Yoganathan, Ajit P; He, Zhaoming; Casey Jones, S
Valvular heart disease is a life-threatening disease that afflicts millions of people worldwide and leads to approximately 250,000 valve repairs and/or replacements each year. Malfunction of a native valve impairs its efficient fluid mechanic/hemodynamic performance. Artificial heart valves have been used since 1960 to replace diseased native valves and have saved millions of lives. Unfortunately, despite four decades of use, these devices are less than ideal and lead to many complications. Many of these complications/problems are directly related to the fluid mechanics associated with the various mechanical and bioprosthetic valve designs. This review focuses on the state-of-the-art experimental and computational fluid mechanics of native and prosthetic heart valves in current clinical use. The fluid dynamic performance characteristics of caged-ball, tilting-disc, bileaflet mechanical valves and porcine and pericardial stented and nonstented bioprostheic valves are reviewed. Other issues related to heart valve performance, such as biomaterials, solid mechanics, tissue mechanics, and durability, are not addressed in this review.
... replacement; Valve repair; Heart valve prosthesis; Mechanical valves; Prosthetic valves ... surgery. Your heart valve has been damaged by infection ( endocarditis ). You have received a new heart valve ...
Hafizah Mokhtar, N.; Abas, Aizat
In this work, an artificial heart valve is designed for use in real heart with further consideration on the effect of thrombosis, vorticity, and stress. The design of artificial heart valve model is constructed by Computer-aided design (CAD) modelling and simulated using Computational fluid dynamic (CFD) software. The effect of blood flow pattern, velocity and vorticity of the artificial heart valve design has been analysed in this research work. Based on the results, the artificial heart valve design shows that it has a Doppler velocity index that is less than the allowable standards for the left heart with values of more than 0.30 and less than 2.2. These values are safe to be used as replacement of the human heart valve.
Steiner, R.M.; Flicker, S.
The development of prosthetic heart valves in the late 1950s ushered in a new era in the treatment of heart disease. The radiologist has an important role to play preoperatively in the diagnosis of valvular heart disease. Radiology is valuable in identification of the implanted prosthetic valve and recognition of complications associated with valve implantation. Radiologists must be familiar with the imaging techniques best suited to evaluate the function of the valve prosthesis in question. In this chapter the authors discuss the radiographic approach to the evaluation of the status of patients for valve replacement and the imaging problems peculiar to the types of valves in current use. The relative value of plain-film radiography, fluoroscopy, videorecording and cinerecording, and aortography is addressed, as well as the potential value of magnetic resonance imaging and subsecond dynamic computed tomography
Your heart has four valves. Normally, these valves open to let blood flow through or out of your heart, and then shut to keep it from flowing ... close tightly. It's one of the most common heart valve conditions. Sometimes it causes regurgitation. Stenosis - when ...
Rogers, Stuart E.; Kwak, Dochan; Kiris, Cetin; Chang, I-Dee
NASA technical memorandum discusses computations of flow of blood through artificial heart and through tilting-disk artificial heart valve. Represents further progress in research described in "Numerical Simulation of Flow Through an Artificial Heart" (ARC-12478). One purpose of research to exploit advanced techniques of computational fluid dynamics and capabilities of supercomputers to gain understanding of complicated internal flows of viscous, essentially incompressible fluids like blood. Another to use understanding to design better artificial hearts and valves.
Govindarajan, V; Udaykumar, H S; Herbertson, L H; Deutsch, S; Manning, K B; Chandran, K B
The fluid dynamics during valve closure resulting in high shear flows and large residence times of particles has been implicated in platelet activation and thrombus formation in mechanical heart valves. Our previous studies with bi-leaflet valves have shown that large shear stresses induced in the gap between the leaflet edge and the valve housing results in relatively high platelet activation levels whereas flow between the leaflets results in shed vortices not conducive to platelet damage. In this study we compare the result of closing dynamics of a tilting disc valve with that of a bi-leaflet valve. The two-dimensional fluid-structure interaction analysis of a tilting disc valve closure mechanics is performed with a fixed grid Cartesian mesh flow solver with local mesh refinement, and a Lagrangian particle dynamic analysis for computation of potential for platelet activation. Throughout the simulation the flow remains in the laminar regime and the flow through the gap width is marked by the development of a shear layer which separates from the leaflet downstream of the valve. Zones of re-circulation are observed in the gap between the leaflet edge and the valve housing on the major orifice region of the tilting disc valve and are seen to be migrating towards the minor orifice region. Jet flow is observed at the minor orifice region and a vortex is formed which sheds in the direction of fluid motion as observed in experiments using PIV measurements. The activation parameter computed for the tilting disc valve, at the time of closure was found to be 2.7 times greater than that of the bi-leaflet mechanical valve and was found to be in the vicinity of the minor orifice region mainly due to the migration of vortical structures from the major to the minor orifice region during the leaflet rebound of the closing phase.
Artiﬁcial heart valves are engineered devices used for replacing diseased or damaged natural valves of the heart. Most commonly used for replacement are mechanical heart valves and biological valves. This paper brieﬂy outlines the evolution, designs employed, materials being used,. and important factors that affect the ...
Full Text Available The mechanical components of the heart, especially the valves and leaflets, are enormous stressed during lifetime. Therefore, those structures undergo different pathophysiological tissue transformations which affect cardiac output and in consequence living comfort of affected patients. These changes may lead to calcific aortic valve stenosis (AVS, the major heart valve disease in humans. The knowledge about changes of the dynamic behaviour during the course of this disease and the possibility of early stage diagnosis is of particular interest and could lead to the development of new treatment strategies and drug based options of prevention or therapy. 4D optical coherence tomography (OCT in combination with high-speed video microscopy were applied to characterize dynamic behaviour of the murine aortic valve and to characterize dynamic properties during artificial stimulation. We present a promising tool to investigate the aortic valve dynamics in an ex vivo disease model with a high spatial and temporal resolution using a multimodal imaging setup.
... Guide: Understanding Your Heart Valve Problem | Spanish Symptom Tracker | Spanish Pre-surgery Checklist | Spanish What Is Heart ... Heart Valves • Heart Valve Problems and Causes • Risks, Signs and Symptoms • Accurate Diagnosis • Treatment Options • Recovery and ...
Hedayat, Mohammadali; Asgharzadeh, Hafez; Borazjani, Iman
Since the advent of heart valve, several valve types such as mechanical and bio-prosthetic valves have been designed. Mechanical Heart Valves (MHV) are durable but suffer from thromboembolic complications that caused by shear-induced platelet activation near the valve region. Bio-prosthetic Heart Valves (BHV) are known for better hemodynamics. However, they usually have a short average life time. Realistic simulations of heart valves in combination with platelet activation models can lead to a better understanding of the potential risk of thrombus formation in such devices. In this study, an Eulerian approach is developed to calculate the platelet activation in three-dimensional simulations of flow through MHV and BHV using a parallel overset-curvilinear immersed boundary technique. A curvilinear body-fitted grid is used for the flow simulation through the anatomic aorta, while the sharp-interface immersed boundary method is used for simulation of the Left Ventricle (LV) with prescribed motion. In addition, dynamics of valves were calculated numerically using under-relaxed strong-coupling algorithm. Finally, the platelet activation results for BMV and MHV are compared with each other.
Heim, Frederic; Durand, Bernard; Chakfe, Nabil
The rapid developments and success in percutaneous vascular surgery over the last two decades with the now common stent grafts implantation, make the noninvasive surgery technique today attractive even for heart valve replacement. Less traumatic for the patient and also less time consuming, percutaneous heart valve replacement is however at its beginning and restricted to end of life patients. The noninvasive procedure expects from the heart valve prosthesis material to be resistant and adapted to folding requirements of the implantation process (catheter). Polyester fabric could be a suited material for heart valve implanted percutaneously. Highly flexible and resistant, polyester fabric proved to be well adapted to the dynamic behavior of a valve and polyester (Dacron) is also widely used for vascular grafts implantation and shows good biocompatibility and durability. However, today there's no data available on long-term durability of fabric used as heart valve material. The purpose of this work is to study the long term behavior of a microdenier polyester fabric construction under combined in vitro flexure and tension fatigue stress. In the novel in vitro testing technique presented, a fabric specimen was subjected to combined flexural and tensile fatigue generated by fluid flow under physiological pressure conditions. The results obtained show how flexural properties change with fatigue time, which reflects directly on the suitability of a fabric in such devices. It was also observed that these fabric structural changes directly influence the in vitro behavior of the textile heart valve prosthesis. (c) 2009 Wiley Periodicals, Inc.
Kwon, Young Joo
This paper addresses the basic concept of MDO methodology and the structural analysis that should be performed in the design process of a mechanical heart valve prosthesis with flat leaflet using MDO methodology. In the structural design of the mechanical heart valve (MHV) prosthesis, the fluid mechanics analysis is executed for the blood flow passing through the leaflets of a mechanical heart valve prosthesis. Thereafter, the rigid body dynamics analysis of the leaflet motion is performed to obtain the structural condition for the structural mechanics analysis of the deformed leaflet. Then the structural mechanics analysis of the deformed leaflet follows to confirm the minimum thickness of the leaflet for the structural durability of the mechanical heart valve prosthesis. This paper shows that the minimum leaflet thickness can be evaluated to be 0.6mm among the suggested thicknesses.
... should be completely closed For example: Watch an animation of mitral valve regurgitation A leaking mitral valve ... Not Alone Popular Articles 1 Understanding Blood Pressure Readings 2 Sodium and Salt 3 Heart Attack Symptoms ...
... ACC guideline for the management of patients with valvular heart disease: executive summary: a report of the American College ... Editorial team. Related MedlinePlus Health Topics Heart Surgery Heart Valve Diseases Browse the Encyclopedia A.D.A.M., Inc. ...
Full Text Available Objective To summarize the experience with resurgery for recurrent valvular heart diseases. Methods From June 2004 to June 2015, 28 patients (15 males and 13 females with ages ranging from 44 to 67 years (55.6±6.5 years with recurrent heart valve disease underwent resurgery. The reasons for resurgery included perivalvular leakage (7 cases, bioprosthetic valve decline (6 cases in mitral valve and 3 in tricuspid valve, mechanical prostheses dysfunction (2cases, infective endocarditis after valve replacement (2 cases, restenosis of repaired native valve (1 case, and severe tricuspid insufficiency after left-side valve surgery (7 cases. Resurgery included mitral valve replacement in 18 patients and tricuspid valve replacement in 10. All the patients underwent third or fourth or even fifth cardiac surgery for valve replacement. Results There were 2 hospital deaths with a mortality of 7.1% (2/28. The main causes of early-stage deaths were low cardiac output syndrome. The main postoperative complications were respiratory failure in 3, low cardiac output syndrome in 2, reexploration for bleeding in 2 and serious infectious shock in 1. All the patients were found with the great improvement in heart function and the re-implanted prostheses worked well during follow-up. Conclusions Although resurgery for recurrent heart valve disease poses a continuing challenge to cardiac surgeon, it could be performed with the satisfactory results. The keys to a successful cardiac resurgery include appropriate operational timing, refined surgical technique and reasonable perioperative managements. DOI: 10.11855/j.issn.0577-7402.2017.01.11
Abad, Cipriano; Urso, Stefano; Gomez, Elsa; De la Vega, Maria
A 68-year-old woman with a history of previous double-valve replacement with On-X mechanical heart valves presented with clinical, echocardiographic and cardiac catheterization signs of obstruction of the On-X tricuspid heart valve prosthesis. The patient was successfully reoperated, but at surgery the valve was seen to be invaded by an abnormal overgrowth of pannus that blocked one of the leaflets. A small amount of non-obstructive fresh thrombus was also observed. The valve was successfully replaced with a biological heart valve prosthesis. The patient was discharged home, and is doing well four months after the operation, when echocardiography demonstrated normal function in the tricuspid valve. The present case represents the first ever report of pannus formation and subsequent dysfunction in an On-X heart valve, and also the first case of tricuspid valve malfunction and obstruction using this type of heart valve substitute.
Hafizah Mokhtar, N.; Abas, Aizat
In this work, two different designs of artificial heart valve were devised and then compared by considering the thrombosis, wear and valve orifice to anatomical orifice ratio of each mechanical heart valve. These different design configurations of bi-leaflet mechanical heart valves model are created through the use of Computer-aided design (CAD) modelling and simulated using Computational fluid dynamic (CFD) software. Design 1 is based on existing conventional bi-leaflet valve and design 2 based on modified bi-leaflet respectively. The flow pattern, velocity, vorticity and stress analysis have been done to justify the best design. Based on results, both of the designs show a Doppler velocity index of less than the allowable standard of 2.2 which is safe to be used as replacement of the human heart valve. However, design 2 shows that it has a lower possibility of cavitation issue which will lead to lower thrombosis and provide good central flow area of blood as compared to design 1.
Khan, Nasir A; Butany, Jagdish; Leong, Shaun W; Rao, Vivek; Cusimano, Robert J; Ross, Heather J
Prosthetic heart valve dysfunction due to thrombus or pannus formation can be a life-threatening complication. The present report describes a 47-year-old woman who developed valvular cardiomyopathy after chorda-sparing mitral valve replacement, and subsequently underwent heart transplantation for progressive heart failure. The explanted mitral valve prosthesis showed significant thrombus and pannus leading to reduced leaflet mobility and valvular stenosis. The present report illustrates the role of the subvalvular apparatus and pannus in prosthesis dysfunction. PMID:19279993
Rogers, Stewart E.; Kwak, Dochan; Kiris, Cetin; Chang, I-Dee
Report discusses computations of blood flow through prosthetic tilting disk valve. Computational procedure developed in simulation used to design better artificial hearts and valves by reducing or eliminating following adverse flow characteristics: large pressure losses, which prevent hearts from working efficiently; separated and secondary flows, which causes clotting; and high turbulent shear stresses, which damages red blood cells. Report reiterates and expands upon part of NASA technical memorandum "Computed Flow Through an Artificial Heart and Valve" (ARC-12983). Also based partly on research described in "Numerical Simulation of Flow Through an Artificial Heart" (ARC-12478).
Clark, G. [Lawrence Livermore National Lab., CA (United States)
Prosthetic heart valves and the many great strides in valve design have been responsible for extending the life spans of many people with serious heart conditions. Even though the prosthetic valves are extremely reliable, they are eventually susceptible to long-term fatigue and structural failure effects expected from mechanical devices operating over long periods of time. The purpose of our work is to classify the condition of in vivo Bjork-Shiley Convexo-Concave (BSCC) heart valves by processing acoustic measurements of heart valve sounds. The structural failures of interest for Bscc valves is called single leg separation (SLS). SLS can occur if the outlet strut cracks and separates from the main structure of the valve. We measure acoustic opening and closing sounds (waveforms) using high sensitivity contact microphones on the patient`s thorax. For our analysis, we focus our processing and classification efforts on the opening sounds because they yield direct information about outlet strut condition with minimal distortion caused by energy radiated from the valve disc.
