Operations manual for the megachannel gamma-ray coincidence system
International Nuclear Information System (INIS)
Ruhter, W.
1977-01-01
To aid in the study of nuclear structures, a megachannel pulse-height coincidence analysis system on a PDP-8 computer was constructed. The system digitizes the energies of coincident gamma-rays and stores the resultant information on a moving-head disk. The system uses a minicomputer to sort and store gamma-gamma coincident information on line. The megachannel system and how to use it are described
Research on digital multi-channel pulse height analysis techniques
International Nuclear Information System (INIS)
Xiao Wuyun; Wei Yixiang; Ai Xianyun; Ao Qi
2005-01-01
Multi-channel pulse height analysis techniques are developing in the direction of digitalization. Based on digital signal processing techniques, digital multi-channel analyzers are characterized by powerful pulse processing ability, high throughput, improved stability and flexibility. This paper analyzes key techniques of digital nuclear pulse processing. With MATLAB software, main algorithms are simulated, such as trapezoidal shaping, digital baseline estimation, digital pole-zero/zero-pole compensation, poles and zeros identification. The preliminary general scheme of digital MCA is discussed, as well as some other important techniques about its engineering design. All these lay the foundation of developing homemade digital nuclear spectrometers. (authors)
Megachannel γ--γ coincidence system using a PDP-8/E computer and moving-head disks
International Nuclear Information System (INIS)
Ruhter, W.D.; Camp, D.C.; Mann, L.G.; Niday, J.B.; Siemens, P.D.
1976-01-01
A megachannel pulse-height analysis system using a PDP-8/E computer and two moving-head disk memories has been developed. The system has a storage capacity of 220 memory locations, is capable of processing 1100 events/s, and provides on-line sorting and disk storage. An X- or Y-pulse-height spectrum in coincidence with one or several arbitrary pulse-height windows can be assembled in core for scope display and spectral analysis within 2 to 20 seconds. Reconstruction of a complete X- or Y-pulse-height spectrum requires about 3 minutes
Data processing for the multichannel pulse height analysis system ND-50/50, (1)
International Nuclear Information System (INIS)
Katagiri, Masaki
1977-03-01
The multichannel pulse height analysis system ND-50/50 manufactured by Nuclear Data Inc. consists of a 4096 channel pulse height analyzer and a small computer PDP-8/L (Digital Equipment Corporation). A conversational interpretive language, MACAL (Multichannel Analyzer CALculator) has been developed to analyze gamma-ray spectra in ND-50/50. It is a modification of the FOCAL (FOrmula CALculator) language of Digital Equipment Corporation. MACAL consists of imperative English commands and mathematical expressions, and has standard mathematical functions and functions for controlling the multichannel pulse height analyzer and I/O devices (high-speed reader, high-speed punch, and digital plotter). With MACAL, the following five programs were prepared : 1) program for controlling the multichannel pulse height analyzer and data handling, 2) program for automatically analyzing gamma-ray spectra obtained by germanium detectors, 3) program for automatically measuring characteristics of germanium detectors, 4) program for calculating the center energy of gamma-ray peaks, and 5) program for plotting the pulse height distribution and diagraming the results obtained with the programs. By combining the programs according to experimental purposes, the system can be highly effective. (auth.)
Organization of pulse-height analysis programs for high event rates
Energy Technology Data Exchange (ETDEWEB)
Cohn, C E [Argonne National Lab., Ill. (USA)
1976-09-01
The ability of a pulse-height analysis program to handle high event rates can be enhanced by organizing it so as to minimize the time spent in interrupt housekeeping. Specifically, the routine that services the data-ready interrupt from the ADC should test whether another event is ready before performing the interrupt return.
Program for the analysis of pulse height spectra and the background from a proportional detector
International Nuclear Information System (INIS)
Flores-Llamas, H.; Yee-Madeira, H.; Contreras-Puente, G.; Zamorano-Ulloa, R.
1991-01-01
A PC-Fortran program is presented for fitting of lineshapes and the analysis of pulse height spectra obtainable with proportional detectors. The common fitting and analysis of pulse height spectra by means of mixed Gaussian lineshapes is readily improved by using Voigt lineshapes. In addition, the background can be evaluated during the fitting process without the need of extra measurements. As an application of the program, a pulse height transmission spectrum accumulated with a static 57 Co source and detected with an argon-metane proportional detector, was least squares fitted to an elaborated complex trial lineshape function containing two Voigt lines plus a straight line. The fitting straight line parameters a and b characterize quantitatively the background. The very good PC-fitting obtained shows that the fitting of experimental spectra with the more realistic Voigt lineshapes is no longer a formidable task and that it is possible to evaluate and subtract the background inherent to the experiment during the fitting process. (orig.)
A multistation pulse height analysis system based on a PDP9L computer
International Nuclear Information System (INIS)
Ellis, P.J.
1976-06-01
A PDP9L computer is used as the basis for a multistation pulse height analysis system. Each station which is equipped with an analog-to-digital converter, a display and a keyboard-printer can perform the basic operations associated with a commercial hard-wired system. The stations are independent of one another but two or more accumulating or displaying simultaneously result in a slight increase in analog-to-digital converter dead time. The keyboard-printer operates in conversational mode with no output forthcoming if incorrect characters are entered. System procedures are initiated from the station keyboard to control the display, accumulate, integrate, and read and write functions as flexibly as possible without excessively increasing the program-to-data ratio for core usage. This system was designed to enable software to be modified when different procedures become necessary, and to enable adaptation of the available equipment to other modes of accumulation and display. (author)
FPGA-based technology for Pulse Height Analysis in nuclear spectrometry system
International Nuclear Information System (INIS)
Andrianiaina, H.; Raoelina Andriambololona; Rajaobelison, J.; Rambolamanana, G.; Roengen, H.
2009-01-01
Facing the rapid technology development applied in nuclear instruments, Madagascar-INSTN has taken measures to estabilish and enhance the national capabilities in their maintenance, calibration, design/modification and repair, which is a key factor in the development of sustainable nuclear technology.The maintenance will be more convenient if all technical resources are made available: this is only possible when the acces to designer side of the instruments is permitted. Research and instrumentation development projects have been developed to support and to fit the local needs: Pulse Height Analysis algorithm is designed within FPGA and VHDL programming technics to build a Multi-Input Multi-Channel Analyzer. Description and measurement results with the MIMCA will be described in the present paper
Fuzzy coarse coding UP-grades A/D converters for pulse-height analysis
Energy Technology Data Exchange (ETDEWEB)
Casoli, P; Maranesi, P [Politecnico di Milano (Italy). Ist. di Ingegneria Nucleare
1983-01-15
Since they were first employed in nuclear pulse-height analysis, analog-to-digital converters show an evolution characterized by a continuous increase in their amplitude resolution. This trend looks as if it will continue in the immediate future but this forecast disagrees with the fact that the circuit complexity of the ADCs realized by conventional techniques rises sharply beyond 12 bit resolution. This paper describes and proposes a new A/D encoding technique. A successive-approximation ADC operating over a limited input range and having correspondingly a reduced resolution is employed. In order to adapt the input signal to this dynamic limitation, the signal is at first roughtly and quickly estimated by a flash converter and then, if necessary, an analog level is subtracted through a DAC. The input digit of this DAC is added to the result of the successive-approximations conversion to get the final correct result. The inherent differential non-linearities of the successive-approximations ADC and of the DAC are avoided by the combined actions of a sliding scale circuit and of an innovative circuit named 'shaker'. The satisfactory performance of the encoder has been experimentally verified through a 13 bit prototype. The simplicity of the proposed technique in comparison to conventional ones leads to the possibility of further raising the standard performance of nuclear spectroscopy ADCs.
Laboratory tests of the Pulse Height Analysis system for Wendelstein 7-X
Kubkowska, M.; Czarnecka, A.; Figacz, W.; Jabłoński, S.; Kaczmarczyk, J.; Krawczyk, N.; Ryć, L.; Biedermann, C.; Koenig, R.; Thomsen, H.; Weller, A.; W7-X Team
2015-10-01
A pulse height analysis (PHA) system has been designed and manufactured for the Wendelstein 7-X stellarator, in such a way as to be already compatible with later quasi-continuous operation requirements. The diagnostic will provide X-ray spectra with energy resolution better than 180 eV . The system has three energy channels: 0.25-20 keV, 0.95-20 keV and 1.5-20 keV . For each channel a separate Silicon Drift Detector (SDD) equipped with a suitably selected beryllium foil is used. The range of the 3 energy channels can be further adapted to particular experiments by moving via a pneumatic actuator additional beryllium filters in front of the fixed ones. The PHA system is intended for measuring impurity species (e.g. C, Fe, Ni), electron temperature and for investigating possible suprathermal tails in the spectra. The system will be installed on the horizontal port AEK50 on W7-X. The SDD detectors, the replaceable filters and the adjustable piezo driven slits which allow to suitably adapt the X-ray signal intensity are mounted inside a vacuum chamber which is connected to the plasma vessel via a gate valve. The on-air diagnostic components are the preamplifiers, the Digital X-Ray Processor (XIA, U.S.A.), a computer, and an X-ray calibration source. For controlling the operation of the entire diagnostic system, as well as, for the data acquisition of the electrical pulses coming a special code was developed. The paper presents the construction of the PHA system for W7-X and the laboratory tests of its mechanical parts together with the information on the code developed to operate the diagnostic. The diagnostic was also tested and characterised by measuring Fe55 spectrum and fluorescence spectra of Ni, Fe, Cr and Cu induced by an X-ray mini-tube.
Simulation of pulse height analysis soft X-ray spectra expected from W7-X
Jabłoński, S.; Czarnecka, A.; Kubkowska, M.; Ryć, L.; Weller, A.; Biedermann, C.; König, R.; W7-X Team
2015-10-01
A computer code named RayX has been developed for checking the performance of a spectroscopy system and optimizing individual parts, like detectors and filters for the pulse height analysis (PHA) diagnostic system designed for the stellarator W7-X. Using the code, the intensity and shape of the X-ray spectra are simulated for different plasma scenarios characterized by varying the temperature and density profiles as well as the electron cyclotron resonance heating (ECRH) power over a wide range. In the simulations of the recorded spectra, the influence of geometrical configuration changes of the diagnostic system (pinhole size, detector size, location of each diagnostic component), of the timing of data collection, as well as of the type and thickness of filters are being investigated. The atomic processes of free-free (bremsstrahlung), free-bound (recombination radiation), and bound-bound emission (line radiation) are considered. For the impurities fractional abundancies of 3% carbon (C), 0.5% oxygen (O) and 0.002% iron (Fe) are taken into account. Information about the number of photons which reach the detector and the current generated inside the detector is given. It is shown that the distance between pinhole and detector has a larger impact on the registered spectra (intensity and total number of photons) than the distance between plasma and pinhole. Based on the results of the simulations, the expected optimal positions of the individual components (pinholes, detectors) were defined for the PHA W7-X diagnostic system. A shorter version of this contribution is due to be published in PoS at: 1st EPS conference on Plasma Diagnostics
International Nuclear Information System (INIS)
Sesnic, S.; Diesso, M.; Hill, K.; Holland, A.; Pohl, F.
1988-01-01
Because soft-x-ray pulse-height-analysis spectra contain chordal information, the electron temperature and the radiation intensity (enhancement factor) measurements do not represent the local values. The correction factors for the electron temperature and the enhancement factor as a function of the temperature and density profile parameters and the energy are obtained. The spectrum distortion due to pulse pileup effects is also evaluated. A set of curves is given from which the distortion of the spectrum can be obtained if the electron temperature, the Be filter thickness, and the electronic parameters of the acquisition system are known. PG 1810,1812 ID 131801CON N X-ray diagnostics TT Profile correction to electron temperature and enhancement factor in soft-x-ray pulse-height-analysis measurements in tokamaks AU S. Sesnic, M. Diesso, K. Hill, and A. Holland LO Princeton University, Plasma Physics Laboratory, P.O. Box 451, Princeton, New Jersey 08543 AU F. Pohl LO Max-Planck Institut fuer Plasmaphysik, 8046-Garching, Federal Republic of Germany SD (Presented on 16 March 1988) AB Because soft-x-ray pulse-height-analysis spectra contain chordal information, the electron temperature and the radiation intensity (enhancement factor) measurements do not represent the local values. The correction factors for the electron temperature and the enhancement factor as a function of the temperature and density profile parameters and the energy are obtained. The spectrum distortion due to pulse pileup effects is also evaluated. A set of curves is given from which the distortion of the spectrum can be obtained if the electron tempe
Automatic analysis algorithm for radionuclide pulse-height data from beta-gamma coincidence systems
International Nuclear Information System (INIS)
Foltz Biegalski, K.M.
2001-01-01
There are two acceptable noble gas monitoring measurement modes for Comprehensive Nuclear-Test-Ban-Treaty (CTBT) verification purposes defined in CTBT/PC/II/WG.B/1. These include beta-gamma coincidence and high-resolution gamma-spectrometry. There are at present no commercial, off-the-shelf (COTS) applications for the analysis of β-γ coincidence data. Development of such software is in progress at the Prototype International Data Centre (PIDC) for eventual deployment at the International Data Centre (IDC). Flowcharts detailing the automatic analysis algorithm for β-γ coincidence data to be coded at the PIDC is included. The program is being written in C with Oracle databasing capabilities. (author)
Energy Technology Data Exchange (ETDEWEB)
Killian, E.W. [Idaho National Engineering Lab., Idaho Falls, ID (United States)
1997-11-01
A technique has been developed at the Idaho National Engineering Laboratory to sum high resolution gamma-ray pulse spectra from systems with multiple Ge detectors. Lockheed Martin Idaho Technologies Company operates a multi-detector spectrometer configuration at the Stored Waste Examination Pilot Plant facility which is used to characterize the radionuclide contents in waste drums destined for shipment to Waste Isolation Pilot Plant. This summing technique was developed to increase the sensitivity of the system, reduce the count times required to properly quantify the radio-nuclides and provide a more consistent methodology for combining data collected from multiple detectors. In spectrometer systems with multiple detectors looking at non homogeneous waste forms it is often difficult to combine individual spectrum analysis results from each detector to obtain a meaningful result for the total waste container. This is particularly true when the counting statistics in each individual spectrum are poor. The spectrum summing technique adds the spectra collected by each detector into a single spectrum which has better counting statistics than each individual spectrum. A normal spectral analysis program can then be used to analyze the sum spectrum to obtain radio-nuclide values which have smaller errors and do not have to be further manipulated to obtain results for the total waste container. 2 refs., 2 figs.
Improving detector signal processing with pulse height analysis in Moessbauer spectrometers
International Nuclear Information System (INIS)
Pechousek, Jiri; Mashlan, Miroslav; Frydrych, Jiri; Jancik, Dalibor; Prochazka, Roman
2007-01-01
A plenty of different programming techniques and instrument solutions are used in the development of Moessbauer spectrometers. Each of them should provide a faster spectrum accumulation process, increased productivity of measurements, decreased nonlinearity of the velocity scale, etc. The well known virtual instrumentation programming method has been used to design a computer-based Moessbauer spectrometer. Hardware solution was based on two commercially-available PCI modules produced by National Instruments Co. Virtual Moessbauer spectrometer is implemented by the graphical programming language LabVIEW 7 Express. This design environment allows to emulate the multichannel analyzer on the digital oscilloscope platform. This is a novel method based on Waveform Peak Detection function which allows detailed analysis of the acquired signal. The optimal treatment of the detector signal from various detector types is achieved by mathematical processing only. As a result, the possibility of an increase of signal/noise ratio is presented.
Cubukcu, Solen; Yücel, Haluk
2016-12-01
In this study, paraffin was selected as a base material and mixed with different amounts of CaSO 4 ·2H 2 O and H 3 BO 3 compounds in order to mimic breast tissue. Slab phantoms were produced with suitable mixture ratios of the additives in the melted paraffin. Subsequently, these were characterized in terms of first half-value layer (HVL) in the mammographic X-ray range using a pulse-height spectroscopic analysis with a CdTe detector. Irradiations were performed in the energy range of 23-35 kV p under broad beam conditions from Mo/Mo and Mo/Rh target/filter combinations. X-ray spectra were acquired with a CdTe detector without and with phantom material interposition in increments of 1 cm thickness and then evaluated to obtain the transmission data. The net integral areas of the spectra for the slabs were used to plot the transmission curves and these curves were fitted to the Archer model function. The results obtained for the slabs were compared with those of standard mammographic phantoms such as CIRS BR series phantoms and polymethylmethacrylate plates (PMMA). From the evaluated transmission curves, the mass attenuation coefficients and HVLs of some mixtures are close to those of the commercially available standard mammography phantoms. Results indicated that when a suitable proportion of H 3 BO 3 and CaSO 4 ·2H 2 O is added to the paraffin, the resulting material may be a good candidate for a breast tissue equivalent phantom.
International Nuclear Information System (INIS)
Guzik, Z.; Borsuk, S.; Plominski, M.; Traczyk, K.
2005-01-01
We present autonomous a 8K-channel miniature device designed for spectroscopy or intensity vs. time measurements. The device (TUKAN-8K-USB) is based on the USB interface, and is contained in a screened separate box - it can be proved either directly from the USB port or from an external DC source (wall adapter of battery). The device may work in two independent operational modes: Multi-Channel Analysis (MCA) and Multi-Channel Scaling (MCS). The crucial MCA component - Peak detect and Hold circuitry - is featuring a novel architecture based on a diamond transistor. Its analog stage can accept analog pulses with front edges down to 100 ns and has a differential linearity below 0.5% (full scale sliding scale averaging). Automatic stops on count in Region-Of-Interest (ROI) and on preset live or real time are implemented. The MCS works at medium speed counting rates (up to 8 MHz), with preset dwell time, number of channels and multi-sweep mode. Each these parameters can also be controlled externally. Digital interfacing is based on four used configurable logical I/O lines. A single CYCLONE EP1C3 Altera FPGA provides all control functions. The USB communication is based on FYDI FIFO controller. The analyzer is equipped with advanced, user-friendly software, which is subjected of another publication. )author)
Borak, Thomas B; Chapman, Phillip L
2014-10-01
Microdosimetric spectra of single event distributions have been used to provide estimates of quality factors for radiation protection of high-LET radiation. In situations with high-dose rates it becomes difficult to measure, record and store energy deposition from single events. An alternative approach is to store random energy deposition events in a sequence of fixed time intervals that does not require identifying from single events. This can be accomplished with a single detector without pulse height analysis. We show the development of the algorithm using expectation analysis of the statistical estimators for moments of lineal energy: ȳf and ȳD. The method was tested using Monte Carlo simulations based on single event distributions measured with spherical tissue equivalent proportional counters where the event sizes spanned more than two orders of magnitude. The evaluation included testing at various mean numbers of events per interval (i.e., dose rate) and numbers of intervals (i.e., total duration). Results of the expectation analysis and Monte Carlo simulation showed that the algorithm corrects for the excess dispersion due to the random number of events in each time interval when the underlying dose rate is constant. It also converges to the correct value when there is a linear trend in dose rate of the duration of the measurement process. Although this system is not applicable for pulsed radiation fields it proved to be robust when applied to measured distributions with single event spectra (PuBe neutrons, Fe ions at 1,000 MeV/nucleon and a power function distribution of single event sizes) with a coefficient of variation of 25% for estimates of ȳD using 100 sampling intervals and 10% using 400 sampling intervals.
Pulse height model for deuterated scintillation detectors
International Nuclear Information System (INIS)
Wang, Haitang; Enqvist, Andreas
2015-01-01
An analytical model of light pulse height distribution for finite deuterated scintillation detectors is created using the impulse approximation. Particularly, the energy distribution of a scattered neutron is calculated based on an existing collision probability scheme for general cylindrical shaped detectors considering double differential cross-sections. The light pulse height distribution is analytically and numerically calculated by convoluting collision sequences with the light output function for an EJ-315 detector from our measurements completed at Ohio University. The model provides a good description of collision histories capturing transferred neutron energy in deuterium-based scintillation materials. The resulting light pulse height distribution details pulse compositions and their corresponding contributions. It shows that probabilities of neutron collision with carbon and deuterium nuclei are comparable, however the light pulse amplitude due to collisions with carbon nuclei is small and mainly located at the lower region of the light pulse distribution axis. The model can explore those neutron interaction events that generate pulses near or below a threshold that would be imposed in measurements. A comparison is made between the light pulse height distributions given by the analytical model and measurements. It reveals a significant probability of a neutron generating a small light pulse due to collisions with carbon nuclei when compared to larger light pulse generated by collisions involving deuterium nuclei. This model is beneficial to understand responses of scintillation materials and pulse compositions, as well as nuclei information extraction from recorded pulses.
The use of logarithmic pulse height and energy scales in organic scintillator spectroscopy
International Nuclear Information System (INIS)
Whittlestone, S.
1980-01-01
The use of logarithmic pulse height and energy scales is advantageous for organic for organic scintillator neutron spectroscopy, providing an expanded dynamic range and economy of computer usage. An experimental logarithmic pulse height analysis system is shown to be feasible. A pulse height spectrum from a neutron measurement has been analysed using linear and logarithmic scales; the latter reduced the computer storage requirements by a factor of 13 and analysis time by 8.7, and there was no degradation of the analysed spectrum. Most of the arguments favouring use of logarithmic scales apply equally well to other types of scintillation spectroscopy. (orig.)
Development of a pulse height analizer
International Nuclear Information System (INIS)
Moreira, E.S.
1984-01-01
The development of a Pulse Height Analizer is described. This equipment is essential to analize data coming from detectors producing information codified in pulse amplitudes. The system developed consist of a Signal Input Module connected to a Controller Module based on a 8085A microprocessor capable to memorize pulses up to 1 uS in 256 channels with a resolution better than 20 mV. A Communication Module with a serial interface is used for data transfer to a host computer using RS232c protocol. The Monitoring and Operation Module consist of a hexadecimal Keybord, a 6 digit 7-segment display and a XY analog output enabling real time visualization of data on a XY monitor. The hardware and the software designed for this low cost system were optimized to obtain a typical dead time of approximately 100 uS. As application, this device was used to adquire curves at the Small Angle X-ray Scattering Laboratory in this Department. The apparatus performance was tested by comparing its data with a Northern Pulse Height Analizer model NS633 output, with favorable results. (Author) [pt
An Apple II -based bidimensional pulse height analyser
International Nuclear Information System (INIS)
Bateman, J.E.; Flesher, A.C.; Honeyman, R.N.; Pritchard, T.E.; Price, W.P.R.
1984-06-01
The implementation of a pulse height analyser function in an Apple II microcomputer using minimal purpose built hardware is described. Except for a small interface module the system consists of two suites of software, one giving a conventional one dimensional analysis on a span of 1024 channels, and the other a two dimensional analysis on a 128 x 128 image format. Using the recently introduced ACCELERATOR coprocessor card the system performs with a dead time per event of less than 50 μS. Full software facilities are provided for display, storage and processing of the data using standard Applesoft BASIC. (author)
Modeling Photo-multiplier Gain and Regenerating Pulse Height Data for Application Development
Aspinall, Michael D.; Jones, Ashley R.
2018-01-01
Systems that adopt organic scintillation detector arrays often require a calibration process prior to the intended measurement campaign to correct for significant performance variances between detectors within the array. These differences exist because of low tolerances associated with photo-multiplier tube technology and environmental influences. Differences in detector response can be corrected for by adjusting the supplied photo-multiplier tube voltage to control its gain and the effect that this has on the pulse height spectra from a gamma-only calibration source with a defined photo-peak. Automated methods that analyze these spectra and adjust the photo-multiplier tube bias accordingly are emerging for hardware that integrate acquisition electronics and high voltage control. However, development of such algorithms require access to the hardware, multiple detectors and calibration source for prolonged periods, all with associated constraints and risks. In this work, we report on a software function and related models developed to rescale and regenerate pulse height data acquired from a single scintillation detector. Such a function could be used to generate significant and varied pulse height data that can be used to integration-test algorithms that are capable of automatically response matching multiple detectors using pulse height spectra analysis. Furthermore, a function of this sort removes the dependence on multiple detectors, digital analyzers and calibration source. Results show a good match between the real and regenerated pulse height data. The function has also been used successfully to develop auto-calibration algorithms.
Development of a PDA Based Portable Pulse Height Analyzer System
International Nuclear Information System (INIS)
Mankheed, Panuphong; Ngernvijit, Narippawaj; Thong-Aram, Decho
2007-08-01
Full text: In this research a portable pulse height analyzer system was developed by application of a Personal Digital Assistant (PDAs) palm Tungsten T model together with Single Chip SCA developed by Department of Nuclear Technology, Chulalongkorn University to be used for education and research works. Capability of the developed system could measure both the energy and the average count rate of gamma rays. The results of this research showed that the gamma energy spectrum analysis of the developed system with a 2? x 2? NaI(Tl) detector could display photo peaks of Cs-137 and Co-60 at channel 57, channel 103, and channel 117 respectively. The energy resolution was found to be 7.14% at energy 661.66 keV of Cs-137
FERDO/FERD, Unfolding of Pulse-Height Spectrometer Spectra
International Nuclear Information System (INIS)
Rust, B.W.; Ingersoll, D.T.; Burrus, W.R.
1985-01-01
1 - Description of problem or function: FERDO and FERD are unfolding codes which can be used to correct observed pulse-height distributions for the non-ideal response of a pulse-height spectrometer or to solve poorly conditioned linear equations. 2 - Method of solution: It is assumed that the response of the spectrometer is given by Ax = b, where A is the spectrometer response function matrix, x is the unknown spectrum, and b is the pulse-height distribution. FERDO does not resolve directly for x but instead solves for p = Wx, where W is a 'window function matrix'. Typically, W is the resolution function of an ideal spectrometer which has a single Gaussian response. The effective resolution of the unfolding solution may be varied by the choice of W. Confidence intervals are found for each element of the solution p
Improved pulse-height store for A/D conversion
Energy Technology Data Exchange (ETDEWEB)
Casoli, P [Montedel S.p.a., Laben Division, Via Bassini 15, Milano, Italy; Maranesi, P [Politecnico di Milano (Italy). Centro Studi Nucleari E. Fermi
1979-11-15
A new pulse-height store is described. Suitable contrivances improve integral linearity and reduce the differential errors that generally occur at signal amplitudes near the lower threshold. No degradations appear at high rates of input events. The electrical specifications of the pulse-height store are determined through a series of measurements described in the final part of the paper. In order to test the circuit in the most significant way, it has been connected to a fast successive-approximation conversion module which uses the sliding-scale technique for channel width equalisation, thus implementing a complete analog-to-digital converter (ADC) for nuclear spectrometry. The performances of the pulse-height store have been deduced from the behavior of the whole system.
Output pulse height distribution of the GM counters
International Nuclear Information System (INIS)
Zhang Songshou; Xiong Jianping
1995-01-01
The GM counters are the radiation detectors most in use. It has special advantages compared with other detectors. This paper introduces the output pulse height distribution of the GM counters, gives the measuring instruments and methods. The measuring results, some discussions, and useful conclusion are given as well
A Fast Time-to-Pulse Height Converter
Energy Technology Data Exchange (ETDEWEB)
Aspelund, O
1962-12-15
A fast time-to-pulse height converter representing a development of Green and Bell's gated beam converter is described. The converter is compatible with 2 input pulses in the stop channel and exhibits excellent linearity and time resolution properties. High stability and large output pulses are obtained by using a large time constant in the converting network.
Digital baseline estimation method for multi-channel pulse height analyzing
International Nuclear Information System (INIS)
Xiao Wuyun; Wei Yixiang; Ai Xianyun
2005-01-01
The basic features of digital baseline estimation for multi-channel pulse height analysis are introduced. The weight-function of minimum-noise baseline filter is deduced with functional variational calculus. The frequency response of this filter is also deduced with Fourier transformation, and the influence of parameters on amplitude frequency response characteristics is discussed. With MATLAB software, the noise voltage signal from the charge sensitive preamplifier is simulated, and the processing effect of minimum-noise digital baseline estimation is verified. According to the results of this research, digital baseline estimation method can estimate baseline optimally, and it is very suitable to be used in digital multi-channel pulse height analysis. (authors)
International Nuclear Information System (INIS)
Tardocchi, M.; Pietropaolo, A.; Andreani, C.; Bracco, A.; D'Angelo, A.; Gorini, G.; Imberti, S.; Senesi, R.; Rhodes, N.J.; Schooneveld, E.M.
2004-01-01
The Resonance Detector Spectrometer was recently revised for neutron spectroscopic studies in the eV energy region. In this technique one makes use of a photon detector to record the gamma emission from analyser foils used as neutron-gamma converters. The pulse-height response of a Cadmium-Zinc-Telluride photon detector to neutron capture emission from 238 U and 197 Au analyser foils was characterised in the neutron energy range 1-200 eV. The experiment was performed on the VESUVIO spectrometer at the ISIS neutron-pulsed source. A biparametric data acquisition, specifically developed for these measurements, allowed the simultaneous measurements of both the neutron time of flight and γ pulse-height spectra. Through the analysis of the γ pulse-height spectra the main components of the signal associated with resonant and non-resonant neutron absorption were identified. It was also shown that, in principle, energy discrimination can be used to improve the signal to background ratio of the neutron time-of-flight measurement
Drift chamber and pulse height readout systems using analog multiplexing
International Nuclear Information System (INIS)
Cisneros, E.L; Kang, H.K.; Hall, J.N.; Larsen, R.S.
1976-11-01
Drift chamber and pulse-height readout systems are being developed for use in a new large scale detector at the SPEAR colliding beam facility. The systems are based upon 32 channels of sample-and-hold together with an analog multiplexer in a single-width CAMAC module. The modules within each crate are scanned by an autonomous controller containing a single ADC and memory plus arithmetic capability for offset, gain and linearity corrections. The drift chamber module has a facility for extracting hit wire information for use in trigger decision circuitry
Fast Computation of Pulse Height Spectra Using SGRD Code
Directory of Open Access Journals (Sweden)
Humbert Philippe
2017-01-01
Full Text Available SGRD (Spectroscopy, Gamma rays, Rapid, Deterministic code is used for fast calculation of the gamma ray spectrum produced by a spherical shielded source and measured by a detector. The photon source lines originate from the radioactive decay of the unstable isotopes. The emission rate and spectrum of these primary sources are calculated using the DARWIN code. The leakage spectrum is separated in two parts, the uncollided component is transported by ray-tracing and the scattered component is calculated using a multigroup discrete ordinates method. The pulsed height spectrum is then simulated by folding the leakage spectrum with the detector response functions which are pre-calculated using MCNP5 code for each considered detector type. An application to the simulation of the gamma spectrum produced by a natural uranium ball coated with plexiglass and measured using a NaI detector is presented.
Control of a pulse height analyzer using an RDX workstation
International Nuclear Information System (INIS)
Montelongo, S.; Hunt, D.N.
1984-12-01
The Nuclear Chemistry Division of Lawrence Livermore National laboratory is in the midst of upgrading its radiation counting facilities to automate data acquisition and quality control. This upgrade requires control of a pulse height analyzer (PHA) from an interactive LSI-11/23 workstation running RSX-11M. The PHA is a micro-computer based multichannel analyzer system providing data acquisition, storage, display, manipulation and input/output from up to four independent acquisition interfaces. Control of the analyzer includes reading and writing energy spectra, issuing commands, and servicing device interrupts. The analyzer communicates to the host system over a 9600-baud serial line using the Digital Data Communications link level Protocol (DDCMP). We relieved the RSX workstation CPU from the DDCMP overhead by implementing a DEC compatible in-house designed DMA serial line board (the ISL-11) to communicate with the analyzer. An RSX I/O device driver was written to complete the path between the analyzer and the RSX system by providing the link between the communication board and an application task. The I/O driver is written to handle several ISL-11 cards all operating in parallel thus providing support for control of multiple analyzers from a single workstation. The RSX device driver, its design and use by application code controlling the analyzer, and its operating environment will be discussed
Application of avalanche photodiode for soft X-ray pulse-height analyses in the Ht-7 tokamak
Shi Yue Jiang; Hu Li Qun; Sun Yan Jun; LiuSheng; Ling Bil
2002-01-01
An avalanche photodiode (APD) has been used as soft X-ray energy pulse-height analysis system for the measurement of the electron temperature on the HT-7 tokamak. The experimental results obtained with the APD with its inferior energy resolution show a little difference compared to the conventional high energy-resolution Si (Li) detector. Both numerical analysis and experimental results prove that the APD is good enough for application of the electron temperature measurement in tokamaks.
Pulse height measurements and electron attachment in drift chambers operated with Xe,CO2 mixtures
Andronic, A
2003-01-01
We present pulse height measurements in drift chambers operated with Xe,CO2 gas mixtures. We investigate the attachment of primary electrons on oxygen and SF6 contaminants in the detection gas. The measurements are compared with simulations of properties of drifting electrons. We present two methods to check the gas quality: gas chromatography and Fe55 pulse height measurements using monitor detectors.
Pulse height measurements and electron attachment in drift chambers operated with Xe,CO2 mixtures
International Nuclear Information System (INIS)
Andronic, A.; Appelshaeuser, H.; Blume, C.; Braun-Munzinger, P.; Bucher, D.; Busch, O.; Ramirez, A.C.A. Castillo; Catanescu, V.; Ciobanu, M.; Daues, H.; Devismes, A.; Emschermann, D.; Fateev, O.; Garabatos, C.; Herrmann, N.; Ivanov, M.; Mahmoud, T.; Peitzmann, T.; Petracek, V.; Petrovici, M.; Reygers, K.; Sann, H.; Santo, R.; Schicker, R.; Sedykh, S.; Shimansky, S.; Simon, R.S.; Smykov, L.; Soltveit, H.K.; Stachel, J.; Stelzer, H.; Tsiledakis, G.; Vulpescu, B.; Wessels, J.P.; Windelband, B.; Winkelmann, O.; Xu, C.; Zaudtke, O.; Zanevsky, Yu.; Yurevich, V.
2003-01-01
We present pulse height measurements in drift chambers operated with Xe,CO 2 gas mixtures. We investigate the attachment of primary electrons on oxygen and SF 6 contaminants in the detection gas. The measurements are compared with simulations of properties of drifting electrons. We present two methods to check the gas quality: gas chromatography and 55 Fe pulse height measurements using monitor detectors
Conversion of ADC pulse heights into MIP units (NUMASS test data, September 1991)
International Nuclear Information System (INIS)
Zhang, Q.P.
1993-04-01
ADC calibration is an important part of the data analysis in particle physics. Data analysis of ADC pulse heights is based on a physical unit which is called MIP (most probable energy loss of Minimum Ionization Particles). This paper describes the ADC calibration work for the September 1991 NUMASS test data, which were produced at CERN at the SPS facility. The Vavilov function is used to fit the energy loss distributions of minimum ionizing particles traversing the time of flight scintillators embedded in the uranium scintillator calorimeter. It is assumed that the energy loss distribution is the same for identical scintillators and the value at the peak of these distributions is used, i.e., the most probable energy loss, as a physical unit to derive the gains of the individual readout channels
Pulse-height defect in single-crystal CVD diamond detectors
Energy Technology Data Exchange (ETDEWEB)
Beliuskina, O.; Imai, N. [The University of Tokyo, Center for Nuclear Study, Wako, Saitama (Japan); Strekalovsky, A.O.; Aleksandrov, A.A.; Aleksandrova, I.A.; Ilich, S.; Kamanin, D.V.; Knyazheva, G.N.; Kuznetsova, E.A.; Mishinsky, G.V.; Pyatkov, Yu.V.; Strekalovsky, O.V.; Zhuchko, V.E. [JINR, Flerov Laboratory of Nuclear Reactions, Dubna, Moscow Region (Russian Federation); Devaraja, H.M. [Manipal University, Manipal Centre for Natural Sciences, Manipal, Karnataka (India); Heinz, C. [II. Physikalisches Institut, Justus-Liebig-Universitaet Giessen, Giessen (Germany); Heinz, S. [II. Physikalisches Institut, Justus-Liebig-Universitaet Giessen, Giessen (Germany); GSI Helmholtzzentrum fuer Schwerionenforschung, Darmstadt (Germany); Hofmann, S.; Kis, M.; Kozhuharov, C.; Maurer, J.; Traeger, M. [GSI Helmholtzzentrum fuer Schwerionenforschung, Darmstadt (Germany); Pomorski, M. [CEA, LIST, Diamond Sensor Laboratory, CEA/Saclay, Gif-sur-Yvette (France)
2017-02-15
The pulse-height versus deposited energy response of a single-crystal chemical vapor deposition (scCVD) diamond detector was measured for ions of Ti, Cu, Nb, Ag, Xe, Au, and of fission fragments of {sup 252} Cf at different energies. For the fission fragments, data were also measured at different electric field strengths of the detector. Heavy ions have a significant pulse-height defect in CVD diamond material, which increases with increasing energy of the ions. It also depends on the electrical field strength applied at the detector. The measured pulse-height defects were explained in the framework of recombination models. Calibration methods known from silicon detectors were modified and applied. A comparison with data for the pulse-height defect in silicon detectors was performed. (orig.)
Investigation of the Pulse Height Distribution of Boron Trifluoride Proportional Counters
Energy Technology Data Exchange (ETDEWEB)
Andersson, I Oe; Malmskog, S
1962-08-15
The report describes a theoretical and experimental investigation of the factors that determine the pulse height distribution of BF{sub 3}, proportional counters irradiated by thermal neutrons. The branching ratio of the {sup 10}B (n,{alpha}) {sup 7}Li reaction for thermal neutrons has been measured.
Time-correlated pulse-height measurements of low-multiplying nuclear materials
Energy Technology Data Exchange (ETDEWEB)
Miller, E.C., E-mail: ericcm@umich.edu [Department of Nuclear Engineering and Radiological Sciences, University of Michigan, Ann Arbor, MI (United States); Dolan, J.L.; Clarke, S.D.; Pozzi, S.A. [Department of Nuclear Engineering and Radiological Sciences, University of Michigan, Ann Arbor, MI (United States); Tomanin, A.; Peerani, P. [European Commission EC-JRC-IPSC, Ispra (Italy); Marleau, P. [Sandia National Laboratories, Livermore, CA (United States); Mattingly, J.K. [North Carolina State University, Raleigh, NC (United States)
2013-11-21
Methods for the determination of the subcritical neutron multiplication of nuclear materials are of interest in the field of nuclear nonproliferation and safeguards. A series of measurements were performed at the Joint Research Center facility in Ispra, Italy to investigate the possibility of using a time-correlated pulse-height (TCPH) analysis to estimate the sub-critical multiplication of nuclear material. The objective of the measurements was to evaluate the effectiveness of this technique, and to benchmark the simulation capabilities of MCNPX-PoliMi/MPPost. In this campaign, two low-multiplication samples were measured: a 1-kg mixed oxide (MOX) powder sample and several low-mass plutonium–gallium (PuGa) disks. The measured results demonstrated that the sensitivity of the TCPH technique could not clearly distinguish samples with very-low levels of multiplication. However, the simulated TCPH distributions agree well with the measured data, within 12% for all cases, validating the simulation capabilities of MCNPX-PoliMi/MPPost. To investigate the potential of the TCPH method for identifying high-multiplication samples, the validated MCNPX-PoliMi/MPPost codes were used to simulate sources of higher multiplications. Lastly, a characterization metric, the cumulative region integral (CRI), was introduced to estimate the level of multiplication in a source. However, this response was shown to be insensitive over the range of multiplications of interest. -- Highlights: •Present results of measurements of MOX fuel and PuGa disks. •Compared measurement results to simulations performed using MCNPX-Polimi and MPPost. •Investigated using correlated γ–n pairs to determine the multiplication of a system.
Optimal Linear Filters for Pulse Height Measurements in the Presence of Noise
International Nuclear Information System (INIS)
Nygaard, K.
1966-07-01
For measurements of nuclear pulse height spectra a linear filter is used between the pulse amplifier and the pulse height recorder so as to improve the signal/noise ratio. The problem of finding the optimal filter is investigated with emphasis on technical realizability. The maximum available signal/noise ratio is theoretically calculated on the basis of all the information which can be found in the output of the pulse amplifier, and on an assumed a priori knowledge of the pulse time of arrival. It is then shown that the maximum available signal/noise ratio can be obtained with practical measurements without any a priori knowledge of pulse time of arrival, and a general description of the optimal linear filter is given. The solution is unique, technically realizable, and based solely on data (noise power spectrum and pulse shape) which can be measured at the output terminals of the pulse amplifier used
Derivation of pulse height to exposure rate conversion functions for aerial radiological surveys
International Nuclear Information System (INIS)
Artuso, J.F.
1985-01-01
A method is described for deriving conversion functions that can be used to convert pulse height spectra taken at altitude to the exposure rate at the 1-m level. An integral equation is set up which involves the integration of a calculated pulse height spectrum multiplied by an unknown conversion function and then set equal to the exposure rate at ground level. This equation is then solved for the conversion function by assuming as a solution a three-term polynomial. Conversion functions have been derived for various source distributions, including surface, uniform, and exponentially distributed sources. These conversion functions are independent of source energy, which means that a conversion can be made without any knowledge of the isotopic content of the source. In the case of a uniform distribution, these conversion functions provide conversions that agree to within 10% with ground truth measurements
Measurement of H'(0.07) with pulse height weighting integration method
International Nuclear Information System (INIS)
Liye, LIU; Gang, JIN; Jizeng, MA
2002-01-01
H'(0.07) is an important quantity for radiation field measurement in health physics. One of the plastic scintillator measurement methods is employing the weak current produced by PMT. However, there are some weaknesses in the current method. For instance: sensitive to environment humidity and temperature, non-linearity energy response. In order to increase the precision of H'(0.07) measurement, a Pulse Height Weighting Integration Method is introduced for its advantages: low noise, high sensitivity, data processable, wide measurement range. Pulse Height Weighting Integration Method seems to be acceptable to measure directional dose equivalent. The representative theoretical energy response of the pre-described method accords with the preliminary experiment result
Single-electron pulse-height spectra in thin-gap parallel-plate chambers
Fonte, Paulo J R; Peskov, Vladimir; Policarpo, Armando
1999-01-01
Single-electron pulse-height spectra were measured in 0.6 and 1.2 mm parallel-plate chambers developed for the TOF system of the ALICE /LHC-HI experiment. Mixtures of Ar with ethane, isobutane, and SF/sub 6/ were studied. The observed spectrum shows a clear peak for all gases, suggesting efficient single-electron detection in thin parallel-plate structures. The pulse-height spectrum can be described by the weighted sum of an exponential and a Polya distribution, the Polya contribution becoming more important at higher gains. Additionally, it was found that the maximum gain, above 10/sup 6/, is limited by the appearance of streamers and depends weakly on the gas composition. The suitability of each mixture for single-electron detection is also quantitatively assessed. (8 refs).
Emission spectrochemical determination of boron in steels with pulse height distribution analyzer
International Nuclear Information System (INIS)
Ito, Minao; Sato, Shoki; Fushida, Hiroshi; Narita, Masanao
1983-01-01
The method for rapid determination of total, acid soluble and insoluble boron was established by using emission spectrochemical apparatus equipped with pulse height distribution analyzer. By using the analyzer, emission intensity can be expressed as different level intensity of pulse height distribution. It was made clear that soluble and insoluble boron had different contribution degree to each intensity and that this contribution degree varied at different pre-spark. Therefore, it is necessary for accurate determination of boron that this contribution degree should be corrected by using two intensities, of which contribution degrees are different. It was found on this two intensities method that total and soluble boron corresponded well to 50 % intensities at zero pre-spark and at 2000 pre-spark and that insoluble boron corresponded well to 70 % intensity at zero pre-spark and 50 % intensity at 2000 pre-spark. (author)
Optimal Linear Filters for Pulse Height Measurements in the Presence of Noise
Energy Technology Data Exchange (ETDEWEB)
Nygaard, K
1966-07-15
For measurements of nuclear pulse height spectra a linear filter is used between the pulse amplifier and the pulse height recorder so as to improve the signal/noise ratio. The problem of finding the optimal filter is investigated with emphasis on technical realizability. The maximum available signal/noise ratio is theoretically calculated on the basis of all the information which can be found in the output of the pulse amplifier, and on an assumed a priori knowledge of the pulse time of arrival. It is then shown that the maximum available signal/noise ratio can be obtained with practical measurements without any a priori knowledge of pulse time of arrival, and a general description of the optimal linear filter is given. The solution is unique, technically realizable, and based solely on data (noise power spectrum and pulse shape) which can be measured at the output terminals of the pulse amplifier used.
Pulse height non-linearity in LaBr3:Ce crystal for gamma ray spectrometry and imaging
International Nuclear Information System (INIS)
Pani, R.; Cinti, M.N.; Pellegrini, R.; Bennati, P.; Ridolfi, S.; Scafe, R.; Orsolini Cencelli, V.; De Notaristefani, F.; Fabbri, A.; Navarria, F.L.; Lanconelli, N.; Moschini, G.; Boccaccio, P.
2011-01-01
In this paper the response in term of pulse height linearity of two Hamamatsu photomultipliers is investigated, when coupled to a LaBr 3 :Ce scintillation crystal. The two photodetectors have high quantum efficiency and in particular 30% for R6231-01 and 42% for R7600-200 tube. The substantial difference is in the dynode structure, linear focused and metal channel for R6231 and R7600 respectively. In this work in order to verify the non-linearity effects on the pulse height distribution, due principally to the high and fast light production of LaBr 3 :Ce scintillator, we propose a 'peak by peak' procedure to calibrate the pulse height distribution. Utilizing a specific fragmentation of the calibration curve in subsets, the calculated energy values are very similar for both PMTs. This result confirmed the potentiality of the procedure to highlight the non-linearity effects on pulse height distribution.
Energy Technology Data Exchange (ETDEWEB)
Andronic, A. E-mail: a.andronic@gsi.de; Appelshaeuser, H.; Blume, C.; Braun-Munzinger, P.; Bucher, D.; Busch, O.; Ramirez, A.C.A. Castillo; Catanescu, V.; Ciobanu, M.; Daues, H.; Devismes, A.; Emschermann, D.; Fateev, O.; Garabatos, C.; Herrmann, N.; Ivanov, M.; Mahmoud, T.; Peitzmann, T.; Petracek, V.; Petrovici, M.; Reygers, K.; Sann, H.; Santo, R.; Schicker, R.; Sedykh, S.; Shimansky, S.; Simon, R.S.; Smykov, L.; Soltveit, H.K.; Stachel, J.; Stelzer, H.; Tsiledakis, G.; Vulpescu, B.; Wessels, J.P.; Windelband, B.; Winkelmann, O.; Xu, C.; Zaudtke, O.; Zanevsky, Yu.; Yurevich, V
2003-02-11
We present pulse height measurements in drift chambers operated with Xe,CO{sub 2} gas mixtures. We investigate the attachment of primary electrons on oxygen and SF{sub 6} contaminants in the detection gas. The measurements are compared with simulations of properties of drifting electrons. We present two methods to check the gas quality: gas chromatography and {sup 55}Fe pulse height measurements using monitor detectors.
Computer model for calculating gamma-ray pulse-height spectra for logging applications
International Nuclear Information System (INIS)
Evans, M.L.
1981-01-01
A generalized computer model has been devised to simulate the emission, transport, and detection of natural gamma radiation from various logging environments. The model yields high-resolution gamma-ray pulse-height spectra that can be used to correct both gross gamma and spectral gamma-ray logs. The technique can help provide corrections to airborne and surface radiometric survey logs for the effects of varying altitude, formation composition, and overburden. Applied to borehole logging, the model can yield estimates of the effects of varying borehole fluid and casing attenuations, as well as varying formation porosity and saturation
Applying a CPLD for Refurbishment of a Multi-channel Pulse Height Analyzer
International Nuclear Information System (INIS)
Leetragunpichitchai, Supalerk; Thong-Aram, Decho; Ploykrachang, Kamontip
2007-08-01
Full text: This research applied a CPLD for construction of a 100 MHz, 2048 channel, Wilkinson type analog to digital converter (ADC) circuits for refurbishment of an original multi-channel pulse height analyzer (PHA) ADC. Introduction of the CPLD could reduce the complexity of the circuits, equipment size and also the power consumption while the operation speed was increased. The linearity test of the ADC was found to be excellent with an R2 = 0.9995 and a maximum pulse rate of 48.828 k cps could be converted in this system. Therefore the developed system was appropriate for replacing the original ADC
Catalogue of response spectra for unfolding in situ gamma-ray pulse-height distributions
International Nuclear Information System (INIS)
Dymke, N.
