anže; Štancar, Žiga; Barbot, Loïc; Fourmentel, Damien; Snoj, Luka
2015-01-01
For efficient utilization of research reactors, such as TRIGA Mark II reactor in Ljubljana, it is important to know neutron flux distribution in the reactor as accurately as possible. The focus of this study is on the neutron flux redistributions due to control rod movements. For analyzing neutron flux redistributions, Monte Carlo calculations of fission rate distributions with the JSI TRIGA reactor model at different control rod configurations have been performed. Sensitivity of the detector response due to control rod movement have been studied. Optimal radial and axial positions of the detector have been determined. Measurements of the axial neutron flux distribution using the CEA manufactured fission chambers have been performed. The experiments at different control rod positions were conducted and compared with the MCNP calculations for a fixed detector axial position. In the future, simultaneous on-line measurements with multiple fission chambers will be performed inside the reactor core for a more accurate on-line power monitoring system. - Highlights: • Neutron flux redistribution due to control rod movement in JSI TRIGA has been studied. • Detector response sensitivity to the control rod position has been minimized. • Optimal radial and axial detector positions have been determined
International Nuclear Information System (INIS)
Goncalves, G.A.; Bogado Leite, S.Q.; Vilhena, M.T. de
2009-01-01
An analytical solution has been obtained for the one-speed stationary neutron transport problem, in an infinitely long cylinder with anisotropic scattering by the decomposition method. Series expansions of the angular flux distribution are proposed in terms of suitably constructed functions, recursively obtainable from the isotropic solution, to take into account anisotropy. As for the isotropic problem, an accurate closed-form solution was chosen for the problem with internal source and constant incident radiation, obtained from an integral transformation technique and the F N method
Neutron flux effect on the fracture toughness behavior of Tihange-III RPV material
International Nuclear Information System (INIS)
Gerard, R.; Chaouadi, R.; Bertolis, D.
2015-01-01
The question whether material test reactor (MTR) data can be used to supplement power reactor pressure vessel (RPV) surveillance data is still debated in the international community and its implications are particularly important in the perspective of long term operation (LTO). However, addressing the flux effect can be confusing if specific material and irradiation variables are not taken into account. This means that the answer to whether there is flux effect or not is neither 'no' nor 'yes' without specifying the application range. Indeed, neutron flux effect was recognized to occur in high Cu-containing steels in the low fluence range. But at high fluence, relevant for long term operation, it becomes difficult to clearly distinguish the differences between high flux and low flux. In this work, we irradiated the low Cu base metal and weld of the Tihange-III surveillance coupon in the BR2 reactor at high flux. The BR2 flux is about two orders of magnitude higher than the flux in the surveillance position. Tensile, Charpy impact and fracture toughness tests were performed on both the surveillance and MTR specimens and compared to assess the neutron flux effect. The results confirm that, at high fluence levels, the flux effect on mechanical properties is not significant, offering therefore the possibility of accelerated irradiation to investigate RPV embrittlement in the high fluence regime relevant for long term operation. (authors)
International Nuclear Information System (INIS)
Koskinas, M.F.
1979-01-01
Experimental and theoretical details of the foil activation method applied to neutrons flux measurements at the IEA-R1 reactor are presented. The thermal - and epithermal - neutron flux were determined form activation measurements of gold, cobalt and manganese foils; and for the fast neutron flux determination, aluminum, iron and nickel foils were used. The measurements of the activity induced in the metal foils were performed using a Ge-Li gamma spectrometry system. In each energy range of the reactor neutron spectrum, the agreement among the experimental flux values obtained using the three kind of materials, indicates the consistency of the theoretical approach and of the nuclear parameters selected. (Author) [pt
Calculation of fast neutron flux in reactor pressure tubes and experimental facilities
Energy Technology Data Exchange (ETDEWEB)
Barnett, P. C. [Canadian General Electric (Canada)
1968-07-15
The computer program EPITHET was used to calculate the fast neutron flux (>1 MeV) in several reactor pressure tubes and experimental facilities in order to compare the fast neutron flux in the different cases and to provide a self-consistent set of flux values which may be used to relate creep strain to fast neutron flux . The facilities considered are shown below together with the calculated fast neutron flux (>1 MeV). Fast flux 10{sup 13} n/cm{sup 2}s: NPD 1.14, Douglas Point 2.66, Pickering 2.89, Gentilly 2.35, SGHWR 3.65, NRU U-1 and U-2 3.25'' pressure tube - 19 element fuel 3.05, NRU U-1 and U-2 4.07'' pressure tube - 28 element fuel 3.18, NRU U-1 and U-2 4.07'' pressure tube - 18 element fuel 2.90, NRX X-5 0.88, PRTR Mk I fuel 2.81, PRTR HPD fuel 3.52, WR-1 2.73, Mk IV creep machine (NRX) 0.85, Mk VI creep machine (NRU) 2.04, Biaxial creep insert (NRU U-49) 2.61.
Radiation protection aspects of a high flux, fast neutron generator
International Nuclear Information System (INIS)
DeLuca, P.M.; Torti, R.P.; Chenevert, G.M.; Tesmer, J.R.; Kelsey, C.A.
1976-01-01
During the development and operation of a gas target, DT neutron generator for use in cancer therapy, two radiation hazards were routinely encountered - personnel exposure to neutrons and to tritium. The principal hazard was irradiation by fast neutrons. By assembling the source below ground level, adding shielding and the use of a controlled access, key identification interlock, the neutron hazard has been reduced. With the present source strength of 2 x 10 12 n/sec, an average neutron dose rate in the control room of 20 mrem/hr was measured- a level compatible with a limited run schedule. The second hazard was exposure to tritium in both gaseous and solid forms. A target inventory of 90 Ci, and overall inventory of 500 Ci, and the need to modify and repair the generator present significant potential hazard due to tritium exposure. The use of protective gloves, wipe tests, urine assays, continuous room air monitoring, and equipment decontamination minimized personnel exposure and effectively confined contamination. The dose due to tritium has been ∼ 0.5 rem/year and negligible spread of contamination has occurred
An observer-theoretic approach to estimating neutron flux distribution
International Nuclear Information System (INIS)
Park, Young Ho; Cho, Nam Zin
1989-01-01
State feedback control provides many advantages such as stabilization and improved transient response. However, when the state feedback control is considered for spatial control of a nuclear reactor, it requires complete knowledge of the distributions of the system state variables. This paper describes a method for estimating the flux spatial distribution using only limited flux measurements. It is based on the Luenberger observer in control theory, extended to the distributed parameter systems such as the space-time reactor dynamics equation. The results of the application of the method to simple reactor models showed that the flux distribution is estimated by the observer very efficiently using information from only a few sensors
Energetic ion diagnostics using neutron flux measurements during pellet injection
Energy Technology Data Exchange (ETDEWEB)
Heidbrink, W.W.
1986-01-01
Neutron measurements during injection of deuterium pellets into deuterium plasmas on the Tokamak Fusion Test Reactor (TFTR) indicate that the fractional increase in neutron emission about 0.5 msec after pellet injection is proportional to the fraction of beam-plasma reactions to total fusion reactions in the unperturbed plasma. These observations suggest three diagnostic applications of neutron measurements during pellet injection: (1) measurement of the beam-plasma reaction rate in deuterium plasmas for use in determining the fusion Q in an equivalent deuterium-tritium plasma, (2) measurement of the radial profile of energetic beam ions by varying the pellet size and velocity, and (3) measurement of the ''temperature'' of ions accelerated during wave heating. 18 refs., 3 figs.
Energetic ion diagnostics using neutron flux measurements during pellet injection
International Nuclear Information System (INIS)
Heidbrink, W.W.
1986-01-01
Neutron measurements during injection of deuterium pellets into deuterium plasmas on the Tokamak Fusion Test Reactor (TFTR) indicate that the fractional increase in neutron emission about 0.5 msec after pellet injection is proportional to the fraction of beam-plasma reactions to total fusion reactions in the unperturbed plasma. These observations suggest three diagnostic applications of neutron measurements during pellet injection: (1) measurement of the beam-plasma reaction rate in deuterium plasmas for use in determining the fusion Q in an equivalent deuterium-tritium plasma, (2) measurement of the radial profile of energetic beam ions by varying the pellet size and velocity, and (3) measurement of the ''temperature'' of ions accelerated during wave heating. 18 refs., 3 figs
On solution to the problem of reactor kinetics with delayed neutrons by Monte Carlo method
International Nuclear Information System (INIS)
Kyncl, Jan
2013-07-01
The initial value problem is addressed for the neutron transport equation and for the system of equations that describe the behaviour of emitters of delayed neutrons. Examination of the solution to this problem is based on several main assumptions concerning the behaviour of macroscopic effective cross-sections describing the reaction of the neutron with the medium, the temperature of medium and the remaining parameters of the equations. Formulation of these assumptions is adequately general and is in agreement with the properties of all known models of the physical quantities involved. Among others, the assumptions admit dependence of the macroscopic effective cross-sections and temperature on spatial coordinates and time that can be arbitrary to a great extent. The problem starts from a set of integro-differential equations. This problem is first transposed into the equivalent problem of solving a linear integral equation for neutron flux. This integral equation is solved by the method of successive iterations and its uniqueness is demonstrated. Numeric solution to the integral equation by Monte Carlo method consists in finding a functional of the exact solution. For this, a random process is set up and some random variables are proposed. Then it is demonstrated that each of these variables is an unbiased estimator of that functional. (author)
Development of network communication function for digitalized neutron flux monitoring instrument
International Nuclear Information System (INIS)
Li Kai; Zhang Liangju; Chen Xiaojun; Li Baoxiang
2002-01-01
It is essential for a digitalized Neutron Flux Monitoring Instrument to communicate with other parts of Instrumentation and Control System in a network environment, and it is fairly different from the case of traditional analogue nuclear instrumentation. How to satisfy all the requirements of different network structure and communication protocol, which might be adopted in different target nuclear power plant, is a key issue in the design and development of a digitalized neutron flux monitoring instrument. The author describes the overall communication scheme, mainly discusses the design idea and the requirements of the communication interfaces and the implementation of the RS-485 interface as an example of the digitalized neutron flux monitoring instrument, which is under development in the institute
International Nuclear Information System (INIS)
Rana, Y.S.; Mishra, Abhishek; Singh, Tej
2016-06-01
Dhruva is a 100 MW th tank type research reactor with natural metallic uranium as fuel and heavy water as coolant, moderator and reflector. The reactor is utilized for production of a large variety of radioisotopes for fulfilling growing demands of various applications in industrial, agricultural and medicinal sectors, and neutron beam research in condensed matter physics. The core consists of two on-power tray rods for radioisotope production and fifteen experimental beam holes for neutron beam research. Recently, a self-serve facility has also been commissioned in one of the through tubes in the reactor for carrying out short term irradiations. To get accurate information about neutron flux spectrum, measurements have been carried out in self-serve facility of Dhruva reactor. The present report describes measurement method, analysis technique and results. Theoretical estimations for neutron flux were also carried out and a comparison between theoretical and experimental results is made. (author)
To the problem of the coherence length of neutrons
International Nuclear Information System (INIS)
Varga, P.
1992-11-01
The challenge of the high accuracy of certain optical measurements, the long coherence length of light provokes one to search for possibilities to enlarge the neutron coherence length. A proposal is made to achieve this by using a five or a four plate Bonse-Hart interferometer. A further problem is, whether the neutron beam is composed of wave packets or of overlapping independent monochromatic waves; it is considered that the former is more likely. (author) 12 refs.; 3 figs
International Nuclear Information System (INIS)
Bosevski, T.
1971-01-01
The polynomial interpolation of neutron flux between the chosen space and energy variables enabled transformation of the integral transport equation into a system of linear equations with constant coefficients. Solutions of this system are the needed values of flux for chosen values of space and energy variables. The proposed improved method for solving the neutron transport problem including the mathematical formalism is simple and efficient since the number of needed input data is decreased both in treating the spatial and energy variables. Mathematical method based on this approach gives more stable solutions with significantly decreased probability of numerical errors. Computer code based on the proposed method was used for calculations of one heavy water and one light water reactor cell, and the results were compared to results of other very precise calculations. The proposed method was better concerning convergence rate, decreased computing time and needed computer memory. Discretization of variables enabled direct comparison of theoretical and experimental results
Development of a Neutron Flux Monitoring System for Sodium-cooled Fast Reactors
Verma, Vasudha
2017-01-01
Safety and reliability are one of the key objectives for future Generation IV nuclear energy systems. The neutron flux monitoring system forms an integral part of the safety design of a nuclear reactor and must be able to detect any irregularities during all states of reactor operation. The work in this thesis mainly concerns the detection of in-core perturbations arising from unwanted movements of control rods with in-vessel neutron detectors in a sodium-cooled fast reactor. Feasibility stud...
Effect of Different Thermal Neutron Fluxes on Blood of Male Mice
International Nuclear Information System (INIS)
Abd El-Latif, A.A.; Saeid, Kh. S.; Abd El-Latif, A.A.; Emara, N.M.; Emara, N.M.
2010-01-01
This work deals with the exposing of male mice to different fluxes of thermal neutron .Investigation has been performed by calculating of thermal neutron fluxes(0.27x10 8 N/cm 2 . 1h , 0.54x10 8 N/cm 2 . 1h, 1.08x10 8 N/cm 2 . 1h, 2.16x10 8 N/cm 2 . 3h and 4.32x10 8 N/cm 2 . 6h) which emitted from neutron irradiation cell with source Ra - Be (α,n) have activity 3 m. Ci made by leybold(55930) . The number and differential leucocytes counts types of white blood cells in million per cubic millimeter (W. B. Cs. mm -3 ) ,the number of platelets mm -3 ,the number of red blood cells in million per cubic millimeter (R. B. Cs. mm -3 ), the hemoglobin in Blood (mg/dl), the lymphocytes ,and the eosiniphil leucocytes in blood decrease with increasing thermal neutron fluxes. But neutrophile and monocytes in blood increase with increasing the thermal neutron fluxes
Feasibility of laser pumping with neutron fluxes from present-day large tokamaks
Energy Technology Data Exchange (ETDEWEB)
Jassby, D.L.
1986-08-01
The minimum fusion-neutron flux needed to observe nuclear-pumped lasing with tokamaks can be reduced substantially by optimizing neutron scattering into the laser cell, located between adjacent toroidal-field coils. The laser lines most readily pumped are probably the /sup 3/He-Ne lines at 0.633 ..mu.. and in the infrared, where the /sup 3/He-Ne gas is excited by energetic ions produced in the /sup 3/He(n,p)T reaction. These lines are expected to lase at the levels of D-T neutron flux foreseen for the TFTR in 1989 (>>10/sup 12/ n/cm/sup 2//s), while amplification should be observable at the existing levels of D-D neutron flux (greater than or equal to 5 x 10/sup 9/ n/cm/sup 2//s). Lasing on the 1.73 ..mu.. and 2.63 ..mu.. transitions of Xe may be observable at the maximum expected levels of D-T neutron flux in TFTR enhanced by scattering.
Feasibility of laser pumping with neutron fluxes from present-day large tokamaks
International Nuclear Information System (INIS)
Jassby, D.L.
1986-08-01
The minimum fusion-neutron flux needed to observe nuclear-pumped lasing with tokamaks can be reduced substantially by optimizing neutron scattering into the laser cell, located between adjacent toroidal-field coils. The laser lines most readily pumped are probably the 3 He-Ne lines at 0.633 μ and in the infrared, where the 3 He-Ne gas is excited by energetic ions produced in the 3 He(n,p)T reaction. These lines are expected to lase at the levels of D-T neutron flux foreseen for the TFTR in 1989 (>>10 12 n/cm 2 /s), while amplification should be observable at the existing levels of D-D neutron flux (≥ 5 x 10 9 n/cm 2 /s). Lasing on the 1.73 μ and 2.63 μ transitions of Xe may be observable at the maximum expected levels of D-T neutron flux in TFTR enhanced by scattering
Measurements of neutron flux distributions in the core of the Ljubljana TRIGA Mark II Reactor
International Nuclear Information System (INIS)
Rant, J.; Ravnik, M.; Mele, I.; Dimic, V.
2008-01-01
Recently the Ljubljana TRIGA Mark II Reactor has been refurbished and upgraded to pulsed operation. To verify the core design calculations using TRIGAP and PULSTR1 codes and to obtain necessary data for future irradiation and neutron beam experiments, an extensive experimental program of neutron flux mapping and neutron field characterization was carried out. Using the existing neutron measuring thimbles complete axial and radial distributions in two radial directions were determined for two different core configurations. For one core configuration the measurements were also carried out in the pulsed mode. For flux distributions thin Cu (relative measurements) and diluted Au wires (absolute values) were used. For each radial position the cadmium ratio was determined in two axial levels. The core configuration was rather uniform, well defined (fresh fuel of a single type, including fuelled followers) and compact (no irradiation channels or gaps), offering unique opportunity to test the computer codes for TRIGA reactor calculations. The neutron flux measuring procedures and techniques are described and the experimental results are presented. The agreement between the predicted and measured power peaking factors are within the error limits of the measurements (<±5%) and calculations (±10%). Power peaking occurs in the B ring, and in the A ring (centre) there is a significant flux depression. (authors)
Flux-Vortex Pinning and Neutron Star Evolution
Indian Academy of Sciences (India)
M. Ali Alpar
2017-09-12
Sep 12, 2017 ... M. ALI ALPAR. Faculty of Engineering and Natural Sciences, Sabancı University, 34956, Istanbul, Turkey. E-mail: ... netic field of the neutron star were B ∼ 109 G. At the ..... across pinning energy barriers by thermal activation.
The measurements of thermal neutron flux distribution in a paraffin ...
Indian Academy of Sciences (India)
outputs and the best agreement was generated using NaI(Tl) by a minimum discrepancy of .... the four possible reactions with neutron absorption, our desire is to form 116m1In, which .... According to Geiger user's manual, its efficiency for.
Absolute measurements of the fast neutron flux in the reactor RA
Energy Technology Data Exchange (ETDEWEB)
Berovic, N; Boreli, F; Dragin, R [Institute of Nuclear Sciences Boris Kidric, Department of physics, Vinca, Beograd (Serbia and Montenegro)
1961-10-15
The absolute neutron flux in the vertical VK-5 hole of the reactor RA was determined by using the {sup 27}Al (n, alpha) {sup 24}Na reaction, and by counting the {sup 24}Na - 2.5 MeV gamma line photopeak activity. A method for the determination of {sigma}{sub eff} as a mean value between the two large limiting cases of neutron spectra is used. The flux at the power level of 5 MW was found to be (2.5{+-}0.9){center_dot}10{sup 12}n/cm{sup 2}sec (author)
Neutron flux density data acquisition system based on LabVIEW
International Nuclear Information System (INIS)
Zhao Yanhui; Zhao Xiuliang; Li Zonglun; Liang Fengyan; Liu Liyan
2011-01-01
In the LabVIEW software, combined with PCI-6251 data acquisition card, VI of neutron flux density data acquisition is realized by DAQmx data acquisition functions. VI is composed of front panel and block diagram. The data collected can be displayed in the forms of the data curve and the data control, and saved in the form of files. Test results show that the frequency of output signal in NI ELVIS can be accurately measured by the system, realizing neutron flux density data acquisition based on LabVIEW. (authors)
Development of a position-sensitive fission counter and measurement of neutron flux distributions
International Nuclear Information System (INIS)
Yamagishi, Hideshi; Soyama, Kazuhiko; Kakuta, Tsunemi
2001-08-01
A position-sensitive fission counter (PSFC) that operates in high neutron flux and high gamma-ray background such as at the side of a power reactor vessel has been developed. Neutron detection using the PSFC with a solenoid electrode is based on a delay-line method. The PSFC that has the outer diameter of 25 mm and the sensitive length of 1000 mm was manufactured for investigation of the performances. The PSFC provided output current pulses that were sufficiently higher than the alpha noise, though the PSFC has a solenoid electrode and large electrode-capacitance. The S/N ratio of PSFC outputs proved to be higher than that of ordinary fission counters with 200 mm sensitive length. A performance test to measure neutron flux distributions by a neutron measuring system with the PSFC was carried out by the side of a graphite pile, W2.4 x H1.4 x L1.2 m, with neutron sources, Am-Be 370 GBq x 2. It was confirmed that the neutron flux distribution was well measured with the system. (author)
A neutronic feasibility study for LEU conversion of the High Flux Beam Reactor (HFBR)
International Nuclear Information System (INIS)
Pond, R.B.; Hanan, N.A.; Matos, J.E.
1997-01-01
A neutronic feasibility study for converting the High Flux Beam Reactor at Brookhaven National Laboratory from HEU to LEU fuel was performed at Argonne National Laboratory. The purpose of this study is to determine what LEU fuel density would be needed to provide fuel lifetime and neutron flux performance similar to the current HEU fuel. The results indicate that it is not possible to convert the HFBR to LEU fuel with the current reactor core configuration. To use LEU fuel, either the core needs to be reconfigured to increase the neutron thermalization or a new LEU reactor design needs to be considered. This paper presents results of reactor calculations for a reference 28-assembly HEU-fuel core configuration and for an alternative 18-assembly LEU-fuel core configuration with increased neutron thermalization. Neutronic studies show that similar in-core and ex-core neutron fluxes, and fuel cycle length can be achieved using high-density LEU fuel with about 6.1 gU/cm 3 in an altered reactor core configuration. However, hydraulic and safety analyses of the altered HFBR core configuration needs to be performed in order to establish the feasibility of this concept. (author)
International Nuclear Information System (INIS)
Ramirez G, T.
1976-01-01
This is a study of the technique for the determination of nitrogen and other elements in wheat flour through activation analysis with fast neutrons from a thermal nuclear reactor. The study begins with an introduction about the basis of the analytical methods, the equipment used in activation analysis and a brief description of the neutrons source. In the study are included the experiments carried out in order to determine the flux form in the site of irradiation, the N-13 half life and the interference due to the sample composition. (author)
TRANSHEX, 2-D Thermal Neutron Flux Distribution from Epithermal Flux in Hexagonal Geometry
International Nuclear Information System (INIS)
Patrakka, E.
1994-01-01
1 - Description of program or function: TRANSHEX is a multigroup integral transport program that determines the thermal scalar flux distribution arising from a known epithermal flux in two- dimensional hexagonal geometry. 2 - Method of solution: The program solves the isotropic collision probability equations for a region-averaged scalar flux by an iterative method. Either a successive over-relaxation or an inner-outer iteration technique is applied. Flat flux collision probabilities between trigonal space regions with white boundary condition are utilized. The effect of epithermal flux is taken into consideration as a slowing-down source that is calculated for a given spatial distribution and 1/E energy dependence of the epithermal flux
Directory of Open Access Journals (Sweden)
Alekseenko Victor
2017-01-01
Full Text Available Some exotic geophysical events are observed by a global net of electron-neutron detectors (en-detectors developed in the framework of the PRISMA EAS project. Our en-detectors running both on the Earth's surface and underground are continuously measuring the environmental thermal neutron flux. Thermal neutrons are in equilibrium with media and are therefore sensitive to many geophysical phenomena, which are exotic for people studying ultra high-energy cosmic rays or carrying out low background experiments deep underground.
Higher order Cambell techniques for neutron flux measurement. Pt. 1
International Nuclear Information System (INIS)
Lux, I.; Baranyai, A.
1982-01-01
An exact mathematical description of arbitrary high order Campbell techniques for measuring particle fluxes is given. The nth order Campbell technique assumes the measurement of the moments of the outcoming voltage up to the nth one. A simple relation is derived among the various moments of the total measured voltage and of the detector signal caused by one incident particle. It is proven that in the monoparticle case combination of the measured moments up to the order n provides an expression proportional to the particle flux and to the nth moment of the detector signal. Generalization to several different particles is given and it is shown that if the flux of the particle causing the largest detector signal is measured with a relative error epsilon in the dc method and the error is due to the signals of other particles, then in the nth order campbelling the error will be of order epsilonsup(n). The effect of a random background on the measured voltage is also investigated and it is established that the nth order campbelling supresses the noise according to the nth power of the relative amplitude of the noise to the signal. The results concerning constant fluxes are generalized to time dependent particle fluxes and a method assuming a Fourier transform of the measured quantities is proposed for their determination. (orig.)
Tritium solid targets for intense D-T neutron production and its related problems
International Nuclear Information System (INIS)
Sumita, Kenji
1988-01-01
This review paper is divided into three parts. Firstly, to attain an intense neutron production rate, the construction of a design with a higher tritium-containing surface and an effective cooling system like a rotating target device are discussed. The maximum attainable intensity based on tritium solid targets shall be estimated regarding planning for future D-T sources. Secondly, on the way to carry out some experiments, an absolute intensity calibration and an angular dependent neutron energy spectrum of the neutron source are essential parameters to analyse the results of the experiments. Sometimes the space dependent neutron spectrum is required as well as the space dependent neutron flux near the targets and irradiation samples. The measurement methods and their examples are reviewed for tritium solid targets. The third part is devoted to discuss the protection to tritium contamination problems due to unavoidable release of tritium gas from targets. Performance and effectiveness of tritium collection systems for intense D-T neutron sources shall be discussed in some examples. Tritium contamination incidents due to the faulted film powder of target surface are also reported in some real incident cases. (author). Abstract only
International Nuclear Information System (INIS)
Akki, T.S.; Benayad, S.A.; Megahid, R.M.
1992-01-01
Measurements have been carried out to study the spatial fluxes and energy distributions of reactor fast neutrons transmitted through two types of heat resistant concretes, serpentine concrete and magnetic lemonite concrete. The physical, chemical and mechanical properties of these concretes were checked by well known techniques. In addition, the effect of heating at temperatures up to 500deg C on the crystaline water content was checked by the method of differential thermal analysis. Measurements were performed using a collimated beam of reactor neutrons emitted from a 10 MW research reactor. The neutron spectra transmitted through concrete barriers of different thickness were measured by a scintillation spectrometer with NE-213 liquid organic scintillator. Discrimination against undesired pulses due to gamma-rays was achieved by a method based on pulse shape discrimination technique. The operating principle of this technique is based on the comparison of two weighted time integrals of the detector signal. The measured pulse amplitude distribution was converted to neutron energy distribution by a computational code based on double differentiation technique. The spectrometer workability and the accuracy of the unfolding technique were checked by measuring the neutron spectra of neutrons from Pu-α-Be and 252 Cf neutron sources. The obtained neutron spectra for the two concretes were used to derive the total cross sections for neutrons of different energies. (orig.)
Experience with neutron flux monitoring systems qualified for post-accident monitoring
International Nuclear Information System (INIS)
Shugars, H.G.; Miller, J.F.
1995-01-01
In this paper we discuss the environmental requirements for excore neutron flux monitors that are qualified for use during and after postulated accidents in Pressurized Water Reactors (PWRs). We emphasize PWRs designed in the United States, which are similar to those used also in parts of Western Europe and Eastern Asia. We then discuss design features of the flux monitoring systems necessary to address the environmental, functional, and regulatory requirements, and the experience with these systems. (author). 9 refs, 2 figs
Calculation of neutron flux and reactivity by perturbation theory at high order
International Nuclear Information System (INIS)
Silva, W.L.P. da; Silva, F.C. da; Thome Filho, Z.D.
1982-01-01
A high order pertubation theory is studied, independent of time, applied to integral parameter calculation of a nuclear reactor. A pertubative formulation, based on flux difference technique, which gives directy the reactivity and neutron flux up to the aproximation order required, is presented. As an application of the method, global pertubations represented by fuel temperature variations, are used. Tests were done aiming to verify the relevancy of the approximation order for several intensities of the pertubations considered. (E.G.) [pt
International Nuclear Information System (INIS)
Abdel Munem, E.; Shukri, A.; Tajuddin, A.A.
2006-01-01
A study of the thermal column of the Malaysian TRIGA Mark II reactor, forming part of a feasibility study for BNCT was proposed in 2001. In the current study, pure metals were used to measure the neutron flux at selected points in the thermal column and the neutron flux determined using SAND-II. Monte Carlo simulation of the thermal column was also carried out. The reactor core was homogenized and calculations of the neutron flux through the graphite stringers performed using MCNP5. The results show good agreement between the measured flux and the MCNP calculated flux. An obvious extension from this is that the MCNP neutron flux output can be utilized as an input spectrum for SAND-II for the flux iteration. (author)
A new detector for the measurement of neutron flux in nuclear reactors
International Nuclear Information System (INIS)
Koch, L.; Labeyrie, J.; Tarassenko, S.
1958-01-01
The detector described is designed for the instantaneous measurement of thermal neutron fluxes, in the presence of high γ ray activity; this detector can withstand temperatures as high as 500 deg. C. It is based on the following principle: radioactive atoms resulting from heavy-nucleus fission are carried by a gas flow to a detector recording their β and γ disintegration. Thermal neutron fluxes as low as few neutrons per cm 2 per second can be measured. This detector may be used to control a nuclear reactor, to plot the thermal flux distribution with an excellent definition (1 mm 2 ) for fluxes higher than 10 8 n/cm 2 /s. The time response of the system to a sharp variation of flux is limited, in case of large fluxes, to the transit time of the gas flow between the fission product emitter and the detector; of the order of one tenth of a sec per meter of piping. The detector may also be applied for spectroscopy of fission products eider than 0,1 s. (author) [fr
Violet and visual flux problems in red giant stars
International Nuclear Information System (INIS)
Faulkner, D.R.