... Guide: Understanding Your Heart Valve Problem | Spanish Symptom Tracker | Spanish Pre-surgery Checklist | Spanish What Is Heart ... Cardiac Arrest: How Are They Different? 7 Warning Signs of a Heart Attack 8 Low Blood Pressure - ...
Ledesma, R.; Zenit, R.; Pulos, G.; Sanchez, E.; Juarez, A.
Blood damage commonly appears in medical valve prothesis. It is a mayor concern for the designers and surgeons. It is well known that this damage and other complications result from the modified fluid dynamics through the replacement valve. To evaluate the performance of prosthetic heart valves, it is necessary to study the flow through them. To conduct this study , we have built a flow channel that emulates cardiac conditions and allows optical access such that a 3D-PIV velocimetry system could be used. The experiments are aimed to reconstruct the downstream structure of the flow through a mechanical and a bio-material tricuspid heart valve prothesis. Preliminary results show that the observed coherent structures can be related with haemolysis and trombosis, illnesses commonly found in valve prothesis recipients. The mean flow, the levels of strain rate and the turbulence intensity generated by the valves can also be directly related to blood damage. In general, bio-material made valves tend to reduce these complications.
Sodian, R; Hoerstrup, S P; Sperling, J S; Daebritz, S H; Martin, D P; Schoen, F J; Vacanti, J P; Mayer, J E
Tissue engineering is a new approach, whereby techniques are being developed to transplant autologous cells onto biodegradable scaffolds to ultimately form new functional tissue in vitro and in vivo. Our laboratory has focused on the tissue engineering of heart valves, and we have fabricated a trileaflet heart valve scaffold from a biodegradable polymer, a polyhydroxyalkanoate. In this experiment we evaluated the suitability of this scaffold material as well as in vitro conditioning to create viable tissue for tissue engineering of a trileaflet heart valve. We constructed a biodegradable and biocompatible trileaflet heart valve scaffold from a porous polyhydroxyalkanoate (Meatabolix Inc, Cambridge, MA). The scaffold consisted of a cylindrical stent (1 x 15 x 20 mm inner diameter) and leaflets (0.3 mm thick), which were attached to the stent by thermal processing techniques. The porous heart valve scaffold (pore size 100 to 240 microm) was seeded with vascular cells grown and expanded from an ovine carotid artery and placed into a pulsatile flow bioreactor for 1, 4, and 8 days. Analysis of the engineered tissue included biochemical examination, enviromental scanning electron microscopy, and histology. It was possible to create a trileaflet heart valve scaffold from polyhydroxyalkanoate, which opened and closed synchronously in a pulsatile flow bioreactor. The cells grew into the pores and formed a confluent layer after incubation and pulsatile flow exposure. The cells were mostly viable and formed connective tissue between the inside and the outside of the porous heart valve scaffold. Additionally, we demonstrated cell proliferation (DNA assay) and the capacity to generate collagen as measured by hydroxyproline assay and movat-stained glycosaminoglycans under in vitro pulsatile flow conditions. Polyhydroxyalkanoates can be used to fabricate a porous, biodegradable heart valve scaffold. The cells appear to be viable and extracellular matrix formation was induced
Steffensen, Maria Charlotte; Maegbaek, Merete Lund; Laurberg, Peter
Increased risk of heart valve disease during treatment with certain dopamine agonists, such as cabergoline, has been observed in patients with Parkinson's disease. The same compound is used to treat hyperprolactinemia, but it is unknown whether this also associates with heart valve disease....
Full Text Available Patients with mechanical valve prostheses require a lifelong anticoagulant treatment. The combined use of Warfarin and low-dose aspirin appears to reduce the risk of valve thrombosis and systemic embolism at a low risk of bleeding. The management of women with prosthetic heart valves during pregnancy poses a particular challenge, as there are no available controlled clinical trials to provide guidelines for effective antithrombotic therapy. Oral anticoagulants, such as Warfarin, cause foetal embryopathy; unfractionated heparin and low-molecular-weight heparin have been reported to be ineffective in preventing thromboembolic complications.This article discusses the available data and the most recent guidelines in the antithrombotic management of patients with prosthetic valves, and antithrombotic therapy in various clinical situations such as pregnant women with prosthetic heart valves, and patients with prosthetic heart valves undergoing noncardiac surgery.
Kostrzewa, Benita; Rybak, Zbigniew
Artificial heart valves can be classified into mechanical and biological. We have three types of mechanical heart valves: caged ball, tilting disc and bileaflet. Mechanical heart valves are made from various materials. They may be produced from metals, ceramics and polymers, e.g.: stainless steel, titanium, silicone, pyrolytic carbon. Biological valves are made from synthetic components (e.g.: PTFE, Dacron) and materials of biological origin (e.g.: cow pericardium, pig heart valve). We have also identified transcatheter aortic valve implantation (TAVI). TAVI may be produced from metals, ceramics and polymers (e.g.: stainless steel, titanium, Dacron) and biological material (e.g.: pig heart valve). This paper describes advantages and disadvantages of different types of artificial heart valves. The lifespan of mechanical valves is 20-30 years and they can be used for patients of any age. Mechanical valves have also disadvantages--anticoagulants are required to prevent thrombosis. Biological valves are made from natural materials, so they do not require prolonged anticoagulation. Their lifetime is 10-15 years, so they are offered to patients over 40 years. Another problem is the occurrence of calcification.
He, Z; Xi, B; Zhu, K; Hwang, N H
The induction of mechanical heart valve (MHV) cavitation was investigated using a 27 mm Medtronic Hall (MH27) tilting disk valve. The MH27 valve was mounted in the mitral position of a simulating pulse flow system, and stroboscopic lighting used to visualize cavitation bubbles on the occluder inflow surface at the instant of valve closure. MHV cavitation was monitored using a digital camera with 0.04 mm/pixel resolution sufficient to render the tiny bubbles clearly visible on the computer monitor screen. Cavitation on MH27 valve was classified as five types according to the time, site and shape of the cavitation bubbles. Valve cavitation occurred at the instant of occluder impact with the valve seat at closing. The impact motion was subdivided into three temporal phases: (i) squeezing flow; (ii) elastic collision; and (iii) leaflet rebound. MHV cavitation caused by vortices was found to be initiated by the squeezing jet and/or by the transvalvular leakage jets. By using a tension wave which swept across the occluder surface immediately upon elastic impact, nuclei in the vortex core were expanded to form cavitation bubbles. Analysis of the shape and location of the cavitation bubbles permitted a better understanding of MHV cavitation mechanisms, based on the fluid dynamics of jet vortex and tension wave propagations.
Full Text Available Patients with replaced heart valve submitted to echocardiographic examination may have symptoms related either to valvular malfunction or ventricular dysfunction from different causes. Clinical examination is not reliable in a prosthetic valve evaluation and the main information regarding its function could be obtained using different cardiac ultrasound modalities. This review provides a description of echocardiographic and Doppler techniques useful in evaluation of prosthetic heart valves. For the interpretation of echocardiography there is a need in special knowledge of prosthesis types and possible reasons of prosthetic function deterioration. Echocardiography allows to reveal valve thrombosis, pannus formation, vegetation and such complications of infective endocarditis as valve ring abscess or dehiscence. Transthoracic echocardiography requires different section plane angles and unconventional views. Transesophageal echocardiography is more often used than in native valve examination due to better visualization of prosthetic valve structure and function. Three-dimensional echocardiography could provide more detailed visual information especially in the assessment of paravalvular regurgitation or valve obstruction.
Lo, Chi-Wen; Liu, Jia-Shing; Li, Chi-Pei; Lu, Po-Chien; Hwang, Ned H
Accelerated testing provides a substantial amount of data on mechanical heart valve durability in a short period of time, but such conditions may not accurately reflect in vivo performance. Cavitation, which occurs during mechanical heart valve closure when local flow field pressure decreases below vapor pressure, is thought to play a role in valve damage under accelerated conditions. The underlying flow dynamics and mechanisms behind cavitation bubble formation are poorly understood. Under physiologic conditions, random perivalvular cavitation is difficult to capture. We applied accelerated testing at a pulse rate of 600 bpm and transvalvular pressure of 120 mm Hg, with synchronized videographs and high-frequency pressure measurements, to study cavitation of the Medtronic Hall Standard (MHS), Medtronic Hall D-16 (MHD), and Omni Carbon (OC) valves. Results showed cavitation bubbles between 340 and 360 micros after leaflet/housing impact of the MHS, MHD, and OC valves, intensified by significant leaflet rebound. Squeeze flow, Venturi, and water hammer effects each contributed to cavitation, depending on valve design.
Hoffmeister, H.M.; Pirschel, J.
To determine the radiographic appearance of prosthetic heart valves 15 different models were investigated in situ. Fluoroscopy with detail radiographs and standard chest-radiographs were used for identification of the specific type of the prosthesis, for determination of details of the valve apparatus and for evaluation of the motion of the valve disc/ball. - Fluoroscopy and routine radiographs provided sufficient information to identify all prosthetic heart valves with exception of the Xenomedica bioprosthesis. If radiopaque material was used for the valve disc/ball, fluoroscopic assessment of the function of the valve was possible. Thus, in most types of disc or ball valves the opening/closing of the valve can be visualized, whereas in bioprosthesis a radiological determination of the mechanical function is not possible. (orig.) [de
Amstrup Funder, Jonas; Christian Danielsen, Carl; Baandrup, Ulrik
BACKGROUND: Heart valves which exist naturally in an extreme-pressure system must have evolved in a way to resist the stresses of high pressure. Giraffes are interesting as they naturally have a blood pressure twice that of humans. Thus, knowledge regarding giraffe heart valves may aid...... in developing techniques to design improved pressure-resistant biological heart valves. METHODS: Heart valves from 12 giraffes and 10 calves were explanted and subjected to either biomechanical or morphological examinations. Strips from the heart valves were subjected to cyclic loading tests, followed...... in giraffes than in calves, which would make giraffe valves more resistant to the high-pressure forces. However, collagen also stiffens and thickens the valves. The mitral leaflets showed similar (but mostly insignificant) trends in strength, stiffness, and collagen content....
Dijkman, P.E.; Fioretta, E.S.; Frese, L.; Pasqualini, F.S.; Hoerstrup, S.P.
The incidence of severe valvular dysfunctions (e.g., stenosis and insufficiency) is increasing, leading to over 300,000 valves implanted worldwide yearly. Clinically used heart valve replacements lack the capacity to grow, inherently requiring repetitive and high-risk surgical interventions during
Umakanthan, Ramanan; Ghanta, Ravi K; Rangaraj, Aravind T; Lee, Lawrence S; Laurence, Rita G; Fox, John A; Mihaljevic, Tomislav; Bolman, Ralph M; Cohn, Lawrence H; Chen, Frederick Y
Open heart surgery is commonly associated with cardiopulmonary bypass and cardioplegic arrest. The attendant risks of cardiopulmonary bypass may be prohibitive in high-risk patients. We present a novel endoscopic technique of performing tricuspid valve repair without cardiopulmonary bypass in a beating ovine heart. Six sheep underwent sternotomy and creation of a right heart shunt to eliminate right atrial and right ventricular blood for clear visualization. The superior vena cava, inferior vena cava, pulmonary artery, and coronary sinus were cannulated, and the blood flow from these vessels was shunted into the pulmonary artery via a roller pump. The posterior leaflet of the tricuspid valve was partially excised to create tricuspid regurgitation, which was confirmed by Doppler echocardiography. A 7.0-mm fiberoptic videoscope was inserted into the right atrium to visualize the tricuspid valve. Under cardioscopic vision, an endoscopic needle driver was inserted into the right atrium, and a concentric stitch was placed along the posterior annulus to bicuspidize the tricuspid valve. Doppler echocardiography confirmed reduction of tricuspid regurgitation. All animals successfully underwent and tolerated the surgical procedure. The right heart shunt generated a bloodless field, facilitating cardioscopic tricuspid valve visualization. The endoscopic stitch resulted in annular plication and functional tricuspid valve bicuspidization, significantly reducing the degree of tricuspid regurgitation. Cardioscopy enables less invasive, beating-heart tricuspid valve surgery in an ovine model. This technique may be useful in performing right heart surgery without cardiopulmonary bypass in high-risk patients.
An improved understanding of the roles of hemodynamic forces play in cardiac development and the pathogenesis of cardiac disease will have significant scientific and clinical impact. I will focus on the role of fluid dynamics in congenital heart disease and aortic valve calcification. Congenital heart defects are the most common form of birth defect. Aortic valve calcification/stenosis is the third leading cause of adult heart disease and the most common form of acquired valvular disease in developed countries. Given the high incidence of these diseases and their associated morbidity and mortality, the potential translational impact of an improved understanding of cardiac hemodynamic forces is very large. Division of Pediatric Cardiology, Rady Children's Hospital, San Diego
Silvestre, Francisco-Javier; Gil-Raga, Irene; Martinez-Herrera, Mayte; Lauritano, Dorina; Silvestre-Rangil, Javier
Patients scheduled for heart valve surgery should be free of any oral infectious disorders that might pose a risk in the postoperative period. Few studies have been made on the dental conditions of such patients prior to surgery. The present study describes the most frequent prior oral diseases in this population group. A prospective, observational case-control study was designed involving 60 patients (30 with heart valve disease and 30 controls, with a mean age of 71 years in both groups). A dental exploration was carried out, with calculation of the DMFT (decayed, missing and filled teeth) index and recording of the periodontal parameters (plaque index, gingival bleeding index, periodontal pocket depth, and attachment loss). The oral mucosa was also examined, and panoramic X-rays were used to identify possible intrabony lesions. Significant differences in bacterial plaque index were observed between the two groups ( p <0.05), with higher scores in the patients with valve disease. Probing depth and the presence of moderate pockets were also greater in the patients with valve disease than among the controls ( p <0.01). Sixty percent of the patients with valve disease presented periodontitis. Patients scheduled for heart valve surgery should be examined for possible active periodontitis before the operation. Those individuals found to have periodontal disease should receive adequate periodontal treatment before heart surgery. Key words: Valve disease, aortic, mitral, heart surgery, periodontitis.
Beelen, M.J.; Neerincx, P.E.; Molengraft, van de M.J.G.
A disposable injection molded bioreactor for growing tissue-engineered heart valves is controlled to mimic the physiological heart cycle. Tissue-engineered heart valves, cultured from human stem cells, are a possible alternative for replacing failing aortic heart valves, where nowadays biological
Hansen, T B; Zwisler, Ann-Dorthe; Berg, S K
BACKGROUND: Owing to a lack of evidence, patients undergoing heart valve surgery have been offered exercise-based cardiac rehabilitation (CR) since 2009 based on recommendations for patients with ischaemic heart disease in Denmark. The aim of this study was to investigate the impact of CR...... expensive outpatient visits. Further studies should investigate the benefits of CR to heart valve surgery patients as part of a formal cost-utility analysis....
Rasmussen, Caroline Elisabeth; Falk, Bo Torkel; Zois, Nora Elisabeth
Autonomic modulation of heart rhythm is thought to influence the pathophysiology of myxomatous mitral valve disease (MMVD).......Autonomic modulation of heart rhythm is thought to influence the pathophysiology of myxomatous mitral valve disease (MMVD)....