1982-01-01
To unfold in situ gamma-ray pulse-height distributions by means of a response matrix technique, the matrix must be in keeping with the measurement geometry, detector size, and energy range to be covered by the measurements. A methodology has been described for determination of standard gamma-ray spectra needed in deriving response matrices and a spectrum catalogue compiled containing graphs and data for the 0-3 MeV (4 x 4 in. NaI(Tl)) and 0-8 MeV (1.5 x 1.5 in. NaI(Tl)) ranges. (author)
Baseline distortion effect on gamma-ray pulse-height spectra in neutron capture experiments
International Nuclear Information System (INIS)
Laptev, A.; Harada, H.; Nakamura, S.; Hori, J.; Igashira, M.; Ohsaki, T.; Ohgama, K.
2005-01-01
A baseline distortion effect due to gamma-flash at neutron time-of-flight measurement using a pulse neutron source has been investigated. Pulses from C 6 D 6 detectors accumulated by flash-ADC were processed with both standard analog-to-digital converter (ADC) and flash-ADC operational modes. A correction factor of gamma-ray yields, due to baseline shift, was quantitatively obtained by comparing the pulse height spectra of the two data-taking modes. The magnitude of the correction factor depends on the time after gamma-flash and has complex time dependence with a changing sign
The development of 128 ch fast multi channel pulse height analyzer for a tokamak plasmas
International Nuclear Information System (INIS)
Kawashima, Hisato; Matoba, Tohru; Ogawa, Toshihide; Kawakami, Tomohide
1985-02-01
A high counting rate multi channel pulse height analyzer was developed and tested to measure the detailed time evolution of X-ray energy spectrun radiated from a tokamak plasmas. Main developing objects of this analyzer are as follows. 1. The maximum counting rate and the minimum time resolution are 4 Mcps and 10 ms, respectively. 2. The energy resolution has ability to distinguish the characterisitic X-ray line. 3. Computer has to be used for operating system. This fast multi channel analyzer is using to measure the Soft X-ray spectrum on JFT-2M tokamak, and is confirmed to be useful for a practical measuring system. (author)
High-speed nuclear quality pulse height analyzer for synchrotron-based applications
International Nuclear Information System (INIS)
Beche, Jean-Francois; Bucher, Jerome J.; Fabris, Lorenzo; Riot, Vincent J.
2001-01-01
A high throughput Pulse Height Analyzer system for synchrotron-based applications requiring high resolution, high processing speed and low dead time has been developed. The system is comprised of a 120ns 12-bit nuclear quality Analog to Digital converter with a self-adaptive fast peak detector-stretcher and a custom-made fast histogramming memory module that records and processes the digitized data. The histogramming module is packaged in a VME or VXI compatible interface. Data is transferred through a fast optical link from the memory interface to a computer. A dedicated data acquisition program matches the hardware characteristics of the histogramming memory module. The data acquisition system allows for two data collection modes: ''standard'' data acquisition mode where the data is accumulated and read in synchronization with an external trigger and ''live'' data acquisition mode where the system operates as a standard Pulse Height Analyzer. The acquisition, standard or live, can be performed on several channels simultaneously. A two-channel prototype has been demonstrated at the Stanford Synchrotron Radiation Laboratory accelerator in conjunction with an X-ray Fluorescence Absorption Spectroscopy experiment. A detailed description of the entire system is given and experimental data is shown
Pulse-height response of silicon surface-barrier detectors to high-energy heavy ions
International Nuclear Information System (INIS)
Smith, G.D.
1973-01-01
The pulse-height defect (PHD) of high-energy heavy ions in silicon surface-barrier detectors can be divided into three components: (1) energy loss in the gold-surface layer, (2) a nuclear-stopping defect, and (3) a defect due to recombination of electron-hole pairs in the plasma created by the heavy ion. The plasma recombination portion of the PHD was the subject of this study using the variation of the PHD with (1) the angle of incidence of incoming heavy ions, and (2) changes in the detector bias. The Tandem Van de Graaff accelerator at Argonne National Laboratory was used to produce scattered beam ions ( 32 S, 35 Cl) and heavy target recoils (Ni, Cu, 98 Mo, Ag, Au) at sufficient energies to produce a significant recombination defect. The results confirm the existence of a recombination zone at the front surface of these detectors and the significance of plasma recombination as a portion of the pulse-height defect. (Diss. Abstr. Int., B)
Energy Technology Data Exchange (ETDEWEB)
Costrell, L; Brueckmann, R E [National Bureau of Standards, Washington, DC (United States)
1962-04-15
The low-duty cycles of many pulsed accelerators make high pulse rates necessary within the bursts in order to accumulate adequate data in a reasonable time. Not only do economic factors, as influenced by expensive machine time, dictate the use of high pulse rates, but purely technical considerations often make experiments unfeasible unless the pulses per burst are so numerous as to exclude the use of conventional pulse-height analysers. For these reasons much effort has been devoted to the development of high-speed pulse-height analysers for use with pulsed accelerators. Much of this work has been directed toward producing what we term a ''charge-storage analyser'' based on work conducted at the National Bureau of Standards. We have developed a charge-storage analyser that operates with the NBS 180-MeV synchrotron on a nuclear-absorption experiment for which the obstacles would otherwise be formidable. The analyser uses temporary electrostatic storage for the accumulation of pulse-height data during the machine bursts. During the dead intervals between bursts the contents of the temporary storage are analysed and transferred into a conventional magnetic core memory. The use of this technique for nanosecond pulses is discussed and data is presented to show its feasibility. (author) [French] Comme la partie du cycle de travail effectivement utilisee dans un grand nombre d'accelerateurs nucleaires puises est tres faible, il faut employer des frequences de repetition elevees pendant les bouffees, si l'on veut accumuler suffisamment de resultats en un temps raisonnable. Non seulement les facteurs economiques - influences par l'importance du temps de fonctionnement d'une machine couteuse - imposent des frequences de repetition elevees mais les experiences sont bien souvent irrealisables pour des raisons purement techniques, a moins que le nombre d'impulsions utilisees ne permettent de se passer d'analyseurs classiques des amplitudes d'impulsion. C'est pourquoi on a
Developments of quad channel pulse height analyzer for radon/thoron measurement
International Nuclear Information System (INIS)
Ashokkumar, P.; Raman, Anand; Babu, D.A.R.; Sharma, D.N.; Topkar, Anita; Mayya, Y.S.
2011-01-01
Radon and thoron are naturally occurring noble radioactive gases, the exposure to which has a linear relationship to lung cancer risk. This paper describes development of an automated Radon/Thoron measurement system using an indigenously developed silicon PIN diode. This system employs the 8051 core architecture based Si-lab microcontroller (C-8051F340) integrated with LCD display, hex key pad, non volatile flash memory besides I/O ports interfaced with humidity-temperature sensors and air sampling pump. Air is sampled through a dehumidifier by using a software controlled dc pump. The positively charged progeny atoms are electro statically collected over the detector surface and the deposited radioactivity is assessed by alpha pulse height discrimination technique. The ionization charges produced due to the interaction of alpha particles in the charge depletion region of the diode which is reverse biased at 40V are collected and measured. The measurement circuit uses a charge sensitive preamplifier developed around a low noise opamp IC. The pulses are further processed through a spectroscopy amplifier to obtain distinct pulse height levels for four of the alpha emitting progenies of Rn and Tn namely 210 Po, 214 Po, 216 Po and 212 Po. These signals are input to the quad channel analyzer which provides four individual TTL pulses corresponding to four nuclides mentioned above. The analyzer outputs are processed by the microcontroller module to obtain the Rn/Tn concentration in Bq/M 3 . This portable system stores one week hourly individual channel data along with the corresponding Rn/Tn concentrations, temperature, humidity and can be transferred to pc. Preliminary studies have indicated that sensitivity as low as 0.50 cph/Bq.m -3 can be achieved by this system. (author)
First result from x-ray pulse height analyzer with radial scanning system for LHD
Muto, Sadatsugu; Morita, Shigeru
2001-01-01
Radial profiles of x-ray spectrum have been successfully obtained using an assembly of x-ray pulse height analyzer in large helical device. The observed profile is obtained from plasma heated by ICRF and neutral beam injection (NBI). As a detector, Si(Li) semiconductor is used with a histogramming memory and analog-to-digital converter (ADC) basically working at high counting rate up to 500 kcps. In routine operation a count rate of 62 kcps has been normally obtained with energy resolution better than 400 eV at iron Kα line. The assembly is equipped with four detectors and a radial scanning system which modulates sight lines of the detectors in major radius direction. The profiles of electron temperature and the intensity of metallic impurities have been obtained with a spatial resolution of a few centimeters. Measured electron temperature is in good agreement with that from Thomson scattering. The system is applicable to steady-state discharge. The design philosophy of the assembly and recent results on the performance tests are also presented.
X-ray imaging with amorphous selenium: Pulse height measurements of avalanche gain fluctuations
International Nuclear Information System (INIS)
Lui, Brian J. M.; Hunt, D. C.; Reznik, A.; Tanioka, K.; Rowlands, J. A.
2006-01-01
Avalanche multiplication in amorphous selenium (a-Se) can provide a large, adjustable gain for active matrix flat panel imagers (AMFPI), enabling quantum noise limited x-ray imaging during both radiography and fluoroscopy. In the case of direct conversion AMFPI, the multiplication factor for each x ray is a function of its depth of interaction, and the resulting variations in gain can reduce the detective quantum efficiency (DQE) of the system. An experimental method was developed to measure gain fluctuations by analyzing images of individual x rays that were obtained using a video camera with an a-Se target operated in avalanche mode. Pulse height spectra (PHS) of the charge produced per x ray were recorded for monoenergetic 30.9, 49.4, and 73.8 keV x-ray sources. The rapid initial decay and long tail of each PHS can be explained by a model in which positive charge dominates the initiation of avalanche. The Swank information factor quantifies the effect of gain fluctuation on DQE and was calculated from the PHS. The information factor was found to be 0.5 for a 25 μm a-Se layer with a maximum gain of ∼300. Changing the energy of the incident x ray influenced the range of the primary photoelectron and noticeably affected the tail of the experimental PHS, but did not significantly change the avalanche Swank factor
The unfolding effects of transfer functions and processing of the pulse height distributions
Directory of Open Access Journals (Sweden)
Avdić Senada
2010-01-01
Full Text Available This paper deals with the improvements of the linear artificial neural network unfolding approach aimed at accurately determining the incident neutron spectrum. The effects of the transfer functions and pre-processing of the simulated pulse height distributions from liquid scintillation detectors on the artificial neural networks performance have been studied. A better energy resolution and higher reliability of the linear artificial neural network technique have been achieved after implementation of the results of this study. The optimized structure of the network was used to unfold both monoenergetic and continuous neutron energy spectra, such as the spectra of 252Cf and 241Am-Be sources, traditionally used in the nuclear safeguards experiments. We have demonstrated that the artificial neural network energy resolution of 0.1 MeV is comparable with the one obtained by the reference maximum likelihood expectation-maximization method which was implemented by using the one step late algorithm. Although the maximum likelihood algorithm provides the unfolded results of higher accuracy, especially for continuous neutron sources, the artificial neural network approach with the improved performances is more suitable for fast and robust determination of the neutron spectra with sufficient accuracy.
Comparison of performance of analog-to-digital converters (ADC) for pulse height analyzers
International Nuclear Information System (INIS)
Suzuki, Shogo; Hirai, Shoji
1981-01-01
In the recent ADCs for pulse height analyzers (PHA), Wilkinson type is a leading one, and those with 50 to 200 MHz clock frequency and 8K channel are mostly on the market. The comparison of the performance of ADCs was conducted under the condition of using these multi-channel PHAs as γ-ray spectrometers. The following three ADCs were compared: ADC incorporated in CANBERRA 8100 MCA, and CANBERRA 8060 ADCs used as the ADCs for 1st and 2nd GAMA systems. In this case, the conversion gains of these ADCs were set at 4K because the memory of 8100 MCA was 4K, and the GAMA systems were also mostly operated at 4K. In this paper, report is made on the measurements of dead time, the change of γ peaks immediately after the power was turned on, the temperature dependence of ADCs the long term stability of ADCs, derivative non-linearity, and integration non-linearity. The dead time of the ADC of 8100 CMA was shorter in the range up to 1000 channels. The temperature dependence was -0.017%/deg C in 8100 MCA, and -0.061 and -0.072%/deg C in GAMA. The long term stability was -0.02%/14 hr in 8100 MCA, and -0.027%/14 hr in 8060 of 1st GAMA. The derivative non-linearity was 0.45% in 8100 MCA, 0.67% in 8060 of 1st GAMA, and the integration non-linearity was 0.075% for 8100 MCA, and 0.085 - 0.1% in 8060 ADCs. Also, the change of γ peaks immediately after the power was turned on for 8100 ADC was better than that of 8060 ADC. (Wakatsuki, Y.)
The development of a digital multi-channel pulse height analysis
International Nuclear Information System (INIS)
Huang Shanshan; Sang Ziru; Liang Futian; Chen Lian; Liang Hao; Jin Ge
2012-01-01
A kind of digital multi-channel analyzer which was developed for Nuclear and Particle Experiment of Undergraduate Student in university was introduced. The input signal is digitalized with a high speed ADC, and feed to a FPGA for seeking peak, recording, spectrometer processing and displaying automatically. The principle of digital searching peck was given in the paper. In order to improve the capability of anti-noise, a dual digital thresholds and dual buffer memories was designed. Aiming at the influence of nonlinear of ADC to the system, a revised algorithm was designed to ensure the channel width uniformity. Since the huge sampling and storaging data of MCA, a method of parallel work of double memory was used, just separated empty transmit data and stored count, reduced the dead time and data overflowing. The host computer can set up and monitor MCA real-time, read and write data from MCA by an USB interface. Contrasts commercial MCA, the MCA has a good result in function and price. (authors)
International Nuclear Information System (INIS)
Shaw, S.E.
1979-01-01
An outline for a simple pulse spectroscope which attaches to a standard laboratory CRO is presented. The peak amplitude voltage of each pulse from the linear amplifier of a single channel X-ray analyser is stored for the duration of one oscilloscope trace. For each amplifier pulse, input from the discriminator is tested and if these is coincidence of pulses the oscilloscope beam is blanked for approximately the first 2 cm of its traverse across the screen. Repetition of pulses forms a pulse height distribution with a rectangular dark area marking the position of the discriminator window. (author)
Energy Technology Data Exchange (ETDEWEB)
Nagaoka, Toshi; Saito, Kimiaki; Moriuchi, Shigeru
1988-09-01
Ground level cosmic ray pulse height spectrum of a 7.5 cm diameter spherical NaI(Tl) scintillation detector was evaluated through stripping operation based on a pulse height spectrum measured on the sea and high precision response functions of the detector for U-series, Th-series and /sup 40/K. The exposure rate calculated from the determined cosmic ray pulse height spectrum was 0.21 R/h eq., which agreed well with that obtained from another method. The shape of the pulse height spectrum showed similarity to that measured at the altitude of 16,000 ft, especially in the energy region of 0 to 3 MeV. The principle of the adopted method is rather plain, however, the reliability of the spectrum is satisfactory. As the pulse height spectrum is a common information to any analytical method for environmental gamma ray using NaI(Tl) scintillation detector, it is expected to be used for simple and precise separation of cosmic ray component involved in the enviromental pulse height spectrum.
International Nuclear Information System (INIS)
Moss, C.E.; Lucas, M.C.; Tisinger, E.W.; Hamm, M.E.
1984-01-01
Our system consists of a LeCroy 3500 data acquisition system with a built-in CAMAC crate and eight bismuth-germanate detectors 7.62 cm in diameter and 7.62 cm long. Gamma-ray pulse-height distributions are acquired simultaneously for up to eight positions. The system was very carefully calibrated and characterized from 0.1 to 8.3 MeV using gamma-ray spectra from a variety of radioactive sources. By fitting the pulse-height distributions from the sources with a function containing 17 parameters, we determined theoretical repsonse functions. We use these response functions to unfold the distributions to obtain flux spectra. A flux-to-dose-rate conversion curve based on the work of Dimbylow and Francis is then used to obtain dose rates. Direct use of measured spectra and flux-to-dose-rate curves to obtain dose rates avoids the errors that can arise from spectrum dependence in simple gamma-ray dosimeter instruments. We present some gamma-ray doses for the Little Boy assembly operated at low power. These results can be used to determine the exposures of the Hiroshima survivors and thus aid in the establishment of radation exposure limits for the nuclear industry
International Nuclear Information System (INIS)
Beutler, D.E.; Halbleib, J.A.; Knott, D.P.
1989-01-01
This paper reports pulse-height distributions in two different types of Ge detectors measured for a variety of medium-energy x-ray bremsstrahlung spectra. These measurements have been compared to predictions using the integrated tiger series (ITS) Monte Carlo electron/photon transport code. In general, the authors find excellent agreement between experiments and predictions using no free parameters. These results demonstrate that the ITS codes can predict the combined bremsstrahlung production and energy deposition with good precision (within measurement uncertainties). The one region of disagreement observed occurs for low-energy (<50 keV) photons using low-energy bremsstrahlung spectra. In this case the ITS codes appear to underestimate the produced and/or absorbed radiation by almost an order of magnitude
Shape of intrinsic alpha pulse height spectra in lanthanide halide scintillators
Wolszczak, W.; Dorenbos, P.
2017-06-01
Internal contamination with actinium-227 and its daughters is a serious drawback in low-background applications of lanthanide-based scintillators. In this work we showed the important role of nuclear γ de-excitations on the shape of the internal alpha spectrum measured in scintillators. We calculated with Bateman equations the activities of contamination isotopes and the time evolution of actinium-227 and its progenies. Next, we measured the intrinsic background spectra of LaBr3(Ce), LaBr3(Ce,Sr) and CeBr3 with a digital spectroscopy technique, and we analyzed them with a pulse shape discrimination method (PSD) and a time-amplitude analysis. Finally, we simulated the α background spectrum with Geant4 tool-kit, consequently taking into account complex α-γ-electron events, the α / β ratio dependence on the α energy, and the electron/γ nonproportionality. We found that α-γ mixed events have higher light yield than expected for alpha particles alone, which leads to overestimation of the α / β ratio when it is measured with internal 227Th and 223Ra isotopes. The time-amplitude analysis showed that the α peaks of 219Rn and 215Po in LaBr3(Ce) and LaBr3(Ce,Sr) are not symmetric. We compared the simulation results with the measured data and provided further evidence of the important role of mixed α-γ-electron events for understanding the shape of the internal α spectrum in scintillators.
International Nuclear Information System (INIS)
Sharma, S.L.; Anil Kumar, G.; Choudhury, R.K.
2006-01-01
Pulse-type ionization chambers are invariably operated in the electron-sensitive mode where the capacitive decay in the detector-circuit during the electron collection produces loss in the pulse-height. In order to understand and appreciate the effect of this capacitive decay on the detector response, we have carried out Monte Carlo simulations of the response of two-electrode parallel plate ionization chambers with and without the capacitive decay keeping shaping time so large that the ballistic deficit is negligibly small. These simulations have been carried out incorporating the physical processes, namely, emission of charged particles from a point radioactive source, the generation of charge carriers in the active volume, separation and acceleration of the charge carriers, transport of the charge carriers, induction of charges on the electrodes, pulse processing by preamplifier-amplifier network, etc. These simulations have shown that the concerned capacitive decay produces appreciable loss in the pulse-height, if the detector-circuit time constant is of the order of maximum electron collection time. We have also carried out measurements on the pulse-height loss due to the capacitive decay in the detector-circuit during the electron collection for a two-electrode parallel plate ionization chamber. The experimental data on the pulse-height loss match reasonably well with the theoretical predictions
International Nuclear Information System (INIS)
Yamagishi, Kojiro
1978-02-01
The 4096 channel Pulse Height Analyzer (PHA) assembled with CAMAC plug-in units has been developed in JAERI. The PHA consists of ADC unit, CRT-display unit, and CAMAC plug-in units, which are memory-controller, MCA-timer, 4K words RAM memory and CRT-driver. The system is on-line connected to FACOM U-200 Mini-Computer through CAMAC interface unit Crate-controller. The softwares for on-line data acquisition of the system have been developed. These are four utility programs written in FORTRAN and two program packages written in assembler language FASP which are CAMAC Program Package and Basic Input/Output Program Package. CAMAC Program Package has 18 subroutine programs for control of CAMAC plug-in units from FACOM U-200 Mini-Computer; and Basic Input/Output Program Package has 26 subroutine programs to input/output data to/from a typewriter, keyboard, cassette magnetic tape and open reel magnetic tape. These subroutine programs are all FORTRAN callable. The PHA with CAMAC system is first outlined, and then usage is described in detail of four utility programs, CAMAC Program Package and Basic Input/Output Program Package. (auth.)
International Nuclear Information System (INIS)
Lin, Huiyun; Chen, Defu; Wang, Min; Lin, Juqiang; Li, Buhong; Xie, Shusen
2011-01-01
Direct measurement of near-infrared (NIR) luminescence around 1270 nm is the golden standard of singlet oxygen ( 1 O 2 ) identification. In this study, the influence of pulse-height discrimination threshold on measurement accuracy of the 1 O 2 luminescence that is generated from the photoirradiation of meso-tetra (N-methyl-4-pyridyl) morphine tetra-tosylate (TMPyP) in aqueous solution was investigated by using our custom-developed detection system. Our results indicate that the discrimination threshold has a significant influence on the absolute 1 O 2 luminescence counts, and the optimal threshold for our detection system is found to be about − 41.2 mV for signal discrimination. After optimization, the derived triplet-state and 1 O 2 lifetimes of TMPyP in aqueous solution are found to be 1.73 ± 0.03 and 3.70 ± 0.04 µs, respectively, and the accuracy of measurement was further independently demonstrated using the laser flash photolysis technique
A novel time-to-pulse height converter for fast-neutron time-of-flight techniques
International Nuclear Information System (INIS)
Christiansen, J.
1962-01-01
An electronic time-to-pulse height converter is described which uses a multiplicative method instead of the usual one of adding overlapping pulses. This is achieved by a coincidence of a linear sawtooth and a sharply clipped needle-pulse. The sawtooth is fed to the grid of a beam-deflecting tube (E80T) and the needle-pulse is applied to the deflecting plates and opens the tube only during a time-interval of about 5.10 -9 s. The plate gets a charge proportional to the time-difference between the start of the sawtooth and the needle pulse. The plate-pulse is stretched and amplified and its height represents a measurement of the time-difference. With this method we got a time resolution of 2τ = 7 x 10 -12 s with artificial pulses, 2τ = 3 x 10 -10 s with Co 60 γ-coincidences by using NE 102 plastic crystals and 2τ = 1.4 x 10 -9 s with 511-keV γ-coincidences using NaI(Te) crystals. The method was also used with pulsed beam techniques. In this case we got from the pulsing RF an 8-Mc, sharply-peaked pulse-sequence, which was fed to the E80T plates. We had a time-resolution of 2τ = 1.1 x 10 -9 s with 4-MeV neutrons using plastic crystals 0.7 in long. Normally the region of linear response was 30 ns but it was possible to go up to 120 ns. (author) [fr
International Nuclear Information System (INIS)
Olko, P.; Moutarde, C.; Segur, P.
1995-01-01
The ultraminiature proportional counters, UMC, unique radiation detectors for monitoring high intensity therapy fields, designed by Kliauga and operated at Columbia University (USA), have yielded a number of pulse height distributions for photons, neutrons and ions at simulated diameters of 5-50 nm. Monte Carlo calculations of the gas gain in such a counter questioned the possibility of achieving proportionally at such low simulated diameters. The response of the UMC has now been modelled taking into account both fluctuations of energy deposited in the counter volume and its calculated gas gain. Energy deposition was calculated using the MOCA-14, MOCA-8 and TRION codes, whereby distributions of ionisations d(j) after irradiations with 137 Cs, 15 MeV neutrons and 7 MeV.amu -1 deuterons were obtained. Monte Carlo calculations of electron avalanches in UMC show that the size of the single-electron avalanche P(n) reaching the anode depends strongly on the location of the primary ionisation within the counter volume. Distributions of the size of electron avalanches for higher numbers of primary ionisations, P *j (n), were obtained by successive convolutions of P(n). Finally, the counter response was obtained by weighting P *j (n) over d(j) distributions. On comparing the measured and calculated spectra it was concluded that the previously proposed single-electron peak calibration method might not be valid for the UMC due to the excessive width and overlap of electron avalanche distributions. Better agreement between the measured and calculated spectra is found if broader electron avalanche distributions than those used in the present calculations, are assumed. (author)
International Nuclear Information System (INIS)
Currie, G.D.; Marshall, M.
1989-03-01
The feasibility of developing a computer program suitable for evaluating the pulse-height spectrum in a gamma-ray detector from a complex geometry source has been examined. A selection of relevant programs, Monte Carlo radiation transport codes, have been identified and their applicability to this study discussed. It is proposed that the computation be performed in two parts: the evaluation of the photon fluence at the detector using a photon transport code, and calculation of the pulse-height distribution from this spectrum using response functions determined with an electron-photon transport code. The two transport codes selected to perform this procedure are MCNP (Monte Carlo Neutron Photon code), and EGS4 (Electron Gamma Shower code). (Author)
Swanson, C.; Jandovitz, P.; Cohen, S. A.
2018-02-01
We measured Electron Energy Distribution Functions (EEDFs) from below 200 eV to over 8 keV and spanning five orders-of-magnitude in intensity, produced in a low-power, RF-heated, tandem mirror discharge in the PFRC-II apparatus. The EEDF was obtained from the x-ray energy distribution function (XEDF) using a novel Poisson-regularized spectrum inversion algorithm applied to pulse-height spectra that included both Bremsstrahlung and line emissions. The XEDF was measured using a specially calibrated Amptek Silicon Drift Detector (SDD) pulse-height system with 125 eV FWHM at 5.9 keV. The algorithm is found to out-perform current leading x-ray inversion algorithms when the error due to counting statistics is high.
International Nuclear Information System (INIS)
Diwan, P.K.; Sharma, V.; Shyam Kumar; Avasthi, D.K.
2005-01-01
The response of ion implanted silicon detector has been studied for heavy ions with Z= 6-28 in the energy range ∼ 0.25-2.5 MeV/u utilizing the 15UD Pelletron Accelerator facility at Nuclear Science Centre, New Delhi, India. The variation of pulse height in ion implanted silicon detector with projectile's atomic number and its energy have been investigated. It has been observed that pulse height-energy calibration for a given projectile is described well by a linear relationship indicating no pulse height defect with the variation in energy for specific Z projectile. Pulse height defect has been found to increase with increasing projectile atomic number. The mean slope of the collected charge signal versus projectile energy depends significantly on the atomic number of the projectile. (author)
A histogram memory plug-in board for IBM-PC based nuclear pulse height analysis applications
International Nuclear Information System (INIS)
Behere, Anita; Ghodgaonkar, M.D.
1989-01-01
The histogram memory PC plug-in board has 8K x 24 dual ported memory with access from PC as well as from on board data acquisition logic. The arbitration control logic monitors the memory access requests from both the sources and honours them on first come first served basis. The data acquisition logic takes only 840 ns. to perform Read-Modify-Write memory operation. The data acquisition logic incorporates ADC interface logic for connecting to a NIM ADC which is normally housed in a NIM system along with other required front-end processing modules. Two interval timers are provided on the board. One of them provides Live Time/Clock Time counting and the other generates a 200 ms interrupt which is used for live spectrum display. The board is fully supported with system and data processing software developed in Turbo Pascal. (author)
Realization of a counter/timer circuit used in digital pulse height analysis in a single chip
International Nuclear Information System (INIS)
Mahmoud, I.I.
2000-01-01
This paper presents a single chip realization of a counter circuit, which is used in random signal processing and nuclear gamma ray spectrometers. The circuit contains a counter to count the repetition rate of a selected pulse train coming from a single channel analyzer circuit. Also, it contains a timer to measure the accumulation period. The timer possesses a predetermined time facility so that processing lasts for a certain adjustable predetermined period. The counter and the timer are synchronized to start and stop simultaneously at the beginning and end of the counting interval. A multiplexed BCD to 7-segment decoder/driver is also included in the circuit. The multiplexing allows the decrease of pin count of the chip.Two stages are designed, simulated for a single channel, however more stages and channels can be added by copying the designed circuits. Schematic flow of Xilinx v.1.2I is used as the design strategy with top-level schematic design containing VHDL and schematic macros
International Nuclear Information System (INIS)
Nagaoka, Toshi; Saito, Kimiaki; Moriuchi, Shigeru
1988-01-01
Ground level cosmic ray pulse height spectrum of a 7.5 cm diameter spherical NaI(Tl) scintillation detector was evaluated through stripping operation based on a pulse height spectrum measured on the sea and high precision response functions of the detector for U-series, Th-series and 40 K. The exposure rate calculated from the determined cosmic ray pulse height spectrum was 0.21 μR/h eq., which agreed well with that obtained from another method. The shape of the pulse height spectrum showed similarity to that measured at the altitude of 16,000 ft, especially in the energy region of 0 to 3 MeV. The principle of the adopted method is rather plain, however, the reliability of the spectrum is satisfactory. As the pulse height spectrum is a common information to any analytical method for environmental gamma ray using NaI(Tl) scintillation detector, it is expected to be used for simple and precise separation of cosmic ray component involved in the enviromental pulse height spectrum. (author)
Swanson, C.; Jandovitz, P.; Cohen, S. A.
2017-10-01
Knowledge of the full x-ray energy distribution function (XEDF) emitted from a plasma over a large dynamic range of energies can yield valuable insights about the electron energy distribution function (EEDF) of that plasma and the dynamic processes that create them. X-ray pulse height detectors such as Amptek's X-123 Fast SDD with Silicon Nitride window can detect x-rays in the range of 200eV to 100s of keV. However, extracting EEDF from this measurement requires precise knowledge of the detector's response function. This response function, including the energy scale calibration, the window transmission function, and the resolution function, can be measured directly. We describe measurements of this function from x-rays from a mono-energetic electron beam in a purpose-built gas-target x-ray tube. Large-Z effects such as line radiation, nuclear charge screening, and polarizational Bremsstrahlung are discussed.
TUKAN—An 8K Pulse Height Analyzer and Multi-Channel Scaler With a PCI or a USB Interface
Guzik, Z.; Borsuk, S.; Traczyk, K.; Plominski, M.
2006-02-01
In this paper we present two types of 8K-channel analyzers designed for spectroscopy and intensity versus time measurements. The first type (Tukan-8K-PCI) incorporates a PCI interface and is designed to be plugged into a PCI slot of a normal PC. The second type (Tukan-8K-USB) incorporates a USB interface. It is mounted in a separate screened box and can be powered either directly from the USB port or from an external dc source (wall adapter or battery). Each type of device may operate in either of two independent operational modes: Multi Channel Analysis (MCA) and Multi-Channel Scaling (MCS). The most crucial component for the MCA mode-the Peak Detect and Hold circuit-is featuring a novel architecture based on a diamond transistor. Its analog stage can accept analog pulses with rise times as short as 100 ns and has a differential linearity below 1% with sliding scale averaging over the full scale. The functionality includes automatic stop on a programmable count in the Region-Of-Interest (ROI) and on preset live- or real time. The MCS mode works at medium counting rates of up to 8 MHz. The dwell time, the number of channels and single or multi-sweep mode may be preset. Each of these parameters can also be controlled externally via four user configurable logical I/O lines. A single Altera FLEX 10KE30 FPGA provides all control functions and incorporates PCI interface. The USB interface is based on FTDI FIFO controller. Advanced and user-friendly software has been developed for the analyzer
Patterson, J. D.; Madanian, H.; Manweiler, J. W.; Lanzerotti, L. J.
2017-12-01
We present the compositional variation in the Solar Energetic Particle (SEP) population in the inner heliosphere over two solar cycles using data from the Ulysses Heliospheric Instrument for Spectra, Composition, and Anisotropy at Low Energies (HISCALE) and Advanced Composition Explorer (ACE) Electron Proton Alpha Monitor (EPAM). The Ulysses mission was active from late 1990 to mid-2009 in a heliopolar orbit inclined by 80° with a perihelion of 1.3 AU and an aphelion of 5.4 AU. The ACE mission has been active since its launch in late 1997 and is in a halo orbit about L1. These two missions provide a total of 27 years of continuous observation in the inner heliosphere with twelve years of simultaneous observation. HISCALE and EPAM data provide species-resolved differential flux and density of SEP between 0.5-5 MeV/nuc. Several ion species (He, C, O, Ne, Si, Fe) are identified using the Pulse Height Analyzer (PHA) system of the Composition Aperture for both instruments. The He density shows a noticeable increase at high solar activity followed by a moderate drop at the quiet time of the solar minimum between cycles 23 and 24. The density of heavier ions (i.e. O and Fe) change minimally with respect to the F10.7 index variations however, certain energy-specific count rates decrease during solar minimum. With Ulysses and ACE observing in different regions of the inner heliosphere, there are significant latitudinal differences in how the O/He ratios vary with the solar cycle. At solar minimum, there is reasonable agreement between the observations from both instruments. At solar max 23, the differences in composition over the course of the solar cycle, and as observed at different heliospheric locations can provide insight to the origins of and acceleration processes differentially affecting solar energetic ions.
Fang, Yuan; Badal, Andreu; Allec, Nicholas; Karim, Karim S; Badano, Aldo
2012-01-01
The authors describe a detailed Monte Carlo (MC) method for the coupled transport of ionizing particles and charge carriers in amorphous selenium (a-Se) semiconductor x-ray detectors, and model the effect of statistical variations on the detected signal. A detailed transport code was developed for modeling the signal formation process in semiconductor x-ray detectors. The charge transport routines include three-dimensional spatial and temporal models of electron-hole pair transport taking into account recombination and trapping. Many electron-hole pairs are created simultaneously in bursts from energy deposition events. Carrier transport processes include drift due to external field and Coulombic interactions, and diffusion due to Brownian motion. Pulse-height spectra (PHS) have been simulated with different transport conditions for a range of monoenergetic incident x-ray energies and mammography radiation beam qualities. Two methods for calculating Swank factors from simulated PHS are shown, one using the entire PHS distribution, and the other using the photopeak. The latter ignores contributions from Compton scattering and K-fluorescence. Comparisons differ by approximately 2% between experimental measurements and simulations. The a-Se x-ray detector PHS responses simulated in this work include three-dimensional spatial and temporal transport of electron-hole pairs. These PHS were used to calculate the Swank factor and compare it with experimental measurements. The Swank factor was shown to be a function of x-ray energy and applied electric field. Trapping and recombination models are all shown to affect the Swank factor.
International Nuclear Information System (INIS)
Borsuk, S.; Kulka, Z.
1989-12-01
A 12-bit spectroscopy analog-to-digital converter (ADC) type SAA (Successive Approximation type with channel width Averaging) intended for multichannel pulse height analyzer SWAN-1 based on IBM PC/XT/AT has been described. Design principles, specifications and measurements of a fundamental SAA-2 converter version are reported. Finally, two next versions of the converter with introduced modifications are discussed. 6 refs., 7 figs. (author)
Energy Technology Data Exchange (ETDEWEB)
Howansky, A; Peng, B; Lubinsky, A; Zhao, W [Stony Brook University, Stony Brook, NY (United States)
2016-06-15
Purpose: Pulse height spectra (PHS) have been used to determine the Swank factor of a scintillator by measuring fluctuations in its light output per x-ray interaction. The Swank factor and x-ray quantum efficiency of a scintillator define the upper limit to its imaging performance, i.e. DQE(0). The Swank factor below the K-edge is dominated by optical properties, i.e. variations in light escape efficiency from different depths of interaction, denoted e(z). These variations can be optimized to improve tradeoffs in x-ray absorption, light yield, and spatial resolution. This work develops a quantitative model for interpreting measured PHS, and estimating e(z) on an absolute scale. The method is used to investigate segmented ceramic GOS scintillators used in multi-slice CT detectors. Methods: PHS of a ceramic GOS plate (1 mm thickness) and segmented GOS array (1.4 mm thick) were measured at 46 keV. Signal and noise propagation through x-ray conversion gain, light escape, detection by a photomultiplier tube and dynode amplification were modeled using a cascade of stochastic gain stages. PHS were calculated with these expressions and compared to measurements. Light escape parameters were varied until modeled PHS agreed with measurements. The resulting estimates of e(z) were used to calculate PHS without measurement noise to determine the inherent Swank factor. Results: The variation in e(z) was 67.2–89.7% in the plate and 40.2–70.8% in the segmented sample, corresponding to conversion gains of 28.6–38.1 keV{sup −1} and 17.1–30.1 keV{sup −1}, respectively. The inherent Swank factors of the plate and segmented sample were 0.99 and 0.95, respectively. Conclusion: The high light escape efficiency in the ceramic GOS samples yields high Swank factors and DQE(0) in CT applications. The PHS model allows the intrinsic optical properties of scintillators to be deduced from PHS measurements, thus it provides new insights for evaluating the imaging performance of
International Nuclear Information System (INIS)
Fenyvesi, A.; Valastyan, I.; Olah, L.; Csikai, J.; Plompen, A.; Jaime, R.; Loevestam, G.; Semkova, V.
2011-01-01
Monoenergetic neutrons were produced at the Van de Graaff accelerator of the EC-JRC-Institute for Reference Materials and Measurements (IRMM, Geel, Belgium). An air-jet cooled D_2-gas target (1.2 bar, ΔE_d = 448 keV) was bombarded with E_d =4976 keV deuterons to produce neutrons up to E_n = 8 MeV energy via the D(d,n)"3He reaction. Higher energy neutrons up to E_n = 18 MeV were produced via the T(d,n)"4He reaction by bombarding a TiT target with E_d =1968 keV deuterons. Pulse height spectra were measured at different neutron energies from E_n = 8 MeV up to E_n = 18 MeV with the NE-213 liquid scintillator based Pulse Height Response Spectrometer (PHRS) of UD-IEP. The energy calibration of the PHRS system has been extended up to E_n = 18 MeV. Pulse height spectra induced by gamma photons have been simulated by the GRESP7 code. Neutron induced pulse height spectra have been simulated by the NRESP7 and MCNP-POLIMI codes. Comparison of the results of measurements and simulations enables the improvement of the parameter set of the function used by us to describe the light output dependence of the resolution of the PHRS system at light outputs of L > 2 light units. Also, it has been shown that the derivation method for unfolding neutron spectra from measured pulse height spectra performs well when relative measurements are done up to E_n = 18 MeV neutron energy. For matrix unfolding purposes, the NRESP7 code has to be preferred to calculate the pulse height response matrix of the PHRS system. Leakage spectra of neutrons behind bismuth slabs of different thicknesses have been measured with the PHRS system by using monoenergetic neutrons. The maximum slab thickness was d = 14 cm. Simulations of the measurements have been carried out with the MCNP-4c code. The necessary nuclear cross-sections were taken from the from the ENDF/B-VII and JEFF.3.1 data libraries. For both libraries, the agreement of measured and simulated neutron spectra is good for the 5 MeV ≤ En ≤ 18 Me
Minicomputer system for radiochemical analysis by coincidence spectrometry
International Nuclear Information System (INIS)
Brauer, F.P.; Fager, J.E.
1979-01-01
Minicomputer-based coincidence analysis methods have been developed for use in performing radiochemical analysis by high-resolution x- and gamma-ray coincidence spectrometry. This paper describes the data-acquisition and analysis methods develolped for qualitative and quantitative analyses of coincidence spectrometric data. Data-acquisition capabilities include both direct multiparameter pulse-height analysis and buffered list-mode acquisition
QUASAR - an interactive program for spectrum analysis in personal computers
International Nuclear Information System (INIS)
Auler, L.T.; Nobrega, J.A.W. da.
1991-11-01
The QUASAR software for the interactive analysis and report of energy (pulse-height) and time (multichannel scaling) spectra is described. The operating instructions as well as the mathematical methods and algorithms used by the program are presented in detail. This program is an extension to the PULSAR program. (author)
Neutron-gamma discrimination by pulse analysis with superheated drop detector
International Nuclear Information System (INIS)
Das, Mala; Seth, S.; Saha, S.; Bhattacharya, S.; Bhattacharjee, P.
2010-01-01
Superheated drop detector (SDD) consisting of drops of superheated liquid of halocarbon is irradiated to neutrons and gamma-rays from 252 Cf fission neutron source and 137 Cs gamma source, respectively, separately. Analysis of pulse height of signals at the neutron and gamma-ray sensitive temperature provides significant information on the identification of neutron and gamma-ray induced events.
Pulse height distribution and radiation tolerance of CVD diamond detectors
Energy Technology Data Exchange (ETDEWEB)
Adam, W.; Berdermann, E.; Bergonzo, P.; Bertuccio, G.; Bogani, F.; Borchi, E.; Brambilla, A.; Bruzzi, M.; Colledani, C.; Conway, J.; Dangelo, P.; Dabrowski, W.; Delpierre, P.; Deneuville, A.; Dulinski, W.; Eijk, B. van; Fallou, A.; Fizzotti, F.; Foulon, F.; Friedl, M.; Gan, K.K.; Gheeraert, E.; Grigoriev, E.; Hallewell, G.; Han, S.; Hartjes, F. E-mail: f.hartjes@nikhef.nl; Hrubec, J.; Husson, D.; Kagan, H.; Kania, D.; Kaplon, J.; Karl, C.; Kass, R.; Krammer, M.; Logiudice, A.; Lu, R.; Manfredotti, C.; Meier, D.; Mishina, M.; Moroni, L.; Oh, A.; Pan, L.S.; Pernicka, M.; Peitz, A.; Pirollo, S.; Polesello, P.; Procario, M.; Riester, J.L.; Roe, S.; Rousseau, L.; Rudge, A.; Russ, J.; Sala, S.; Sampietro, M.; Schnetzer, S.; Sciortino, S.; Stelzer, H.; Stone, R.; Suter, B.; Tapper, R.J.; Tesarek, R.; Trawick, M.; Trischuk, W.; D.Tromson,; Vittone, E.; Walsh, A.M.; Wedenig, R.; Weilhammer, P.; White, C.; Zeuner, W.; Zoeller, M.; Fenyvesi, A.; Molnar, J.; Sohler, D
2000-06-01
The paper reviews measurements of the radiation tolerance of CVD diamond for irradiation with 24 GeV/c protons, 300 MeV/c pions and 1 MeV neutrons. For proton and neutron irradiation, the measured charge signal spectrum is compared with the spectrum calculated by a model. Irradiation by particles causes radiation damage leading to a decrease of the charge signal. However, both the measurements and the outcome from the model shows that for tracker applications this drawback is at least partly counterbalanced by a narrowing of the distribution curve of the charge signal. In addition, we observed after proton irradiation at the charge signal spectrum a decrease of the number of small signals. As a result, the efficiency of a CVD diamond tracker is less affected by irradiation than the mean charge signal.
Improved stabilization scheme for computerized pulse-height analyzers
International Nuclear Information System (INIS)
Cohn, C.E.
1982-01-01
A stabilization scheme has been adapted from a 24-bit to a 16-bit computer. A precision pulse generator produces tagged reference pulses alternately of high and low amplitude. Gain and zero corrections are obtained from these and applied to each normal event. Provisions are made to make the system robust against pileup corruption of the reference pulses. (orig.)
Pulse height distribution and radiation tolerance of CVD diamond detectors
Adam, W; Bergonzo, P; Bertuccio, G; Bogani, F; Borchi, E; Brambilla, A; Bruzzi, Mara; Colledani, C; Conway, J; D'Angelo, P; Dabrowski, W; Delpierre, P A; Deneuville, A; Dulinski, W; van Eijk, B; Fallou, A; Fizzotti, F; Foulon, F; Friedl, M; Gan, K K; Gheeraert, E; Grigoriev, E; Hallewell, G D; Han, S; Hartjes, F G; Hrubec, Josef; Husson, D; Kagan, H; Kania, D R; Kaplon, J; Karl, C; Kass, R; Krammer, Manfred; Lo Giudice, A; Lü, R; Manfredotti, C; Meier, D; Mishina, M; Moroni, L; Oh, A; Pan, L S; Pernicka, Manfred; Peitz, A; Pirollo, S; Polesello, P; Procario, M; Riester, J L; Roe, S; Rousseau, L; Rudge, A; Russ, J; Sala, S; Sampietro, M; Schnetzer, S R; Sciortino, S; Stelzer, H; Stone, R; Suter, B; Tapper, R J; Tesarek, R J; Trawick, M L; Trischuk, W; Tromson, D; Vittone, E; Walsh, A M; Wedenig, R; Weilhammer, Peter; White, C; Zeuner, W; Zöller, M; Fenyvesi, A; Molnár, J; Sohler, D
2000-01-01
The paper reviews measurements of the radiation tolerance of CVD diamond for irradiation with 24 GeV/c protons, 300 MeV/c pions and 1 MeV neutrons. For proton and neutron irradiation, the measured charge signal spectrum is compared with the spectrum calculated by a model. Irradiation by particles causes radiation damage leading to a decrease of the charge signal. However, both the measurements and the outcome from the model shows that for tracker applications this drawback is at least partly counterbalanced by a narrowing of the distribution curve of the charge signal. In addition, we observed after proton irradiation at the charge signal spectrum a decrease of the number of small signals. As a result, the efficiency of a CVD diamond tracker is less affected by irradiation than the mean charge signal. (11 refs).
Pulse height distribution and radiation tolerance of CVD diamond detectors
International Nuclear Information System (INIS)
Adam, W.; Berdermann, E.; Bergonzo, P.; Bertuccio, G.; Bogani, F.; Borchi, E.; Brambilla, A.; Bruzzi, M.; Colledani, C.; Conway, J.; Dangelo, P.; Dabrowski, W.; Delpierre, P.; Deneuville, A.; Dulinski, W.; Eijk, B. van; Fallou, A.; Fizzotti, F.; Foulon, F.; Friedl, M.; Gan, K.K.; Gheeraert, E.; Grigoriev, E.; Hallewell, G.; Han, S.; Hartjes, F.; Hrubec, J.; Husson, D.; Kagan, H.; Kania, D.; Kaplon, J.; Karl, C.; Kass, R.; Krammer, M.; Logiudice, A.; Lu, R.; Manfredotti, C.; Meier, D.; Mishina, M.; Moroni, L.; Oh, A.; Pan, L.S.; Pernicka, M.; Peitz, A.; Pirollo, S.; Polesello, P.; Procario, M.; Riester, J.L.; Roe, S.; Rousseau, L.; Rudge, A.; Russ, J.; Sala, S.; Sampietro, M.; Schnetzer, S.; Sciortino, S.; Stelzer, H.; Stone, R.; Suter, B.; Tapper, R.J.; Tesarek, R.; Trawick, M.; Trischuk, W.; D.Tromson,; Vittone, E.; Walsh, A.M.; Wedenig, R.; Weilhammer, P.; White, C.; Zeuner, W.; Zoeller, M.; Fenyvesi, A.; Molnar, J.; Sohler, D.