1989-01-01
Red giant stars are sites of many astrophysically interesting processes and are important links to late stages of stellar evolution and the chemical history of the galaxy. Much of what is known about stars comes from their spectra, which are formed in the outer layers (atmospheres). Unfortunately the low temperatures in red giant atmospheres promote the formation of many molecules, and the resultant complexity of the spectra has slowed progress in obtaining good models of these objects and leaves many unanswered questions. Several of these problems are investigated. Spectra of red giants provide a natural classification according to composition: M stars are oxygen rich, C stars are carbon rich, while S stars are intermediate. One long standing problem with C stars has been the explanation of the severe lack of energy flux in the violet and near ultraviolet part of their spectrum, generally attributed to an unusual opacity. Results show that one source, SiC, is untenable, while the case for the other, C3, is severely weakened. Synthetic spectra from atmospheric models are compared to spectra of TX Psc, a C star, to show that the contribution of thousands of atomic lines are probably responsible for the violet and ultraviolet flux deficiency. The agreement between the synthetic spectra and observations is very good. K and M type stars also have a violet flux deficiency, though it is less severe than with carbon stars
Measurement of the North-South asymmetry in the solar proton albedo neutron flux
International Nuclear Information System (INIS)
Ifedili, S.O.
1979-01-01
The solar proton albedo neutron flux in the range 10 -2 --10 7 eV measured by a neutron detector on board the Ogo 6 satellite was examined for north-south asymmetry. For the solar proton event of December 19, 1969, the S/N ratio of the solar proton albedo neutron rate at geomagnetic latitude lambda>70 0 was 1.61 +- 0.27 during the event, while for the November 2, 1969, event at 40 0 0 and altitudes ranging from 700 km to 800 km the solar proton albedo neutron rate was 0.40 +- 0.10 count/s in the north and 0.00 +- 0.10 count/s in the south. During the solar proton event of December 18, 1969, the N/S ratio of the solar proton albedo neutron rate at lambda>70 0 was 1.00 +- 0.26. The results are consistent with the expected N-S asymmetry in the solar proton flux. An interplanetary proton anisotropy with the interplanetary magnetic field polarity away from the sun corresponded to larger fluxes of solar proton albedo neutrons at the north polar cap than at the south, while an interplanetary proton anisotropy with the interplanetary magnetic field polarity toward the sun corresponded to larger fluxes of solar proton albedo neutrons at the south polar cap than at the north. This evidence favors the direct access of solar protons to the earth's polar caps via the merged interplanetary and geomagnetic field lines
Fundamental Problems of Neutron Physics at the Spallation Neutron Source at the ORNL
International Nuclear Information System (INIS)
Gudkov, Vladimir
2008-01-01
We propose to provide theoretical support for the experimental program in fundamental neutron physics at the SNS. This includes the study of neutron properties, neutron beta-decay, parity violation effects and time reversal violation effects. The main purpose of the proposed research is to work on theoretical problems related to experiments which have a high priority at the SNS. Therefore, we will make a complete analysis of beta-decay process including calculations of radiative corrections and recoil corrections for angular correlations for polarized neutron decay, with an accuracy better that is supposed to be achieved in the planning experiments. Based on the results of the calculations, we will provide analysis of sensitivity of angular correlations to be able to search for the possible extensions of the Standard model. Also we will help to plan other experiments to address significant problems of modern physics and will work on their theoretical support.
Noise and its application to neutron flux measurements
International Nuclear Information System (INIS)
Sabate Puigmal, Pedro.
1984-08-01
Fission Counter's (FC) fundamental principles were studied, operating this neutron detector as pulses generator (AC modes) and fluctuant current (DC modes). Power spectral series were obtained in DC modes, corresponding to: alpha activity of the FC neutron converter, gamma exposition in Co 60 radiation field, and only neutronic field. These noise spectra were correlated with those obtained from the FC in RA-3 critical facility, at different reactor power levels. These experiments allow to verify that, in DC mode, the power noise is very weakly dependent of the reactor gamma field, over a wide range of reactor working power, and that this range is strongly dependent of the detector's position with respect to the core's position. The frequency band of measurement is not critical. The results suggest that it is possible to develop a simple and compact measurement chain for nuclear reactors control. This would be obtained with an adequate combination of the FC operation ranges in AC and DC modes. Approximately ten decades in working power would be thus controlled with this unique type of detector (Campbellian method). A locally devised commercial detector (CFPT9) was used in these tests, and several of the most useful positions of the FC were determined. Frequency band from 150 Hz to 150 KHz was investigated. (M.E.L.) [es
Energy Technology Data Exchange (ETDEWEB)
Bitelli, Ulysses d' Utra; Mura, Luiz Ernesto Credidio; Santos, Diogo Feliciano dos; Jerez, Rogerio; Mura, Luis Felipe Liamos, E-mail: ubitelli@ipen.br, E-mail: credidiomura@gmail.com [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2013-07-01
This paper presents the neutron energy spectrum in the central position of a neutron flux trap assembled in the core center of the research nuclear reactor IPEN/MB-01 obtained by an unfolding method. To this end, have been used several different types of activation foils (Au, Sc, Ti, Ni, and plates) which have been irradiated in the central position of the reactor core (setting number 203) at a reactor power level of 64.57 ±2.91 watts . The activation foils were counted by solid-state detector HPGe (gamma spectrometry). The experimental data of nuclear reaction rates (saturated activity per target nucleus) and a neutron spectrum estimated by a reactor physics computer code are the main input data to get the most suitable neutron spectrum in the irradiation position obtained through SANDBP code: a neutron spectra unfolding code that use an iterative adjustment method. The adjustment resulted in 3.85 ± 0.14 10{sup 9} n cm{sup -2} s{sup -1} for the integral neutron flux, 2.41 ± 0.01 10{sup 9} n cm{sup -2} s{sup -1} for the thermal neutron flux, 1.09 ± 0.02 10{sup 9} n cm{sup -2} s{sup -1} for intermediate neutron flux and 3.41± 0.02 10{sup 8} n cm{sup -2} s{sup -1} for the fast neutrons flux. These results can be used to verify and validate the nuclear reactor codes and its associated nuclear data libraries, besides show how much is effective the use of a neutron flux trap in the nuclear reactor core to increase the thermal neutron flux without increase the operation reactor power level. The thermal neutral flux increased 4.04 ± 0.21 times compared with the standard configuration of the reactor core. (author)
Energy Technology Data Exchange (ETDEWEB)
Mas, P; Sciers, P [Commissariat a l' Energie Atomique, Grenoble (France). Centre d' Etudes Nucleaires
1964-07-01
The device described in this report is working from this following principle: an element, the thermal neutron activate of which has a short period and is emitter of a {beta} particle, is surrounded with a dielectric which is itself surrounded by a conducting metal. {beta} particles are stopped in the conducting metal. Thus an electric current rises between the emitter element and the collector metal. The experimental study has been-done with rhodium as emitter. The linearity as function of thermal flux of the power of the pile, the accuracy, the efficiency of the device have been measured and are satisfactory. Just the stability of the measure equipment is not very good. This method gives, us at moment the instantaneous flux and the integrated flux, is well adapted for the measurements in pile to execute on loops. It is possible to continue to perfect the process. (authors) [French] Le dispositif decrit dans ce rapport fonctionne d'apres le principe suivant: un corps, dont le produit d'activation en neutrons thermiques a une courte periode et est emetteur d'une particule {beta}, est entoure d'un dielectrique lui-meme entoure d'un metal conducteur. Les particules {beta} sont arretees dans le metal conducteur. Il y a ainsi naissance d'un courant electrique entre le corps emetteur et le metal collecteur. L'etude experimentale a ete menee avec le rhodium comme emetteur. La linearite en fonction du flux thermique, de la puissance de la pile, la precision, le rendement du dispositif ont ete mesures et sont satisfaisants. Seule la stabilite de l'appareillage de mesure n'est pas tres bonne. Cette methode, qui peut nous donner a tout instant le flux instantane et le flux integre, est tout a fait adaptee aux mesures en pile a effectuer. sur des boucles. On peut poursuivre la mise au point du procede. (auteurs)
International Nuclear Information System (INIS)
Parsons, R.J.
1983-01-01
By using a combination of Monte-Carlo and diffusion theory techniques, the behaviour of the thermal neutron flux during fluid injection is studied. It is shown that the change in neutron flux induced by the fluid injection, is equal to the neutron flux due to a certain thermal neutron source distribution. Using this result, a method of estimating the size of an elliptical invaded region is given. This choice of region shape is not a necessity but a convenience and it is possible that the method may be generalised to include higher order shapes. (author)
Characteristics of Fabricated SiC Neutron Detectors for Neutron Flux Monitoring
Energy Technology Data Exchange (ETDEWEB)
Kim, Han Soo; Ha, Jang Ho; Park, Se Hwan; Lee, Kyu Hong [Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of); Lee, Cheol Ho [Hanyang University, Seoul (Korea, Republic of)
2011-05-15
An SPND (Self-powered Neutron Detector) is commonly used for neutron detection in NPP (Nuclear Power Plant) by virtue of un-reactivity for gamma-rays. But it has a drawback, which is that it cannot detect neutrons in real time due to beta emissions (about > 48 s) after reactions between neutrons and {sup 103}Rh in an SPND. And Generation IV reactors such as MSR (Molten-salt reactor), SFR (Sodium-cooled fast reactor), and GFR (Gas-cooled fast reactor) are designed to compact size and integration type. For GEN IV reactor, neutron monitor also must be compact-sized to apply such reactor easily and much more reliable. The wide band-gap semiconductors such as SiC, AlN, and diamond make them an attractive alternative in applications in harsh environments by virtue of the lower operating voltage, faster charge-collection times compared with gas-filled detectors, and compact size.1) In this study, two PIN-type SiC semiconductor neutron detectors, which are for fast neutron detection by elastic and inelastic scattering SiC atoms and for thermal neutron detection by charged particle emissions of 6LiF reaction, were designed and fabricated for NPP-related applications. Preliminary tests such as I-V and alpha response were performed and neutron responses at ENF in HANARO research reactor were also addressed. The application feasibility of the fabricated SiC neutron detector as an in-core neutron monitor was discussed
Thermal–hydraulic analysis of a candidate design for ITER divertor neutron flux monitor (DNFM)
Energy Technology Data Exchange (ETDEWEB)
Tanchuk, Victor, E-mail: Victor.Tanchuk@sintez.niiefa.spb.su [Scientific Technical Center SINTEZ, D.V. Efremov Institute, 196641 St. Petersburg (Russian Federation); Alexandrov, Evgeny [Institution “Project Center ITER”, 1, Akademika Kurchatova sq., 123182 Moscow (Russian Federation); Batyunin, Alexander; Kashchuk, Yuri [State Research Center of Russian Federation Troitsk Institute for Innovation and Fusion Research, ul. Pushkovykh, vladenie 12, 142190 Troitsk, Moscow Region (Russian Federation); Korban, Svetlana; Lyublin, Boris [Scientific Technical Center SINTEZ, D.V. Efremov Institute, 196641 St. Petersburg (Russian Federation); Obudovsky, Sergey [State Research Center of Russian Federation Troitsk Institute for Innovation and Fusion Research, ul. Pushkovykh, vladenie 12, 142190 Troitsk, Moscow Region (Russian Federation); Senik, Konstantin [Scientific Technical Center SINTEZ, D.V. Efremov Institute, 196641 St. Petersburg (Russian Federation)
2013-10-15
The key role in direct measurement of the ITER fusion power is assigned to the neutron diagnostic system for measurement of total neutron flux of the D–D and D–T fusion reaction with the help of a neutron flux monitor located under the divertor dome. High plasma heat loads in this position implies stringent requirements for the detector design and its cooling system to ensure the required temperature operation regime of the neutron detector. The paper describes the neutron flux monitor design developed in close collaboration with IO ITER diagnostic division. Two numerical models (hydraulic and thermal) built up to simulate the water flow in the cooling system and the temperature state of detector components are also presented and discussed. The numerical investigations carried out on the developed models have shown that only good thermal contact between the shell of the detector blocks and water-cooled casing of the monitor (fit, brazing) will provide the required temperature operation regimes of the most temperature-sensitive IFC electrodes. The obtained high temperature of the detector supports makes necessary an auxiliary direct cooling of the supports or their redesign so as to provide their higher thermal conductivity.
Thermal–hydraulic analysis of a candidate design for ITER divertor neutron flux monitor (DNFM)
International Nuclear Information System (INIS)
Tanchuk, Victor; Alexandrov, Evgeny; Batyunin, Alexander; Kashchuk, Yuri; Korban, Svetlana; Lyublin, Boris; Obudovsky, Sergey; Senik, Konstantin
2013-01-01
The key role in direct measurement of the ITER fusion power is assigned to the neutron diagnostic system for measurement of total neutron flux of the D–D and D–T fusion reaction with the help of a neutron flux monitor located under the divertor dome. High plasma heat loads in this position implies stringent requirements for the detector design and its cooling system to ensure the required temperature operation regime of the neutron detector. The paper describes the neutron flux monitor design developed in close collaboration with IO ITER diagnostic division. Two numerical models (hydraulic and thermal) built up to simulate the water flow in the cooling system and the temperature state of detector components are also presented and discussed. The numerical investigations carried out on the developed models have shown that only good thermal contact between the shell of the detector blocks and water-cooled casing of the monitor (fit, brazing) will provide the required temperature operation regimes of the most temperature-sensitive IFC electrodes. The obtained high temperature of the detector supports makes necessary an auxiliary direct cooling of the supports or their redesign so as to provide their higher thermal conductivity
Temporal variation of the neutron flux in the carousel facility of a TRIGA reactor
International Nuclear Information System (INIS)
Jacimovic, R.; Stegnar, P.; Trkov, A.
2003-01-01
In this work we focused on identifying quantitatively the temporal (time-dependent) variation of neutron flux in the carousel facility (CF) of TRIGA reactor at the 'Jozef Stefan' Institute (IJS) for core No. 176, set up in April 2002. The measurements are based on neutron detectors (ionisation chambers), which surround the graphite reflector of the reactor core. In principle, the variations of the neutron flux produce a systematic error in the results obtained by absolute or 'quasi' absolute measuring techniques (such as neutron activation analysis (NAA) by the k 0 -standardization method), which assume constant conditions during irradiation. The results of our study show that for typical irradiation of 20 hours in channels of the CF aligned in the direction of the ionisation chamber (safety channel) the time-dependent variation of the neutron flux is about 6-8%. In the k 0 method, which we are using for routine work at the IJS, this variation introduced a systematic error in the results up to 4.6%, depending on the half-life of investigated radionuclide. (author)
Characterization of magnetic degradation mechanism in a high-neutron-flux environment
Energy Technology Data Exchange (ETDEWEB)
Samin, Adib; Qiu, Jie [Nuclear Engineering Program, Department of Mechanical and Aerospace, The Ohio State University, Columbus, OH 43210 (United States); Hattrick-Simpers, Jason; Dai-Hattrick, Liyang [Department of Chemical Engineering, University of South Carolina, Columbia, SC 29208 (United States); Zheng, Yuan F. [Department of Electrical and Computer Engineering, The Ohio State University, Columbus, OH 43210 (United States); Cao, Lei, E-mail: Cao.152@osu.edu [Nuclear Engineering Program, Department of Mechanical and Aerospace, The Ohio State University, Columbus, OH 43210 (United States)
2014-09-01
Radiation-induced demagnetization of permanent magnets can result in the failure of magnet-based devices operating in high-radiation environments. To understand the mechanism underlying demagnetization, Nd-Fe-B magnets were irradiated with fast and fast plus thermal neutrons at fluences of 10{sup 12}, 10{sup 13}, 10{sup 14}, and 10{sup 15} n/cm{sup 2}, respectively. After irradiation, magnetic flux losses were shown to increase with the fluence. Compared with samples irradiated only with fast neutrons, the samples exposed to the fast plus thermal neutrons have higher magnetic flux losses, which is attributed to the thermal neutron capture reaction of boron. Hysteresis loops of the Nd-Fe-B magnets reveal a slightly increase in the coercivity after irradiation. Full remagnetization of the samples after irradiation was possible, which indicates that structural damage is unlikely an important factor in the demagnetization process at these levels of neutron flux and fluence. Finally, we performed a preliminary Molecular Dynamic (MD) simulation on a cube of ions to obtain a better understanding of the thermal spike mechanism.
“Influence Method” applied to measure a moderated neutron flux
International Nuclear Information System (INIS)
Rios, I.J.; Mayer, R.E.
2016-01-01
The “Influence Method” is conceived for the absolute determination of a nuclear particle flux in the absence of known detector efficiency. This method exploits the influence of the presence of one detector, in the count rate of another detector when they are placed one behind the other and define statistical estimators for the absolute number of incident particles and for the efficiency. The method and its detailed mathematical description were recently published (Rios and Mayer, 2015 [1]). In this article we apply it to the measurement of the moderated neutron flux produced by an "2"4"1AmBe neutron source surrounded by a light water sphere, employing a pair of "3He detectors. For this purpose, the method is extended for its application where particles arriving at the detector obey a Poisson distribution and also, for the case when efficiency is not constant over the energy spectrum of interest. Experimental distributions and derived parameters are compared with theoretical predictions of the method and implications concerning the potential application to the absolute calibration of neutron sources are considered. - Highlights: • “Influence Method” applied to measure a moderated neutron flux. • Effective efficiency defined independently of calibration sources. • Neutron sources calibration discussion.
One procedure for determination of the neutron flux in the nuclear reactor fuel
International Nuclear Information System (INIS)
Bulovic, V.; Krtil, J.; Maksimovic, Z.; Martinc, R.
1979-09-01
Possibility of determination of the neutron flux in the fuel of a heavy water reactor has been examined. In determination of the flux an iterative procedure was used to compare calculated and measured contents of several fission products. The former contents were determined by calculation of the burning process balance and the latter by non-destructive gamma-spectrometric analysis of fuel. The obtained results prove the possibility of such determination of not only the average value of the flux but also of the change of its intensity during utilization of fuel (author) [sr
Supplementary neutron flux calculations for the ORNL pool critical assembly pressure vessel facility
Energy Technology Data Exchange (ETDEWEB)
Maerker, R.E.; Maudlin, P.J.
1981-02-01
A three-dimensional Monte Carlo calculation was performed to estimate the neutron flux in the 8/7 configuration of the ORNL Pool Critical Assembly Pressure Vessel Facility. The calculational tool was the multigroup transport code MORSE operated in the adjoint mode. The MORSE flux results compared well with those using a previously adopted procedure for constructing a three-dimensional flux from one- and two-dimensional discrete ordinates calculations using the DOT-IV code. This study concluded that use of these discrete ordinates constructions in previous calculations is sufficiently accurate and does not account for the existing discrepancies between calculation and experiment.
Supplementary neutron flux calculations for the ORNL pool critical assembly pressure vessel facility
International Nuclear Information System (INIS)
Maerker, R.E.; Maudlin, P.J.
1981-02-01
A three-dimensional Monte Carlo calculation was performed to estimate the neutron flux in the 8/7 configuration of the ORNL Pool Critical Assembly Pressure Vessel Facility. The calculational tool was the multigroup transport code MORSE operated in the adjoint mode. The MORSE flux results compared well with those using a previously adopted procedure for constructing a three-dimensional flux from one- and two-dimensional discrete ordinates calculations using the DOT-IV code. This study concluded that use of these discrete ordinates constructions in previous calculations is sufficiently accurate and does not account for the existing discrepancies between calculation and experiment
International Nuclear Information System (INIS)
Hoogenboom, J.E.
1980-01-01
1 - Description of problem or function: FOCUS enables the calculation of any quantity related to neutron transport in reactor or shielding problems, but was especially designed to calculate differential quantities, such as point values at one or more of the space, energy, direction and time variables of quantities like neutron flux, detector response, reaction rate, etc. or averages of such quantities over a small volume of the phase space. Different types of problems can be treated: systems with a fixed neutron source which may be a mono-directional source located out- side the system, and Eigen function problems in which the neutron source distribution is given by the (unknown) fundamental mode Eigen function distribution. Using Monte Carlo methods complex 3- dimensional geometries and detailed cross section information can be treated. Cross section data are derived from ENDF/B, with anisotropic scattering and discrete or continuous inelastic scattering taken into account. Energy is treated as a continuous variable and time dependence may also be included. 2 - Method of solution: A transformed form of the adjoint Boltzmann equation in integral representation is solved for the space, energy, direction and time variables by Monte Carlo methods. Adjoint particles are defined with properties in some respects contrary to those of neutrons. Adjoint particle histories are constructed from which estimates are obtained of the desired quantity. Adjoint cross sections are defined with which the nuclide and reaction type are selected in a collision. The energy after a collision is selected from adjoint energy distributions calculated together with the adjoint cross sections in advance of the actual Monte Carlo calculation. For multiplying systems successive generations of adjoint particles are obtained which will die out for subcritical systems with a fixed neutron source and will be kept approximately stationary for Eigen function problems. Completely arbitrary problems can
Measurement of thermal neutron distribution of flux in the fuel element cluster - Progress report
International Nuclear Information System (INIS)
Krcevinac, S.B.; Takac, S.M.
1966-12-01
Relative distribution of thermal neutron flux in the fuel element cluster (19 UO 2 rods with Zr-II cladding, D 2 O moderator and coolant) was measured by newly developed cell perturbation method. The obtained values of mean density ratios are compared to the results obtained by TER-I code using Amouyal - Benoist model [sr
The neutrons flux density calculations by Monte Carlo code for the double heterogeneity fuel
International Nuclear Information System (INIS)
Gurevich, M.I.; Brizgalov, V.I.
1994-01-01
This document provides the calculation technique for the fuel elements which consists of the one substance as a matrix and the other substance as the corn embedded in it. This technique can be used in the neutron flux density calculation by the universal Monte Carlo code. The estimation of accuracy is presented too. (authors). 6 refs., 1 fig
International Nuclear Information System (INIS)
Wasastjerna, F.; Lux, I.
1980-03-01
A transmission probability method implemented in the program TPHEX is described. This program was developed for the calculation of neutron flux distributions in hexagonal light water reactor fuel assemblies. The accuracy appears to be superior to diffusion theory, and the computation time is shorter than that of the collision probability method. (author)
Measurement of thermal neutron flux spatial distribution in the IEA-R1 reactor core
International Nuclear Information System (INIS)
D'Utra Bitelli, U.
1993-01-01
This work presents the spatial thermal neutron flux in IEA-R1 reactor obtained by activation foils methods. These measurements were made in 27 fuel elements of the reactor core (165 B configuration). The results are important to compare with theoretical values, power calibration and safety analysis. (author)
Discussion about modeling the effects of neutron flux exposure for nuclear reactor core analysis
International Nuclear Information System (INIS)
Vondy, D.R.
1986-04-01
Methods used to calculate the effects of exposure to a neutron flux are described. The modeling of the nuclear-reactor core history presents an analysis challenge. The nuclide chain equations must be solved, and some of the methods in use for this are described. Techniques for treating reactor-core histories are discussed and evaluated
Neutron spectra measurements and neutron flux monitoring for radiation damage purposes
International Nuclear Information System (INIS)
Osmera, B.; Petr, J.; Racek, J.; Rumler, C.; Turzik, Z.; Franc, L.; Holman, M.; Hogel, J.; Kovarik, K.; Marik, P.; Vespalec, R.; Albert, D.; Hansen, V.; Vogel, W.
1979-09-01
Neutron spectra were measured for the TR-0, WWR-S and SR-0 experimental reactors using the recoil proton method, 6 Li spectrometry, scintillation spectrometry and activation detectors in a variety of conditions. Neutron fluence was also measured and calculated. (M.S.)
Seasonal and Lunar Month Periods Observed in Natural Neutron Flux at High Altitude
Stenkin, Yuri; Alekseenko, Victor; Cai, Zeyu; Cao, Zhen; Cattaneo, Claudio; Cui, Shuwang; Giroletti, Elio; Gromushkin, Dmitry; Guo, Cong; Guo, Xuewen; He, Huihai; Liu, Ye; Ma, Xinhua; Shchegolev, Oleg; Vallania, Piero; Vigorito, Carlo; Zhao, Jing
2017-07-01
Air radon concentration measurement is useful for research on geophysical effects, but it is strongly sensitive to site geology and many geophysical and microclimatic processes such as wind, ventilation, air humidity and so on inducing very big fluctuations on the concentration of radon in air. On the contrary, monitoring the radon concentration in soil by measuring the thermal neutron flux reduces environmental effects. In this paper, we report some experimental results on the natural thermal neutron flux as well as on the concentration of air radon and its variations at 4300 m asl. These results were obtained with unshielded thermal neutron scintillation detectors (en-detectors) and radon monitors located inside the ARGO-YBJ experimental hall. The correlation of these variations with the lunar month and 1-year period is undoubtedly confirmed. A method for earthquake prediction provided by a global net of en-detectors is currently under study.
Measurement and calculation of fast neutron flux in a zero-energy reactor
International Nuclear Information System (INIS)
Day, D.H.; Fox, W.N.; Hyder, H.R.
1963-05-01
An activation technique for measuring relative fast neutron fluxes is described which has some advantages over the normal method using U238 fission. The technique is based on the formation of Rh 103 after inelastic scattering of neutrons above 100 keV in energy. This isomer decays with a 57.4 minute half-life giving an easily measurable γ-activity. The energy dependence of the inelastic scattering cross-section of Rh 103 is similar to that of the fission cross-section of U 238 thus making the results of direct relevance to reactor calculations. Using the Rh 103 activation technique, measurements have been made of the fast neutron flux distribution in a typical pressure tube heavy water lattice and are compared in this report with theoretical calculations using the MONTE CARLO method. (author)
The analysis of RPV fast neutron flux calculation for PWR with three-dimensional SN method
International Nuclear Information System (INIS)
Yang Shouhai; Chen Yixue; Wang Weijin; Shi Shengchun; Lu Daogang
2011-01-01
Discrete ordinates (S N ) method is one of the most widely used method for reactor pressure vessel (RPV) design. As the fast development of computer CPU speed and memory capacity and consummation of three-dimensional discrete-ordinates method, it is mature for 3-D S N method to be used to engineering design for nuclear facilities. This work was done specifically for PWR model, with the results of 3-D core neutron transport calculation by 3-D core calculation, 3-D RPV fast neutron flux distribution obtain by 3-D S N method were compared with gained by 1-D and 2-D S N method and the 3-D Monte Carlo (MC) method. In this paper, the application of three-dimensional S N method in calculating RPV fast neutron flux distribution for pressurized water reactor (PWR) is presented and discussed. (authors)
High flux polarized neutrons triple-axis spectrometer: 2T (LLB-Saclay)
International Nuclear Information System (INIS)
Bourges, Ph.; Hennion, B.; Sidis, Y.; Boutrouille, Ph.; Baroni, P.
1999-01-01
A description of the performance of the newly designed thermal beam triple-axis spectrometer, 2T at LLB (Saclay) is given. The beam tube will be increased to 50 x 120 mm 2 (HxV) before the monochromator. A gain of about a factor 2 on the neutron flux at the monitor position is expected after this operation, scheduled on April/May 1999. Polarized neutrons beam option will be installed on this triple axis. The polarization is obtained using high quality heusler crystals recently grown at ILL. The size of both heusler monochromator and analyzer have been chosen to fully cover the beam size. The monochromator (analyzer) will be equipped with a vertical (horizontal) curvature. The flux of the polarized beam on the detector is then expected to be 5 times better than IN20 at ILL (best existing polarized neutrons triple-axis on thermal beam) with incident energy upto 75 MeV. (author)
An analytical approach for a nodal scheme of two-dimensional neutron transport problems
International Nuclear Information System (INIS)
Barichello, L.B.; Cabrera, L.C.; Prolo Filho, J.F.
2011-01-01
Research highlights: → Nodal equations for a two-dimensional neutron transport problem. → Analytical Discrete Ordinates Method. → Numerical results compared with the literature. - Abstract: In this work, a solution for a two-dimensional neutron transport problem, in cartesian geometry, is proposed, on the basis of nodal schemes. In this context, one-dimensional equations are generated by an integration process of the multidimensional problem. Here, the integration is performed for the whole domain such that no iterative procedure between nodes is needed. The ADO method is used to develop analytical discrete ordinates solution for the one-dimensional integrated equations, such that final solutions are analytical in terms of the spatial variables. The ADO approach along with a level symmetric quadrature scheme, lead to a significant order reduction of the associated eigenvalues problems. Relations between the averaged fluxes and the unknown fluxes at the boundary are introduced as the usually needed, in nodal schemes, auxiliary equations. Numerical results are presented and compared with test problems.