Pistecky, P.; Havlik, P.; Van Kasteren, J.
The action of any pump will start to decline when the valves no longer close properly. The same goes for the heart, the pump that maintains the circulation in our vascular system. Consequently, a major field of focus of open heart surgery is the repair or replacement of heart valves. Petr Havl a
Victoria McIntyre, BASc Chemical Engineering (2018 candidate
Full Text Available Infective endocarditis (IE is an infection that does not usually respond rapidly to treatment, often because its early symptoms are non-specific. The diseased valves (native or bioprosthetic may be calcified and the thrombotic vegetations on them typically friable and embolize easily. Left untreated IE leads to damage to the infected valve and to congestive heart failure (CHF. Its treatment usually requires heart valve replacement. Our 69-year-old patient had IE, and underwent aortic valve replacement (AVR with a bioprosthesis. This case stresses the complications of IE and its tendency to recur in patients with bioprosthetic heart valves (BHV who previously had IE.
Prosthetic heart valve (PHV) dysfunction is an infrequent but potentially life-threatening disease with a heterogeneous clinical presentation. Patients with PHV dysfunction clinically can present with symptoms of congestive heart failure (dyspnea, fatigue, edema), fever, angina pectoris, dizziness
Sibilitz, K L; Berg, S K; Thygesen, L C
investigated. RESULTS: After valve surgery, the self-reported health was lower (Short Form-36 (SF-36) Physical Component Scale (PCS): 44.5 vs. 50.6 and Mental Component Scale (MCS): 51.9 vs. 55.0, pClinical signs......BACKGROUND: After heart valve surgery, knowledge on long-term self-reported health status and readmission is lacking. Thus, the optimal strategy for out-patient management after surgery remains unclear. METHODS: Using a nationwide survey with linkage to Danish registers with one year follow-up, we...... included all adults 6-12months after heart valve surgery irrespective of valve procedure, during Jan-June 2011 (n=867). Participants completed a questionnaire regarding health-status (n=742), and answers were compared with age- and sex-matched healthy controls. Readmission rates and mortality were...
Brody, Sarah; Anilkumar, Thapasimuthu; Liliensiek, Sara; Last, Julie A; Murphy, Christopher J; Pandit, Abhay
A fully effective prosthetic heart valve has not yet been developed. A successful tissue-engineered valve prosthetic must contain a scaffold that fully supports valve endothelial cell function. Recently, topographic features of scaffolds have been shown to influence the behavior of a variety of cell types and should be considered in rational scaffold design and fabrication. The basement membrane of the aortic valve endothelium provides important parameters for tissue engineering scaffold design. This study presents a quantitative characterization of the topographic features of the native aortic valve endothelial basement membrane; topographical features were measured, and quantitative data were generated using scanning electron microscopy (SEM), atomic force microscopy (AFM), transmission electron microscopy (TEM), and light microscopy. Optimal conditions for basement membrane isolation were established. Histological, immunohistochemical, and TEM analyses following decellularization confirmed basement membrane integrity. SEM and AFM photomicrographs of isolated basement membrane were captured and quantitatively analyzed. The basement membrane of the aortic valve has a rich, felt-like, 3-D nanoscale topography, consisting of pores, fibers, and elevations. All features measured were in the sub-100 nm range. No statistical difference was found between the fibrosal and ventricular surfaces of the cusp. These data provide a rational starting point for the design of extracellular scaffolds with nanoscale topographic features that mimic those found in the native aortic heart valve basement membrane.
Ilina, Anna; Lasso, Andras; Jolley, Matthew A.; Wohler, Brittany; Nguyen, Alex; Scanlan, Adam; Baum, Zachary; McGowan, Frank; Fichtinger, Gabor
PURPOSE: Patient-specific heart and valve models have shown promise as training and planning tools for heart surgery, but physically realistic valve models remain elusive. Available proprietary, simulation-focused heart valve models are generic adult mitral valves and do not allow for patient-specific modeling as may be needed for rare diseases such as congenitally abnormal valves. We propose creating silicone valve models from a 3D-printed plastic mold as a solution that can be adapted to any individual patient and heart valve at a fraction of the cost of direct 3D-printing using soft materials. METHODS: Leaflets of a pediatric mitral valve, a tricuspid valve in a patient with hypoplastic left heart syndrome, and a complete atrioventricular canal valve were segmented from ultrasound images. A custom software was developed to automatically generate molds for each valve based on the segmentation. These molds were 3D-printed and used to make silicone valve models. The models were designed with cylindrical rims of different sizes surrounding the leaflets, to show the outline of the valve and add rigidity. Pediatric cardiac surgeons practiced suturing on the models and evaluated them for use as surgical planning and training tools. RESULTS: Five out of six surgeons reported that the valve models would be very useful as training tools for cardiac surgery. In this first iteration of valve models, leaflets were felt to be unrealistically thick or stiff compared to real pediatric leaflets. A thin tube rim was preferred for valve flexibility. CONCLUSION: The valve models were well received and considered to be valuable and accessible tools for heart valve surgery training. Further improvements will be made based on surgeons' feedback.
Jung, Young mee; Kim, Soo Hyun
Artificial heart valves are used in valvular heart diseases, but these valves have disadvantages that they cannot grow, repair and remodel. In current study, the strategies to development of in vitro cultured functional tissue by tissue engineering is available to heart valve disease. In the point of using viable autolougous cells, tissue engineered heart valves have some advantage to include that they can repair, remodel, and grow. Because heart valve is placed under the strong shear stress condition by pumping of heart, the durability of tissue-engineered heart valves is now questionable. The purpose of the study is to evaluate of the durability of tissue engineered heart valve with surface modified scaffolds under hemodynamic conditions
Kamioka, Norihiko; Caughron, Hope; Corrigan, Frank; Block, Peter; Babaliaros, Vasilis
Currently, there are no recommendations regarding the selection of valve type for a transcatheter heart valve (THV)-in-THV procedure. A supra-annular valve design may be superior in that it results in a larger effective orifice area and may have a lower chance of valve thrombosis after THV-in-THV. In this report, we describe the use of a supra-annular valve strategy for an early degenerated THV. © 2018 Wiley Periodicals, Inc.
Pavone, Luigi Michele; Spina, Anna; Lo Muto, Roberta; Santoro, Dionea; Mastellone, Vincenzo; Avallone, Luigi
Multiple evidence demonstrate a role for serotonin and its transporter SERT in heart valve development and disease. By utilizing a Cre/loxP system driven by SERT gene expression, we recently demonstrated a regionally restricted distribution of SERT-expressing cells in developing mouse heart. In order to characterize the cell types exhibiting SERT expression within the mouse heart valves at early developmental stages, in this study we performed immunohistochemistry for Islet1 (Isl1) and connexin-43 (Cx-43) on heart sections from SERT Cre/+ ;ROSA26R embryos previously stained with X-gal. We observed the co-localization of LacZ staining with Isl1 labelling in the outflow tract, the right ventricle and the conal region of E11.5 mouse heart. Cx-43 labelled cells co-localized with LacZ stained cells in the forming atrioventricular valves. These results demonstrate the cardiomyocyte phenotype of SERT-expressing cells in heart valves of the developing mouse heart, thus suggesting an active role of SERT in early heart valve development.
Johansen, Peter; Engholt, Henrik; Tang, Mariann; Nybo, Rasmus F; Rasmussen, Per D; Nielsen-Kudsk, Jens Erik
Patients with degraded bioprosthetic heart valves (BHV) who are not candidates for valve replacement may benefit from transcatheter valve-in-valve (VIV) therapy. However, in smaller-sized surgical BHV the resultant orifice may become too narrow. To overcome this, the valve frame can be fractured by a high-pressure balloon prior to VIV. However, knowledge on fracture pressures and mechanics are prerequisites. The aim of this study was to identify the fracture pressures needed in BHV, and to describe the fracture mechanics. Commonly used BHV of small sizes were mounted on a high-pressure balloon situated in a biplane fluoroscopic system with a high-speed camera. The instant of fracture was captured along with the balloon pressure. The valves were inspected for material protrusion and later dissected for fracture zone investigation and description. The valves with a polymer frame fractured at a lower pressure (8-10 atm) than those with a metal stent (19-26 atm). None of the fractured valves had elements protruding. VIV procedures in small-sized BHV may be performed after prior fracture of the valve frame by high-pressure balloon dilatation. This study provides tentative guidelines for expected balloon sizes and pressures for valve fracturing.
Habets, Jesse; Mali, Willem P.T.M.; Budde, Ricardo P.J.; Brink, Renee B.A. van den; Uijlings, Ruben; Spijkerboer, Anje M.; Chamuleau, Steven A.J.
Patients with prosthetic heart valves may require assessment for coronary artery disease. We assessed whether valve artefacts hamper coronary artery assessment by multidetector CT. ECG-gated or -triggered CT angiograms were selected from our PACS archive based on the presence of prosthetic heart valves. The best systolic and diastolic axial reconstructions were selected for coronary assessment. Each present coronary segment was scored for the presence of valve-related artefacts prohibiting coronary artery assessment. Scoring was performed in consensus by two observers. Eighty-two CT angiograms were performed on a 64-slice (n = 27) or 256-slice (n = 55) multidetector CT. Eighty-nine valves and five annuloplasty rings were present. Forty-three out of 1160 (3.7%) present coronary artery segments were non-diagnostic due to valve artefacts (14/82 patients). Valve artefacts were located in right coronary artery (15/43; 35%), left anterior descending artery (2/43; 5%), circumflex artery (14/43; 32%) and marginal obtuse (12/43; 28%) segments. All cobalt-chrome containing valves caused artefacts prohibiting coronary assessment. Biological and titanium-containing valves did not cause artefacts except for three specific valve types. Most commonly implanted prosthetic heart valves do not hamper coronary assessment on multidetector CT. Cobalt-chrome containing prosthetic heart valves preclude complete coronary artery assessment because of severe valve artefacts. circle Most commonly implanted prosthetic heart valves do not hamper coronary artery assessment circle Prosthetic heart valve composition determines the occurrence of prosthetic heart valve-related artefacts circle Bjoerk-Shiley and Sorin tilting disc valves preclude diagnostic coronary artery segment assessment. (orig.)
Full Text Available Using the electrospinning process the geometry of a heart valve is not replicable by just one manufacturing process. To produce heart valve scaffolds the heart valve leaflets and the vessel have to be produced in separated spinning processes. For the final product of a heart valve they have to be mated afterwards. In this work an already existing three-axes laser was enhanced to laser weld those scaffolds. The automation control software is based on the robot operating system (ROS. The mechatronically control is done by an Arduino Mega. A graphical user interface (GUI is written with Python and Kivy.
Kaasi, Andreas; Cestari, Idágene A.; Stolf, Noedir A G.
The 'biomimetic' approach to tissue engineering usually involves the use of a bioreactor mimicking physiological parameters whilst supplying nutrients to the developing tissue. Here we present a new heart valve bioreactor, having as its centrepiece a ventricular assist device (VAD), which exposes...... chamber. Subsequently, applied vacuum to the pneumatic chamber causes the blood chamber to fill. A mechanical heart valve was placed in the VAD's inflow position. The tissue engineered (TE) valve was placed in the outflow position. The VAD was coupled in series with a Windkessel compliance chamber...
Rajani, Ronak; Khattar, Rajdeep; Chiribiri, Amedeo; Victor, Kelly; Chambers, John
Unidentified heart valve disease is associated with a significant morbidity and mortality. It has therefore become important to accurately identify, assess and monitor patients with this condition in order that appropriate and timely intervention can occur. Although echocardiography has emerged as the predominant imaging modality for this purpose, recent advances in cardiac magnetic resonance and cardiac computed tomography indicate that they may have an important contribution to make. The current review describes the assessment of regurgitant and stenotic heart valves by multimodality imaging (echocardiography, cardiac computed tomography and cardiac magnetic resonance) and discusses their relative strengths and weaknesses
Rajani, Ronak, E-mail: Dr.R.Rajani@gmail.com [Department of Cardiology, St. Thomas’ Hospital, London (United Kingdom); Khattar, Rajdeep [Department of Cardiology, Royal Brompton Hospital, London (United Kingdom); Chiribiri, Amedeo [Divisions of Imaging Sciences, The Rayne Institute, St. Thomas' Hospital, London (United Kingdom); Victor, Kelly; Chambers, John [Department of Cardiology, St. Thomas’ Hospital, London (United Kingdom)
Unidentified heart valve disease is associated with a significant morbidity and mortality. It has therefore become important to accurately identify, assess and monitor patients with this condition in order that appropriate and timely intervention can occur. Although echocardiography has emerged as the predominant imaging modality for this purpose, recent advances in cardiac magnetic resonance and cardiac computed tomography indicate that they may have an important contribution to make. The current review describes the assessment of regurgitant and stenotic heart valves by multimodality imaging (echocardiography, cardiac computed tomography and cardiac magnetic resonance) and discusses their relative strengths and weaknesses.
Samiei, Niloufar; Amirsardari, Mandana; Rezaei, Yousef; Parsaee, Mozhgan; Kashfi, Fahimeh; Hantoosh Zadeh, Sedigheh; Beikmohamadi, Somayeh; Fouladi, Masoumeh; Hosseini, Saeid; Peighambari, Mohammad Mehdi; Mohebbi, Ahmad
Physiologic changes during pregnancy can deteriorate or improve patients' hemodynamic status in the setting of valvular heart disease. There are sparse data regarding the effect of pregnancy on valve hemodynamics in normal pregnant women with known valvular heart disease. In a prospective study from July 2014 to January 2016, a total of 52 normal pregnant women who had mitral stenosis, aortic stenosis, or a history of mitral valve or aortic valve replacements were assessed. All patients underwent echocardiographic examinations and hemodynamic parameters were measured for both the mitral valve and aortic valve at first, second, and third trimesters. The parameters included mean gradient, peak gradient, mean gradient/heart rate, peak gradient/heart rate, pressure halftime, dimensionless velocity index, and valve area. Although most hemodynamic parameters (i.e., mean gradient, peak gradient, mean gradient/heart rate, and peak gradient/heart rate) increased approximately 50% from first to second trimester and first to third trimester (p 0.05). The ratio of changes between trimesters for valve area and dimensionless velocity index were comparable. No clinical decompensations were observed except for 3 and 7 cases of deterioration to functional class II at second and third trimesters, respectively. In conclusion, during a full-term and uncomplicated pregnancy, mitral and aortic valve gradients increase without significant changes in valve area that are more marked between the second and first trimester than between the third and second trimester. Copyright © 2016 Elsevier Inc. All rights reserved.
Willems, I. E.; Havenith, M. G.; Smits, J. F.; Daemen, M. J.
Heart valves are an important denominator of the function of the heart but detailed studies of structural alterations of heart valves after hemodynamic changes are lacking. Structural alterations of heart valves, including DNA synthesis, collagen mRNA, and protein concentration were measured in
Gataulin, Yakov A.; Khorobrov, Svyatoslav V.; Yukhnev, Andrey D.