2000-01-01
The paper reviews measurements of the radiation tolerance of CVD diamond for irradiation with 24 GeV/c protons, 300 MeV/c pions and 1 MeV neutrons. For proton and neutron irradiation, the measured charge signal spectrum is compared with the spectrum calculated by a model. Irradiation by particles causes radiation damage leading to a decrease of the charge signal. However, both the measurements and the outcome from the model shows that for tracker applications this drawback is at least partly counterbalanced by a narrowing of the distribution curve of the charge signal. In addition, we observed after proton irradiation at the charge signal spectrum a decrease of the number of small signals. As a result, the efficiency of a CVD diamond tracker is less affected by irradiation than the mean charge signal
The analysis of the gamma-ray pulseheight spectra resulting from the NaI detector
International Nuclear Information System (INIS)
Huang Zhengde; Zhang Guishan; Chen Qun; Cao Zhong
1990-01-01
The analysis of the Gamma-ray pulse-height spectra resulting from NaI detector is described by using weighted least square iteration. The computer program has the function of searching for Gamma-ray peak automatically. It can be used in the analysis of continuous, discrete or their superposition spectra. Besides, there are some function of the spectrum smooth,the correction of the shift in gain and zero energy channel intercept. Some results of the computer program are presented
An integrated acquisition, display, and analysis system
International Nuclear Information System (INIS)
Ahmad, T.; Huckins, R.J.
1987-01-01
The design goal of the ND9900/Genuie was to integrate a high performance data acquisition and display subsystem with a state-of-the-art 32-bit supermicrocomputer. This was achieved by integrating a Digital Equipment Corporation MicroVAX II CPU board with acquisition and display controllers via the Q-bus. The result is a tightly coupled processing and analysis system for Pulse Height Analysis and other applications. The system architecture supports distributed processing, so that acquisition and display functions are semi-autonomous, making the VAX concurrently available for applications programs
International Nuclear Information System (INIS)
Ruhter, W.D.
1977-12-01
A megachannel pulse-height analysis system using a 32,000-word PDP-8/E minicomputer and two moving-head disk memories was developed. The system has a storage capacity of 2 18 events at any of 2 20 data locations, is capable of processing 1,040 events/s, and provides on-line sorting and disk storage. An X- or Y- pulse-height spectrum in coincidence with one to four arithmetically combined pulse-height windows can be assembled in core for scope display and spectral analysis within 2 to 20 seconds. The software for the system was written extensively in machine language. Excellent energy and timing resolution were achieved. An energy resolution of 2.3 keV fwhm and a timing resolution of 8.5 ns fwhm were obtained for 60 Co at a singles rate of 10,000 counts/s. The prompt timing peak remained Gaussian down to the fwtm by gating the TAC with SCAs which discriminated against low-energy events. The coincidence electronics also allowed on-line subtraction of accidental coincidence events. Alignment of the detectors and tests of the system's performance were made by measuring solid-angle correction factors for the Ge(Li) detectors through the correlation measurements of the 0 + --2 + --0 + cascade in 106 Pd, and by measuring the angular spread of positron annihilation radiation coincidences. Results were in excellent agreement with theoretical solid-angle correction factors calculated for the detector and source sizes used. Directional correlation data were obtained for gamma-ray cascades in 134 Ba and 110 Cd. Analyses of these data gave E2/M1 mixing ratios which are in excellent agreement with results obtained by other investigators. The improved resolution and data processing capabilities of this system gave new results for the 563-keV transition in 134 Ba and the 1505-keV transition in 110 Cd. E2/M1 mixing ratios of 13.3/sup +2.3/sub -1.8/ for the 563-keV transition and -1.24 +- 0.20 for the 1505-keV transition were measured
Neutron radiative capture methods for surface elemental analysis
Trombka, J.I.; Senftle, F.; Schmadebeck, R.
1970-01-01
Both an accelerator and a 252Cf neutron source have been used to induce characteristic gamma radiation from extended soil samples. To demonstrate the method, measurements of the neutron-induced radiative capture and activation gamma rays have been made with both Ge(Li) and NaI(Tl) detectors, Because of the possible application to space flight geochemical analysis, it is believed that NaI(Tl) detectors must be used. Analytical procedures have been developed to obtain both qualitative and semiquantitative results from an interpretation of the measured NaI(Tl) pulse-height spectrum. Experiment results and the analytic procedure are presented. ?? 1970.
International Nuclear Information System (INIS)
Blunden, A.; O'Prey, D.G.; Tait, W.H.
1983-01-01
A method is described for the separation of a composite pulse-height spectrum into its unresolved component parts, which belong to a set of measured library spectra. The method allows real-time estimation giving running estimates during acquisition of the spectrum, minimises computation space, especially for a number of parallel calculations, estimates in advance the rms errors, and produces a significance measure for the hypothesis that the composite contains only the library spectra. Least squares curve-fitting, and other methods, can be compared, with the formalism developed, allowing analytical comparison of the effect of detector energy resolution and detection efficiency. A rational basis for the choice between the various methods of spectrum analysis follows from the theory, minimising rms estimation errors. The method described is applicable for very low numbers of counts and poor resolution. (orig.)
A low-cost multichannel pulse-height analyzer PHA 256 using single-chip microcomputer
International Nuclear Information System (INIS)
Koehler, M.; Meiling, W.
1985-01-01
The PHA 256 multichannel analyzer on the base of the U8820 single-chip microcomputer applied for radiation measurements, for example in monitoring systems with scintillation detectors, is described. The analyzer contains a power supply unit and 7 boards, namely, the processor board; data and program memory; 8-bit analog-to-digital converter; driver to display device; keyboard with 23 function keys; pulse amplifier and high-voltage supply (up to 2 kV). Software used provides preprocessing of spectra supported by following functions: addition and subtraction of different spectra, spectrum monitoring by use of a 5-point-algorithm, calculation of peak areas with linearly interpolated background
The conversion of a pulse height analyzer to a time analyzer
International Nuclear Information System (INIS)
Adib, M.; Salama, M.; Abdel-Kawy, A.; Faragalla, S.; Sayed, E.M.; Hamouda, I.
1975-01-01
A transistorized time unit circuit has been designed to convert a 400-channel amplitude analyser (RIDL model 34-12B) to a 400 channel time analyser. The obtained time analyser has channel widths of 24.4 μs, 48.8 μs, 97.6 μs. Experimental determination of the stability of the obtained time analyser was carried out by measuring statistically distributed pulses at constant rate. The stability of the time analyser for long time operation was found to be not less than 0.14%. The linearity of the time scale calibration was checked by measuring the transmission of neutrons through a polycrystalline powder of nickel. This method was found to give a more precise calibration of the time scale than the method based on using the transmission of the elements with well known resonances. From the experimental values obtained for Bragg's cut-off positions for nickel, the non linearity was calculated and found not to exceed 0.3%
Pulse-height loss in the signal readout circuit of compound semiconductor detectors
Nakhostin, M.; Hitomi, K.
2018-06-01
Compound semiconductor detectors such as CdTe, CdZnTe, HgI2 and TlBr are known to exhibit large variations in their charge collection times. This paper considers the effect of such variations on the measurement of induced charge pulses by using resistive feedback charge-sensitive preamplifiers. It is shown that, due to the finite decay-time constant of the preamplifiers, the capacitive decay during the signal readout leads to a variable deficit in the measurement of ballistic signals and a digital pulse processing method is employed to correct for it. The method is experimentally examined by using sampled pulses from a TlBr detector coupled to a charge-sensitive preamplifier with 150 μs of decay-time constant and 20 % improvement in the energy resolution of the detector at 662 keV is achieved. The implications of the capacitive decay on the correction of charge-trapping effect by using depth-sensing technique are also considered.
Energy Technology Data Exchange (ETDEWEB)
Christiansen, J [Physikalisches Staatsinstitut, Hamburg, Federal Republic of Germany (Germany)
1962-04-15
An electronic time-to-pulse height converter is described which uses a multiplicative method instead of the usual one of adding overlapping pulses. This is achieved by a coincidence of a linear sawtooth and a sharply clipped needle-pulse. The sawtooth is fed to the grid of a beam-deflecting tube (E80T) and the needle-pulse is applied to the deflecting plates and opens the tube only during a time-interval of about 5.10{sup -9} s. The plate gets a charge proportional to the time-difference between the start of the sawtooth and the needle pulse. The plate-pulse is stretched and amplified and its height represents a measurement of the time-difference. With this method we got a time resolution of 2{tau} = 7 x 10{sup -12}s with artificial pulses, 2{tau} = 3 x 10{sup -10}s with Co{sup 60} {gamma}-coincidences by using NE 102 plastic crystals and 2{tau} = 1.4 x 10{sup -9} s with 511-keV {gamma}-coincidences using NaI(Te) crystals. The method was also used with pulsed beam techniques. In this case we got from the pulsing RF an 8-Mc, sharply-peaked pulse-sequence, which was fed to the E80T plates. We had a time-resolution of 2{tau} = 1.1 x 10{sup -9} s with 4-MeV neutrons using plastic crystals 0.7 in long. Normally the region of linear response was 30 ns but it was possible to go up to 120 ns. (author) [French] L'auteur decrit un convertisseur electronique temps en amplitude d'impulsions fonctionnant d'apres une methode de multiplication au lieu de la methode habituelle qui consiste a ajouter des impulsions qui se chevauchent. On fait coincider une impulsion lineaire en dents de scie avec une impulsion en forme d'aiguille fortement ecretee. L'impulsion en dents de scie est envoyee sur la grille d'un tube a faisceaux electroniques diriges (E80T) ; et l'impulsion en forme d'aiguille est appliquee aux plaques de deflexion et n'ouvre le tube que pendant un intervalle de temps d'environ 5 {center_dot} 10{sup -9} s. La plaque recoit une charge proportionnelle a la difference de
Progress Towards Improved Analysis of TES X-ray Data Using Principal Component Analysis
Busch, S. E.; Adams, J. S.; Bandler, S. R.; Chervenak, J. A.; Eckart, M. E.; Finkbeiner, F. M.; Fixsen, D. J.; Kelley, R. L.; Kilbourne, C. A.; Lee, S.-J.;
2015-01-01
The traditional method of applying a digital optimal filter to measure X-ray pulses from transition-edge sensor (TES) devices does not achieve the best energy resolution when the signals have a highly non-linear response to energy, or the noise is non-stationary during the pulse. We present an implementation of a method to analyze X-ray data from TESs, which is based upon principal component analysis (PCA). Our method separates the X-ray signal pulse into orthogonal components that have the largest variance. We typically recover pulse height, arrival time, differences in pulse shape, and the variation of pulse height with detector temperature. These components can then be combined to form a representation of pulse energy. An added value of this method is that by reporting information on more descriptive parameters (as opposed to a single number representing energy), we generate a much more complete picture of the pulse received. Here we report on progress in developing this technique for future implementation on X-ray telescopes. We used an 55Fe source to characterize Mo/Au TESs. On the same dataset, the PCA method recovers a spectral resolution that is better by a factor of two than achievable with digital optimal filters.
Pulse pileup effects of plasma electron temperature measurements by soft x-ray energy analysis
International Nuclear Information System (INIS)
Dyer, G.R.; Neilson, G.H.; Kelley, G.G.
1978-10-01
The electron temperature of hot plasmas is conveniently derived from bremsstrahlung spectra obtained by pulse-height analysis using a lithium-compensated silicon detector. Time-resolved temperature measurements require high counting rates, with ultimate rate limited by pulse pileup. To evaluate this limit, spectral distortion due to pileup and consequent effects on temperature determination are investigated. Expressions for distorted spectra are derived as functions of Maxwellian temperature and pileup fraction for both square and triangular pulse shapes. A comparison of temperatures obtained from distorted spectra with actual values indicates that measurements with less than 10% error can be made in the absence of line radiation, even from spectra containing 40% pileup
Mercury determination in dentist's hair and nails by instrumental neutron activation analysis
International Nuclear Information System (INIS)
Mazzilli, B.; Munita, C.S.
1986-01-01
The mercury in scalp hair and fingernails of a group of dentists who usually handle with this toxic element in their profession is determined. The results were obtained by instrumental neutron activation analysis. The experimental procedure was based on the evaluation of the 197 Hg photopeak area, whose half life is 65h. After at a neutron fluxo of 5x10 12 n.cm -2 .sec -1 , the activity of the samples were measured by using a solid state Ge (Li) detector coupled to a 4.096 channel pulse height analyser. (M.A.C.) [pt
Lemarchand, Guillermo A.
1996-06-01
Project META (Megachannel ExtraTerrestrial Assay), a full-sky survey for artificial narrow-band signals, has been conducted from the Harvard/Smithsonian 26 m radiotelescope at Agassiz Station and from one of the two 30 m radiotelescopes of the Instituto Argentino de Radioastronomia (IAR). The search was performed near the 1420 MHz line of neutral hydrogen, and its second harmonic, using two 8.4 X 10(superscript 6) channel Fourier spectrometers of 0.05 Hz resolution and 400 kHz of instantaneous bandwidth. The observing frequency was corrected both for motions with respect to three astronomical inertial frames, and for the effect of Earth's rotation, which provides a characteristic changing signature for narrow-band signals of extraterrestrial origin. Among the 6 X 10(superscript 13) spectral channels searched in the northern hemisphere, Horowitz and Sagan reported 37 candidates events exceeding the average threshold of 1.7 X 10(superscript -23) W m(superscript -2), while in the southern hemisphere among 2 X 10(superscript 13) spectral channels analyzed we found 19 events exceeding the same threshold. The strongest signals that survive culling for terrestrial interference lie in or near the Galactic Plane. The first high resolution southern target search around 71 stars (-90 degrees intelligence. It is showed that these narrow-band non-repeating 'events' found by Project META can be generated by (a) radiometer noise fluctuations, (b) a population of constant galactic sources which undergo deep fading and amplification due to interstellar scintillation, consistent with ETI transmissions and (c) real, transient signals of either terrestrial or extraterrestrial origin. The Bayesian test shows that hypothesis (b) and (c) are both highly preferred to (a), but the first two are about equally likely. Using this analysis we discuss the best observing strategies to determine the real origin of these 'events'.
International Nuclear Information System (INIS)
Shimoyama, Susumu; Noda, Yasuko
2000-01-01
A portable RI X-ray fluorescence spectrometer having a weight of 1800 g was constructed by using a radiation annular source ( 241 Am sealed up with ceramics, 1.85 MBq), a small Si-PIN photodiode X-ray detector mounted on a Peltier cooling device, an amplifier and a pocket-type multi-channel pulse-height analyzer. It could also be applied to a non-destructive analysis of the blue colorant in the original Japanese votive picture 'Rashoumon-zu', offered to shrine in 1682. Consequently, it was clarified that the blue colorant (X) was 'Smalt' containing Ga, Fe, Co, Ni and As as major elements. (author)
Computerized radionuclidic analysis in production facilities
International Nuclear Information System (INIS)
Gibbs, A.
1978-03-01
The Savannah River Plant Laboratories Department has been using a dual computer system to control all radionuclidic pulse height analyses since 1971. This computerized system analyzes 7000 to 8000 samples per month and has allowed the counting room staff to be reduced from three persons to one person. More reliable process information is being returned to the production facilities and for environmental evaluations and being returned faster, even though the sample load has more than tripled. This information is now more easily retrievable for other evaluations. The computer is also used for mass spectrometer data reduction and for quality control data analysis. The basic system is being expanded by interfacing microcomputers which provide data input from all of the laboratory modules for quality assurance programs
Operator's guide for VAXGAP, a gamma-ray spectrum analysis package
International Nuclear Information System (INIS)
Killian, E.W.; Femec, D.A.
1992-08-01
This report describes the gamma-ray analysis program VAXGAP, which has continually evolved at the Idaho National Engineering Laboratory over a period of several years. It is an integrated suite of computer programs for performing analyses of pulse-height spectra from high-resolution gamma-ray spectrometers and assorted support functions. VAXGAP programs operate on Digital Equipment Corporation (DEC) VAX computers running the VMS operating system, and were written in VAX FORTRAN and DEC Digital Command Language (DCL). These programs make use of DEC GKS and ReGIS for graphical output on standard terminals and printers, and DEC windows for graphics on workstations and terminals that support the X Window System protocol. This report addresses the use of VAXGAP for data acquisition and control, energy scale calibration, and real-time analyses of background and sample pulse-height spectra. Also addressed are the creation and use efficiency tables and isotope libraries, manipulation of spectrum files and their contents, and graphical display of on-going acquisitions, saved spectra, and mathematical fits to spectral peaks
Neutron activation analysis of trace elements in foodstuffs
International Nuclear Information System (INIS)
Schelenz, R.; Fischer, E.
1976-05-01
A neutron activation method for multielement determination in biological material was developed. The individual steps of the method include radiochemical processing as well as nondestructive techniques. In order to develop a high resolution gamma spectrometric method the indispensable assumptions were the application of Ge(Li)-semiconductor detectors, multi-channel pulse height analyzers and the use of electronic data evaluation with mini-computers for the automatic evaluation of complex gamma spectra. After radiochemical separation (RNAA) 33 elements can be determined in biological materials and by application of nondestructive, purely instrumental techniques (INAA) 25 elements. The time required for the analysis of 33 elements can be determined in biological materials and by application of nondestructive, purely instrumental techniques (INAA) 25 elements. The time required for the analysis of 33 elements is 4 days. The neutron activation method is used routinely for the determination of trace elements in foodstuffs and in the field of nutrition research. (orig.) [de
Microcomputer-based pneumatic controller for neutron activation analysis
International Nuclear Information System (INIS)
Byrd, J.S.; Sand, R.J.
1976-10-01
A microcomputer-based pneumatic controller for neutron activation analysis was designed and built at the Savannah River Laboratory for analysis of large numbers of geologic samples for locating potential supplies of uranium ore for the National Uranium Resource Evaluation program. In this system, commercially available microcomputer logic modules are used to transport sample capsules through a network of pressurized air lines. The logic modules are interfaced to pneumatic valves, solenoids, and photo-optical detectors. The system operates from programs stored in firmware (permanent software). It also commands a minicomputer and a hard-wired pulse height analyzer for data collection and bookkeeping tasks. The advantage of the system is that major system changes can be implemented in the firmware with no hardware changes. This report describes the hardware, firmware, and software for the electronics system
International Nuclear Information System (INIS)
Matsuda, Yatsuka; Mamuro, Tetsuo
1975-01-01
An analytical technique for the simultaneous measurements of the atmospheric concentrations of SO 2 gas and sulphur absorbed by aerosol particles has been developed. Aerosol particles are collected on membrane filter and at the same time SO 2 gas is captured on alkali impregnated filter. The sulphur content in each filter is measured by an energy dispersive X -ray fluorescence spectrometer consisting of a Si(Li) semiconductor detector connected to a multi-channel pulse height analyzer and an excitation source of 55 Fe. Two methods are acceptable for the determination of the sulphur content in impregnated filter by X-ray fluorescence analysis. In the first method X-ray fluorescence analysis is made after the collected sulphur gas diffused and distributed uniformly enough throughout the filter, and in the second method X-ray fluorescence analysis gas to be finished before the diffusion of the collected sulphur becomes appreciable. (author)
Analysis of pulse-shape discrimination techniques for BC501A using GHz digital signal processing
International Nuclear Information System (INIS)
Rooney, B.D.; Dinwiddie, D.R.; Nelson, M.A.; Rawool-Sullivan, Mohini W.
2001-01-01
A comparison study of pulse-shape analysis techniques was conducted for a BC501A scintillator using digital signal processing (DSP). In this study, output signals from a preamplifier were input directly into a 1 GHz analog-to-digital converter. The digitized data obtained with this method was post-processed for both pulse-height and pulse-shape information. Several different analysis techniques were evaluated for neutron and gamma-ray pulse-shape discrimination. It was surprising that one of the simplest and fastest techniques resulted in some of the best pulse-shape discrimination results. This technique, referred to here as the Integral Ratio technique, was able to effectively process several thousand detector pulses per second. This paper presents the results and findings of this study for various pulse-shape analysis techniques with digitized detector signals.
International Nuclear Information System (INIS)
Velasquez, A.A.; Trujillo, J.M.; Morales, A.L.; Tobon, J.E.; Gancedo, J.R.; Reyes, L.
2005-01-01
A multichannel analyzer (MCA) and a single channel-analyzer (SCA) for Moessbauer spectrometry application have been designed and built. Both systems include low-cost digital and analog components. A microcontroller manages, either in PHA or MCS mode, the data acquisition, data storage and setting of the pulse discriminator limits. The user can monitor the system from an external PC through the serial port with the RS232 communication protocol. A graphic interface made with the LabVIEW software allows the user to adjust digitally the lower and upper limits of the pulse discriminator, and to visualize as well as save the PHA spectra in a file. The system has been tested using a 57Co radioactive source and several iron compounds, yielding satisfactory results. The low cost of its design, construction and maintenance make this equipment an attractive choice when assembling a Moessbauer spectrometer
Energy Technology Data Exchange (ETDEWEB)
Dewberry, R.; Sigg, R.; Salaymeh, S.
2009-03-23
The Savannah River Site (SRS) Burial Ground had a container labeled as Box 33 for which they had no reliable solid waste stream designation. The container consisted of an outer box of dimensions 48-inch x 46-inch x 66-inch and an inner box that contained high density and high radiation dose material. From the outer box Radiation Control measured an extremity dose rate of 22 mrem/h. With the lid removed from the outer box, the maximum dose rate measured from the inner box was 100 mrem/h extremity and 80 mrem/h whole body. From the outer box the material was sufficiently high in density that the Solid Waste Management operators were unable to obtain a Co-60 radiograph of the contents. Solid Waste Management requested that the Analytical Development Section of Savannah River National Laboratory perform a {gamma}-ray assay of the item to evaluate the radioactive content and possibly to designate a solid waste stream. This paper contains the results of three models used to analyze the measured {gamma}-ray data acquired in an unusual configuration.
International Nuclear Information System (INIS)
Ballini, J.-P.; Cazes, P.; Turpin, P.-Y.
1976-01-01
Analysing the histogram of anode pulse amplitudes allows a discussion of the hypothesis that has been proposed to account for the statistical processes of secondary multiplication in a photomultiplier. In an earlier work, good agreement was obtained between experimental and reconstructed spectra, assuming a first dynode distribution including two Poisson distributions of distinct mean values. This first approximation led to a search for a method which could give the weights of several Poisson distributions of distinct mean values. Three methods have been briefly exposed: classical linear regression, constraint regression (d'Esopo's method), and regression on variables subject to error. The use of these methods gives an approach of the frequency function which represents the dispersion of the punctual mean gain around the whole first dynode mean gain value. Comparison between this function and the one employed in Polya distribution allows the statement that the latter is inadequate to describe the statistical process of secondary multiplication. Numerous spectra obtained with two kinds of photomultiplier working under different physical conditions have been analysed. Then two points are discussed: - Does the frequency function represent the dynode structure and the interdynode collection process. - Is the model (the multiplication process of all dynodes but the first one, is Poissonian) valid whatever the photomultiplier and the utilization conditions. (Auth.)
International Nuclear Information System (INIS)
Dewberry, R.; Sigg, R.; Salaymeh, S.
2009-01-01
The Savannah River Site (SRS) Burial Ground had a container labeled as Box 33 for which they had no reliable solid waste stream designation. The container consisted of an outer box of dimensions 48-inch x 46-inch x 66-inch and an inner box that contained high density and high radiation dose material. From the outer box Radiation Control measured an extremity dose rate of 22 mrem/h. With the lid removed from the outer box, the maximum dose rate measured from the inner box was 100 mrem/h extremity and 80 mrem/h whole body. From the outer box the material was sufficiently high in density that the Solid Waste Management operators were unable to obtain a Co-60 radiograph of the contents. Solid Waste Management requested that the Analytical Development Section of Savannah River National Laboratory perform a γ-ray assay of the item to evaluate the radioactive content and possibly to designate a solid waste stream. This paper contains the results of three models used to analyze the measured γ-ray data acquired in an unusual configuration
International Nuclear Information System (INIS)
Greim, L.; Motamedi, K.; Niedergesaess, R.
1976-01-01
A computer code evaluating experimental data of neutron activation analysis (NAA) for determination of atomic abundancies is described. The experimental data are, beside a probe designation, the probe weight, irradiation parameters and a Ge(Li)-pulse-height-spectrum from the activity measurement. The organisation of the necessary nuclear data, comprising all methods of activation in reactor-irradiations, is given. Furthermore the automatic evaluation of spectra, the designation of the resulting peaks to nuclei and the calculation of atomic abundancies are described. The complete evaluation of a spectrum with many lines, e.g. 100 lines of 20 nuclei, takes less than 1 minute machine-time on the TR 440 computer. (orig.) [de
International Nuclear Information System (INIS)
Spitz, H.B.; Buschbom, R.; Rieksts, G.A.; Palmer, H.E.
1985-01-01
A new method has been developed to reliably analyze pulse height-energy spectra obtained from measurements employing high resolution germanium detectors. The method employs a simple data transformation and smoothing function to calculate background and identify photopeaks and isotopic analysis. This technique is elegant in its simplicity because it avoids dependence upon complex spectrum deconvolution, stripping, or other least-square-fitting techniques which complicate the assessment of measurement reliability. A moving median was chosen for data smoothing because, unlike moving averages, medians are not dominated by extreme data points. Finally, peaks are identified whenever the difference between the background spectrum and the transformed spectrum exceeds a pre-determined number of standard deviations
Automatic particle-size analysis of HTGR nuclear fuel microspheres
International Nuclear Information System (INIS)
Mack, J.E.
1977-01-01
An automatic particle-size analyzer (PSA) has been developed at ORNL for measuring and counting samples of nuclear fuel microspheres in the diameter range of 300 to 1000 μm at rates in excess of 2000 particles per minute, requiring no sample preparation. A light blockage technique is used in conjunction with a particle singularizer. Each particle in the sample is sized, and the information is accumulated by a multi-channel pulse height analyzer. The data are then transferred automatically to a computer for calculation of mean diameter, standard deviation, kurtosis, and skewness of the distribution. Entering the sample weight and pre-coating data permits calculation of particle density and the mean coating thickness and density. Following this nondestructive analysis, the sample is collected and returned to the process line or used for further analysis. The device has potential as an on-line quality control device in processes dealing with spherical or near-spherical particles where rapid analysis is required for process control
Velásquez, A. A.; Gancedo, J. R.; Trujillo, J. M.; Morales, A. L.; Tobón, J. E.; Reyes, L.
2005-04-01
A multichannel analyzer (MCA) and a single channel-analyzer (SCA) for Mössbauer spectrometry application have been designed and built. Both systems include low-cost digital and analog components. A microcontroller manages, either in PHA or MCS mode, the data acquisition, data storage and setting of the pulse discriminator limits. The user can monitor the system from an external PC through the serial port with the RS232 communication protocol. A graphic interface made with the LabVIEW software allows the user to adjust digitally the lower and upper limits of the pulse discriminator, and to visualize as well as save the PHA spectra in a file. The system has been tested using a 57Co radioactive source and several iron compounds, yielding satisfactory results. The low cost of its design, construction and maintenance make this equipment an attractive choice when assembling a Mössbauer spectrometer.
International Nuclear Information System (INIS)
Ono, Masayoshi; Tasaki, Seiji; Okamoto, Sunao
1989-01-01
A data acquisition system having the various modern electronic devices was designed and tested for practical use of neutron time-of-flight (TOF) measurements with multiple counters. The system is principally composed of TOF logic units (load-able up to 128 units) with a control unit and a conventional micro-computer. The TOF logic unit (main memory, 2048 ch, 24 bits/ch) demonstrates about 1.7 times higher efficiency for neutron counting rate per channel than the one by a conventional TOF logic unit. Meanwhile, some data-access functions of the TOF logic unit were applied to position sensitive analyzer of one-dimensional neutron PSD for small angle scattering. The analyzer was tested with use of pulse generator. The result shows good linearity. (author)
Status of neutron activation analysis in developing countries
International Nuclear Information System (INIS)
Chatt, A.
1996-01-01
The 60th anniversary of the discovery of neutron activation analysis (NAA) by Hevesy and Levi is being celebrated in 1996. With the availability of nuclear reactors capable of producing fluxes of the order of 10 12 to 10 14 n/cm 2 s, the development of high-resolution and high-efficiency conventional and anticoincidence gamma-ray detectors, multichannel pulse-height analyzers, and personal computer-based softwares, NAA has become an extremely valuable analytical technique, especially for the simultaneous determinations of multielement concentrations. This technique can be used in a number of ways, depending on the nature of the matrix, the major elements in the sample, and on the elements of interest. In most cases, several elements can be determined without any chemical pretreatment of the sample; the technique is then called instrumental NAA (INAA). In other cases, an element can be concentrated from an interfering matrix prior to irradiation; the technique is then termed preconcentration NAA (PNAA). In opposite instances, the irradiation is followed by a chemical separation of the desired element; the technique is then called radiochemical NAA (RNAA). All three forms of NAA can provide elemental concentrations of high accuracy and precision with excellent sensitivity. The number of research reactors in developing countries has increased steadily from 17 in 1955 through 71 in 1975 to 89 in 1995. Low flux reactors such as SLOWPOKE and the Chinese MNSR are primarily used for NAA
Determination of trace elements in Turkish tea leaves by instrumental neutron activation analysis
International Nuclear Information System (INIS)
Demiralp, R.
1986-01-01
The human body continuously assimilates a variety of inorganic elements from food and the environment. Some of these elements are closely related to human health and disorder. Tea is one of the most popular stimulating beverages which is consumed by low and high income family groups in many countries. Instrumental neutron activation analysis is one of the preferred methods because information on a large number of elements can be obtained simultaneously. Five packets each of the seven commonly used brands of tea were obtained from the market. In order to determine the transfer of trace elements into the drinkable portion about 2-3g. of the tea leaves were boiled in hot water for 2 min. After filtration the used tea leaves were dried at 65 deg. C in an oven and a portion, about 200mg was used for analysis. Samples and standards were irradiated 10 min. and 2 hrs. at pneumatic system and central thimble in the TRIGA MARK-II research reactor. After irradiation, the activities of samples and standards were measured with a aoaxial Ge detector coupled to a spectroscopic amplifier. A Canbera 90 model multi- channel analyzer with an 8K memory was used for pulse height analysis. The system has a resolution of 2.0 KeV. for the 1332.5 KeV gamma ray of 60 Co. The activity of the sample and standard was compared and the element contst of the sample was calculated. (author)
Principal component analysis of solar flares in the soft X-ray flux
International Nuclear Information System (INIS)
Teuber, D.L.; Reichmann, E.J.; Wilson, R.M.; National Aeronautics and Space Administration, Huntsville, AL
1979-01-01
Principal component analysis is a technique for extracting the salient features from a mass of data. It applies, in particular, to the analysis of nonstationary ensembles. Computational schemes for this task require the evaluation of eigenvalues of matrices. We have used EISPACK Matrix Eigen System Routines on an IBM 360-75 to analyze full-disk proportional-counter data from the X-ray event analyzer (X-REA) which was part of the Skylab ATM/S-056 experiment. Empirical orthogonal functions have been derived for events in the soft X-ray spectrum between 2.5 and 20 A during different time frames between June 1973 and January 1974. Results indicate that approximately 90% of the cumulative power of each analyzed flare is contained in the largest eigenvector. The first two largest eigenvectors are sufficient for an empirical curve-fit through the raw data and a characterization of solar flares in the soft X-ray flux. Power spectra of the two largest eigenvectors reveal a previously reported periodicity of approximately 5 min. Similar signatures were also obtained from flares that are synchronized on maximum pulse-height when subjected to a principal component analysis. (orig.)
Neutron activation analysis of single human hairs and possible applications for forensic purposes
International Nuclear Information System (INIS)
Dybczynski, R.; Boboli, K.
1975-01-01
A new analytical procedure that enables to determine more than 10 trace elements in single, 3 cm human hair segments by neutron activation analysis (NAA) was elaborated. Application of a special washing procedure of hair (before irradiation) made possible to lower sodium content by two orders of magnitude without affecting trace element content. After irradiation in the thermal neutron flux of about 10 14 n.cm -2 .sec -1 the activity of hair was measured with 70 cm 3 Ge(Li) detector coupled to 4000 channel pulse height analyser. As an alternative method, a destructive version of NAA with ion exchange group separation of radionuclides was applied. It was found that sometimes high gradients of element concentration along the hair length exist and that there is without any doubt a distinct influence of environmental factor on the content of some trace elements in hair. The criminalistic aspects of hair analysis were also studied using new statistical criterion for elimination (identification). Both possibilities and limitations of the method are discussed. (author)
Study on x-ray spectroscopic analysis by the use of nuclear radiations from radioisotopes
International Nuclear Information System (INIS)
Tominaga, Hiroshi
1977-02-01
The basis and practice of analytical application of the characteristic X rays of low atomic number elements excited by mainly alpha rays from radioisotopes were studied. Some alpha emitters were proved to be advantageously usable to those elements, in particular, very low atomic number elements, despite their accompanying radiations. Operating characteristics of gas flow proportional counters were fully investigated, and besides, the characteristics of pulse-height distribution continuum in those counters and semiconductor detectors were comparatively studied as the detector for the low energy X rays. For calibration in quantitative analysis by the X-ray spectroscopy, a new simple semitheoretical method for thin specimens is proposed and on the other hand general formulae for matrix correction in thick specimens was derived. An industrial X-ray analyzer to simultaneously analyze four main elements of cement raw material mixtures was constructed with the use of a 210 Po alpha-ray source and gain-controlled proportional counters. Problems in practical analysis for the process control were solved and the analyzer has been used routinely in the process for more than ten years. Another analytical application to airborne dust is described as a typical example of application of the proposed calibration method for thin specimens. (auth.)
International Nuclear Information System (INIS)
Cardoso, Antonio
1976-01-01
Neutron Activation Analysis, using a Californium-252 neutron source, has been applied for the determination of manganese in ores such as pyrolusite, rodonite (manganese silicate)' and blending used in dry-batteries The favorable nuclear properties of manganese, such as high thermal neutron cross-section for the reaction 55 Mn (n.gamma) 56 Mn, high concentration of manganese in the matrix and short half - life of 56 Mn, are an ideal combination for non-destructive analysis of manganese in ores. Samples and standards of manganese dioxide were irradiated for about 20 minutes, followed by a 4 to 15 minutes decay and counted in a single channel pulse-height discrimination using a NaI(Tl) scintillation detector. Counting time was equal to 10 minutes. The interference of nuclear reactions 56 Fe(n,p) 56 Mn and 59 Co (n, α) 56 were studied, as well as problems in connection with neutron shadowing during irradiation, gamma-rays attenuation during counting and influence of granulometry of samples. One sample,was also analysed by wet-chemical method (sodium bismuthate) in order to compare results. As a whole, i t was shown that the analytical method of neutron activation for manganese in ores and blending, is a method simple, rapid and with good precision and accuracy. (author)
Past, present, future of neutron activation analysis
International Nuclear Information System (INIS)
Guinn, V.P.
1991-01-01
Although the method of NAA was originated in 1936, use of the method was minimal for about the next ten years - due to the lack of a high-flux source of thermal neutrons. When such a copious source of neutrons - the nuclear reactor - became available, commencing in 1945, the power of the method to determine many elements, quantitatively, even though present in samples at concentrations as low as ppb, attracted many scientists. During about the next 20 years (1945-1965), NAA provided the first major results on trace elements (essential and toxic) in biomedical materials. These studies constituted the first major achievement of NAA - and such studies continue. Due to the initial lack of a high-efficiency detector of gamma radiation, work during the early part of this period had to be carried out by tedious radiochemical separations. Starting in the early 1950's, with the advent of the NaI(Tl) scintillation detector and multichannel pulse-height analyzers, the purely-instrumental form of the method (INAA) was born, and research and applications mushroomed. In the 1960s, two additional important applications of NAA were its vital use in the semiconductor industry and in the analysis of the lunar rocks. The INAA method in this period was greatly advanced by the availability of high-resolution germanium detectors. Applications in environmental studies, archaeology, and forensic chemistry were also of note. INAA is now a mature method, but still improving. As for the future, what is still needed are less expensive high-flux neutron sources, simple automated systems, gamma-ray detectors of even better resolution, and faster electronics. The field of NAA has been, and is, blessed by a large number of outstanding scientists, world-wide, all intrigued by the atomic nucleus and radioactive decay
International Nuclear Information System (INIS)
Kim, Jong-Yun; Choi, Yong Suk; Park, Yong Joon; Jung, Sung-Hee
2009-01-01
Neutron spectrometry, based on the scattering of high energy fast neutrons from a radioisotope and slowing-down by the light hydrogen atoms, is a useful technique for non-destructive, quantitative measurement of hydrogen content because it has a large measuring volume, and is not affected by temperature, pressure, pH value and color. The most common choice for radioisotope neutron source is 252 Cf or 241 Am-Be. In this study, 252 Cf with a neutron flux of 6.3x10 6 n/s has been used as an attractive neutron source because of its high flux neutron and weak radioactivity. Pulse-height neutron spectra have been obtained by using in-house built radioisotopic neutron spectrometric system equipped with 3 He detector and multi-channel analyzer, including a neutron shield. As a preliminary study, polyethylene block (density of ∼0.947 g/cc and area of 40 cmx25 cm) was used for the determination of hydrogen content by using multivariate calibration models, depending on the thickness of the block. Compared with the results obtained from a simple linear calibration model, partial least-squares regression (PLSR) method offered a better performance in a quantitative data analysis. It also revealed that the PLSR method in a neutron spectrometric system can be promising in the real-time, online monitoring of the powder process to determine the content of any type of molecules containing hydrogen nuclei.
Analysis of uranium, thorium, and potassium in the soil and rocks in northwestern Taiwan
International Nuclear Information System (INIS)
Shyong, J.; Wu, C.
1984-01-01
The contents of uranium, thorium, and potassium in terrestrial samples in northwestern Taiwan were determined by field survey and core sampling techniques. NaI(Tl) scintillation survey meters were applied to field survey. Analysis of radionuclides in the soil and rocks was performed by a 60 cm 3 Ge(Li) detector connected with a 4096-channel pulse height analyzer and a minicomputer PDP 11/04. Computer programs were used to identify energies of photopeaks, to integrate peak areas, and to evaluate the contents of radionuclides to ppm order. All the factors such as counting efficiency, statistical uncertainty, dead time, etc. had been taken into consideration. Natural terrestrial radiation exposure rates ranging from 6.5 to 20.5 uR.hr -1 were observed in 65 villages or hamlets. The average concentration of each of uranium, thorium, and potassium was 3.7+.0.8 ppm, 12.0+.2.8 ppm, and 1.3+.0.5 percent respectively
Evaluation of the health status of six volunteers from the Mars 500 project using pulse analysis.
Shi, Hong-Zhi; Fan, Quan-Chun; Gao, Jian-Yi; Liu, Jun-Lian; Bai, Gui-E; Mi, Tao; Zhao, Shuang; Liu, Yu; Xu, Dong; Guo, Zhi-Feng; Li, Yong-Zhi
2017-08-01
To comprehensively evaluate the health status of 6 volunteers from the Mars 500 Project through analyzing their pulse graphs and determining the changes in cardiovascular function, degree of fatigue and autonomic nervous function. Six volunteers were recruited; all were male aged 26-38 years (average 31.83±4.96 years). Characteristic parameters reflflecting the status of cardiovascular functions were extracted, which included left ventricular contraction, vascular elasticity and peripheral resistance. The degree of fatigue was determined depending on the difference between the calendar age and biological age, which was calculated through the analysis of blood pressure value and characteristic parameters. Based on the values of pulse height variation and pulse time variation on a 30-s pulse graph, autonomic nervous function was evaluated. All parameters examined were marked on an equilateral polygon to form an irregular polygon of the actual fifigure, then health status was evaluated based on the coverage area of the actual fifigure. The results demonstrated: (1) volunteers developed weakened pulse power, increased vascular tension and peripheral resistance, and slight decreased ventricular systolic function; (2) the degree of fatigue was basically mild or moderate; and (3) autonomic nervous function was excited but generally balanced. These volunteers were in the state of sub-health. According to Chinese medicine theories, such symptoms are mainly caused by the weakening of healthy qi, Gan (Liver) failing in free coursing, and disharmony between Gan and Wei (Stomach), which manifests as a weak and string-like pulse.
Applications of short lived nuclides in activation analysis, problems and progress
Energy Technology Data Exchange (ETDEWEB)
Grass, F [Atominstitut, Vienna (Austria)
1976-07-01
Short lived nuclides or isomeric transitions, respectively would have some advantages over long lived ones. Although we published a paper concerning a germanium-determination in iron meteorites some years ago, only few laboratories use this technique, the main reason being that the high matrix activity disturbs the measurement of energy-spectra. A multichannel analyzer in the time sequence mode enables Li-8 determination by a purely instrumental method which is therefore used more frequently. In the time sequence mode much higher counting rates up to 10 - 50 MHz are processed then by taking energy-spectra. This is the reason why activation analysis with short lived isomeric states is seldom applied when counting rate and pulse height are to be detected simultaneously. Exceptional difficulties are encountered in measurement of samples activated by a reactor pulse. Further difficulties arise from the fact that an optimal expelling time depends on the half life of the nuclide, and is more critical if the half life is short and the full width half maximum of the reactor pulse is small. Commercial Ge-Li-detectors can be used only at low counting rates, so that samples with high matrix activities cannot be measured. Modifying the electronic system enables registration of samples with high matrix activities. For short lived nuclides emitting hard beta-rays, e.g. B-12 or Li-8, a Cerenkov-detector is optimal. These problems are discussed in examples. (author)
Passive gamma analysis of the boiling-water-reactor assemblies
Energy Technology Data Exchange (ETDEWEB)
Vo, D., E-mail: ducvo@lanl.gov [Los Alamos National Laboratory, Los Alamos, NM (United States); Favalli, A. [Los Alamos National Laboratory, Los Alamos, NM (United States); Grogan, B. [Oak Ridge National Laboratory, Oak Ridge, TN (United States); Jansson, P. [Uppsala University, Uppsala (Sweden); Liljenfeldt, H. [Oak Ridge National Laboratory, Oak Ridge, TN (United States); Mozin, V. [Lawrence Livermore National Laboratory, Livermore, CA (United States); Schwalbach, P. [European Atomic Energy Community (EURATOM), Luxemburg (Luxembourg); Sjöland, A. [Swedish Nuclear Fuel and Waste Management Company, Stockholm (Sweden); Tobin, S.; Trellue, H. [Los Alamos National Laboratory, Los Alamos, NM (United States); Vaccaro, S. [European Atomic Energy Community (EURATOM), Luxemburg (Luxembourg)
2016-09-11
This research focused on the analysis of a set of stationary passive gamma measurements taken on the spent nuclear fuel assemblies from a boiling water reactor (BWR) using pulse height analysis data acquisition. The measurements were performed on 25 different BWR assemblies in 2014 at Sweden's Central Interim Storage Facility for Spent Nuclear Fuel (Clab). This study was performed as part of the Next Generation of Safeguards Initiative–Spent Fuel project to research the application of nondestructive assay (NDA) to spent fuel assemblies. The NGSI–SF team is working to achieve the following technical goals more easily and efficiently than in the past using nondestructive assay (NDA) measurements of spent fuel assemblies: (1) verify the initial enrichment, burnup, and cooling time of facility declaration; (2) detect the diversion or replacement of pins, (3) estimate the plutonium mass, (4) estimate the decay heat, and (5) determine the reactivity of spent fuel assemblies. The final objective of this project is to quantify the capability of several integrated NDA instruments to meet the aforementioned goals using the combined signatures of neutrons, gamma rays, and heat. This report presents a selection of the measured data and summarizes an analysis of the results. Specifically, trends in the count rates measured for spectral lines from the following isotopes were analyzed as a function of the declared burnup and cooling time: {sup 137}Cs, {sup 154}Eu, {sup 134}Cs, and to a lesser extent, {sup 106}Ru and {sup 144}Ce. From these measured count rates, predictive algorithms were developed to enable the estimation of the burnup and cooling time. Furthermore, these algorithms were benchmarked on a set of assemblies not included in the standard assemblies set used by this research team.
16-channel analog store and multiplexer unit
Energy Technology Data Exchange (ETDEWEB)
Brossard, M; Kulka, Z [Clermont-Ferrand-2 Univ., 63 - Aubiere (France). Lab. de Physique Corpusculaire
1979-03-15
A 16-channel analog store and multiplexer unit is described. The unit enables storing and selection of analog information which is then digitally encoded by single ADC. This solution becomes economically attractive particularly in multidetector pulse height analysis systems.
International Nuclear Information System (INIS)
Barbosa, J.B.S.
1985-11-01
The basic principles involved in Si(Li) system used in X-ray spectrometry is described. It also demonstrates its application in the energy range where the resolution is better than that characteristic of conventional spectrometers. The theoretical principles underlying the interaction between the electromagnetic radiation and matter, and a review on semiconductors are presented at first. It emphasizes the fluorescence phenomenon and the process of photon detection by semiconductor crystals whose properties and characteristics allow, in the specific case of Si-crystal, the confection of detectors with large sensitivity volume useful for X-ray spectrometry. In addition, the components of the Si(Li) system are described individually, with special attention to the operating aspects, and to the parameters affecting the quality of pulse height spectrum. Finally, the spectrometer performance is experimentally evaluated though the quantitative analyses of rare-earth element oxides (La, Ce, Pr, Nd). It should be stressed that this research indicates that the X-ray emission-transmission analysis is the most adequate method under the activation conditions provided by the spectrometer, where Am 241 emissor UPSILON of 60KeV is the photon source for the fluorescence. Therefore, the experimental work was extended in order to include all the necessary treatment. (Author) [pt
International Nuclear Information System (INIS)
Tanaka, Keiichi; Odawara, Akikazu; Nagata, Atsushi; Ikeda, Masanori; Baba, Yukari; Nakayama, Satoshi; Chinone, Kazuo
2006-01-01
A cryogen-free energy-dispersive spectrometer (EDS) using a transition edge sensor (TES) was developed for material analysis. This system can maintain a temperature at 130 mK within 30 μK, and has good energy resolution (19 eV for Mn-Kα) for long-time measurement with a drift in the DC level of less than 0.02 eV/min. This system utilizes a dilution refrigerator (φ 272 mmxheight 572 mm) and has a snout (370 mm long and φ25 mm) similar to that in a conventional EDS system. The dilution refrigerator is pre-cooled by a GM refrigerator. A flexible tube between the dilution refrigerator and GM refrigerator damps the mechanical vibration of the GM refrigerator. Two shields (4 and 80 K) thermally protect the Cu rod (φ8 mm) cooled to be 100 mK. Windows composed of polyimide+Al film allow X-ray detection above the C-Kα line. A TES (6 mmx6 mm) and array SQUID amplifier (1.5 mmx3 mm) are mounted on top of the Cu rod. For Mn-Kα, the pulse height is 5.5 μA and decay time (τ eff ) is 90 μs. The maximum count rate (1/20 τ eff ) is estimated at about 500 cps
Spectral analysis of colour-quenched and chemically quenched C 14 samples
International Nuclear Information System (INIS)
Grau Malonda, A.; Scott Guillearrd, P.E.
1987-01-01
Pairs of pulse height distribution curves, of C-14 samples, colour quenched and chemically quenched were obtained. The possibility to choose a counting window in order to obtain the counting efficiency curves, for both type of quenching was studied. (author). 7 figs., 7 refs
Spectral analysis of colour-quenched and chemically quenched C-14 samples
International Nuclear Information System (INIS)
Scott, P. E.; Grau, A.
1987-01-01
In this paper pairs of pulse height distribution curves, of C-14 samples, colour-quenched and chemically quenched was obtained. The possibility to choose a counting window in order to obtain the counting efficiency curves, for both type of quenching was studied. (Author) 7 refs
Utilization of neutrons in nuclear data measurements and bulk sample analysis
International Nuclear Information System (INIS)
Jonah, S. A.