Neutron flux characterization using Lr-115 NTD and binary glass metal as converter
International Nuclear Information System (INIS)
Alvarado, R.; Palacios, D.; Sajo B, L.; Greaves, E.; Barros, H.; Nemeth, P.; Goncalves, I. F.
2010-01-01
The installed neutron irradiation cell at the Simon Bolivar University Laboratory is employed frequently for applications where the neutron field should be monitored. The ratio of epithermal and thermal neutron flux were determined by the cadmium difference method in three different irradiation locations of a graphite thermalized 252 Cf source. Monte Carlo simulation provided the expected thermal neutron field inside the irradiation cell and the nuclear trace detectors sensitivity. The passive device is a Kodak-Pathe Lr-115 externally coated with a thin (0,1 mm) binary glass metal strip Fe 75 B 25 where the 10 B(n,α) 7 Li reaction converts free neutrons into alpha particles that are registered as etched tracks. The results allow to determine the neutron flux in an easy, reproducible and effective way, with the advantage of a low self-absorption effect due to the converter material. Improved etching process of the Lr-115 nuclear track detectors and the software parameters for track analysis are included. (Author)
Neutron flux measurement in the central channel (XC-1) of TRIGA 14 MW LEU core
International Nuclear Information System (INIS)
BARBOS, D.; BUSUIOC, P.; ROTH, Cs.; PAUNOIU, C.
2008-01-01
The TRIGA 14 MW reactor, operated by Institute for Nuclear Research Pitesti, Romania, is a pool type reactor, and has a rectangular shape which holds fuel bundles and is surrounded with beryllium reflectors. Each fuel bundle is composed of 25 nuclear fuel rods. The TRIGA 14 MW reactor was commissioned 28 years ago with HEU fuel rods. The conversion was gradually achieved, starting in February 1992 and completed in March 2006. The full conversion of the 14 MW TRIGA Research Reactor was completed in May 2006 and each step of the conversion was achieved by removal of HEU fuel, replaced by LEU fuel, accompanied by a large set of theoretical evaluation and physical measurements intended to confirm the performances of gradual conversion. After the core full conversion, a program of measurements and comparisons with previous results of core physics and measurements is underway, allowing data acquisition for normal operation, demonstration of safety and economics of the converted core. Neutron flux spectrum measurements in the XC in the XC-1 water 1 water-filled channel were performed using multi multi-foil activation techniques. The neutron spectra and flux are obtained by unfolding from measured reaction rates using SAND II computer code. The integral neutron flux value for LEU core is greater of 13% than for the standard HEU core. Also thermal neutron flux value for converted LEU core is smaller by 0.38% than for the standard HEU core. These differences appear because the foil activation detectors have been irradiated using a pneumatic rabbit having a diameter of 32 mm, whereas foil irradiations in standard HEU core has been performed with a pneumatic rabbit having a diameter of 14 mm, and therefore the neutron spectra in LEU core is less thermalized and the weight of fast neutron is greater
Finite element method for solving neutron transport problems
International Nuclear Information System (INIS)
Ferguson, J.M.; Greenbaum, A.
1984-01-01
A finite element method is introduced for solving the neutron transport equations. Our method falls into the category of Petrov-Galerkin solution, since the trial space differs from the test space. The close relationship between this method and the discrete ordinate method is discussed, and the methods are compared for simple test problems
Neutron targets of Moscow meson facility status, problems, prospects
Energy Technology Data Exchange (ETDEWEB)
Sidorkin, S.; Koptelov, E.; Perekrestenko, A.; Stavissky, Y.; Trushkin, V.; Sobolevsky, N. [Institute for Nuclear Research RAS, 60-th October Anniversary Prospect, Moscow (Russian Federation)
2001-03-01
The status, problems and possible perspectives of target complexes of the Moscow meson factory is described in the report. The results of test proton beam session to neutron source are analysed. Some technical features of targets and expected modes in the nearest sessions are stated. (author)
Spatial domain decomposition for neutron transport problems
International Nuclear Information System (INIS)
Yavuz, M.; Larsen, E.W.
1989-01-01
A spatial Domain Decomposition method is proposed for modifying the Source Iteration (SI) and Diffusion Synthetic Acceleration (DSA) algorithms for solving discrete ordinates problems. The method, which consists of subdividing the spatial domain of the problem and performing the transport sweeps independently on each subdomain, has the advantage of being parallelizable because the calculations in each subdomain can be performed on separate processors. In this paper we describe the details of this spatial decomposition and study, by numerical experimentation, the effect of this decomposition on the SI and DSA algorithms. Our results show that the spatial decomposition has little effect on the convergence rates until the subdomains become optically thin (less than about a mean free path in thickness)
Reactivity And Neutron Flux At Silicide Fuel Element In The Core Of RSG-GAS
International Nuclear Information System (INIS)
Hamzah, Amir
2000-01-01
In order to 4.8 and 5.2 gr U/cm exp 3 loading of U 3 Si 2 --Al fuel plates characterization, he core reactivity change and neutron flux depression had been done. Control rod calibration method was used to reactivity change measurement and neutron flux distribution was measured using foil activation method. Measurement of insertion of A-type of testing fuel element with U-loading above cannot be done due to technical reason, so the measurement using full type silicide fuel element of 2.96 gr U/cm exp 3 loading. The reactivity change measurement result of insertion in A-9 and C-3 is + 2.67 cent. The flux depression at silicide fuel in A-9 is 1.69 times bigger than oxide and in C-3 is 0.68 times lower than oxide
Use of Germanium as comparator and integral monitor of neutron flux in activation analysis
International Nuclear Information System (INIS)
Furnari, Juan C.; Cohen, Isaac M.; Arribere, Maria A.; Kestelman, Abraham J.
1997-01-01
The possibility of using germanium as monitor of the thermal and epithermal components of the neutron flux, and comparator in parametric activation analysis, is discussed. The advantages and drawbacks associated to the use of this element are commented on, and the comparison with zirconium, in terms of the determination relative error, is performed. The utilisation of germanium as integral flux monitor, including the fast component of the neutron spectrum, is also discussed. Data corresponding to measurements of k 0 factor for the most relevant gamma transitions from Ge-75 and Be-77 are presented, as well as the results of the reference material analysis, employing germanium as flux monitor and comparator in a simultaneous way. (author). 8 refs., 3 figs., 2 tabs
Neutron flux variations near the Earth’s crust. A possible tectonic activity detection
Directory of Open Access Journals (Sweden)
B. M. Kuzhevskij
2003-01-01
Full Text Available The present work contains some results of observations of neutron flux variations near the Earth’s surface. The Earth’s crust is determined to be a significant source of thermal and slow neutrons, originated from the interaction between the nuclei of the elements of the Earth’s crust and the atmosphere and α-particles, produced by decay of radioactive gases (Radon, Thoron and Actinon. In turn, variations of radioactive gases exhalation is connected with geodynamical processes in the Earth’s crust, including tectonic activity. This determined relation between the processes in the Earth’s crust and neutrons’ flux allow to use variations of thermal and slow neutrons’ flux in order to observe increasing tectonic activity and to develop methods for short-term prediction of natural hazards.
ZZ CAD, 51 Neutron-Group, 25 Gamma-Group Albedo Data for 4 Materials from DOT Flux
International Nuclear Information System (INIS)
1992-01-01
A - Description of problem or function: Format: BREESE tape-writing program, MORSE; Number of groups: 51 neutron, 25 gamma-ray group albedo data. Nuclides: 1) 12 inches of water. 2) 12 inches of ordinary concrete. 3) 9 inches of carbon steel (SA508). 4) 1/2 inches of steel over 12 inches of concrete. (O, Ca, Al, C, Si, H, K, Mg, Fe, Na, Mn); Origin: DOT angular flux tape. CAD is a set of 51 neutron, 25 gamma-ray group albedo data for the following four materials: 1) 12 inches of water. 2) 12 inches of ordinary concrete. 3) 9 inches of carbon steel (SA508). 4) 1/2 inches of steel over 12 inches of concrete. The differential angular albedos are a function of the five incident polar directions and 30 reflected directions. B - Method of solution: The data has been generated from a DOT angular flux tape using the code CARP (abstract PSR-0131). C - Restrictions on the complexity of the problem: Since the amount of data is so large, it is necessary to run CARP, using the group reduction option, in order to run a problem on most computers
Analytical calculations of neutron slowing down and transport in the constant-cross-section problem
International Nuclear Information System (INIS)
Cacuci, D.G.
1978-01-01
Some aspects of the problem of neutron slowing down and transport in an infinite medium consisting of a single nuclide that scatters elastically and isotropically and has energy-independent cross sections were investigated. The method of singular eigenfunctions was applied to the Boltzmann equation governing the Laplace transform (with respect to the lethargy variable) of the neutron flux. A new sufficient condition for the convergence of the coefficients of the expansion of the scattering kernel in Legendre polynomials was rigorously derived for this energy-dependent problem. Formulas were obtained for the lethargy-dependent spatial moments of the scalar flux that are valid for medium to large lethargies. In deriving these formulas, use was made of the well-known connection between the spatial moments of the Laplace-transformed scalar flux and the moments of the flux in the ''eigenvalue space.'' The calculations were greatly aided by the construction of a closed general expression for these ''eigenvalue space'' moments. Extensive use was also made of the methods of combinatorial analysis and of computer evaluation, via FORMAC, of complicated sequences of manipulations. For the case of no absorption it was possible to obtain for materials of any atomic weight explicit corrections to the age-theory formulas for the spatial moments M/sub 2n/(u) of the scalar flux that are valid through terms of the order of u -5 . The evaluation of the coefficients of the powers of n, as explicit functions of the nuclear mass, is one of the end products of this investigation. In addition, an exact expression for the second spatial moment, M 2 (u), valid for arbitrary (constant) absorption, was derived. It is now possible to calculate analytically and rigorously the ''age'' for the constant-cross-section problem for arbitrary (constant) absorption and nuclear mass. 5 figures, 1 table
Analytical calculations of neutron slowing down and transport in the constant-cross-section problem
International Nuclear Information System (INIS)
Cacuci, D.G.
1978-04-01
Aspects of the problem of neutron slowing down and transport in an infinite medium consisting of a single nuclide that scatters elastically and isotropically and has energy-independent cross sections were investigated. The method of singular eigenfunctions was applied to the Boltzmann Equation governing the Laplace transform (with respect to the lethargy variable) of the neutron flux. A new sufficient condition for the convergence of the coefficients of the expansion of the scattering kernel in Legendre polynomials was rigorously derived for this energy-dependent problem. Formulas were obtained for the lethargy-dependent spatial moments of the scalar flux that are valid for medium to large lethargies. Use was made of the well-known connection between the spatial moments of the Laplace-transformed scalar flux and the moments of the flux in the ''eigenvalue space.'' The calculations were aided by the construction of a closed general expression for these ''eigenvalue space'' moments. Extensive use was also made of the methods of combinatorial analysis and of computer evaluation of complicated sequences of manipulations. For the case of no absorption it was possible to obtain for materials of any atomic weight explicit corrections to the age-theory formulas for the spatial moments M/sub 2n/(u) of the scalar flux that are valid through terms of the order of u -5 . The evaluation of the coefficients of the powers of n, as explicit functions of the nuclear mass, represent one of the end products of this investigation. In addition, an exact expression for the second spatial moment, M 2 (u), valid for arbitrary (constant) absorption, was derived. It is now possible to calculate analytically and rigorously the ''age'' for the constant-cross-section problem for arbitrary (constant) absorption and nuclear mass. 5 figures, 1 table
Analytical calculations of neutron slowing down and transport in the constant-cross-section problem
Energy Technology Data Exchange (ETDEWEB)
Cacuci, D.G.
1978-04-01
Aspects of the problem of neutron slowing down and transport in an infinite medium consisting of a single nuclide that scatters elastically and isotropically and has energy-independent cross sections were investigated. The method of singular eigenfunctions was applied to the Boltzmann Equation governing the Laplace transform (with respect to the lethargy variable) of the neutron flux. A new sufficient condition for the convergence of the coefficients of the expansion of the scattering kernel in Legendre polynomials was rigorously derived for this energy-dependent problem. Formulas were obtained for the lethargy-dependent spatial moments of the scalar flux that are valid for medium to large lethargies. Use was made of the well-known connection between the spatial moments of the Laplace-transformed scalar flux and the moments of the flux in the ''eigenvalue space.'' The calculations were aided by the construction of a closed general expression for these ''eigenvalue space'' moments. Extensive use was also made of the methods of combinatorial analysis and of computer evaluation of complicated sequences of manipulations. For the case of no absorption it was possible to obtain for materials of any atomic weight explicit corrections to the age-theory formulas for the spatial moments M/sub 2n/(u) of the scalar flux that are valid through terms of the order of u/sup -5/. The evaluation of the coefficients of the powers of n, as explicit functions of the nuclear mass, represent one of the end products of this investigation. In addition, an exact expression for the second spatial moment, M/sub 2/(u), valid for arbitrary (constant) absorption, was derived. It is now possible to calculate analytically and rigorously the ''age'' for the constant-cross-section problem for arbitrary (constant) absorption and nuclear mass. 5 figures, 1 table.
Principles and problems in neutron nuclear data evaluation
International Nuclear Information System (INIS)
Schmidt, J.J.
1967-01-01
The history of neutron nuclear data evaluation is briefly summarized. The physical problems involved in nuclear data evaluation, such as discrepancies and inconsistencies between different experimental data sets and gaps in experimental information, are discussed. The discrepancies in the capture cross-section data for molybdenum and iron are chosen to illustrate the great difficulties in systematizing and automatizing the evaluation process. The technical problems of data evaluation, such as computer storage and the establishment of nuclear data files, are not discussed. (author)
Application of Walsh functions to neutron transport problems. I. Theory
International Nuclear Information System (INIS)
Seed, T.J.; Albrecht, R.W.
1976-01-01
An approximation to the neutron transport equation is made by representing the angular flux with an expansion of the angular dependence in the orthogonal, complete, and binary valued sets of Walsh function. The Walsh approximation is applied to the one-speed, isotropic-scattering, rectangular-geometry form of the neutron transport equation. Sets of partial differential equations for the expansion coefficients are derived along with appropriate boundary conditions for their solution. The sets of the Walsh expansion to one- and two-dimensional forms of the transport equation are also obtained. The two-dimensional expansion coefficient equations are shown to be not only hyperbolic but also transformable to a set of S/sub N/-like equations that are coupled only through the scattering term. Such transformal sets of equations are termed Walsh-derived quadrature sets
Rehm, C
2002-01-01
Next-generation spallation neutron source facilities will offer instruments with unprecedented capabilities through simultaneous enhancement of source power and usage of advanced optical components. The Spallation Neutron Source (SNS), already under construction at Oak Ridge National Laboratory and scheduled to be completed by 2006, will provide greater than an order of magnitude more effective source flux than current state-of-the-art facilities, including the most advanced research reactors. An additional order of magnitude gain is expected through the use of new optical devices and instrumentation concepts. Many instrument designs require supermirror neutron guides with very high critical angles for total reflection. In this contribution, we will discuss how the performance of a modern neutron-scattering instrument depends on the efficiency of these supermirrors. We summarize current limitations of supermirror coatings and outline ideas for enhancing their performance, particularly for improving the reflec...
International Nuclear Information System (INIS)
Perez, G.; Martin, G.; Ceballos, C.; Padron, I.; Shtejer, K.; Perez, N.; Guibert, R.; Ledo, L.M.; Cruz Inclan, Carlos
2001-01-01
The Havana High Flux Neutron Generator facility is an intense neutron source based on a 20 mA duoplasmatron ion source and a 250 kV high voltage power supply. It has been installed in the Neutron Generator Laboratory at the Center of Applied Technologies and Nuclear Research in 1997. This paper deal outlined the future applied program to be carried out in this facility in the next years. The Applied Research Program consists on install two nuclear analytic techniques: the PELAN technique which uses the neutron generator in the pulse mode and the Low Energy PIXE technique which uses the same facility as a low energy proton accelerator for PIXE analysis. (author)
International Nuclear Information System (INIS)
Goncalves, Glenio A.; Bodmann, Bardo; Bogado, Sergio; Vilhena, Marco T.
2008-01-01
Analytical solutions for neutron transport in cylindrical geometry is available for isotropic problems, but to the best of our knowledge for anisotropic problems are not available, yet. In this work, an analytical solution for the neutron transport equation in an infinite cylinder assuming anisotropic scattering is reported. Here we specialize the solution, without loss of generality, for the linearly anisotropic problem using the combined decomposition and HTS N methods. The key feature of this method consists in the application of the decomposition method to the anisotropic problem by virtue of the fact that the inverse of the operator associated to isotropic problem is well know and determined by the HTS N approach. So far, following the idea of the decomposition method, we apply this operator to the integral term, assuming that the angular flux appearing in the integrand is considered to be equal to the HTS N solution interpolated by polynomial considering only even powers. This leads to the first approximation for an anisotropic solution. Proceeding further, we replace this solution for the angular flux in the integral and apply again the inverse operator for the isotropic problem in the integral term and obtain a new approximation for the angular flux. This iterative procedure yields a closed form solution for the angular flux. This methodology can be generalized, in a straightforward manner, for transport problems with any degree of anisotropy. For the sake of illustration, we report numerical simulations for linearly anisotropic transport problems. (author)
Monte Carlo methods for flux expansion solutions of transport problems
International Nuclear Information System (INIS)
Spanier, J.
1999-01-01
Adaptive Monte Carlo methods, based on the use of either correlated sampling or importance sampling, to obtain global solutions to certain transport problems have recently been described. The resulting learning algorithms are capable of achieving geometric convergence when applied to the estimation of a finite number of coefficients in a flux expansion representation of the global solution. However, because of the nonphysical nature of the random walk simulations needed to perform importance sampling, conventional transport estimators and source sampling techniques require modification to be used successfully in conjunction with such flux expansion methods. It is shown how these problems can be overcome. First, the traditional path length estimators in wide use in particle transport simulations are generalized to include rather general detector functions (which, in this application, are the individual basis functions chosen for the flus expansion). Second, it is shown how to sample from the signed probabilities that arise as source density functions in these applications, without destroying the zero variance property needed to ensure geometric convergence to zero error
Energy Technology Data Exchange (ETDEWEB)
Kurosawa, M. [Toshiba Corp., Yokohama (Japan); Tsukiyama, T.; Hayashi, K. [Hitachi Engineering Co. Ltd., Hitachi-shi (Japan)
2001-07-01
A neutron and gamma flux distribution around the core of BWR commercial plant in Japan was calculated, using a three-dimensional transport code, TORT in DOORS32 code system. In the external of the core, the bottom of the model was at an elevation of 150 cm below the bottom of active fuel, the top of the model was at an elevation of the top of the shroud head dome and the radial part of the model was to the outside of the reactor pressure vessel. The top guide beams were modeled explicitly to obtain the neutron and gamma flux distribution both in the beams and outside beams. The each control rod guide tube was also modeled with homogeneous region which included the blade wing and poison tubes so that we could obtain the neutron and gamma flux distribution around the each control rod guide tube. The calculation model mentioned above needed very large memory size which exceeded a few decade giga-bytes. As the using the splicing/coupling method had uncertainly at the splicing/coupling boundary, in this work the calculation was performed without this splicing/coupling method. On the other hand, radioactivity data were measured for a few pieces of the top guide beam, shroud and in-core monitor guide tube in the same plant which was analyzed in the above calculation. So the calculation results were able to be compared with those measured data as benchmarking and at the end of this task, the C/M values at these measured points were obtained and calculation model using TORT was evaluated. (authors)
Çeçen, Yiğit; Gülümser, Tuğçe; Yazgan, Çağrı; Dapo, Haris; Üstün, Mahmut; Boztosun, Ismail
2017-09-01
In cancer treatment, high energy X-rays are used which are produced by linear accelerators (LINACs). If the energy of these beams is over 8 MeV, photonuclear reactions occur between the bremsstrahlung photons and the metallic parts of the LINAC. As a result of these interactions, neutrons are also produced as secondary radiation products (γ,n) which are called photoneutrons. The study aims to map the photoneutron flux distribution within the LINAC bunker via neutron activation analysis (NAA) using indium-cadmium foils. Irradiations made at different gantry angles (0°, 90°, 180° and 270°) with a total of 91 positions in the Philips SLI-25 linear accelerator treatment room and location-based distribution of thermal neutron flux was obtained. Gamma spectrum analysis was carried out with high purity germanium (HPGe) detector. Results of the analysis showed that the maximum neutron flux in the room occurred at just above of the LINAC head (1.2x105 neutrons/cm2.s) which is compatible with an americium-beryllium (Am-Be) neutron source. There was a 90% decrease of flux at the walls and at the start of the maze with respect to the maximum neutron flux. And, just in front of the LINAC door, inside the room, neutron flux was measured less than 1% of the maximum.
Sabaibang, S.; Lekchaum, S.; Tipayakul, C.
2015-05-01
This study is a part of an on-going work to develop a computational model of Thai Research Reactor (TRR-1/M1) which is capable of accurately predicting the neutron flux level and spectrum. The computational model was created by MCNPX program and the CT (Central Thimble) in-core irradiation facility was selected as the location for validation. The comparison was performed with the typical flux measurement method routinely practiced at TRR-1/M1, that is, the foil activation technique. In this technique, gold foil is irradiated for a certain period of time and the activity of the irradiated target is measured to derive the thermal neutron flux. Additionally, the flux measurement with SPND (self-powered neutron detector) was also performed for comparison. The thermal neutron flux from the MCNPX simulation was found to be 1.79×1013 neutron/cm2s while that from the foil activation measurement was 4.68×1013 neutron/cm2s. On the other hand, the thermal neutron flux from the measurement using SPND was 2.47×1013 neutron/cm2s. An assessment of the differences among the three methods was done. The difference of the MCNPX with the foil activation technique was found to be 67.8% and the difference of the MCNPX with the SPND was found to be 27.8%.
Directory of Open Access Journals (Sweden)
Çeçen Yiğit
2017-01-01
Full Text Available In cancer treatment, high energy X-rays are used which are produced by linear accelerators (LINACs. If the energy of these beams is over 8 MeV, photonuclear reactions occur between the bremsstrahlung photons and the metallic parts of the LINAC. As a result of these interactions, neutrons are also produced as secondary radiation products (γ,n which are called photoneutrons. The study aims to map the photoneutron flux distribution within the LINAC bunker via neutron activation analysis (NAA using indium-cadmium foils. Irradiations made at different gantry angles (0°, 90°, 180° and 270° with a total of 91 positions in the Philips SLI-25 linear accelerator treatment room and location-based distribution of thermal neutron flux was obtained. Gamma spectrum analysis was carried out with high purity germanium (HPGe detector. Results of the analysis showed that the maximum neutron flux in the room occurred at just above of the LINAC head (1.2x105 neutrons/cm2.s which is compatible with an americium-beryllium (Am-Be neutron source. There was a 90% decrease of flux at the walls and at the start of the maze with respect to the maximum neutron flux. And, just in front of the LINAC door, inside the room, neutron flux was measured less than 1% of the maximum.
Cosmogenic excess of 40K and the flux of fast neutrons in meteorites
International Nuclear Information System (INIS)
Stegmann, W.; Begemann, F.
1975-01-01
Results are reported of a mass spectrometric investigation of the content and isotopic composition of potassium from the mesosiderite Emery (silicate phase) and the chondrite Elenovka (bulk). Normalized to the Nier value of 39 K/ 41 K = 13.47, the 40 K in Emery (K-content 220 +- 25 ppm) was found to be enriched by (4.03 +- 0.30)%, the potassium from Elenovka (760 +- 50 ppm) to be indistinguishable from terrestrial potassium. Evidence is presented that the excess 40 K in the silicates from Emery (Ca-content 6.06 weight %) has been produced essentially by secondary cosmic ray neutrons via the 40 Ca(n,p)-reaction. The total excess of (2.57 +- 0.39) x 10 14 40 K-atoms/gCa together with the excitation function of the 40 Ca(n,p)-reaction and the neutron flux spectrum of Arnold, Honda and Lal yields a dose of fast neutrons (2 MeV 16 neutrons/cm 2 and an average flux during the cosmic ray exposure age T = (134 +- 12) Myrs of PHI = (17.4 +- 3.1) neutrons/cm 2 sec. (orig./BJ) [de
Calculation of neutron and gamma-ray flux-to-dose-rate conversion factors
International Nuclear Information System (INIS)
Kwon, S.G.; Lee, S.Y.; Yook, C.C.
1981-01-01
This paper presents flux-to-dose-rate conversion factors for neutrons and gamma rays based on the American National Standard Institute (ANSI) N666. These data are used to calculate the dose rate distribution of neutron and gamma ray in radiation fields. Neutron flux-to-dose-rate conversion factors for energies from 2.5 x 10 -8 to 20 MeV are presented; the corresponding energy range for gamma rays is 0.01 to 15 MeV. Flux-to-dose-rate conversion factors were calculated, under the assumption that radiation energy distribution has nonlinearity in the phantom, have different meaning from those values obtained by monoenergetic radiation. Especially, these values were determined with the cross section library. The flux-to-dose-rate conversion factors obtained in this work were in a good agreement to the values presented by ANSI. Those data will be useful for the radiation shielding analysis and the radiation dosimetry in the case of continuous energy distributions. (author)
International Nuclear Information System (INIS)
Mussaeva, M.A.; Zinov'ev, V.; Ibragimova, E.M.; Muminov, M.I.
2006-01-01
Full text: For carrying out experiments in the channels of nuclear reactor, it is necessary to know the distribution of neutron flux and the intensity of accompanying gamma-radiation both in the working and quenched regimes. Dosimetric parameter of transparent dielectrics is based on the effect of monotonous changing of optical absorption or luminescence under neutrons and/or gamma-radiation. While the radioactivity induced in an element monitor is proportional only to a neutron fluence beginning from a threshold energy. Therefore the aim of this work was to determine the values of neutron and gamma-component fluxes separately and evaluate the contribution of each into the defect production in dielectrics. We used very pure quartz glass of KU-1 type, produced in Russian State Optical Institute by fusion from SiCl 4 in the mixed flow of O 2 +H 2 (impurities of Cl and OH up to 10 -2 % and the rest - below 10 -4 %), SiO 2 glasses with 30 % Ba, and also pure Ni wire. Since under irradiation in the working reactor samples were undergone mixed neutron and gamma fluxes, we suggested determination of intensity of gamma-radiation from radio-nuclides (products of uranium fission) after quenching the reactor by the current of ionization chamber and glass dosimeters. Samples of SiO 2 -BaO together with Ni monitors were irradiated for 1 hour in 18 channels of the active zone of the working reactor both in the sealed ampoules and in the contact with water of the 1-st cooling circuit at 40 deg C. The linear dependence of the induced optical density on the absorbed dose of n 0 + γ-radiation was obtained. Ni -monitors not sensitive to γ-radiation gained the induced radioactivity proportional to the absorbed energy of neutron flux above 1 MeV. Neutron fluxes in the 18 channels varied from 9.53·10 11 to 1.21·10 13 cm -2 s -1 corresponding to fluences from 3.43·10 15 to 4.3·10 16 cm -2 . Optical density of band 215 nm ascertained to E ' - center, which is ≡ Si * near oxygen
International Nuclear Information System (INIS)
Marques, Andre Luis Ferreira; Ting, Daniel Kao Sun; Mendonca, Arlindo Gilson
1996-01-01
A calculation methodology of Flux Depression, Self-Shielding and Cadmium Factors is presented, using the ANISN code, for experiments conducted at the IPEN/MB-01 Research Reactor. The correction factors were determined considering thermal neutron flux and 0.125 e 0.250 mm diameter of 197 Au wires. (author)
Measure of thermal neutron flux in the IPEN/MB-01 reactor using 197 Au wire activation detectors
International Nuclear Information System (INIS)
Marques, Andre Luis Ferreira
1995-01-01
This dissertation has aimed at developing a neutron flux measurement technique by means of detectors activation analysis. The main task of this work was the implementation of this thermal neutron flux measurement technique, using gold wires as activation detectors in the IPEN/MB-01 reactor core. The neutron thermal flux spatial distribution was obtained by gold wire activation technique, with wire diameters of 0.125 mm and 0.250 mm in seven selected reactor experimental channels. The values of thermal flux were about 10 9 neutrons/cm 2 .s. This experiment has been the first one conducted with gold wires in the IPEN/MB-01 reactor, being this technique implemented for use by experiments in flux mapping of the core
International Nuclear Information System (INIS)
Soneda, N.; Dohi, K.; Nishida, K.; Nomoto, A.; Iwasaki, M.; Tsuno, S.; Akiyama, T.; Watanabe, S.; Ohta, T.