Bileaflet mechanical valves are most commonly used for heart valve replacement. Nowadays swirling blood flow is registered in different parts of the cardiovascular system: left ventricle, aorta, arteries and veins. In present contribution for the first time the physiological swirling flow inlet conditions are used for numerical simulation of aortic bileaflet mechanical heart valve hemodynamics. Steady 3-dimensional continuity and RANS equations are employed to describe blood motion. The Menter SST model is used to simulate turbulence effects. Boundary conditions are corresponded to systolic peak flow. The domain was discretized into hybrid tetrahedral and hexahedral mesh with an emphasis on wall boundary layer. A system of equations was solved in Ansys Fluent finite-volume package. Noticeable changes in the flow structure caused by inlet swirl are shown. The swirling flow interaction with the valve leaflets is analyzed. A central orifice jet changes its cross-section shape, which leads to redistribution of wall shear stress on the leaflets. Transvalvular pressure gradient and area-averaged leaflet wall shear stress increase. Physiological swirl intensity noticeably reduces downstream of the valve.
Yaghoubi, A.R.; Raeesi, K.
Assessment of surgical pathology of excised heart valves in a referral hospital in Iran in a five years period. This retrospective descriptive study was done from 2002 to 2005 in Rajaie heart center in Tehran, Iran. Surgery and pathology records of patients who underwent valve replacement or repair surgery were reviewed. Of 1563 patients 738 (47.2%) underwent mitral, 565 (36.1%) aortic, and 215 (14%) multivalve operation. Most common pathology of mitral valve was rheumatic (68%), while degenerative calcific pathology was dominant in aortic valve (52%). Rheumatic involvement was 46%, and degenerative pathology was common in tricuspid and pulmonary valves (50% and 67%, respectively). Time trend analysis shows no significant variation in excised valves pathology or pattern from 2002 to 2005 (p=0.112). Rheumatic pathology in excised heart valves is still common in this referral heart center in Iran, and no obvious change in this pattern was found during a 5 years period. (author)
Voldstedlund, Marianne; Fuursted, Kurt; Bruun, Niels Eske
The degree to which the results of valve culture depend on different laboratory procedures as well as other factors is unknown. The aim of this study was to compare the results of heart valve culture at 2 different endocarditis centres in order to clarify this.......The degree to which the results of valve culture depend on different laboratory procedures as well as other factors is unknown. The aim of this study was to compare the results of heart valve culture at 2 different endocarditis centres in order to clarify this....
Dobrova, N B; Agafonov, A V; Barbarash, L S; Zavalishin, N N; Neniukov, A K
Hydraulic characteristics of heart valve bioprostheses mounted on supporting structures of various rigidity have been studied under physiologic conditions. An actual mobility of the supporting structures made of different polymers is determined. Static and dynamic components of the support displacements have been shown to develop as the bioprosthesis is under the load, the dynamic component being strongly dependent upon the rigidity of fastening the bioprosthesis on the axis. It is noted that considerable improvements in hydraulic characteristics of bioprostheses are achieved through the use of flexible supporting structures.
Kwon, Murray H; Shemin, Richard J
Tricuspid valve regurgitation (TVR) in the orthotopic heart transplant (OHT) recipient is quite common and has varied clinical sequelae. In its severest forms, it can lead to right-sided failure symptoms indistinguishable from that seen in native heart TVR disease. While certain implantation techniques are widely recognized to reduce the risk of TVR in the cardiac allograft, concomitant tricuspid annuloplasty, while having advocates, is not currently accepted as a routinely established adjunct. Decisions to surgically correct TVR in the OHT recipient must be made carefully, as certain clinical scenarios have high risk of failure. Like in the native heart, anatomic etiologies typically have the greatest chances for success compared to functional etiologies. While repair options have been utilized, there is emerging data to support replacement as the more durable option. While mechanical prostheses are impractical in the heart transplant recipient, biologic valves offer the advantage of continued access to the right ventricle for biopsies in addition to acceptable durability in the low pressure system of the right side.
In this thesis a flexible leaflet polymeric prosthetic aortic heart valve was designed, manufactured, and tested. The prosthesis was designed with the aim of overcoming the need for anticoagulant therapy, which is required for current mechanical prostheses; while also having lifelong durability, which current bioprosthetic heart valves are not able to achieve. Inspired by the anisotropic architecture of collagen in the natural valve, a shortlist of polystyrene based block copolymers (BCPs), w...
Sordelli, Chiara; Severino, Sergio; Ascione, Luigi; Coppolino, Pasquale; Caso, Pio
Patients submitted to valve replacement with mechanical or biological prosthesis, may present symptoms related either to valvular malfunction or ventricular dysfunction from other causes. Because a clinical examination is not sufficient to evaluate a prosthetic valve, several diagnostic methods have been proposed to assess the functional status of a prosthetic valve. This review provides an overview of echocardiographic and Doppler techniques useful in evaluation of prosthetic heart valves. Compared to native valves, echocardiographic evaluation of prosthetic valves is certainly more complex, both for the examination and the interpretation. Echocardiography also allows discriminating between intra- and/or peri-prosthetic regurgitation, present in the majority of mechanical valves. Transthoracic echocardiography (TTE) requires different angles of the probe with unconventional views. Transesophageal echocardiography (TEE) is the method of choice in presence of technical difficulties. Three-dimensional (3D)-TEE seems to be superior to 2D-TEE, especially in the assessment of paravalvular leak regurgitation (PVL) that it provides improved localization and analysis of the PVL size and shape. PMID:28465917
Kheradvar, Arash; Groves, Elliott M; Falahatpisheh, Ahmad; Mofrad, Mohammad K; Hamed Alavi, S; Tranquillo, Robert; Dasi, Lakshmi P; Simmons, Craig A; Jane Grande-Allen, K; Goergen, Craig J; Baaijens, Frank; Little, Stephen H; Canic, Suncica; Griffith, Boyce
In this final portion of an extensive review of heart valve engineering, we focus on the computational methods and experimental studies related to heart valves. The discussion begins with a thorough review of computational modeling and the governing equations of fluid and structural interaction. We then move onto multiscale and disease specific modeling. Finally, advanced methods related to in vitro testing of the heart valves are reviewed. This section of the review series is intended to illustrate application of computational methods and experimental studies and their interrelation for studying heart valves.
Lund, K.; Sibilitz, Kirstine Lærum; Kikkenborg Berg, Selina
physical activity levels 6-12 months after heart valve surgery and (1) survival, (2) hospital readmission 18-24 months after surgery and (3) participation in exercise-based cardiac rehabilitation. METHODS: Prospective cohort study with registry data from The CopenHeart survey, The Danish National Patient......OBJECTIVES: Increased physical activity predicts survival and reduces risk of readmission in patients with coronary heart disease. However, few data show how physical activity is associated with survival and readmission after heart valve surgery. Objective were to assess the association between...... Register and The Danish Civil Registration System of 742 eligible patients. Physical activity was quantified with the International Physical Activity Questionnaire and analysed using Kaplan-Meier analysis and Cox regression and logistic regression methods. RESULTS: Patients with a moderate to high physical...
Vlimmeren, van M.A.A.
Heart valve tissue engineering is a promising technique to overcome the drawbacks of currently used mechanical and prosthetic heart valve replacements. Tissue engineered (TE) heart valves are viable and autologous implants that have the capacity to grow, remodel and repair throughout a patient’s
Bluestein, D; Einav, S
Many problems and complications associated with heart valves are related to the dynamic behavior of the valve and the resultant unsteady flow patterns. An accurate depiction of the spatial and temporal velocity and rms distributions imparts better understanding of flow related valve complications, and may be used as a guideline in valve design. While the generalized correlation between increased turbulence level and the severity of the stenosis is well established, few studies addressed the issue of the intermittent nature of turbulence and its timing in the cardiac cycle, and almost none assessed the effect of a progressive stenosis on the flow characteristics through heart valves. In this experimental work we simulated the type of flow which is present in normal and stenosed valves and conducted a comprehensive investigation of valve hemodynamics, valvular turbulence and morphology under varying degrees of stenosis. The characteristics of valves and stenoses were simulated closely, to achieve the flow conditions that initiate turbulent flow conditions. Laser Doppler anemometry (LDA) measurements were carried out in a pulse duplicator system distal to trileaflet polyurethane prosthetic heart valves, installed at mitral and aortic positions. The effect of the degree of the stenosis was comparatively studied through the structure of the turbulent jets emerging from normal and stenotic heart valves. Maximum turbulence level was achieved during the decelerating phase and correlated to the severity of the stenosis, followed by relaminarization of the flow during the acceleration phase. The intermittent nature of the turbulence emphasized the importance of realizing the timing of the turbulence production and its spatial location for optimizing current valve designs. The plug flow through the normal aortic valve prosthesis was replaced by jet like behavior for a 65% stenosis, with the jet becoming narrower and stronger for a 90% stenosis. The morphology of the velocity
Sibilitz, Kirstine Lærum; Berg, Selina Kikkenborg; Tang, Lars Hermann
BACKGROUND: Exercise-based cardiac rehabilitation may benefit heart valve surgery patients. We conducted a systematic review to assess the evidence for the use of exercise-based intervention programmes following heart valve surgery. OBJECTIVES: To assess the benefits and harms of exercise......-based cardiac rehabilitation compared with no exercise training intervention, or treatment as usual, in adults following heart valve surgery. We considered programmes including exercise training with or without another intervention (such as a psycho-educational component). SEARCH METHODS: We searched...... handsearched Web of Science, bibliographies of systematic reviews and trial registers (ClinicalTrials.gov, Controlled-trials.com, and The World Health Organization International Clinical Trials Registry Platform). SELECTION CRITERIA: We included randomised clinical trials that investigated exercise...
Jahnavi, S., E-mail: email@example.com [Stem Cell and Molecular Biology Laboratory, Department of Biotechnology, Indian Institute of Technology Madras, Chennai 600036, TN (India); Tissue Culture Laboratory, Biomedical Technology Wing, Sree Chitra Tirunal Institute for Medical Sciences and Technology, Poojappura, Trivandrum, Kerala 695012 (India); Kumary, T.V., E-mail: firstname.lastname@example.org [Tissue Culture Laboratory, Biomedical Technology Wing, Sree Chitra Tirunal Institute for Medical Sciences and Technology, Poojappura, Trivandrum, Kerala 695012 (India); Bhuvaneshwar, G.S., E-mail: email@example.com [Trivitron Innovation Centre, Department of Engineering Design, Indian Institute of Technology Madras, Chennai 600036, TN (India); Natarajan, T.S., E-mail: firstname.lastname@example.org [Conducting Polymer laboratory, Department of Physics, Indian Institute of Technology, Madras, Chennai 600036, TN (India); Verma, R.S., E-mail: email@example.com [Stem Cell and Molecular Biology Laboratory, Department of Biotechnology, Indian Institute of Technology Madras, Chennai 600036, TN (India)
Current treatment strategy for end stage valve disease involves either valvular repair or replacement with homograft/mechanical/bioprosthetic valves. In cases of recurrent stenosis/ regurgitation, valve replacement is preferred choice of treatment over valvular repair. Currently available mechanical valves primarily provide durability whereas bioprosthetic valves have superior tissue compatibility but both lack remodelling and regenerative properties making their utility limited in paediatric patients. With advances in tissue engineering, attempts have been made to fabricate valves with regenerative potential using various polymers, decellularized tissues and hybrid scaffolds. To engineer an ideal heart valve, decellularized bovine pericardium extracellular matrix (DBPECM) is an attractive biocompatible scaffold but has weak mechanical properties and rapid degradation. However, DBPECM can be modified with synthetic polymers to enhance its mechanical properties. In this study, we developed a Bio-Hybrid scaffold with non-cross linked DBPECM in its native structure coated with a layer of Polycaprolactone-Chitosan (PCL-CH) nanofibers that displayed superior mechanical properties. Surface and functional studies demonstrated integration of PCL-CH to the DBPECM with enhanced bio and hemocompatibility. This engineered Bio-Hybrid scaffold exhibited most of the physical, biochemical and functional properties of the native valve that makes it an ideal scaffold for fabrication of cardiac valve with regenerative potential. - Highlights: • A Bio-Hybrid scaffold was fabricated with PCL-CH blend and DBPECM. • PCL-CH functionally interacted with decellularized matrix without cross linking. • Modified scaffold exhibited mechanical properties similar to native heart valve. • Supported better fibroblast and endothelial cell adhesion and proliferation. • The developed scaffold can be utilized for tissue engineering of heart valve.
Issa, Issa Farah; Poulsen, Steen Hvitfeldt; Waziri, Farhad
OBJECTIVES: Concern has been raised regarding the long-term durability of the Mitroflow biological heart valve prosthesis. Our aim was to assess the incidence of structural valve degeneration (SVD) for the Mitroflow bioprosthesis in a nationwide study in Denmark including all patients alive......: A total of 173 patients were diagnosed with SVD by echocardiography. Of these, 64 (11%) patients had severe SVD and 109 (19%) patients moderate SVD. Severe SVD was associated with the age of the prosthesis and small prosthesis size [Size 21: hazard ratio (95% confidence interval, CI) 2.72 (0.97-8.56), P...
K. Yu. Klyshnikov
Full Text Available The aim of the study was to perform a series of in vitro tests of a prototype of the developing heart valve prosthesis to evaluate its functional characteristics. Materials and methods. In this work we have used the frames and full prototypes of the prosthesis, consisting of a stent-like stainless steel support frame with mounted biological leaflets and cover. The authors evaluated the calculated and experimental forces necessary for the displacement of the sutureless implanted prosthesis using the test machine under uniaxial tension. The risk of defects and damages to the supporting framework as a result of implantation was evaluated by scanning electron microscopy. The hydrodynamic characteristics of the prosthesis were investigated under physiological conditions and «valvein-valve» implantation. Evaluation of the ergonomics and applicability of the proposed construction on the cadaver heart model of cattle was carried out. Results. As a result of the forces assessment, it was found that the force required to shear the prosthesis was 3.12 ± 0.37 N, while the calculated value was 1.7 N, which is significantly lower than the obtained value. The comparison of the images obtained with small and large magnifications demonstrated the absence of critical surface defects. Additional analysis under the super-large magnifications also did not reveal problem areas. During the hydrodynamic study, it was shown that the average transplant gradient increased slightly from 2.8–3.4 to 3.2–4.5 mm Hg for the initial prosthesis and the «valve-in-valve» complex, respectively. The decrease of the effective orifice area was 6–9% relative to the initial one. Evaluation of the implantation technique demonstrated the consistency of the approach: the use of the developed holder in combination with the balloon implantation system made it possible to position the prosthesis throughout the procedure. Conclusion. The series of tests demonstrates the consistency
Del Gaudio, Costantino; Gasbarroni, Pier Luca; Romano, Giovanni Paolo
End-stage failing heart valves are currently replaced by mechanical or biological prostheses. Both types positively contribute to restore the physiological function of native valves, but a number of drawbacks limits the expected performances. In order to improve the outcome, tissue engineering can offer an alternative approach to design and fabricate innovative heart valves capable to support the requested function and to promote the formation of a novel, viable and correctly operating physiological structure. This potential result is particularly critical if referred to the aortic valve, being the one mainly exposed to structural and functional degeneration. In this regard, the here proposed study presents the fabrication and in vitro characterization of a bioresorbable electrospun heart valve prosthesis using the particle image velocimetry technique either in physiological and pathological fluid dynamic conditions. The scaffold was designed to reproduce the aortic valve geometry, also mimicking the fibrous structure of the natural extracellular matrix. To evaluate its performances for possible implantation, the flow fields downstream the valve were accurately investigated and compared. The experimental results showed a correct functionality of the device, supported by the formation of vortex structures at the edge of the three cusps, with Reynolds stress values below the threshold for the risk of hemolysis (which can be comprised in the range 400-4000N/m(2) depending on the exposure period), and a good structural resistance to the mechanical loads generated by the driving pressure difference. Copyright © 2016 Elsevier Ltd. All rights reserved.