1995-01-01
Experimental investigations were carried out with neutrons in the fields of neutron data measurements and bulk sample analysis based on the interactions of neutron interactions required in the investigations together with some salient features of the sources employed are enumerated. Excitation cross section curves and isomeric cross section ratio of 58 Ni(n,p) 58 Co m , g reaction over the neutron energy range of between 5 and 15 MeV were determined using the activation analysis technique in combination with high-resolution gamma spectroscopy. Characteristics of the incident neutrons produced via the D-T reaction of a neutron generator and D-D reaction of a cyclotron were determined experimentally to account for the contributing effects of background neutrons especially in the 5-13 MeV neutron energy range where existing data are scanty and rather discrepant. The measured data agree well with calculated data using nuclear models but deviate significantly from the recommended data based on existing literature data. The measured δ act and δ m /δ g data made it possible to determine the cross section curve for 58 Ni(n,p) 58 Co m reaction. Furthermore the flux density distributions of thermal and primary fast neutrons in different configurations of bulk samples consisting of water, graphite and coal together with the attenuation characteristics were determined by the activation analysis and pulse height response spectrometry techniques. From the results obtained, an experimental geometry has been proposed for on-line elemental analysis of coal and other minerals. Similarly the total hydrogen content and 0+C/H atomic ratio in household and motor oils as well as crude oil samples of different origins were measured by an improved experimental arrangement based on the thermal neutron reflection technique. A detection limit of 0.12 w % was obtained for hydrogen indicating the possible adaptation of this technique for quality control of petroleum products
DEFF Research Database (Denmark)
Mathiesen, Brian Vad; Liu, Wen; Zhang, Xiliang
2014-01-01
three major technological changes: energy savings on the demand side, efficiency improvements in energy production, and the replacement of fossil fuels by various sources of renewable energy. Consequently, the analysis of these systems must include strategies for integrating renewable sources...
Energy Technology Data Exchange (ETDEWEB)
Cardoso, Antonio
1976-07-01
Neutron Activation Analysis, using a Californium-252 neutron source, has been applied for the determination of manganese in ores such as pyrolusite, rodonite (manganese silicate)' and blending used in dry-batteries The favorable nuclear properties of manganese, such as high thermal neutron cross-section for the reaction {sup 55}Mn (n.gamma){sup 56} Mn, high concentration of manganese in the matrix and short half - life of {sup 56}Mn, are an ideal combination for non-destructive analysis of manganese in ores. Samples and standards of manganese dioxide were irradiated for about 20 minutes, followed by a 4 to 15 minutes decay and counted in a single channel pulse-height discrimination using a NaI(Tl) scintillation detector. Counting time was equal to 10 minutes. The interference of nuclear reactions {sup 56}Fe(n,p){sup 56}Mn and {sup 59} Co (n, {alpha}){sup 56} were studied, as well as problems in connection with neutron shadowing during irradiation, gamma-rays attenuation during counting and influence of granulometry of samples. One sample,was also analysed by wet-chemical method (sodium bismuthate) in order to compare results. As a whole, i t was shown that the analytical method of neutron activation for manganese in ores and blending, is a method simple, rapid and with good precision and accuracy. (author)
Energy Technology Data Exchange (ETDEWEB)
Cardoso, Antonio
1976-07-01
Neutron Activation Analysis, using a Californium-252 neutron source, has been applied for the determination of manganese in ores such as pyrolusite, rodonite (manganese silicate)' and blending used in dry-batteries The favorable nuclear properties of manganese, such as high thermal neutron cross-section for the reaction {sup 55}Mn (n.gamma){sup 56} Mn, high concentration of manganese in the matrix and short half - life of {sup 56}Mn, are an ideal combination for non-destructive analysis of manganese in ores. Samples and standards of manganese dioxide were irradiated for about 20 minutes, followed by a 4 to 15 minutes decay and counted in a single channel pulse-height discrimination using a NaI(Tl) scintillation detector. Counting time was equal to 10 minutes. The interference of nuclear reactions {sup 56}Fe(n,p){sup 56}Mn and {sup 59} Co (n, {alpha}){sup 56} were studied, as well as problems in connection with neutron shadowing during irradiation, gamma-rays attenuation during counting and influence of granulometry of samples. One sample,was also analysed by wet-chemical method (sodium bismuthate) in order to compare results. As a whole, i t was shown that the analytical method of neutron activation for manganese in ores and blending, is a method simple, rapid and with good precision and accuracy. (author)
Uncertainty analysis in the simulation of an HPGe detector using the Monte Carlo Code MCNP5
International Nuclear Information System (INIS)
Gallardo, Sergio; Pozuelo, Fausto; Querol, Andrea; Verdu, Gumersindo; Rodenas, Jose; Ortiz, J.; Pereira, Claubia
2013-01-01
A gamma spectrometer including an HPGe detector is commonly used for environmental radioactivity measurements. Many works have been focused on the simulation of the HPGe detector using Monte Carlo codes such as MCNP5. However, the simulation of this kind of detectors presents important difficulties due to the lack of information from manufacturers and due to loss of intrinsic properties in aging detectors. Some parameters such as the active volume or the Ge dead layer thickness are many times unknown and are estimated during simulations. In this work, a detailed model of an HPGe detector and a petri dish containing a certified gamma source has been done. The certified gamma source contains nuclides to cover the energy range between 50 and 1800 keV. As a result of the simulation, the Pulse Height Distribution (PHD) is obtained and the efficiency curve can be calculated from net peak areas and taking into account the certified activity of the source. In order to avoid errors due to the net area calculation, the simulated PHD is treated using the GammaVision software. On the other hand, it is proposed to use the Noether-Wilks formula to do an uncertainty analysis of model with the main goal of determining the efficiency curve of this detector and its associated uncertainty. The uncertainty analysis has been focused on dead layer thickness at different positions of the crystal. Results confirm the important role of the dead layer thickness in the low energy range of the efficiency curve. In the high energy range (from 300 to 1800 keV) the main contribution to the absolute uncertainty is due to variations in the active volume. (author)
Uncertainty analysis in the simulation of an HPGe detector using the Monte Carlo Code MCNP5
Energy Technology Data Exchange (ETDEWEB)
Gallardo, Sergio; Pozuelo, Fausto; Querol, Andrea; Verdu, Gumersindo; Rodenas, Jose, E-mail: sergalbe@upv.es [Universitat Politecnica de Valencia, Valencia, (Spain). Instituto de Seguridad Industrial, Radiofisica y Medioambiental (ISIRYM); Ortiz, J. [Universitat Politecnica de Valencia, Valencia, (Spain). Servicio de Radiaciones. Lab. de Radiactividad Ambiental; Pereira, Claubia [Universidade Federal de Minas Gerais (UFMG), Belo Horizonte, MG (Brazil). Departamento de Engenharia Nuclear
2013-07-01
A gamma spectrometer including an HPGe detector is commonly used for environmental radioactivity measurements. Many works have been focused on the simulation of the HPGe detector using Monte Carlo codes such as MCNP5. However, the simulation of this kind of detectors presents important difficulties due to the lack of information from manufacturers and due to loss of intrinsic properties in aging detectors. Some parameters such as the active volume or the Ge dead layer thickness are many times unknown and are estimated during simulations. In this work, a detailed model of an HPGe detector and a petri dish containing a certified gamma source has been done. The certified gamma source contains nuclides to cover the energy range between 50 and 1800 keV. As a result of the simulation, the Pulse Height Distribution (PHD) is obtained and the efficiency curve can be calculated from net peak areas and taking into account the certified activity of the source. In order to avoid errors due to the net area calculation, the simulated PHD is treated using the GammaVision software. On the other hand, it is proposed to use the Noether-Wilks formula to do an uncertainty analysis of model with the main goal of determining the efficiency curve of this detector and its associated uncertainty. The uncertainty analysis has been focused on dead layer thickness at different positions of the crystal. Results confirm the important role of the dead layer thickness in the low energy range of the efficiency curve. In the high energy range (from 300 to 1800 keV) the main contribution to the absolute uncertainty is due to variations in the active volume. (author)
Cascaded systems analysis of photon counting detectors.
Xu, J; Zbijewski, W; Gang, G; Stayman, J W; Taguchi, K; Lundqvist, M; Fredenberg, E; Carrino, J A; Siewerdsen, J H
2014-10-01
Photon counting detectors (PCDs) are an emerging technology with applications in spectral and low-dose radiographic and tomographic imaging. This paper develops an analytical model of PCD imaging performance, including the system gain, modulation transfer function (MTF), noise-power spectrum (NPS), and detective quantum efficiency (DQE). A cascaded systems analysis model describing the propagation of quanta through the imaging chain was developed. The model was validated in comparison to the physical performance of a silicon-strip PCD implemented on an experimental imaging bench. The signal response, MTF, and NPS were measured and compared to theory as a function of exposure conditions (70 kVp, 1-7 mA), detector threshold, and readout mode (i.e., the option for coincidence detection). The model sheds new light on the dependence of spatial resolution, charge sharing, and additive noise effects on threshold selection and was used to investigate the factors governing PCD performance, including the fundamental advantages and limitations of PCDs in comparison to energy-integrating detectors (EIDs) in the linear regime for which pulse pileup can be ignored. The detector exhibited highly linear mean signal response across the system operating range and agreed well with theoretical prediction, as did the system MTF and NPS. The DQE analyzed as a function of kilovolt (peak), exposure, detector threshold, and readout mode revealed important considerations for system optimization. The model also demonstrated the important implications of false counts from both additive electronic noise and charge sharing and highlighted the system design and operational parameters that most affect detector performance in the presence of such factors: for example, increasing the detector threshold from 0 to 100 (arbitrary units of pulse height threshold roughly equivalent to 0.5 and 6 keV energy threshold, respectively), increased the f50 (spatial-frequency at which the MTF falls to a value of
Cascaded systems analysis of photon counting detectors
International Nuclear Information System (INIS)
Xu, J.; Zbijewski, W.; Gang, G.; Stayman, J. W.; Taguchi, K.; Carrino, J. A.; Lundqvist, M.; Fredenberg, E.; Siewerdsen, J. H.
2014-01-01
Purpose: Photon counting detectors (PCDs) are an emerging technology with applications in spectral and low-dose radiographic and tomographic imaging. This paper develops an analytical model of PCD imaging performance, including the system gain, modulation transfer function (MTF), noise-power spectrum (NPS), and detective quantum efficiency (DQE). Methods: A cascaded systems analysis model describing the propagation of quanta through the imaging chain was developed. The model was validated in comparison to the physical performance of a silicon-strip PCD implemented on an experimental imaging bench. The signal response, MTF, and NPS were measured and compared to theory as a function of exposure conditions (70 kVp, 1–7 mA), detector threshold, and readout mode (i.e., the option for coincidence detection). The model sheds new light on the dependence of spatial resolution, charge sharing, and additive noise effects on threshold selection and was used to investigate the factors governing PCD performance, including the fundamental advantages and limitations of PCDs in comparison to energy-integrating detectors (EIDs) in the linear regime for which pulse pileup can be ignored. Results: The detector exhibited highly linear mean signal response across the system operating range and agreed well with theoretical prediction, as did the system MTF and NPS. The DQE analyzed as a function of kilovolt (peak), exposure, detector threshold, and readout mode revealed important considerations for system optimization. The model also demonstrated the important implications of false counts from both additive electronic noise and charge sharing and highlighted the system design and operational parameters that most affect detector performance in the presence of such factors: for example, increasing the detector threshold from 0 to 100 (arbitrary units of pulse height threshold roughly equivalent to 0.5 and 6 keV energy threshold, respectively), increased the f 50 (spatial-frequency at
Cascaded systems analysis of photon counting detectors
Energy Technology Data Exchange (ETDEWEB)
Xu, J.; Zbijewski, W.; Gang, G.; Stayman, J. W. [Department of Biomedical Engineering, Johns Hopkins University, Baltimore, Maryland 21205 (United States); Taguchi, K.; Carrino, J. A. [Russell H. Morgan Department of Radiology, Johns Hopkins University, Baltimore, Maryland 21205 (United States); Lundqvist, M.; Fredenberg, E. [Philips Healthcare, Solna 171 41 (Sweden); Siewerdsen, J. H., E-mail: jeff.siewerdsen@jhu.edu [Department of Biomedical Engineering, Johns Hopkins University, Baltimore, Maryland 21205 (United States); Russell H. Morgan Department of Radiology, Johns Hopkins University, Baltimore, Maryland 21205 (United States)
2014-10-15
Purpose: Photon counting detectors (PCDs) are an emerging technology with applications in spectral and low-dose radiographic and tomographic imaging. This paper develops an analytical model of PCD imaging performance, including the system gain, modulation transfer function (MTF), noise-power spectrum (NPS), and detective quantum efficiency (DQE). Methods: A cascaded systems analysis model describing the propagation of quanta through the imaging chain was developed. The model was validated in comparison to the physical performance of a silicon-strip PCD implemented on an experimental imaging bench. The signal response, MTF, and NPS were measured and compared to theory as a function of exposure conditions (70 kVp, 1–7 mA), detector threshold, and readout mode (i.e., the option for coincidence detection). The model sheds new light on the dependence of spatial resolution, charge sharing, and additive noise effects on threshold selection and was used to investigate the factors governing PCD performance, including the fundamental advantages and limitations of PCDs in comparison to energy-integrating detectors (EIDs) in the linear regime for which pulse pileup can be ignored. Results: The detector exhibited highly linear mean signal response across the system operating range and agreed well with theoretical prediction, as did the system MTF and NPS. The DQE analyzed as a function of kilovolt (peak), exposure, detector threshold, and readout mode revealed important considerations for system optimization. The model also demonstrated the important implications of false counts from both additive electronic noise and charge sharing and highlighted the system design and operational parameters that most affect detector performance in the presence of such factors: for example, increasing the detector threshold from 0 to 100 (arbitrary units of pulse height threshold roughly equivalent to 0.5 and 6 keV energy threshold, respectively), increased the f{sub 50} (spatial
Method and device for X-ray fluorescence analysis
International Nuclear Information System (INIS)
Jagoutz, E.; Palme, C.
1978-01-01
In the x-ray fluorescence analyzer the useful signal can be completely separated from the spurious signals, and especially the pulse can be determined. For this purpose the output of the radiation detector is connected with a multichannel pulse height discriminator. The measured signal determined in the pulse heigth discriminator may be indicated by a visual display or processed by a computer (coincidence circuits). (DG) [de
Analysis of airborne radiometric data. Volume 3. Topical reports
International Nuclear Information System (INIS)
Reed, J.H.; Shreve, D.C.; Sperling, M.; Woolson, W.A.
1978-05-01
This volume consists of four topical reports: a general discussion of the philosophy of unfolding spectra with continuum and discrete components, a mathematical treatment of the effects of various physical parameters on the uncollided gamma-ray spectrum at aircraft elevations, a discussion of the application of the unfolding code MAZNAI to airborne data, and a discussion of the effects of the nonlinear relationship between energy deposited and pulse height in NaI(T1) detectors
International Nuclear Information System (INIS)
Camp, D.C.; Ruhter, W.D.; Benjamin, S.
1979-01-01
Energy-dispersive x-ray fluorescence analysis can be used for quantitative on-line monitoring of the product concentrations in single- or dual-element process streams in a reprocessing plant. The 122-keV gamma ray from 57 Co is used to excite the K x-rays of uranium and/or plutonium in nitric acid solution streams. A collimated HPGe detector is used to measure the excited x-ray intensities. Net solution radioactivity may be measured by eclipsing the exciting radiation, or by measuring it simultaneously with a second detector. The technique is nondestructive and noninvasive, and is easily adapted directly to pipes containing the solution of interest. The dynamic range of the technique extends from below 1 to 500 g/l. Measurement times depend on concentration, but better than 1% counting statistics can be obtained in 100 s for 400 g/l concentrations, and in 1000 s for as little as 10 g/l. Calibration accuracies of 0.3% or better over the entire dynamic range can be achieved easily using carefully prepared standards. Computer-based analysis equipment allows concentration changes in flowing streams to be dynamically monitored. Changes in acid normality of the stream will affect the concentration determined, hence it must also be determined by measuring the intensity of a transmitted 57 Co beam. The computer/disk-based pulse-height analysis system allows all necessary calculations to be done on-line. Experimental requirements for an in-plant installation or a test and evaluation are discussed
Determination of gold and indium in sea water by neutron activation analysis
International Nuclear Information System (INIS)
Tateno, Yukio; Ohta, Naoichi
1979-01-01
A combination of electrodeposition on graphite with neutron activation analysis was used for the determination of gold and indium in sea water. At a potential of -0.70 V vs. the silver-silver chloride electrode, gold and indium were electrolyzed on to a graphite electrode (1.1 cm phi x 0.2 cm) from 100 ml of 0.5 M sodium chloride. Recovery yield of gold was constant at pH from 1 to 3 and was independent of the initial concentration of gold, (0.01 -- 1) ppb. For a 72-h electrolysis at pH 2 the recovery yield of gold was 92%, while that of indium was 32%. The graphite electrode was exposed to a thermal neutron flux of 5.1 x 10 11 or 1.5 x 10 12 n cm -2 s -1 : 5 min exposure for indium and 6 to 12 h for gold. After appropriate decay periods the activities of 198 Au and sup(116m)In were measured for 2000 s and 300 s, respectively, with a 4000-channel pulse-height analyser and a Ge(Li) detector. The total amount of gold in 1 l of a sea water sample (Tokyo Bay) was (0.023 +- 0.001)μg, in which nonelectrolyzable gold was estimated to be 0.005 μg. Indium concentration in the sample was too low to be determined by the present method. Detection limit for indium was 1 ppb. (author)
Energy Technology Data Exchange (ETDEWEB)
Scott, P E; Grau, A
1987-07-01
In this paper pairs of pulse height distribution curves, of C-14 samples, colour-quenched and chemically quenched was obtained. The possibility to choose a counting window in order to obtain the counting efficiency curves, for both type of quenching was studied. (Author) 7 refs.
Energy Technology Data Exchange (ETDEWEB)
Pottier, R; Berger, R [Commissariat a l' Energie Atomique, Centre d' Etudes Nucleaires de Fontenay-aux-Roses, 92 (France)
1966-06-01
A procedure is described for the qualitative and quantitative spectrographic analysis of radioactive sources containing two alpha-gamma emitters having the same alpha energy or two beta-gamma emitters having the same gamma energy. The main apparatus is a multichannel pulse-height analyzer including a coincidence circuit. The principle of the method, the synoptic scheme, the electronic device, the type of sources, and the precautions to be taken or the corrections to take into account are reported. The results obtained in solving the three following problems are discussed as examples of applications of the method: analysis of {sup 241}Am in alpha-gamma sources containing {sup 238}Pu; analysis of {sup 237}Np in beta-gamma sources containing {sup 239}Pu; and analysis of {sup 106}Ru-{sup 106}Rh in beta-gamma sources containing {sup 95}Zr-{sup 95}Nb. (authors) [French] Dans ce. rapport, on presente une methode d'analyse spectrographique qualitative et quantitative de sources radioactives contenant deux emetteurs alpha-gamma de meme energie alpha et deux emetteurs beta-gamma de meme energie gamma. L'organe principal est un analyseur d'amplitude a 400 canaux comprenant un circuit de coincidence. On decrit le principe de la methode, le schema synoptique, l'appareillage, le type des sources, les precautions a prendre ou les corrections a faire. On discute les resultats obtenus dans la solution des trois problemes suivants traites a titre d'application de la methode: 1. analyse d'americium 241 en presence de plutonium 238; 2. analyse de neptunium 237 en presence de plutonium 239; 3. analyse de ruthenium 106-rhodium 106 en presence de zirconium 95-niobium 95. (auteurs)
Rietveld analysis and electronic bands structure on Tc superconductors systems
International Nuclear Information System (INIS)
Aldea, N.; Tiusan, C. V.; Sandu, V.
1999-01-01
A procedure for simultaneous refinement of structural and micro-structural disorder parameters for polycrystalline YBa 2 Cu 3 O 7-x system is proposed. It is based on Rietveld method combined with Fourier analysis for broadened peaks Another purpose of this paper consists in electronic structure determination studied by using the self-consistent Tight Binding Linear Muffin-Tin Orbital Atomic Spheres Approximation TB-LMTO-ASA methods. The Rietveld method uses an analytical function that describes the profiles, usually pseudo-Voigt (pV) or Pearson VII (PVII). The parameters of the analytical profiles describe its amplitude, position and peak shape. The full width at half maximum (FWHM) is supposed to vary with the diffraction angle in agreement with the Caglioti, Paoletti and Ricci's relationship. The best structural parameters are determined in the least squares sense by the minimisation a classical residual using the Marquardt method. In this case, the peak profiles were modelled by the pseudo-Voigt function corrected by the instrumental asymmetry. The physical information obtained are: scale factor, lattice parameters, atomic position and displacements, atomic occupation numbers, temperature factor (isotropy or anisotropy), preferred orientation parameter, crystalline size and micro-strain along different crystallographic directions, distributions of crystallite size and micro-strain functions. This procedure was implemented on computer code and it has a friendly graphical interface based on pull down menus technique. From the experimental point of view the X-ray diffraction data were collected using a horizontal powder diffractometer in the Bragg-Brentano (BB) geometry with a Ni filtered CuKα, λ = 1.54178 A, at room temperature using a DRON 2 set-up. The diffraction profiles were measured with a proportional gas detector, a single channel pulse-height discrimination and a standard associated counting circuit. The electronic band calculations are based on the TB
Pocket PC-based portable gamma-ray spectrometer
Directory of Open Access Journals (Sweden)
Kamontip Ploykrachang
2011-04-01
Full Text Available A portable gamma-ray spectrometer based on a Pocket PC has been developed. A 12-bit pipeline analog-to-digitalconverter (ADC associated with an implemented pulse height histogram function on field programmable gate array (FPGAoperating at 15 MHz is employed for pulse height analysis from built-in pulse amplifier. The system, which interfaces withthe Pocket PC via an enhanced RS-232 serial port under the microcontroller facilitation, is utilized for spectrum acquisition,display and analysis. The pulse height analysis capability of the system was tested and it was found that the ADC integralnonlinearity of ±0.45% was obtained with the throughput rate at 160 kcps. The overall system performance was tested usinga PIN photodiode-CsI(Tl crystal coupled scintillation detector and gamma standard radioactive sources of Cs-137 andCo-60. Low cost and the compact system size as a result of the implemented logical function are also discussed.
Analysis of a neutron scattering integral experiment on iron for neutron energies from 1 to 15 MeV
International Nuclear Information System (INIS)
Cramer, S.N.; Oblow, E.M.
1976-11-01
Monte Carlo calculations were made to analyze the results of an integral experiment with an iron sample to determine the adequacy of neutron scattering cross section data for iron. The experimental results analyzed included energy-dependent NE-213 detector count rates at a scattering angle of 90 deg and pulse-height spectra for scattered neutrons produced in an iron ring pulsed with a 1- to 20-MeV neutron source. The pulse-height data were unfolded to generate secondary neutron spectra at 90 deg as a function of incident neutron energy. Multigroup Monte Carlo calculations using the MORSE code and ENDF/B-IV cross sections were made to analyze all reported results. Discrepancies between calculated and measured responses were found for inelastic scattering reactions in the range from 1 to 4 MeV. These results were related to deficiencies in ENDF/B-IV iron cross section data
International Nuclear Information System (INIS)
Fedorko, I.
2004-01-01
In this work, I present a measurement of the cross section of W production at collider Tevatron times branching fraction for W → eν (sigma(pp'bar' → W) x BF(W → eν)) with electron reconstructed in the forward region of the detector CDF, using combined calorimetric and tracking information. This is the first CDF measurement in the forward rapidity range and is the first step for study of W properties at large η. For the period Run II, started from autumn 2001, was made upgrade of CDF detector. The forward region (for pseudorapidity |η| > 1) was strongly affected by this upgrade. Mainly due to new silicon tracking system and new forward calorimeter. With combination of tracking detectors SVXII and ISL it is now possible to reconstruct '3D' tracks. The analysis starts from calorimeter-based selection criteria finished with sample of W candidates. This selection is followed by matching '3D' tracks (to remove remaining background) with reconstructed cluster in electromagnetic calorimeter, which is measuring not only energy, but also position of electromagnetic object by Preshower detector (part of calorimeter). Besides MET PEM trigger and tracking efficiencies were established as a helpful numbers for other analysis in forward region. The measured value of the σ x BF(W → eν) is 2.874 ± 0.034(stat) ± 0.167(sys) ± 0.172(lum)nb for data sample of integrated luminosity 64 pb -1 , taken from February 2002 until the January 2003 shutdown. Presented value is in agreement with measurements performed by CDF in the central region and with theoretical estimates. Analysis of the photomultiplier (PMT) pulse height spectra from faint light sources (usually called the single-photoelectron spectra) is of a great importance because it reveals many features and can be used to find relevant parameters of PMTs. A deconvolution method is based on a sophisticated photomultiplier response function, which takes into account also photoeffect on first dynode and non
Large Area X-Ray Proportional Counter (LAXPC) Instrument on ...
Indian Academy of Sciences (India)
P. C. Agrawal
2017-06-19
Jun 19, 2017 ... high time resolution X-ray observations in 3–80 keV energy band with moderate energy ... able to detect low-and high-frequency QPOs (up ... black hole binaries will provide a tool to probe .... cessed for pulse height analysis.
Runaway electrons in the TRIAM-1 tokamak
International Nuclear Information System (INIS)
Satoh, Takemichi; Nakamura, Kazuo; Toi, Kazuo; Nakamura, Yukio; Hiraki, Naoji
1981-01-01
Pulse height analysis of soft X-rays is carried out in the TRIAM-1 tokamak. The electron temperatures determined from the soft X-ray spectrum agree well with those from Thomson scattering. It is observed that low-energy runaway (slideaway) electrons appear in the high-current-density discharges. (author)
Soft x-ray measurements in the TRIAM-1 tokamak
Energy Technology Data Exchange (ETDEWEB)
Satoh, T; Toi, K; Nakamura, K; Nakamura, Y; Hiraki, N [Kyushu Univ., Fukuoka (Japan). Research Inst. for Applied Mechanics
1981-07-01
Soft X-ray pulse height analysis system has been designed and constructed for measurements of electron distribution function and impurity with high spatial resolution (0.5 cm) and temporal resolution (2 msec) in the TRIAM-1 tokamak. The experimental results about electron temperature, enhancement factor, Z sub(eff) and runaway electrons are presented and discussed.
Runaway electrons in the TRIAM-1 tokamak
Energy Technology Data Exchange (ETDEWEB)
Satoh, T; Nakamura, K; Toi, K; Nakamura, Y; Hiraki, N [Kyushu Univ., Fukuoka (Japan). Research Inst. for Applied Mechanics
1981-09-01
Pulse height analysis of soft X-rays is carried out in the TRIAM-1 tokamak. The electron temperatures determined from the soft X-ray spectrum agree well with those from Thomson scattering. It is observed that low-energy runaway (slideaway) electrons appear in the high-current-density discharges.
DEFF Research Database (Denmark)
Chin, Catherine; Engelsholm, Rasmus Dybbro; Moselund, Peter Morten
2017-01-01
to 2200 nm, and fast photo detectors, to record 800 consecutive pulses. Peaks from these pulses are first extracted, then distribution of their pulse height histogram (PHH) is constructed. Analysis using higher-order moments about the mean (variance, skewness and kurtosis) showed that: (1) around the pump...
Teaching aspects of the instrumental neutron activation analysis advance prediction computer program
International Nuclear Information System (INIS)
Guinn, V.P.
1990-01-01
The INAA APCP was developed, and is extensively used, primarily as a very useful guide for efficient INAA work. From the exact or approximate elemental composition of a material, it calculates the proper sample weight, and all details of the germanium gamma-ray pulse height spectrum for any and all input flux, detector, irradiation time, decay time, and counting time conditions specified. For each set of conditions, it also prints out the five largest induced activities and the five largest gamma-ray emitters at five different decay periods. The program also provides an excellent educational device for students in a radiochemistry course
International Nuclear Information System (INIS)
Seidel, K.; Freiesleben, H.; Poenitz, E.; Klix, A.; Unholzer, S.; Batistoni, P.; Fischer, U.; Leichtle, D.
2006-01-01
The nuclear parameters of a breeding blanket, such as tritium production rate, nuclear heating, activation and dose rate, are calculated by integral folding of an energy dependent cross section (or coefficient) with the neutron (or gamma-ray) flux energy spectra. The uncertainties of the designed parameters are determined by the uncertainties of both the cross section data and the flux spectra obtained by transport calculations. Also the analysis of possible discrepancies between measured and calculated integral nuclear parameter represents a two-step procedure. First, the energy region and the amount of flux discrepancies has to be found out and second, the cross section data have to be checked. To this end, neutron and gamma-ray flux spectra in a mock-up of the EU Helium-Cooled Pebble Bed (HCPB) breeder Test Blanket Module (TBM), irradiated with 14 MeV neutrons, were measured and analysed by means of Monte Carlo transport calculations. The flux spectra were determined for the energy ranges that are relevant for the most important nuclear parameters of the TBM, which are the tritium production rate and the shielding capability. The fast neutron flux which determines the tritium production on 7 Li and dominates the shield design was measured by the pulse-height distribution obtained from an organic liquid scintillation detector. Simultaneously, the gamma-ray flux spectra were measured. The neutron flux at lower energies, down to thermal, which determines the tritium production on 6 Li, was measured with time-of-arrival spectroscopy. For this purpose, the TUD neutron generator was operated in pulsed mode (pulse width 10 μs, frequency 1 kHz) and the neutrons arriving at a 3 He proportional counter in the mock-up were recorded as a function of time after the source neutron pulse. The spectral distributions for the two positions in the mock-up, where measurements were carried out, were calculated with the Monte Carlo code MCNP, version 5, and nuclear data from the
Analysis of the propagation of neutrons and gamma-rays from the fast neutron source reactor YAYOI
Energy Technology Data Exchange (ETDEWEB)
Yoshida, Shigeo, E-mail: neutron@keyaki.cc.u-tokai.ac.jp [Department of Energy Science and Engineering, School of Engineering, Tokai University, Hiratsuka, Kanagawa 259-1292 (Japan); Murata, Isao [Division of Electrical, Electronic and Information Engineering, Osaka University, Suita, Osaka 565-0871 (Japan); Nakagawa, Tsutomu; Saito, Isao [Nuclear Professional School, School of Engineering, The University of Tokyo, Tokai-mura, Naka-gun, Ibaraki 319-1188 (Japan)
2011-10-01
The skyshine effect is crucial for designing appropriate shielding. To investigate the skyshine effect, the propagation of neutrons was measured and analyzed at the fast neutron source reactor YAYOI. Pulse height spectra and dose distributions of neutron and secondary gamma-ray were measured outside YAYOI, and analyzed with MCNP-5 and JENDL-3.3. Comparison with the experimental results showed good agreement. Also, a semi-empirical formula was successfully derived to describe the dose distribution. The formulae can be used to predict the skyshine effect at YAYOI, and will be useful for estimating the skyshine effect and designing the shield structure for fusion facilities.
Dual parameter analysis of CsI:Tl/PMT with a digital oscilloscope
Devol, T A; Fjeld, R A
1999-01-01
Scintillation pulses from alpha-particle and gamma-ray excited CsI:Tl were recorded with a photomultiplier tube and digital oscilloscope card in single parameter (pulse shape or pulse height) and dual parameter (simultaneous pulse shape and pulse height) modes. A 2.8x1.3x1.3 cm sup 3 CsI:Tl crystal was coupled to a Burle 8850 photomultiplier tube (PMT) and excited with sup 1 sup 4 sup 8 Gd (E subalpha=3.18 MeV) and sup 1 sup 3 sup 7 Cs (E subgamma=0.662 MeV). The timing pulses (50 OMEGA terminated dynode signal) from the PMT base were directly input to a GaGe CompuScope 8012A/PCI sup T sup M (12-bit, 50 MHz dual channel digital oscilloscope card) that resided in a P6-300 MHz personal computer. Individual digitized pulses were processed via a LabVIEW sup T sup M V 4.1 interface to the oscilloscope. The ratio of the charge collected over a short time interval to the total charge collected was used for the single parameter pulse shape spectrum while the total charge collected was used for the single parameter pu...
{sup 3}He detector analysis of some special shielding materials
Energy Technology Data Exchange (ETDEWEB)
Avdic, S; Pesic, M [Boris Kidric, Institute of Nuclear Sciences, Beograd (Yugoslavia); Marinkovic, P [ETF Belgrade Univ. (Yugoslavia)
1990-07-01
The shielding properties of commercial materials of reactor Experiments, Inc. (R/X) were analyzed at the facility which includes bare heavy water experimental reactor RB with external neutron converter ENC, The fast neutron spectrum measurements in energy range from 1 MeV to 10 MeV was performed using ORTEC semiconductor neutron detector with He{sup 3} in diode coincidence arrangement. The neutron spectra have been evaluated from measured pulse-height distribution using numerical code HE3 for computation of detector efficiency in a collimated neutron beam. The neutron dose rates behind ENC with and without sample R/X material were determined using cubic spline interpolation routine for calculating the corresponding flux-dose rate conversion factors. Satisfactory shielding properties of the examined material in a fast neutron field in measurements and calculations are demonstrated. (author)
Liquid Scintillation High Resolution Spectral Analysis
Energy Technology Data Exchange (ETDEWEB)
Grau Carles, A.; Grau Malonda, A.
2010-08-06
The CIEMAT/NIST and the TDCR methods in liquid scintillation counting are based on the determination of the efficiency for total counting. This paper tries to expand these methods analysing the pulse-height spectrum of radionuclides. To reach this objective we have to generalize the equations used in the model and to analyse the influence of ionization and chemical quench in both spectra and counting efficiency. We present equations to study the influence of different photomultipliers response in systems with one, two or three photomultipliers. We study the effect of the electronic noise discriminator level in both spectra and counting efficiency. The described method permits one to study problems that up to now was not possible to approach, such as the high uncertainty in the standardization of pure beta-ray emitter with low energy when we apply the TDCR method, or the discrepancies in the standardization of some electron capture radionuclides, when the CIEMAT/NIST method is applied. (Author) 107 refs.
Liquid Scintillation High Resolution Spectral Analysis
International Nuclear Information System (INIS)
Grau Carles, A.; Grau Malonda, A.
2010-01-01
The CIEMAT/NIST and the TDCR methods in liquid scintillation counting are based on the determination of the efficiency for total counting. This paper tries to expand these methods analysing the pulse-height spectrum of radionuclides. To reach this objective we have to generalize the equations used in the model and to analyse the influence of ionization and chemical quench in both spectra and counting efficiency. We present equations to study the influence of different photomultipliers response in systems with one, two or three photomultipliers. We study the effect of the electronic noise discriminator level in both spectra and counting efficiency. The described method permits one to study problems that up to now was not possible to approach, such as the high uncertainty in the standardization of pure beta-ray emitter with low energy when we apply the TDCR method, or the discrepancies in the standardization of some electron capture radionuclides, when the CIEMAT/NIST method is applied. (Author) 107 refs.
International Nuclear Information System (INIS)
von Goeler, S.; Hill, K.W.; Bitter, M.
1982-12-01
A short description of the x-ray diagnostic preparation for the TFTR tokamak is given. The x-ray equipment consists of the limiter x-ray monitoring system, the soft x-ray pulse-height-analysis-system, the soft x-ray imaging system and the x-ray crystal spectrometer. Particular attention is given to the radiation protection of the x-ray systems from the neutron environment
Multi parametric system for the acquisition and processing of nuclear data on a personal computer
International Nuclear Information System (INIS)
Toledo Acosta, R. B.; Osorio Deliz, J. F.; Arista Romeu, E.; Perez Sanchez, R.; Lopez Torres, E.
1997-01-01
A four-parameter Multi parametric System for the acquisition and processing of nuclear data is described. It is characterized for its flexibility and relatively low cost, also guaranteeing a high acquisition capacity. The system allows to be utilized in a multi parametric manner, in pulse height analysis or in any combination of both for parameter. The hardware and the software of the system are described. A general explanation of the operation and the characteristics of the system is offered. (author) [es
A probabilistic model for x-ray PHA data
International Nuclear Information System (INIS)
Diesso, M.; Hill, K.
1986-01-01
In this paper, a mathematical model of the data produced by a single-arm x-ray pulse height analyzer (PHA) system is developed. Given an assumption on the electron temperature and density profiles, a maximum likelihood technique is applied to calculate the peak electron temperature and enhancement factor of the plasma. This method is currently being used in the analysis of x-ray data from the tokamak fusion test reactor (TFTR); sample results are presented
International Nuclear Information System (INIS)
Eisler, P.; Youl, S.; Lwin, T.; Nelson, G.
1983-01-01
Simultaneous fitting of peaks and background functions from gamma-ray spectrometry using multichannel pulse height analysis is considered. The specific case of Gaussian peak and exponential background is treated in detail with respect to simultaneous estimation of both functions by using a technique which incorporates maximum likelihood method as well as a graphical method. Theoretical expressions for the standard errors of the estimates are also obtained. The technique is demonstrated for two experimental data sets. (orig.)
A peak value searching method of the MCA based on digital logic devices
International Nuclear Information System (INIS)
Sang Ziru; Huang Shanshan; Chen Lian; Jin Ge
2010-01-01
Digital multi-channel analyzers play a more important role in multi-channel pulse height analysis technique. The direction of digitalization are characterized by powerful pulse processing ability, high throughput, improved stability and flexibility. This paper introduces a method of searching peak value of waveform based on digital logic with FPGA. This method reduce the dead time. Then data correction offline can improvement the non-linearity of MCA. It gives the α energy spectrum of 241 Am. (authors)
Investigation of runaway electrons in the PRETEXT Tokamak
International Nuclear Information System (INIS)
Eckstrand, S.A.
1981-01-01
High energy (0.2 to 0.4 MeV) runaway electrons have been studied in PRETEXT discharges by detecting the hard x-ray bremsstrahlung radiation produced when they escape from the discharge and strike the limiter. A pulse height analysis system, which included pileup rejection circuitry because of the high count rate, recorded both the amplitude and arrival time of each pulse
A microcomputer based multi parametric system for nuclear data acquisition and processing
International Nuclear Information System (INIS)
Toledo Acosta B, Rene; Osorio Deliz F, Juan; Arista Romeu, Eduardo; Perez Sanchez, Reinaldo; Lopes Torres, E.
1997-01-01
A four-parameter Multi parametric System for the acquisition and processing of nuclear data is described. It is characterized for its flexibility and relatively low cost, also guaranteeing a high acquisition capacity. The system allows to be utilized in a multi parametric manner, in pulse height analysis or in many combination of both for parameter. It is described the hardware and the software of the system
A New Waveform Signal Processing Method Based on Adaptive Clustering-Genetic Algorithms
International Nuclear Information System (INIS)
Noha Shaaban; Fukuzo Masuda; Hidetsugu Morota
2006-01-01
We present a fast digital signal processing method for numerical analysis of individual pulses from CdZnTe compound semiconductor detectors. Using Maxi-Mini Distance Algorithm and Genetic Algorithms based discrimination technique. A parametric approach has been used for classifying the discriminated waveforms into a set of clusters each has a similar signal shape with a corresponding pulse height spectrum. A corrected total pulse height spectrum was obtained by applying a normalization factor for the full energy peak for each cluster with a highly improvements in the energy spectrum characteristics. This method applied successfully for both simulated and real measured data, it can be applied to any detector suffers from signal shape variation. (authors)
Applications of Boron Loaded Scintillating Fibers as NDA Tools for Nuclear Safeguards
International Nuclear Information System (INIS)
Mayo, D.R.; Ensslin, N.; Grazioso, R.F.; Heger, A.S.; Mercer, D.J.; Miller, M.C.; Russo, P.A.; Sweet, M.R.
1997-01-01
Nuclear safeguards and nonproliferation rely on nondestructive analytical tools for prompt and noninvasive detection, verification, and quantitative analysis of nuclear materials in demanding environments. A new tool based on the detection of correlated neutrons in narrow time windows is being investigated to fill the niche created by the current limitations of the existing methods based on polyethylene moderated 3 He gas proportional tubes. Commercially produced Boron-loaded ( 10 B) plastic scintillating fibers are one such technology under consideration. The fibers can be configured in a system to have high efficiency, short neutron die-away, pulse height sensitivity, and mechanical flexibility. Various configurations of the fibers with high density polyethylene have been considered which calculationally result in high efficiency detectors with short die-away times. A discussion of the design considerations and calculations of the detector efficiency, die-away time, and simulated pulse height spectra along with preliminary test results are presented
Applications of boron-loaded scintillating fibers as NDA tools for nuclear safeguards
International Nuclear Information System (INIS)
Mayo, Douglas R.; Ensslin, Norbert; Mercer, David J.; Miller, Michael C.; Russo, Phyllis A.; Sweet, Martin R.; Grazioso, Ronald F.; Heger, A. Sharif
1998-01-01
Nuclear safeguards and nonproliferation rely on nondestructive analytical tools for prompt and noninvasive detection, verification, and quantitative analysis of nuclear materials in demanding environments. A new tool based on the detection of correlated neutrons in narrow time windows is being investigated to fill the niche created by the current limitations of the existing methods based on polyethylene moderated 3 He gas proportional tubes. Commercially produced Boron-loaded ( 10 B) plastic scintillating fibers are one such technology under consideration. The fibers can be configured in a system to have high efficiency, short neutron die-away, pulse height sensitivity, and mechanical flexibility. Various configurations of the fibers with high density polyethylene have been considered which calculationally result in high efficiency detectors with short die-away times. A discussion of the design considerations and calculations of the detector efficiency, die-away time, and simulated pulse height spectra along with preliminary test results are presented
International Nuclear Information System (INIS)
Saito, T.; Kiwamoto, Y.; Honda, T.; Kasugai, A.; Kurihara, K.; Miyoshi, S.
1991-01-01
An apparatus for the measurement of high-energy electrons (10--500 keV) with a silicon surface-barrier detector is described. The apparatus has special features. In particular, a fast CAMAC transient digitizer is used to directly record the wave form of a pulse train from the detector and then pulse heights are analyzed with a computer instead of on a conventional pulse height analyzer. With this method the system is capable of detecting electrons with a count rate as high as ∼300--400 kilocounts/s without serious deterioration of performance. Moreover, piled up signals are reliably eliminated from analysis. The system has been applied to measure electron-cyclotron-resonance-heating-induced end loss electrons in the GAMMA 10 tandem mirror and has yielded information relating to electron heating and diffusion in velocity space
Decision analysis multicriteria analysis
International Nuclear Information System (INIS)
Lombard, J.
1986-09-01
The ALARA procedure covers a wide range of decisions from the simplest to the most complex one. For the simplest one the engineering judgement is generally enough and the use of a decision aiding technique is therefore not necessary. For some decisions the comparison of the available protection option may be performed from two or a few criteria (or attributes) (protection cost, collective dose,...) and the use of rather simple decision aiding techniques, like the Cost Effectiveness Analysis or the Cost Benefit Analysis, is quite enough. For the more complex decisions, involving numerous criteria or for decisions involving large uncertainties or qualitative judgement the use of these techniques, even the extended cost benefit analysis, is not recommended and appropriate techniques like multi-attribute decision aiding techniques are more relevant. There is a lot of such particular techniques and it is not possible to present all of them. Therefore only two broad categories of multi-attribute decision aiding techniques will be presented here: decision analysis and the outranking analysis
The application of a microstrip gas counter to energy-dispersive x-ray fluorescence analysis
International Nuclear Information System (INIS)
Veloso, J.F.C.A.; Santos, J.M.F. dos; Conde, C.A.N.
1996-01-01
Performance characteristics of a microstrip gas counter operated as a x-ray fluorescence spectrometer are reported. Gas amplification as a function of microstrip anode-cathode voltage was measured, and the breakdown threshold voltage was determined in pure xenon. The detector temporal stability and the effect of gas purity were assessed. Energy resolution and linearity, detection efficiency, and uniformity of spatial response in the 2- to 60-keV x-ray energy range were determined from the pulse-height distributions of the fluorescence x-ray spectra induced in a variety of single- and multi-element sample materials. Energy resolution similar to conventional proportional counters was achieved at 6 keV
X-ray analysis of electron Bernstein wave heating in MST
Energy Technology Data Exchange (ETDEWEB)
Seltzman, A. H., E-mail: seltzman@wisc.edu; Anderson, J. K.; DuBois, A. M.; Almagri, A.; Forest, C. B. [Department of Physics, University of Wisconsin, Madison, Wisconsin 53706 (United States)
2016-11-15
A pulse height analyzing x-ray tomography system has been developed to detect x-rays from electron Bernstein wave heated electrons in the Madison symmetric torus reversed field pinch (RFP). Cadmium zinc telluride detectors are arranged in a parallel beam array with two orthogonal multi-chord detectors that may be used for tomography. In addition a repositionable 16 channel fan beam camera with a 55° field of view is used to augment data collected with the Hard X-ray array. The chord integrated signals identify target emission from RF heated electrons striking a limiter located 12° toroidally away from the RF injection port. This provides information on heated electron spectrum, transport, and diffusion. RF induced x-ray emission from absorption on harmonic electron cyclotron resonances in low current (<250 kA) RFP discharges has been observed.
Gamma, x-ray reduction system. Volume I: gamma
International Nuclear Information System (INIS)
Zimmer, W.H.
1976-01-01
The starting premises for this data reduction system were (a) the individual researcher needs all the accuracy that can be achieved but he has neither the time nor the inclination to learn how to achieve it, and (b) if the data reduction system is to be centralized the people operating it will want to minimize conversation with the computer. This is a working system. All spectral data are stored on Data General 4234 discs after background normalization and strip. Storage is initiated from magnetic tapes loaded by detached pulse height analyzers or directly from Scorpio pulse height analyzers. The only restrictions placed on the individual researchers are that the pulse height analyzer energy scale be set up consistently, that a recovery standard be run at least once every day of use, and the total acquisition system be calibrated to its range of use. In many instances, and if desirable, the latter is provided as a service. At the time of writing this gamma data reduction system is actively being used to calculate net peak areas, activities with or without time correction, activations analysis results, counting precisions, and dynamic limits of detection for the spectral data output of 17 detached pulse height analyzers. To all modes of data reduction are applied background subtraction, random summing correction, detector recovery factor correction, peak interfernce correction (second-order product interference for activation analysis), geometry function correction, acquisition time decay corrections, external and internal sorber correction. All of this is accomplished and a customer report typed in a readable format after a halfline of noninteractive instruction
International Nuclear Information System (INIS)
2008-05-01
This book introduces energy and resource technology development business with performance analysis, which has business division and definition, analysis of current situation of support, substance of basic plan of national energy, resource technique development, selection of analysis index, result of performance analysis by index, performance result of investigation, analysis and appraisal of energy and resource technology development business in 2007.
Modeling crosstalk in silicon photomultipliers
International Nuclear Information System (INIS)
Gallego, L; Rosado, J; Blanco, F; Arqueros, F
2013-01-01
Optical crosstalk seriously limits the photon-counting resolution of silicon photomultipliers. In this work, realistic analytical models to describe the crosstalk effects on the response of these photodetectors are presented and compared with experimental data. The proposed models are based on the hypothesis that each pixel of the array has a finite number of available neighboring pixels to excite via crosstalk. Dead-time effects and geometrical aspects of the propagation of crosstalk between neighbors are taken into account in the models for different neighborhood configurations. Simple expressions to account for crosstalk effects on the pulse-height spectrum as well as to evaluate the excess noise factor due to crosstalk are also given. Dedicated measurements were carried out under both dark-count conditions and pulsed illumination. Moreover, the influence of afterpulsing on the measured pulse-height spectrum was studied, and a measurement of the recovery time of pixels was reported. High-resolution pulse-height spectra were obtained by means of a detailed waveform analysis, and the results have been used to validate our crosstalk models.
White-light Detection for Nanoparticle Sizing with the TSI Ultrafine Condensation Particle Counter
International Nuclear Information System (INIS)
Dick, William D.; McMurry, Peter H.; Weber, Rodney J.; Quant, Frederick R.