2011-01-01
Neutron irradiation embrittlement of reactor pressure vessel (RPV) steels is of great concern for the long term operation of light water reactors. In particular, the embrittlement of the RPV steels of pressurized water reactors (PWRs) at very high fluences beyond 6*10 19 n/cm 2 , E > 1 MeV, needs to be understood in more depth because materials irradiated in material test reactors (MTRs) to such high fluences show larger shifts than predicted by current embrittlement correlation equations available worldwide. The primary difference between the irradiation conditions of MTRs and surveillance capsules is the neutron flux. The neutron flux of MTR is typically more than one order of magnitude higher than that of surveillance capsule, but it is not necessarily clear if this difference in neutron flux causes difference in mechanical properties of RPV. In this paper, we perform direct comparison, in terms of mechanical property and microstructure, between the materials irradiated in surveillance capsules and MTRs to clarify the effect of flux at very high fluences and fluxes. We irradiate the archive materials of some of the commercial reactors in Japan in the MTR, LVR-15, of NRI Rez, Czech Republic. Charpy impact test results of the MTR-irradiated materials are compared with the data from surveillance tests. The comparison of the results of microstructural analyses by means of atom probe tomography is also described to demonstrate the similarity / differences in surveillance and MTR-irradiated materials in terms of solute atom behavior. It appears that high Cu material irradiated in a MTR presents larger shifts than those of surveillance data, while low Cu materials present similar embrittlement. The microstructural changes caused by MTR irradiation and surveillance irradiation are clearly different
International Nuclear Information System (INIS)
Muniz, Rafael Oliveira Rondon
2010-01-01
IPEN facility for researches in BNCT (Boron Neutron Capture Therapy) uses IEA-R1 reactor's irradiation channel number 3, where there is a mixed radiation field - neutrons and gamma. The researches in progress require the radiation fields, in the position of the irradiation of sample, to have in its composition maximized thermal neutrons component and minimized, fast and epithermal neutron flux and gamma radiation. This work was developed with the objective of evaluating whether the present radiation field in the facility is suitable for BNCT researches. In order to achieve this objective, a methodology for the dosimetry of thermal neutrons and gamma radiation in mixed fields of high doses, which was not available in IPEN, was implemented in the Center of Nuclear Engineering of IPEN, by using thermoluminescent dosimeters - TLDs 400, 600 and 700. For the measurements of thermal and epithermal neutron flux, activation detectors of gold were used applying the cadmium ratio technique. A cylindrical phantom composed by acrylic discs was developed and tested in the facility and the DOT 3.5. computational code was used in order to obtain theoretical values of neutron flux and the dose along phantom. In the position corresponding to about half the length of the cylinder of the phantom, the following values were obtained: thermal neutron flux (2,52 ± 0,06).10 8 n/cm 2 s, epithermal neutron flux (6,17 ± 0,26).10 7 .10 6 n/cm 2 s, absorbed dose due to thermal neutrons (4,2 ± 1,8)Gy and (10,1 ± 1,3)Gy due to gamma radiation. The obtained values show that the fluxes of thermal and epithermal neutrons flux are appropriate for studies in BNCT, however, the dose due to gamma radiation is high, indicating that the facility should be improved. (author)
Measurement of moderated neutron fluxes using Au-La-Co-Mn sandwich detectors
International Nuclear Information System (INIS)
Costa, L.
1967-03-01
In order to measure the energy spectra of moderated neutrons in reactors or in their protections, it is possible to improve resonating detectors by using three of them together: the difference between the activities of the exterior detectors and that of the central detector is proportional to the neutron flux at the resonance energy. The sensitivity obtained is lower but the energy selectivity is better than with simple detectors. The object of this work has been to develop this technique for four substances: gold, lanthanum, cobalt and manganese which have resonance energies at 4.9 - 73.5 - 132 and 337 eV. (author) [fr
International intercomparison on the neutron flux density spectrum just before the REAL-80 project
International Nuclear Information System (INIS)
Ertek, C.
1981-06-01
This work briefly presents the results of the international intercomparison on the neutron flux density spectrum just before the REAL-80 intercomparison project. Some of the results of this intercomparison with a smaller number of laboratories will be also reflected in the REAL-80 project, therefore, it has some significant issues. This work is performed within the IAEA programme on standardization of reactor radiation measurements, one of the important objectives of which is the assistance of laboratories in Member States to implement or intercompare the multiple foil activation techniques for different neutron field measurements
Measurement of Neutron and Muon Fluxes 100~m Underground with the SciBath Detector
Energy Technology Data Exchange (ETDEWEB)
Garrison, Lance [Indiana Univ., Bloomington, IN (United States)
2014-01-01
The SciBath detector is an 80 liter liquid scintillator detector read out by a three dimensional grid of 768 wavelength-shifting fibers. Initially conceived as a fine-grained charged particle detector for neutrino studies that could image charged particle tracks in all directions, it is also sensitive to fast neutrons (15-200 MeV). In fall of 2011 the apparatus performed a three month run to measure cosmic-induced muons and neutrons 100~meters underground in the FNAL MINOS near-detector area. Data from this run has been analyzed and resulted in measurements of the cosmic muon flux as \
Fission neutrons experiments, evaluation, modeling and open problems
Kornilov, Nikolay
2014-01-01
Although the fission of heavy nuclei was discovered over 75 years ago, many problems and questions still remain to be addressed and answered. The reader will be presented with an old, but persistent problem of this field: The contradiction between Prompt Fission Neutron (PFN) spectra measured with differential (microscopic) experiments and integral (macroscopic and benchmark) experiments (the Micro-Macro problem). The difference in average energy is rather small ~3% but it is stable and we cannot explain the difference due to experimental uncertainties. Can we measure the PFN spectrum with hig
Neutron flux measurement and thermal power calibration of the IAN-R1 TRIGA reactor
Energy Technology Data Exchange (ETDEWEB)
Sarta Fuentes, Jose A.; Castiblanco Bohorquez, Luis A
2008-10-29
The IAN-R1 TRIGA reactor in Colombia was initially fueled with MTR-HEU enriched to 93% U-235, operated since 1965 at 10 kW, and was upgraded to 30 kW in 1980. General Atomics achieved in 1997 the conversion of HEU fuel to LEU fuel TRIGA type, and upgraded the reactor power to 100 kW. Since the IAN-R1 TRIGA reactor was in an extended shutdown during seven years, it was necessary to repeat some results of the commissioning test conducted in 1997. The thermal power calibration was carried out using the calorimetric method. The reactor was operated approximately at 20 kW during 3.5 hours, with manual power corrections since the automatic control system failed and with the forced refrigeration off. During the calorimetric experiment, the pool temperature was measured with a RTD which is installed near to the core. The dates were collected in intervals of 30 minutes. For establishing thermal power reactor, the water temperature versus the running were registered. For a calculated tank volume of 16 m{sup 3}, the tank constant calculated for the IAN-R1 TRIGA reactor is 0.0539 C/kW-hr. The reactor power determined was 19 kW. The core configuration is a rectangular grid plate that holds a combination of 4-rod and 3-rod clusters. The core contains 50 fuel rods with LEU fuel TRIGA (UZr H1.6) type enriched to 19.7%. The radial reflector consists of twenty graphite elements six of which are used for isotope production. The top an bottom reflectors are the cylindrical graphite end reflectors which are installed above and below of the active fuel section in each fuel rod. The spatial dependence of thermal neutron flux was measured axially in the 3-rod clusters 4C, 3D, 5E and in the 4F graphite element. The spatial distribution of the thermal neutron was determined using a self-powered detector and the absolute value of thermal neutron flux was determined by a gold activation detector. The (n, b- ) reaction is applied to determine the relative spatial distribution of thermal
International Nuclear Information System (INIS)
Khattab, K.
2005-03-01
The Miniature Neutron Source Reactor (MNSR) in Syria has five inner irradiation sites in the annulus Beryllium reflectors to analyze the unknown samples using the Neutron Activation Analysis technique and to produce medium and short half life isotopes. The fast neutron flux spectrum has a special importance in the MNSR reactor physics where this spectrum is required to measure the fast neutron flux in the MNSR inner irradiation sites. Hence, calculation of the fast neutron flux spectrum in the MNSR inner irradiation site is conducted in this work using the WIMSD4 code. The energy range is divided in the WIMSD4 to 69 energy groups. The first six energy groups represent the fast neutron ranging from 0.5 to 10 MeV. To calculate the fast neutron flux spectrum in the MNSR inner irradiation site using the WIMSD4 code, the MNSR is modeled as a super unit cell. This cell consists of three regions which are: the homogenized core, annulus Beryllium, and water. The fast neutron spectrum is calculated also using the U 235 fission neutron spectrum approximation. The U 235 fission neutron spectrum agrees very good with the WIMSD4 results when neutron energy exceeds 1 MeV, but it fails when the neutron energy ranges from 0.5 to 1 MeV. The WIMSD4 code is used as well to calculate the microscopic fission cross sections for the U 238 using six energy groups where a unit cell of U 238 is used since the U 238 is usually used to measure the fast neutron flux in the reactor. The macroscopic fission cross sections for the U 238 are calculated first then the microscopic fission cross sections are calculated knowing the U 238 atomic density. (Author)
International Nuclear Information System (INIS)
Khattab, K.
2006-01-01
The Miniature Neutron Source Reactor (MNSR) in Syria has five inner irradiation sites in the annulus Beryllium reflectors to analyze the unknown samples using the Neutron Activation Analysis technique and to produce medium and short half life isotopes. The fast neutron flux spectrum has a special importance in the MNSR reactor physics where this spectrum is required to measure the fast neutron flux in the MNSR inner irradiation sites. Hence, calculation of the fast neutron flux spectrum in the MNSR inner irradiation site is conducted in this work using the WIMSD4 code. The energy range is divided in the WIMSD4 to 69 energy groups. The first six energy groups represent the fast neutron ranging from 0.5 to 10 MeV. To calculate the fast neutron flux spectrum in the MNSR inner irradiation site using the WIMSD4 code, the MNSR is modeled as a super unit cell. This cell consists of three regions which are: the homogenized core, annulus Beryllium, and water. The fast neutron spectrum is calculated also using the U 235 fission neutron spectrum approximation. The U 235 fission neutron spectrum agrees very good with the WIMSD4 results when neutron energy exceeds 1 MeV, but it fails when the neutron energy ranges from 0.5 to 1 MeV. The WIMSD4 code is used as well to calculate the microscopic fission cross sections for the U 238 using six energy groups where a unit cell of U 238 is used since the U 238 is usually used to measure the fast neutron flux in the reactor. The macroscopic fission cross sections for the U 238 are calculated first then the microscopic fission cross sections are calculated knowing the U 238 atomic density. (Author)
International Nuclear Information System (INIS)
Khattab, K.; Ghazi, N.; Omar, H.
2007-01-01
A 3-D neutronic model, using the WIMSD4 and CITATION codes, for the Syrian Miniature Neutron source Reactor (MNSR) is used to calculate the axial thermal neutron flux non-uniform factor in the inner irradiation capsule. The calculated result is 4%. A copper wire is used to measure the axial thermal neutron flux non-uniform factor in the inner irradiation capsule to be compared with the calculated result. The measured result is 5%. Good agreement between the measured and calculated results is obtained. (author)
International Nuclear Information System (INIS)
Khattab, K.; Ghazi, N.; Omar, H.
2007-01-01
A 3-D neutronic model, using the WIMSD4 and CITATION codes, for the Syrian Miniature Neutron Source Reactor (MNSR) is used to calculate the axial thermal neutron flux non-uniform factor in the inner irradiation capsule. The calculated result is 4%. A copper wire is used to measure the axial thermal neutron flux non-uniform factor in the inner irradiation capsule to be compared with the calculated result. The measured result is 5%. Good agreement between the measured and calculated results is obtained
Directory of Open Access Journals (Sweden)
Avramović Ivana
2007-01-01
Full Text Available The H5B is a concept of an accelerator-driven sub-critical research facility (ADSRF being developed over the last couple of years at the Vinča Institute of Nuclear Sciences, Belgrade, Serbia. Using well-known computer codes, the MCNPX and MCNP, this paper deals with the results of a tar get study and neutron flux calculations in the sub-critical core. The neutron source is generated by an interaction of a proton or deuteron beam with the target placed inside the sub-critical core. The results of the total neutron flux density escaping the target and calculations of neutron yields for different target materials are also given here. Neutrons escaping the target volume with the group spectra (first step are used to specify a neutron source for further numerical simulations of the neutron flux density in the sub-critical core (second step. The results of the calculations of the neutron effective multiplication factor keff and neutron generation time L for the ADSRF model have also been presented. Neutron spectra calculations for an ADSRF with an uranium tar get (highest values of the neutron yield for the selected sub-critical core cells for both beams have also been presented in this paper.
Hexagonal tube behaviour in fuel assemblies under neutron flux in a French fast neutron reactor core
International Nuclear Information System (INIS)
Bernard, A.; Ammann, P.
This paper presents what is obtained in the field of the interpretation by calculation of the post irradiation examination of hexagonal tubes, and in the field of prevision by calculation of the behaviour of hexagonal tubes under fast flux [fr
International Nuclear Information System (INIS)
Kim, M.G.
1980-01-01
The multigroup neutron diffusion equations with the constraint of specified power distributions are investigated by the application of the straight-line method which can be considered as the limiting case of zero mesh space in the finite difference method. The standard partial differential form of the diffusion equation is reduced to sets of ordinary differential equations and then converted into sets of integral equations by using Green's functions defined on the pseudo straight lines. Coupling of each straight line to the adjacent lines arises from the application of a three-point central difference formula. The interfaces encountered between two regions are taken into account by imposing the continuity conditions for the grown fluxes and net currents with Taylor expansions of internal fluxes at the interface positions. A few sample problems are selected to test the validity of the method. It is found that the proposed method of solution is similar to the finite Fourier sine transform. Numerical results for the solutions obtained by the method of straight lines are compared with the results of the exact analytical solutions for simple geometries. These comparisons indicate that the proposed method is compatible with the analytical method, and in some problems considered the straight-line solutions are much more efficient than the exact solutions. The method is also extended to the reactor kinetics problem by expressing the kinetics parameters in terms of the basis functions which are used to obtain the solutions of the steady-state neutron diffusion equations
International Nuclear Information System (INIS)
Abdurashitov, D.N.; Gavrin, V.N.; Kalikhov, A.V.; Matushko, V.L.; Shikhin, A.A.; Yants, V.E.; Zaborskaya, O.S.
2001-01-01
The principle of operation, design, registration system and main characteristics of a fast neutron spectrometer are described. The spectrometer is intended for direct measurements of ultra-low fluxes of fast neutrons. It is sensitive to neutron fluxes of 10 -7 cm -2 · s -1 and lower. The detection efficiency of fast neutrons with simultaneous energy measurement was determined from Monte-Carlo simulation to be equal to 0.11 ± 0.01. The background counting rate in the detector corresponds to a neutron flux of (6.5 ± 2.1) · 10 -7 cm -2 · s -1 in the range of 1.0-11.0 MeV. The natural neutron flux from the surrounding mine rock at the depth of 4600 meters of water equivalent was measured to be (7.3 ± 2.4) · 10 -7 cm -2 · s -1 in the interval 1.0 -11.0 MeV. The flux of fast neutrons in the SAGE main room was measured to be 2.3 · 10 -7 cm -2 · s -1 in 1.0 - 11.0 MeV energy range
Abdurashitov, J N; Kalikhov, A V; Matushko, V L; Shikhin, A A; Yants, V E; Zaborskaia, O S
2002-01-01
The principle of operation, design, registration system and main characteristics of a fast neutron spectrometer are described. The spectrometer is intended for direct measurements of ultra low fluxes of fast neutrons. It is sensitive to neutron fluxes of 10 sup - sup 7 cm sup - sup 2 s sup - sup 1 and lower. The detection efficiency of fast neutrons with simultaneous energy measurement was determined from Monte-Carlo simulation to be equal to 0.11+-0.01. The background counting rate in the detector corresponds to a neutron flux of (6.5+-2.1)x10 sup - sup 7 cm sup - sup 2 s sup - sup 1 in the range 1.0-11.0 MeV. The natural neutron flux from the surrounding mine rock at the depth of 4600 m of water equivalent was measured to be (7.3+-2.4)x10 sup - sup 7 cm sup - sup 2 s sup - sup 1 in the range 1.0-11.0 MeV. The flux of fast neutrons in the SAGE main room was measured to be <2.3x10 sup - sup 7 cm sup - sup 2 s sup - sup 1 in 1.0-11.0 MeV energy range.
International Nuclear Information System (INIS)
Khattab, K.; Omar, H.; Ghazi, N.
2007-01-01
The effect of water and fuel temperature increase and changes in the control rod positions on the spatial neutron flux distribution in the Syrian Miniature Neutron Source Reactor (MNSR) is discussed. The cross sections of all the reactor components at different temperatures are generated using the WIMSD4 code. These group constants are used then in the CITATION code to calculate the special neutron flux distribution using four energy groups. This work shows that water and fuel temperature increase in the reactor during the reactor daily operating time does not affect the spatial neutron flux distribution in the reactor. Changing the control rod position does not affect as well the spatial neutron flux distribution except in the region around the control rod position. This stability in the spatial neutron flux distribution, especially in the inner and outer irradiation sites, makes MNSR as a good tool for the neutron activation analysis (NAA) technique and production of radioisotopes with medium or short half lives during the reactor daily operating time. (author)
Evaluation of neutron flux density and power density with SPN-detectors and micro calorimeters
International Nuclear Information System (INIS)
Gehre, G.; Rindelhardt, U.; Seidenkranz, T.; Hogel, J.; Jirousek, V.; Vazek, J.
1983-02-01
During investigations with a special equipped fuel assembly in the Rheinsberg nuclear power station the neutron flux and the power density were evaluated from measurements with SPN-detectors and micro calorimeters. The reliability of both detector types, their measurement accuracy under different physical conditions and the usefulness of the developed calculation models are discussed in detail. The thermal flux and the power density evaluated with SPND's agree well with theoretical results. The values obtained through micro calorimeter measurements are systematic lower by about 18%. This deviation is probably a result of differences in the used calculation models. (author)
The hydrogen anomaly problem in neutron Compton scattering
Karlsson, Erik B.
2018-03-01
Neutron Compton scattering (also called ‘deep inelastic scattering of neutrons’, DINS) is a method used to study momentum distributions of light atoms in solids and liquids. It has been employed extensively since the start-up of intense pulsed neutron sources about 25 years ago. The information lies primarily in the width and shape of the Compton profile and not in the absolute intensity of the Compton peaks. It was therefore not immediately recognized that the relative intensities of Compton peaks arising from scattering on different isotopes did not always agree with values expected from standard neutron cross-section tables. The discrepancies were particularly large for scattering on protons, a phenomenon that became known as ‘the hydrogen anomaly problem’. The present paper is a review of the discovery, experimental tests to prove or disprove the existence of the hydrogen anomaly and discussions concerning its origin. It covers a twenty-year-long history of experimentation, theoretical treatments and discussions. The problem is of fundamental interest, since it involves quantum phenomena on the subfemtosecond time scale, which are not visible in conventional thermal neutron scattering but are important in Compton scattering where neutrons have two orders of magnitude times higher energy. Different H-containing systems show different cross-section deficiencies and when the scattering processes are followed on the femtosecond time scale the cross-section losses disappear on different characteristic time scales for each H-environment. The last section of this review reproduces results from published papers based on quantum interference in scattering on identical particles (proton or deuteron pairs or clusters), which have given a quantitative theoretical explanation both regarding the H-cross-section reduction and its time dependence. Some new explanations are added and the concluding chapter summarizes the conditions for observing the specific quantum
Measurement of the epithermal neutron flux of the Argonauta reactor by the Sandwich method
International Nuclear Information System (INIS)
Nascimento, H.M.
1973-01-01
A common method of obtaining information about the neutron spectrum in the energy range of 1 eV to a few keV is by using resonance sandwich detectors. A sandwich detector is usually made up of three foils placed one on top of the other, each having the same thickness and being made of the same material which has a pronounced absorption resonance. To make an adequate evaluation, the sandwich method was compared with one using an isolated detector. The results obtained from approximate theoretical calculations were checked experimentally, using In, Au and Mn foils, in an isotropic 1/E flux in the Argonaut Reactor at I.E.N. As practical application of this method, the deviation from a 1/E spectrum of the epithermal neutron flux in the core and external graphite reflector of the Argonaut Reactor has been measured with the sandwich foils previously calibrated in a 1/E spectrum. (author)
Energy Technology Data Exchange (ETDEWEB)
Kim, B. G.; Kang, Y. H.; Cho, M. S.; Joo, K. N.; Choi, M. H.; Park, S. J.; Shin, Y. T.; Oh, J. M.; Kim, Y. J
2004-03-01
As a part of the development of instrumentation technologies for a nuclear fuel irradiation test in HANARO(High-flux Advanced Nuclear Application Reactor), a study is performed to measure and evaluate the neutron flux at the same position as the nuclear fuel during irradiation test using the SPND(Self Powered Neutron Detector). To perform this study, rhodium type SPNDs and amplifier are selected suitable to irradiation test, and the selected SPNDs are installed in instrumented fuel capsule(02F-11K). The irradiation test using a instrumented fuel capsule are performed in the OR5 vertical hole of HANARO for about 54 days, and SPND output signals are acquired successfully during irradiation test. Acquired SPND signals are analyzed and evaluated as a reliable data by COSMOS Code. This will be utilized for the fuel related research together with fuel center temperature and reactor operation data.
Neutron flux of 100kW in the irradiation terminals of the IPR-R1 Triga Reactor
International Nuclear Information System (INIS)
Zangirolami, Dante Marco
2009-01-01
In this work, it was carried out a study of the neutron flux in the IPR-R1 TRIGA reactor irradiation facilities: rotary specimen rack (RSR), pneumatic transfer tube two (PTT2) and the central thimble (CT). The objective was to obtain the neutron flux profile on the RSR, which has forty irradiation positions, and also values for the thermal and epithermal neutron fluxes of some RSR positions and also of the PTT2 and of the CT facility. It was applied the neutron activation analysis of a reference material, Al-Au (0.1%) alloy. Irradiations were performed on 16 different dates. It was concluded that for the RSR, the average value of thermal and epithermal neutron fluxes depends on the vertical position of the reactor control rods. Neutron flux variations along the RSR form a characteristic profile, whose values depend on the location of the irradiation position in the reactor core and on the control rods vertical position. In the RSR, the obtained values of thermal and epithermal neutron flux were (8.1 +- 0.3) x 10 11 n.cm -2 .s -1 , and (3.4 +- 0.2)x10 10 n.cm -2 .s -1 , respectively. For the PTT2 and the CT, the values for the epithermal neutron flux were respectively (3.3 +- 0.2) x 10 9 n.cm -2 .s -1 and (2.6 +- 0.1) x 10 11 n.cm -2 .s -1 . For these facilities, the thermal neutron flux was estimated, and the obtained values were (2.4 +- 0.2) x 10 11 n.cm -2 .s -1 and (2.8 +- 0.1)x10 12 n.cm -2 .s -1 for the PTT2 and the CT, respectively. (author)
Neutron flux and annealing effects on irradiation hardening of RPV materials
Chaouadi, R.; Gérard, R.
2011-11-01
This paper aims to examine an eventual effect of neutron flux, sometimes referred to as dose rate effect, on irradiation hardening of a typical A533B reactor pressure vessel steel. Tensile tests on both low flux (reactor surveillance data) and high flux (BR2 reactor) were performed in a large fluence range. The obtained results indicate two features. First, the surveillance data exhibit a constant (˜90 MPa) higher yield strength than the high flux data. However, this difference cannot be explained from a flux effect but most probably from differences in the initial tensile properties. The hardening kinetic of both low and high flux is the same. Annealing at low temperature, 345 °C/40 h, to eventually reveal unstable matrix damage did not affect both BR2 and surveillance specimens. This is confirmed by other annealing experimental data including both tensile and hardness measurements and tensile data on A508 forging and weld. It is suggested that the absence of flux effect on the tensile properties while different radiation-induced microstructures can be attributed to thermal ageing effects.
Neutron Flux Variation in the Nigeria Research Reactor-1 (NIRR-1)
International Nuclear Information System (INIS)
Yahaya, M.; Ahmed, Y.A.
2013-01-01
In order to ascertain the level of flux variation in one of the inner irradiation channels of the Nigeria Research Reactor-1 (NIRR-1), the irradiation container used for routine activation analysis was employed with copper wires as flux monitors. Measurements were carried out with copper wires arranged in axial direction to determine the thermal neutron flux at selected positions using absolute foil activation method. Our results show that there exists a slight flux variation from one position to another ranging from (4.57±0.21) x 10 11 to (5.20± 0.20) x 10 11 cm -2 s -1 .Individual foil shows slight flux variation from one position to another in the same irradiation container but they all pointed toward a level of consistency in variation in spite of the recent installation of the cadmium lined irradiation channel. The values obtained in this work are in good agreement with the previously measured value of (5.14±0.24) x 10 11 cm -2 s -1 after commissioning of NIRR-1 (Jonah et al., 2005). This shows that the cadmium lined installation does not affect the flux stability. In order to improve the accuracy of NAA using NIRR-l facility, there is need for flux corrections to be made by MNSR users during NAA particularly for samples in the axial position for long irradiation.
TMI-2 source and intermediate range neutron flux monitors data report
International Nuclear Information System (INIS)
McCormick, R.D.
1986-03-01
This is a report on the preparation of data from the TMI-2 excore source and intermediate range neutron flux monitors for inclusion into the TMI Data Base. The sources of the as-recorded data are discussed as well as the process of transforming these data into digital form. The corrections to the as-recorded data are given and the data quality classification and uncertainty are established. The identifiers attached to each data set in the TMI Data Base are given
Neutron flux distribution measurement in the elementary cell of the RB reactor
Energy Technology Data Exchange (ETDEWEB)
Takac, S [Boris Kidric Institute of Nuclear Sciences Vinca, Beograd (Yugoslavia)
1963-04-15
The distribution of thermal neutrons was measured in the elementary cell with dysprosium foils for different lattice pitches and the results obtained were given. From the distributions measured average fluxes were determined for every single zone. By using the published data concerning effective absorption cross sections thermal utilization factors were calculated and their changes given as functions of lattice pitches: l = 8.0 cm; 11.3 cm and 17.9 cm. (author)
Design of an automatic sample changer for the measurement of neutron flux by gamma spectrometry
International Nuclear Information System (INIS)
Gago, Javier; Bruna, Ruben; Baltuano, Oscar; Montoya, Eduardo; Descreaux, Killian
2014-01-01
This paper presents calculus, selection and components design for the construction of an automatic system in order to measure neutron flux in a working nuclear reactor by the gamma spectrometry technique using samples irradiated on the RP-10 nucleus. This system will perform the measurement of interchanging 100 samples in a programed and automatic way, reducing operation time by the user and obtaining more accurate measures. (authors).
Application of AC servo motor on the in-core neutron flux instrumentation system
International Nuclear Information System (INIS)
Du Xiaoguang; Wang Mingtao
2010-01-01
The application of ac servo motor in the In-Core Neutron Flux Instrumentation System is described. The hardware component of ac servo motor control system is different from the dc motor control system. The effect of two control system on the instrumentation system is compared. The ac servo motor control system can improve the accuracy of the motion control, optimize the speed control and increase the reliability. (authors)
Koltick, D.; McConchie, S.; Sword, E.
2008-04-01
The design and performance of a pulseable neutron source utilizing a D-T neutron generator and a depleted uranium reflector are presented. Approximately half the generator's 14 MeV neutron flux is used to produce a fission-like neutron spectrum similar to 252Cf. For every 14 MeV neutron entering the reflector, more than one fission-like neutron is reflected back across the surface of the reflector. Because delayed neutron production is more than two orders of magnitude below the prompt neutron production, the source takes full advantage of the generator's pulsed mode capability. Applications include all elemental characterization systems using neutron-induced gamma-ray spectroscopy. The source simultaneously emits 14 MeV neutrons optimal to excite fast neutron-induced gamma-ray signals, such as from carbon and oxygen, and fission-like neutrons optimal to induce neutron capture gamma-ray signals, such as from hydrogen, nitrogen, and chlorine. Experiments were performed, which compare well to Monte Carlo simulations, showing that the uranium reflector enhances capture signals by up to a factor of 15 compared to the absence of a reflector.
Preliminary neutron design of the flux flatter for silicon doping at the RA10
International Nuclear Information System (INIS)
Cintas, A.; Bazzana, S.
2012-01-01
The neutron transmutation doping of silicon (NTD) is one of the facilities under development for the RA10 project. In order to obtain high quality semiconductor, commercial requirements of NTD include achieving high axial and radial uniformity in the silicon targets. Axial uniformity is achieved locating a neutron screen around the Si ingot, obtaining a flat axial distribution of the dopant concentration. We present the neutron design of this screen, also known as flux flattener. MCNP5 was used to model the screen design. We have reached a satisfactory preliminary screen design after numerous iterations. The fluctuation in the axial distribution of the reaction capture rate ( 30 Si(n,γ) 31 Si) is under ≠1,5%, which is the required level by the semiconductor industry to accept the final product (author)
Theoretical and experimental study of collectrons for epithermal neutron flux in reactors
International Nuclear Information System (INIS)
Agu, M.N.