M. Senthilkumar (Newgen Imaging) 1461 1996 Oct 15 13:05:22
BHUVANESHWAR. Biomedical Technology Wing, Sree Chitra Tirunal Institute for Medical Sciences. & Technology ... affect the performance of mechanical heart valves. The clinical performance of ... those who cannot be put under anticoagulant therapy, like women who may still wish to bear children, or hemolytic patients.
Strong, M.; O'Neal, P.D.; Gage, H.N.; Moogk, M.
The Northwest Tissue Center established a human heart valve program in 199 1. It is one of four non-profit tissue banks and one for-profit program that recover and process heart valves in the United States. During the eight years in which the Northwest Tissue Center has been involved in heart valve banking, there have been a total of 673 hearts procured for processing. The age of the donors ranged from <1 to 44 years with a mean of 26.2 years, 66% werw male,and 6.5% of the hearts procered were discarded due to a variety of medical and criteria reason. The primary reasons for differal were questions of possible cancer and questions of high risk behavior/social history. Of the 1,264 cardiovascular tissues processed, 6% were lost because of donor history, 17% were lost because of microbiology results, and 5% were lost because of donor serology . There were total a total of 190 aortic valves and 48 pulmonic conduits transplanted over this time period. The mean age of the recipients was 23.4 with a median or 23 years; 102 of the recipients were less than one year of age. Males comprised 62% of the recipients. Since 1993, there has been a clear shift towards more use of pulmonic valves over aortic valves as a results of the acceptance of the Ross procedure. Early in the program, reports were received from surgeons that some heart valves appeared to have cracks in the conduits. Experimentations in the laboratory led to the discovery that thawing too rapidly would result in cracking of these materials. Packaging was designed to reduce the rate of thawing and this has resolved the problem with cracking. The heart valve program at the Northwest Tissue Center has been very successful in providing the necessary valves for patients in the Northwest Region of the United States
Edwards, Maria-Benedicta; Taylor, Kenneth M
Changes in the age profile of the United Kingdom population and improvements in preoperative and postoperative care have resulted in increasing numbers of very elderly patients undergoing heart valve replacement (HVR) operations. Although HVR operations in nonagenarians are relatively uncommon, the demand for cardiac operations in this age group may increase over time. Outcomes after HVR operations in nonagenarians have not been well described yet. Therefore, the aim of this study was to determine outcomes in terms of early mortality and long-term survival in 35 nonagenarians after HVR operation. Data from the United Kingdom Heart Valve Registry were analyzed and nonagenarian patients were identified. Additional analyzed data include gender, valve position, valve type, valve size, operative priority, follow-up time, and date and cause of death. Kaplan-Meier actuarial curves were calculated to determine accurate 30-day mortality and long-term survival. On average five HVR operations are performed annually in the United Kingdom in nonagenarians with equal numbers of males and females. Aortic valve replacement with a bioprosthetic valve was the most common operation and 86% were elective admissions. Fourteen patients died within the review period; mean time to death was 402 days. Overall 30-day mortality was 17%, which was higher for males compared with females; females also displayed better long-term survival. HVR operations in nonagenarians carry a significantly higher risk of early mortality and reduced long-term survival. Despite increases in the age profile of the population, elective HVR operation with patients aged 90 years or older is likely to remain an infrequent surgical procedure reserved for very carefully selected patients.
Full Text Available In this investigation, scanning electron microscopy was used to characterize the microstructure on the surfaces of animal heart valve cusps/leaflets. The results showed that though these surfaces appear smooth to the naked eye, they are actually comprised of a double hierarchical structure consisting of a cobblestone-like microstructure and nano-cilia along with mastoids with a directional arrangement. Such nanostructures could play a very important role in the hemocompatibility characteristics of heart valves. On this basis, the model of the microstructure was constructed and theoretical analysis was used to obtain optimal geometric parameters for the rough surface of artificial valve cusps/leaflets. This model may help improve reconstructive techniques and it may be beneficial in the design and fabrication of valve substitutes or partial substitutes. Namely, the model may help ameliorate heart valve replacement surgery.
Hansen, Tina Birgitte; Zwisler, Ann Dorthe Olsen; Kikkenborg Berg, Selina
collected data over 18 months (April 2013-October 2014). We recruited nine patients undergoing heart valve surgery from a randomized trial, CopenHeartVR and conducted 27 individual narrative interviews at 2-3 weeks, 3-4 months and 8-9 months after surgery. FINDINGS: Following heart valve surgery...
Pollmann, Agathe Gerwina Elena; Frederiksen, Marianne; Prescott, Eva
PURPOSE: Evidence of the effect of cardiac rehabilitation (CR) after heart valve surgery is scarce, but nevertheless CR is recommended for this group of patients. Therefore, this study assessed the effect of CR on exercise capacity, cardiovascular risk factors, and long-term mortality and morbidity...... ((Equation is included in full-text article.)O2peak) or 6-minute walk test (6MWT). A composite endpoint of all-cause mortality and hospital admission due to myocardial infarction, stroke, heart failure, endocarditis, revascularization, or reoperation was used to assess the hazard ratio between CR attenders...
Suchá, D.; Symersky, Petr; Tanis, W; Mali, Willem P Th M; Leiner, Tim; van Herwerden, LA; Budde, Ricardo P J
Echocardiography and fluoroscopy are the main techniques for prosthetic heart valve (PHV) evaluation, but because of specific limitations they may not identify the morphological substrate or the extent of PHV pathology. Cardiac computed tomography (CT) and magnetic resonance imaging (MRI) have
Valvular heart disease accounts for a substantial part of the cardiovascular disease worldwide with an estimated prevalence of 2.5% in the Western population aged <65 years and over 13% in the population aged >75 years. Surgical prosthetic heart valve (PHV) replacement is the indicated therapy for
Geemen, van D.
Valvular heart disease is a major health problem worldwide causing morbidity and mortality. Heart valve replacement is frequently applied to avoid serious cardiac, pulmonary, or systemic problems. However, the current replacements do not consist of living tissue and, consequently, cannot grow,
An experimental procedure has been developed for rigorous characterization of the fracture resistance and fatigue crack extension in pyrolytic carbon for prosthetic heart valve application. Experiments were conducted under sustained and cyclic loading in a simulated biological environment using Carbomedics Pyrolite carbon. While the material was shown to have modest fracture toughness, it exhibited excellent resistance to subcritical crack growth. The crack growth kinetics in pyrolytic carbon were formulated using a phenomenological description. A fatigue threshold was observed below which the crack growth rate diminishes. A damage tolerance concept based on fracture mechanics was used to develop an engineering design approach for mechanical heart valve prostheses. In particular, a new quantity, referred to as the safe-life index, was introduced to assess the design adequacy against subcritical crack growth in brittle materials. In addition, a weakest-link statistical description of the fracture strength is provided and used in the design of component proof-tests. It is shown that the structural reliability of mechanical heart valves can be assured by combining effective flaw detection and manufacturing quality control with adequate damage tolerance design.
Hansen, Tina; Zwisler, Ann Dorthe; Berg, Selina Kikkenborg
and effect differences were presented in a cost-effectiveness plane and were transformed into net benefit and presented in cost-effectiveness acceptability curves. Results No statistically significant differences were found in total societal costs (-1609 Euros; 95% CI: -6162 to 2942 Euros) or in quality......Background While cardiac rehabilitation in patients with ischaemic heart disease and heart failure is considered cost-effective, this evidence may not be transferable to heart valve surgery patients. The aim of this study was to investigate the cost-effectiveness of cardiac rehabilitation following...... heart valve surgery. Design We conducted a cost-utility analysis based on a randomised controlled trial of 147 patients who had undergone heart valve surgery and were followed for 6 months. Methods Patients were randomised to cardiac rehabilitation consisting of 12 weeks of physical exercise training...
Türkoğlu, İbrahim; Arslan, Ahmet; İlkay, Erdoğan
In this pa per, a decision s up port system is presented for interpretation of the Doppler signals of the heart valve diseases based on the pattern recognition. This paper especially deals with the feature extraction from measured Doppler signal waveforms at the heart valve using the Doppler Ultrasound. Wavelet transforms and power spectrum estimate by Yule-Walker AR method are used to feature extract from the Doppler signals on the timefrequency domain. Wavelet entropy method is applied to ...
Kikkenborg Berg, Selina; Zwisler, Ann-Dorthe; Pedersen, Birthe D.
Heart valve disease is becoming a public health problem due to increasing life expectancy and new treatment methods. Patients are at risk of developing depression, anxiety or post-traumatic stress disorder after heart valve surgery. To better plan proper care, describing and understanding patients...
Kaminsky, R; Dumont, K; Weber, H; Schroll, M; Verdonck, P
The aim of this study was to validate the 2D computational fluid dynamics (CFD) results of a moving heart valve based on a fluid-structure interaction (FSI) algorithm with experimental measurements. Firstly, a pulsatile laminar flow through a monoleaflet valve model with a stiff leaflet was visualized by means of Particle Image Velocimetry (PIV). The inflow data sets were applied to a CFD simulation including blood-leaflet interaction. The measurement section with a fixed leaflet was enclosed into a standard mock loop in series with a Harvard Apparatus Pulsatile Blood Pump, a compliance chamber and a reservoir. Standard 2D PIV measurements were made at a frequency of 60 bpm. Average velocity magnitude results of 36 phase-locked measurements were evaluated at every 10 degrees of the pump cycle. For the CFD flow simulation, a commercially available package from Fluent Inc. was used in combination with inhouse developed FSI code based on the Arbitrary Lagrangian-Eulerian (ALE) method. Then the CFD code was applied to the leaflet to quantify the shear stress on it. Generally, the CFD results are in agreement with the PIV evaluated data in major flow regions, thereby validating the FSI simulation of a monoleaflet valve with a flexible leaflet. The applicability of the new CFD code for quantifying the shear stress on a flexible leaflet is thus demonstrated.
Akhtar, R.P.; Khan, J.S.; Abid, A.R.; Gardezi, S.J.R.
To study patients with mechanical heart valves undergoing non-cardiac surgery and their anticoagulation management during these procedures. Patients with mechanical heart valves undergoing non-cardiac surgical operation during this period, were included. Their anticoagulation was monitored and anticoagulation related complications were recorded. In this study, 507 consecutive patients with a mechanical heart valve replacement were followed-up. Forty two (8.28%) patients underwent non-cardiac surgical operations of which 24 (57.1%) were for abdominal and non-abdominal surgeries, 5 (20.8%) were emergency and 19 (79.2%) were planned. There were 18 (42.9%) caesarean sections for pregnancies. Among the 24 procedures, there were 7(29.1%) laparotomies, 7(29.1%) hernia repairs, 2 (8.3%) cholecystectomies, 2 (8.3%) hysterectomies, 1(4.1%) craniotomy, 1(4.1%) spinal surgery for neuroblastoma, 1(4.1%) ankle fracture and 1(4.1%) carbuncle. No untoward valve or anticoagulation related complication was seen during this period. Patients with mechanical valve prosthesis on life-long anticoagulation, if managed properly, can undergo any type of noncardiac surgical operation with minimal risk. (author)
Lu, Qiyu; Sun, Yi; Duan, Yuyin; Li, Bin; Xia, Jianming; Yu, Songhua; Zhang, Guimin
Valvular heart disease is a leading cause of cardiovascular mortality, especially in China. More than a half of valvular heart diseases are caused by acute rheumatic fever. microRNA is involved in many physiological and pathological processes. However, the miRNA profile of the rheumatic valvular heart disease is unknown. This research is to discuss microRNAs and their target gene pathways involved in rheumatic heart valve disease. Serum miRNA from one healthy individual and four rheumatic heart disease patients were sequenced. Specific differentially expressed miRNAs were quantified by Q-PCR in 40 patients, with 20 low-to-moderate rheumatic mitral valve stenosis patients and 20 severe mitral valve stenosis patients. The target relationship between certain miRNA and predicted target genes were analysis by Luciferase reporter assay. The IL-1β and IL1R1 expression levels were analyzed by immunohistochemistry and western blot in the mitral valve from surgery of mitral valve replacement. The results showed that 13 and 91 miRNAs were commonly upregulated or downregulated in all four patients. Nine miRNAs, 1 upregulated and 8 downregulated, that had a similar fold change in all 4 patients were selected for quantitative PCR verification. The results showed similar results from miRNA sequencing. Within these 9 tested miRNAs, hsa-miR-205-3p and hsa-miR-3909 showed a low degree of dispersion between the members of each group. Hsa miR-205-3p and hsa-miR-3909 were predicted to target the 3'UTR of IL-1β and IL1R1 respectively. This was verified by luciferase reporter assays. Immunohistochemistry and Western blot results showed that the mitral valve from rheumatic valve heart disease showed higher levels of IL- 1β and IL1R1 expression compared with congenital heart valve disease. This suggested a difference between rheumatic heart valve disease and other types of heart valve diseases, with more inflammatory responses in the former. In the present study, by next generation
Murphy, David; Dasi, Lakshmi; Yoganathan, Ajit; Glezer, Ari
The time-periodic closing of bileaflet mechanical heart valves is accompanied by a strong flow transient that is associated with the formation of a counter-rotating vortex pair near the b-datum line of leaflet edges. The strong transitory shear that is generated by these vortices may be damaging to blood elements and may result in platelet activation. In the present work, these flow transients are mitigated using miniature vortex generator arrays that are embedded on the surface of the leaflets. Two vortex generator designs were investigated: one design comprised staggered rectangular fins and the other one staggered hemispheres. The closing transients in the absence and presence of the passive vortex generators are characterized using phase locked PIV measurements. The study utilizes a 25 mm St. Jude Medical valve placed in the aortic position of the Georgia Tech left heart simulator. Measurements of the velocity field in the center plane of the leaflets demonstrate that the dynamics of the transient vortices that precede the formation of the leakage jets can be significantly altered and controlled by relatively simple passive modifications of existing valve designs. Human blood experiments validated the effectiveness of miniature vortex generators in reducing thrombus formation by over 42 percent.