2000-01-01
Several of the most common methods for measuring nanoparticle size distributions employ the ultrafine condensation particle counter (UCPC) for detection purposes. Among these methods, the pulse height analysis (PHA) technique, in which the optical response of the UCPC detector is related to initial particle diameter in the 3-10 nm range, prevails in applications where fast sampling is required or for which concentrations of nanoparticles are frequently very low. With the PHA technique, white light is required for particle illumination in order to obtain a monotonic relationship between initial particle diameter and optical response (pulse height). However, the popular, commercially available TSI Model 3025A UCPC employs a laser for particle detection. Here, we report on a novel white-light detection system developed for the 3025A UCPC that involves minimal alteration to the instrument and preserves normal counting operation. Performance is illustrated with pulse height spectra produced by differential mobility analyzer (DMA) - generated calibration aerosols in the 3-50 nm range
Neutron and photon spectrometry with liquid scintillation detectors in mixed fields
Klein, H
2002-01-01
Liquid scintillation detectors of type NE213 or BC501A are well suited and routinely used for spectrometry in mixed n-gamma-fields. Neutron- and photon-induced pulse height spectra may be simultaneously recorded making use of the n/gamma-discrimination capability based on pulse shape analysis. The light output functions for the detected secondary charged particles, i.e. electrons, positrons, protons and other charged reaction products, and the pulse height resolution function must carefully be determined. This can be done experimentally, in part via an iterative procedure by comparison with calculations. The response functions can then be reliably calculated by Monte Carlo simulations. Photon response functions calculated with the PHRESP code, which was developed on the basis of the EGS4+PRESTA program package, are in very good agreement with calibrations up to 17 MeV, both in shape and absolute scale. Similarly, neutron response functions calculated with the NRESP7 code well describe the pulse height spectra...
YAP scintillators for resonant detection of epithermal neutrons at pulsed neutron sources
International Nuclear Information System (INIS)
Tardocchi, M.; Gorini, G.; Pietropaolo, A.; Andreani, C.; Senesi, R.; Rhodes, N.; Schooneveld, E. M.
2004-01-01
Recent studies indicate the resonance detector (RD) technique as an interesting approach for neutron spectroscopy in the electron volt energy region. This work summarizes the results of a series of experiments where RD consisting of YAlO 3 (YAP) scintillators were used to detect scattered neutrons with energy in the range 1-200 eV. The response of YAP scintillators to radiative capture γ emission from a 238 U analyzer foil was characterized in a series of experiments performed on the VESUVIO spectrometer at the ISIS pulsed neutron source. In these experiments a biparametric data acquisition allowed the simultaneous measurements of both neutron time-of-flight and γ pulse height (energy) spectra. The analysis of the γ pulse height and neutron time of flight spectra permitted to identify and distinguish the signal and background components. These measurements showed that a significant improvement in the signal-to-background ratio can be achieved by setting a lower level discrimination on the pulse height at about 600 keV equivalent photon energy. Present results strongly indicate YAP scintillators as the ideal candidate for neutron scattering studies with epithermal neutrons at both very low (<5 deg.) and intermediate scattering angles
International Nuclear Information System (INIS)
Jae, Myeong Gi; Lee, Won Seong; Kim, Ha Hyeok
1989-02-01
This book give description of electronic engineering such as circuit element and device, circuit analysis and logic digital circuit, the method of electrochemistry like conductometry, potentiometry and current measuring, spectro chemical analysis with electromagnetic radiant rays, optical components, absorption spectroscopy, X-ray analysis, atomic absorption spectrometry and reference, chromatography like gas-chromatography and liquid-chromatography and automated analysis on control system evaluation of automated analysis and automated analysis system and reference.
Energy Technology Data Exchange (ETDEWEB)
Jae, Myeong Gi; Lee, Won Seong; Kim, Ha Hyeok
1989-02-15
This book give description of electronic engineering such as circuit element and device, circuit analysis and logic digital circuit, the method of electrochemistry like conductometry, potentiometry and current measuring, spectro chemical analysis with electromagnetic radiant rays, optical components, absorption spectroscopy, X-ray analysis, atomic absorption spectrometry and reference, chromatography like gas-chromatography and liquid-chromatography and automated analysis on control system evaluation of automated analysis and automated analysis system and reference.
Energy Technology Data Exchange (ETDEWEB)
Kim, Seung Jae; Seo, Seong Gyu
1995-03-15
This textbook deals with instrumental analysis, which consists of nine chapters. It has Introduction of analysis chemistry, the process of analysis and types and form of the analysis, Electrochemistry on basic theory, potentiometry and conductometry, electromagnetic radiant rays and optical components on introduction and application, Ultraviolet rays and Visible spectrophotometry, Atomic absorption spectrophotometry on introduction, flame emission spectrometry and plasma emission spectrometry. The others like infrared spectrophotometry, X-rays spectrophotometry and mass spectrometry, chromatography and the other instrumental analysis like radiochemistry.
International Nuclear Information System (INIS)
Kim, Seung Jae; Seo, Seong Gyu
1995-03-01
This textbook deals with instrumental analysis, which consists of nine chapters. It has Introduction of analysis chemistry, the process of analysis and types and form of the analysis, Electrochemistry on basic theory, potentiometry and conductometry, electromagnetic radiant rays and optical components on introduction and application, Ultraviolet rays and Visible spectrophotometry, Atomic absorption spectrophotometry on introduction, flame emission spectrometry and plasma emission spectrometry. The others like infrared spectrophotometry, X-rays spectrophotometry and mass spectrometry, chromatography and the other instrumental analysis like radiochemistry.
Neutron activation analysis of arsenic and antimony in human hair
International Nuclear Information System (INIS)
Kanda, Yukio; Isono, Hideo; Kozuka, Hiroshi.
1975-01-01
A radiochemical neutron activation method for the determination of trace amounts of arsenic and antimony in human hair samples is studied. The sample of hair (100 mg) irradiated for 5 hours with a neutron flux of 2.1x10 12 n/cm 2 s was decomposed with a sulfuric-nitric acid mixture after addition of each 5 mg of arsenic and antimony as carrier. Arsine and stibine were evolved from the solution of decomposed hair by reduction with 3 g of granular zinc and were absorbed in 0.1N iodine solution for half an hour. Metal arsenic was separated from iodine solution by precipitation with sodium hypophosphite, followed by precipitation of antimony as sulfide with thioacetamide. These precipitates were dissolved and their gamma-ray spectra were measured with a well type 3''x3'' NaI(TI) detector equipped with a 200 channel pulse-height analyzer. After the measurement of gamma-ray spectra, the chemical yields were determined by colorimetric methods. The relative standard deviations were 7% and 4% for 0.01 μg As and 0.024 μg Sb, respectively. The sensitivity of this method was estimated to be 1x10 -3 μg for arsenic and 2x10 -3 μg for antimony. (auth.)
... page: //medlineplus.gov/ency/article/003741.htm Sensitivity analysis To use the sharing features on this page, please enable JavaScript. Sensitivity analysis determines the effectiveness of antibiotics against microorganisms (germs) ...
McShane, Edward James
2013-01-01
This text surveys practical elements of real function theory, general topology, and functional analysis. Discusses the maximality principle, the notion of convergence, the Lebesgue-Stieltjes integral, function spaces and harmonic analysis. Includes exercises. 1959 edition.
Cerebrospinal fluid analysis ... Analysis of CSF can help detect certain conditions and diseases. All of the following can be, but ... An abnormal CSF analysis result may be due to many different causes, ... Encephalitis (such as West Nile and Eastern Equine) Hepatic ...
... analysis URL of this page: //medlineplus.gov/ency/article/003627.htm Semen analysis To use the sharing features on this page, please enable JavaScript. Semen analysis measures the amount and quality of a man's semen and sperm. Semen is ...
Kantorovich, L V
1982-01-01
Functional Analysis examines trends in functional analysis as a mathematical discipline and the ever-increasing role played by its techniques in applications. The theory of topological vector spaces is emphasized, along with the applications of functional analysis to applied analysis. Some topics of functional analysis connected with applications to mathematical economics and control theory are also discussed. Comprised of 18 chapters, this book begins with an introduction to the elements of the theory of topological spaces, the theory of metric spaces, and the theory of abstract measure space
International Nuclear Information System (INIS)
Berman, M.; Bischof, L.M.; Breen, E.J.; Peden, G.M.
1994-01-01
This paper provides an overview of modern image analysis techniques pertinent to materials science. The usual approach in image analysis contains two basic steps: first, the image is segmented into its constituent components (e.g. individual grains), and second, measurement and quantitative analysis are performed. Usually, the segmentation part of the process is the harder of the two. Consequently, much of the paper concentrates on this aspect, reviewing both fundamental segmentation tools (commonly found in commercial image analysis packages) and more advanced segmentation tools. There is also a review of the most widely used quantitative analysis methods for measuring the size, shape and spatial arrangements of objects. Many of the segmentation and analysis methods are demonstrated using complex real-world examples. Finally, there is a discussion of hardware and software issues. 42 refs., 17 figs
Thiemann, Francis C.
Semiotic analysis is a method of analyzing signs (e.g., words) to reduce non-numeric data to their component parts without losing essential meanings. Semiotics dates back to Aristotle's analysis of language; it was much advanced by nineteenth-century analyses of style and logic and by Whitehead and Russell's description in this century of the role…
Indian Academy of Sciences (India)
Dimensional analysis is a useful tool which finds important applications in physics and engineering. It is most effective when there exist a maximal number of dimensionless quantities constructed out of the relevant physical variables. Though a complete theory of dimen- sional analysis was developed way back in 1914 in a.
Bravená, Helena
2009-01-01
This bacherlor thesis deals with the importance of job analysis for personnel activities in the company. The aim of this work is to find the most suitable method of job analysis in a particular enterprise, and continues creating descriptions and specifications of each job.
International Nuclear Information System (INIS)
Francois, P.
1996-01-01
We undertook a study programme at the end of 1991. To start with, we performed some exploratory studies aimed at learning some preliminary lessons on this type of analysis: Assessment of the interest of probabilistic incident analysis; possibility of using PSA scenarios; skills and resources required. At the same time, EPN created a working group whose assignment was to define a new approach for analysis of incidents on NPPs. This working group gave thought to both aspects of Operating Feedback that EPN wished to improve: Analysis of significant incidents; analysis of potential consequences. We took part in the work of this group, and for the second aspects, we proposed a method based on an adaptation of the event-tree method in order to establish a link between existing PSA models and actual incidents. Since PSA provides an exhaustive database of accident scenarios applicable to the two most common types of units in France, they are obviously of interest for this sort of analysis. With this method we performed some incident analyses, and at the same time explores some methods employed abroad, particularly ASP (Accident Sequence Precursor, a method used by the NRC). Early in 1994 EDF began a systematic analysis programme. The first, transient phase will set up methods and an organizational structure. 7 figs
Energy Technology Data Exchange (ETDEWEB)
Francois, P
1997-12-31
We undertook a study programme at the end of 1991. To start with, we performed some exploratory studies aimed at learning some preliminary lessons on this type of analysis: Assessment of the interest of probabilistic incident analysis; possibility of using PSA scenarios; skills and resources required. At the same time, EPN created a working group whose assignment was to define a new approach for analysis of incidents on NPPs. This working group gave thought to both aspects of Operating Feedback that EPN wished to improve: Analysis of significant incidents; analysis of potential consequences. We took part in the work of this group, and for the second aspects, we proposed a method based on an adaptation of the event-tree method in order to establish a link between existing PSA models and actual incidents. Since PSA provides an exhaustive database of accident scenarios applicable to the two most common types of units in France, they are obviously of interest for this sort of analysis. With this method we performed some incident analyses, and at the same time explores some methods employed abroad, particularly ASP (Accident Sequence Precursor, a method used by the NRC). Early in 1994 EDF began a systematic analysis programme. The first, transient phase will set up methods and an organizational structure. 7 figs.
International Nuclear Information System (INIS)
Dutra, C.V.; Gomes, C.B.
1990-01-01
This work is about the X-ray fluorescence aplication in geology. It's showing the X-ray origin and excitation. About the instrumentation this work shows the following: X-ray tubes, colimators, analysers crystals, detectors, amplifiers, pulse height selector, and others electronic components. By X-ray fluorescente are done quantitative and qualitative geological analysis and this work shows this analysis and its detection limits. The problems determination is the example. In this work was done yet the comparative analysis of the various instrumental methods in geochemistry. (C.G.) [pt
Khabaza, I M
1960-01-01
Numerical Analysis is an elementary introduction to numerical analysis, its applications, limitations, and pitfalls. Methods suitable for digital computers are emphasized, but some desk computations are also described. Topics covered range from the use of digital computers in numerical work to errors in computations using desk machines, finite difference methods, and numerical solution of ordinary differential equations. This book is comprised of eight chapters and begins with an overview of the importance of digital computers in numerical analysis, followed by a discussion on errors in comput
International Nuclear Information System (INIS)
Warner, M.
1987-01-01
What is the current state of quantitative trace analytical chemistry? What are today's research efforts? And what challenges does the future hold? These are some of the questions addressed at a recent four-day symposium sponsored by the National Bureau of Standards (NBS) entitled Accuracy in Trace Analysis - Accomplishments, Goals, Challenges. The two plenary sessions held on the first day of the symposium reviewed the history of quantitative trace analysis, discussed the present situation from academic and industrial perspectives, and summarized future needs. The remaining three days of the symposium consisted of parallel sessions dealing with the measurement process; quantitation in materials; environmental, clinical, and nutrient analysis; and advances in analytical techniques
Goodstein, R L
2010-01-01
Recursive analysis develops natural number computations into a framework appropriate for real numbers. This text is based upon primary recursive arithmetic and presents a unique combination of classical analysis and intuitional analysis. Written by a master in the field, it is suitable for graduate students of mathematics and computer science and can be read without a detailed knowledge of recursive arithmetic.Introductory chapters on recursive convergence and recursive and relative continuity are succeeded by explorations of recursive and relative differentiability, the relative integral, and
Tao, Terence
2016-01-01
This is part one of a two-volume book on real analysis and is intended for senior undergraduate students of mathematics who have already been exposed to calculus. The emphasis is on rigour and foundations of analysis. Beginning with the construction of the number systems and set theory, the book discusses the basics of analysis (limits, series, continuity, differentiation, Riemann integration), through to power series, several variable calculus and Fourier analysis, and then finally the Lebesgue integral. These are almost entirely set in the concrete setting of the real line and Euclidean spaces, although there is some material on abstract metric and topological spaces. The book also has appendices on mathematical logic and the decimal system. The entire text (omitting some less central topics) can be taught in two quarters of 25–30 lectures each. The course material is deeply intertwined with the exercises, as it is intended that the student actively learn the material (and practice thinking and writing ri...
Tao, Terence
2016-01-01
This is part two of a two-volume book on real analysis and is intended for senior undergraduate students of mathematics who have already been exposed to calculus. The emphasis is on rigour and foundations of analysis. Beginning with the construction of the number systems and set theory, the book discusses the basics of analysis (limits, series, continuity, differentiation, Riemann integration), through to power series, several variable calculus and Fourier analysis, and then finally the Lebesgue integral. These are almost entirely set in the concrete setting of the real line and Euclidean spaces, although there is some material on abstract metric and topological spaces. The book also has appendices on mathematical logic and the decimal system. The entire text (omitting some less central topics) can be taught in two quarters of 25–30 lectures each. The course material is deeply intertwined with the exercises, as it is intended that the student actively learn the material (and practice thinking and writing ri...
International Nuclear Information System (INIS)
1988-01-01
Basic studies in nuclear analytical techniques include the examination of underlying assumptions and the development and extention of techniques involving the use of ion beams for elemental and mass analysis. 1 ref., 1 tab
Energy Technology Data Exchange (ETDEWEB)
2016-06-01
Fact sheet summarizing NREL's techno-economic analysis and life-cycle assessment capabilities to connect research with future commercial process integration, a critical step in the scale-up of biomass conversion technologies.
Gasinski, Leszek
2005-01-01
Hausdorff Measures and Capacity. Lebesgue-Bochner and Sobolev Spaces. Nonlinear Operators and Young Measures. Smooth and Nonsmooth Analysis and Variational Principles. Critical Point Theory. Eigenvalue Problems and Maximum Principles. Fixed Point Theory.
DEFF Research Database (Denmark)
Bauer-Gottwein, Peter; Riegels, Niels; Pulido-Velazquez, Manuel
2017-01-01
Hydroeconomic analysis and modeling provides a consistent and quantitative framework to assess the links between water resources systems and economic activities related to water use, simultaneously modeling water supply and water demand. It supports water managers and decision makers in assessing...... trade-offs between different water uses, different geographic regions, and various economic sectors and between the present and the future. Hydroeconomic analysis provides consistent economic performance criteria for infrastructure development and institutional reform in water policies and management...... organizations. This chapter presents an introduction to hydroeconomic analysis and modeling, and reviews the state of the art in the field. We review available economic water-valuation techniques and summarize the main types of decision problems encountered in hydroeconomic analysis. Popular solution strategies...
Schiffrin, Deborah
1990-01-01
Summarizes the current state of research in conversation analysis, referring primarily to six different perspectives that have developed from the philosophy, sociology, anthropology, and linguistics disciplines. These include pragmatics; speech act theory; interactional sociolinguistics; ethnomethodology; ethnography of communication; and…
Gorsuch, Richard L
2013-01-01
Comprehensive and comprehensible, this classic covers the basic and advanced topics essential for using factor analysis as a scientific tool in psychology, education, sociology, and related areas. Emphasizing the usefulness of the techniques, it presents sufficient mathematical background for understanding and sufficient discussion of applications for effective use. This includes not only theory but also the empirical evaluations of the importance of mathematical distinctions for applied scientific analysis.
International Nuclear Information System (INIS)
1959-01-01
Radioactivation analysis is the technique of radioactivation analysis of the constituents of a very small sample of matter by making the sample artificially radioactive. The first stage is to make the sample radioactive by artificial means, e.g. subject it to neutron bombardment. Once the sample has been activated, or made radioactive, the next task is to analyze the radiations given off by the sample. This analysis would indicate the nature and quantities of the various elements present in the sample. The reason is that the radiation from a particular radioisotope. In 1959 a symposium on 'Radioactivation Analysis' was organized in Vienna by the IAEA and the Joint Commission on Applied Radioactivity (ICSU). It was pointed out that there are certain factors creating uncertainties and elaborated how to overcome them. Attention was drawn to the fact that radioactivation analysis had proven a powerful tool tackling fundamental problems in geo- and cosmochemistry, and a review was given of the recent work in this field. Because of its extreme sensitivity radioactivation analysis had been principally employed for trace detection and its most extensive use has been in control of semiconductors and very pure metals. An account of the experience gained in the USA was given, where radioactivation analysis was being used by many investigators in various scientific fields as a practical and useful tool for elemental analyses. Much of this work had been concerned with determining sub microgramme and microgramme concentration of many different elements in samples of biological materials, drugs, fertilizers, fine chemicals, foods, fuels, glass, ceramic materials, metals, minerals, paints, petroleum products, resinous materials, soils, toxicants, water and other materials. In addition to these studies, radioactivation analysis had been used by other investigators to determine isotopic ratios of the stable isotopes of some of the elements. Another paper dealt with radioactivation
Energy Technology Data Exchange (ETDEWEB)
NONE
1959-07-15
Radioactivation analysis is the technique of radioactivation analysis of the constituents of a very small sample of matter by making the sample artificially radioactive. The first stage is to make the sample radioactive by artificial means, e.g. subject it to neutron bombardment. Once the sample has been activated, or made radioactive, the next task is to analyze the radiations given off by the sample. This analysis would indicate the nature and quantities of the various elements present in the sample. The reason is that the radiation from a particular radioisotope. In 1959 a symposium on 'Radioactivation Analysis' was organized in Vienna by the IAEA and the Joint Commission on Applied Radioactivity (ICSU). It was pointed out that there are certain factors creating uncertainties and elaborated how to overcome them. Attention was drawn to the fact that radioactivation analysis had proven a powerful tool tackling fundamental problems in geo- and cosmochemistry, and a review was given of the recent work in this field. Because of its extreme sensitivity radioactivation analysis had been principally employed for trace detection and its most extensive use has been in control of semiconductors and very pure metals. An account of the experience gained in the USA was given, where radioactivation analysis was being used by many investigators in various scientific fields as a practical and useful tool for elemental analyses. Much of this work had been concerned with determining sub microgramme and microgramme concentration of many different elements in samples of biological materials, drugs, fertilizers, fine chemicals, foods, fuels, glass, ceramic materials, metals, minerals, paints, petroleum products, resinous materials, soils, toxicants, water and other materials. In addition to these studies, radioactivation analysis had been used by other investigators to determine isotopic ratios of the stable isotopes of some of the elements. Another paper dealt with radioactivation
DEFF Research Database (Denmark)
Brænder, Morten; Andersen, Lotte Bøgh
2014-01-01
Based on our 2013-article, ”Does Deployment to War Affect Soldiers' Public Service Motivation – A Panel Study of Soldiers Before and After their Service in Afghanistan”, we present Panel Analysis as a methodological discipline. Panels consist of multiple units of analysis, observed at two or more...... in research settings where it is not possible to distribute units of analysis randomly or where the independent variables cannot be manipulated. The greatest disadvantage in regard to using panel studies is that data may be difficult to obtain. This is most clearly vivid in regard to the use of panel surveys...... points in time. In comparison with traditional cross-sectional studies, the advantage of using panel studies is that the time dimension enables us to study effects. Whereas experimental designs may have a clear advantage in regard to causal inference, the strength of panel studies is difficult to match...
Loeb, Peter A
2016-01-01
This textbook is designed for a year-long course in real analysis taken by beginning graduate and advanced undergraduate students in mathematics and other areas such as statistics, engineering, and economics. Written by one of the leading scholars in the field, it elegantly explores the core concepts in real analysis and introduces new, accessible methods for both students and instructors. The first half of the book develops both Lebesgue measure and, with essentially no additional work for the student, general Borel measures for the real line. Notation indicates when a result holds only for Lebesgue measure. Differentiation and absolute continuity are presented using a local maximal function, resulting in an exposition that is both simpler and more general than the traditional approach. The second half deals with general measures and functional analysis, including Hilbert spaces, Fourier series, and the Riesz representation theorem for positive linear functionals on continuous functions with compact support....
Scott, L Ridgway
2011-01-01
Computational science is fundamentally changing how technological questions are addressed. The design of aircraft, automobiles, and even racing sailboats is now done by computational simulation. The mathematical foundation of this new approach is numerical analysis, which studies algorithms for computing expressions defined with real numbers. Emphasizing the theory behind the computation, this book provides a rigorous and self-contained introduction to numerical analysis and presents the advanced mathematics that underpin industrial software, including complete details that are missing from most textbooks. Using an inquiry-based learning approach, Numerical Analysis is written in a narrative style, provides historical background, and includes many of the proofs and technical details in exercises. Students will be able to go beyond an elementary understanding of numerical simulation and develop deep insights into the foundations of the subject. They will no longer have to accept the mathematical gaps that ex...
Rao, G Shanker
2006-01-01
About the Book: This book provides an introduction to Numerical Analysis for the students of Mathematics and Engineering. The book is designed in accordance with the common core syllabus of Numerical Analysis of Universities of Andhra Pradesh and also the syllabus prescribed in most of the Indian Universities. Salient features: Approximate and Numerical Solutions of Algebraic and Transcendental Equation Interpolation of Functions Numerical Differentiation and Integration and Numerical Solution of Ordinary Differential Equations The last three chapters deal with Curve Fitting, Eigen Values and Eigen Vectors of a Matrix and Regression Analysis. Each chapter is supplemented with a number of worked-out examples as well as number of problems to be solved by the students. This would help in the better understanding of the subject. Contents: Errors Solution of Algebraic and Transcendental Equations Finite Differences Interpolation with Equal Intervals Interpolation with Unequal Int...
Jacques, Ian
1987-01-01
This book is primarily intended for undergraduates in mathematics, the physical sciences and engineering. It introduces students to most of the techniques forming the core component of courses in numerical analysis. The text is divided into eight chapters which are largely self-contained. However, with a subject as intricately woven as mathematics, there is inevitably some interdependence between them. The level of difficulty varies and, although emphasis is firmly placed on the methods themselves rather than their analysis, we have not hesitated to include theoretical material when we consider it to be sufficiently interesting. However, it should be possible to omit those parts that do seem daunting while still being able to follow the worked examples and to tackle the exercises accompanying each section. Familiarity with the basic results of analysis and linear algebra is assumed since these are normally taught in first courses on mathematical methods. For reference purposes a list of theorems used in the t...
DiBenedetto, Emmanuele
2016-01-01
The second edition of this classic textbook presents a rigorous and self-contained introduction to real analysis with the goal of providing a solid foundation for future coursework and research in applied mathematics. Written in a clear and concise style, it covers all of the necessary subjects as well as those often absent from standard introductory texts. Each chapter features a “Problems and Complements” section that includes additional material that briefly expands on certain topics within the chapter and numerous exercises for practicing the key concepts. The first eight chapters explore all of the basic topics for training in real analysis, beginning with a review of countable sets before moving on to detailed discussions of measure theory, Lebesgue integration, Banach spaces, functional analysis, and weakly differentiable functions. More topical applications are discussed in the remaining chapters, such as maximal functions, functions of bounded mean oscillation, rearrangements, potential theory, a...
Electron temperature diagnostics in the RFX reversed field pinch experiment
International Nuclear Information System (INIS)
Bartiromo, R.; Carraro, L.; Marrelli, L.; Murari, A.; Pasqualotto, R.; Puiatti, M.E.; Scarin, P.; Valisa, M.; Franz, P.; Martin, P.; Zabeo, L.
2000-01-01
The paper presents an integrated approach to the problem of electron temperature diagnostics of the plasma in a reversed field pinch. Three different methods, sampling different portions of the electron distribution function, are adopted, namely Thomson scattering, soft X-ray spectroscopy by pulse-height analysis and filtered soft X-ray intensity ratio. A careful analysis of the different sources of systematic errors is performed and a novel statistical approach is adopted to mutually validate the three independent measurements. A satisfactory agreement is obtained over a large range of experimental conditions, indicating that in the plasma core the energy distribution function is well represented by a maxwellian. (author)
VHDL implementation on histogram with ADC CAMAC module
International Nuclear Information System (INIS)
Ruby Santhi, R.; Satyanarayana, V.V.V.; Ajith Kumar, B.P.
2007-01-01
Modern nuclear spectroscopy systems the data acquisition and analysis in experimental science have been undergoing major changes because of faster speed and higher resolution. The CAMAC module which is described here is FPGA based 8K x 24 bit Histogram Memory integrated with ADC on a single board has been designed and fabricated. This module accepts input from Spectroscopy Amplifier for Pulse Height Analysis and offers all features single spectra for a few selected parameters. These on line histograms are to monitor the progress of the experiments during on line experiments
International Nuclear Information System (INIS)
Romli
1997-01-01
Cluster analysis is the name of group of multivariate techniques whose principal purpose is to distinguish similar entities from the characteristics they process.To study this analysis, there are several algorithms that can be used. Therefore, this topic focuses to discuss the algorithms, such as, similarity measures, and hierarchical clustering which includes single linkage, complete linkage and average linkage method. also, non-hierarchical clustering method, which is popular name K -mean method ' will be discussed. Finally, this paper will be described the advantages and disadvantages of every methods
Rockafellar, Ralph Tyrell
2015-01-01
Available for the first time in paperback, R. Tyrrell Rockafellar's classic study presents readers with a coherent branch of nonlinear mathematical analysis that is especially suited to the study of optimization problems. Rockafellar's theory differs from classical analysis in that differentiability assumptions are replaced by convexity assumptions. The topics treated in this volume include: systems of inequalities, the minimum or maximum of a convex function over a convex set, Lagrange multipliers, minimax theorems and duality, as well as basic results about the structure of convex sets and
Brezinski, C
2012-01-01
Numerical analysis has witnessed many significant developments in the 20th century. This book brings together 16 papers dealing with historical developments, survey papers and papers on recent trends in selected areas of numerical analysis, such as: approximation and interpolation, solution of linear systems and eigenvalue problems, iterative methods, quadrature rules, solution of ordinary-, partial- and integral equations. The papers are reprinted from the 7-volume project of the Journal of Computational and Applied Mathematics on '/homepage/sac/cam/na2000/index.html<
International Nuclear Information System (INIS)
Biehl, F.A.
1984-05-01
This paper presents the criteria, previous nuclear experience in space, analysis techniques, and possible breakup enhancement devices applicable to an acceptable SP-100 reentry from space. Reactor operation in nuclear-safe orbit will minimize the radiological risk; the remaining safeguards criteria need to be defined. A simple analytical point mass reentry technique and a more comprehensive analysis method that considers vehicle dynamics and orbit insertion malfunctions are presented. Vehicle trajectory, attitude, and possible breakup enhancement devices will be integrated in the simulation as required to ensure an adequate representation of the reentry process
Aggarwal, Charu C
2013-01-01
With the increasing advances in hardware technology for data collection, and advances in software technology (databases) for data organization, computer scientists have increasingly participated in the latest advancements of the outlier analysis field. Computer scientists, specifically, approach this field based on their practical experiences in managing large amounts of data, and with far fewer assumptions- the data can be of any type, structured or unstructured, and may be extremely large.Outlier Analysis is a comprehensive exposition, as understood by data mining experts, statisticians and
Everitt, Brian S; Leese, Morven; Stahl, Daniel
2011-01-01
Cluster analysis comprises a range of methods for classifying multivariate data into subgroups. By organizing multivariate data into such subgroups, clustering can help reveal the characteristics of any structure or patterns present. These techniques have proven useful in a wide range of areas such as medicine, psychology, market research and bioinformatics.This fifth edition of the highly successful Cluster Analysis includes coverage of the latest developments in the field and a new chapter dealing with finite mixture models for structured data.Real life examples are used throughout to demons
Snell, K S; Langford, W J; Maxwell, E A
1966-01-01
Elementary Analysis, Volume 2 introduces several of the ideas of modern mathematics in a casual manner and provides the practical experience in algebraic and analytic operations that lays a sound foundation of basic skills. This book focuses on the nature of number, algebraic and logical structure, groups, rings, fields, vector spaces, matrices, sequences, limits, functions and inverse functions, complex numbers, and probability. The logical structure of analysis given through the treatment of differentiation and integration, with applications to the trigonometric and logarithmic functions, is
International Nuclear Information System (INIS)
Baron, J.H.; Nunez McLeod, J.; Rivera, S.S.
1997-01-01
This book contains a selection of research works performed in the CEDIAC Institute (Cuyo National University) in the area of Risk Analysis, with specific orientations to the subjects of uncertainty and sensitivity studies, software reliability, severe accident modeling, etc. This volume presents important material for all those researches who want to have an insight in the risk analysis field, as a tool to solution several problems frequently found in the engineering and applied sciences field, as well as for the academic teachers who want to keep up to date, including the new developments and improvements continuously arising in this field [es
Alan Gallegos
2002-01-01
Watershed analyses and assessments for the Kings River Sustainable Forest Ecosystems Project were done on about 33,000 acres of the 45,500-acre Big Creek watershed and 32,000 acres of the 85,100-acre Dinkey Creek watershed. Following procedures developed for analysis of cumulative watershed effects (CWE) in the Pacific Northwest Region of the USDA Forest Service, the...
Freund, Rudolf J; Sa, Ping
2006-01-01
The book provides complete coverage of the classical methods of statistical analysis. It is designed to give students an understanding of the purpose of statistical analyses, to allow the student to determine, at least to some degree, the correct type of statistical analyses to be performed in a given situation, and have some appreciation of what constitutes good experimental design
International Nuclear Information System (INIS)
Unterberger, A.
1987-01-01
We study the Klein-Gordon symbolic calculus of operators acting on solutions of the free Klein-Gordon equation. It contracts to the Weyl calculus as c→∞. Mathematically, it may also be considered as a pseudodifferential analysis on the unit ball of R n [fr
International Nuclear Information System (INIS)
Woodard, K.
1985-01-01
The objectives of this paper are to: Provide a realistic assessment of consequences; Account for plant and site-specific characteristics; Adjust accident release characteristics to account for results of plant-containment analysis; Produce conditional risk curves for each of five health effects; and Estimate uncertainties
DEFF Research Database (Denmark)
Hjørland, Birger
2017-01-01
The domain-analytic approach to knowledge organization (KO) (and to the broader field of library and information science, LIS) is outlined. The article reviews the discussions and proposals on the definition of domains, and provides an example of a domain-analytic study in the field of art studies....... Varieties of domain analysis as well as criticism and controversies are presented and discussed....
International Nuclear Information System (INIS)
Rhoades, W.A.; Dray, B.J.
1970-01-01
The effect of Gadolinium-155 on the prompt kinetic behavior of a zirconium hydride reactor has been deduced, using experimental data from the SNAPTRAN machine. The poison material makes the temperature coefficient more positive, and the Type IV sleeves were deduced to give a positive coefficient above 1100 0 F. A thorough discussion of the data and analysis is included. (U.S.)
International Nuclear Information System (INIS)
Saadi, Radouan; Marah, Hamid
2014-01-01
This report presents results related to Tritium analysis carried out at the CNESTEN DASTE in Rabat (Morocco), on behalf of Senegal, within the framework of the RAF7011 project. It describes analytical method and instrumentation including general uncertainty estimation: Electrolytic enrichment and liquid scintillation counting; The results are expressed in Tritium Unit (TU); Low Limit Detection: 0.02 TU
Miller, Rupert G
2011-01-01
A concise summary of the statistical methods used in the analysis of survival data with censoring. Emphasizes recently developed nonparametric techniques. Outlines methods in detail and illustrates them with actual data. Discusses the theory behind each method. Includes numerous worked problems and numerical exercises.
Koornneef, M.; Alonso-Blanco, C.; Stam, P.
2006-01-01
The Mendelian analysis of genetic variation, available as induced mutants or as natural variation, requires a number of steps that are described in this chapter. These include the determination of the number of genes involved in the observed trait's variation, the determination of dominance
DEFF Research Database (Denmark)
Nielsen, Kirsten
2010-01-01
The first part of this article presents the characteristics of Hebrew poetry: features associated with rhythm and phonology, grammatical features, structural elements like parallelism, and imagery and intertextuality. The second part consists of an analysis of Psalm 121. It is argued that assonan...
Adrian Ioana; Tiberiu Socaciu
2013-01-01
The article presents specific aspects of management and models for economic analysis. Thus, we present the main types of economic analysis: statistical analysis, dynamic analysis, static analysis, mathematical analysis, psychological analysis. Also we present the main object of the analysis: the technological activity analysis of a company, the analysis of the production costs, the economic activity analysis of a company, the analysis of equipment, the analysis of labor productivity, the anal...
International Nuclear Information System (INIS)
Smith, M.; Jones, D.R.
1991-01-01
The goal of exploration is to find reserves that will earn an adequate rate of return on the capital invested. Neither exploration nor economics is an exact science. The authors must therefore explore in those trends (plays) that have the highest probability of achieving this goal. Trend analysis is a technique for organizing the available data to make these strategic exploration decisions objectively and is in conformance with their goals and risk attitudes. Trend analysis differs from resource estimation in its purpose. It seeks to determine the probability of economic success for an exploration program, not the ultimate results of the total industry effort. Thus the recent past is assumed to be the best estimate of the exploration probabilities for the near future. This information is combined with economic forecasts. The computer software tools necessary for trend analysis are (1) Information data base - requirements and sources. (2) Data conditioning program - assignment to trends, correction of errors, and conversion into usable form. (3) Statistical processing program - calculation of probability of success and discovery size probability distribution. (4) Analytical processing - Monte Carlo simulation to develop the probability distribution of the economic return/investment ratio for a trend. Limited capital (short-run) effects are analyzed using the Gambler's Ruin concept in the Monte Carlo simulation and by a short-cut method. Multiple trend analysis is concerned with comparing and ranking trends, allocating funds among acceptable trends, and characterizing program risk by using risk profiles. In summary, trend analysis is a reality check for long-range exploration planning
DEFF Research Database (Denmark)
The 19th Scandinavian Conference on Image Analysis was held at the IT University of Copenhagen in Denmark during June 15-17, 2015. The SCIA conference series has been an ongoing biannual event for more than 30 years and over the years it has nurtured a world-class regional research and development...... area within the four participating Nordic countries. It is a regional meeting of the International Association for Pattern Recognition (IAPR). We would like to thank all authors who submitted works to this year’s SCIA, the invited speakers, and our Program Committee. In total 67 papers were submitted....... The topics of the accepted papers range from novel applications of vision systems, pattern recognition, machine learning, feature extraction, segmentation, 3D vision, to medical and biomedical image analysis. The papers originate from all the Scandinavian countries and several other European countries...
Helson, Henry
2010-01-01
This second edition has been enlarged and considerably rewritten. Among the new topics are infinite product spaces with applications to probability, disintegration of measures on product spaces, positive definite functions on the line, and additional information about Weyl's theorems on equidistribution. Topics that have continued from the first edition include Minkowski's theorem, measures with bounded powers, idempotent measures, spectral sets of bounded functions and a theorem of Szego, and the Wiener Tauberian theorem. Readers of the book should have studied the Lebesgue integral, the elementary theory of analytic and harmonic functions, and the basic theory of Banach spaces. The treatment is classical and as simple as possible. This is an instructional book, not a treatise. Mathematics students interested in analysis will find here what they need to know about Fourier analysis. Physicists and others can use the book as a reference for more advanced topics.
Bray, Hubert L; Mazzeo, Rafe; Sesum, Natasa
2015-01-01
This volume includes expanded versions of the lectures delivered in the Graduate Minicourse portion of the 2013 Park City Mathematics Institute session on Geometric Analysis. The papers give excellent high-level introductions, suitable for graduate students wishing to enter the field and experienced researchers alike, to a range of the most important areas of geometric analysis. These include: the general issue of geometric evolution, with more detailed lectures on Ricci flow and Kähler-Ricci flow, new progress on the analytic aspects of the Willmore equation as well as an introduction to the recent proof of the Willmore conjecture and new directions in min-max theory for geometric variational problems, the current state of the art regarding minimal surfaces in R^3, the role of critical metrics in Riemannian geometry, and the modern perspective on the study of eigenfunctions and eigenvalues for Laplace-Beltrami operators.
Freitag, Eberhard
2005-01-01
The guiding principle of this presentation of ``Classical Complex Analysis'' is to proceed as quickly as possible to the central results while using a small number of notions and concepts from other fields. Thus the prerequisites for understanding this book are minimal; only elementary facts of calculus and algebra are required. The first four chapters cover the essential core of complex analysis: - differentiation in C (including elementary facts about conformal mappings) - integration in C (including complex line integrals, Cauchy's Integral Theorem, and the Integral Formulas) - sequences and series of analytic functions, (isolated) singularities, Laurent series, calculus of residues - construction of analytic functions: the gamma function, Weierstrass' Factorization Theorem, Mittag-Leffler Partial Fraction Decomposition, and -as a particular highlight- the Riemann Mapping Theorem, which characterizes the simply connected domains in C. Further topics included are: - the theory of elliptic functions based on...
International Nuclear Information System (INIS)
Quinn, C.A.
1983-01-01
The article deals with spectrographic analysis and the analytical methods based on it. The theory of spectrographic analysis is discussed as well as the layout of a spectrometer system. The infrared absorption spectrum of a compound is probably its most unique property. The absorption of infrared radiation depends on increasing the energy of vibration and rotation associated with a covalent bond. The infrared region is intrinsically low in energy thus the design of infrared spectrometers is always directed toward maximising energy throughput. The article also considers atomic absorption - flame atomizers, non-flame atomizers and the source of radiation. Under the section an emission spectroscopy non-electrical energy sources, electrical energy sources and electrical flames are discussed. Digital computers form a part of the development on spectrographic instrumentation
Cheng, Lizhi; Luo, Yong; Chen, Bo
2014-01-01
This book could be divided into two parts i.e. fundamental wavelet transform theory and method and some important applications of wavelet transform. In the first part, as preliminary knowledge, the Fourier analysis, inner product space, the characteristics of Haar functions, and concepts of multi-resolution analysis, are introduced followed by a description on how to construct wavelet functions both multi-band and multi wavelets, and finally introduces the design of integer wavelets via lifting schemes and its application to integer transform algorithm. In the second part, many applications are discussed in the field of image and signal processing by introducing other wavelet variants such as complex wavelets, ridgelets, and curvelets. Important application examples include image compression, image denoising/restoration, image enhancement, digital watermarking, numerical solution of partial differential equations, and solving ill-conditioned Toeplitz system. The book is intended for senior undergraduate stude...
International Nuclear Information System (INIS)
Hwang, Hun
2007-02-01
This book explains potentiometry, voltametry, amperometry and basic conception of conductometry with eleven chapters. It gives the specific descriptions on electrochemical cell and its mode, basic conception of electrochemical analysis on oxidation-reduction reaction, standard electrode potential, formal potential, faradaic current and faradaic process, mass transfer and overvoltage, potentiometry and indirect potentiometry, polarography with TAST, normal pulse and deferential pulse, voltammetry, conductometry and conductometric titration.
International Nuclear Information System (INIS)
Badwe, R.A.
1999-01-01
The primary endpoint in the majority of the studies has been either disease recurrence or death. This kind of analysis requires a special method since all patients in the study experience the endpoint. The standard method for estimating such survival distribution is Kaplan Meier method. The survival function is defined as the proportion of individuals who survive beyond certain time. Multi-variate comparison for survival has been carried out with Cox's proportional hazard model
DEFF Research Database (Denmark)
Andersen, Lars
This book contains the lecture notes for the 9th semester course on elastodynamics. The first chapter gives an overview of the basic theory of stress waves propagating in viscoelastic media. In particular, the effect of surfaces and interfaces in a viscoelastic material is studied, and different....... Thus, in Chapter 3, an alternative semi-analytic method is derived, which may be applied for the analysis of layered half-spaces subject to moving or stationary loads....
Mucha, Hans-Joachim; Sofyan, Hizir
2000-01-01
As an explorative technique, duster analysis provides a description or a reduction in the dimension of the data. It classifies a set of observations into two or more mutually exclusive unknown groups based on combinations of many variables. Its aim is to construct groups in such a way that the profiles of objects in the same groups are relatively homogenous whereas the profiles of objects in different groups are relatively heterogeneous. Clustering is distinct from classification techniques, ...
International Nuclear Information System (INIS)
Garbarino, J.R.; Steinheimer, T.R.; Taylor, H.E.
1985-01-01
This is the twenty-first biennial review of the inorganic and organic analytical chemistry of water. The format of this review differs somewhat from previous reviews in this series - the most recent of which appeared in Analytical Chemistry in April 1983. Changes in format have occurred in the presentation of material concerning review articles and the inorganic analysis of water sections. Organic analysis of water sections are organized as in previous reviews. Review articles have been compiled and tabulated in an Appendix with respect to subject, title, author(s), citation, and number of references cited. The inorganic water analysis sections are now grouped by constituent using the periodic chart; for example, alkali, alkaline earth, 1st series transition metals, etc. Within these groupings the references are roughly grouped by instrumental technique; for example, spectrophotometry, atomic absorption spectrometry, etc. Multiconstituent methods for determining analytes that cannot be grouped in this manner are compiled into a separate section sorted by instrumental technique. References used in preparing this review were compiled from nearly 60 major journals published during the period from October 1982 through September 1984. Conference proceedings, most foreign journals, most trade journals, and most government publications are excluded. References cited were obtained using the American Chemical Society's Chemical Abstracts for sections on inorganic analytical chemistry, organic analytical chemistry, water, and sewage waste. Cross-references of these sections were also included. 860 references
Energy Technology Data Exchange (ETDEWEB)
None
1980-06-01
The Energy Policy and Conservation Act (EPCA) mandated that minimum energy efficiency standards be established for classes of refrigerators and refrigerator-freezers, freezers, clothes dryers, water heaters, room air conditioners, home heating equipment, kitchen ranges and ovens, central air conditioners, and furnaces. EPCA requires that standards be designed to achieve the maximum improvement in energy efficiency that is technologically feasible and economically justified. Following the introductory chapter, Chapter Two describes the methodology used in the economic analysis and its relationship to legislative criteria for consumer product efficiency assessment; details how the CPES Value Model systematically compared and evaluated the economic impacts of regulation on the consumer, manufacturer and Nation. Chapter Three briefly displays the results of the analysis and lists the proposed performance standards by product class. Chapter Four describes the reasons for developing a baseline forecast, characterizes the baseline scenario from which regulatory impacts were calculated and summarizes the primary models, data sources and assumptions used in the baseline formulations. Chapter Five summarizes the methodology used to calculate regulatory impacts; describes the impacts of energy performance standards relative to the baseline discussed in Chapter Four. Also discussed are regional standards and other program alternatives to performance standards. Chapter Six describes the procedure for balancing consumer, manufacturer, and national impacts to select standard levels. Details of models and data bases used in the analysis are included in Appendices A through K.
The DEIS high energy muon spectrometer. II. The data acquisition system
International Nuclear Information System (INIS)
Allkofer, O.C.; Dau, W.D.; Faehnders, E.; Jokisch, H.; Kaleschke, G.P.; Klemke, G.; Sauerland, K.; Schmidtke, G.; Uhr, R.C.; Bella, G.; Oren, Y.; Virni, U.; Seidman, A.
1977-01-01
The whole spectrometer is read out and controlled on-line via a CAMAC-system by a minicomputer. The magnetostrictive read out signals of 66 magnetostrictive read out wands of the wire spark chambers are digitized by 20-MHz-scalers which can store up to 8 sparks per chamber. The time-of-flight of the muon, the pulse heights of the scintillation counters, the time of event are also recorded. The on-line-computer makes reliability checks of the data and stores them together with monitor data about magnetic field, gas and high voltage system, etc. on magnetic tape for off-line analysis. (author)
Multifunctional data acquisition system based on USB and FPGA
International Nuclear Information System (INIS)
Huang Tuchen; Gong Hui; Shao Beibei
2013-01-01
A multifunctional data acquisition system based on USB and FPGA was developed. The system has four analog inputs digitalized by fast ADC. Based on flexibility of FPGA, different functions can be implemented such as waveform sampling, pulse counting, multi-channel pulse height analysis, and charge division readout process. The hardware communicates with host PC via USB interface. The Labview based user soft ware initializes the hardware, configures the running parameters, reads and processes the data as well as displays the result online. (authors)
Computer-controlled back scattering and sputtering-experiment using a heavy-ion-accelerator
International Nuclear Information System (INIS)
Becker, H.; Birnbaum, M.; Degenhardt, K.H.; Mertens, P.; Tschammer, V.
1978-12-01
Control and data acquisition of a PDP 11/40 computer and CAMAC instrumentation are reported for an experiment that has been developed to measure sputtering in yields and energy losses for heavy 100 - 300 keV ions in thin metal foils. Besides a quadrupole mass filter or a bending magnet, a multichannel analyser is coupled to the computer, so that also pulse height analysis can be performed under computer control. CAMAC instrumentation and measuring programs are built in a modular form to enable an easy application to other experimental problems. (orig.) 891 KBE/orig. 892 BRE
The effect of ICRF antenna phasing on metal impurities in TFTR
International Nuclear Information System (INIS)
Stevens, J.E.; Bush, C.; Colestock, P.L.; Oak Ridge National Lab., TN; AN Ukrainskoj SSR, Kharkov
1989-07-01
ICRF power levels of up to 2.8 MW were achieved during the 1988 experimental run on TFTR. Metal impurity concentrations (Ti, Cr, Fe, Ni) and Z eff were monitored during ICRF heating by x-ray pulse height analysis and uv spectroscopy. Antenna phasing was the key variable affecting ICRF performance. No increase in metallic impurities was observed for P rf approx lt 2.8 MW with the antenna straps 0-Π, while a measurable increase in titanium (Faraday screen material) was observed for P rf approx gt 1.0 MW with 0-0 phasing. 18 refs., 8 figs
Detection of explosives by neutron scattering
International Nuclear Information System (INIS)
Brooks, F.D.; Buffler, A.; Allie, M.S.; Nchodu, M.R.; Bharuth-Ram, K.
1998-01-01
For non-intrusive detection of hidden explosives or other contraband such as narcotics a fast neutron scattering analysis (FNSA) technique is proposed. An experimental arrangement uses a collimated, pulsed beam of neutrons directed at the sample. Scattered neutrons are detected by liquid scintillation counters at different scattering angles. A scattering signature is derived from two-parameter data, counts vs pulse height and time-of-flight measured for each element (H, C, N or O) at each of two scattering angles and two neutron energies. The elemental signatures are very distinctive and constitute a good response matrix for unfolding elemental components from the scattering signatures measured for different compounds
International Nuclear Information System (INIS)
Veshchev, E. A.; Ozaki, T.; Goncharov, P. R.; Sudo, S.