1986-01-01
A theoretical study of nuclear reactions and electric charge displacements arising in sensitivity to thermal and epithermal neutrons in collectrons allowed a computer code conception. Collectrons in Rhodium, Silver, Cobalt, Hafnium, Erbium, Gadolinium and Holmium have been tested in different radiation fields given by neutron or gamma filters irradiated in different places of Melusine and Siloe reactors. Some emitters were covered with different steel, nickel or zircaloy thicknesses. Theoretical and experimental results are consistent; that validate the computer code and show possibilities and necessity of covering collectron emitters to reduce or cancel the gamma sensitivity and to improve response instantaneity. A selective measurement of epithermal neutron flux can by this way, made by associating two types of collectrons [fr
Fourier convergence analysis applied to neutron diffusion Eigenvalue problem
International Nuclear Information System (INIS)
Lee, Hyun Chul; Noh, Jae Man; Joo, Hyung Kook
2004-01-01
Fourier error analysis has been a standard technique for the stability and convergence analysis of linear and nonlinear iterative methods. Though the methods can be applied to Eigenvalue problems too, all the Fourier convergence analyses have been performed only for fixed source problems and a Fourier convergence analysis for Eigenvalue problem has never been reported. Lee et al proposed new 2-D/1-D coupling methods and they showed that the new ones are unconditionally stable while one of the two existing ones is unstable at a small mesh size and that the new ones are better than the existing ones in terms of the convergence rate. In this paper the convergence of method A in reference 4 for the diffusion Eigenvalue problem was analyzed by the Fourier analysis. The Fourier convergence analysis presented in this paper is the first one applied to a neutronics eigenvalue problem to the best of our knowledge
Development of Coincidence Method for Determination Thermal Neutron Flux on RSG-GAS
International Nuclear Information System (INIS)
Bakhri, Syaiful; Hamzah, Amir
2004-01-01
The research to develop detection radiation system using coincidence method has been done to determine thermal neutron flux in RS1 and RS2 irradiation facilities RSG-GAS. At this research has arranged beta-gamma coincidence equipment system and parameter of measurement according to Au-198 beta-gamma spectrum. Gold foils that have irradiated for period of time, counted, and the activities of radiation is analyzed to get neutron flux. Result of research indicate that systems measurement of absolute activity with gamma beta coincidence method functioning well and can be applied at activity measurement of gold foil for irradiation facility characterization. The results show that thermal neutron flux in RS1 and RS2, respectively is 2.007E+12 n/cm 2 s and 2.147E+12 n/cm 2 s. To examine the system performance, the result was compared to measure activity using high resolution of Hp Ge detector and achieved discrepancy is about 1.26% and 6.70%. (author)
International Nuclear Information System (INIS)
Bosevski, T.
1986-01-01
An improved collision probability method for thermal-neutron-flux calculation in a cylindrical reactor cell has been developed. Expanding the neutron flux and source into a series of even powers of the radius, one' gets a convenient method for integration of the one-energy group integral transport equation. It is shown that it is possible to perform an analytical integration in the x-y plane in one variable and to use the effective Gaussian integration over another one. Choosing a convenient distribution of space points in fuel and moderator the transport matrix calculation and cell reaction rate integration were condensed. On the basis of the proposed method, the computer program DISKRET for the ZUSE-Z 23 K computer has been written. The suitability of the proposed method for the calculation of the thermal-neutron-flux distribution in a reactor cell can be seen from the test results obtained. Compared with the other collision probability methods, the proposed treatment excels with a mathematical simplicity and a faster convergence. (author)
Radiation Hardness tests with neutron flux on different Silicon photomultiplier devices
Cattaneo, P. W.; Cervi, T.; Menegolli, A.; Oddone, M.; Prata, M.; Prata, M. C.; Rossella, M.
2017-07-01
Radiation hardness is an important requirement for solid state readout devices operating in high radiation environments common in particle physics experiments. The MEG II experiment, at PSI, Switzerland, investigates the forbidden decay μ+ → e+ γ. Exploiting the most intense muon beam of the world. A significant flux of non-thermal neutrons (kinetic energy Ek>= 0.5 MeV) is present in the experimental hall produced along the beam-line and in the hall itself. We present the effects of neutron fluxes comparable to the MEG II expected doses on several Silicon Photomultiplier (SiPMs). The tested models are: AdvanSiD ASD-NUV3S-P50 (used in MEG II experiment), AdvanSiD ASD-NUV3S-P40, AdvanSiD ASD-RGB3S-P40, Hamamatsu and Excelitas C30742-33-050-X. The neutron source is the thermal Sub-critical Multiplication complex (SM1) moderated with water, located at the University of Pavia (Italy). We report the change of SiPMs most important electric parameters: dark current, dark pulse frequency, gain, direct bias resistance, as a function of the integrated neutron fluency.
High-Flux Neutron Generator Facility for Geochronology and Nuclear Physics Research
Waltz, Cory; HFNG Collaboration
2015-04-01
A facility based on a next-generation, high-flux D-D neutron generator (HFNG) is being commissioned at UC Berkeley. The generator is designed to produce monoenergetic 2.45 MeV neutrons at outputs exceeding 1011 n/s. The HFNG is designed around two RF-driven multi-cusp ion sources that straddle a titanium-coated copper target. D + ions, accelerated up to 150 keV from the ion sources, self-load the target and drive neutron generation through the d(d,n)3 He fusion reaction. A well-integrated cooling system is capable of handling beam power reaching 120 kW impinging on the target. The unique design of the HFNG target permits experimental samples to be placed inside the target volume, allowing the samples to receive the highest neutron flux (1011 cm-2 s-1) possible from the generator. In addition, external beams of neutrons will be available simultaneously, ranging from thermal to 2.45 MeV. Achieving the highest neutron yields required carefully designed schemes to mitigate back-streaming of high energy electrons liberated from the cathode target by deuteron bombardment. The proposed science program is focused on pioneering advances in the 40 Ar/39 Ar dating technique for geochronology, new nuclear data measurements, basic nuclear science, and education. An end goal is to become a user facility for researchers. This work is supported by NSF Grant No. EAR-0960138, U.S. DOE LBNL Contract No. DE-AC02-05CH11231, U.S. DOE LLNL Contract No. DE-AC52-07NA27344, and UC Office of the President Award 12-LR-238745.
Criticality problems in energy dependent neutron transport theory
International Nuclear Information System (INIS)
Victory, H.D. Jr.
1979-01-01
The criticality problem is considered for energy dependent neutron transport in an isotropically scattering, homogeneous slab. Under a positivity assumption on the scattering kernel, an expression can be found relating the thickness of the slab to a parameter characterizing production by fission. This is accomplished by exploiting the Perron-Frobenius-Jentsch characterization of positive operators (i.e. those leaving invariant a normal, reproducing cone in a Banach space). It is pointed out that those techniques work for classes of multigroup problems were the Case singular eigenfunction approach is not as feasible as in the one-group theory, which is also analyzed
The problem of the black plate with zero thickness and finite width in neutron transport theory
International Nuclear Information System (INIS)
Benoist, Pierre.
1979-08-01
A black plate with zero thickness, finite width and infinite height, imbedded in an infinite and homogeneous medium which scatters and absorbs neutrons, is considered. The problem is time-independent and the neutrons, which are supposed to have a unique speed, are issued, either from a current at infinity (problem A), or from a uniform source (problem B). It is shown that the Csub(N) method seems to be particularly well suited to the resolution of this 'two-dimensional Milne problem'. A particular interest is attached to the determination of the radius R of the black cylinder leading to the same polar behaviour of the flux at infinity as the plate (criterion 1), or absorbing the same number of neutrons as the plate (criterion 2). In this preliminary report, values of R are calculated in various limit cases: the width of the plate being taken equal to one, l being the mean free path and c the number of secondaries par collision in the outer medium, R is calculated at first in the limit l → 0 (for c = 1) by the theory of Musklelishvili, and then in the limit l → infinity (whatever c is) and c → 0 (whatever l is). In the limit c → 1 (whatever l is), R is shown to be the same in problems A and B and criteria 1 and 2. On the other hand, whatever l and c are; the values of R obtained in the problem A with the criterion 2 and in the problem B with the criterion 1 are shown to be equal. All these results allow henceforth a reasonable interpolation which can be useful in the practice [fr
Thermal neutron flux measurements in the rotary specimen rack of the IPR-R1 TRIGA reactor
Energy Technology Data Exchange (ETDEWEB)
Souza, Rose Mary G. do Prado; Rodrigues, Rogério R.; Souza, Luiz Claudio A., E-mail: souzarm@cdtn.br, E-mail: rrr@cdtn.br, E-mail: lcas@cdtn.br [Centro de Desenvolvimento da Tecnologia Nuclear (CDTN/CNEN-MG), Belo Horizonte, MG (Brazil)
2017-07-01
The thermal neutron flux in the rotary specimen rack of the IPR-R1 TRIGA reactor at the Nuclear Technology Development Center (CDTN), Belo Horizonte, Brazil, has been measured by the neutron activation method, using bare and cadmium covered gold foils. Those foils were irradiated in the rotary specimen rack with the reactor at 100 kW. The reactor core configuration has 63 fuel elements, composed of 59 original aluminum-clad elements and 4 stainless steel-clad fuel elements. The gamma activities of the foils were measured using Ge spectrometer. The perturbations of the thermal neutron flux caused by the introduction of an absorbing foil into the medium were considered in order to obtain accurate determination of the flux. The thermal neutron flux obtained was 7.4 x 10{sup 11} n.cm{sup -2}.s{sup -1}. (author)
International Nuclear Information System (INIS)
Shimada, Masashi; Hatano, Y.; Calderoni, P.; Oda, T.; Oya, Y.; Sokolov, M.; Zhang, K.; Cao, G.; Kolasinski, R.; Sharpe, J.P.
2011-01-01
With the Japan-US joint research project Tritium, Irradiations, and Thermofluids for America and Nippon (TITAN), an initial set of tungsten samples (99.99% purity, A.L.M.T. Co.) were irradiated by high flux neutrons at 323 K to 0.025 dpa in High Flux Isotope Reactor (HFIR) at Oak Ridge National Laboratory (ORNL). Subsequently, one of the neutron-irradiated tungsten samples was exposed to a high-flux deuterium plasma (ion flux: 5 x 10 21 m -2 s -1 , ion fluence: 4 x 10 25 m -2 ) in the Tritium Plasma Experiment (TPE) at Idaho National Laboratory (INL). The deuterium retention in the neutron-irradiated tungsten was 40% higher in comparison to the unirradiated tungsten. The observed broad desorption spectrum from neutron-irradiated tungsten and associated TMAP modeling of the deuterium release suggest that trapping occurs in the bulk material at more than three different energy sites.
Shimada, Masashi; Hatano, Y.; Calderoni, P.; Oda, T.; Oya, Y.; Sokolov, M.; Zhang, K.; Cao, G.; Kolasinski, R.; Sharpe, J. P.
2011-08-01
With the Japan-US joint research project Tritium, Irradiations, and Thermofluids for America and Nippon (TITAN), an initial set of tungsten samples (99.99% purity, A.L.M.T. Co.) were irradiated by high flux neutrons at 323 K to 0.025 dpa in High Flux Isotope Reactor (HFIR) at Oak Ridge National Laboratory (ORNL). Subsequently, one of the neutron-irradiated tungsten samples was exposed to a high-flux deuterium plasma (ion flux: 5 × 1021 m-2 s-1, ion fluence: 4 × 1025 m-2) in the Tritium Plasma Experiment (TPE) at Idaho National Laboratory (INL). The deuterium retention in the neutron-irradiated tungsten was 40% higher in comparison to the unirradiated tungsten. The observed broad desorption spectrum from neutron-irradiated tungsten and associated TMAP modeling of the deuterium release suggest that trapping occurs in the bulk material at more than three different energy sites.
Energy Technology Data Exchange (ETDEWEB)
Bryant, Rebecca [Bryant Research, LLC; Kszos, Lynn A [ORNL
2011-03-01
Oak Ridge National Laboratory (ORNL) offers the scientific community unique access to two types of world-class neutron sources at a single site - the Spallation Neutron Source (SNS) and the High Flux Isotope Reactor (HFIR). The 85-MW HFIR provides one of the highest steady-state neutron fluxes of any research reactor in the world, and the SNS is one of the world's most intense pulsed neutron beams. Management of these two resources is the responsibility of the Neutron Sciences Directorate (NScD). NScD commissioned this survey research to develop baseline information regarding awareness of and perceptions about neutron science. Specific areas of investigative interest include the following: (1) awareness levels among those in the scientific community about the two neutron sources that ORNL offers; (2) the level of understanding members of various scientific communities have regarding benefits that neutron scattering techniques offer; and (3) any perceptions that negatively impact utilization of the facilities. NScD leadership identified users of two light sources in North America - the Advanced Photon Source (APS) at Argonne National Laboratory and the National Synchrotron Light Source (NSLS) at Brookhaven National Laboratory - as key publics. Given the type of research in which these scientists engage, they would quite likely benefit from including the neutron techniques available at SNS and HFIR among their scientific investigation tools. The objective of the survey of users of APS, NSLS, SNS, and HFIR was to explore awareness of and perceptions regarding SNS and HFIR among those in selected scientific communities. Perceptions of SNS and FHIR will provide a foundation for strategic communication plan development and for developing key educational messages. The survey was conducted in two phases. The first phase included qualitative methods of (1) key stakeholder meetings; (2) online interviews with user administrators of APS and NSLS; and (3) one
International Nuclear Information System (INIS)
Bryant, Rebecca; Kszos, Lynn A.
2011-01-01
Oak Ridge National Laboratory (ORNL) offers the scientific community unique access to two types of world-class neutron sources at a single site - the Spallation Neutron Source (SNS) and the High Flux Isotope Reactor (HFIR). The 85-MW HFIR provides one of the highest steady-state neutron fluxes of any research reactor in the world, and the SNS is one of the world's most intense pulsed neutron beams. Management of these two resources is the responsibility of the Neutron Sciences Directorate (NScD). NScD commissioned this survey research to develop baseline information regarding awareness of and perceptions about neutron science. Specific areas of investigative interest include the following: (1) awareness levels among those in the scientific community about the two neutron sources that ORNL offers; (2) the level of understanding members of various scientific communities have regarding benefits that neutron scattering techniques offer; and (3) any perceptions that negatively impact utilization of the facilities. NScD leadership identified users of two light sources in North America - the Advanced Photon Source (APS) at Argonne National Laboratory and the National Synchrotron Light Source (NSLS) at Brookhaven National Laboratory - as key publics. Given the type of research in which these scientists engage, they would quite likely benefit from including the neutron techniques available at SNS and HFIR among their scientific investigation tools. The objective of the survey of users of APS, NSLS, SNS, and HFIR was to explore awareness of and perceptions regarding SNS and HFIR among those in selected scientific communities. Perceptions of SNS and FHIR will provide a foundation for strategic communication plan development and for developing key educational messages. The survey was conducted in two phases. The first phase included qualitative methods of (1) key stakeholder meetings; (2) online interviews with user administrators of APS and NSLS; and (3) one-on-one interviews
Domain decomposition methods for the neutron diffusion problem
International Nuclear Information System (INIS)
Guerin, P.; Baudron, A. M.; Lautard, J. J.
2010-01-01
The neutronic simulation of a nuclear reactor core is performed using the neutron transport equation, and leads to an eigenvalue problem in the steady-state case. Among the deterministic resolution methods, simplified transport (SPN) or diffusion approximations are often used. The MINOS solver developed at CEA Saclay uses a mixed dual finite element method for the resolution of these problems. and has shown his efficiency. In order to take into account the heterogeneities of the geometry, a very fine mesh is generally required, and leads to expensive calculations for industrial applications. In order to take advantage of parallel computers, and to reduce the computing time and the local memory requirement, we propose here two domain decomposition methods based on the MINOS solver. The first approach is a component mode synthesis method on overlapping sub-domains: several Eigenmodes solutions of a local problem on each sub-domain are taken as basis functions used for the resolution of the global problem on the whole domain. The second approach is an iterative method based on a non-overlapping domain decomposition with Robin interface conditions. At each iteration, we solve the problem on each sub-domain with the interface conditions given by the solutions on the adjacent sub-domains estimated at the previous iteration. Numerical results on parallel computers are presented for the diffusion model on realistic 2D and 3D cores. (authors)
International Nuclear Information System (INIS)
Osorio, A.M.; Hadler, J.C.; Iunes, P.J.; Paulo, S.R. de
1993-06-01
In order to use the fission track dating method the flux gradient was verified within the sample holder, in some irradiation positions of the IEA-R1 reactor at IPEN/CNEN, Sao Paulo. The fission track dating method considers only the thermal neutron fission tracks, to subtract the other contributions sample irradiations with a cadmium cover was performed. The neutron flux cadmium influence was studied. (author)
International Nuclear Information System (INIS)
Koch, R.; Findeisen, A.
1986-04-01
The multigroup neutron diffusion theory code SNAP-3D has been used to perform time independent neutron flux and power calculations of the 10 MW Rossendorf research reactor of the type WWR-SM. The report describes these calculations, as well as the actual reactor configuration, some details of the code SNAP-3D, and two- and three-dimensional reactor models. For evaluating the calculations some flux values and control rod worths have been compared with those of measurements. (author)
Khattab, K; Sulieman, I
2009-04-01
The MCNP-4C code, based on the probabilistic approach, was used to model the 3D configuration of the core of the Syrian miniature neutron source reactor (MNSR). The continuous energy neutron cross sections from the ENDF/B-VI library were used to calculate the thermal and fast neutron fluxes in the inner and outer irradiation sites of MNSR. The thermal fluxes in the MNSR inner irradiation sites were also measured experimentally by the multiple foil activation method ((197)Au (n, gamma) (198)Au and (59)Co (n, gamma) (60)Co). The foils were irradiated simultaneously in each of the five MNSR inner irradiation sites to measure the thermal neutron flux and the epithermal index in each site. The calculated and measured results agree well.
Basiri, H.; Tavakoli-Anbaran, H.
2018-01-01
Am-Be neutrons source is based on (α, n) reaction and generates neutrons in the energy range of 0-11 MeV. Since the thermal neutrons are widely used in different fields, in this work, we investigate how to improve the source configuration in order to increase the thermal flux. These suggested changes include a spherical moderator instead of common cylindrical geometry, a reflector layer and an appropriate materials selection in order to achieve the maximum thermal flux. All calculations were done by using MCNP1 Monte Carlo code. Our final results indicated that a spherical paraffin moderator, a layer of beryllium as a reflector can efficiently increase the thermal neutron flux of Am-Be source.
Computation of 3D neutron fluxes in one pin hexagonal cell
International Nuclear Information System (INIS)
Prabha, Hem; Marleau, Guy
2013-01-01
Highlights: ► Computations of 3D neutron fluxes in one pin hexagonal cell is performed by Carlvik’s method of collision probability. ► Carlvik’s method requires computation of track lengths in the geometry. ► Equations are developed to compute tracks, in 2D and 3D, in hexagons and are implemented in a program HX7. ► The program HX7 is implemented in NXT module of the code DRAGON, where tracks in pins are computed. ► The tracks are plotted and fluxes are compared with the EXCELT module of the code DRAGON. - Abstract: In this paper we are presenting the method of computation of three dimensional (3D) neutron fluxes in one pin hexagonal cell. Carlvik’s collision probability method of solving neutron transport equation for computing fluxes has been used here. This method can consider exact geometrical details of the given geometry. While using this method, track length computations are required to be done. We have described here the method of computing tracks in one 3D hexagon. A program HX7 has been developed for this purpose. This program has been implemented in the NXT module of the code DRAGON, where tracks in the pins are computed. For computing tracks in 3D, first we use the tracks computed in the two dimensions (2D) and then we project them in the third dimension. We have developed equations for this purpose. In both the regions, fuel pin as well as in the moderator surrounding the pin the fluxes are assumed to be uniform. A uniform source is assumed in the moderator region. Reflecting boundary conditions are applied on all the sides as well as on the top and bottom surfaces. One group 2D and 3D fluxes are compared with the respective results obtained by the EXCELT module of DRAGON. To check the computations, tracks are plotted and errors in the computations are obtained. It is observed by using both the modules EXCELT and NXT that the fluxes in the pins converge faster and in the moderator region fluxes converge very slowly
Pellet by pellet neutron flux calculations coupled with nodal expansion method
International Nuclear Information System (INIS)
Aldo, Dall'Osso
2003-01-01
We present a technique whose aim is to replace 2-dimensional pin by pin de-homogenization, currently done in core reactor calculations with the nodal expansion method (NEM), by a 3-dimensional finite difference diffusion calculation. This fine calculation is performed as a zoom in each node taking as boundary conditions the results of the NEM calculations. The size of fine mesh is of the order of a fuel pellet. The coupling between fine and NEM calculations is realised by an albedo like boundary condition. Some examples are presented showing fine neutron flux shape near control rods or assembly grids. Other fine flux behaviour as the thermal flux rise in the fuel near the reflector is emphasised. In general the results show the interest of the method in conditions where the separability of radial and axial directions is not granted. (author)
The neutron small-angle camera D11 at the high-flux reactor, Grenoble
International Nuclear Information System (INIS)
Ibel, K.
1976-01-01
The neutron small-angle scattering system at the high-flux reactor in Grenoble consists of three major parts: the supply of cold neutrons via bent neutron guides; the small-angle camera D11; and the data handling facilities. The camera D11 has an overall length of 80 m. The effective length of the camera is variable. The full length of the collimator before the fixed sample position can be reduced by movable neutron guides; the second flight path of 40 m full length contains detector sites in various positions. Thus a large range of momentum transfers can be used with the same relative resolution. Scattering angles between 5 x 10 -4 and 0.5 rad and neutron wavelengths from 0.2 to 2.0 nm are available. A large-area position-sensitive detector is used which allows simultaneous recording of intensities scattered at different angles; it is a multiwire proportional chamber. 3808 elements of 1 cm 2 are arranged in a two-dimensional matrix. (Auth.)
Geslot, B.; Vermeeren, L.; Filliatre, P.; Lopez, A. Legrand; Barbot, L.; Jammes, C.; Bréaud, S.; Oriol, L.; Villard, J.-F.
2011-03-01
Flux monitoring is of great interest for experimental studies in material testing reactors. Nowadays, only the thermal neutron flux can be monitored on line, e.g., using fission chambers or self-powered neutron detectors. In the framework of the Joint Instrumentation Laboratory between SCK-CEN and CEA, we have developed a fast neutron detector system (FNDS) capable of measuring on line the local high-energy neutron flux in fission reactor core and reflector locations. FNDS is based on fission chambers measurements in Campbelling mode. The system consists of two detectors, one detector being mainly sensitive to fast neutrons and the other one to thermal neutrons. On line data processing uses the CEA depletion code DARWIN in order to disentangle fast and thermal neutrons components, taking into account the isotopic evolution of the fissile deposit. The first results of FNDS experimental test in the BR2 reactor are presented in this paper. Several fission chambers have been irradiated up to a fluence of about 7 × 1020 n/cm2. A good agreement (less than 10% discrepancy) was observed between FNDS fast flux estimation and reference flux measurement.
Energy Technology Data Exchange (ETDEWEB)
Geslot, B.; Filliatre, P.; Barbot, L.; Jammes, C.; Breaud, S.; Oriol, L.; Villard, J.-F. [CEA, DEN, Cadarache, SPEx/LDCI, F-13108 Saint-Paul-lez-Durance (France); Vermeeren, L. [SCK-CEN, Boeretang 200, B-2400 Mol (Belgium); Lopez, A. Legrand [CEA, DEN, Saclay, SIREN/LECSI, F-91400 Saclay (France)
2011-03-15
Flux monitoring is of great interest for experimental studies in material testing reactors. Nowadays, only the thermal neutron flux can be monitored on line, e.g., using fission chambers or self-powered neutron detectors. In the framework of the Joint Instrumentation Laboratory between SCK-CEN and CEA, we have developed a fast neutron detector system (FNDS) capable of measuring on line the local high-energy neutron flux in fission reactor core and reflector locations. FNDS is based on fission chambers measurements in Campbelling mode. The system consists of two detectors, one detector being mainly sensitive to fast neutrons and the other one to thermal neutrons. On line data processing uses the CEA depletion code DARWIN in order to disentangle fast and thermal neutrons components, taking into account the isotopic evolution of the fissile deposit. The first results of FNDS experimental test in the BR2 reactor are presented in this paper. Several fission chambers have been irradiated up to a fluence of about 7 x 10{sup 20} n/cm{sup 2}. A good agreement (less than 10% discrepancy) was observed between FNDS fast flux estimation and reference flux measurement.
International Nuclear Information System (INIS)
Geslot, B.; Filliatre, P.; Barbot, L.; Jammes, C.; Breaud, S.; Oriol, L.; Villard, J.-F.; Vermeeren, L.; Lopez, A. Legrand
2011-01-01
Flux monitoring is of great interest for experimental studies in material testing reactors. Nowadays, only the thermal neutron flux can be monitored on line, e.g., using fission chambers or self-powered neutron detectors. In the framework of the Joint Instrumentation Laboratory between SCK-CEN and CEA, we have developed a fast neutron detector system (FNDS) capable of measuring on line the local high-energy neutron flux in fission reactor core and reflector locations. FNDS is based on fission chambers measurements in Campbelling mode. The system consists of two detectors, one detector being mainly sensitive to fast neutrons and the other one to thermal neutrons. On line data processing uses the CEA depletion code DARWIN in order to disentangle fast and thermal neutrons components, taking into account the isotopic evolution of the fissile deposit. The first results of FNDS experimental test in the BR2 reactor are presented in this paper. Several fission chambers have been irradiated up to a fluence of about 7 x 10 20 n/cm 2 . A good agreement (less than 10% discrepancy) was observed between FNDS fast flux estimation and reference flux measurement.
Directory of Open Access Journals (Sweden)
Burnham Steven
2017-01-01
Full Text Available Neutron activation analysis is widely used for identification of elements and their quantities even in trace amounts in the samples of almost any type. The challenges in detecting trace amounts of particular elements are often associated with the neutron flux produced at the research reactors. Low-flux neutron activation analysis usually presents the biggest challenge when analyzing trace quantities of elements with lower magnitude of radiative capture cross-sections. In this paper, we present the methodology and the quantified uncertainties associated with the detection of trace amounts of cobalt and europium, using as an example concrete aggregates. Recent growing interest is in improving structural concrete (increasing its strength but reducing its activation in nuclear power plant environments. Aside from buildings, structural concrete is also used as a biological shield in nuclear power plant that become radioactive after exposure to neutron flux. Due to radiative capture interactions, artificial radionuclides are generated to high enough concentrations that classify concrete as low-level radioactive waste at the time of the plant's decommissioning. Disposal of this concrete adds to the expense of nuclear power plant financing and its construction. Three radionuclides, 60Co, 152Eu, and 154Eu, account for 99 % of total residual radioactivity of nuclear power plant decommissioned concrete. IAEA document RS-G-1.7, Application of the Concepts of Exclusion, Exemption, and Clearance, specifies clearance levels of radionuclides specific activities: a specific activity lower than 0.1 Bqg-1 for 60Co and 152Eu, and 154Eu allows for a concrete to be recycled after decommissioning of the nuclear power plant. Therefore, low-flux neutron activation analysis is used to test the detection limits of trace elements in samples of cement, coarse, and fine concrete aggregates. These samples are irradiated at the University of Utah's 100 kW TRIGA Reactor at
International Nuclear Information System (INIS)
Dias, M.S.; Koskinas, M.F.; Berretta, J.R.; Fratin, L.; Botelho, S.
1990-01-01
The thermal, epithermal and fast neutron fluxes have been determined experimentally by the activation foil method at position GI, located near the IEA-R1 reactor core. The reactions used were 197 Au (n,gamma) 198 Au, for thermal and epithermal neutrons and 27 Na (n,alpha) 24 Na, for fast neutrons. The activities were measured by the 4π(PC)β-γ coincidence method. (author)
Energy Technology Data Exchange (ETDEWEB)
Moran, Jean M. [Univ. of Idaho, Idaho Falls, ID (United States)
1992-02-01
Calculations of radiation flux and dose distributions for Boron Neutron Capture Therapy (BNCT) of brain tumors are typically performed using sophisticated three-dimensional analytical models based on either a homogeneous approximation or a simplified few-region approximation to the actual highly-heterogeneous geometry of the irradiation volume. Such models should be validated by comparison with calculations using detailed models in which all significant macroscopic tissue heterogeneities and geometric structures are explicitly represented as faithfully as possible. This work describes a validation exercise for BNCT of canine brain tumors. Geometric measurements of the canine anatomical structures of interest for this work were performed by dissecting and examining two essentially identical Labrador Retriever heads. Chemical analyses of various tissue samples taken during the dissections were conducted to obtain measurements of elemental compositions for tissues of interest. The resulting geometry and tissue composition data were then used to construct a detailed heterogeneous calculational model of the Labrador Retriever head. Calculations of three-dimensional radiation flux distributions pertinent to BNCT were performed for the model using the TORT discrete-ordinates radiation transport code. The calculations were repeated for a corresponding volume-weighted homogeneous tissue model. Comparison of the results showed that the peak neutron and photon flux magnitudes were quite similar for the two models (within 5%), but that the spatial flux profiles were shifted in the heterogeneous model such that the fluxes in some locations away from the peak differed from the corresponding fluxes in the homogeneous model by as much as 10-20%. Differences of this magnitude can be therapeutically significant, emphasizing the need for proper validation of simplified treatment planning models.