Oomen, Pim J A; Holland, Maria A; Bouten, Carlijn V C; Kuhl, Ellen; Loerakker, Sandra
Tissue growth and remodeling are known to govern mechanical homeostasis in biological tissue, but their relative contributions to homeostasis remain unclear. Here, we use mechanical models, fueled by experimental findings, to demonstrate that growth and remodeling have different effects on heart valve stretch homeostasis during physiological postnatal development. Two developmental stages were considered: early-stage (from infant to adolescent) and late-stage (from adolescent to adult) development. Our models indicated that growth and remodeling play opposing roles in preserving tissue stretch and with time. During early-stage development, excessive tissue stretch was decreased by tissue growth and increased by remodeling. In contrast, during late-stage development tissue stretch was decreased by remodeling and increased by growth. Our findings contribute to an improved understanding of native heart valve adaptation throughout life, and are highly relevant for the development of tissue-engineered heart valves.
Zhao Shihua; Lu Minjie; Zhang Yan; Jiang Shiliang; Liu Yuqing; Zhang Puhong
Objective: To assess the diagnostic value of the magnetic resonance imaging (MRI) for heart valve disease qualitatively and quantitatively. Methods: From 18th Sep, 2004 to 30th Jun, 2005, 56 consecutive patients underwent MR scanning with multiple sequences, including two-dimensional dark and bright sequences, K-space segmented TrueFISP and FLASH cine sequences, as well as velocity-encoded cine MR(VEC-MR). Morphologic and functional parameters were applied to assess the disease qualitatively and quantitatively. For quantitative analysis, Doppler echocardiography was compared to evaluate the reliability of VEC-MR in assessing the severity of aortic valve disease. Correlations coefficient was analyzed by a statistic software (SPSS 13.0), P sq =0.951, P=0.01 for AS and R=0.965, R sq =0.932, P<0.01 for AI). Conclusion: Heart valve diseases can be qualitatively and quantitatively evaluated by MR multiple sequences, especially in aortic valve disease. (authors)
Converse, Gabriel L; Buse, Eric E; Neill, Kari R; McFall, Christopher R; Lewis, Holley N; VeDepo, Mitchell C; Quinn, Rachael W; Hopkins, Richard A
Heart valve tissue engineering offers the promise of improved treatments for congenital heart disorders; however, widespread clinical availability of a tissue engineered heart valve (TEHV) has been hindered by scientific and regulatory concerns, including the lack of a disposable, bioreactor system for nondestructive valve seeding and mechanical conditioning. Here we report the design for manufacture and the production of full scale, functional prototypes of such a system. To evaluate the efficacy of this bioreactor as a tool for seeding, ovine aortic valves were decellularized and subjected to seeding with human mesenchymal stem cells (hMSC). The effects of pulsatile conditioning using cyclic waveforms tuned to various negative and positive chamber pressures were evaluated, with respect to the seeding of cells on the decellularized leaflet and the infiltration of seeded cells into the interstitium of the leaflet. Infiltration of hMSCs into the aortic valve leaflet was observed following 72 h of conditioning under negative chamber pressure. Additional conditioning under positive pressure improved cellular infiltration, while retaining gene expression within the MSC-valve interstitial cell phenotype lineage. This protocol resulted in a subsurface pilot population of cells, not full tissue recellularization. © 2015 Wiley Periodicals, Inc. J Biomed Mater Res Part B: Appl Biomater, 105B: 249-259, 2017. © 2015 Wiley Periodicals, Inc.
Erdelyi, M Jr [Orszagos Roentgen es Sugarfizikai Intezet, Budapest (Hungary)
The possible causes of the disfunction of artificial valves are detailed. Routine X-ray examinations of the chest only rarely reveal direct signs of disfunction, but the haemodynamic alterations (dilated heart chambers, or veins) must be considered as important indirect signs. The characteristics and possibly the cause of the valve insufficience can be analyzed on the basis of kino- and spot radiograms. The symptoms observed by these methods are reported in detail.
Niinimäki, Eetu; Mennander, Ari A; Paavonen, Timo; Kholová, Ivana
Inflammation-associated lymphangiogenesis (IAL) has been identified as part of several acute and chronic inflammation. Sparse data exist on lymphatics during endocarditis. Fifty-two patients with surgically resected valves were included. Endocarditis was present in 18 aortic and 10 mitral valves. Controls consisted of 15 degenerative aortic and 9 degenerative mitral valves. There were 22 males with endocarditis and 17 males in controls. The mean age was 58 (SD 15) years with endocarditis vs. 62 (SD 13) years for controls. Lymphatics were detected by podoplanin antibody immunohistochemistry and morphometrical analysis was performed. The lymphatic density in endocarditis was 833 (SD 529) vessels/mm(2) (range 0-1707) as compared with 39 (SD 60) vessels/mm(2) (range 0-250) in controls (p=0.000). In endocarditis, the mean lymphatic size was 153 (SD 372) μm(2) ranging from 1 to 2034μm(2), whereas it was 30 (SD 29) μm(2), with maximum 90μm(2) and minimum 2μm(2) in controls (p=0.000). IAL is increased in valves with endocarditis as compared with controls. Lymphatics in heart valves may provide a novel means for treatment strategies against endocarditis. Copyright © 2016 Elsevier Ireland Ltd. All rights reserved.
Vossler, John D; Min Ju, Young; Williams, J Koudy; Goldstein, Steven; Hamlin, James; Lee, Sang Jin; Yoo, James J; Atala, Anthony
The long term efficacy of tissue based heart valve grafts may be limited by progressive degeneration characterized by immune mediated inflammation and calcification. To avoid this degeneration, decellularized heart valves with functionalized surfaces capable of rapid in vivo endothelialization have been developed. The aim of this study is to examine the capacity of CD133 antibody-conjugated valve tissue to capture circulating endothelial progenitor cells (EPCs). Decellularized human pulmonary valve tissue was conjugated with CD133 antibody at varying concentrations and exposed to CD133 expressing NTERA-2 cl.D1 (NT2) cells in a microflow chamber. The amount of CD133 antibody conjugated on the valve tissue surface and the number of NT2 cells captured in the presence of shear stress was measured. Both the amount of CD133 antibody conjugated to the valve leaflet surface and the number of adherent NT2 cells increased as the concentration of CD133 antibody present in the surface immobilization procedure increased. The data presented in this study support the hypothesis that the rate of CD133(+) cell adhesion in the presence of shear stress to decellularized heart valve tissue functionalized by CD133 antibody conjugation increases as the quantity of CD133 antibody conjugated to the tissue surface increases.
van Slooten, Ymkje J.; Melle, van Joost P.; Freling, Hendrik G.; Bouma, Berto J.; van Dijk, Arie P. J.; Jongbloed, Monique R. M.; Post, Martijn C.; Sieswerda, Gertjan T.; in 't Veld, Anna Huis; Ebels, Tjark; Voors, Adriaan A.; Pieper, Petronella G.
Objectives To report the prevalence of aortic valve prosthesis patient mismatch (PPM) in an adult population with congenital heart disease (CHD) and its impact on exercise capacity. Adults with congenital heart disease (ACHD) with a history of aortic valve replacement may outgrow their prosthesis
van Slooten, Ymkje J.; van Melle, Joost P.; Freling, Hendrik G.; Bouma, Berto J.; van Dijk, Arie Pj; Jongbloed, Monique Rm; Post, Martijn C.; Sieswerda, Gertjan T.; Huis In 't Veld, Anna; Ebels, Tjark; Voors, Adriaan A.; Pieper, Petronella G.
To report the prevalence of aortic valve prosthesis-patient mismatch (PPM) in an adult population with congenital heart disease (CHD) and its impact on exercise capacity. Adults with congenital heart disease (ACHD) with a history of aortic valve replacement may outgrow their prosthesis later in
Østergaard, Lauge; Valeur, Nana; Ihlemann, Nikolaj
Aims: Patients with left-sided heart valve replacement are considered at high-risk of infective endocarditis (IE). However, data on the incidence and risk factors associated with IE are sparse. Methods and results: Through Danish administrative registries, we identified patients who underwent left.......35-2.15), and cardiac implantable electronic device (CIED) (HR = 1.57, 95% CI 1.19-2.06) were among factors associated with an increased risk of IE. Conclusion: Infective endocarditis after left-sided heart valve replacement is not uncommon and occurs in about 1/20 over 10 years. Male, bioprosthetic valve, and heart...
Gragnano, Felice; Crisci, Mario; Bigazzi, Maurizio Cappelli; Bianchi, Renatomaria; Sperlongano, Simona; Natale, Francesco; Fimiani, Fabio; Concilio, Claudia; Cesaro, Arturo; Pariggiano, Ivana; Diana, Vincenzo; Limongelli, Giuseppe; Cirillo, Plinio; Russo, Mariagiovanna; Golia, Enrica; Calabrò, Paolo
von Willebrand Factor (vWF) is a well-known mediator of hemostasis and vascular inflammation. Its dynamic modulation in the bloodstream, according to hemodynamic conditions, makes it an appealing biomarker in patients with valvular heart disease (VHD). Recent studies highlight the close connection between vWF and VHD, with possible implications in the pathogenesis of VHD, promoting valve aging and calcification or favoring the development of infective endocarditis. Moreover, vWF has been recently proposed as a new diagnostic and prognostic tool in patients with valve stenosis or regurgitation, showing a strict correlation with severity of valve disease, outcome, and bleeding (Heyde syndrome). A novel role for vWF is also emerging in patients undergoing percutaneous or surgical valve repair/replacement to select and stratify patients, evaluate periprocedural bleeding risk, and detect procedural complications. We also report our single-center experience, suggesting, for the first time, possible clinical implications for vWF in percutaneous mitral valve repair (MitraClip). This review summarizes recent advances in the role of vWF in VHD with an updated overview going from bench to operating room.
Kortsmit, J.; Driessen, N.J.B.; Rutten, M.C.M.; Baaijens, F.P.T.
Despite recent progress, mechanical behavior of tissue-engineered heart valves still needs improvement when native aortic valves are considered as a benchmark. Although it is known that cyclic straining enhances tissue formation, optimal loading protocols have not been defined yet. To obtain a
Traeger, Brad; Srivatsa, Sanjay S.; Beussman, Kevin M.; Wang, Yechun; Suzen, Yildirim B.; Rybicki, Frank J.; Mazur, Wojciech; Miszalski-Jamka, Tomasz
Aortic stenosis is the most common valvular heart disease. Assessing the contribution of the valve as a portion to total ventricular load is essential for the aging population. A CT scan for one patient was used to create one in vivo tricuspid aortic valve geometry and assessed with computational fluid dynamics (CFD). CFD simulated the pressure, velocity, and flow rate, which were used to assess the Gorlin formula and continuity equation, current clinical diagnostic standards. The results demonstrate an underestimation of the anatomic orifice area (AOA) by Gorlin formula and overestimation of AOA by the continuity equation, using peak velocities, as would be measured clinically by Doppler echocardiography. As a result, we suggest that the Gorlin formula is unable to achieve the intended estimation of AOA and largely underestimates AOA at the critical low-flow states present in heart failure. The disparity in the use of echocardiography with the continuity equation is due to the variation in velocity profile between the outflow tract and the valve orifice. Comparison of time-averaged orifice areas by Gorlin and continuity with instantaneous orifice areas by planimetry can mask the errors of these methods, which is a result of the assumption that the blood flow is inviscid.
Lopez, A.; Ledesma, R.; Zenit, R.; Pulos, G.
The hemodynamic properties of prosthetic heart valves can cause blood damage and platelet activation due to the non- physiological flow patterns. Blood recirculation and elevated shear stresses are believed to be responsible for these complications. The objective of this study is to identify and quantify the conditions for which recirculation and high stress zones appear. We have performed a comparative study between a mechanical monoleaflet and biological valve. In order to generate the flow conditions to test the prosthesis, we have built a hydraulic circuit which reproduces the human systemic circulation, on the basis of the Windkessel model. This model is based on an electrical analogy which consists of an arterial resistance and compliance. Using PIV 3D- Stereo measurements, taken downstream from the prosthetic heart valves, we have reconstructed the full phase-averaged tridimensional velocity field. Preliminary results show that critical zones are more prominent in mechanical prosthesis, indicating that valves made with bio-materials are less likely to produce blood trauma. This is in accordance with what is generally found in the literature.
Raghav, Vrishank; Sastry, Sudeep; Saikrishnan, Neelakantan
Experimental flow field characterization is a critical component of the assessment of the hemolytic and thrombogenic potential of heart valve substitutes, thus it is important to identify best practices for these experimental techniques. This paper presents a brief review of commonly used flow assessment techniques such as Particle image velocimetry (PIV), Laser doppler velocimetry, and Phase contrast magnetic resonance imaging and a comparison of these methodologies. In particular, recommendations for setting up planar PIV experiments such as recommended imaging instrumentation, acquisition and data processing are discussed in the context of heart valve flows. Multiple metrics such as residence time, local velocity and shear stress that have been identified in the literature as being relevant to hemolysis and thrombosis in heart valves are discussed. Additionally, a framework for uncertainty analysis and data reporting for PIV studies of heart valves is presented in this paper. It is anticipated that this paper will provide useful information for heart valve device manufacturers and researchers to assess heart valve flow fields for the potential for hemolysis and thrombosis.
Akhtar, R.P.; Abid, A.R.
To compare the follow-up results of double valve replacement (DVR) i.e. mitral valve replacement (MVR) and aortic valve replacement (AVR) vs. isolated MVR or AVR for rheumatic heart disease. Study Design: An interventional qausi-experimental study. Prospective follow-up of 493 patients with mechanical heart valves was carried out using clinical assessment, international normalized ratio and echocardiography. Patients were divided into three groups: group I having MVR, group II having AVR and group III having DVR. Survival, time and causes of mortality, and frequency of valve thrombosis, haemorrhage and cerebrovascular haemorrhage was noted in the three groups and described as proportions. Actuarial survival was analyzed by Kaplan-Meier method. There were 493 with 287 (58.3%) in group I, 87 (17.6%) in group II and 119 (24.1%) in group III. Total follow-up was 2429.2 patient (pt)-years. Of 77 (15.6%) deaths, 19 (3.8%) were in-hospital and 58 (11.8%) were late. In-hospital mortality was highest 4 (4.6%) in group II followed by 5 (4.2%) group III and 10 (3.5%) group I. Late deaths were 39 (13.4%) in group I, 9 (10.2%) in group II and 10 (8.3%) in group III. The total actuarial survival was 84.4% with survival of 83%, 85.1%, 87.4% in groups I, II and III respectively. On follow-up valve thrombosis occurred in 12 (0.49%/pt-years) patients; 9 (0.67%/pt-years) group I, 1 (0.22%/pt-years) in group II and 2 (0.31%/pt-years) in group III. Severe haemorrhage occurred in 19 (0.78%/pt-years); 14 in (1.04%/pt-years) in group I, 3 (0.66%/pt-years) group II and 2 (0.31%/pt-years) in group III. Cerebrovascular accidents occurred in 34 (1.3%/pt-years); 26 (1.95%/pt-years) in group I and 4 in groups II (0.89%/pt-years) and III (0.62%/pt-years) each. In patients with rheumatic heart disease having combined mitral and aortic valve disease DVR should be performed whenever indicated as it has similar in-hospital mortality and better late survival as compared to isolated aortic or mitral
Habets, Jesse; Budde, Ricardo P.; Symersky, Petr; van den Brink, Renee B.; de Mol, Bas A.; Mali, Willem P.; van Herwerden, Lex A.; Chamuleau, Steven A.