2006-01-01
A new multichannel diagnostic for fast ion distribution studies has been developed and successfully tested on the Large Helical Device (LHD) in different plasma heating conditions. The diagnostic is based on a linear array AXUV detector consisting of 20 segments, charge sensitive preamplifiers, and a set of pulse height analysis channels. The main advantage of this system is the possibility to make time, energy, and angle-resolved measurements of charge exchange neutral particles in a single plasma discharge. This feature makes the new diagnostic a very helpful and powerful tool intended to contribute to the understanding of fast ion behavior in a complex helical plasma geometry like the one of LHD
High rate amplifier-digitizer system for liquid argon calorimeters
International Nuclear Information System (INIS)
Droege, T.F.; Lobkowicz, F.; Fukushima, Y.
1978-01-01
A low-cost charge amplifier for a liquid argon photon detector and a new method for pulse height analysis are described. This scheme is suitable for high-energy photon detection with high counting rate. Samples of preamplifer output are taken just before and just after the arrival of the charge from the detector. The difference of these samples provides a stable pedestal and rejects low frequency noise. Short two-pulse resolving time (approximately equal to 200ns) is achieved. 6 refs
Advanced instrumentation for nuclear monitoring
International Nuclear Information System (INIS)
Armantrout, G.; McGibbon, A.; Swierkowski, S.; Sherohman, J.; Yee, J.
1975-01-01
Research on semiconductor radiation detectors is described. Computational models to calculate the energy band structure, carrier mobility, and carrier lifetime of proposed detector materials are presented; and a computer spectrum simulation that accurately predicts the potential performance of the materials as detectors is given. Also, a self contained, field-portable spectrometer for laboratory-grade pulse-height analysis of gamma-ray spectra suitable for use under extreme environmental conditions and isolated locations by personnel not trained in electronics is reported. (auth)
A high resolution 16 k multi-channel analyzer PC add-on card
International Nuclear Information System (INIS)
Kulkarni, C.P.; Paulson, Molly; Vaidya, P.P.
2001-01-01
This paper describes the system details of a 16 K channel resolution Multi-Channel Analyzer (MCA) developed at Electronics Division, BARC, which is used in high resolution nuclear spectroscopy systems for pulse height analysis. The high resolution data acquisition PC add-on card is architectured using a state of the art digital circuit design technology which makes use of a Field Programmable Gate Array (FPGA), and some of the most modern and advanced analog counterparts like low power, high speed and high precision comparators, Op-amps, ADCs and DACs etc. The 16 K MCA card gives an economic, compact, and low power alternative for nuclear pulse spectroscopy use. (author)
A method for synthesizing response functions of NaI detectors to gamma rays
International Nuclear Information System (INIS)
Sie, S.H.
1978-08-01
A simple method of parametrizing the response function of NaI detectors to gamma rays is described, based on decomposition of the pulse-height spectrum into components associated with the actual detection processes. Smooth dependence of the derived parameters on the gamma-ray energy made it possible to generate a lineshape for any gamma-ray energy by suitable interpolation techniques. The method is applied in analysis of spectra measured with a 7.6 x 7.6 cm NaI detector in continuum gamma-ray study following (HI,xn) reaction
Application of NaI(Tl) scintillation detector to measurement of tritium concentration
International Nuclear Information System (INIS)
Nishizawa, Kazushige; Endo, Yoshio; Shinagawa, Mutsuaki
1977-01-01
Measurements of tritium concentration in various media have been successfully performed by detecting with an ordinary NaI(Tl) scintillation counter the bremsstrahlung emitted from tritium decay. To distinguish meaningful signals from noises emanating mainly from thermoelectrons present in the phototube, differences in rise time were discriminated by means of an electronic circuit, instead of separation by pulse height analysis. This pulse-shape discriminator successfully reduced noise counts below 18 keV in a multichannel analyzer from 100 cpm to 4 cpm, which permitted direct counting of tritiated water as dilute as 1 μCi/ml, without requiring any complicated preparation. (auth.)
An automatic sample changer for gamma spectrometry
International Nuclear Information System (INIS)
Andrews, D.J.
1984-01-01
An automatic sample changer for gamma spectrometry is described which is designed for large-volume, low radioactivity environmental samples of various sizes up to maximum dimensions 100 mm diameter x 60 mm high. The sample changer is suitable for use with most existing gamma spectrometry systems which utilize GeLi or NaI detectors in vertical mode, in conjunction with a pulse height analyzer having auto-cycle and suitable data output facilities; it is linked to a Nuclear Data ND 6620 computer-based analysis system. (U.K.)
The detection and estimation of spurious pulses
International Nuclear Information System (INIS)
1976-01-01
Spurious pulses which may interfere with the counting of particles can sometimes easily be detected by integral counting as a function of amplification or by pulse-height analysis. However, in order to estimate their count rate, more elaborate methods based on their time relationship are needed. Direct techniques (delayed coincidences, use of a multichannel analyser in time mode, time-to-amplitude conversion) and gating techniques (simple subtraction, correlation counting, pulsed sources, modulo counting) are discussed. These techniques are compared to each other and their application to various detectors is studied as well as the influence of a dead time on spurious pulses
Efficiency enhancements for MCP-based beta autoradiography imaging
Lees, J E
2002-01-01
We describe three approaches to increase the beta detection efficiency of microchannel plate detectors for biological beta autoradiography:(a)reversing the microchannel plate (MCP) bias polarity, changing the conventional high negative voltage on the input MCP to a grounded input, (b) a reduction in MCP pore size from 12.5 to 6 mu m, (c) using a CsI coating as an efficient secondary electron emitter. We also present our first measurements of double-tracer ( sup 3 H and sup 1 sup 4 C) imaging using pulse height analysis to distinguish between isotopes.
Newell, Homer E
2006-01-01
When employed with skill and understanding, vector analysis can be a practical and powerful tool. This text develops the algebra and calculus of vectors in a manner useful to physicists and engineers. Numerous exercises (with answers) not only provide practice in manipulation but also help establish students' physical and geometric intuition in regard to vectors and vector concepts.Part I, the basic portion of the text, consists of a thorough treatment of vector algebra and the vector calculus. Part II presents the illustrative matter, demonstrating applications to kinematics, mechanics, and e
Brand, Louis
2006-01-01
The use of vectors not only simplifies treatments of differential geometry, mechanics, hydrodynamics, and electrodynamics, but also makes mathematical and physical concepts more tangible and easy to grasp. This text for undergraduates was designed as a short introductory course to give students the tools of vector algebra and calculus, as well as a brief glimpse into these subjects' manifold applications. The applications are developed to the extent that the uses of the potential function, both scalar and vector, are fully illustrated. Moreover, the basic postulates of vector analysis are brou
International Nuclear Information System (INIS)
Lee, A.T.; Cabrera, B.; Dougherty, B.L.; Penn, M.J.; Pronko, J.G.; Tamura, S.
1996-01-01
We present measurements of the ballistic-phonon component resulting from nuclear and electron recoils in silicon at ∼380 mK. The detectors used for these experiments consist of a 300-μm-thick monocrystal of silicon instrumented with superconducting titanium transition-edge sensors. These sensors detect the initial wavefront of athermal phonons and give a pulse height that is sensitive to changes in surface-energy density resulting from the focusing of ballistic phonons. Nuclear recoils were generated by neutron bombardment of the detector. A Van de Graaff proton accelerator and a thick 7 Li target were used. Pulse-height spectra were compared for neutron, x-ray, and γ-ray events. A previous analysis of this data set found evidence for an increase in the ballistic-phonon component for nuclear recoils compared to electron recoils at a 95% confidence level. An improved understanding of the detector response has led to a change in the result. In the present analysis, the data are consistent with no increase at the 68% confidence level. This change stems from an increase in the uncertainty of the result rather than a significant change in the central value. The increase in ballistic phonon energy for nuclear recoils compared to electron recoils as a fraction of the total phonon energy (for equal total phonon energy events) was found to be 0.024 +0.041 -0.055 (68% confidence level). This result sets a limit of 11.6% (95% confidence level) on the ballistic phonon enhancement for nuclear recoils predicted by open-quote open-quote hot spot close-quote close-quote and electron-hole droplet models, which is the most stringent to date. To measure the ballistic-phonon component resulting from electron recoils, the pulse height as a function of event depth was compared to that of phonon simulations. (Abstract Truncated)
SPE analysis of high efficiency PMTs for the DEAP-3600 dark matter detector
International Nuclear Information System (INIS)
Olsen, Kevin; Hallin, Aksel
2011-01-01
The Dark matter Experiment using Argon Pulse-shape discrimination is a collaborative effort to develop a next-generation, tonne-scale dark matter detector at SNOLAB. The detector will feature a single-phase liquid argon (LAr) target surrounded by an array of 266 photomultiplier tubes (PMTs). A new high-efficiency Hamamatsu R877-100 PMT has been delivered to the University of Alberta for evaluation by the DEAP collaboration. The increase in efficiency could lead to a much greater light yield, but other experiments have reported a slower rise time. We have placed the PMT in a small dark box and had a base and preamplifier designed to be used with either an oscilloscope or a multi-channel analyzer. With this setup we have demonstrated the PMT's ability to distinguish single photo-electrons (SPE) and characterized the PMT by measuring the SPE pulse height spectrum, the peak-to-valley ratio, the dark pulse rate, the baseline, time resolution and SPE efficiency for varying the high voltage supplied to the PMT.
The XRF spectrometer and the selection of analysis conditions (instrumental variables)
International Nuclear Information System (INIS)
Willis, J.P.
2002-01-01
Full text: This presentation will begin with a brief discussion of EDXRF and flat- and curved-crystal WDXRF spectrometers, contrasting the major differences between the three types. The remainder of the presentation will contain a detailed overview of the choice and settings of the many instrumental variables contained in a modern WDXRF spectrometer, and will discuss critically the choices facing the analyst in setting up a WDXRF spectrometer for different elements and applications. In particular it will discuss the choice of tube target (when a choice is possible), the kV and mA settings, tube filters, collimator masks, collimators, analyzing crystals, secondary collimators, detectors, pulse height selection, X-ray path medium (air, nitrogen, vacuum or helium), counting times for peak and background positions and their effect on counting statistics and lower limit of detection (LLD). The use of Figure of Merit (FOM) calculations to objectively choose the best combination of instrumental variables also will be discussed. This presentation will be followed by a shorter session on a subsequent day entitled - A Selection of XRF Conditions - Practical Session, where participants will be given the opportunity to discuss in groups the selection of the best instrumental variables for three very diverse applications. Copyright (2002) Australian X-ray Analytical Association Inc
Pulse shape analysis based on similarity and neural network with digital-analog fusion method
International Nuclear Information System (INIS)
Mardiyanto, M.P.; Uritani, A.; Sakai, H.; Kawarabayashi, J.; Iguchi, T.
2000-01-01
Through the measurement of 22 Na γ-rays, it has been demonstrated that the correction process was well done by fusing the similarity values with the pulse heights measured by the analog system, where at least four improvements in the energy spectrum characteristics were recognized, i.e., the increase of the peak-to-valley ratio, the photopeak area, the photopeak sharpness without discarding any events, and the 1,275 keV γ-ray photopeak was seen. The use of a slow digitizer was the main problem for this method. However, it can be solved easily using a faster digitizer. The fusion method was also applied for the beta-gamma mixed spectra separation. Mixed spectra of beta-gamma of the 137 Cs- 90 Sr mixed source could be separated well. We made a comparison between the energy spectrum of 137 Cs as a result of independent measurement with the result of the separation. After being compared, both FWHM agreed quite well. However, there was a slight difference between the two spectra on the peak-to-valley ratio. This separation method is simple and useful so that it can be applied for many other similar applications. (S.Y.)
Abbott, Stephen
2015-01-01
This lively introductory text exposes the student to the rewards of a rigorous study of functions of a real variable. In each chapter, informal discussions of questions that give analysis its inherent fascination are followed by precise, but not overly formal, developments of the techniques needed to make sense of them. By focusing on the unifying themes of approximation and the resolution of paradoxes that arise in the transition from the finite to the infinite, the text turns what could be a daunting cascade of definitions and theorems into a coherent and engaging progression of ideas. Acutely aware of the need for rigor, the student is much better prepared to understand what constitutes a proper mathematical proof and how to write one. Fifteen years of classroom experience with the first edition of Understanding Analysis have solidified and refined the central narrative of the second edition. Roughly 150 new exercises join a selection of the best exercises from the first edition, and three more project-sty...
DEFF Research Database (Denmark)
Moore, R; Brødsgaard, I; Miller, ML
1997-01-01
A quantitative method for validating qualitative interview results and checking sample parameters is described and illustrated using common pain descriptions among a sample of Anglo-American and mandarin Chinese patients and dentists matched by age and gender. Assumptions were that subjects were ...... of covalidating questionnaires that reflect results of qualitative interviews are recommended in order to estimate sample parameters such as intersubject agreement, individual subject accuracy, and minimum required sample sizes.......A quantitative method for validating qualitative interview results and checking sample parameters is described and illustrated using common pain descriptions among a sample of Anglo-American and mandarin Chinese patients and dentists matched by age and gender. Assumptions were that subjects were...... of consistency in use of descriptors within groups, validity of description, accuracy of individuals compared with others in their group, and minimum required sample size were calculated using Cronbach's alpha, factor analysis, and Bayesian probability. Ethnic and professional differences within and across...
International Nuclear Information System (INIS)
Iorio, A.F.; Crespi, J.C.
1987-01-01
After ten years of operation at the Atucha I Nuclear Power Station a gear belonging to a pressurized heavy water reactor refuelling machine, failed. The gear box was used to operate the inlet-outlet heavy-water valve of the machine. Visual examination of the gear device showed an absence of lubricant and that several gear teeth were broken at the root. Motion was transmitted with a speed-reducing device with controlled adjustable times in order to produce a proper fitness of the valve closure. The aim of this paper is to discuss the results of the gear failure analysis in order to recommend the proper solution to prevent further failures. (Author)
International Nuclear Information System (INIS)
1988-01-01
In a search for correlations between the elemental composition of trace elements in human stones and the stone types with relation to their growth pattern, a combined PIXE and x-ray diffraction spectrometry approach was implemented. The combination of scanning PIXE and XRD has proved to be an advance in the methodology of stone analysis and may point to the growth pattern in the body. The exact role of trace elements in the formation and growth of urinary stones is not fully understood. Efforts are thus continuing firstly to solve the analytical problems concerned and secondly to design suitable experiments that would provide information about the occurrence and distribution of trace elements in urine. 1 fig., 1 ref
Computer-based system for acquisition of nuclear well log data
International Nuclear Information System (INIS)
Meisner, J.E.
1983-01-01
There is described a computer-based well logging system, for acquiring nuclear well log data, including gamma ray energy spectrum and neutron population decay rate data, and providing a real-time presentation of the data on an operator's display based on a traversal by a downhole instrument of a prescribed borehole depth interval. The system has a multichannel analyzer including a pulse height analyzer and a memory. After a spectral gamma ray pulse signal coming from a downhole instrument over a logging cable is amplified and conditioned, the pulse height analyzer converts the pulse height into a digital code by peak detection, sample-and-hold action, and analog-to-digital conversion. The digital code defines the address of a memory location or channel, corresponding to a particular gamma ray energy and having a count value to be incremented. The spectrum data is then accessed by the system central processing unit (CPU) for analysis, and routed to the operator's display for presentation as a plot of relative gamma ray emissions activity versus energy level. For acquiring neutron decay rate data, the system has a multichannel scaling unit including a memory and a memory address generator. After a burst of neutrons downhole, thermal and epithermal neutron detector pulses build up and die away. Using the neutron source trigger as an initializing reference, the address generator produces a sequence of memory address codes, each code addressing the memory for a prescribed period of time, so as to define a series of time slots. A detector pulse signal produced during a time slot results in the incrementing of the count value in an address memory location. (author)
International Nuclear Information System (INIS)
Straub, W.A.
1987-01-01
This review is the seventh in the series compiled by using the Dialog on-line CA Search facilities at the Information Resource Center of USS Technical Center covering the period from Oct. 1984 to Nov. 1, 1986. The quest for better surface properties, through the application of various electrochemical and other coating techniques, seems to have increased and reinforces the notion that only through the value added to a steel by proper finishing steps can a major supplier hope to compete profitably. The detection, determination, and control of microalloying constituents has also been generating a lot of interest as evidenced by the number of publications devoted to this subject in the last few years. Several recent review articles amplify on the recent trends in the application of modern analytical technology to steelmaking. Another review has been devoted to the determination of trace elements and the simultaneous determination of elements in metals by mass spectrometry, atomic absorption spectrometry, and multielement emission spectrometry. Problems associated with the analysis of electroplating wastewaters have been reviewed in a recent publication that has described the use of various spectrophotometric methods for this purpose. The collection and treatment of analytical data in the modern steel making environment have been extensively reviewed emphasis on the interaction of the providers and users of the analytical data, its quality, and the cost of its collection. Raw material treatment and beneficiation was the dominant theme
Bhatia, Rajendra
1997-01-01
A good part of matrix theory is functional analytic in spirit. This statement can be turned around. There are many problems in operator theory, where most of the complexities and subtleties are present in the finite-dimensional case. My purpose in writing this book is to present a systematic treatment of methods that are useful in the study of such problems. This book is intended for use as a text for upper division and gradu ate courses. Courses based on parts of the material have been given by me at the Indian Statistical Institute and at the University of Toronto (in collaboration with Chandler Davis). The book should also be useful as a reference for research workers in linear algebra, operator theory, mathe matical physics and numerical analysis. A possible subtitle of this book could be Matrix Inequalities. A reader who works through the book should expect to become proficient in the art of deriving such inequalities. Other authors have compared this art to that of cutting diamonds. One first has to...
International Nuclear Information System (INIS)
Thomas, R.E.
1982-03-01
An evaluation is made of the suitability of analytical and statistical sampling methods for making uncertainty analyses. The adjoint method is found to be well-suited for obtaining sensitivity coefficients for computer programs involving large numbers of equations and input parameters. For this purpose the Latin Hypercube Sampling method is found to be inferior to conventional experimental designs. The Latin hypercube method can be used to estimate output probability density functions, but requires supplementary rank transformations followed by stepwise regression to obtain uncertainty information on individual input parameters. A simple Cork and Bottle problem is used to illustrate the efficiency of the adjoint method relative to certain statistical sampling methods. For linear models of the form Ax=b it is shown that a complete adjoint sensitivity analysis can be made without formulating and solving the adjoint problem. This can be done either by using a special type of statistical sampling or by reformulating the primal problem and using suitable linear programming software
International Nuclear Information System (INIS)
Mihailescu, L.C.; Borella, A.; Massimi, C.; Schillebeeckx, P.
2009-01-01
The relatively long dead time in conventional data acquisition systems that provide simultaneously the pulse height and the time information for the detected events hinders cross-section measurements with high count rates. This is the case for capture cross-section measurements at the time-of-flight facility GELINA using high radioactive samples or thick samples of materials having strong resonances. Either the high average count rate (e.g. due to the radioactivity of the sample) or the high instantaneous count rate for strong resonances results in a large dead time correction. One solution to reduce the impact of the dead time is the use of a data acquisition system based on fast digitizers. The performances of two commercial digitizers (CAEN N172B and Acqiris DC282), coupled to a C 6 D 6 scintillator, have been tested in terms of pulse height linearity and resolution, dead time and time resolution. The signal processing was done on-line obtaining simultaneously the pulse height and time information for each detected event. With both digitizers a comparable pulse height linearity and resolution has been obtained as with a conventional system. The total dead time of both digital systems is at least a factor 5 shorter than the one for the conventional system. The main difference in performance between the two digitizers is the time resolution. For a relatively large scintillator, a time resolution of about 2 ns has been achieved with the DC282 module and the conventional system while the time resolution was limited to 15 ns with the CAEN N1728B module. For most nuclei a 15 ns time resolution is sufficient to perform resonance shape analysis. Therefore, the CAEN N1728B module can be used for the majority of capture cross-section measurements at GELINA. However, for nuclei with low level density, for which the resolved resonance region extends to the keV-region, a better time resolution is required and the Acqiris DC282 module has to be used.
Energy Technology Data Exchange (ETDEWEB)
Ibsen, Lars Bo; Liingaard, M.
2006-12-15
This technical report concerns the basic theory and principles for experimental modal analysis. The sections within the report are: Output-only modal analysis software, general digital analysis, basics of structural dynamics and modal analysis and system identification. (au)
Theoretical numerical analysis a functional analysis framework
Atkinson, Kendall
2005-01-01
This textbook prepares graduate students for research in numerical analysis/computational mathematics by giving to them a mathematical framework embedded in functional analysis and focused on numerical analysis. This helps the student to move rapidly into a research program. The text covers basic results of functional analysis, approximation theory, Fourier analysis and wavelets, iteration methods for nonlinear equations, finite difference methods, Sobolev spaces and weak formulations of boundary value problems, finite element methods, elliptic variational inequalities and their numerical solu
International Nuclear Information System (INIS)
2003-08-01
This book deals with analysis of heat transfer which includes nonlinear analysis examples, radiation heat transfer, analysis of heat transfer in ANSYS, verification of analysis result, analysis of heat transfer of transition with automatic time stepping and open control, analysis of heat transfer using arrangement of ANSYS, resistance of thermal contact, coupled field analysis such as of thermal-structural interaction, cases of coupled field analysis, and phase change.
Information security risk analysis
Peltier, Thomas R
2001-01-01
Effective Risk AnalysisQualitative Risk AnalysisValue AnalysisOther Qualitative MethodsFacilitated Risk Analysis Process (FRAP)Other Uses of Qualitative Risk AnalysisCase StudyAppendix A: QuestionnaireAppendix B: Facilitated Risk Analysis Process FormsAppendix C: Business Impact Analysis FormsAppendix D: Sample of ReportAppendix E: Threat DefinitionsAppendix F: Other Risk Analysis OpinionsIndex
International Nuclear Information System (INIS)
Son, Seung Hui
2004-02-01
This book deals with information technology and business process, information system architecture, methods of system development, plan on system development like problem analysis and feasibility analysis, cases for system development, comprehension of analysis of users demands, analysis of users demands using traditional analysis, users demands analysis using integrated information system architecture, system design using integrated information system architecture, system implementation, and system maintenance.
International Nuclear Information System (INIS)
Marshall, J.H. III.
1979-01-01
The active reduction of the number of analyzed events with pulse amplitudes which pileup has distorted improves measurement accuracy and response time in an apparatus for neutron-capture-based on-line elemental analysis of bulk substances. Within the apparatus, the analyzed bulk substance is exposed to neutrons, and neutron capture generates prompt gamma rays therefrom. A detector interacts with some of these gamma rays to produce electrical signals used to measure their energy spectrum by pulse-height analysis. Circuits associated with this pulse-height analysis also detect the pileup of the signals of two or more independent gamma rays using one or more of several techniques. These techniques include multiple outputs from a special amplifier-discriminator system, which has been optimized for low pulse-pair resolving time and may have adaptive thresholds, and the requirement that the relative amplitudes of the outputs of slow and fast amplifiers be consistent with a single event producing both outputs. Pulse-width measurements are also included in the pileup detection
Multichord time-resolved electron temperature measurements by the x-ray absorber-foil method on TFTR
International Nuclear Information System (INIS)
Kiraly, J.; Bitter, M.; Efthimion, P.
1985-09-01
Absorber foils have been installed in the TFTR X-Ray Imaging System to permit measurement of the electron temperature along 10 to 30 chords spaced at 5-12.5 cm with a time resolution of less than 100 μs. The technique uses the ratio of x-ray fluxes transmitted through two different foils. The ratio depends mainly on electron temperature. Simulations show that strong impurity line radiation can distort this ratio. To correct for these effects, special beryllium-scandium filters are employed to select the line-free region between 2 and 4.5 keV. Other filter pairs allow corrections for Fe L and Ni L line radiation as well as Ti K and Ni K emission. Good accuracy is also obtained with simple beryllium filters, provided that impurity corrections are incorporated in the analysis, taking line intensities from the x-ray pulse-height analysis diagnostic. A description of modeling calculations and a comparison of temperature values from this diagnostic with data from the x-ray pulse height analysis, the electron cyclotron emission, and the Thomson scattering diagnostics are presented. Several applications of the absorber foil electron temperature diagnostic on TFTR are discussed
Multichord time-resolved electron temperature measurements by the x-ray absorber-foil method on TFTR
Energy Technology Data Exchange (ETDEWEB)
Kiraly, J.; Bitter, M.; Efthimion, P.; von Goeler, S.; Grek, B.; Hill, K.W.; Johnson, D.; McGuire, K.; Sauthoff, N.; Sesnic, S.
1985-09-01
Absorber foils have been installed in the TFTR X-Ray Imaging System to permit measurement of the electron temperature along 10 to 30 chords spaced at 5-12.5 cm with a time resolution of less than 100 ..mu..s. The technique uses the ratio of x-ray fluxes transmitted through two different foils. The ratio depends mainly on electron temperature. Simulations show that strong impurity line radiation can distort this ratio. To correct for these effects, special beryllium-scandium filters are employed to select the line-free region between 2 and 4.5 keV. Other filter pairs allow corrections for Fe L and Ni L line radiation as well as Ti K and Ni K emission. Good accuracy is also obtained with simple beryllium filters, provided that impurity corrections are incorporated in the analysis, taking line intensities from the x-ray pulse-height analysis diagnostic. A description of modeling calculations and a comparison of temperature values from this diagnostic with data from the x-ray pulse height analysis, the electron cyclotron emission, and the Thomson scattering diagnostics are presented. Several applications of the absorber foil electron temperature diagnostic on TFTR are discussed.
International Nuclear Information System (INIS)
Klix, A.; Batistoni, P.; Boettger, R.; Lebrun-Grandie, D.; Fischer, U.; Henniger, J.; Leichtle, D.; Villari, R.
2010-01-01
Fast neutron and gamma-ray flux spectra and time-of-arrival spectra of slow neutrons have been measured in a neutronics mock-up of the European Helium-Cooled Lithium-Lead Test Blanket Module with the aim to validate nuclear cross-section data. The mock-up was irradiated with fusion peak neutrons from the DT neutron generator of the Technical University of Dresden. A well characterized cylindrical NE-213 scintillator was inserted into two positions in the LiPb/EUROFER assembly. Pulse height spectra from neutrons and gamma-rays were recorded from the NE-213 output. The spectra were then unfolded with experimentally obtained response matrices of the NE-213 detector. Time-of-arrival spectra of slow neutrons were measured with a 3 He counter placed in the mock-up, and the neutron generator was operated in pulsed mode. Monte Carlo calculations using the MCNP code and nuclear cross-section data from the JEFF-3.1.1 and FENDL-2.1 libraries were performed and the results are compared with the experimental results. A good agreement of measurement and calculation was found with some deviations in certain energy intervals.
International Nuclear Information System (INIS)
Moeller, G.; Presser, G.; Staehler, J.
1981-01-01
Pulse heights, timing properties and detection efficiencies of an MWPC were measured with 5.5 MeV alpha particles for different counting gases at low pressures. The pulse heights show a striking nonmonotonic dependence on the gas pressure that can be explained by a simple model of the amplification process at high reduced electric fields. The consequences of the observed pressure dependence of pulse heights for the detection of heavy ions with low pressure MWPCs are discussed. (orig.)
Nondestructive inspection of chemical warfare based on API-TOF
International Nuclear Information System (INIS)
Wang Xinhua; Zheng Pu; He Tie; An Li; Yang Jie; Fan Yu
2013-01-01
Background: Real-time, fast, accurate, nondestructive inspection (NDI) and quantitative analysis for chemical warfare are very imperative for chemical defense, anti-terror and nation security. Purpose: Associated Particles Technique (APT)/Neutron Time of Flight (TOF) has been developed for non-invasive inspection of sealed containers with chemical warfare agents. Methods: A prototype equipment for chemical warfare is consisted of an APT neutron generator with a 3×3 matrix of semiconductor detectors of associated alpha-particles, the shielding protection of neutron and gamma-ray, arrayed NaI(Tl)-based detectors of gamma-rays, fully-digital data acquisition electronics, data analysis, decision-making software, support platform and remote control system. Inelastic scattering gamma-ray pulse height spectra of sarin, VX, mustard gas and adamsite induced by 14-MeV neutron are measured. The energies of these gamma rays are used to identify the inelastic scattering elements, and the intensities of the peaks at these energies are used to reveal their concentrations. Results: The characteristic peaks of inelastic scattering gamma-ray pulse height spectra show that the prototype equipment can fast and accurately inspect chemical warfare. Conclusion: The equipment can be used to detect not only chemical warfare agents but also other hazardous materials, such as chemical/toxic/drug materials, if their chemical composition is in any way different from that of the surrounding materials. (authors)
Analysis of Project Finance | Energy Analysis | NREL
Analysis of Project Finance Analysis of Project Finance NREL analysis helps potential renewable energy developers and investors gain insights into the complex world of project finance. Renewable energy project finance is complex, requiring knowledge of federal tax credits, state-level incentives, renewable
International Nuclear Information System (INIS)
Wright, A.C.D.
2002-01-01
This paper discusses the safety analysis fundamentals in reactor design. This study includes safety analysis done to show consequences of postulated accidents are acceptable. Safety analysis is also used to set design of special safety systems and includes design assist analysis to support conceptual design. safety analysis is necessary for licensing a reactor, to maintain an operating license, support changes in plant operations
An example of multidimensional analysis: Discriminant analysis
International Nuclear Information System (INIS)
Lutz, P.
1990-01-01
Among the approaches on the data multi-dimensional analysis, lectures on the discriminant analysis including theoretical and practical aspects are presented. The discrimination problem, the analysis steps and the discrimination categories are stressed. Examples on the descriptive historical analysis, the discrimination for decision making, the demonstration and separation of the top quark are given. In the linear discriminant analysis the following subjects are discussed: Huyghens theorem, projection, discriminant variable, geometrical interpretation, case for g=2, classification method, separation of the top events. Criteria allowing the obtention of relevant results are included [fr
... Sources Ask Us Also Known As Sperm Analysis Sperm Count Seminal Fluid Analysis Formal Name Semen Analysis This ... semen Viscosity—consistency or thickness of the semen Sperm count—total number of sperm Sperm concentration (density)—number ...
Papageorgiou, Nikolaos S
2009-01-01
Offers an examination of important theoretical methods and procedures in applied analysis. This book details the important theoretical trends in nonlinear analysis and applications to different fields. It is suitable for those working on nonlinear analysis.
Soft x-ray measurements from the PDX tokamak
International Nuclear Information System (INIS)
Silver, E.H.; Bitter, M.; Brau, K.; Eames, D.; Greenberger, A.; Hill, K.W.; Meade, D.M.; Roney, W.; Sauthoff, N.R.; von Goeler, S.
1982-05-01
Temporally and spatially-resolved profiles of the PDX soft x-ray spectra have been measured during single tokamak pulses of circular and divertor plasmas with a recently developed pulse height analyzer. This detection system incorporates an array of five vertically displaced sets of lithium-drifted silicon detectors, each consisting of three independent channels optimized for rapid data collection in adjacent energy regions. Simultaneous measurement of x-ray emission integrated along five chords of the plasma cross section can thereby be achieved. Abel inversion of these data yields temporally-resolved radial profiles of the local electron temperature from the slope of the continuum, concentrations of high-Z impurities from the characteristic line intensities, and a measure of Z/sub eff/ from the continuum intensity. The techniques of x-ray pulse height analysis, with illustrations featuring the results from the initial PDX circular plasma experiments are discussed in detail. In addition, comparisons between circular and divertor plasmas on PDX, derived from the x-ray measurements, are also presented
International Nuclear Information System (INIS)
Goursky, V.; Thenes, P.
1969-01-01
This multipurpose unit is designed to accomplish one of the following functions: - gated window amplifier, - Analog memory and - Amplitude-to-time converter. The first function is mainly devoted to improve the poor resolution of pulse-height analyzers with a small number of channels. The analog memory, a new function in the standard range of plug-in modules, is capable of performing a number of operations: 1) fixed delay, or variable delay dependent on an external parameter (application to the analog processing of non-coincident pulses), 2) de-randomiser to increase the efficiency of the pulse height analysis in a spectrometry experiment, 3) linear multiplexer to allow an analyser to serve as many spectrometry devices as memory elements that it possesses. Associated with a coding scaler, this unit, if used as a amplitude-to-time converter, constitutes a Wilkinson A.D.C with a capability of 10 bits (or more) and with a 100 MHz clock frequency. (authors) [fr
International Nuclear Information System (INIS)
Sharma, Amandeep; Singh, Karamjit; Singh, Bhajan; Sandhu, B.S.
2011-01-01
The response function of gamma detector is an important factor for spectrum analysis because some photons and secondary electrons may escape the detector volume before fully depositing their energy, of course destroys the ideal delta function response. An inverse matrix approach, for unfolding of observed pulse-height distribution to a true photon spectrum, is used for construction of experimental response function by formulating a 40 x 40 matrix with bin mesh (E 1/2 ) of 0.025 (MeV) 1/2 for the present measurements. A tomographic scanner system, operating in a non-destructive and non-invasive way, is also presented for inspection of density variation in any object. The incoherent scattered intensity of 662 keV gamma photons, obtained by unfolding (deconvolution) the experimental pulse-height distribution of NaI(Tl) scintillation detector, provides the desired information. The method is quite sensitive, for showing inclusion of medium Z (atomic number) material (iron) in low Z material (aluminium) and detecting a void of ∼2 mm in size for iron block, to investigate the inhomogeneities in the object. Also, the grey scale images (using 'MATLAB') are shown to visualise the presence of defects/inclusion in metal samples.
Shape analysis in medical image analysis
Tavares, João
2014-01-01
This book contains thirteen contributions from invited experts of international recognition addressing important issues in shape analysis in medical image analysis, including techniques for image segmentation, registration, modelling and classification, and applications in biology, as well as in cardiac, brain, spine, chest, lung and clinical practice. This volume treats topics such as, anatomic and functional shape representation and matching; shape-based medical image segmentation; shape registration; statistical shape analysis; shape deformation; shape-based abnormity detection; shape tracking and longitudinal shape analysis; machine learning for shape modeling and analysis; shape-based computer-aided-diagnosis; shape-based medical navigation; benchmark and validation of shape representation, analysis and modeling algorithms. This work will be of interest to researchers, students, and manufacturers in the fields of artificial intelligence, bioengineering, biomechanics, computational mechanics, computationa...
Organization Search Go Search Polar Go MMAB SST Analysis Main page About MMAB Our Mission Our Personnel EMC Branches Global Climate & Weather Modeling Mesoscale Modeling Marine Modeling and Analysis Contact EMC (RTG_SST_HR) analysis For a regional map, click the desired area in the global SST analysis and anomaly maps
Foundations of factor analysis
Mulaik, Stanley A
2009-01-01
Introduction Factor Analysis and Structural Theories Brief History of Factor Analysis as a Linear Model Example of Factor AnalysisMathematical Foundations for Factor Analysis Introduction Scalar AlgebraVectorsMatrix AlgebraDeterminants Treatment of Variables as Vectors Maxima and Minima of FunctionsComposite Variables and Linear Transformations Introduction Composite Variables Unweighted Composite VariablesDifferentially Weighted Composites Matrix EquationsMulti
International Nuclear Information System (INIS)
PECH, S.H.
2000-01-01
This report describes the methodology used in conducting the K Basins Hazard Analysis, which provides the foundation for the K Basins Final Safety Analysis Report. This hazard analysis was performed in accordance with guidance provided by DOE-STD-3009-94, Preparation Guide for U. S. Department of Energy Nonreactor Nuclear Facility Safety Analysis Reports and implements the requirements of DOE Order 5480.23, Nuclear Safety Analysis Report
Quantitative analysis chemistry
International Nuclear Information System (INIS)
Ko, Wansuk; Lee, Choongyoung; Jun, Kwangsik; Hwang, Taeksung
1995-02-01
This book is about quantitative analysis chemistry. It is divided into ten chapters, which deal with the basic conception of material with the meaning of analysis chemistry and SI units, chemical equilibrium, basic preparation for quantitative analysis, introduction of volumetric analysis, acid-base titration of outline and experiment examples, chelate titration, oxidation-reduction titration with introduction, titration curve, and diazotization titration, precipitation titration, electrometric titration and quantitative analysis.
Energy Technology Data Exchange (ETDEWEB)
PECH, S.H.
2000-08-23
This report describes the methodology used in conducting the K Basins Hazard Analysis, which provides the foundation for the K Basins Final Safety Analysis Report. This hazard analysis was performed in accordance with guidance provided by DOE-STD-3009-94, Preparation Guide for U. S. Department of Energy Nonreactor Nuclear Facility Safety Analysis Reports and implements the requirements of DOE Order 5480.23, Nuclear Safety Analysis Report.
International Nuclear Information System (INIS)
WEBB, R.H.
1999-01-01
This report describes the methodology used in conducting the K Basins Hazard Analysis, which provides the foundation for the K Basins Safety Analysis Report (HNF-SD-WM-SAR-062/Rev.4). This hazard analysis was performed in accordance with guidance provided by DOE-STD-3009-94, Preparation Guide for U. S. Department of Energy Nonreactor Nuclear Facility Safety Analysis Reports and implements the requirements of DOE Order 5480.23, Nuclear Safety Analysis Report
Philipp Mayring
2000-01-01
The article describes an approach of systematic, rule guided qualitative text analysis, which tries to preserve some methodological strengths of quantitative content analysis and widen them to a concept of qualitative procedure. First the development of content analysis is delineated and the basic principles are explained (units of analysis, step models, working with categories, validity and reliability). Then the central procedures of qualitative content analysis, inductive development of ca...
RELIABILITY ANALYSIS OF BENDING ELIABILITY ANALYSIS OF ...
African Journals Online (AJOL)
eobe
Reliability analysis of the safety levels of the criteria slabs, have been .... was also noted [2] that if the risk level or β < 3.1), the ... reliability analysis. A study [6] has shown that all geometric variables, ..... Germany, 1988. 12. Hasofer, A. M and ...
DTI analysis methods : Voxel-based analysis
Van Hecke, Wim; Leemans, Alexander; Emsell, Louise
2016-01-01
Voxel-based analysis (VBA) of diffusion tensor imaging (DTI) data permits the investigation of voxel-wise differences or changes in DTI metrics in every voxel of a brain dataset. It is applied primarily in the exploratory analysis of hypothesized group-level alterations in DTI parameters, as it does
Analysis of Precision of Activation Analysis Method
DEFF Research Database (Denmark)
Heydorn, Kaj; Nørgaard, K.
1973-01-01
The precision of an activation-analysis method prescribes the estimation of the precision of a single analytical result. The adequacy of these estimates to account for the observed variation between duplicate results from the analysis of different samples and materials, is tested by the statistic T...
Hazard Analysis Database Report
Grams, W H
2000-01-01
The Hazard Analysis Database was developed in conjunction with the hazard analysis activities conducted in accordance with DOE-STD-3009-94, Preparation Guide for U S . Department of Energy Nonreactor Nuclear Facility Safety Analysis Reports, for HNF-SD-WM-SAR-067, Tank Farms Final Safety Analysis Report (FSAR). The FSAR is part of the approved Authorization Basis (AB) for the River Protection Project (RPP). This document describes, identifies, and defines the contents and structure of the Tank Farms FSAR Hazard Analysis Database and documents the configuration control changes made to the database. The Hazard Analysis Database contains the collection of information generated during the initial hazard evaluations and the subsequent hazard and accident analysis activities. The Hazard Analysis Database supports the preparation of Chapters 3 ,4 , and 5 of the Tank Farms FSAR and the Unreviewed Safety Question (USQ) process and consists of two major, interrelated data sets: (1) Hazard Analysis Database: Data from t...
Santiago, John
2013-01-01
Circuits overloaded from electric circuit analysis? Many universities require that students pursuing a degree in electrical or computer engineering take an Electric Circuit Analysis course to determine who will ""make the cut"" and continue in the degree program. Circuit Analysis For Dummies will help these students to better understand electric circuit analysis by presenting the information in an effective and straightforward manner. Circuit Analysis For Dummies gives you clear-cut information about the topics covered in an electric circuit analysis courses to help
Cluster analysis for applications
Anderberg, Michael R
1973-01-01
Cluster Analysis for Applications deals with methods and various applications of cluster analysis. Topics covered range from variables and scales to measures of association among variables and among data units. Conceptual problems in cluster analysis are discussed, along with hierarchical and non-hierarchical clustering methods. The necessary elements of data analysis, statistics, cluster analysis, and computer implementation are integrated vertically to cover the complete path from raw data to a finished analysis.Comprised of 10 chapters, this book begins with an introduction to the subject o
Directory of Open Access Journals (Sweden)
Mahsa Noori Asl
2013-01-01
Full Text Available Compton-scattered photons included within the photopeak pulse-height window result in the degradation of SPECT images both qualitatively and quantitatively. The purpose of this study is to evaluate and compare six scatter correction methods based on setting the energy windows in 99m Tc spectrum. SIMIND Monte Carlo simulation is used to generate the projection images from a cold-sphere hot-background phantom. For evaluation of different scatter correction methods, three assessment criteria including image contrast, signal-to-noise ratio (SNR and relative noise of the background (RNB are considered. Except for the dual-photopeak window (DPW method, the image contrast of the five cold spheres is improved in the range of 2.7-26%. Among methods considered, two methods show a nonuniform correction performance. The RNB for all of the scatter correction methods is ranged from minimum 0.03 for DPW method to maximum 0.0727 for the three energy window (TEW method using trapezoidal approximation. The TEW method using triangular approximation because of ease of implementation, good improvement of the image contrast and the SNR for the five cold spheres, and the low noise level is proposed as most appropriate correction method.
Activation analysis in food analysis. Pt. 9
International Nuclear Information System (INIS)
Szabo, S.A.
1992-01-01
An overview is presented on the application of activation analysis (AA) techniques for food analysis, as reflected at a recent international conference titled Activation Analysis and its Applications. The most popular analytical techniques include instrumental neutron AA, (INAA or NAA), radiochemical NAA (RNAA), X-ray fluorescence analysis and mass spectrometry. Data are presented for the multielemental NAA of instant soups, for elemental composition of drinking water in Iraq, for Na, K, Mn contents of various Indian rices, for As, Hg, Sb and Se determination in various seafoods, for daily microelement takeup in China, for the elemental composition of Chinese teas. Expected development trends in AA are outlined. (R.P.) 24 refs.; 8 tabs
Joint fluid analysis; Joint fluid aspiration ... El-Gabalawy HS. Synovial fluid analysis, synovial biopsy, and synovial pathology. In: Firestein GS, Budd RC, Gabriel SE, McInnes IB, O'Dell JR, eds. Kelly's Textbook of ...
International Nuclear Information System (INIS)
Burgess, R.L.
1978-01-01
Progress is reported on the following research programs: analysis and modeling of ecosystems; EDFB/IBP data center; biome analysis studies; land/water interaction studies; and computer programs for development of models
Confirmatory Composite Analysis
Schuberth, Florian; Henseler, Jörg; Dijkstra, Theo K.
2018-01-01
We introduce confirmatory composite analysis (CCA) as a structural equation modeling technique that aims at testing composite models. CCA entails the same steps as confirmatory factor analysis: model specification, model identification, model estimation, and model testing. Composite models are
Introductory numerical analysis
Pettofrezzo, Anthony J
2006-01-01
Written for undergraduates who require a familiarity with the principles behind numerical analysis, this classical treatment encompasses finite differences, least squares theory, and harmonic analysis. Over 70 examples and 280 exercises. 1967 edition.
Gap Analysis: Application to Earned Value Analysis
Langford, Gary O.; Franck, Raymond (Chip)
2008-01-01
Sponsored Report (for Acquisition Research Program) Earned Value is regarded as a useful tool to monitor commercial and defense system acquisitions. This paper applies the theoretical foundations and systematics of Gap Analysis to improve Earned Value Management. As currently implemented, Earned Value inaccurately provides a higher value for the work performed. This preliminary research indicates that Earned Value calculations can be corrected. Value Analysis, properly defined and enacted,...
Importance-performance analysis based SWOT analysis
Phadermrod, Boonyarat; Crowder, Richard M.; Wills, Gary B.
2016-01-01
SWOT analysis, a commonly used tool for strategic planning, is traditionally a form of brainstorming. Hence, it has been criticised that it is likely to hold subjective views of the individuals who participate in a brainstorming session and that SWOT factors are not prioritized by their significance thus it may result in an improper strategic action. While most studies of SWOT analysis have only focused on solving these shortcomings separately, this study offers an approach to diminish both s...
Discourse analysis and Foucault's
Directory of Open Access Journals (Sweden)
Jansen I.
2008-01-01
Full Text Available Discourse analysis is a method with up to now was less recognized in nursing science, althoughmore recently nursing scientists are discovering it for their purposes. However, several authors have criticized thatdiscourse analysis is often misinterpreted because of a lack of understanding of its theoretical backgrounds. In thisarticle, I reconstruct Foucault’s writings in his “Archaeology of Knowledge” to provide a theoretical base for futurearchaeological discourse analysis, which can be categorized as a socio-linguistic discourse analysis.
Yuan, Ying; MacKinnon, David P.
2009-01-01
This article proposes Bayesian analysis of mediation effects. Compared to conventional frequentist mediation analysis, the Bayesian approach has several advantages. First, it allows researchers to incorporate prior information into the mediation analysis, thus potentially improving the efficiency of estimates. Second, under the Bayesian mediation analysis, inference is straightforward and exact, which makes it appealing for studies with small samples. Third, the Bayesian approach is conceptua...
Fitzmaurice, Garrett M; Ware, James H
2012-01-01
Praise for the First Edition "". . . [this book] should be on the shelf of everyone interested in . . . longitudinal data analysis.""-Journal of the American Statistical Association Features newly developed topics and applications of the analysis of longitudinal data Applied Longitudinal Analysis, Second Edition presents modern methods for analyzing data from longitudinal studies and now features the latest state-of-the-art techniques. The book emphasizes practical, rather than theoretical, aspects of methods for the analysis of diverse types of lo
Regression analysis by example
Chatterjee, Samprit
2012-01-01
Praise for the Fourth Edition: ""This book is . . . an excellent source of examples for regression analysis. It has been and still is readily readable and understandable."" -Journal of the American Statistical Association Regression analysis is a conceptually simple method for investigating relationships among variables. Carrying out a successful application of regression analysis, however, requires a balance of theoretical results, empirical rules, and subjective judgment. Regression Analysis by Example, Fifth Edition has been expanded
2014-01-01
M.Ing. (Electrical & Electronic Engineering) One of the most important steps to be taken before a site is to be selected for the extraction of wind energy is the analysis of the energy within the wind on that particular site. No wind energy analysis system exists for the measurement and analysis of wind power. This dissertation documents the design and development of a Wind Energy Analysis System (WEAS). Using a micro-controller based design in conjunction with sensors, WEAS measure, calcu...
Slice hyperholomorphic Schur analysis
Alpay, Daniel; Sabadini, Irene
2016-01-01
This book defines and examines the counterpart of Schur functions and Schur analysis in the slice hyperholomorphic setting. It is organized into three parts: the first introduces readers to classical Schur analysis, while the second offers background material on quaternions, slice hyperholomorphic functions, and quaternionic functional analysis. The third part represents the core of the book and explores quaternionic Schur analysis and its various applications. The book includes previously unpublished results and provides the basis for new directions of research.