International Nuclear Information System (INIS)
Moran, J.M.
1992-02-01
Calculations of radiation flux and dose distributions for Boron Neutron Capture Therapy (BNCT) of brain tumors are typically performed using sophisticated three-dimensional analytical models based on either a homogeneous approximation or a simplified few-region approximation to the actual highly-heterogeneous geometry of the irradiation volume. Such models should be validated by comparison with calculations using detailed models in which all significant macroscopic tissue heterogeneities and geometric structures are explicitly represented as faithfully as possible. This work describes a validation exercise for BNCT of canine brain tumors. Geometric measurements of the canine anatomical structures of interest for this work were performed by dissecting and examining two essentially identical Labrador Retriever heads. Chemical analyses of various tissue samples taken during the dissections were conducted to obtain measurements of elemental compositions for tissues of interest. The resulting geometry and tissue composition data were then used to construct a detailed heterogeneous calculational model of the Labrador Retriever head. Calculations of three-dimensional radiation flux distributions pertinent to BNCT were performed for the model using the TORT discrete-ordinates radiation transport code. The calculations were repeated for a corresponding volume-weighted homogeneous tissue model. Comparison of the results showed that the peak neutron and photon flux magnitudes were quite similar for the two models (within 5%), but that the spatial flux profiles were shifted in the heterogeneous model such that the fluxes in some locations away from the peak differed from the corresponding fluxes in the homogeneous model by as much as 10-20%. Differences of this magnitude can be therapeutically significant, emphasizing the need for proper validation of simplified treatment planning models
International Nuclear Information System (INIS)
Drozdowicz, K.
1999-01-01
Macroscopic parameters for a description of the thermal neutron transport in finite volumes are considered. A very good correspondence between the theoretical and experimental parameters of hydrogenous media is attained. Thermal neutrons in the medium possess an energy distribution, which is dependent on the size (characterized by the geometric buckling) and on the neutron transport properties of the medium. In a hydrogenous material the thermal neutron transport is dominated by the scattering cross section which is strongly dependent on energy. A monoenergetic treatment of the thermal neutron group (admissible for other materials) leads in this case to a discrepancy between theoretical and experimental results. In the present paper the theoretical definitions of the pulsed thermal neutron parameters (the absorption rate, the diffusion coefficient, and the diffusion cooling coefficient) are based on Nelkin's analysis of the decay of a neutron pulse. Problems of the experimental determination of these parameters for a hydrogenous medium are discussed. A theoretical calculation of the pulsed parameters requires knowledge of the scattering kernel. For thermal neutrons it is individual for each hydrogenous material because neutron scattering on hydrogen nuclei bound in a molecule is affected by the molecular dynamics (characterized with internal energy modes which are comparable to the incident neutron energy). Granada's synthetic model for slow-neutron scattering is used. The complete up-dated formalism of calculation of the energy transfer scattering kernel after this model is presented in the paper. An influence of some minor variants within the model on the calculated differential and integral neutron parameters is shown. The theoretical energy-dependent scattering cross section (of Plexiglas) is compared to experimental results. A particular attention is paid to the calculation of the diffusion cooling coefficient. A solution of an equation, which determines the
Nodal methods for problems in fluid mechanics and neutron transport
International Nuclear Information System (INIS)
Azmy, Y.Y.
1985-01-01
A new high-accuracy, coarse-mesh, nodal integral approach is developed for the efficient numerical solution of linear partial differential equations. It is shown that various special cases of this general nodal integral approach correspond to several high efficiency nodal methods developed recently for the numerical solution of neutron diffusion and neutron transport problems. The new approach is extended to the nonlinear Navier-Stokes equations of fluid mechanics; its extension to these equations leads to a new computational method, the nodal integral method which is implemented for the numerical solution of these equations. Application to several test problems demonstrates the superior computational efficiency of this new method over previously developed methods. The solutions obtained for several driven cavity problems are compared with the available experimental data and are shown to be in very good agreement with experiment. Additional comparisons also show that the coarse-mesh, nodal integral method results agree very well with the results of definitive ultra-fine-mesh, finite-difference calculations for the driven cavity problem up to fairly high Reynolds numbers
International Nuclear Information System (INIS)
Jurkevicius, A.; Remeikis, V.
2001-01-01
The isotopic composition of hafnium in the radial neutron flux sensor of the RBMK-1500 reactor, the rates of the neutron absorption on Hf isotopes and the neutron spectrum in the sensor were numerically modeled. The sequence SAS2 (Shielding Analysis Sequence) from the package SCALE 4.3 was used for calculations. It has been obtained that the main neutron absorber 167 Er isotope practically burns up completely at the 18 MW d/kgU burnup depth, and at that time the capture rate of thermal neutrons in erbium decreases ten-fold. The average neutron flux density was calculated 7.6*10 13 neutrons. Cm -2 S -1 in the RBMK-1500 reactor grating, when the nuclear fuel enriched with 235 U by 2.4% and with Er by 0.4% is used in a fuel assembly. When the sensor burnup reaches 28 MW d/kgU, the neutron absorption rate of 178 Hf exceeds the rate of 177 Hf. The overall neutron absorption rate in hafnium decreases 2.53 times due to the sensor burnup to 56 MW d/kgU. The corrective factors ξ d (I) at different integral flux I of the sensors were calculated. The obtained dependence ξ d (I) calculated numerically was compared to the experimental one determined by processing repeated calibration results of Hf sensors in RBMK-1500 reactors, as well as compared to the theoretical one currently used in the Ignalina NPP special mathematical algorithms. (author)
A sensitivity study on neutron flux variation due to 10B concentration in dose calculation for BNCT
International Nuclear Information System (INIS)
Jung, Sang Hoon
2006-02-01
The effects of inclusion of 10 B concentration on neutron flux and dose in dose calculation were studied. In order to provide the quantitative effects of inclusion of 10 B concentrations on depressions of neutron and photon flux and dose, the fluxes and doses with voxel head phantoms for various 10 B concentrations homogeneously distributed were calculated by using MCNPX simulations. A lithium target system and beam shaping assembly, which have been developed at the Hanyang University, were used as epithermal neutron beam. The calculation results show that the neutron flux at the center of the head phantom decreases by approximately 5.4% per 10 ppm of 10 B concentration in comparison with the neutron flux in the case of boron-free. It was also observed that the tissue dose at the center of the head phantom is depressed by approximately 4.7% per 10 ppm of the 10 B concentration and the tumor dose by approximately 5.3% per 10 ppm. According to depth of tumors, it was observed that the depressions of the doses in the tumors are ranged in 3.7 ∼ 9.2%. The dose calculations in the case of boron-free show that it is overestimated in comparison with the dose calculations in the cases of the inclusion of 10 B concentrations for the normal tissue and the tumors. Therefore, in dose calculation for BNCT, the depressions of neutron flux and dose should be considered. The results in this study are available to setting up the depression ratios which can be used for converting neutron and gamma fluxes and doses in phantom with boron free into the fluxes and doses in phantom with inclusion of 10 B concentrations in treatment. It is expected that the depression ratios is practicable to dose evaluation for BNCT
Energy Technology Data Exchange (ETDEWEB)
Patil, B.J., E-mail: bjp@physics.unipune.ernet.i [Department of Physics, University of Pune, Pune 411 007 (India); Chavan, S.T.; Pethe, S.N.; Krishnan, R. [SAMEER, IIT Powai Campus, Mumbai 400 076 (India); Bhoraskar, V.N. [Department of Physics, University of Pune, Pune 411 007 (India); Dhole, S.D., E-mail: sanjay@physics.unipune.ernet.i [Department of Physics, University of Pune, Pune 411 007 (India)
2010-10-15
A 6 MeV Race track Microtron (an electron accelerator) based pulsed neutron source has been designed specifically for the elemental analysis of short lived activation products where the low neutron flux requirement is desirable. The bremsstrahlung radiation emitted by impinging 6 MeV electron on the e-{gamma} primary target, was made to fall on the {gamma}-n secondary target to produce neutrons. The optimisation of bremsstrahlung and neutron producing target along with their spectra were estimated using FLUKA code. The measurement of neutron flux was carried out by activation of vanadium and the measured fluxes were 1.1878 x 10{sup 5}, 0.9403 x 10{sup 5}, 0.7428 x 10{sup 5}, 0.6274 x 10{sup 5}, 0.5659 x 10{sup 5}, 0.5210 x 10{sup 5} n/cm{sup 2}/s at 0{sup o}, 30{sup o}, 60{sup o}, 90{sup o}, 115{sup o}, 140{sup o} respectively. The results indicate that the neutron flux was found to be decreased as increase in the angle and in good agreement with the FLUKA simulation.
Radiation problems expected for the German spallation neutron source
International Nuclear Information System (INIS)
Goebel, K.
1981-01-01
The German project for the construction of a Spallation Neutron Source with high proton beam power (5.5 MW) will have to cope with a number of radiation problems. The present report describes these problems and proposes solutions for keeping exposures for the staff and release of activity and radiation into the environment as low as reasonably achievable. It is shown that the strict requirements of the German radiation protection regulations can be met. The main problem will be the exposure of maintenance personnel to remanent gamma radiation, as is the case at existing proton accelerators. Closed ventilation and cooling systems will reduce the release of (mainly short-lived) activity to acceptable levels. Shielding requirements for different sections are discussed, and it is demonstrated by calculations and extrapolations from experiments that fence-post doses well below 150 mrem/y can be obtained at distances of the order of 100 metres from the principal source points. The radiation protection system proposed for the Spallation Neutron Source is discussed, in particular the needs for monitor systems and a central radiation protection data base and alarm system. (orig.)
On the problem of bound states of pions and neutrons
International Nuclear Information System (INIS)
Gudima, K.K.; Karnaukhov, V.A.
1992-01-01
The problem of existence of the bound states of negative pions and neutrons has been widely discussed for the last years. It is considered possibilities of the experimental observation of pion-neutron clusters, if they do exist, in nucleus-nucleus collisions. The yields of exotic fragments π -Z n A in the interactions of 12 C and 56 Fe with 208 Pb at the energies from 0.3 to 3.7 GeV per nucleon are calculated. For 40 Ar+ 238 U and 139 La+ 238 U collisions the calculations were performed at the energied of 1.8 GeV and 1.3 GeV per nucleon, respectively. These calculations were performed in the framework of the coalescence mechanism with the differential cross sections for pion and neutron production generated by firestreak model. The differential cross sections for production of π -1 n -2 , π -2 N 2 , π - n 4 , π -4 n 6 , and π -12 n 6 were calculated. It is shown that the use of very heavy projectiles like 56 Fe and 139 La has a great advantage in the experimental search for the exotic clusters. 20 refs.; 8 figs
International Nuclear Information System (INIS)
Tamaki, S.; Sakai, M.; Yoshihashi, S.; Manabe, M.; Zushi, N.; Murata, I.; Hoashi, E.; Kato, I.; Kuri, S.; Oshiro, S.; Nagasaki, M.; Horiike, H.
2015-01-01
Mock-up experiment for development of accelerator based neutron source for Osaka University BNCT project was carried out at Birmingham University, UK. In this paper, spatial distribution of neutron flux intensity was evaluated by foil activation method. Validity of the design code system was confirmed by comparing measured gold foil activities with calculations. As a result, it was found that the epi-thermal neutron beam was well collimated by our neutron moderator assembly. Also, the design accuracy was evaluated to have less than 20% error. - Highlights: • Accelerator based neutron source for BNCT is being developed in Osaka University. • Mock-up experiment was carried out at Birmingham University, UK. • Neutronics performance of our assembly was evaluated from gold foil activation. • Gold foil activation was determined by using HPGe detectors. • Validity of the neutronics design code system was confirmed.
International Nuclear Information System (INIS)
Khelifi, R.; Idiri, Z.; Bode, P.
2002-01-01
The CITATION code based on neutron diffusion theory was used for flux calculations inside voluminous samples in prompt gamma activation analysis with an isotopic neutron source (Am-Be). The code uses specific parameters related to the energy spectrum source and irradiation system materials (shielding, reflector). The flux distribution (thermal and fast) was calculated in the three-dimensional geometry for the system: air, polyethylene and water cuboidal sample (50x50x50 cm). Thermal flux was calculated in a series of points inside the sample. The results agreed reasonably well with observed values. The maximum thermal flux was observed at a distance of 3.2 cm while CITATION gave 3.7 cm. Beyond a depth of 7.2 cm, the thermal flux to fast flux ratio increases up to twice and allows us to optimise the detection system position in the scope of in-situ PGAA
Parallel preconditioned conjugate gradient algorithm applied to neutron diffusion problem
International Nuclear Information System (INIS)
Majumdar, A.; Martin, W.R.
1992-01-01
Numerical solution of the neutron diffusion problem requires solving a linear system of equations such as Ax = b, where A is an n x n symmetric positive definite (SPD) matrix; x and b are vectors with n components. The preconditioned conjugate gradient (PCG) algorithm is an efficient iterative method for solving such a linear system of equations. In this paper, the authors describe the implementation of a parallel PCG algorithm on a shared memory machine (BBN TC2000) and on a distributed workstation (IBM RS6000) environment created by the parallel virtual machine parallelization software
Energy Technology Data Exchange (ETDEWEB)
Yavar, A.R. [School of Applied Physics, Faculty of Science and Technology, University Kebangsaan Malaysia (UKM), Bangi, Selangor 43600 (Malaysia); Sarmani, S. [School of Chemical Sciences and Food Technology, Faculty of Science and Technology, University Kebangsaan Malaysia (UKM), Bangi, Selangor 43600 (Malaysia); Wood, A.K. [Analytical Chemistry Application Group, Industrial Technology Division, Malaysian Nuclear Agency (MNA), Bangi, Kajang, Selangor 43000 (Malaysia); Fadzil, S.M. [School of Applied Physics, Faculty of Science and Technology, University Kebangsaan Malaysia (UKM), Bangi, Selangor 43600 (Malaysia); Masood, Z. [Analytical Chemistry Application Group, Industrial Technology Division, Malaysian Nuclear Agency (MNA), Bangi, Kajang, Selangor 43000 (Malaysia); Khoo, K.S., E-mail: khoo@ukm.m [School of Applied Physics, Faculty of Science and Technology, University Kebangsaan Malaysia (UKM), Bangi, Selangor 43600 (Malaysia)
2011-02-15
The Malaysian Nuclear Agency (MNA) research reactor, commissioned in 1982, is a TRIGA Mark II swimming pool type reactor. When the core configuration changed in June 2009, it became essential to re-determine such neutron flux parameters as thermal to epithermal neutron flux ratio (f), epithermal neutron flux shape factor ({alpha}), thermal neutron flux ({phi}{sub th}) and epithermal neutron flux ({phi}{sub epi}) in the irradiation positions of MNA research reactor in order to guarantee accuracy in the application of k{sub 0}-neutron activation analysis (k{sub 0}-NAA).The f and {alpha} were determined using the bare bi-isotopic monitor and bare triple monitor methods, respectively; Au and Zr monitors were utilized in present study. The results for four irradiation positions are presented and discussed in the present work. The calculated values of f and {alpha} ranged from 33.49 to 47.33 and -0.07 to -0.14, respectively. The {phi}{sub th} and the {phi}{sub epi} were measured as 2.03 x 10{sup 12} (cm{sup -2} s{sup -1}) and 6.05 x 10{sup 10} (cm{sup -2} s{sup -1}) respectively. These results were compared to those of previous studies at this reactor as well as to those of reactors in other countries. The results indicate a good conformity with other findings.
International Nuclear Information System (INIS)
Tomiyoshi, Irene Akemy
1982-01-01
In this work, the possibility of the indirect determination of the integrated neutron flux, through the change of isotopic ratios of cadmium and gadolinium was investigated. The samples of cadmium we/e gadolinium were irradiated in the IEA-Rl reactor. These elements were chosen because they have high thermal neutron absorption cross section which permit the change in the isotopic composition during a short irradiation time to be measured accurately. The isotopic ratios were measured with a thermionic mass spectrometer the silica-gel technique and arrangement with single filament were used for the cadmium analysis, where as the oxi - reduction technique and arrangement with double filaments were used for gadolinium analysis. The mass fractionation effects for cadmium and gadolinium were corrected respectively by the exponential and potential expansion of the isotopic fractionation factor per atomic mass unit. The flux values supplied by the Centro de Operacao e Utilizacao do Reator de Pesquisas do IPEN were extrapolated. These values and the integrated flux values obtained experimentally were compared. (author)
International Nuclear Information System (INIS)
Rehm, C.
2001-01-01
Next-generation spallation neutron source facilities will offer instruments with unprecedented capabilities through simultaneous enhancement of source power and usage of advanced optical components. The Spallation Neutron Source (SNS), already under construction at Oak Ridge National Laboratory and scheduled to be completed by 2006, will provide greater than an order of magnitude more effective source flux than current state-of-the-art facilities, including the most advanced research reactors. An additional order of magnitude gain is expected through the use of new optical devices and instrumentation concepts. Many instrument designs require supermirror (SM) neutron guides with very high critical angles for total reflection. In this contribution, they discuss how the performance of modern neutron scattering instruments depends on the efficiency of these supermirrors. They outline ideas for enhancing the performance of the SM coatings, particularly for improving the reflectivity at the position of the critical wave vector transfer. A simulation program has been developed which allows different approaches for SM designs to be studied. Possible instrument performance gains are calculated for the example of the SNS reflectometer
Marginal thermal-neutron peak fluxes in systems with modulation of reactivity
International Nuclear Information System (INIS)
Alekseev, N.I.; Stolypin, V.S.
1978-01-01
A possibility of obtaining high (including marginal) thermal neutron peak fluxes PHIsub(m) in a light water trap of a pulsed fast reactor with modulation of reactivity has been studied. The dependences of sub(m) on the subcriticality and supercriticality as well as on the supercritical state duration have been calculated on stepped variations of the reactivity. The calculations show that PHIsub(m) of about 7.3x10 18 neutron/cm 2 xs with the effective pulse duration of approximately 150 μc, pulse frequency of approximately 1 Hz and at fuel temperature of approximately 1300 deg C can be obtained with the reactor. The comparative calculations show that sub(m) is 1.5 times higher than that of a booster obtained using a ''meson plant'' (designs of the booster and the reactor are equivalent). The neutron background between pulses in the reactor is much lower than in the booster, and there is no need for a power injector in the reactor altogether. Meanwhile the maximum attainable PHIsub(m) for the booster and the reactor are the same and equal approximately 2x10 19 neutron/cm 2 xs
High count problems in elemental analysis using pulsed neutron inelastic scattering
Energy Technology Data Exchange (ETDEWEB)
Vartsky, D; Wielopolski, L; Ellis, K J; Cohn, S H [Brookhaven National Lab., Upton, NY (USA). Medical Dept.
1983-03-01
Elemental analysis by neutron inelastic scattering using a miniature intense pulsed neutron source ('Zetatron') was evaluated. The particular problems associated with detector pulse-pile-up during the neutron burst and the limited ability of the analyzer to process on average more than one detector pulse per neutron burst were examined. The severity of these problems is described and a solution using a multiple ADC system is proposed.
American National Standard: neutron and gamma-ray flux-to-dose rate factors
International Nuclear Information System (INIS)
Anon.
1977-01-01
This Standard presents data recommended for computing biological dose rates due to neutron and gamma-ray radiation fields. Neutron flux-to-dose-rate conversion factors for energies from 2.5 x 10 -8 to 20 MeV are given; the energy range for the gamma-ray conversion factors is 0.01 to 15 MeV. Specifically, this Standard is intended for use by shield designers to calculate wholebody dose rates to radiation workers and the general public. Establishing dose-rate limits is outside the scope of this Standard. Use of this Standard in cases where the dose equivalents are far in excess of occupational exposure guidelines is not recommended
International Nuclear Information System (INIS)
Brixy, H.; Hecker, R.; Serpekian, T.; Benninghofen, G.; Serafin, N.; Spillekothen, H.G.
1982-06-01
The development is described of the large-area SPN-detector as an out of core power monitoring system. Gadolinium or cobalt was used as the emitter. Response functions of the gadolinium SPN-detector were found with regard to the reactor power, the effect of the gamma field, its short-term behaviour following reactor shutdown and long-term behaviour during reactor operation. It was shown that a detector of 0.1 mm emitter thickness can withstand an integral thermal neutron flux of 2.10 20 nvt almost without efficiency loss thus indicating that the large-area gadolinium SPN-detector is a suitable means for power monitoring in large HTGR's
International Nuclear Information System (INIS)
Merz, Stefan; Djuricic, Mile; Villa, Mario; Boeck, Helmuth; Steinhauser, Georg
2011-01-01
The activation of the biological shield is an important process for waste management considerations of nuclear facilities. The final activity can be estimated by modeling using the neutron flux density rather than the radiometric approach of activity measurements. Measurement series at the TRIGA reactor Vienna reveal that the flux density next to the biological shield is in the order of 10 9 cm -2 s -1 at maximum power; but it is strongly influenced by reactor installations. The data allow the estimation of the final waste categorization of the concrete according to the Austrian legislation. - Highlights: → Neutron activation is an important process for the waste management of nuclear facilities. → Biological shield of the TRIGA reactor Vienna has been topic of investigation. → Flux values allow a categorization of the concrete concerning radiation protection legislation. → Reactor installations are of great importance as neutron sources into the biological shield. → Every installation shows distinguishable flux profiles.
VSOP, Neutron Spectra, 2-D Flux Synthesis, Fuel Management, Thermohydraulics Calculation
International Nuclear Information System (INIS)
Teuchert, E.; Haas, K.A.
1995-01-01
1 - Description of problem or function: VSOP (Very Superior Old Programs) is a system of codes linked together for the simulation of reactor life histories. It comprises neutron cross section libraries and processing routines, repeated neutron spectrum evaluation, 2-D and 3-D diffusion calculation, depletion and shut-down features, in- core and out-of-pile fuel management, fuel cycle cost analysis, and thermal hydraulics (steady state and transient). Various techniques have been employed to accelerate the iterative processes and to optimize the internal data transfer. The code system has been used extensively for comparison studies of thermal reactors, their fuel cycles, thermal transients, and safety assessment. Besides its use in research and development work for the Gas Cooled High Temperature Reactor, the system has been applied successfully to Light Water and Heavy Water Reactors, MAGNOX, and RBMK. 2 - Method of solution: The nuclear data for 184 isotopes are contained in two libraries. Fast and epithermal data in a 68 group GAM-I structure have been prepared mainly from ENDF/B-V and JEF-1. Resonance cross section data are given as input. Thermal data in a 30 group THERMOS structure have been collapsed from a 96 group THERMALIZATION (GATHER) library by a relevant neutron energy spectrum generated by the THERMALIZATION code. Graphite scattering matrices are based on the Young phonon spectrum in graphite. The neutron spectrum is calculated by a combination of the GAM and THERMOS codes. They can simultaneously be employed for many core regions differing in temperature, burnup, and fuel element lay-out. The thermal cell code THERMOS has been extended to treat the grain structure of the coated particles inside the fuel elements, and the epithermal GAM code uses modified cross sections for the resonance absorbers prepared from double heterogeneous ZUT-DGL calculations. The diffusion module of the code is CITATION with 2 - 8 energy groups. It provides the neutron
International Nuclear Information System (INIS)
Luiz, Leandro C.; Crispim, Verginia R.; Ferreira, Francisco J. O.
2017-01-01
The presence of crust on the inner walls of metallic ducts impairs transportation because crust completely or partially hinders the passage of fluid to the processing unit and causes damage to equipment connected to the production line. Its localization is crucial. With the development of the electronic imaging system installed at the Argonauta/Nuclear Engineering Institute (IEN)/National Nuclear Energy Commission (CNEN) reactor, it became possible to visualize crust in the interior of metallic piping of small diameter using real-time neutron radiography images obtained with a low neutron flux. The obtained images showed the resistance offered by crust on the passage of water inside the pipe. No discrepancy of the flow profile at the bottom of the pipe, before the crust region, was registered. However, after the passage of liquid through the pipe, images of the disturbances of the flow were clear and discrepancies in the flow profile were steep. This shows that this technique added the assembled apparatus was efficient for the visualization of the crust and of the two-phase flows
Energy Technology Data Exchange (ETDEWEB)
Luiz, Leandro C.; Crispim, Verginia R. [Nuclear Engineering Program, Federal University of Rio de Janeiro, Rio de Janeiro (Brazil); Ferreira, Francisco J. O. [National Nuclear Energy Commission, CNEN/IEN, Division Reactors, Rio de Janeiro (Brazil)
2017-06-15
The presence of crust on the inner walls of metallic ducts impairs transportation because crust completely or partially hinders the passage of fluid to the processing unit and causes damage to equipment connected to the production line. Its localization is crucial. With the development of the electronic imaging system installed at the Argonauta/Nuclear Engineering Institute (IEN)/National Nuclear Energy Commission (CNEN) reactor, it became possible to visualize crust in the interior of metallic piping of small diameter using real-time neutron radiography images obtained with a low neutron flux. The obtained images showed the resistance offered by crust on the passage of water inside the pipe. No discrepancy of the flow profile at the bottom of the pipe, before the crust region, was registered. However, after the passage of liquid through the pipe, images of the disturbances of the flow were clear and discrepancies in the flow profile were steep. This shows that this technique added the assembled apparatus was efficient for the visualization of the crust and of the two-phase flows.
International Nuclear Information System (INIS)
Boulyga, E.G.; Boulyga, S.F.
2000-01-01
A computer program is described, which calculates the induced activities of isotopes after irradiation in a known neutron field, thermal and epithermal neutron fluxes from the measured induced activities and from nuclear data of 2-4 monitor nuclides as well as the element concentrations in samples irradiated together with the monitors. The program was developed for operation in Windows 3.1 (or higher). The application of the program for neutron activation analysis allows to simplify the experimental procedure and to reduce the time. The program was tested by measuring different types of standard reference materials at the FRJ-2 (Research Centre, Juelich, Germany) and Triga (University Mainz, Germany) reactors. Comparison of neutron flux parameters calculated by this program with those calculated by a VAX program developed at the Research Centre, Juelich was done. The results of testing seem to be satisfactory. (author)
Energy Technology Data Exchange (ETDEWEB)
Obradovic, D; Bosevski, T [Institut za nuklearne nauke Boris Kidric, Vinca, Beograd (Yugoslavia)
1964-07-01
Application of semiconductor detectors for measuring neutron flux distribution is about 10 times faster than measurements by activation foils and demands significantly lower reactor power. Following corrections are avoided: mass of activation foils which influences the self shielding, nuclear decay during activity measurements; counter dead-time. It is possible to control the measured data during experiment and repeat measurements if needed. Precision of the measurement is higher since it is possible to choose the wanted statistics. The method described in this paper is applied for measurements at the RB reactor. It is concluded that the method is suitable for fast measurements but the activation analysis is still indispensable.
Energy Technology Data Exchange (ETDEWEB)
Huseynov, E.; Garibli, A., E-mail: elchin.huse@yahoo.com [National Nuclear Research Center, Department of Nanotechnology and Radiation Material Science, 1073, Inshaatchilar pr. 4, Baku (Azerbaijan)
2016-11-01
It has been reviewed the frequency dependencies of electrical conductivity of nanoparticles affected by neutron flux at different times and initial state, at various constant temperatures such as 100, 200, 300 and 400 K. Measurements have been carried out at each temperature at the different 97 values of frequency in the 1 Hz - 1 MHz range. From interdependence between real and imaginary parts of electrical conductivity it has been determined the type of conductivity. Moreover, in the work it is given the mechanism of electrical conductivity according to the obtained results. (Author)
Detailed studies of Minor Actinide transmutation-incineration in high-intensity neutron fluxes
International Nuclear Information System (INIS)
Bringer, O.; Al Mahamid, I.; Blandin, C.; Chabod, S.; Chartier, F.; Dupont, E.; Fioni, G.; Isnard, H.; Letourneau, A.; Marie, F.; Mutti, P.; Oriol, L.; Panebianco, S.; Veyssiere, C.