Prosthetic heart valve (PHV) dysfunction is a rare, but potentially life-threatening, complication. In clinical practice, PHV dysfunction poses a diagnostic dilemma. Echocardiography and fluoroscopy are the imaging techniques of choice and are routinely used in daily practice. However, these
Habets, Jesse; Budde, Ricardo P.; Symersky, Petr; van den Brink, Renee B.; de Mol, Bas A.; Mali, Willem P.; van Herwerden, Lex A.; Chamuleau, Steven A.
Prosthetic heart valve (PHV) dysfunction is a rare, but potentially life-threatening, complication. In clinical practice, PHV dysfunction poses a diagnostic dilemma. Echocardiography and fluoroscopy are the imaging techniques of choice and are routinely used in daily practice. However, these
Bachmann, R.; Juengehuelsing, M.; Schicha, H.; Deutsch, H.J.; Sechtem, U.; Hilger, H.H.
Artifical valve prostheses are often regarded as a contraindication for magnetic resonance imaging (MRI), although preliminary in vitro studies suggested, that patients with these metallic implants might safely undergo MR examination. This study reports on the experience with a group of 89 patients with 100 heart valve prostheses who were examined by spin-echo MR and gradient-echo MR. MR examination was performed in all patients without complications. The spin-echo sequence showed advantages in the depiction of anatomical structures like paravalvular abcesses. Anatomical structures adjacent to the artificial valve were clearly visivle and the metal components of the valves showes no or only small artifacts. Artifacts were accentuated when using gradient-echo sequences. Gradient-echo sequences provided valuable information regarding the presence of valvular insufficiency. Physiological valvular regurgitation was easy to differentiate from pathological paravalvular or transvalvular regurgitation. These results demonstrate that patients with artificial valve prostheses can be imaged by MR without risk and that prosthesis-induced artifacts do no interfere with image interpretation. (orig.) [de
Kuntze, CEE; Blackstone, EH; Ebels, T
Background. This study was designed to revise and substantiate previous inferences, based on short-term follow-up, about differences in the incidence of anticoagulant-related events after heart valve replacement among patients who had been randomly assigned to receive either a Bjork-Shiley,
Dobrova, N B; Zaretskiĭ, Iu V
Bench tests for 38 models of artificial heart valves (AHV) with different design parameters allowed us to decide in favour of the valves with reduced eccentricity (compared to the serial AHV of the EMAHV type) according to its resistance in the constant flow. Out of the compatibility checks of the design parameters tested it was concluded that the disk did not make the complete calculated angle when rotated. The dependence of AHV resistance on the disk rotation angle showed that there is no necessity to increase that angle more than 70 degrees for the mitral valve and more than 75 degrees for the aortic AHV.
Akhtar, Raja Parvez; Abid, Abdul Rehman; Zafar, Hasnain; Gardezi, Syed Javed Raza; Waheed, Abdul; Khan, Jawad Sajid
To study patients with mechanical heart valves undergoing non-cardiac surgery and their anticoagulation management during these procedures. It was a cohort study. The study was conducted at the Department of Cardiac Surgery, Punjab Institute of Cardiology, Lahore and Department of Surgery, Services Institute of Medical Sciences, Lahore, from September 1994 to June 2006. Patients with mechanical heart valves undergoing non-cardiac surgical operation during this period, were included. Their anticoagulation was monitored and anticoagulation related complications were recorded. In this study, 507 consecutive patients with a mechanical heart valve replacement were followed-up. Forty two (8.28%) patients underwent non-cardiac surgical operations of which 24 (57.1%) were for abdominal and non-abdominal surgeries, 5 (20.8%) were emergency and 19 (79.2%) were planned. There were 18 (42.9%) caesarean sections for pregnancies. Among the 24 procedures, there were 7(29.1%) laparotomies, 7(29.1%) hernia repairs, 2 (8.3%) cholecystectomies, 2 (8.3%) hysterectomies, 1(4.1%) craniotomy, 1(4.1%) spinal surgery for neuroblastoma, 1(4.1%) ankle fracture and 1(4.1%) carbuncle. No untoward valve or anticoagulation related complication was seen during this period. Patients with mechanical valve prosthesis on life-long anticoagulation, if managed properly, can undergo any type of non-cardiac surgical operation with minimal risk.
Ryan S. Hennessy, MD
Full Text Available Summary: Sterilization of grafts is essential. Supercritical carbon dioxide, electrolyzed water, gamma radiation, ethanol-peracetic acid, and hydrogen peroxide techniques were compared for impact on sterility and mechanical integrity of porcine decellularized aortic valves. Ethanol-peracetic acidâ and supercritical carbon dioxideâtreated valves were found to be sterile using histology, microbe culture, and electron microscopy assays. The cusp tensile properties of supercritical carbon dioxideâtreated valves were higher compared with valves treated with other techniques. Superior sterility and integrity was found in the decellularized valves treated with supercritical carbon dioxide sterilization. This sterilization technique may hold promise for other decellularized soft tissues. Key Words: decellularized, decontamination, heart valve, tensile properties, tissue engineering
Moreno, Sandra; Parra, Beatriz; Botero, Javier E; Moreno, Freddy; Vásquez, Daniel; Fernández, Hugo; Alba, Sandra; Gallego, Sara; Castillo, Gilberto; Contreras, Adolfo
Periodontitis is an infectious disease that affects the support tissue of the teeth and it is associated with different systemic diseases, including cardiovascular disease. Microbiological studies facilitate the detection of microorganisms from subgingival and cardiovascular samples. To describe the cultivable periodontal microbiota and the presence of microorganisms in heart valves from patients undergoing valve replacement surgery in a clinic in Cali. We analyzed 30 subgingival and valvular tissue samples by means of two-phase culture medium, supplemented blood agar and trypticase soy agar with antibiotics. Conventional PCR was performed on samples of valve tissue. The periodontal pathogens isolated from periodontal pockets were: Fusobacterium nucleatum (50%), Prevotella intermedia/ nigrescens (40%), Campylobacter rectus (40%), Eikenella corrodens (36.7%), Gram negative enteric bacilli (36.7%), Porphyromonas gingivalis (33.3%), and Eubacterium spp. (33.3%). The pathogens isolated from the aortic valve were Propionibacterium acnes (12%), Gram negative enteric bacilli (8%), Bacteroides merdae (4%), and Clostridium bifermentans (4%), and from the mitral valve we isolated P. acnes and Clostridium beijerinckii. Conventional PCR did not return positive results for oral pathogens and bacterial DNA was detected only in two samples. Periodontal microbiota of patients undergoing surgery for heart valve replacement consisted of species of Gram-negative bacteria that have been associated with infections in extraoral tissues. However, there is no evidence of the presence of periodontal pathogens in valve tissue, because even though there were valve and subgingival samples positive for Gram-negative enteric bacilli, it is not possible to maintain they corresponded to the same phylogenetic origin.
Hasler, D.; Landolt, A.; Obrist, D.
The instantaneous three-dimensional velocity field past a bioprosthetic heart valve was measured using tomographic particle image velocimetry. Two digital cameras were used together with a mirror setup to record PIV images from four different angles. Measurements were conducted in a transparent silicone phantom with a simplified geometry of the aortic root. The refraction indices of the silicone phantom and the working fluid were matched to minimize optical distortion from the flow field to the cameras. The silicone phantom of the aorta was integrated in a flow loop driven by a piston pump. Measurements were conducted for steady and pulsatile flow conditions. Results of the instantaneous, ensemble and phase-averaged flow field are presented. The three-dimensional velocity field reveals a flow topology, which can be related to features of the aortic valve prosthesis.
Tremblay, Catherine; Ruel, Jean; Bourget, Jean-Michel; Laterreur, Véronique; Vallières, Karine; Tondreau, Maxime Y; Lacroix, Dan; Germain, Lucie; Auger, François A
Tissue engineering appears as a promising option to create new heart valve substitutes able to overcome the serious drawbacks encountered with mechanical substitutes or tissue valves. The objective of this article is to present the construction method of a new entirely biological stentless aortic valve using the self-assembly method and also a first assessment of its behavior in a bioreactor when exposed to a pulsatile flow. A thick tissue was created by stacking several fibroblast sheets produced with the self-assembly technique. Different sets of custom-made templates were designed to confer to the thick tissue a three-dimensional (3D) shape similar to that of a native aortic valve. The construction of the valve was divided in two sequential steps. The first step was the installation of the thick tissue in a flat preshaping template followed by a 4-week maturation period. The second step was the actual cylindrical 3D forming of the valve. The microscopic tissue structure was assessed using histological cross sections stained with Masson's Trichrome and Picrosirius Red. The thick tissue remained uniformly populated with cells throughout the construction steps and the dense extracellular matrix presented corrugated fibers of collagen. This first prototype of tissue-engineered heart valve was installed in a bioreactor to assess its capacity to sustain a light pulsatile flow at a frequency of 0.5 Hz. Under the light pulsed flow, it was observed that the leaflets opened and closed according to the flow variations. This study demonstrates that the self-assembly method is a viable option for the construction of complex 3D shapes, such as heart valves, with an entirely biological material.
von Segesser, L K; Stauffer, J C; Delabays, A; Chassot, P G
Tricuspid regurgitation is relatively common. Due to the progress made in echocardiography, its diagnosis is in general made readily and in reliable fashion. Basically one has to distinguish between functional tricuspid valve regurgitation due to volume and/or pressure overload of the right ventricle with intact valve structures versus tricuspid valve regurgitation due to pathologic valve structures. The clear identification of the regurgitation mechanism is of prime importance for the treatment. Functional tricuspid valve regurgitation can often be improved by medical treatment of heart failure, and eventually a tricuspid valve plasty can solve the problem. However, the presence of pathologic tricuspid valve structures makes in general more specific plastic surgical procedures and even prosthetic valve replacements necessary. A typical example for a structural tricuspid valve regurgitation is the case of a traumatic papillary muscle rupture. Due to the sudden onset, this pathology is not well tolerated and requires in general surgical reinsertion of the papillary muscle. In contrast, tricuspid valve regurgitation resulting from chronic pulmonary embolism with pulmonary artery hypertension, can be improved by pulmonary artery thrombendarteriectomy and even completely cured with an additional tricuspid annuloplasty. However, tricuspid regurgitations due to terminal heart failure are not be addressed with surgery directed to tricuspid valve repair or replacement. Heart transplantation, dynamic cardiomyoplasty or mechanical circulatory support should be evaluated instead.
Shafique, H.; Chaudhry, A.; Ayyub, M.
To evaluate the complications and outcome of anticoagulation therapy in pregnant females with valvular heart diseases. All pregnant females with prosthetic heart valves admitted in Armed Forces Institute of Cardiology from Jan 2004 to Dec 2004 were included in this study Basic demographic data including age, duration of pregnancy and complications observed were recorded. Warfarin was replaced with un-fractionated heparin (UFH) in first trimester and after that warfarin was continued with a targeted INR between 2.0-3.0. At 36 weeks warfarin was stopped and UFH was added; however, if patient went into spontaneous labour before this then immediate caesarian section was performed and UFH was restarted 4-6 hours after delivery along with oral warfarin. Out of 21 patients, sixteen (76.1%) had mitral valve diseases and five (23.9%) had both mitral and atrial. Majority (42.3%)of patients were in age group 26-30 years. Eleven (52.2%) reported in 9th month of gestation. Complications observed were hypertension (1), transient ischaemic attacks (1), pulmonary embolism (1), haemoptysis (1) and abortion (1). All patients, except one had successful completion of pregnancy. No case of foetal abnormality was seen. In 76% patients, daily dose of warfarin was <5 mg. Thrombo-prophylaxis in pregnancy with warfarin and UFH with an INR of 2.0-3.0 is effective in preventing thrombotic complications in females with mechanical valves without resulting in increase hemorrhagic complications. (author)
Full Text Available Introduction: Rheumatic heart disease still remains one of the leading causes of congestive heart failure and death owing to valvular pathologies, in developing countries. Valve replacement still remains the treatment of choice in such patients.The aim of this study wasto analyze the postoperative outcome of double valve replacement (Mitral and Aortic in patients of rheumatic heart disease. Materials and Methods: Between 1988 and 2008, 300 patients of rheumatic heart disease underwent double (Mitral and Aortic valve replacement with Starr Edwards valve or St Jude mechanical valve prosthesis were implanted. These patients were studied retrospectively for preoperative data and postoperative outcome including causes of early and late deaths and the data was analyzed statistically. Results: The 30-day hospital death rate was 11.3% andlate death occurred in 11.6%. Anticoagulant regimen was followed to maintain the target pro-thrombin time at 1.5 times the control value. The actuarial survival (exclusive of hospital mortality was 92.4%, 84.6%, and 84.4%, per year at 5, 10, and 20 years, respectively Conclusions: In view of the acknowledged advantageof superior durability, increased thromboresistance in our patient population, and its cost effectiveness the Starr-Edwards ball valve or St. Jude valve is the mechanical prosthesis of choice for advanced combined valvular disease. The low-intensity anticoagulant regimen has offered suffcient protection against thromboembolism as well as hemorrhage.
LAURENS, RRP; WIT, HP; EBELS, T
In a randomised study, we investigated the sound production of mechanical heart valve prostheses and the complaints related to this sound. The CarboMedics, Bjork-Shiley monostrut and StJude Medical prostheses were compared. A-weighted levels of the pulse-like sound produced by the prosthesis were
A human pericardium biopolymeric scaffold for autologous heart valve tissue engineering: cellular and extracellular matrix structure and biomechanical properties in comparison with a normal aortic heart valve.
Straka, Frantisek; Schornik, David; Masin, Jaroslav; Filova, Elena; Mirejovsky, Tomas; Burdikova, Zuzana; Svindrych, Zdenek; Chlup, Hynek; Horny, Lukas; Daniel, Matej; Machac, Jiri; Skibová, Jelena; Pirk, Jan; Bacakova, Lucie
The objective of our study was to compare the cellular and extracellular matrix (ECM) structure and the biomechanical properties of human pericardium (HP) with the normal human aortic heart valve (NAV). HP tissues (from 12 patients) and NAV samples (from 5 patients) were harvested during heart surgery. The main cells in HP were pericardial interstitial cells, which are fibroblast-like cells of mesenchymal origin similar to the valvular interstitial cells in NAV tissue. The ECM of HP had a statistically significantly (p structures of the two tissues, the dense part of fibrous HP (49 ± 2%) and the lamina fibrosa of NAV (47 ± 4%), was similar. In both tissues, the secant elastic modulus (Es) was significantly lower in the transversal direction (p structure and has the biomechanical properties required for a tissue from which an autologous heart valve replacement may be constructed.