Computational movement analysis
Laube, Patrick
2014-01-01
This SpringerBrief discusses the characteristics of spatiotemporal movement data, including uncertainty and scale. It investigates three core aspects of Computational Movement Analysis: Conceptual modeling of movement and movement spaces, spatiotemporal analysis methods aiming at a better understanding of movement processes (with a focus on data mining for movement patterns), and using decentralized spatial computing methods in movement analysis. The author presents Computational Movement Analysis as an interdisciplinary umbrella for analyzing movement processes with methods from a range of fi
Thompson, Cheryl Bagley
2009-01-01
This 13th article of the Basics of Research series is first in a short series on statistical analysis. These articles will discuss creating your statistical analysis plan, levels of measurement, descriptive statistics, probability theory, inferential statistics, and general considerations for interpretation of the results of a statistical analysis.
Trend Analysis Using Microcomputers.
Berger, Carl F.
A trend analysis statistical package and additional programs for the Apple microcomputer are presented. They illustrate strategies of data analysis suitable to the graphics and processing capabilities of the microcomputer. The programs analyze data sets using examples of: (1) analysis of variance with multiple linear regression; (2) exponential…
Yuan, Ying; MacKinnon, David P.
2009-01-01
In this article, we propose Bayesian analysis of mediation effects. Compared with conventional frequentist mediation analysis, the Bayesian approach has several advantages. First, it allows researchers to incorporate prior information into the mediation analysis, thus potentially improving the efficiency of estimates. Second, under the Bayesian…
Automation of activation analysis
International Nuclear Information System (INIS)
Ivanov, I.N.; Ivanets, V.N.; Filippov, V.V.
1985-01-01
The basic data on the methods and equipment of activation analysis are presented. Recommendations on the selection of activation analysis techniques, and especially the technique envisaging the use of short-lived isotopes, are given. The equipment possibilities to increase dataway carrying capacity, using modern computers for the automation of the analysis and data processing procedure, are shown
Cuesta, Hector
2013-01-01
Each chapter of the book quickly introduces a key 'theme' of Data Analysis, before immersing you in the practical aspects of each theme. You'll learn quickly how to perform all aspects of Data Analysis.Practical Data Analysis is a book ideal for home and small business users who want to slice & dice the data they have on hand with minimum hassle.
Mathematical analysis fundamentals
Bashirov, Agamirza
2014-01-01
The author's goal is a rigorous presentation of the fundamentals of analysis, starting from elementary level and moving to the advanced coursework. The curriculum of all mathematics (pure or applied) and physics programs include a compulsory course in mathematical analysis. This book will serve as can serve a main textbook of such (one semester) courses. The book can also serve as additional reading for such courses as real analysis, functional analysis, harmonic analysis etc. For non-math major students requiring math beyond calculus, this is a more friendly approach than many math-centric o
Foundations of mathematical analysis
Johnsonbaugh, Richard
2010-01-01
This classroom-tested volume offers a definitive look at modern analysis, with views of applications to statistics, numerical analysis, Fourier series, differential equations, mathematical analysis, and functional analysis. Upper-level undergraduate students with a background in calculus will benefit from its teachings, along with beginning graduate students seeking a firm grounding in modern analysis. A self-contained text, it presents the necessary background on the limit concept, and the first seven chapters could constitute a one-semester introduction to limits. Subsequent chapters discuss
Analysis in usability evaluations
DEFF Research Database (Denmark)
Følstad, Asbjørn; Lai-Chong Law, Effie; Hornbæk, Kasper
2010-01-01
While the planning and implementation of usability evaluations are well described in the literature, the analysis of the evaluation data is not. We present interviews with 11 usability professionals on how they conduct analysis, describing the resources, collaboration, creation of recommendations......, and prioritization involved. The interviews indicate a lack of structure in the analysis process and suggest activities, such as generating recommendations, that are unsupported by existing methods. We discuss how to better support analysis, and propose four themes for future research on analysis in usability...
DEFF Research Database (Denmark)
Bøving, Kristian Billeskov; Simonsen, Jesper
2004-01-01
This article documents how log analysis can inform qualitative studies concerning the usage of web-based information systems (WIS). No prior research has used http log files as data to study collaboration between multiple users in organisational settings. We investigate how to perform http log...... analysis; what http log analysis says about the nature of collaborative WIS use; and how results from http log analysis may support other data collection methods such as surveys, interviews, and observation. The analysis of log files initially lends itself to research designs, which serve to test...... hypotheses using a quantitative methodology. We show that http log analysis can also be valuable in qualitative research such as case studies. The results from http log analysis can be triangulated with other data sources and for example serve as a means of supporting the interpretation of interview data...
Amir Farbin
The ATLAS Analysis Model is a continually developing vision of how to reconcile physics analysis requirements with the ATLAS offline software and computing model constraints. In the past year this vision has influenced the evolution of the ATLAS Event Data Model, the Athena software framework, and physics analysis tools. These developments, along with the October Analysis Model Workshop and the planning for CSC analyses have led to a rapid refinement of the ATLAS Analysis Model in the past few months. This article introduces some of the relevant issues and presents the current vision of the future ATLAS Analysis Model. Event Data Model The ATLAS Event Data Model (EDM) consists of several levels of details, each targeted for a specific set of tasks. For example the Event Summary Data (ESD) stores calorimeter cells and tracking system hits thereby permitting many calibration and alignment tasks, but will be only accessible at particular computing sites with potentially large latency. In contrast, the Analysis...
Multivariate analysis with LISREL
Jöreskog, Karl G; Y Wallentin, Fan
2016-01-01
This book traces the theory and methodology of multivariate statistical analysis and shows how it can be conducted in practice using the LISREL computer program. It presents not only the typical uses of LISREL, such as confirmatory factor analysis and structural equation models, but also several other multivariate analysis topics, including regression (univariate, multivariate, censored, logistic, and probit), generalized linear models, multilevel analysis, and principal component analysis. It provides numerous examples from several disciplines and discusses and interprets the results, illustrated with sections of output from the LISREL program, in the context of the example. The book is intended for masters and PhD students and researchers in the social, behavioral, economic and many other sciences who require a basic understanding of multivariate statistical theory and methods for their analysis of multivariate data. It can also be used as a textbook on various topics of multivariate statistical analysis.
International Nuclear Information System (INIS)
Actis, Oxana; Brodski, Michael; Erdmann, Martin; Fischer, Robert; Hinzmann, Andreas; Mueller, Gero; Muenzer, Thomas; Plum, Matthias; Steggemann, Jan; Winchen, Tobias; Klimkovich, Tatsiana
2010-01-01
VISPA is a development environment for high energy physics analyses which enables physicists to combine graphical and textual work. A physics analysis cycle consists of prototyping, performing, and verifying the analysis. The main feature of VISPA is a multipurpose window for visual steering of analysis steps, creation of analysis templates, and browsing physics event data at different steps of an analysis. VISPA follows an experiment-independent approach and incorporates various tools for steering and controlling required in a typical analysis. Connection to different frameworks of high energy physics experiments is achieved by using different types of interfaces. We present the look-and-feel for an example physics analysis at the LHC and explain the underlying software concepts of VISPA.
Cost benefit analysis cost effectiveness analysis
International Nuclear Information System (INIS)
Lombard, J.
1986-09-01
The comparison of various protection options in order to determine which is the best compromise between cost of protection and residual risk is the purpose of the ALARA procedure. The use of decision-aiding techniques is valuable as an aid to selection procedures. The purpose of this study is to introduce two rather simple and well known decision aiding techniques: the cost-effectiveness analysis and the cost-benefit analysis. These two techniques are relevant for the great part of ALARA decisions which need the use of a quantitative technique. The study is based on an hypothetical case of 10 protection options. Four methods are applied to the data
International Nuclear Information System (INIS)
Sommer, S; Tinh Tran, T.
2008-01-01
Washington Safety Management Solutions, LLC developed web-based software to improve the efficiency and consistency of hazard identification and analysis, control selection and classification, and to standardize analysis reporting at Savannah River Site. In the new nuclear age, information technology provides methods to improve the efficiency of the documented safety analysis development process which includes hazard analysis activities. This software provides a web interface that interacts with a relational database to support analysis, record data, and to ensure reporting consistency. A team of subject matter experts participated in a series of meetings to review the associated processes and procedures for requirements and standard practices. Through these meetings, a set of software requirements were developed and compiled into a requirements traceability matrix from which software could be developed. The software was tested to ensure compliance with the requirements. Training was provided to the hazard analysis leads. Hazard analysis teams using the software have verified its operability. The software has been classified as NQA-1, Level D, as it supports the analysis team but does not perform the analysis. The software can be transported to other sites with alternate risk schemes. The software is being used to support the development of 14 hazard analyses. User responses have been positive with a number of suggestions for improvement which are being incorporated as time permits. The software has enforced a uniform implementation of the site procedures. The software has significantly improved the efficiency and standardization of the hazard analysis process
Functional analysis and applications
Siddiqi, Abul Hasan
2018-01-01
This self-contained textbook discusses all major topics in functional analysis. Combining classical materials with new methods, it supplies numerous relevant solved examples and problems and discusses the applications of functional analysis in diverse fields. The book is unique in its scope, and a variety of applications of functional analysis and operator-theoretic methods are devoted to each area of application. Each chapter includes a set of problems, some of which are routine and elementary, and some of which are more advanced. The book is primarily intended as a textbook for graduate and advanced undergraduate students in applied mathematics and engineering. It offers several attractive features making it ideally suited for courses on functional analysis intended to provide a basic introduction to the subject and the impact of functional analysis on applied and computational mathematics, nonlinear functional analysis and optimization. It introduces emerging topics like wavelets, Gabor system, inverse pro...
DEFF Research Database (Denmark)
Bemman, Brian; Meredith, David
it with a “ground truth” analysis of the same music pro- duced by a human expert (see, in particular, [5]). In this paper, we explore the problem of generating an encoding of the musical surface of a work automatically from a systematic encoding of an analysis. The ability to do this depends on one having...... an effective (i.e., comput- able), correct and complete description of some aspect of the structure of the music. Generating the surface struc- ture of a piece from an analysis in this manner serves as a proof of the analysis' correctness, effectiveness and com- pleteness. We present a reductive analysis......In recent years, a significant body of research has focused on developing algorithms for computing analyses of mu- sical works automatically from encodings of these works' surfaces [3,4,7,10,11]. The quality of the output of such analysis algorithms is typically evaluated by comparing...
Fundamentals of functional analysis
Farenick, Douglas
2016-01-01
This book provides a unique path for graduate or advanced undergraduate students to begin studying the rich subject of functional analysis with fewer prerequisites than is normally required. The text begins with a self-contained and highly efficient introduction to topology and measure theory, which focuses on the essential notions required for the study of functional analysis, and which are often buried within full-length overviews of the subjects. This is particularly useful for those in applied mathematics, engineering, or physics who need to have a firm grasp of functional analysis, but not necessarily some of the more abstruse aspects of topology and measure theory normally encountered. The reader is assumed to only have knowledge of basic real analysis, complex analysis, and algebra. The latter part of the text provides an outstanding treatment of Banach space theory and operator theory, covering topics not usually found together in other books on functional analysis. Written in a clear, concise manner,...
DEFF Research Database (Denmark)
This book provides an in-depth introduction and overview of current research in computational music analysis. Its seventeen chapters, written by leading researchers, collectively represent the diversity as well as the technical and philosophical sophistication of the work being done today...... on well-established theories in music theory and analysis, such as Forte's pitch-class set theory, Schenkerian analysis, the methods of semiotic analysis developed by Ruwet and Nattiez, and Lerdahl and Jackendoff's Generative Theory of Tonal Music. The book is divided into six parts, covering...... music analysis, the book provides an invaluable resource for researchers, teachers and students in music theory and analysis, computer science, music information retrieval and related disciplines. It also provides a state-of-the-art reference for practitioners in the music technology industry....
Analysis apparatus and method of analysis
International Nuclear Information System (INIS)
1976-01-01
A continuous streaming method developed for the excution of immunoassays is described in this patent. In addition, a suitable apparatus for the method was developed whereby magnetic particles are automatically employed for the consecutive analysis of a series of liquid samples via the RIA technique
International Nuclear Information System (INIS)
Dougherty, E.M.; Fragola, J.R.
1988-01-01
The authors present a treatment of human reliability analysis incorporating an introduction to probabilistic risk assessment for nuclear power generating stations. They treat the subject according to the framework established for general systems theory. Draws upon reliability analysis, psychology, human factors engineering, and statistics, integrating elements of these fields within a systems framework. Provides a history of human reliability analysis, and includes examples of the application of the systems approach
Emission spectrochemical analysis
International Nuclear Information System (INIS)
Rives, R.D.; Bruks, R.R.
1983-01-01
The emission spectrochemical method of analysis based on the fact that atoms of elements can be excited in the electric arc or in the laser beam and will emit radiation with characteristic wave lengths is considered. The review contains the data on spectrochemical analysis, of liquids geological materials, scheme of laser microprobe. The main characteristics of emission spectroscopy, atomic absorption spectroscopy and X-ray fluorescent analysis, are aeneralized
International Nuclear Information System (INIS)
Crawford, H.J.; Lindstrom, P.J.
1983-06-01
Our analysis program LULU has proven very useful in all stages of experiment analysis, from prerun detector debugging through final data reduction. It has solved our problem of having arbitrary word length events and is easy enough to use that many separate experimenters are now analyzing with LULU. The ability to use the same software for all stages of experiment analysis greatly eases the programming burden. We may even get around to making the graphics elegant someday
Mastering Clojure data analysis
Rochester, Eric
2014-01-01
This book consists of a practical, example-oriented approach that aims to help you learn how to use Clojure for data analysis quickly and efficiently.This book is great for those who have experience with Clojure and who need to use it to perform data analysis. This book will also be hugely beneficial for readers with basic experience in data analysis and statistics.
Fast neutron activation analysis
International Nuclear Information System (INIS)
Pepelnik, R.
1986-01-01
Since 1981 numerous 14 MeV neutron activation analyses were performed at Korona. On the basis of that work the advantages of this analysis technique and therewith obtained results are compared with other analytical methods. The procedure of activation analysis, the characteristics of Korona, some analytical investigations in environmental research and material physics, as well as sources of systematic errors in trace analysis are described. (orig.) [de
Crisan, Dan
2011-01-01
"Stochastic Analysis" aims to provide mathematical tools to describe and model high dimensional random systems. Such tools arise in the study of Stochastic Differential Equations and Stochastic Partial Differential Equations, Infinite Dimensional Stochastic Geometry, Random Media and Interacting Particle Systems, Super-processes, Stochastic Filtering, Mathematical Finance, etc. Stochastic Analysis has emerged as a core area of late 20th century Mathematics and is currently undergoing a rapid scientific development. The special volume "Stochastic Analysis 2010" provides a sa
The ATLAS Analysis Architecture
International Nuclear Information System (INIS)
Cranmer, K.S.
2008-01-01
We present an overview of the ATLAS analysis architecture including the relevant aspects of the computing model and the major architectural aspects of the Athena framework. Emphasis will be given to the interplay between the analysis use cases and the technical aspects of the architecture including the design of the event data model, transient-persistent separation, data reduction strategies, analysis tools, and ROOT interoperability
Circuit analysis with Multisim
Baez-Lopez, David
2011-01-01
This book is concerned with circuit simulation using National Instruments Multisim. It focuses on the use and comprehension of the working techniques for electrical and electronic circuit simulation. The first chapters are devoted to basic circuit analysis.It starts by describing in detail how to perform a DC analysis using only resistors and independent and controlled sources. Then, it introduces capacitors and inductors to make a transient analysis. In the case of transient analysis, it is possible to have an initial condition either in the capacitor voltage or in the inductor current, or bo
Textile Technology Analysis Lab
Federal Laboratory Consortium — The Textile Analysis Labis built for evaluating and characterizing the physical properties of an array of textile materials, but specifically those used in aircrew...
DEFF Research Database (Denmark)
Sørensen, Olav Jull
2009-01-01
The review presents the book International Market Analysis: Theories and Methods, written by John Kuiada, professor at Centre of International Business, Department of Business Studies, Aalborg University. The book is refreshingly new in its way of looking at a classical problem. It looks at market...... analysis from the point of vie of ways of thinking about markets. Furthermore, the book includes the concept of learning in the analysis of markets og how the way we understand business reality influneces our choice of methodology for market analysis....
Chemical Security Analysis Center
Federal Laboratory Consortium — In 2006, by Presidential Directive, DHS established the Chemical Security Analysis Center (CSAC) to identify and assess chemical threats and vulnerabilities in the...
Geospatial Data Analysis Facility
Federal Laboratory Consortium — Geospatial application development, location-based services, spatial modeling, and spatial analysis are examples of the many research applications that this facility...
National Research Council Canada - National Science Library
Gilbert, John
1984-01-01
... quantification methods used in the analysis of mycotoxins in foods - Confirmation and quantification of trace organic food contaminants by mass spectrometry-selected ion monitoring - Chemiluminescence...
Federal Laboratory Consortium — FUNCTION: Uses state-of-the-art instrumentation for qualitative and quantitative analysis of organic and inorganic compounds, and biomolecules from gas, liquid, and...
Thermogravimetric Analysis Laboratory
Federal Laboratory Consortium — At NETL’s Thermogravimetric Analysis Laboratory in Morgantown, WV, researchers study how chemical looping combustion (CLC) can be applied to fossil energy systems....
Sensitivity and uncertainty analysis
Cacuci, Dan G; Navon, Ionel Michael
2005-01-01
As computer-assisted modeling and analysis of physical processes have continued to grow and diversify, sensitivity and uncertainty analyses have become indispensable scientific tools. Sensitivity and Uncertainty Analysis. Volume I: Theory focused on the mathematical underpinnings of two important methods for such analyses: the Adjoint Sensitivity Analysis Procedure and the Global Adjoint Sensitivity Analysis Procedure. This volume concentrates on the practical aspects of performing these analyses for large-scale systems. The applications addressed include two-phase flow problems, a radiative c
Automated multispectra alpha spectrometer and data reduction system
International Nuclear Information System (INIS)
Hochel, R.C.
1975-12-01
A complete hardware and software package for the accumulation and rapid analysis of multiple alpha pulse height spectra has been developed. The system utilizes a 4096-channel analyzer capable of accepting up to sixteen inputs from solid-state surface barrier detectors via mixer-router modules. The analyzer is interfaced to a desk-top programmable calculator and thermal line printer. A chained software package including spectrum printout, peak analysis, plutonium-238 and plutonium-239 data reduction, and automatic energy calibration routines was written. With the chained program a complete printout, peak analysis, and plutonium data reduction of a 512-channel alpha spectrum are obtained in about three minutes with an accuracy within five percent of hand analyses
Hox, J.J.; Maas, C.J.M.; Lensvelt-Mulders, G.J.L.M.
2004-01-01
The goal of meta-analysis is to integrate the research results of a number of studies on a specific topic. Characteristic for meta-analysis is that in general only the summary statistics of the studies are used and not the original data. When the published research results to be integrated
International Nuclear Information System (INIS)
Hahn, A.A.
1994-11-01
The complexity of instrumentation sometimes requires data analysis to be done before the result is presented to the control room. This tutorial reviews some of the theoretical assumptions underlying the more popular forms of data analysis and presents simple examples to illuminate the advantages and hazards of different techniques
Activation analysis. Detection limits
International Nuclear Information System (INIS)
Revel, G.
1999-01-01
Numerical data and limits of detection related to the four irradiation modes, often used in activation analysis (reactor neutrons, 14 MeV neutrons, photon gamma and charged particles) are presented here. The technical presentation of the activation analysis is detailed in the paper P 2565 of Techniques de l'Ingenieur. (A.L.B.)
SMART performance analysis methodology
International Nuclear Information System (INIS)
Lim, H. S.; Kim, H. C.; Lee, D. J.
2001-04-01
To ensure the required and desired operation over the plant lifetime, the performance analysis for the SMART NSSS design is done by means of the specified analysis methodologies for the performance related design basis events(PRDBE). The PRDBE is an occurrence(event) that shall be accommodated in the design of the plant and whose consequence would be no more severe than normal service effects of the plant equipment. The performance analysis methodology which systematizes the methods and procedures to analyze the PRDBEs is as follows. Based on the operation mode suitable to the characteristics of the SMART NSSS, the corresponding PRDBEs and allowable range of process parameters for these events are deduced. With the developed control logic for each operation mode, the system thermalhydraulics are analyzed for the chosen PRDBEs using the system analysis code. Particularly, because of different system characteristics of SMART from the existing commercial nuclear power plants, the operation mode, PRDBEs, control logic, and analysis code should be consistent with the SMART design. This report presents the categories of the PRDBEs chosen based on each operation mode and the transition among these and the acceptance criteria for each PRDBE. It also includes the analysis methods and procedures for each PRDBE and the concept of the control logic for each operation mode. Therefore this report in which the overall details for SMART performance analysis are specified based on the current SMART design, would be utilized as a guide for the detailed performance analysis
Contrast analysis : A tutorial
Haans, A.
2018-01-01
Contrast analysis is a relatively simple but effective statistical method for testing theoretical predictions about differences between group means against the empirical data. Despite its advantages, contrast analysis is hardly used to date, perhaps because it is not implemented in a convenient
Interactive Controls Analysis (INCA)
Bauer, Frank H.
1989-01-01
Version 3.12 of INCA provides user-friendly environment for design and analysis of linear control systems. System configuration and parameters easily adjusted, enabling INCA user to create compensation networks and perform sensitivity analysis in convenient manner. Full complement of graphical routines makes output easy to understand. Written in Pascal and FORTRAN.
Marketing research cluster analysis
Directory of Open Access Journals (Sweden)
Marić Nebojša
2002-01-01
Full Text Available One area of applications of cluster analysis in marketing is identification of groups of cities and towns with similar demographic profiles. This paper considers main aspects of cluster analysis by an example of clustering 12 cities with the use of Minitab software.
SWOT ANALYSIS - CHINESE PETROLEUM
Directory of Open Access Journals (Sweden)
Chunlan Wang
2014-01-01
Full Text Available This article was written in early December 2013,combined with the historical development andthe latest data on the Chinese Petroleum carried SWOT- analysis. This paper discusses corporate resources, cost, management and external factorssuch as the political environment and the marketsupply and demand, conducted a comprehensiveand profound analysis.
de Roon, F.A.; Nijman, T.E.; Ter Horst, J.R.
2000-01-01
In this paper we evaluate applications of (return based) style analysis.The portfolio and positivity constraints imposed by style analysis are useful in constructing mimicking portfolios without short positions.Such mimicking portfolios can be used, e.g., to construct efficient portfolios of mutual
F.A. de Roon (Frans); T.E. Nijman (Theo); B.J.M. Werker
2000-01-01
textabstractIn this paper we evaluate applications of (return based) style analysis. The portfolio and positivity constraints imposed by style analysis are useful in constructing mimicking portfolios without short positions. Such mimicking portfolios can be used e.g. to construct efficient
Directory of Open Access Journals (Sweden)
Satu Elo
2014-02-01
Full Text Available Qualitative content analysis is commonly used for analyzing qualitative data. However, few articles have examined the trustworthiness of its use in nursing science studies. The trustworthiness of qualitative content analysis is often presented by using terms such as credibility, dependability, conformability, transferability, and authenticity. This article focuses on trustworthiness based on a review of previous studies, our own experiences, and methodological textbooks. Trustworthiness was described for the main qualitative content analysis phases from data collection to reporting of the results. We concluded that it is important to scrutinize the trustworthiness of every phase of the analysis process, including the preparation, organization, and reporting of results. Together, these phases should give a reader a clear indication of the overall trustworthiness of the study. Based on our findings, we compiled a checklist for researchers attempting to improve the trustworthiness of a content analysis study. The discussion in this article helps to clarify how content analysis should be reported in a valid and understandable manner, which would be of particular benefit to reviewers of scientific articles. Furthermore, we discuss that it is often difficult to evaluate the trustworthiness of qualitative content analysis studies because of defective data collection method description and/or analysis description.
Schraagen, J.M.C.
2000-01-01
Cognitive task analysis is defined as the extension of traditional task analysis techniques to yield information about the knowledge, thought processes and goal structures that underlie observable task performance. Cognitive task analyses are conducted for a wide variety of purposes, including the
DEFF Research Database (Denmark)
Damkilde, Lars
2007-01-01
Limit State analysis has a long history and many prominent researchers have contributed. The theoretical foundation is based on the upper- and lower-bound theorems which give a very comprehensive and elegant formulation on complicated physical problems. In the pre-computer age Limit State analysis...... also enabled engineers to solve practical problems within reinforced concrete, steel structures and geotechnics....
Verhoosel, C.V.; Scott, M.A.; Borden, M.J.; Borst, de R.; Hughes, T.J.R.; Mueller-Hoeppe, D.; Loehnert, S.; Reese, S.
2011-01-01
Isogeometric analysis is a versatile tool for failure analysis. On the one hand, the excellent control over the inter-element continuity conditions enables a natural incorporation of continuum constitutive relations that incorporate higher-order strain gradients, as in gradient plasticity or damage.
DEFF Research Database (Denmark)
Durbin, Richard; Eddy, Sean; Krogh, Anders Stærmose
This book provides an up-to-date and tutorial-level overview of sequence analysis methods, with particular emphasis on probabilistic modelling. Discussed methods include pairwise alignment, hidden Markov models, multiple alignment, profile searches, RNA secondary structure analysis, and phylogene...
International Nuclear Information System (INIS)
Arien, B.
2000-01-01
The objective of SCK-CEN's programme on reactor safety is to develop expertise in probabilistic and deterministic reactor safety analysis. The research programme consists of two main activities, in particular the development of software for reliability analysis of large systems and participation in the international PHEBUS-FP programme for severe accidents. Main achievements in 1999 are reported
Factorial Analysis of Profitability
Georgeta VINTILA; Ilie GHEORGHE; Ioana Mihaela POCAN; Madalina Gabriela ANGHEL
2012-01-01
The DuPont analysis system is based on decomposing the profitability ratio in factors of influence. This paper describes the factorial analysis of profitability based on the DuPont system. Significant importance is given to the impact on various indicators on the shares value and profitability.
Spool assembly support analysis
International Nuclear Information System (INIS)
Norman, B.F.
1994-01-01
This document provides the wind/seismic analysis and evaluation for the pump pit spool assemblies. Hand calculations were used for the analysis. UBC, AISC, and load factors were used in this evaluation. The results show that the actual loads are under the allowable loads and all requirements are met
International Nuclear Information System (INIS)
Hansen, J.D.
1976-01-01
This article discusses the partial wave analysis of two, three and four meson systems. The difference between the two approaches, referred to as amplitude and Ascoli analysis is discussed. Some of the results obtained with these methods are shown. (B.R.H.)
Enabling interdisciplinary analysis
L. M. Reid
1996-01-01
'New requirements for evaluating environmental conditions in the Pacific Northwest have led to increased demands for interdisciplinary analysis of complex environmental problems. Procedures for watershed analysis have been developed for use on public and private lands in Washington State (Washington Forest Practices Board 1993) and for federal lands in the Pacific...
Shot loading platform analysis
International Nuclear Information System (INIS)
Norman, B.F.
1994-01-01
This document provides the wind/seismic analysis and evaluation for the shot loading platform. Hand calculations were used for the analysis. AISC and UBC load factors were used in this evaluation. The results show that the actual loads are under the allowable loads and all requirements are met
Marketing research cluster analysis
Marić Nebojša
2002-01-01
One area of applications of cluster analysis in marketing is identification of groups of cities and towns with similar demographic profiles. This paper considers main aspects of cluster analysis by an example of clustering 12 cities with the use of Minitab software.
Towards Cognitive Component Analysis
DEFF Research Database (Denmark)
Hansen, Lars Kai; Ahrendt, Peter; Larsen, Jan
2005-01-01
Cognitive component analysis (COCA) is here defined as the process of unsupervised grouping of data such that the ensuing group structure is well-aligned with that resulting from human cognitive activity. We have earlier demonstrated that independent components analysis is relevant for representing...
Satu Elo; Maria Kääriäinen; Outi Kanste; Tarja Pölkki; Kati Utriainen; Helvi Kyngäs
2014-01-01
Qualitative content analysis is commonly used for analyzing qualitative data. However, few articles have examined the trustworthiness of its use in nursing science studies. The trustworthiness of qualitative content analysis is often presented by using terms such as credibility, dependability, conformability, transferability, and authenticity. This article focuses on trustworthiness based on a review of previous studie...
Interaction Analysis and Supervision.
Amidon, Edmund
This paper describes a model that uses interaction analysis as a tool to provide feedback to a teacher in a microteaching situation. The author explains how interaction analysis can be used for teacher improvement, describes the category system used in the model, the data collection methods used, and the feedback techniques found in the model. (JF)
Activation analysis. Chapter 4
International Nuclear Information System (INIS)
1976-01-01
The principle, sample and calibration standard preparation, activation by neutrons, charged particles and gamma radiation, sample transport after activation, activity measurement, and chemical sample processing are described for activation analysis. Possible applications are shown of nondestructive activation analysis. (J.P.)
Donahue, Craig J.; Rais, Elizabeth A.
2009-01-01
This lab experiment illustrates the use of thermogravimetric analysis (TGA) to perform proximate analysis on a series of coal samples of different rank. Peat and coke are also examined. A total of four exercises are described. These are dry exercises as students interpret previously recorded scans. The weight percent moisture, volatile matter,…
Ian M. Franks; Mike Hughes
2004-01-01
This book addresses and appropriately explains the notational analysis of technique, tactics, individual athlete/team exercise and work-rate in sport. The book offers guidance in: developing a system, analyzes of data, effective coaching using notational performance analysis and modeling sport behaviors. It updates and improves the 1997 edition
Directory of Open Access Journals (Sweden)
Ian M. Franks
2004-06-01
Full Text Available This book addresses and appropriately explains the notational analysis of technique, tactics, individual athlete/team exercise and work-rate in sport. The book offers guidance in: developing a system, analyzes of data, effective coaching using notational performance analysis and modeling sport behaviors. It updates and improves the 1997 edition
Allain, Rhett
2016-05-01
We currently live in a world filled with videos. There are videos on YouTube, feature movies and even videos recorded with our own cameras and smartphones. These videos present an excellent opportunity to not only explore physical concepts, but also inspire others to investigate physics ideas. With video analysis, we can explore the fantasy world in science-fiction films. We can also look at online videos to determine if they are genuine or fake. Video analysis can be used in the introductory physics lab and it can even be used to explore the make-believe physics embedded in video games. This book covers the basic ideas behind video analysis along with the fundamental physics principles used in video analysis. The book also includes several examples of the unique situations in which video analysis can be used.
Ramsay, J O
1997-01-01
Scientists today collect samples of curves and other functional observations. This monograph presents many ideas and techniques for such data. Included are expressions in the functional domain of such classics as linear regression, principal components analysis, linear modelling, and canonical correlation analysis, as well as specifically functional techniques such as curve registration and principal differential analysis. Data arising in real applications are used throughout for both motivation and illustration, showing how functional approaches allow us to see new things, especially by exploiting the smoothness of the processes generating the data. The data sets exemplify the wide scope of functional data analysis; they are drwan from growth analysis, meterology, biomechanics, equine science, economics, and medicine. The book presents novel statistical technology while keeping the mathematical level widely accessible. It is designed to appeal to students, to applied data analysts, and to experienced researc...
Systems engineering and analysis
Blanchard, Benjamin S
2010-01-01
For senior-level undergraduate and first and second year graduate systems engineering and related courses. A total life-cycle approach to systems and their analysis. This practical introduction to systems engineering and analysis provides the concepts, methodologies, models, and tools needed to understand and implement a total life-cycle approach to systems and their analysis. The authors focus first on the process of bringing systems into being--beginning with the identification of a need and extending that need through requirements determination, functional analysis and allocation, design synthesis, evaluation, and validation, operation and support, phase-out, and disposal. Next, the authors discuss the improvement of systems currently in being, showing that by employing the iterative process of analysis, evaluation, feedback, and modification, most systems in existence can be improved in their affordability, effectiveness, and stakeholder satisfaction.
International Nuclear Information System (INIS)
Ishii, Keizo
1997-01-01
Elemental analysis based on the particle induced x-ray emission (PIXE) is a novel technique to analyze trace elements. It is a very simple method, its sensitivity is very high, multiple elements in a sample can be simultaneously analyzed and a few 10 μg of a sample is enough to be analyzed. Owing to these characteristics, the PIXE analysis is now used in many fields (e.g. biology, medicine, dentistry, environmental pollution, archaeology, culture assets etc.). Fundamentals of the PIXE analysis are described here: the production of characteristic x-rays and inner shell ionization by heavy charged particles, the continuous background in PIXE spectrum, quantitative formulae of the PIXE analysis, the detection limit of PIXE analysis, etc. (author)
International Nuclear Information System (INIS)
Porten, D.R.; Crowe, R.D.
1994-01-01
The purpose of this accident safety analysis is to document in detail, analyses whose results were reported in summary form in the K Basins Safety Analysis Report WHC-SD-SNF-SAR-001. The safety analysis addressed the potential for release of radioactive and non-radioactive hazardous material located in the K Basins and their supporting facilities. The safety analysis covers the hazards associated with normal K Basin fuel storage and handling operations, fuel encapsulation, sludge encapsulation, and canister clean-up and disposal. After a review of the Criticality Safety Evaluation of the K Basin activities, the following postulated events were evaluated: Crane failure and casks dropped into loadout pit; Design basis earthquake; Hypothetical loss of basin water accident analysis; Combustion of uranium fuel following dryout; Crane failure and cask dropped onto floor of transfer area; Spent ion exchange shipment for burial; Hydrogen deflagration in ion exchange modules and filters; Release of Chlorine; Power availability and reliability; and Ashfall
International Nuclear Information System (INIS)
Goetz, A.; Gerring, M.; Svensson, O.; Brockhauser, S.
2012-01-01
Data Analysis Workbench (DAWB) is a new software tool being developed at the ESRF. Its goal is to provide a tool for both online data analysis which can be used on the beamlines and for offline data analysis which users can use during experiments or take home. The tool includes support for data visualization and work-flows. work-flows allow algorithms which exploit parallel architectures to be designed from existing high level modules for data analysis in combination with data collection. The workbench uses Passerelle as the work-flow engine and EDNA plug-ins for data analysis. Actors talking to Tango are used for sending commands to a limited set of hardware to start existing data collection algorithms. A Tango server allows work-flows to be executed from existing applications. There are scripting interfaces to Python, Javascript and SPEC. The current state at the ESRF is the workbench is in test on a selected number of beamlines. (authors)
J Olive, David
2017-01-01
This text presents methods that are robust to the assumption of a multivariate normal distribution or methods that are robust to certain types of outliers. Instead of using exact theory based on the multivariate normal distribution, the simpler and more applicable large sample theory is given. The text develops among the first practical robust regression and robust multivariate location and dispersion estimators backed by theory. The robust techniques are illustrated for methods such as principal component analysis, canonical correlation analysis, and factor analysis. A simple way to bootstrap confidence regions is also provided. Much of the research on robust multivariate analysis in this book is being published for the first time. The text is suitable for a first course in Multivariate Statistical Analysis or a first course in Robust Statistics. This graduate text is also useful for people who are familiar with the traditional multivariate topics, but want to know more about handling data sets with...
Field, Michael
2017-01-01
This book provides a rigorous introduction to the techniques and results of real analysis, metric spaces and multivariate differentiation, suitable for undergraduate courses. Starting from the very foundations of analysis, it offers a complete first course in real analysis, including topics rarely found in such detail in an undergraduate textbook such as the construction of non-analytic smooth functions, applications of the Euler-Maclaurin formula to estimates, and fractal geometry. Drawing on the author’s extensive teaching and research experience, the exposition is guided by carefully chosen examples and counter-examples, with the emphasis placed on the key ideas underlying the theory. Much of the content is informed by its applicability: Fourier analysis is developed to the point where it can be rigorously applied to partial differential equations or computation, and the theory of metric spaces includes applications to ordinary differential equations and fractals. Essential Real Analysis will appeal t...
Real analysis and applications
Botelho, Fabio Silva
2018-01-01
This textbook introduces readers to real analysis in one and n dimensions. It is divided into two parts: Part I explores real analysis in one variable, starting with key concepts such as the construction of the real number system, metric spaces, and real sequences and series. In turn, Part II addresses the multi-variable aspects of real analysis. Further, the book presents detailed, rigorous proofs of the implicit theorem for the vectorial case by applying the Banach fixed-point theorem and the differential forms concept to surfaces in Rn. It also provides a brief introduction to Riemannian geometry. With its rigorous, elegant proofs, this self-contained work is easy to read, making it suitable for undergraduate and beginning graduate students seeking a deeper understanding of real analysis and applications, and for all those looking for a well-founded, detailed approach to real analysis.
Nonactivation interaction analysis. Chapter 5
International Nuclear Information System (INIS)
1976-01-01
Analyses are described including the alpha scattering analysis, beta absorption and scattering analysis, gamma and X-ray absorption and scattering analysis, X-ray fluorescence analysis, neutron absorption and scattering analysis, Moessbauer effect application and an analysis based on the application of radiation ionizing effects. (J.P.)
Is activation analysis still active?
International Nuclear Information System (INIS)
Chai Zhifang
2001-01-01
This paper reviews some aspects of neutron activation analysis (NAA), covering instrumental neutron activation analysis (INAA), k 0 method, prompt gamma-ray neutron activation analysis (PGNAA), radiochemical neutron activation analysis (RNAA) and molecular activation analysis (MAA). The comparison of neutron activation analysis with other analytical techniques are also made. (author)
International Nuclear Information System (INIS)
González Caballero, I; Cuesta Noriega, A; Rodríguez Marrero, A; Fernández del Castillo, E
2012-01-01
The analysis of the complex LHC data usually follows a standard path that aims at minimizing not only the amount of data but also the number of observables used. After a number of steps of slimming and skimming the data, the remaining few terabytes of ROOT files hold a selection of the events and a flat structure for the variables needed that can be more easily inspected and traversed in the final stages of the analysis. PROOF arises at this point as an efficient mechanism to distribute the analysis load by taking advantage of all the cores in modern CPUs through PROOF Lite, or by using PROOF Cluster or PROOF on Demand tools to build dynamic PROOF cluster on computing facilities with spare CPUs. However using PROOF at the level required for a serious analysis introduces some difficulties that may scare new adopters. We have developed the PROOF Analysis Framework (PAF) to facilitate the development of new analysis by uniformly exposing the PROOF related configurations across technologies and by taking care of the routine tasks as much as possible. We describe the details of the PAF implementation as well as how we succeeded in engaging a group of CMS physicists to use PAF as their daily analysis framework.
Hazard Analysis Database Report
Energy Technology Data Exchange (ETDEWEB)
GAULT, G.W.
1999-10-13
The Hazard Analysis Database was developed in conjunction with the hazard analysis activities conducted in accordance with DOE-STD-3009-94, Preparation Guide for US Department of Energy Nonreactor Nuclear Facility Safety Analysis Reports, for the Tank Waste Remediation System (TWRS) Final Safety Analysis Report (FSAR). The FSAR is part of the approved TWRS Authorization Basis (AB). This document describes, identifies, and defines the contents and structure of the TWRS FSAR Hazard Analysis Database and documents the configuration control changes made to the database. The TWRS Hazard Analysis Database contains the collection of information generated during the initial hazard evaluations and the subsequent hazard and accident analysis activities. The database supports the preparation of Chapters 3,4, and 5 of the TWRS FSAR and the USQ process and consists of two major, interrelated data sets: (1) Hazard Evaluation Database--Data from the results of the hazard evaluations; and (2) Hazard Topography Database--Data from the system familiarization and hazard identification.
Containment vessel stability analysis
International Nuclear Information System (INIS)
Harstead, G.A.; Morris, N.F.; Unsal, A.I.
1983-01-01
The stability analysis for a steel containment shell is presented herein. The containment is a freestanding shell consisting of a vertical cylinder with a hemispherical dome. It is stiffened by large ring stiffeners and relatively small longitudinal stiffeners. The containment vessel is subjected to both static and dynamic loads which can cause buckling. These loads must be combined prior to their use in a stability analysis. The buckling loads were computed with the aid of the ASME Code case N-284 used in conjunction with general purpose computer codes and in-house programs. The equations contained in the Code case were used to compute the knockdown factors due to shell imperfections. After these knockdown factors were applied to the critical stress states determined by freezing the maximum dynamic stresses and combining them with other static stresses, a linear bifurcation analysis was carried out with the aid of the BOSOR4 program. Since the containment shell contained large penetrations, the Code case had to be supplemented by a local buckling analysis of the shell area surrounding the largest penetration. This analysis was carried out with the aid of the NASTRAN program. Although the factor of safety against buckling obtained in this analysis was satisfactory, it is claimed that the use of the Code case knockdown factors are unduly conservative when applied to the analysis of buckling around penetrations. (orig.)
International Nuclear Information System (INIS)
Thompson, W.A. Jr.
1979-11-01
This paper briefly describes WASH 1400 and the Lewis report. It attempts to define basic concepts such as risk and risk analysis, common mode failure, and rare event. Several probabilistic models which go beyond the WASH 1400 methodology are introduced; the common characteristic of these models is that they recognize explicitly that risk analysis is time dependent whereas WASH 1400 takes a per demand failure rate approach which obscures the important fact that accidents are time related. Further, the presentation of a realistic risk analysis should recognize that there are various risks which compete with one another for the lives of the individuals at risk. A way of doing this is suggested
International Nuclear Information System (INIS)
Kartiwa Sumadi; Yayah Rohayati
1996-01-01
The 'monazit' analytical program has been set up for routine work of Rare Earth Elements analysis in the monazite and xenotime minerals samples. Total relative error of the analysis is very low, less than 2.50%, and the reproducibility of counting statistic and stability of the instrument were very excellent. The precision and accuracy of the analytical program are very good with the maximum percentage relative are 5.22% and 1.61%, respectively. The mineral compositions of the 30 monazite samples have been also calculated using their chemical constituents, and the results were compared to the grain counting microscopic analysis
Methods of Multivariate Analysis
Rencher, Alvin C
2012-01-01
Praise for the Second Edition "This book is a systematic, well-written, well-organized text on multivariate analysis packed with intuition and insight . . . There is much practical wisdom in this book that is hard to find elsewhere."-IIE Transactions Filled with new and timely content, Methods of Multivariate Analysis, Third Edition provides examples and exercises based on more than sixty real data sets from a wide variety of scientific fields. It takes a "methods" approach to the subject, placing an emphasis on how students and practitioners can employ multivariate analysis in real-life sit
Dunham, Ken
2014-01-01
The rapid growth and development of Android-based devices has resulted in a wealth of sensitive information on mobile devices that offer minimal malware protection. This has created an immediate demand for security professionals that understand how to best approach the subject of Android malware threats and analysis.In Android Malware and Analysis, Ken Dunham, renowned global malware expert and author, teams up with international experts to document the best tools and tactics available for analyzing Android malware. The book covers both methods of malware analysis: dynamic and static.This tact
Aven, Terje
2012-01-01
Foundations of Risk Analysis presents the issues core to risk analysis - understanding what risk means, expressing risk, building risk models, addressing uncertainty, and applying probability models to real problems. The author provides the readers with the knowledge and basic thinking they require to successfully manage risk and uncertainty to support decision making. This updated edition reflects recent developments on risk and uncertainty concepts, representations and treatment. New material in Foundations of Risk Analysis includes:An up to date presentation of how to understand, define and
International Nuclear Information System (INIS)
Ko, Myeong Su; Kim, Tae Hwa; Park, Gyu Hyeon; Yang, Jong Beom; Oh, Chang Hwan; Lee, Kyoung Hye
2010-04-01
This textbook describes instrument analysis in easy way with twelve chapters. The contents of the book are pH measurement on principle, pH meter, pH measurement, examples of the experiments, centrifugation, Absorptiometry, Fluorescent method, Atomic absorption analysis, Gas-chromatography, Gas chromatography-mass spectrometry, High performance liquid chromatography liquid chromatograph-mass spectrometry, Electrophoresis on practical case and analysis of the result and examples, PCR on principle, device, application and examples and Enzyme-linked immunosorbent assay with indirect ELISA, sandwich ELISA and ELISA reader.
International Nuclear Information System (INIS)
Williams, Mike; Egede, Ulrik; Paterson, Stuart
2011-01-01
The distributed analysis experience to date at LHCb has been positive: job success rates are high and wait times for high-priority jobs are low. LHCb users access the grid using the GANGA job-management package, while the LHCb virtual organization manages its resources using the DIRAC package. This clear division of labor has benefitted LHCb and its users greatly; it is a major reason why distributed analysis at LHCb has been so successful. The newly formed LHCb distributed analysis support team has also proved to be a success.
Factor analysis and scintigraphy
International Nuclear Information System (INIS)
Di Paola, R.; Penel, C.; Bazin, J.P.; Berche, C.
1976-01-01
The goal of factor analysis is usually to achieve reduction of a large set of data, extracting essential features without previous hypothesis. Due to the development of computerized systems, the use of largest sampling, the possibility of sequential data acquisition and the increase of dynamic studies, the problem of data compression can be encountered now in routine. Thus, results obtained for compression of scintigraphic images were first presented. Then possibilities given by factor analysis for scan processing were discussed. At last, use of this analysis for multidimensional studies and specially dynamic studies were considered for compression and processing [fr
Energy Technology Data Exchange (ETDEWEB)
Ko, Myeong Su; Kim, Tae Hwa; Park, Gyu Hyeon; Yang, Jong Beom; Oh, Chang Hwan; Lee, Kyoung Hye
2010-04-15
This textbook describes instrument analysis in easy way with twelve chapters. The contents of the book are pH measurement on principle, pH meter, pH measurement, examples of the experiments, centrifugation, Absorptiometry, Fluorescent method, Atomic absorption analysis, Gas-chromatography, Gas chromatography-mass spectrometry, High performance liquid chromatography liquid chromatograph-mass spectrometry, Electrophoresis on practical case and analysis of the result and examples, PCR on principle, device, application and examples and Enzyme-linked immunosorbent assay with indirect ELISA, sandwich ELISA and ELISA reader.
DEFF Research Database (Denmark)
Raket, Lars Lau
We propose a direction it the field of statistics which we will call functional object analysis. This subfields considers the analysis of functional objects defined on continuous domains. In this setting we will focus on model-based statistics, with a particularly emphasis on mixed......-effect formulations, where the observed functional signal is assumed to consist of both fixed and random functional effects. This thesis takes the initial steps toward the development of likelihood-based methodology for functional objects. We first consider analysis of functional data defined on high...
Bayesian nonparametric data analysis
Müller, Peter; Jara, Alejandro; Hanson, Tim
2015-01-01
This book reviews nonparametric Bayesian methods and models that have proven useful in the context of data analysis. Rather than providing an encyclopedic review of probability models, the book’s structure follows a data analysis perspective. As such, the chapters are organized by traditional data analysis problems. In selecting specific nonparametric models, simpler and more traditional models are favored over specialized ones. The discussed methods are illustrated with a wealth of examples, including applications ranging from stylized examples to case studies from recent literature. The book also includes an extensive discussion of computational methods and details on their implementation. R code for many examples is included in on-line software pages.
International Nuclear Information System (INIS)
Ramirez T, J.J.; Lopez M, J.; Sandoval J, A.R.; Villasenor S, P.; Aspiazu F, J.A.