2006-01-01
The Mini-INCA project is dedicated to the measurement of incineration-transmutation chains and potentials of minor actinides in high-intensity thermal neutron fluxes. In this context, new types of detectors and methods of analysis have been developed. The 241 Am and 232 Th transmutation-incineration chains have been studied and several capture and fission cross sections measured very precisely, showing some discrepancies with existing data or evaluated data. An impact study was made on different based-like GEN-IV reactors. It underlines the necessity to proceed to precise measurements for a large number of minor-actinides that contribute to these future incineration scenarios. (authors)
Measurements of neutron fluxes and cadmium ratio at equilibrium core in JRR-3M
International Nuclear Information System (INIS)
Ohtomo, Akitoshi; Sasajima, Fumio; Ishida, Takuya; Shigemoto, Masamitsu; Takahashi, Hidetake; Maejima, Takeshi; Sekine, Katsunori.
1993-08-01
Construction and characteristics tests of JRR-3M (Modified JRR-3) had been completed on October 1990, and the reactor reached to equilibrium core in July 1991. Measurements of neutron flux and cadmium ratio in Hydraulic irradiation facility (HR) and Pneumatic irradiation facility (PN) at 20 MW reactor power were carried out for the equilibrium core from May to August 1991 and for the latest core in April 1993. The results at the equilibrium core and the latest core are described in this paper. (author)
Neutron radiography imaging with 2-dimensional photon counting method and its problems
International Nuclear Information System (INIS)
Ikeda, Y.; Kobayashi, H.; Niwa, T.; Kataoka, T.
1988-01-01
A ultra sensitive neutron imaging system has been deviced with a 2-dimensional photon counting camara (ARGUS 100). The imaging system is composed by a 2-dimensional single photon counting tube and a low background vidicon followed with an image processing unit and frame memories. By using the imaging system, electronic neutron radiography (NTV) has been possible under the neutron flux less than 3 x 10 4 n/cm 2 ·s. (author)
International Nuclear Information System (INIS)
Asaoka, Takumi; Watanabe, N.
1976-01-01
In achieving a high peak thermal neutron flux in hydrogenous moderators installed in repetitive pulse reactors, the core-moderator arrangement can play as much an important role as the moderator design itself. However, the effect of the former has not been adequately emphasized to date, while a rather extensive study has been made on the latter. The present study concerns with a core-moderator system parameter optimization for a repetitive accelerator pulsed fast reactor. The results have shown that small differences in the arrangement resulting from the optimizations of various parameters are significant and the effects can be summed up to give an increase in the peak thermal flux by a factor of about two. (auth.)
Numerical studies of the flux-to-current ratio method in the KIPT neutron source facility
International Nuclear Information System (INIS)
Cao, Y.; Gohar, Y.; Zhong, Z.
2013-01-01
The reactivity of a subcritical assembly has to be monitored continuously in order to assure its safe operation. In this paper, the flux-to-current ratio method has been studied as an approach to provide the on-line reactivity measurement of the subcritical system. Monte Carlo numerical simulations have been performed using the KIPT neutron source facility model. It is found that the reactivity obtained from the flux-to-current ratio method is sensitive to the detector position in the subcritical assembly. However, if multiple detectors are located about 12 cm above the graphite reflector and 54 cm radially, the technique is shown to be very accurate in determining the k eff this facility in the range of 0.75 to 0.975. (authors)
International Nuclear Information System (INIS)
Ishikawa, M.; Itoga, T.; Okuji, T.; Nakhostin, M.; Shinohara, K.; Hayashi, T.; Sukegawa, A.; Baba, M.; Nishitani, T.
2006-01-01
A line-integrated neutron emission profile is routinely measured using the radial neutron collimator system in JT-60U tokamak. Stilbene neuron detectors (SNDs), which combine a stilbene organic crystal scintillation detector (SD) with an analog neutron-gamma pulse shape discrimination (PSD) circuit, have been used to measure collimated neutron flux. Although the SND has many advantages as a neutron detector, the maximum count rate is limited up to ∼1x10 5 counts/s due to the analog PSD circuit. To overcome this issue, a digital signal processing system (DSPS) using a flash analog-to-digital converter (Acqiris DC252, 8 GHz, 10 bits) has been developed at Cyclotron and Radioisotope Center in Tohoku University. In this system anode signals from photomultiplier of the SD are directory stored and digitized. Then, the PSD between neutrons and gamma rays is performed using software. The DSPS has been installed in the vertical neutron collimator system in JT-60U and applied to deuterium experiments. It is confirmed that the PSD is sufficiently performed and collimated neutron flux is successfully measured with count rate up to ∼5x10 5 counts/s without the effect of pileup of detected pulses. The performance of the DSPS as a neutron detector, which supersedes the SND, is demonstrated
Increasing the neutron flux study for the TRR-II core design
International Nuclear Information System (INIS)
Chen, C.-H.; Yang, J.-T.; Chou, Y.-C.
1999-01-01
The maximum unperturbed thermal flux of the originally proposed core design, which is a 6x6 square arrangement with power level of 20 MW and has been presented at the 6th Meeting of IGORR, for the TRR-II reactor is about 2.0x10 14 n/cm 2 -sec. However, it is no longer satisfied the user's requirement, that is, it must reach at least 2.5x10 14 n/cm 2 -sec. In order to enhance the thermal neutron flux, one of the most effective ways is to increase the average power density. Therefore, two new designs with more compact cores are then proposed and studied. One is 5x6 rectangular arrangement with power of 20 MW; the other one is 5x5 square arrangement with power of 16 MW. It is for sure that both core designs can satisfy thermal hydraulic safety limits. The designed parameters related to neutronics are listed and compared fundamentally. According to our calculation, although both cores have similar average power density, the results show that the 5x6/20 MW design has the maximum unperturbed thermal flux in the D 2 O region about 2.7x10 14 n/cm 2 -sec, and the 5x5/16 MW design has 2.5x10 14 n/cm 2 -sec. The maximum thermal flux in the neighborhood of the longer side of the 5x6 core is about 7% higher than the one in the neighborhood of any side of the 5x5 core. This 'long-side effect' gives the 5x6/20 MW core design an advantage of the utilization of the thermal neutron flux in the D 2 O region. In addition, the 5x5 core is also more sensitive to the reactivity change on account of in-core irradiation test facilities. Therefore, under overall considerations the 5x6/20 MW core design is chosen for further detailed design. (author)
International Nuclear Information System (INIS)
Buckel, G.
1983-01-01
The objectives are the development, testing and cultivation of reliable, efficient and user-optimized neutron-physical calculation methods and conformity with users' requirements concerning design of power reactors, planning and analysis of experiments necessary for their protection as well as research on physical key problems. A short outline of available computing programmes for the following objectives is given: - Provision of macroscopic group constants, - Calculation of neutron flux distribution in transport theory and diffusion approximation, - Evaluation of neutron flux-distribution, - Execution of disturbance calculations for the determination reactivity coefficients, and - graphical representation of results. (orig./RW) [de
Energy Technology Data Exchange (ETDEWEB)
Kocic, A [Institute of nuclear sciences Boris Kidric, Vinca, Beograd (Yugoslavia)
1974-07-01
Anisotropy of neutron elastic scattering is a problem of special importance in solving the Boltzmann transport equation numerically. This is not the case when Monte Carlo method is applied. Estimation of the influence of elastic scattering anisotropy on the neutron flux is treated in order to justify the application of Monte Carlo method which is computer time consuming. Correlation procedure was applied for the study of this influence. One group case was used as an example to enable comparison of other methods.
Neutron flux calculation and fluence in the encircling of the core and vessel of a reactor BWR
International Nuclear Information System (INIS)
Martinez C, E.
2011-01-01
One of the main objectives related to the safety of any nuclear power plant, including the nuclear power plant of Laguna Verde is to ensure the structural integrity of reactor pressure vessel. To identify and quantify the damage caused by neutron irradiation in the vessel of any nuclear reactor, it is necessary to know both the neutron flux and the neutron fluence that the vessel has been receiving during its operation lifetime, and that the damage observed by mechanical testing are products of microstructural effects induced by neutron irradiation; therefore, it is important the study and prediction of the neutron flux in order to have a better understanding of the damage that these materials are receiving. The calculation here described uses the DORT code, which solves the neutron transport equation in discrete ordinates in two dimensions (x-y, r-θ and r-z), according to a regulatory guide, it should make an approximation of the neutron flux in three dimensions by the so called synthesis method. It is called in that way because it achieves a representation of 3 Dimensional neutron flux combining or summarizing the fluxes calculated by DORT r-θ, r-z and r. This work presents the application of synthesis method, according to Regulatory Guide 1190, to determine the 3 Dimensional fluxes in internal BWR reactor using three different spatial meshes. The results of the neutron flux and fluence, using three different meshes in the directions r, θ and z were compared with results reported in the literature obtaining a difference not larger than 9.61%, neutron flux reached its maximum, 1.58 E + 12 n/cm 2 s, at a height H 4 (239.07 cm) and angle 32.236 o in the core shroud and 4.00 E + 09 n/cm 2 s at a height H 4 and angle 35.27 o in the inner wall of the reactor vessel, positions that are consistent to within ±10% over the ones reported in the literature. (Author)
Energy Technology Data Exchange (ETDEWEB)
Carle, R.; Mazancourt, T. de [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1957-07-01
We sought to determine the behavior of the thermal, resonant and rapid neutron fluxes in the multiplier-reflector transition region, in the two principal directions of the system. We have also measured the variation of these different fluxes in the body of the multiplier medium in a canal filled with graphite and in an empty canal. The results are given in the form of curves representing: - the variation of the ratio of the thermal flux to the rapid flux in axial and radial transitions - the behavior of the thermal and resonant fluxes and the variation of their ratio in the same regions. (author) [French] Nous avons cherche a determiner le comportement des differents flux, thermique, resonnant et rapide a la transition milieu multiplicateur-reflecteur dans les deux directions principales du reseau. Nous avons egalement mesure la variation de ces differents flux au sein du milieu multiplicateur dans un canal rempli de graphite et dans un canal vide. Les resultats sont donnes sous forme de courbe representant: - La variation du rapport du flux thermique au flux rapide aux transitions axiale et radiale - L'allure des flux thermique et resonnant et la variation de leur rapport dans les memes regions. (auteur)
Energy Technology Data Exchange (ETDEWEB)
Carle, R; Mazancourt, T de [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1957-07-01
We sought to determine the behavior of the thermal, resonant and rapid neutron fluxes in the multiplier-reflector transition region, in the two principal directions of the system. We have also measured the variation of these different fluxes in the body of the multiplier medium in a canal filled with graphite and in an empty canal. The results are given in the form of curves representing: - the variation of the ratio of the thermal flux to the rapid flux in axial and radial transitions - the behavior of the thermal and resonant fluxes and the variation of their ratio in the same regions. (author) [French] Nous avons cherche a determiner le comportement des differents flux, thermique, resonnant et rapide a la transition milieu multiplicateur-reflecteur dans les deux directions principales du reseau. Nous avons egalement mesure la variation de ces differents flux au sein du milieu multiplicateur dans un canal rempli de graphite et dans un canal vide. Les resultats sont donnes sous forme de courbe representant: - La variation du rapport du flux thermique au flux rapide aux transitions axiale et radiale - L'allure des flux thermique et resonnant et la variation de leur rapport dans les memes regions. (auteur)
International Nuclear Information System (INIS)
Seidel, K.; Freiesleben, H.; Poenitz, E.; Klix, A.; Unholzer, S.; Batistoni, P.; Fischer, U.; Leichtle, D.
2006-01-01
The nuclear parameters of a breeding blanket, such as tritium production rate, nuclear heating, activation and dose rate, are calculated by integral folding of an energy dependent cross section (or coefficient) with the neutron (or gamma-ray) flux energy spectra. The uncertainties of the designed parameters are determined by the uncertainties of both the cross section data and the flux spectra obtained by transport calculations. Also the analysis of possible discrepancies between measured and calculated integral nuclear parameter represents a two-step procedure. First, the energy region and the amount of flux discrepancies has to be found out and second, the cross section data have to be checked. To this end, neutron and gamma-ray flux spectra in a mock-up of the EU Helium-Cooled Pebble Bed (HCPB) breeder Test Blanket Module (TBM), irradiated with 14 MeV neutrons, were measured and analysed by means of Monte Carlo transport calculations. The flux spectra were determined for the energy ranges that are relevant for the most important nuclear parameters of the TBM, which are the tritium production rate and the shielding capability. The fast neutron flux which determines the tritium production on 7 Li and dominates the shield design was measured by the pulse-height distribution obtained from an organic liquid scintillation detector. Simultaneously, the gamma-ray flux spectra were measured. The neutron flux at lower energies, down to thermal, which determines the tritium production on 6 Li, was measured with time-of-arrival spectroscopy. For this purpose, the TUD neutron generator was operated in pulsed mode (pulse width 10 μs, frequency 1 kHz) and the neutrons arriving at a 3 He proportional counter in the mock-up were recorded as a function of time after the source neutron pulse. The spectral distributions for the two positions in the mock-up, where measurements were carried out, were calculated with the Monte Carlo code MCNP, version 5, and nuclear data from the
International Nuclear Information System (INIS)
Widarto
2002-01-01
Determination and analysis of neutron flux measurements on radial piercing beam port have been done as completion experimental data document and progressing on utilization of the Kartini research reactor purposes. The analysis and determination of the neutron flux have been carried out by using Au foils detector neutron activation analysis method which put on the radius of cross section (19 cm) and a long of radial piercing beam port (310 cm) Based on the calculation, distribution of the thermal neutron flux is around (8.3 ± 0.9) x 10 5 ncm -2 s -1 to (6.8 ± 0.5) x 10 7 ncm -2 s -1 and fast neutron is (5.0 ± 0.2) x 10 5 ncm -2 s -1 to (1.43 ± 0.6) x 10 7 ncm -2 s -1 . Analyzing by means of curve fitting method could be concluded that the neutron flux distribution on radial piercing beam port has profiled as a polynomial curve. (author)
Analysis of JKT01 Neutron Flux Detector Measurements In RSG-GAS Reactor Using LabVIEW
Rokhmadi; Nur Rachman, Agus; Sujarwono; Taryo, Taswanda; Sunaryo, Geni Rina
2018-02-01
The RSG-GAS Reactor, one of the Indonesia research reactors and located in Serpong, is owned by the National Nuclear Energy Agency (BATAN). The RSG-GAS reactor has operated since 1987 and some instrumentation and control systems are considered to be degraded and ageing. It is therefore, necessary to evaluate the safety of all instrumentation and controls and one of the component systems to be evaluated is the performance of JKT01 neutron flux detector. Neutron Flux Detector JKT01 basically detects neutron fluxes in the reactor core and converts it into electrical signals. The electrical signal is then forwarded to the amplifier (Amplifier) to become the input of the reactor protection system. One output of it is transferred to the Main Control Room (RKU) showing on the analog meter as an indicator used by the reactor operator. To simulate all of this matter, a program to simulate the output of the JKT01 Neutron Flux Detector using LabVIEW was developed. The simulated data is estimated using a lot of equations also formulated in LabVIEW. The calculation results are also displayed on the interface using LabVIEW available in the PC. By using this simulation program, it is successful to perform anomaly detection experiments on the JKT01 detector of RSG-GAS Reactor. The simulation results showed that the anomaly JKT01 neutron flux using electrical-current-base are respectively, 1.5×,1.7× and 2.0×.
International Nuclear Information System (INIS)
Yamamoto, Kazuyoshi; Kumada, Hiroaki; Kishi, Toshiaki; Torii, Yoshiya; Horiguchi, Yoji
2001-03-01
To evaluate nitrogen dose, boron dose and gamma-ray dose occurred by neutron capture reaction of the hydrogen at the medical irradiation, two-dimensional distribution of the thermal neutron flux is very important because these doses are proportional to the thermal neutron distribution. This report describes the measurement of the two-dimensional thermal neutron distribution in a head water phantom by neutron beams of the JRR-4 and evaluation of the dose distribution characteristic. Thermal neutron flux in the phantom was measured by gold wire placed in the spokewise of every 30 degrees in order to avoid the interaction. Distribution of the thermal neutron flux was also calculated using two-dimensional Lagrange's interpolation program (radius, angle direction) developed this time. As a result of the analysis, it was confirmed to become distorted distribution which has annular peak at outside of the void, though improved dose profile of the deep direction was confirmed in the case which the radiation field in the phantom contains void. (author)
International Nuclear Information System (INIS)
Shafii, M. Ali; Su'ud, Zaki; Waris, Abdul; Kurniasih, Neny; Ariani, Menik; Yulianti, Yanti
2010-01-01
Nuclear reactor design and analysis of next-generation reactors require a comprehensive computing which is better to be executed in a high performance computing. Flat flux (FF) approach is a common approach in solving an integral transport equation with collision probability (CP) method. In fact, the neutron flux distribution is not flat, even though the neutron cross section is assumed to be equal in all regions and the neutron source is uniform throughout the nuclear fuel cell. In non-flat flux (NFF) approach, the distribution of neutrons in each region will be different depending on the desired interpolation model selection. In this study, the linear interpolation using Finite Element Method (FEM) has been carried out to be treated the neutron distribution. The CP method is compatible to solve the neutron transport equation for cylindrical geometry, because the angle integration can be done analytically. Distribution of neutrons in each region of can be explained by the NFF approach with FEM and the calculation results are in a good agreement with the result from the SRAC code. In this study, the effects of the mesh on the k eff and other parameters are investigated.
Energy Technology Data Exchange (ETDEWEB)
Vermeeren, L. [SCK/CEN, B-2400 Mol (Belgium)
2001-07-01
Self-powered neutron detectors (SPND) are well suited to monitor continuously the neutronic operating conditions of driver fuel of research reactors and to follow its burnup evolution. This is of particular importance when advanced or new MTR fuel designs need to be qualified. We have developed a detailed MCNP-4B based Monte Carlo approach for the calculation of neutron sensitivities of SPNDs. Results for the neutron sensitivity of a Rh SPND are in excellent agreement with experimental data recently obtained at the BR2 research reactor. A critical comparison of the Monte Carlo results with results from standard analytical methods reveals an important deficiency of the analytical methods in the description of the electron transport efficiency. Our calculation method allows a reliable on-line determination of the absolute in-pile neutron flux. (author)
International Nuclear Information System (INIS)
Vermeeren, L.
2001-01-01
Self-powered neutron detectors (SPND) are well suited to monitor continuously the neutronic operating conditions of driver fuel of research reactors and to follow its burnup evolution. This is of particular importance when advanced or new MTR fuel designs need to be qualified. We have developed a detailed MCNP-4B based Monte Carlo approach for the calculation of neutron sensitivities of SPNDs. Results for the neutron sensitivity of a Rh SPND are in excellent agreement with experimental data recently obtained at the BR2 research reactor. A critical comparison of the Monte Carlo results with results from standard analytical methods reveals an important deficiency of the analytical methods in the description of the electron transport efficiency. Our calculation method allows a reliable on-line determination of the absolute in-pile neutron flux. (author)
Measurement and calculation of spatial and energetic neutron flux in the IEA-R1 reactor core
International Nuclear Information System (INIS)
Bittelli, U.D.
1988-01-01
This work presents spatial and energetic flux distribution measured in the IEA-R1 reactor core. The thermal neutron flux was measured by gold activation foils (bare and covered with cadmium) in the fuel element number 108 (reaction: 197 Au(n,γ) 198 Au) at 451W overall reactor power. The fast neutron flux was measured by indium activation foils (reaction: 115 In(n,n') 115m In) in the fuel elements number 94 at 4510W overall reactor power. The neutron energy spectrum was adjusted by SAND II code with the data produced by the irradiation of seven activation detectors in the fuel element number 94 at 4510 W overall reactor power. The following reactions were used: 58 Fe(n,γ) 59 Fe, 232 Th(n,γ) 233 Th, 197 Au(n,γ) 198 Au, 59 Co(n,γ) 60 Co, 54 Fe(n,p) 54 Mn, 24 Mg(n,p) 24 Na, 47 Ti(n,p) 47 Sc, 48 Ti(n,p) 48 Sc and 115 In(n,n') 115m In. The experimental results compared to those obtained by CITATION (spatial distribution flux) and HAMMER (energetic distribution flux) code, showed good agreement. The results presented in this work are a good contribution for a better knowledge of spatial and energetic neutron flux distribution in the IEA-R1 reactor core, besides that the experimental procedure is easily applicable to another situations. (autor) [pt
Analysis of Neutron Flux Distribution in Rsg-Gas Reactor With U-Mo Fuels
Directory of Open Access Journals (Sweden)
Taswanda Taryo
2004-01-01
Full Text Available The use of U-Mo fuels in research reactors seems to be promising and, recently, world researchers have carried out these such activities actively. The National Nuclear Energy Agency (BATAN which owns RSG-GAS reactor available in Serpong Research Center for Atomic Energy should anticipate this trend. It is, therefore, this research work on the use of U-Mo fuels in RSG-GAS reactor should be carried out. The work was focused on the analysis of neutron flux distribution in the RSG-GAS reactor using different content of molybdenum in U-Mo fuels. To begin with, RSG-GAS reactor core model was developed and simulated into X, Y and Z dimensions. Cross section of materials based on the developed cells of standard and control fuels was then generated using WIMS-D5-B. The criticality calculations were finally carried out applying BATAN-2DIFF code. The results showed that the neutron flux distribution obtained in U-Mo-fuel-based RSG-GAS core is very similar to those achieved in the 300-gram sillicide-fuel-based RSG-GAS reactor core. Indeed, the utilization of the U-Mo RSG-GAS core can be very similar to that of the high-density sillicide reactor core and even could be better in the future.
High heat flux components with Be armour before and after neutron irradiation
International Nuclear Information System (INIS)
Lodato, A.; Derz, H.; Duwe, R.; Linke, J.; Roedig, M.
2000-01-01
Beryllium/copper mock-ups produced by different joining techniques have been tested in the electron beam facility JUDITH (Juelich Divertor Test Facility in Hot Cells) at Forschungszentrum Juelich. The experiments described in this paper represent the conclusive part of a test program started in 1994. The properties of non-irradiated Be/Cu joints have been characterised in a previous test campaign. Post-irradiation tests are now being carried out to investigate the neutron damage on the joints. The neutron irradiation on selected mock-ups has been carried out in the High Flux Reactor (HFR) at Petten (The Netherlands). Parametric finite element thermal analyses have been carried out to establish the allowable heat flux value to be applied during the tests. Screening tests up to power densities of ∼7 MW/m 2 and thermal fatigue tests up to 1000 cycles have been performed. None of these mock-ups showed any indication of failure. Post-mortem analyses (metallography, SEM) have also been conducted
Flux weighted method for solution of stiff neutron dynamic equations and its application
International Nuclear Information System (INIS)
Li Huiyun; Jiao Huixian
1987-12-01
To analyze reactivity event for nuclear power plants, it is necessary to solve the neutron dynamic equations, which is a group of typical stiff constant differential equations. Very small time steps could only be adopted when the group of equations is solved by common methods. However, a large time steps might be selected if the Flux Weighted Medthod introduced in this paper is used. Generally, weighted factor θ i1 is set as a constant. Naturally, this treatment method can decrease the accuracy of calculation for the increase of the steadiness of solving the equations. An accurate theoretical formula of 4 x 4 matrix of θ i1 is rigorously derived so that the accuracy of calculation is ensured, as well as the steadiness of solved equations is increased. This method have the advantage over classical Runge-kutta Method and other methods. The time steps could be increased by a factor of 1 ∼ 3 orders of magnitude so as to save a lot of computating time. The programe solving neutron dynamic equation, which is prepared by using Flux Weighted Method, could be sued for real time analog of training simulator, as well as for analysis and computation of reactivity event (including rod jumping out event)
Neutron and Gamma Fluxes and dpa Rates for HFIR Vessel Beltline Region (Present and Upgrade Designs)
Energy Technology Data Exchange (ETDEWEB)
Blakeman, E.D.
2001-01-11
The Oak Ridge National Laboratory (ORNL) High Flux Isotope Reactor (HFIR) is currently undergoing an upgrading program, a part of which is to increase the diameters of two of the four radiation beam tubes (HB-2 and HB-4). This change will cause increased neutron and gamma radiation dose rates at and near locations where the tubes penetrate the vessel wall. Consequently, the rate of radiation damage to the reactor vessel wall at those locations will also increase. This report summarizes calculations of the neutron and gamma flux (particles/cm{sup 2}/s) and the dpa rate (displacements/atom/s) in iron at critical locations in the vessel wall. The calculated dpa rate values have been recently incorporated into statistical damage evaluation codes used in the assessment of radiation induced embrittlement. Calculations were performed using models based on the discrete ordinates methodology and utilizing ORNL two-dimensional and three-dimensional discrete ordinates codes. Models for present and proposed beam tube designs are shown and their results are compared. Results show that for HB-2, the dpa rate in the vessel wall where the tube penetrates the vessel will be increased by {approximately}10 by the proposed enlargement. For HB-4, a smaller increase of {approximately}2.6 is calculated.
Baers, L. B.; Gutierrez, T. Rivero; Mendoza, R. A. Carrillo; Santana, G. Jimenez
1993-08-01
The second (conventional variance or Campbell signal), the third, and the modified fourth order central signal moments associated with the amplified and filtered currents from two electrodes of an ex-core neutron sensitive fission detector were measured versus the reactor power of the 1-MW TRIGA reactor in Mexico City. Two channels of a high-speed (400-MHz) multiplexing data sampler and an analog-to-digital converter with 12-b resolution and 1-Mword buffer memory were used. The data were further retrieved into a PC, and estimates for autocorrelation and cross-correlation moments up to the fifth order, coherence, skewness, excess, etc., quantities were calculated offline. Five-mode operation of the detector was achieved, including conventional counting rates and currents in agreement with theory and the authors' previous results with analog techniques. The signals are proportional to the neutron flux and reactor power in some flux ranges. The suppression of background noise is improved and the lower limit of the measurement range is extended as the order of moment is increased, in agreement with theory.
Performance modeling of parallel algorithms for solving neutron diffusion problems
International Nuclear Information System (INIS)
Azmy, Y.Y.; Kirk, B.L.
1995-01-01
Neutron diffusion calculations are the most common computational methods used in the design, analysis, and operation of nuclear reactors and related activities. Here, mathematical performance models are developed for the parallel algorithm used to solve the neutron diffusion equation on message passing and shared memory multiprocessors represented by the Intel iPSC/860 and the Sequent Balance 8000, respectively. The performance models are validated through several test problems, and these models are used to estimate the performance of each of the two considered architectures in situations typical of practical applications, such as fine meshes and a large number of participating processors. While message passing computers are capable of producing speedup, the parallel efficiency deteriorates rapidly as the number of processors increases. Furthermore, the speedup fails to improve appreciably for massively parallel computers so that only small- to medium-sized message passing multiprocessors offer a reasonable platform for this algorithm. In contrast, the performance model for the shared memory architecture predicts very high efficiency over a wide range of number of processors reasonable for this architecture. Furthermore, the model efficiency of the Sequent remains superior to that of the hypercube if its model parameters are adjusted to make its processors as fast as those of the iPSC/860. It is concluded that shared memory computers are better suited for this parallel algorithm than message passing computers
A continuously self regenerating high-flux neutron-generator facility
Rogers, A. M.; Becker, T. A.; Bernstein, L. A.; van Bibber, K.; Bleuel, D. L.; Chen, A. X.; Daub, B. H.; Goldblum, B. L.; Firestone, R. B.; Leung, K.-N.; Renne, P. R.; Waltz, C.
2013-10-01
A facility based on a next-generation, high-flux D-D neutron generator (HFNG) is being constructed at UC Berkeley. The current generator, designed around two RF-driven multicusp deuterium ion sources, is capable of producing a neutron output of >1011 n/s. A specially designed titanium-coated copper target located between the ion sources accelerates D+ ions up to 150 keV, generating 2.45 MeV neutrons through the d(d,3He)n fusion reaction. Deuterium in the target is self loaded and regenerating through ion implantation, enabling stable and continuous long-term operation. The proposed science program is focused on pioneering advances in the 40Ar/39Ar dating technique for geochronology, new nuclear data measurements, basic nuclear science research including statistical model studies of radiative-strength functions and level densities, and education. An overview of the facility and its unique capabilities as well as first measurements from the HFNG commissioning will be presented. Work supported by NSF Grant No. EAR-0960138, U.S. DOE LBL Contract No. DE-AC02-05CH11231, and U.S. DOE LLNL Contract No. DE-AC52-07NA27344.
Activation analysis of stainless steel flux monitors using 252Cf neutron sources
International Nuclear Information System (INIS)
Williams, J.G.; Newton, T.H. Jr.; Cogburn, C.O.
1984-01-01
Activation analysis was performed on stainless steel beads from a chain which is used in reactor pressure vessel surveillance experiments at the Arkansas Power and Light Company reactors. The beads allow monitoring of two fast and three thermal neutron induced reactions: 58 Ni(n,p) 58 Co, 54 Fe(n,p) 54 Mn, 58 Fe(n,γ) 59 Fe, 59 Co(n,γ) 60 Co and 50 Cr(n,γ) 51 Cr. The analysis was performed using 12 beads from various positions along 5 different batches of chain and standard materials in an H 2 O moderator tank using two intense californium sources which had a total neutron emission rate of 3.97 x 10 10 /s. Semiconductor gamma spectrometers were used to count the products of the above reactions in the specimens. The percentage by weight of the iron, chromium and cobalt in the beads were found to be 62.1%, 20.2% and 0.120%, respectively. The excellent uniformity found in the bead compositions demonstrates the reproducibility of the experimental techniques and enhances considerably the value of the beads as neutron flux montitors
Measuring neutron fluences and gamma/x-ray fluxes with CCD cameras
International Nuclear Information System (INIS)
Yates, G.J.; Smith, G.W.; Zagarino, P.; Thomas, M.C.
1991-01-01
The capability to measure bursts of neutron fluences and gamma/x-ray fluxes directly with charge coupled device (CCD) cameras while being able to distinguish between the video signals produced by these two types of radiation, even when they occur simultaneously, has been demonstrated. Volume and area measurements of transient radiation-induced pixel charge in English Electric Valve (EEV) Frame Transfer (FT) charge coupled devices (CCDs) from irradiation with pulsed neutrons (14 MeV) and Bremsstrahlung photons (4--12 MeV endpoint) are utilized to calibrate the devices as radiometric imaging sensors capable of distinguishing between the two types of ionizing radiation. Measurements indicate ∼.05 V/rad responsivity with ≥1 rad required for saturation from photon irradiation. Neutron-generated localized charge centers or ''peaks'' binned by area and amplitude as functions of fluence in the 10 5 to 10 7 n/cm 2 range indicate smearing over ∼1 to 10% of CCD array with charge per pixel ranging between noise and saturation levels
International Nuclear Information System (INIS)
Gambarini, G.; Martini, M.; Meinardi, F.; Raffaglio, C.; Salvadori, P.; Scacco, A.; Sichirollo, A.E.
1996-01-01
Thermoluminescent dosemeters (TLD), widely experimented and utilized in personal dosimetry, have some advantageous characteristics which induce one to employ them also in radiotherapy. The new radiotherapy techniques are aimed at selectively depositing a high dose in cancerous tissues. This goal is reached by utilising both conventional and other more recently proposed radiation, such as thermal neutrons and heavy charged particles. In these inhomogeneous radiation fields a reliable mapping of the spatial distribution of absorbed dose is desirable, and the utilized dosemeters have to give such a possibility without notably perturbing the radiation field with the materials of the dosemeters themselves. TLDs, for their small dimension and their tissue equivalence for most radiation, give good support in the mapping of radiation fields. After exposure to the high fluxes of therapeutic beams, some commercial TL dosemeters have shown a loss of reliability. An investigation has therefore be performed, both on commercial and on laboratory made phosphors, in order to investigate their behaviour in such radiation fields. In particular the thermal neutron and gamma ray mixed field of the thermal column of a nuclear reactor, of interest for Boron Neutron Capture Therapy (B.N.C.T.) and a proton beam, of interest for proton therapy, were considered. Here some results obtained with new TL phosphors exposed in such radiation fields are presented, after a short description of some radiation damage effect on commercial LiF TLDs exposed in the (n th ,γ) field of the thermal column of a reactor. (author)
International Nuclear Information System (INIS)
Filho, J. F. P.; Barichello, L. B.
2013-01-01
In this work, an analytical discrete ordinates method is used to solve a nodal formulation of a neutron transport problem in x, y-geometry. The proposed approach leads to an important reduction in the order of the associated eigenvalue systems, when combined with the classical level symmetric quadrature scheme. Auxiliary equations are proposed, as usually required for nodal methods, to express the unknown fluxes at the boundary introduced as additional unknowns in the integrated equations. Numerical results, for the problem defined by a two-dimensional region with a spatially constant and isotropically emitting source, are presented and compared with those available in the literature. (authors)
Bortolussi, S.; Protti, N.; Ferrari, M.; Postuma, I.; Fatemi, S.; Prata, M.; Ballarini, F.; Carante, M. P.; Farias, R.; González, S. J.; Marrale, M.; Gallo, S.; Bartolotta, A.; Iacoviello, G.; Nigg, D.; Altieri, S.
2018-01-01
University of Pavia is equipped with a TRIGA Mark II research nuclear reactor, operating at a maximum steady state power of 250 kW. It has been used for many years to support Boron Neutron Capture Therapy (BNCT) research. An irradiation facility was constructed inside the thermal column of the reactor to produce a sufficient thermal neutron flux with low epithermal and fast neutron components, and low gamma dose. In this irradiation position, the liver of two patients affected by hepatic metastases from colon carcinoma were irradiated after borated drug administration. The facility is currently used for cell cultures and small animal irradiation. Measurements campaigns have been carried out, aimed at characterizing the neutron spectrum and the gamma dose component. The neutron spectrum has been measured by means of multifoil neutron activation spectrometry and a least squares unfolding algorithm; gamma dose was measured using alanine dosimeters. Results show that in a reference position the thermal neutron flux is (1.20 ± 0.03) ×1010 cm-2 s-1 when the reactor is working at the maximum power of 250 kW, with the epithermal and fast components, respectively, 2 and 3 orders of magnitude lower than the thermal component. The ratio of the gamma dose with respect to the thermal neutron fluence is 1.2 ×10-13 Gy/(n/cm2).
International Nuclear Information System (INIS)
Taryo, Taswanda
2002-01-01
Multi Purpose Reactor G.A. Siwabessy (RSG-GAS) reactor core possesses 4 fission-chamber detectors to measure intermediate power level of RSG-GAS reactor. Another detector, also fission-chamber detector, is intended to measure power level of RSG-GAS reactor. To investigate influence of space to the neutron flux values for each detector measuring intermediate and power levels has been carried out. The calculation was carried out using combination of WIMS/D4 and CITATION-3D code and focused on calculation of neutron flux at different detector location of RSG-GAS typical working core various scenarios. For different scenarios, all calculation results showed that each detector, located at different location in the RSG-GAS reactor core, causes different neutron flux occurred in the reactor core due to spatial time effect
Directory of Open Access Journals (Sweden)
Strugalska-Gola Elzbieta
2017-01-01
Full Text Available This work was performed within the international project “Energy plus Transmutation of Radioactive Wastes” (E&T - RAW for investigations of energy production and transmutation of radioactive waste of the nuclear power industry. 89Y (Yttrium 89 samples were located in the Quinta assembly in order to measure an average high neutron flux density in three different energy ranges using deuteron and proton beams from Dubna accelerators. Our analysis showed that the neutron density flux for the neutron energy range 20.8 - 32.7 MeV is higher than for the neutron energy range 11.5 - 20.8 MeV both for protons with an energy of 0.66 GeV and deuterons with an energy of 2 GeV, while for deuteron beams of 4 and 6 GeV we did not observe this.
Energy Technology Data Exchange (ETDEWEB)
Strugar, P; Raisic, N; Obradovic, D; Jovanovic, S [Institute of Nuclear Sciences Boris Kidric, Vinca, Beograd (Serbia and Montenegro)
1965-05-01
A series of critical experiments were performed on the RB reactor in order to determine the thermal neutron flux increase in the central horizontal reflector formed by a split reactor core. The objectives of these experiments were to study the possibilities of improving the thermal neutron flux characteristics of the neutron beam in the horizontal beam tube of the RA research reactor. The construction of RA reactor enables to split the core in two, to form a central horizontal reflector in front of the beam tube. This is achieved by replacing 2% enriched uranium slugs in the fuel channel by dummy aluminium slugs. The purpose of the first series of experiments was to study the gain in thermal neutron component inside the horizontal reflector and the loss of reactivity as a function of the lattice pitch and central reflector thickness.
International Nuclear Information System (INIS)
Ertek, C.
1981-02-01
This work is a continuation of the work performed within the IAEA programme on standardization of reactor radiation measurements, one of the important objectives of which is the assistance to laboratories in Member States to implement or intercompare the multiple foil activation techniques for different neutron field measurements. The importance of these techniques is well recognized. In CESNEF-FERMI Politecnico di Milano, Italy, they have installed near the core of a water boiler of 50kW, a neutron filter made of B 4 C in order to obtain a neutron flux density spectrum that could be of utility in intercalibration problems connected with irradiation in fast assemblies. Dr. V. Sangiust from CESNEF kindly sent the input guess neutron flux density spectrum and a series of measured reaction rates to be treated by the IAEA Seibersdorf laboratory using the SAND-II and the WINDOWS unfolding codes. The meaningful comparison using partly the same ENDF/B IV cross section data is performed. In the present work we extended the investiga tion using WINDOWS unfolding code for different numbers of activation fo ils or reaction rates
Describing function theory as applied to thermal and neutronic problems
International Nuclear Information System (INIS)
Nassersharif, B.
1983-01-01
Describing functions have traditionally been used to obtain the solutions of systems of ordinary differential equations. In this work the describing function concept has been extended to include nonlinear, distributed parameter partial differential equations. A three-stage solution algorithm is presented which can be applied to any nonlinear partial differential equation. Two generalized integral transforms were developed as the T-transform for the time domain and the B-transform for the spatial domain. The thermal diffusion describing function (TDDF) is developed for conduction of heat in solids and a general iterative solution along with convergence criteria is presented. The proposed solution method is used to solve the problem of heat transfer in nuclear fuel rods with annular fuel pellets. As a special instance the solid cylindrical fuel pellet is examined. A computer program is written which uses the describing function concept for computing fuel pin temperatures in the radial direction during reactor transients. The second problem investigated was the neutron diffusion equation which is intrinsically different from the first case. Although, for most situations, it can be treated as a linear differential equation, the describing function method is still applicable. A describing function solution is derived for two possible cases: constant diffusion coefficient and variable diffusion coefficient. Two classes of describing functions are defined for each case which portray the leakage and absorption phenomena. For the specific case of a slab reactor criticality problem the comparison between analytical and describing function solutions revealed an excellent agreement
Boundary element methods applied to two-dimensional neutron diffusion problems
International Nuclear Information System (INIS)
Itagaki, Masafumi
1985-01-01
The Boundary element method (BEM) has been applied to two-dimensional neutron diffusion problems. The boundary integral equation and its discretized form have been derived. Some numerical techniques have been developed, which can be applied to critical and fixed-source problems including multi-region ones. Two types of test programs have been developed according to whether the 'zero-determinant search' or the 'source iteration' technique is adopted for criticality search. Both programs require only the fluxes and currents on boundaries as the unknown variables. The former allows a reduction in computing time and memory in comparison with the finite element method (FEM). The latter is not always efficient in terms of computing time due to the domain integral related to the inhomogeneous source term; however, this domain integral can be replaced by the equivalent boundary integral for a region with a non-multiplying medium or with a uniform source, resulting in a significant reduction in computing time. The BEM, as well as the FEM, is well suited for solving irregular geometrical problems for which the finite difference method (FDM) is unsuited. The BEM also solves problems with infinite domains, which cannot be solved by the ordinary FEM and FDM. Some simple test calculations are made to compare the BEM with the FEM and FDM, and discussions are made concerning the relative merits of the BEM and problems requiring future solution. (author)
High-accuracy determination of the neutron flux in the new experimental area nTOF-EAR2 at CERN
Energy Technology Data Exchange (ETDEWEB)
Sabate-Gilarte, M. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Universidad de Sevilla, Departamento de Fisica Atomica, Molecular y Nuclear, Sevilla (Spain); Barbagallo, M.; Colonna, N.; Damone, L.; Belloni, F.; Mastromarco, M.; Tagliente, G.; Variale, V. [Istituto Nazionale di Fisica Nucleare, Sezione di Bari, Bari (Italy); Gunsing, F.; Berthoumieux, E.; Diakaki, M.; Papaevangelou, T.; Dupont, E. [Universite Paris-Saclay, CEA Irfu, Gif-sur-Yvette (France); Zugec, P.; Bosnar, D. [University of Zagreb, Department of Physics, Faculty of Science, Zagreb (Croatia); Vlachoudis, V.; Aberle, O.; Brugger, M.; Calviani, M.; Cardella, R.; Cerutti, F.; Chiaveri, E.; Ferrari, A.; Kadi, Y.; Losito, R.; Macina, D.; Montesano, S.; Rubbia, C. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Chen, Y.H.; Audouin, L.; Tassan-Got, L. [Centre National de la Recherche Scientifique/IN2P3 - IPN, Orsay (France); Stamatopoulos, A.; Kokkoris, M.; Tsinganis, A.; Vlastou, R. [National Technical University of Athens (NTUA), Athens (Greece); Lerendegui-Marco, J.; Cortes-Giraldo, M.A.; Guerrero, C.; Quesada, J.M. [Universidad de Sevilla, Departamento de Fisica Atomica, Molecular y Nuclear, Sevilla (Spain); Villacorta, A. [University of Salamanca, Salamanca (Spain); Cosentino, L.; Finocchiaro, P.; Piscopo, M. [INFN, Laboratori Nazionali del Sud, Catania (Italy); Musumarra, A. [INFN, Laboratori Nazionali del Sud, Catania (Italy); Universita di Catania, Dipartimento di Fisica, Catania (Italy); Andrzejewski, J.; Gawlik, A.; Marganiec, J.; Perkowski, J. [University of Lodz, Lodz (Poland); Becares, V.; Balibrea, J.; Cano-Ott, D.; Garcia, A.R.; Gonzalez, E.; Martinez, T.; Mendoza, E. [Centro de Investigaciones Energeticas Medioambientales y Tecnologicas (CIEMAT), Madrid (Spain); Bacak, M.; Weiss, C. [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Technische Universitaet Wien, Wien (Austria); Baccomi, R.; Milazzo, P.M. [Istituto Nazionale di Fisica Nucleare, Sezione di Trieste, Trieste (Italy); Barros, S.; Ferreira, P.; Goncalves, I.F.; Vaz, P. [Instituto Superior Tecnico, Lisbon (Portugal); Becvar, F.; Krticka, M.; Valenta, S. [Charles University, Prague (Czech Republic); Beinrucker, C.; Goebel, K.; Heftrich, T.; Reifarth, R.; Schmidt, S.; Weigand, M.; Wolf, C. [Goethe University Frankfurt, Frankfurt (Germany); Billowes, J.; Frost, R.J.W.; Ryan, J.A.; Smith, A.G.; Warren, S.; Wright, T. [University of Manchester, Manchester (United Kingdom); Caamano, M.; Deo, K.; Duran, I.; Fernandez-Dominguez, B.; Leal-Cidoncha, E.; Paradela, C.; Robles, M.S. [University of Santiago de Compostela, Santiago de Compostela (Spain); Calvino, F.; Casanovas, A.; Riego-Perez, A. [Universitat Politecnica de Catalunya, Barcelona (Spain); Castelluccio, D.M.; Lo Meo, S. [Agenzia Nazionale per le Nuove Tecnologie (ENEA), Bologna (Italy); Istituto Nazionale di Fisica Nucleare, Sezione di Bologna, Bologna (Italy); Cortes, G.; Mengoni, A. [Agenzia Nazionale per le Nuove Tecnologie (ENEA), Bologna (Italy); Domingo-Pardo, C.; Tain, J.L. [Universidad de Valencia, Instituto de Fisica Corpuscular, Valencia (Spain); Dressler, R.; Heinitz, S.; Kivel, N.; Maugeri, E.A.; Schumann, D. [Paul Scherrer Institut (PSI), Villingen (Switzerland); Furman, V.; Sedyshev, P. [Joint Institute for Nuclear Research (JINR), Dubna (Russian Federation); Gheorghe, I.; Glodariu, T.; Mirea, M.; Oprea, A. [Horia Hulubei National Institute of Physics and Nuclear Engineering, Magurele (Romania); Goverdovski, A.; Ketlerov, V.; Khryachkov, V. [Institute of Physics and Power Engineering (IPPE), Obninsk (Russian Federation); Griesmayer, E.; Jericha, E.; Kavrigin, P.; Leeb, H. [Technische Universitaet Wien, Wien (Austria); Harada, H.; Kimura, A. [Japan Atomic Energy Agency (JAEA), Tokai-mura (Japan); Hernandez-Prieto, A. [European Organization for Nuclear Research (CERN), Geneva (CH); Universitat Politecnica de Catalunya, Barcelona (ES); Heyse, J.; Schillebeeckx, P. [European Commission, Joint Research Centre, Geel (BE); Jenkins, D.G. [University of York, York (GB); Kaeppeler, F. [Karlsruhe Institute of Technology, Karlsruhe (DE); Katabuchi, T. [Tokyo Institute of Technology, Tokyo (JP); Lederer, C.; Lonsdale, S.J.; Woods, P.J. [University of Edinburgh, School of Physics and Astronomy, Edinburgh (GB); Licata, M.; Massimi, C.; Vannini, G. [Istituto Nazionale di Fisica Nucleare, Sezione di Bologna, Bologna (IT); Universita di Bologna, Dipartimento di Fisica e Astronomia, Bologna (IT); Mastinu, P. [Istituto Nazionale di Fisica Nucleare, Sezione di Legnaro, Legnaro (IT); Matteucci, F. [Istituto Nazionale di Fisica Nucleare, Sezione di Trieste, Trieste (IT); Universita di Trieste, Dipartimento di Astronomia, Trieste (IT); Mingrone, F. [European Organization for Nuclear Research (CERN), Geneva (CH); Istituto Nazionale di Fisica Nucleare, Sezione di Bologna, Bologna (IT); Nolte, R. [Physikalisch-Technische Bundesanstalt (PTB), Braunschweig (DE); Palomo-Pinto, F.R. [Universidad de Sevilla, Dept. Ingenieria Electronica, Escuela Tecnica Superior de Ingenieros, Sevilla (ES); Patronis, N. [University of Ioannina, Ioannina (GR); Pavlik, A. [University of Vienna, Faculty of Physics, Vienna (AT); Porras, J.I. [University of Granada, Granada (ES); Praena, J. [Universidad de Sevilla, Departamento de Fisica Atomica, Molecular y Nuclear, Sevilla (ES); University of Granada, Granada (ES); Rajeev, K.; Rout, P.C.; Saxena, A.; Suryanarayana, S.V. [Bhabha Atomic Research Centre (BARC), Mumbai (IN); Rauscher, T. [University of Hertfordshire, Centre for Astrophysics Research, Hatfield (GB); University of Basel, Department of Physics, Basel (CH); Tarifeno-Saldivia, A. [Universitat Politecnica de Catalunya, Barcelona (ES); Universidad de Valencia, Instituto de Fisica Corpuscular, Valencia (ES); Ventura, A. [Istituto Nazionale di Fisica Nucleare, Sezione di Bologna, Bologna (IT); Wallner, A. [Australian National University, Canberra (AU)
2017-10-15
A new high flux experimental area has recently become operational at the nTOF facility at CERN. This new measuring station, nTOF-EAR2, is placed at the end of a vertical beam line at a distance of approximately 20 m from the spallation target. The characterization of the neutron beam, in terms of flux, spatial profile and resolution function, is of crucial importance for the feasibility study and data analysis of all measurements to be performed in the new area. In this paper, the measurement of the neutron flux, performed with different solid-state and gaseous detection systems, and using three neutron-converting reactions considered standard in different energy regions is reported. The results of the various measurements have been combined, yielding an evaluated neutron energy distribution in a wide energy range, from 2 meV to 100 MeV, with an accuracy ranging from 2%, at low energy, to 6% in the high-energy region. In addition, an absolute normalization of the nTOF-EAR2 neutron flux has been obtained by means of an activation measurement performed with {sup 197}Au foils in the beam. (orig.)
One procedure for determination of the neutron flux in the nuclear reactor fuel
Energy Technology Data Exchange (ETDEWEB)
Bulovic, V; Krtil, J; Maksimovic, Z; Martinc, R [Institute of nuclear sciences Boris Kidric, Vinca, Beograd (Yugoslavia)
1979-09-15
Possibility of determination of the neutron flux in the fuel of a heavy water reactor has been examined. In determination of the flux an iterative procedure was used to compare calculated and measured contents of several fission products. The former contents were determined by calculation of the burning process balance and the latter by non-destructive gamma-spectrometric analysis of fuel. The obtained results prove the possibility of such determination of not only the average value of the flux but also of the change of its intensity during utilization of fuel (author) U radu je ispitivana mogucnost odredjivanja fluksa neutrona u gorivu teskovodnog nuklearnog reaktora. Pri odredjivanju fluksa koriscen je iterativni postupak, u kome se porede izracunati i izmereni sadrzaji nekoliko gama-radioaktivnih fisionih produkata. Prvi sadrzaji su odredjivani preko proracuna bilansa procesa sagorevanja, a drugi - nedestruktivnom gama-spektrometrijskom analizom goriva. Dobijeni rezultati potvrdjuju mogucnost ovakvog odredjivanja ne samo srednje vrednosti fluksa, vec i promene njegovog intenziteta u toku koriscenja goriva (author)
International Nuclear Information System (INIS)
Ghassoun, Jillali; Jehoauni, Abdellatif
2000-01-01
In practice, the estimation of the flux obtained by Fredholm integral equation needs a truncation of the Neuman series. The order N of the truncation must be large in order to get a good estimation. But a large N induces a very large computation time. So the conditional Monte Carlo method is used to reduce time without affecting the estimation quality. In a previous works, in order to have rapid convergence of calculations it was considered only weakly diffusing media so that has permitted to truncate the Neuman series after an order of 20 terms. But in the most practical shields, such as water, graphite and beryllium the scattering probability is high and if we truncate the series at 20 terms we get bad estimation of flux, so it becomes useful to use high orders in order to have good estimation. We suggest two simple techniques based on the conditional Monte Carlo. We have proposed a simple density of sampling the steps for the random walk. Also a modified stretching factor density depending on a biasing parameter which affects the sample vector by stretching or shrinking the original random walk in order to have a chain that ends at a given point of interest. Also we obtained a simple empirical formula which gives the neutron flux for a medium characterized by only their scattering probabilities. The results are compared to the exact analytic solution, we have got a good agreement of results with a good acceleration of convergence calculations. (author)
Comparisons of Measured and Calculated Neutron Fluxes in Laminated iron and Heavy Water
Energy Technology Data Exchange (ETDEWEB)
Aalto, E
1964-10-15
Measurements of neutron fluxes have been performed in configurations depicting the regions extending radially and axially outwards from the core of a PHWR reactor in order to test the accuracy of the available methods in shield design on thin alternating laminae of Fe and D{sub 2}O. A 'dry' experimental set-up was constructed, i.e. the D{sub 2}O was contained in flat tanks made of Al. The first set of measurements was performed through solid Fe and D{sub 2}O layers, and only the results of these experiments are described in this report. The set-up allowed measurements also in a mock-up of a reactor top penetrated by D{sub 2}O or air-filled channels (to be reported later). The results are compared to fluxes calculated by the British 18-group removal-diffusion method and by the NRN method developed at AE. The results show that the values predicted may be expected to be within a factor of 2 from the true values in most cases. The predicted relative flux distributions follow the observed ones with a very good accuracy in spite of the apparent misuse of diffusion theory for the thin regions in question. Finally, it is shown that the predicted change in the fast spectrum while penetrating these set-ups should be confirmable with certain threshold detectors.
International Nuclear Information System (INIS)
Graul, J.; Mueller, R.G.; Wagner, E.
1975-05-01
The applicability of semiconductor detectors for high thermal neutron flux densities is theoretically estimated and experimentally examined. For good thermal stability and low radiation capture rate silicon carbide is used as semiconductor material, produced in polycristalline layers to achieve high radiation resistance. The relations between crystallinity, photoelectric sensitivity and radiation resistance are shown. The radiation resistance of polycrystalline SiC-probes is approximately 100 times greater than that of conventional single crystal radiation detectors. For thermal neutron measurement they can be used in the flux range of approx. 10 10 13 (cm -2 sec -1 ) with operation times of 1.6 a >= tsub(b,max) >= 30 d, resp. (orig.) [de
International Nuclear Information System (INIS)
Lambri, O.A.; Salvatierra, L.M.; Sanchez, F.A.; Matteo, C.L.; Sorichetti, P.A.; Celauro, C.A.
2005-01-01
Neutron irradiation at room temperature were performed on EPDM (ethylene-propylene-diene monomer) in two different nuclear reactors at different fluxes. The effect of the irradiation on the chain arrangement in the polymer, as a function of the dose is discussed. Different crystal concentrations and crystal shapes, developed by chemi-crystallisation, are obtained depending on the neutron dose. In addition the radiation damage degree in the polymer depends both on the dose and the flux level. Dynamical mechanical analysis, swelling studies, X-ray diffraction, differential thermal analysis and infrared studies were employed as experimental techniques
International Nuclear Information System (INIS)
Moraes, O.M.G. de.
1984-01-01
A systematic study about the registration characteristics of synthetic fused quartz 'Suprasil I' use as a detector of fission fragments under high flux of reactor neutrons and the effects of irradiation on it was performed. Fission fragments of 252 Cf, gamma radiation doses of of 60 Co up to 150 MGy, and integrated neutrons fluxes up to 10 20 n/cm 2 were used. A model to explain the effects on track registration and development characteristics of 'Suprasil I' irradiated on reactors were proposed, based on the obtained results for efficiency an for annealing. (C.G.C.) [pt
Miller, M E; Mariani, L E; Gonçalves-Carralves, M L Sztejnberg; Skumanic, M; Thorp, S I
2004-11-01
A novel system to determine thermal neutron flux in real time during NCT treatments was developed in the National Atomic Energy Commission of Argentina. The system is based on a special self-powered detector that can be implanted in patients owing to its small size and biocompatibility. High voltage is not required to operate this kind of detectors, which is a considerable advantage in terms of medical uses. By choosing the appropriate materials, it was possible to obtain a prototype with thermal neutron sensitivity providing for an adequate signal level in typical NCT thermal fluxes. It was also possible to minimize gamma response in order to neglect its contribution.
Energy Technology Data Exchange (ETDEWEB)
Miller, M.E. E-mail: miller@cae.cnea.gov.ar; Mariani, L.E.; Sztejnberg Goncalves-Carralves, M.L.; Skumanic, M.; Thorp, S.I
2004-11-01
A novel system to determine thermal neutron flux in real time during NCT treatments was developed in the National Atomic Energy Commission of Argentina. The system is based on a special self-powered detector that can be implanted in patients owing to its small size and biocompatibility. High voltage is not required to operate this kind of detectors, which is a considerable advantage in terms of medical uses. By choosing the appropriate materials, it was possible to obtain a prototype with thermal neutron sensitivity providing for an adequate signal level in typical NCT thermal fluxes. It was also possible to minimize gamma response in order to neglect its contribution.
International Nuclear Information System (INIS)
Fourmentel, D.
2013-01-01
A new Material Testing Reactor (MTR), the Jules Horowitz Reactor (JHR), is under construction at the CEA Cadarache (French Alternatives Energies and Atomic Energy Commission). From 2016 this new MTR will be a new facility for the nuclear research on materials and fuels. The quality of the experiments to be conducted in this reactor is largely linked to the good knowledge of the irradiation conditions. Since 2009, a new research program called IN-CORE1 'Instrumentation for Nuclear radiations and Calorimetry Online in Reactor' is under progress between CEA and Aix-Marseille University in the framework of a joint laboratory LIMMEX2. This program aims to improve knowledge of the neutron and photon fluxes in the RJH core, with one hand, an innovative instrumentation performing mapping of experimental locations, and on the other hand by coupling neutron flux, photon flux and thermal measurements. Neutron flux expected in the JHR core is about 10 15 n.cm -2 .s -1 and nuclear heating up to 20 W.g -1 for a nominal power of 100 MWth. One of the challenges is to identify sensors able to measure such fluxes in JHR experimental conditions and to determine how to analyse the signals delivered by these sensors with the most appropriate methods. The thesis is part of this ambitious program and aims to study the potential and the interest of the combination of radiation measurements in the prospect of a better assessment of the levels of neutron flux, gamma radiation and nuclear heating in the JHR experimental locations. The first step of IN-CORE program is to develop and operate an instrumented device called CARMEN-1 adapted to the mapping of the OSIRIS reactor, then to develop a second version called CARMEN-2 dedicated to experiments in the JHR core, especially for its start-up. This experiment was the opportunity to test all the radiation sensors which could meet the needs of JHR, including recently developed sensors. Reference neutron measurements are performed by activation
International Nuclear Information System (INIS)
2012-01-01
On October, 30 - November, 2 in State Scientific Center of the Russian Federation - Institute for Physics and Power Engineering named after A.I. Leypunsky a seminar Neutron-physical problems of nuclear power engineering - Neutronika-2012 took place. On the seminar the following problems were discussed: justification of neutron-physical characteristics of reactor facilities and innovation projects; constant support of neutron-physical calculations of nuclear power installations; numerical simulation during solving reactor physics problems; simulation of neutron-physical processes in reactor facilities by Monte Carlo method; development and verification of programs for reactor facilities neutron-physical calculations; algorithms and programs for solving nonstationary problems of neutron-physical calculation of nuclear reactors; analysis of integral and reactor experiments, experimental database; justification of nuclear and radiation safety of fuel cycle [ru