2001-01-01
An elemental analysis, metallographic and of phases was realized in order to determine the oxidation states of Fe contained in three metallic pieces: block, plate and cylinder of unknown material. Results are presented from the elemental analysis which was carried out in the Tandem Accelerator of ININ by Proton induced X-ray emission (PIXE). The phase analysis was carried out by X-ray diffraction which allowed to know the type of alloy or alloys formed. The combined application of nuclear techniques with metallographic techniques allows the integral characterization of industrial metals. (Author)
Ash, Robert B; Lukacs, E
1972-01-01
Real Analysis and Probability provides the background in real analysis needed for the study of probability. Topics covered range from measure and integration theory to functional analysis and basic concepts of probability. The interplay between measure theory and topology is also discussed, along with conditional probability and expectation, the central limit theorem, and strong laws of large numbers with respect to martingale theory.Comprised of eight chapters, this volume begins with an overview of the basic concepts of the theory of measure and integration, followed by a presentation of var
Iremonger, M J
1982-01-01
BASIC Stress Analysis aims to help students to become proficient at BASIC programming by actually using it in an important engineering subject. It also enables the student to use computing as a means of learning stress analysis because writing a program is analogous to teaching-it is necessary to understand the subject matter. The book begins by introducing the BASIC approach and the concept of stress analysis at first- and second-year undergraduate level. Subsequent chapters contain a summary of relevant theory, worked examples containing computer programs, and a set of problems. Topics c
Fundamentals of mathematical analysis
Paul J Sally, Jr
2013-01-01
This is a textbook for a course in Honors Analysis (for freshman/sophomore undergraduates) or Real Analysis (for junior/senior undergraduates) or Analysis-I (beginning graduates). It is intended for students who completed a course in "AP Calculus", possibly followed by a routine course in multivariable calculus and a computational course in linear algebra. There are three features that distinguish this book from many other books of a similar nature and which are important for the use of this book as a text. The first, and most important, feature is the collection of exercises. These are spread
Systems analysis-independent analysis and verification
Energy Technology Data Exchange (ETDEWEB)
Badin, J.S.; DiPietro, J.P. [Energetics, Inc., Columbia, MD (United States)
1995-09-01
The DOE Hydrogen Program is supporting research, development, and demonstration activities to overcome the barriers to the integration of hydrogen into the Nation`s energy infrastructure. Much work is required to gain acceptance of hydrogen energy system concepts and to develop them for implementation. A systems analysis database has been created that includes a formal documentation of technology characterization profiles and cost and performance information. Through a systematic and quantitative approach, system developers can understand and address important issues and thereby assure effective and timely commercial implementation. This project builds upon and expands the previously developed and tested pathway model and provides the basis for a consistent and objective analysis of all hydrogen energy concepts considered by the DOE Hydrogen Program Manager. This project can greatly accelerate the development of a system by minimizing the risk of costly design evolutions, and by stimulating discussions, feedback, and coordination of key players and allows them to assess the analysis, evaluate the trade-offs, and to address any emerging problem areas. Specific analytical studies will result in the validation of the competitive feasibility of the proposed system and identify system development needs. Systems that are investigated include hydrogen bromine electrolysis, municipal solid waste gasification, electro-farming (biomass gasifier and PEM fuel cell), wind/hydrogen hybrid system for remote sites, home electrolysis and alternate infrastructure options, renewable-based electrolysis to fuel PEM fuel cell vehicle fleet, and geothermal energy used to produce hydrogen. These systems are compared to conventional and benchmark technologies. Interim results and findings are presented. Independent analyses emphasize quality, integrity, objectivity, a long-term perspective, corporate memory, and the merging of technical, economic, operational, and programmatic expertise.
Plasma data analysis using statistical analysis system
International Nuclear Information System (INIS)
Yoshida, Z.; Iwata, Y.; Fukuda, Y.; Inoue, N.
1987-01-01
Multivariate factor analysis has been applied to a plasma data base of REPUTE-1. The characteristics of the reverse field pinch plasma in REPUTE-1 are shown to be explained by four independent parameters which are described in the report. The well known scaling laws F/sub chi/ proportional to I/sub p/, T/sub e/ proportional to I/sub p/, and tau/sub E/ proportional to N/sub e/ are also confirmed. 4 refs., 8 figs., 1 tab
Summary Analysis: Hanford Site Composite Analysis Update
Energy Technology Data Exchange (ETDEWEB)
Nichols, W. E. [CH2M HILL Plateau Remediation Company, Richland, WA (United States); Lehman, L. L. [CH2M HILL Plateau Remediation Company, Richland, WA (United States)
2017-06-05
The Hanford Site’s currently maintained Composite Analysis, originally completed in 1998, requires an update. A previous update effort was undertaken by the U.S. Department of Energy (DOE) in 2001-2005, but was ended before completion to allow the Tank Closure & Waste Management Environmental Impact Statement (TC&WM EIS) (DOE/EIS-0391) to be prepared without potential for conflicting sitewide models. This EIS was issued in 2012, and the deferral was ended with guidance in memorandum “Modeling to Support Regulatory Decision Making at Hanford” (Williams, 2012) provided with the aim of ensuring subsequent modeling is consistent with the EIS.
The feasibility study of BGO crystal applied to natural gamma ray spectral logging in deep wells
International Nuclear Information System (INIS)
Li Wen; Zhang Jingyi
1992-01-01
The experimental research on the feasibility of BGO crystal application to natural gamma ray spectral logging was performed under laboratory conditions. The natural radioactive standard sources and heater case were used to simulate the realistic logging environment. The experimental results of BGO detector show that the spectrum stabilization can be effectively realized, and the temperature effect of energy resolution has no obvious impact on spectrum analysis when the temperature of detector ranging from 20 degree C to 60 degree C. The temperature effect of the output pulse height is the main cause of the system error in spectrum analysis, which will surely be reduced in realistic logging by the low temperature gradient. The experimental results of CsI(Na) detector support this conclusion. A suggestion for further work is proposed at the end of this paper
The feasibility study of BGO crystal applied to natural gamma ray spectral logging in deep wells
Energy Technology Data Exchange (ETDEWEB)
Wen, Li; Jingyi, Zhang [Qinghua Univ., Beijing (China). Dept. of Engineering Physics; Chongfan, He; Jingying, Liao [Shanghai Silicate Institute (China)
1992-09-01
The experimental research on the feasibility of BGO crystal application to natural gamma ray spectral logging was performed under laboratory conditions. The natural radioactive standard sources and heater case were used to simulate the realistic logging environment. The experimental results of BGO detector show that the spectrum stabilization can be effectively realized, and the temperature effect of energy resolution has no obvious impact on spectrum analysis when the temperature of detector ranging from 20 degree C to 60 degree C. The temperature effect of the output pulse height is the main cause of the system error in spectrum analysis, which will surely be reduced in realistic logging by the low temperature gradient. The experimental results of CsI(Na) detector support this conclusion. A suggestion for further work is proposed at the end of this paper.
Reddy, V R; Reddy, T G; Reddy, P Y; Reddy, K R
2003-01-01
An AC modulation technique is described to convert stochastic signal variations into an amplitude variation and its retrieval through Fourier analysis. It is shown that this AC detection of signals of stochastic processes when processed through auto- and cross-correlation techniques improve the signal-to-noise ratio; the correlation techniques serve a similar purpose of frequency and phase filtering as that of phase-sensitive detection. A few model calculations applied to nuclear spectroscopy measurements such as Angular Correlations, Mossbauer spectroscopy and Pulse Height Analysis reveal considerable improvement in the sensitivity of signal detection. Experimental implementation of the technique is presented in terms of amplitude variations of harmonics representing the derivatives of normal spectra. Improved detection sensitivity to spectral variations is shown to be significant. These correlation techniques are general and can be made applicable to all the fields of particle counting where measurements ar...
Historical review of transuranic analytical methodology
International Nuclear Information System (INIS)
Wessman, R.A.; Leventhal, L.
1977-01-01
The methods used for the analysis of the transuranic elements neptunium, plutonium, americium, and curium are reviewed. Particular attention is given to those procedures suitable for low-level environmental and biological assessment. The preparation of such samples, derived from a variety of sources and incorporated in different masses and volumes of soils, water, vegetation, and biological matrices, is described. A comprehensive discussion of matrix dissolution and equilibration techniques indicates that complete dissolution employing tracer techniques is to be preferred. Leaching of large soil samples is satisfactory for worldwide fallout. Individual and sequential separation and purification procedures, including precipitation, extraction, and ion exchange methods, are described. Sample preparation techniques for radiometric assay, such as stippling, evaporation, and plating, are reviewed. For both tracer and nuclide assay, electroplating provides the thin samples suitable for alpha pulse height analysis. Past and current counting methods for the transuranic alpha, beta, and gamma emitters are reviewed
A new multiparametry system for the acquisition, correlation and processing of nuclear data
International Nuclear Information System (INIS)
Toledo Acosta, Rene; Osorio Deliz, Juan F.; Arista Romeu, Eduardo; Arteche Diaz, Raul; Lopez Torres, Ernesto
1999-01-01
A four spectrometric channels multiparametry System for the acquisition, correlation and processing of nuclear data is described. The design was based upon an TMS320-5x DSP substituting the one developed with the 80C186 microprocessor. The current configuration is similar to the previous one since consists of a PC incorporates board, an signals adapter and the software for the processing of the information.The system can be used for the multiparametry analysis, in pulse height analysis or in any combination of both according to the necessities. The System facilitates the stabilization of the zeros of the A/D converters. The block outline of the electronic design is included and they stand out some functions of the system
He, Jingrui
2012-01-01
This book focuses on rare category analysis where the majority classes have smooth distributions and the minority classes exhibit the compactness property. It focuses on challenging cases where the support regions of the majority and minority classes overlap.
Longitudinal categorical data analysis
Sutradhar, Brajendra C
2014-01-01
This is the first book in longitudinal categorical data analysis with parametric correlation models developed based on dynamic relationships among repeated categorical responses. This book is a natural generalization of the longitudinal binary data analysis to the multinomial data setup with more than two categories. Thus, unlike the existing books on cross-sectional categorical data analysis using log linear models, this book uses multinomial probability models both in cross-sectional and longitudinal setups. A theoretical foundation is provided for the analysis of univariate multinomial responses, by developing models systematically for the cases with no covariates as well as categorical covariates, both in cross-sectional and longitudinal setups. In the longitudinal setup, both stationary and non-stationary covariates are considered. These models have also been extended to the bivariate multinomial setup along with suitable covariates. For the inferences, the book uses the generalized quasi-likelihood as w...
International Nuclear Information System (INIS)
1981-09-01
Suggestion are made concerning the method of the fault tree analysis, the use of certain symbols in the examination of system failures. This purpose of the fault free analysis is to find logical connections of component or subsystem failures leading to undesirable occurrances. The results of these examinations are part of the system assessment concerning operation and safety. The objectives of the analysis are: systematical identification of all possible failure combinations (causes) leading to a specific undesirable occurrance, finding of reliability parameters such as frequency of failure combinations, frequency of the undesirable occurrance or non-availability of the system when required. The fault tree analysis provides a near and reconstructable documentation of the examination. (orig./HP) [de
Denker, A; Rauschenberg, J; Röhrich, J; Strub, E
2006-01-01
Materials analysis with ion beams exploits the interaction of ions with the electrons and nuclei in the sample. Among the vast variety of possible analytical techniques available with ion beams we will restrain to ion beam analysis with ion beams in the energy range from one to several MeV per mass unit. It is possible to use either the back-scattered projectiles (RBS – Rutherford Back Scattering) or the recoiled atoms itself (ERDA – Elastic Recoil Detection Analysis) from the elastic scattering processes. These techniques allow the simultaneous and absolute determination of stoichiometry and depth profiles of the detected elements. The interaction of the ions with the electrons in the sample produces holes in the inner electronic shells of the sample atoms, which recombine and emit X-rays characteristic for the element in question. Particle Induced X-ray Emission (PIXE) has shown to be a fast technique for the analysis of elements with an atomic number above 11.
DEFF Research Database (Denmark)
Vatrapu, Ravi; Mukkamala, Raghava Rao; Hussain, Abid
2016-01-01
, conceptual and formal models of social data, and an analytical framework for combining big social data sets with organizational and societal data sets. Three empirical studies of big social data are presented to illustrate and demonstrate social set analysis in terms of fuzzy set-theoretical sentiment...... automata and agent-based modeling). However, when it comes to organizational and societal units of analysis, there exists no approach to conceptualize, model, analyze, explain, and predict social media interactions as individuals' associations with ideas, values, identities, and so on. To address...... analysis, crisp set-theoretical interaction analysis, and event-studies-oriented set-theoretical visualizations. Implications for big data analytics, current limitations of the set-theoretical approach, and future directions are outlined....
PWR systems transient analysis
International Nuclear Information System (INIS)
Kennedy, M.F.; Peeler, G.B.; Abramson, P.B.
1985-01-01
Analysis of transients in pressurized water reactor (PWR) systems involves the assessment of the response of the total plant, including primary and secondary coolant systems, steam piping and turbine (possibly including the complete feedwater train), and various control and safety systems. Transient analysis is performed as part of the plant safety analysis to insure the adequacy of the reactor design and operating procedures and to verify the applicable plant emergency guidelines. Event sequences which must be examined are developed by considering possible failures or maloperations of plant components. These vary in severity (and calculational difficulty) from a series of normal operational transients, such as minor load changes, reactor trips, valve and pump malfunctions, up to the double-ended guillotine rupture of a primary reactor coolant system pipe known as a Large Break Loss of Coolant Accident (LBLOCA). The focus of this paper is the analysis of all those transients and accidents except loss of coolant accidents
Full closure strategic analysis.
2014-07-01
The full closure strategic analysis was conducted to create a decision process whereby full roadway : closures for construction and maintenance activities can be evaluated and approved or denied by CDOT : Traffic personnel. The study reviewed current...
Electrical Subsurface Grounding Analysis
International Nuclear Information System (INIS)
J.M. Calle
2000-01-01
The purpose and objective of this analysis is to determine the present grounding requirements of the Exploratory Studies Facility (ESF) subsurface electrical system and to verify that the actual grounding system and devices satisfy the requirements
DEFF Research Database (Denmark)
Skrypnyuk, Nataliya; Nielson, Flemming; Pilegaard, Henrik
2009-01-01
We present the ongoing work on the pathway analysis of a stochastic calculus. Firstly we present a particular stochastic calculus that we have chosen for our modeling - the Interactive Markov Chains calculus, IMC for short. After that we specify a few restrictions that we have introduced into the...... into the syntax of IMC in order to make our analysis feasible. Finally we describe the analysis itself together with several theoretical results that we have proved for it.......We present the ongoing work on the pathway analysis of a stochastic calculus. Firstly we present a particular stochastic calculus that we have chosen for our modeling - the Interactive Markov Chains calculus, IMC for short. After that we specify a few restrictions that we have introduced...
Canonical Information Analysis
DEFF Research Database (Denmark)
Vestergaard, Jacob Schack; Nielsen, Allan Aasbjerg
2015-01-01
is replaced by the information theoretical, entropy based measure mutual information, which is a much more general measure of association. We make canonical information analysis feasible for large sample problems, including for example multispectral images, due to the use of a fast kernel density estimator......Canonical correlation analysis is an established multivariate statistical method in which correlation between linear combinations of multivariate sets of variables is maximized. In canonical information analysis introduced here, linear correlation as a measure of association between variables...... for entropy estimation. Canonical information analysis is applied successfully to (1) simple simulated data to illustrate the basic idea and evaluate performance, (2) fusion of weather radar and optical geostationary satellite data in a situation with heavy precipitation, and (3) change detection in optical...
Qualitative Data Analysis Strategies
Greaves, Kristoffer
2014-01-01
A set of concept maps for qualitative data analysis strategies, inspired by Corbin, JM & Strauss, AL 2008, Basics of qualitative research: Techniques and procedures for developing grounded theory, 3rd edn, Sage Publications, Inc, Thousand Oaks, California.
Statistical data analysis handbook
National Research Council Canada - National Science Library
Wall, Francis J
1986-01-01
It must be emphasized that this is not a text book on statistics. Instead it is a working tool that presents data analysis in clear, concise terms which can be readily understood even by those without formal training in statistics...
Fanfani, Alessandra; Sanches, Jose Afonso; Andreeva, Julia; Bagliesi, Giusepppe; Bauerdick, Lothar; Belforte, Stefano; Bittencourt Sampaio, Patricia; Bloom, Ken; Blumenfeld, Barry; Bonacorsi, Daniele; Brew, Chris; Calloni, Marco; Cesini, Daniele; Cinquilli, Mattia; Codispoti, Giuseppe; D'Hondt, Jorgen; Dong, Liang; Dongiovanni, Danilo; Donvito, Giacinto; Dykstra, David; Edelmann, Erik; Egeland, Ricky; Elmer, Peter; Eulisse, Giulio; Evans, Dave; Fanzago, Federica; Farina, Fabio; Feichtinger, Derek; Fisk, Ian; Flix, Josep; Grandi, Claudio; Guo, Yuyi; Happonen, Kalle; Hernandez, Jose M; Huang, Chih-Hao; Kang, Kejing; Karavakis, Edward; Kasemann, Matthias; Kavka, Carlos; Khan, Akram; Kim, Bockjoo; Klem, Jukka; Koivumaki, Jesper; Kress, Thomas; Kreuzer, Peter; Kurca, Tibor; Kuznetsov, Valentin; Lacaprara, Stefano; Lassila-Perini, Kati; Letts, James; Linden, Tomas; Lueking, Lee; Maes, Joris; Magini, Nicolo; Maier, Gerhild; McBride, Patricia; Metson, Simon; Miccio, Vincenzo; Padhi, Sanjay; Pi, Haifeng; Riahi, Hassen; Riley, Daniel; Rossman, Paul; Saiz, Pablo; Sartirana, Andrea; Sciaba, Andrea; Sekhri, Vijay; Spiga, Daniele; Tuura, Lassi; Vaandering, Eric; Vanelderen, Lukas; Van Mulders, Petra; Vedaee, Aresh; Villella, Ilaria; Wicklund, Eric; Wildish, Tony; Wissing, Christoph; Wurthwein, Frank
2009-01-01
The CMS experiment expects to manage several Pbytes of data each year during the LHC programme, distributing them over many computing sites around the world and enabling data access at those centers for analysis. CMS has identified the distributed sites as the primary location for physics analysis to support a wide community with thousands potential users. This represents an unprecedented experimental challenge in terms of the scale of distributed computing resources and number of user. An overview of the computing architecture, the software tools and the distributed infrastructure is reported. Summaries of the experience in establishing efficient and scalable operations to get prepared for CMS distributed analysis are presented, followed by the user experience in their current analysis activities.
NOAA's Inundation Analysis Tool
National Oceanic and Atmospheric Administration, Department of Commerce — Coastal storms and other meteorological phenomenon can have a significant impact on how high water levels rise and how often. The inundation analysis program is...
Multidimensional nonlinear descriptive analysis
Nishisato, Shizuhiko
2006-01-01
Quantification of categorical, or non-numerical, data is a problem that scientists face across a wide range of disciplines. Exploring data analysis in various areas of research, such as the social sciences and biology, Multidimensional Nonlinear Descriptive Analysis presents methods for analyzing categorical data that are not necessarily sampled randomly from a normal population and often involve nonlinear relations. This reference not only provides an overview of multidimensional nonlinear descriptive analysis (MUNDA) of discrete data, it also offers new results in a variety of fields. The first part of the book covers conceptual and technical preliminaries needed to understand the data analysis in subsequent chapters. The next two parts contain applications of MUNDA to diverse data types, with each chapter devoted to one type of categorical data, a brief historical comment, and basic skills peculiar to the data types. The final part examines several problems and then concludes with suggestions for futu...
Brackney, Larry; Parker, Andrew; Long, Nicholas; Metzger, Ian; Dean, Jesse; Lisell, Lars
2016-04-12
A building energy analysis system includes a building component library configured to store a plurality of building components, a modeling tool configured to access the building component library and create a building model of a building under analysis using building spatial data and using selected building components of the plurality of building components stored in the building component library, a building analysis engine configured to operate the building model and generate a baseline energy model of the building under analysis and further configured to apply one or more energy conservation measures to the baseline energy model in order to generate one or more corresponding optimized energy models, and a recommendation tool configured to assess the one or more optimized energy models against the baseline energy model and generate recommendations for substitute building components or modifications.
Water Quality Analysis Simulation
The Water Quality analysis simulation Program, an enhancement of the original WASP. This model helps users interpret and predict water quality responses to natural phenomena and man-made pollution for variious pollution management decisions.
... Plasma Free Metanephrines Platelet Count Platelet Function Tests Pleural Fluid Analysis PML-RARA Porphyrin Tests Potassium Prealbumin ... is being tested? Synovial fluid is a thick liquid that acts as a lubricant for the body's ...
Hytönen, Tuomas; Veraar, Mark; Weis, Lutz
The present volume develops the theory of integration in Banach spaces, martingales and UMD spaces, and culminates in a treatment of the Hilbert transform, Littlewood-Paley theory and the vector-valued Mihlin multiplier theorem. Over the past fifteen years, motivated by regularity problems in evolution equations, there has been tremendous progress in the analysis of Banach space-valued functions and processes. The contents of this extensive and powerful toolbox have been mostly scattered around in research papers and lecture notes. Collecting this diverse body of material into a unified and accessible presentation fills a gap in the existing literature. The principal audience that we have in mind consists of researchers who need and use Analysis in Banach Spaces as a tool for studying problems in partial differential equations, harmonic analysis, and stochastic analysis. Self-contained and offering complete proofs, this work is accessible to graduate students and researchers with a background in functional an...
Analysis Streamlining in ATLAS
Heinrich, Lukas; The ATLAS collaboration
2018-01-01
We present recent work within the ATLAS collaboration centrally provide tools to facilitate analysis management and highly automated container-based analysis execution in order to both enable non-experts to benefit from these best practices as well as the collaboration to track and re-execute analyses indpendently, e.g. during their review phase. Through integration with the ATLAS GLANCE system, users can request a pre-configured, but customizable version control setup, including continuous integration for automated build and testing as well as continuous Linux Container image building for software preservation purposes. As analyses typically require many individual steps, analysis workflow pipelines can then be defined using such images and the yadage workflow description language. The integration into the workflow exection service REANA allows the interactive or automated reproduction of the main analysis results by orchestrating a large number of container jobs using the Kubernetes. For long-term archival,...
Wolff, Thomas H; Shubin, Carol
2003-01-01
This book demonstrates how harmonic analysis can provide penetrating insights into deep aspects of modern analysis. It is both an introduction to the subject as a whole and an overview of those branches of harmonic analysis that are relevant to the Kakeya conjecture. The usual background material is covered in the first few chapters: the Fourier transform, convolution, the inversion theorem, the uncertainty principle and the method of stationary phase. However, the choice of topics is highly selective, with emphasis on those frequently used in research inspired by the problems discussed in the later chapters. These include questions related to the restriction conjecture and the Kakeya conjecture, distance sets, and Fourier transforms of singular measures. These problems are diverse, but often interconnected; they all combine sophisticated Fourier analysis with intriguing links to other areas of mathematics and they continue to stimulate first-rate work. The book focuses on laying out a solid foundation for fu...
Water Quality Analysis Simulation
U.S. Environmental Protection Agency — The Water Quality analysis simulation Program, an enhancement of the original WASP. This model helps users interpret and predict water quality responses to natural...
Federal Laboratory Consortium — Provides engineering design of aircraft components, subsystems and installations using Pro/E, Anvil 1000, CADKEY 97, AutoCAD 13. Engineering analysis tools include...
CSIR Research Space (South Africa)
Khuluse, S
2009-04-01
Full Text Available ) determination of the distribution of the damage and (iii) preparation of products that enable prediction of future risk events. The methodology provided by extreme value theory can also be a powerful tool in risk analysis...
Ziemer, William P
2017-01-01
This first year graduate text is a comprehensive resource in real analysis based on a modern treatment of measure and integration. Presented in a definitive and self-contained manner, it features a natural progression of concepts from simple to difficult. Several innovative topics are featured, including differentiation of measures, elements of Functional Analysis, the Riesz Representation Theorem, Schwartz distributions, the area formula, Sobolev functions and applications to harmonic functions. Together, the selection of topics forms a sound foundation in real analysis that is particularly suited to students going on to further study in partial differential equations. This second edition of Modern Real Analysis contains many substantial improvements, including the addition of problems for practicing techniques, and an entirely new section devoted to the relationship between Lebesgue and improper integrals. Aimed at graduate students with an understanding of advanced calculus, the text will also appeal to mo...
DEFF Research Database (Denmark)
Fischer, Paul; Hilbert, Astrid
2012-01-01
We introduce a platform which supplies an easy-to-handle, interactive, extendable, and fast analysis tool for time series analysis. In contrast to other software suits like Maple, Matlab, or R, which use a command-line-like interface and where the user has to memorize/look-up the appropriate...... commands, our application is select-and-click-driven. It allows to derive many different sequences of deviations for a given time series and to visualize them in different ways in order to judge their expressive power and to reuse the procedure found. For many transformations or model-ts, the user may...... choose between manual and automated parameter selection. The user can dene new transformations and add them to the system. The application contains efficient implementations of advanced and recent techniques for time series analysis including techniques related to extreme value analysis and filtering...
International Nuclear Information System (INIS)
Holland, W.E.
1980-02-01
A method was developed to determine if boron-loaded polymeric material contained enriched boron or natural boron. A prototype analyzer was constructed, and initial planning was done for an actual analysis facility
Stakeholder Analysis Worksheet
Stakeholder Analysis WorksheetA worksheet that can be used to document potential stakeholder groups, the information or expertise they hold, the role that they can play, their interests or concerns about the HIA
Energy Technology Data Exchange (ETDEWEB)
Arent, D.; Benioff, R.; Mosey, G.; Bird, L.; Brown, J.; Brown, E.; Vimmerstedt, L.; Aabakken, J.; Parks, K.; Lapsa, M.; Davis, S.; Olszewski, M.; Cox, D.; McElhaney, K.; Hadley, S.; Hostick, D.; Nicholls, A.; McDonald, S.; Holloman, B.
2006-10-01
This paper presents the results of energy market analysis sponsored by the Department of Energy's (DOE) Weatherization and International Program (WIP) within the Office of Energy Efficiency and Renewable Energy (EERE). The analysis was conducted by a team of DOE laboratory experts from the National Renewable Energy Laboratory (NREL), Oak Ridge National Laboratory (ORNL), and Pacific Northwest National Laboratory (PNNL), with additional input from Lawrence Berkeley National Laboratory (LBNL). The analysis was structured to identify those markets and niches where government can create the biggest impact by informing management decisions in the private and public sectors. The analysis identifies those markets and niches where opportunities exist for increasing energy efficiency and renewable energy use.
Principles of Fourier analysis
Howell, Kenneth B
2001-01-01
Fourier analysis is one of the most useful and widely employed sets of tools for the engineer, the scientist, and the applied mathematician. As such, students and practitioners in these disciplines need a practical and mathematically solid introduction to its principles. They need straightforward verifications of its results and formulas, and they need clear indications of the limitations of those results and formulas.Principles of Fourier Analysis furnishes all this and more. It provides a comprehensive overview of the mathematical theory of Fourier analysis, including the development of Fourier series, "classical" Fourier transforms, generalized Fourier transforms and analysis, and the discrete theory. Much of the author''s development is strikingly different from typical presentations. His approach to defining the classical Fourier transform results in a much cleaner, more coherent theory that leads naturally to a starting point for the generalized theory. He also introduces a new generalized theory based ...
Quantitative investment analysis
DeFusco, Richard
2007-01-01
In the "Second Edition" of "Quantitative Investment Analysis," financial experts Richard DeFusco, Dennis McLeavey, Jerald Pinto, and David Runkle outline the tools and techniques needed to understand and apply quantitative methods to today's investment process.
Energy Technology Data Exchange (ETDEWEB)
Wood, William Monford [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2015-02-23
Presenting a systematic study of the standard analysis of rod-pinch radiographs for obtaining quantitative measurements of areal mass densities, and making suggestions for improving the methodology of obtaining quantitative information from radiographed objects.
Biodiesel Emissions Analysis Program
Using existing data, the EPA's biodiesel emissions analysis program sought to quantify the air pollution emission effects of biodiesel for diesel engines that have not been specifically modified to operate on biodiesel.
Introduction to global analysis
Kahn, Donald W
2007-01-01
This text introduces the methods of mathematical analysis as applied to manifolds, including the roles of differentiation and integration, infinite dimensions, Morse theory, Lie groups, and dynamical systems. 1980 edition.
Biorefinery Sustainability Analysis
DEFF Research Database (Denmark)
J. S. M. Silva, Carla; Prunescu, Remus Mihail; Gernaey, Krist
2017-01-01
This chapter deals with sustainability analysis of biorefinery systems in terms of environmental and socio-economic indicators . Life cycle analysis has methodological issues related to the functional unit (FU), allocation , land use and biogenic carbon neutrality of the reference system and of t......This chapter deals with sustainability analysis of biorefinery systems in terms of environmental and socio-economic indicators . Life cycle analysis has methodological issues related to the functional unit (FU), allocation , land use and biogenic carbon neutrality of the reference system...... and of the biorefinery-based system. Socio-economic criteria and indicators used in sustainability frameworks assessment are presented and discussed. There is not one single methodology that can aptly cover the synergies of environmental, economic, social and governance issues required to assess the sustainable...
Pesticide Instrumental Analysis
International Nuclear Information System (INIS)
Samir, E.; Fonseca, E.; Baldyga, N.; Acosta, A.; Gonzalez, F.; Felicita, F.; Tomasso, M.; Esquivel, D.; Parada, A.; Enriquez, P.; Amilibia, M.
2012-01-01
This workshop was the evaluation of the pesticides impact on the vegetable matrix with the purpose to determine the analysis by GC / M S. The working material were lettuce matrix, chard and a mix of green leaves and pesticides.
Kansas Data Access and Support Center — The Kansas GAP Analysis Land Cover database depicts 43 land cover classes for the state of Kansas. The database was generated using a two-stage hybrid classification...
Perspectives in shape analysis
Bruckstein, Alfred; Maragos, Petros; Wuhrer, Stefanie
2016-01-01
This book presents recent advances in the field of shape analysis. Written by experts in the fields of continuous-scale shape analysis, discrete shape analysis and sparsity, and numerical computing who hail from different communities, it provides a unique view of the topic from a broad range of perspectives. Over the last decade, it has become increasingly affordable to digitize shape information at high resolution. Yet analyzing and processing this data remains challenging because of the large amount of data involved, and because modern applications such as human-computer interaction require real-time processing. Meeting these challenges requires interdisciplinary approaches that combine concepts from a variety of research areas, including numerical computing, differential geometry, deformable shape modeling, sparse data representation, and machine learning. On the algorithmic side, many shape analysis tasks are modeled using partial differential equations, which can be solved using tools from the field of n...
National Research Council Canada - National Science Library
1998-01-01
.... Establishing proper job procedures is one of the benefits of conducting a job hazard analysis carefully studying and recording each step of a job, identifying existing or potential job hazards...
Main: Nucleotide Analysis [KOME
Lifescience Database Archive (English)
Full Text Available Nucleotide Analysis Japonica genome blast search result Result of blastn search against jap...onica genome sequence kome_japonica_genome_blast_search_result.zip kome_japonica_genome_blast_search_result ...
Schuller, Björn W
2013-01-01
This book provides the reader with the knowledge necessary for comprehension of the field of Intelligent Audio Analysis. It firstly introduces standard methods and discusses the typical Intelligent Audio Analysis chain going from audio data to audio features to audio recognition. Further, an introduction to audio source separation, and enhancement and robustness are given. After the introductory parts, the book shows several applications for the three types of audio: speech, music, and general sound. Each task is shortly introduced, followed by a description of the specific data and methods applied, experiments and results, and a conclusion for this specific task. The books provides benchmark results and standardized test-beds for a broader range of audio analysis tasks. The main focus thereby lies on the parallel advancement of realism in audio analysis, as too often today’s results are overly optimistic owing to idealized testing conditions, and it serves to stimulate synergies arising from transfer of ...
Scientific stream pollution analysis
National Research Council Canada - National Science Library
Nemerow, Nelson Leonard
1974-01-01
A comprehensive description of the analysis of water pollution that presents a careful balance of the biological,hydrological, chemical and mathematical concepts involved in the evaluation of stream...
International Nuclear Information System (INIS)
Andreeva, J; Maier, G; Spiga, D; Calloni, M; Colling, D; Fanzago, F; D'Hondt, J; Maes, J; Van Mulders, P; Villella, I; Klem, J; Letts, J; Padhi, S; Sarkar, S
2010-01-01
During normal data taking CMS expects to support potentially as many as 2000 analysis users. Since the beginning of 2008 there have been more than 800 individuals who submitted a remote analysis job to the CMS computing infrastructure. The bulk of these users will be supported at the over 40 CMS Tier-2 centres. Supporting a globally distributed community of users on a globally distributed set of computing clusters is a task that requires reconsidering the normal methods of user support for Analysis Operations. In 2008 CMS formed an Analysis Support Task Force in preparation for large-scale physics analysis activities. The charge of the task force was to evaluate the available support tools, the user support techniques, and the direct feedback of users with the goal of improving the success rate and user experience when utilizing the distributed computing environment. The task force determined the tools needed to assess and reduce the number of non-zero exit code applications submitted through the grid interfaces and worked with the CMS experiment dashboard developers to obtain the necessary information to quickly and proactively identify issues with user jobs and data sets hosted at various sites. Results of the analysis group surveys were compiled. Reference platforms for testing and debugging problems were established in various geographic regions. The task force also assessed the resources needed to make the transition to a permanent Analysis Operations task. In this presentation the results of the task force will be discussed as well as the CMS Analysis Operations plans for the start of data taking.
Invitation to classical analysis
Duren, Peter
2012-01-01
This book gives a rigorous treatment of selected topics in classical analysis, with many applications and examples. The exposition is at the undergraduate level, building on basic principles of advanced calculus without appeal to more sophisticated techniques of complex analysis and Lebesgue integration. Among the topics covered are Fourier series and integrals, approximation theory, Stirling's formula, the gamma function, Bernoulli numbers and polynomials, the Riemann zeta function, Tauberian theorems, elliptic integrals, ramifications of the Cantor set, and a theoretical discussion of differ
Analysis of irradiated materials
International Nuclear Information System (INIS)
Bellamy, B.A.
1988-01-01
Papers presented at the UKAEA Conference on Materials Analysis by Physical Techniques (1987) covered a wide range of techniques as applied to the analysis of irradiated materials. These varied from reactor component materials, materials associated with the Authority's radwaste disposal programme, fission products and products associated with the decommissioning of nuclear reactors. An invited paper giving a very comprehensive review of Laser Ablation Microprobe Mass Spectroscopy (LAMMS) was included in the programme. (author)
Oktavianto, Digit
2013-01-01
This book is a step-by-step, practical tutorial for analyzing and detecting malware and performing digital investigations. This book features clear and concise guidance in an easily accessible format.Cuckoo Malware Analysis is great for anyone who wants to analyze malware through programming, networking, disassembling, forensics, and virtualization. Whether you are new to malware analysis or have some experience, this book will help you get started with Cuckoo Sandbox so you can start analysing malware effectively and efficiently.
International Nuclear Information System (INIS)
Niehaus, F.
1988-01-01
In this paper, the risks of various energy systems are discussed considering severe accidents analysis, particularly the probabilistic safety analysis, and probabilistic safety criteria, and the applications of these criteria and analysis. The comparative risk analysis has demonstrated that the largest source of risk in every society is from daily small accidents. Nevertheless, we have to be more concerned about severe accidents. The comparative risk analysis of five different energy systems (coal, oil, gas, LWR and STEC (Solar)) for the public has shown that the main sources of risks are coal and oil. The latest comparative risk study of various energy has been conducted in the USA and has revealed that the number of victims from coal is 42 as many than victims from nuclear. A study for severe accidents from hydro-dams in United States has estimated the probability of dam failures at 1 in 10,000 years and the number of victims between 11,000 and 260,000. The average occupational risk from coal is one fatal accident in 1,000 workers/year. The probabilistic safety analysis is a method that can be used to assess nuclear energy risks, and to analyze the severe accidents, and to model all possible accident sequences and consequences. The 'Fault tree' analysis is used to know the probability of failure of the different systems at each point of accident sequences and to calculate the probability of risks. After calculating the probability of failure, the criteria for judging the numerical results have to be developed, that is the quantitative and qualitative goals. To achieve these goals, several systems have been devised by various countries members of AIEA. The probabilistic safety ana-lysis method has been developed by establishing a computer program permit-ting to know different categories of safety related information. 19 tabs. (author)
International Nuclear Information System (INIS)
Arien, B.
1998-01-01
The objective of SCK-CEN's programme on reactor safety is to develop expertise in probabilistic and deterministic reactor safety analysis. The research programme consists of four main activities, in particular the development of software for reliability analysis of large systems and participation in the international PHEBUS-FP programme for severe accidents, the development of an expert system for the aid to diagnosis; the development and application of a probabilistic reactor dynamics method. Main achievements in 1999 are reported
Brady, Sir Michael; Highnam, Ralph; Irving, Benjamin; Schnabel, Julia A
2016-10-01
Cancer is one of the world's major healthcare challenges and, as such, an important application of medical image analysis. After a brief introduction to cancer, we summarise some of the major developments in oncological image analysis over the past 20 years, but concentrating those in the authors' laboratories, and then outline opportunities and challenges for the next decade. Copyright © 2016 Elsevier B.V. All rights reserved.
International Nuclear Information System (INIS)
Tibari, Elghali; Taous, Fouad; Marah, Hamid
2014-01-01
This report presents results related to stable isotopes analysis carried out at the CNESTEN DASTE in Rabat (Morocco), on behalf of Senegal. These analyzes cover 127 samples. These results demonstrate that Oxygen-18 and Deuterium in water analysis were performed by infrared Laser spectroscopy using a LGR / DLT-100 with Autosampler. Also, the results are expressed in δ values (‰) relative to V-SMOW to ± 0.3 ‰ for oxygen-18 and ± 1 ‰ for deuterium.
Oden, J Tinsley
2010-01-01
The textbook is designed to drive a crash course for beginning graduate students majoring in something besides mathematics, introducing mathematical foundations that lead to classical results in functional analysis. More specifically, Oden and Demkowicz want to prepare students to learn the variational theory of partial differential equations, distributions, and Sobolev spaces and numerical analysis with an emphasis on finite element methods. The 1996 first edition has been used in a rather intensive two-semester course. -Book News, June 2010
Griffel, DH
2002-01-01
A stimulating introductory text, this volume examines many important applications of functional analysis to mechanics, fluid mechanics, diffusive growth, and approximation. Detailed enough to impart a thorough understanding, the text is also sufficiently straightforward for those unfamiliar with abstract analysis. Its four-part treatment begins with distribution theory and discussions of Green's functions. Essentially independent of the preceding material, the second and third parts deal with Banach spaces, Hilbert space, spectral theory, and variational techniques. The final part outlines the
Ivan Stosic; Drasko Nikolic; Aleksandar Zdravkovic
2012-01-01
The main purpose of this paper is to examine the impact of the current Serbian macro-environment on the businesses through the implementation of PEST analysis as a framework for assessing general or macro environment in which companies are operating. The authors argue the elements in presented PEST analysis indicate that the current macro-environment is characterized by the dominance of threats and weaknesses with few opportunities and strengths. Consequently, there is a strong need for faste...
Forensic neutron activation analysis
International Nuclear Information System (INIS)
Kishi, T.
1987-01-01
The progress of forensic neutron activation analysis (FNAA) in Japan is described. FNAA began in 1965 and during the past 20 years many cases have been handled; these include determination of toxic materials, comparison examination of physical evidences (e.g., paints, metal fragments, plastics and inks) and drug sample differentiation. Neutron activation analysis is applied routinely to the scientific criminal investigation as one of multielement analytical techniques. This paper also discusses these routine works. (author) 14 refs
Probabilistic Structural Analysis Program
Pai, Shantaram S.; Chamis, Christos C.; Murthy, Pappu L. N.; Stefko, George L.; Riha, David S.; Thacker, Ben H.; Nagpal, Vinod K.; Mital, Subodh K.
2010-01-01
NASA/NESSUS 6.2c is a general-purpose, probabilistic analysis program that computes probability of failure and probabilistic sensitivity measures of engineered systems. Because NASA/NESSUS uses highly computationally efficient and accurate analysis techniques, probabilistic solutions can be obtained even for extremely large and complex models. Once the probabilistic response is quantified, the results can be used to support risk-informed decisions regarding reliability for safety-critical and one-of-a-kind systems, as well as for maintaining a level of quality while reducing manufacturing costs for larger-quantity products. NASA/NESSUS has been successfully applied to a diverse range of problems in aerospace, gas turbine engines, biomechanics, pipelines, defense, weaponry, and infrastructure. This program combines state-of-the-art probabilistic algorithms with general-purpose structural analysis and lifting methods to compute the probabilistic response and reliability of engineered structures. Uncertainties in load, material properties, geometry, boundary conditions, and initial conditions can be simulated. The structural analysis methods include non-linear finite-element methods, heat-transfer analysis, polymer/ceramic matrix composite analysis, monolithic (conventional metallic) materials life-prediction methodologies, boundary element methods, and user-written subroutines. Several probabilistic algorithms are available such as the advanced mean value method and the adaptive importance sampling method. NASA/NESSUS 6.2c is structured in a modular format with 15 elements.
Directory of Open Access Journals (Sweden)
Iulian N. BUJOREANU
2011-01-01
Full Text Available Sensitivity analysis represents such a well known and deeply analyzed subject that anyone to enter the field feels like not being able to add anything new. Still, there are so many facets to be taken into consideration.The paper introduces the reader to the various ways sensitivity analysis is implemented and the reasons for which it has to be implemented in most analyses in the decision making processes. Risk analysis is of outmost importance in dealing with resource allocation and is presented at the beginning of the paper as the initial cause to implement sensitivity analysis. Different views and approaches are added during the discussion about sensitivity analysis so that the reader develops an as thoroughly as possible opinion on the use and UTILITY of the sensitivity analysis. Finally, a round-up conclusion brings us to the question of the possibility of generating the future and analyzing it before it unfolds so that, when it happens it brings less uncertainty.
International Nuclear Information System (INIS)
Grimanis, A.P.
1985-01-01
A review of research and development on NAA as well as examples of applications of this method are presented, taken from work carried out over the last 21 years at the Radioanalytical Laboratory of the Department of Chemistry in the Greek Nuclear Research Center ''Demokritos''. Improved and faster radiochemical NAA methods have been developed for the determination of Au, Ni, Cl, As, Cu, U, Cr, Eu, Hg and Mo in several materials, for the simultaneous determination of Br and I; Mg, Sr and Ni; As and Cu; As, Sb and Hg; Mn, Sr and Ba; Cd and Zn; Se and As; Mo and Cr in biological materials. Instrumental NAA methods have also been developed for the determination of Ag, Cl and Na in lake waters, Al, Ca, Mg and V in wines, 7 trace elements in biological materials, 17 trace elements in sediments and 20 minor and trace elements in ceramics. A comprehensive computer program for routine activation analysis using Ge(Li) detectors have been worked out. A rather extended charged-particle activation analysis program is carried out for the last 10 years, including particle induced X-ray emission (PIXE) analysis, particle induced prompt gamma-ray emission analysis (PIGE), other nuclear reactions and proton activation analysis. A special neutron activation method, the delayed fission neutron counting method is used for the analysis of fissionable elements, as U, Th, Pu, in samples of the whole nuclear fuel cycle including geological, enriched and nuclear safeguards samples
Integrated genetic analysis microsystems
International Nuclear Information System (INIS)
Lagally, Eric T; Mathies, Richard A
2004-01-01
With the completion of the Human Genome Project and the ongoing DNA sequencing of the genomes of other animals, bacteria, plants and others, a wealth of new information about the genetic composition of organisms has become available. However, as the demand for sequence information grows, so does the workload required both to generate this sequence and to use it for targeted genetic analysis. Microfabricated genetic analysis systems are well poised to assist in the collection and use of these data through increased analysis speed, lower analysis cost and higher parallelism leading to increased assay throughput. In addition, such integrated microsystems may point the way to targeted genetic experiments on single cells and in other areas that are otherwise very difficult. Concomitant with these advantages, such systems, when fully integrated, should be capable of forming portable systems for high-speed in situ analyses, enabling a new standard in disciplines such as clinical chemistry, forensics, biowarfare detection and epidemiology. This review will discuss the various technologies available for genetic analysis on the microscale, and efforts to integrate them to form fully functional robust analysis devices. (topical review)
INTERNAL ENVIRONMENT ANALYSIS TECHNIQUES
Directory of Open Access Journals (Sweden)
Caescu Stefan Claudiu
2011-12-01
Full Text Available Theme The situation analysis, as a separate component of the strategic planning, involves collecting and analysing relevant types of information on the components of the marketing environment and their evolution on the one hand and also on the organization’s resources and capabilities on the other. Objectives of the Research The main purpose of the study of the analysis techniques of the internal environment is to provide insight on those aspects that are of strategic importance to the organization. Literature Review The marketing environment consists of two distinct components, the internal environment that is made from specific variables within the organization and the external environment that is made from variables external to the organization. Although analysing the external environment is essential for corporate success, it is not enough unless it is backed by a detailed analysis of the internal environment of the organization. The internal environment includes all elements that are endogenous to the organization, which are influenced to a great extent and totally controlled by it. The study of the internal environment must answer all resource related questions, solve all resource management issues and represents the first step in drawing up the marketing strategy. Research Methodology The present paper accomplished a documentary study of the main techniques used for the analysis of the internal environment. Results The special literature emphasizes that the differences in performance from one organization to another is primarily dependant not on the differences between the fields of activity, but especially on the differences between the resources and capabilities and the ways these are capitalized on. The main methods of analysing the internal environment addressed in this paper are: the analysis of the organizational resources, the performance analysis, the value chain analysis and the functional analysis. Implications Basically such
Professionalizing Intelligence Analysis
Directory of Open Access Journals (Sweden)
James B. Bruce
2015-09-01
Full Text Available This article examines the current state of professionalism in national security intelligence analysis in the U.S. Government. Since the introduction of major intelligence reforms directed by the Intelligence Reform and Terrorism Prevention Act (IRTPA in December, 2004, we have seen notable strides in many aspects of intelligence professionalization, including in analysis. But progress is halting, uneven, and by no means permanent. To consolidate its gains, and if it is to continue improving, the U.S. intelligence community (IC should commit itself to accomplishing a new program of further professionalization of analysis to ensure that it will develop an analytic cadre that is fully prepared to deal with the complexities of an emerging multipolar and highly dynamic world that the IC itself is forecasting. Some recent reforms in intelligence analysis can be assessed against established standards of more fully developed professions; these may well fall short of moving the IC closer to the more fully professionalized analytical capability required for producing the kind of analysis needed now by the United States.
Harmonic and geometric analysis
Citti, Giovanna; Pérez, Carlos; Sarti, Alessandro; Zhong, Xiao
2015-01-01
This book presents an expanded version of four series of lectures delivered by the authors at the CRM. Harmonic analysis, understood in a broad sense, has a very wide interplay with partial differential equations and in particular with the theory of quasiconformal mappings and its applications. Some areas in which real analysis has been extremely influential are PDE's and geometric analysis. Their foundations and subsequent developments made extensive use of the Calderón–Zygmund theory, especially the Lp inequalities for Calderón–Zygmund operators (Beurling transform and Riesz transform, among others) and the theory of Muckenhoupt weights. The first chapter is an application of harmonic analysis and the Heisenberg group to understanding human vision, while the second and third chapters cover some of the main topics on linear and multilinear harmonic analysis. The last serves as a comprehensive introduction to a deep result from De Giorgi, Moser and Nash on the regularity of elliptic partial differen...
Hansson, Sven Ove; Aven, Terje
2014-07-01
This article discusses to what extent risk analysis is scientific in view of a set of commonly used definitions and criteria. We consider scientific knowledge to be characterized by its subject matter, its success in developing the best available knowledge in its fields of study, and the epistemic norms and values that guide scientific investigations. We proceed to assess the field of risk analysis according to these criteria. For this purpose, we use a model for risk analysis in which science is used as a base for decision making on risks, which covers the five elements evidence, knowledge base, broad risk evaluation, managerial review and judgment, and the decision; and that relates these elements to the domains experts and decisionmakers, and to the domains fact-based or value-based. We conclude that risk analysis is a scientific field of study, when understood as consisting primarily of (i) knowledge about risk-related phenomena, processes, events, etc., and (ii) concepts, theories, frameworks, approaches, principles, methods and models to understand, assess, characterize, communicate, and manage risk, in general and for specific applications (the instrumental part). © 2014 Society for Risk Analysis.