Vignesh Vellayappan, Muthu; Balaji, Arunpandian; Priyadarshini Subramanian, Aruna; Aruna John, Agnes; Jaganathan, Saravana Kumar; Murugesan, Selvakumar; Mohandas, Hemanth; Supriyanto, Eko; Yusof, Mustafa
Cardiovascular disease claims millions of lives every year throughout the world. Biomaterials are used widely for the treatment of this fatal disease. With the advent of nanotechnology, the use of nanocomposites has become almost inevitable in the field of biomaterials. The versatile properties of nanocomposites, such as improved durability and biocompatibility, make them an ideal choice for various biomedical applications. Among the various nanocomposites, polyhedral oligomeric silsesquioxane-poly(carbonate-urea)urethane, bacterial cellulose with polyvinyl alcohol, carbon nanotubes, graphene oxide and nano-hydroxyapatite nanocomposites have gained popularity as putative choices for biomaterials in cardiovascular applications owing to their superior properties. In this review, various studies performed utilizing these nanocomposites for improving the mechanical strength, anti-calcification potential and hemocompatibility of heart valves are reviewed and summarized. The primary motive of this work is to shed light on the emerging nanocomposites for heart valve applications. Furthermore, we aim to promote the prospects of these nanocomposites in the campaign against cardiovascular diseases.
Vellayappan, Muthu Vignesh; Balaji, Arunpandian; Subramanian, Aruna Priyadarshini; John, Agnes Aruna; Jaganathan, Saravana Kumar; Supriyanto, Eko; Yusof, Mustafa; Murugesan, Selvakumar; Mohandas, Hemanth
Cardiovascular disease claims millions of lives every year throughout the world. Biomaterials are used widely for the treatment of this fatal disease. With the advent of nanotechnology, the use of nanocomposites has become almost inevitable in the field of biomaterials. The versatile properties of nanocomposites, such as improved durability and biocompatibility, make them an ideal choice for various biomedical applications. Among the various nanocomposites, polyhedral oligomeric silsesquioxane-poly(carbonate-urea)urethane, bacterial cellulose with polyvinyl alcohol, carbon nanotubes, graphene oxide and nano-hydroxyapatite nanocomposites have gained popularity as putative choices for biomaterials in cardiovascular applications owing to their superior properties. In this review, various studies performed utilizing these nanocomposites for improving the mechanical strength, anti-calcification potential and hemocompatibility of heart valves are reviewed and summarized. The primary motive of this work is to shed light on the emerging nanocomposites for heart valve applications. Furthermore, we aim to promote the prospects of these nanocomposites in the campaign against cardiovascular diseases. (review)
Harpa Marius Mihai
Full Text Available Background: We hypothesized that an ideal heart valve replacement would be acellular valve root scaffolds seeded with autologous stem cells. To test this hypothesis, we prepared porcine acellular pulmonary valves, seeded them with autologous adipose derived stem cells (ADSCs and implanted them in sheep and compared them to acellular valves.
Goffin, Y; Grandmougin, D; Van Hoeck, B
The heart valve bank of the European Homograft Bank has been set up in 1988 to meet the growing demand of cardiac surgeons for various sized and quality controlled cryopreserved homografts. Heart valve donors less than 60 years of age were classified in 3 categories: multiorgan donors with non transplantable hearts, recipients of cardiac transplantation and non beating heart cadavers with a warm ischemic time of less than 6 hours. Past history and biology were checked for transmissible diseases. Preparation, progressive freezing and storage in liquid nitrogen vapors, and quality control were according to the standards of the Belgian Ministry of Health. From end January 1989 to end May 1994, 989 homograft valves were cryopreserved (514 pulmonary, 475 aortic and 3 mitral) whereas 962 valves were discarded. The first cause of rejection being a major macroscopic lesion (41.48%). 138 hearts accepted at inspection were contaminated and 43 cases remained so after antibiotics. 38 cases were positive for hepatitis B or C. Complication at distribution and thawing included 10 instances of bag rupture and 15 of transversal fracture through the wall of the conduit. 477 aortic, 474 pulmonary valves as well as one mitral were implanted between May 1989 and May 1994, either for left or right ventricular outflow tract reconstruction. In the left ventricular outflow tract series 111 aortic and 23 pulmonary homograft valves were used in cases of native endocarditis, prosthetic endocarditis or recurrent endocarditis after homograft implantation. 9.6% of the requests could no be satisfied. Regular follow up information was available from 382 implants-40.1% only. The assessment of 5 years operation of the heart valve bank indicates: 1) the efficiency of selecting, cryopreserving and allocating quality controlled homograft valves from a large pool of donor hearts provided by a network of hospitals; 2) the difficulty of obtaining regular follow up information on the implants.
... most cases, doctors don't know why a heart valve fails to develop properly, so it isn't something you could have prevented. Calcium buildup on the valve. With age, heart valves may accumulate deposits of calcium (aortic valve ...
Stamm, Christof; Pasic, Miralem; Buz, Semih; Hetzer, Roland
A patient required emergency mitral valve replacement and extracorporeal membrane oxygenation (ECMO) support for acute biventricular failure. The left ventricular (LV) vent inserted via the left upper pulmonary vein induced thrombotic immobilization of a prosthetic valve leaflet, with significant intra-prosthesis regurgitation after ECMO explantation. Therefore, the left atrium was opened on the beating heart during conventional extracorporeal circulation, all prosthesis leaflets were excised and a 29-mm expandable Edwards Sapien prosthesis was inserted within the scaffold of the original prosthesis under direct vision. This case illustrates the benefits and potential problems of LV venting on ECMO support, and a rapid and safe way of replacing the prosthesis leaflets in a critical situation. © The Author 2015. Published by Oxford University Press on behalf of the European Association for Cardio-Thoracic Surgery. All rights reserved.
Ye Xia; Shao Yunliang; Zhou Ming; Li Jian; Cai Lan
In order to seek the method to improve the hemo-compatibility of artificial mechanical heart valve, the surface of rabbit's heart valve was observed using the scanning electron microscopy (SEM). The results showed that the dual-scale structure which consists of cobblestones-like structure of 8 μm in underside diameter and 3 μm in height, and the fine cilia of about 150 nm in diameter, was helpful to the hemo-compatibility of the heart valve. Therefore, the polydimethylsiloxane (PDMS) surface with hierarchical micro-structure was fabricated using femtosecond laser fabrication technique and soft lithography. At the same time, the tests of apparent contact angle and platelet adhesion on both smooth and textured PDMS surfaces were carried out to study their wettability and hemo-compatibility. The results demonstrated that the surface with textured structure displayed more excellent wettabililty and anti-coagulation property than that of smooth surface. The apparent contact angle of textured surface enhanced from 113.1 deg. to 163.6 deg. and the amount of adsorbed platelet on such surface was fewer, no distortion and no activation were found.
van den Brink, Renee B. A.
Transesophageal echocardiography (TEE) is especially suitable for examination of prosthetic valves because of the proximity of the esophagus to the heart and absence of interference with lungs and ribs. This article reviews normal and abnormal morphologic characteristics of prosthetic valves such as
... the left ventricle from flowing backward. A defective heart valve fails to either open or close fully. Risk factors Mitral valve stenosis is less common today than it once was because the most common cause, ... other heart valve problems, mitral valve stenosis can strain your ...
Lincoln, Joy; Kist, Ralf; Scherer, Gerd; Yutzey, Katherine E
Heart valve structures derived from mesenchymal cells of the endocardial cushions (ECs) are composed of highly organized cell lineages and extracellular matrix. Sox9 is a transcription factor required for both early and late stages of cartilage formation that is also expressed in the developing valves of the heart. The requirements for Sox9 function during valvulogenesis and adult valve homeostasis in mice were examined by conditional inactivation of Sox9 using Tie2-cre and Col2a1-cre transgenes. Sox9(flox/flox);Tie2-cre mice die before E14.5 with hypoplastic ECs, reduced cell proliferation and altered extracellular matrix protein (ECM) deposition. Sox9(flox/flox);Col2a1-cre mice die at birth with thickened heart valve leaflets, reduced expression of cartilage-associated proteins and abnormal ECM patterning. Thickened valve leaflets and calcium deposits, characteristic of valve disease, are observed in heterozygous adult Sox9(flox/+);Col2a1-cre mice. Therefore, Sox9 is required early in valve development for expansion of the precursor cell population and later is required for normal expression and distribution of valvular ECM proteins. These data indicate that Sox9 is required for early and late stages of valvulogenesis and identify a potential role for Sox9 in valve disease mechanisms.
Albrecht, Helmi; Lim, Yeong Phang; Manning, Linda
A survey of 24 international heart valve banks was conducted to acquire information on heart valve processing techniques used and outcomes achieved. The objective was to provide an overview of heart valve banking activities for tissue bankers, tissue banking associations, and regulatory bodies worldwide. Despite similarities found for basic manufacturing processes, distinct differences in procedural details were also identified. The similarities included (1) use of sterile culture media for procedures, (2) antibiotic decontamination, (3) use of dimethyl sulfoxide (DMSO) as a cryoprotectant, (4) controlled rate freezing for cryopreservation, and (5) storage at ultralow temperatures of below −135°C. Differences in procedures included (1) type of sterile media used, (2) antibiotics combination, (3) temperature and duration used for bioburden reduction, (4) concentration of DMSO used for cryopreservation, and (5) storage duration for released allografts. For most banks, the primary reasons why allografts failed to meet release criteria were positive microbiological culture and abnormal morphology. On average, 85% of allografts meeting release criteria were implanted, with valve size and type being the main reasons why released allografts were not used clinically. The wide variation in percentage of allografts meeting release requirements, despite undergoing validated manufacturing procedures, justifies the need for regular review of important outcomes as cited in this paper, in order to encourage comparison and improvements in the HVBs' processes. PMID:24163756
Louw, J W; Kinsley, R H; Dion, R A; Colsen, P R; Girdwood, R W
The results of 170 emergency heart valve procedures performed during a 4 1/2-year period were analyzed. Five pathological groups of patients were recognized: those with infective endocarditis (Group 1, 28 patients); acute rheumatic carditis (Group 2, 43 patients); previous valve operation (Group 3, 29 patients); acute-on-chronic cardiac disease (Group 4, 67 patients); and miscellaneous conditions (Group 5, 3 patients). Mitral, aortic, and multiple valve procedures were performed on 58, 65, and 44 patients, respectively. The most common functional lesion was regurgitation. Hospital mortality was highest in Groups 3 (34%) and 4 (31%). By contrast, among the hospital survivors, the highest rate of attrition was in Group 2. Myocardial failure was the predominat cause of death. In view of the hopeless prognosis without operation, the 52% overall 3-year actuarial survival is a gratifying salvage. Unnecessary procrastination can only jeopardize the prospects for surgical cure.
Morse, D.; Steiner, R.M.; Fernandez, J.
This book contains 10 chapters. Some of the chapter titles are: The development of artificial heart valves: Introduction and historical perspective; The radiology of prosthetic heart valves; The evaluation of patients for prosthetic valve implantation; Pathology of cardiac valve replacement; and Bioengineering of mechanical and biological heart valve substitutes
Valvular heart disease is accompanied by a high mortality/morbidity and often requires prosthetic heart valve (PHV) replacement in order to improve quality of life and survival. The major drawback of both mechanical and biological PHV implantations is development of dysfunction, which is a life
Full Text Available Background: Although atrial fibrillation (AF, a common arrhythmia, frequently presents in patients with underlying valvular disease, its hemodynamic contributions are not fully understood. The present work aimed to computationally study how physical conditions imposed by pathologic valvular anatomy act on AF hemodynamics. Methods: We simulated AF with different severity grades of left-sided valvular diseases and compared the cardiovascular effects that they exert during AF, compared to lone AF. The fluid dynamics model used here has been recently validated for lone AF and relies on a lumped parameterization of the four heart chambers, together with the systemic and pulmonary circulation. The AF modelling involves: (i irregular, uncorrelated and faster heart rate; (ii atrial contractility dysfunction. Three different grades of severity (mild, moderate, severe were analyzed for each of the four valvulopathies (AS, aortic stenosis, MS, mitral stenosis, AR, aortic regurgitation, MR, mitral regurgitation, by varying–through the valve opening angle–the valve area. Results: Regurgitation was hemodynamically more relevant than stenosis, as the latter led to inefficient cardiac flow, while the former introduced more drastic fluid dynamics variation. Moreover, mitral valvulopathies were more significant than aortic ones. In case of aortic valve diseases, proper mitral functioning damps out changes at atrial and pulmonary levels. In the case of mitral valvulopathy, the mitral valve lost its regulating capability, thus hemodynamic variations almost equally affected regions upstream and downstream of the valve. In particular, the present study revealed that both mitral and aortic regurgitation strongly affect hemodynamics, followed by mitral stenosis, while aortic stenosis has the least impact among the analyzed valvular diseases. Discussion: The proposed approach can provide new mechanistic insights as to which valvular pathologies merit more aggressive
Fuchs, Andreas; De Backer, Ole; Brooks, Matthew
AIMS: This study aimed to assess the potential relationship between subclinical leaflet thickening and stent frame geometry in patients who underwent aortic valve replacement with a self-expanding transcatheter heart valve (THV). METHODS AND RESULTS: Seventy-five patients with a self-expanding THV....... CONCLUSIONS: Regional THV stent frame underexpansion is associated with an increased risk of leaflet thickening. Post-dilatation of self-expanding THV as well as a supra-annular valve position seem to reduce the occurrence of this phenomenon....
Schaefer, Andreas; Treede, Hendrik; Seiffert, Moritz; Deuschl, Florian; Schofer, Niklas; Schneeberger, Yvonne; Blankenberg, Stefan; Reichenspurner, Hermann; Schaefer, Ulrich; Conradi, Lenard
Paravalvular leakage (PVL) is a known complication of transcatheter aortic valve implantation (TAVI) and is associated with poor outcome. Besides balloon-post-dilatation, valve-in-valve (ViV) procedures can be taken into consideration to control this complication. Herein we present initial experience with use of the latest generation balloon-expandable Edwards Sapien 3® (S3) transcatheter heart valve (THV) for treatment of failing THVs. Between 01/2014 and 12/2014 three patients (two male, age: 71-80 y, log EUROScore I: 11.89 - 32.63) with failing THVs were refered to our institution for further treatment. THV approach with secondary implantation of an S3 was chosen after mutual agreement of the local interdisciplinary heart team at an interval of 533-1119 days from the index procedure. The performed procedures consisted of: S3 in Sapien XT, JenaValve and CoreValve. Successful transfemoral implantation with significant reduction of PVL was achieved in all cases. No intraprocedural complications occurred regarding placement of the S3 with a postprocedural effective orifice area (EOA) of 1.5-2.5 cm(2) and pressure gradients of max/mean 14/6-36/16 mmHg. 30-day mortality was 0%. At the latest follow-up of 90-530 days, all patients are alive and well with satisfactory THV function. Regarding VARC-2 criteria one major bleeding and one TIA was reported. In the instance of moderate or severe aortic regurgitation after TAVI, S3 ViV deployment is an excellent option to reduce residual regurgitation to none or mild. For further assertions concerning functional outcomes long-term results have to be awaited.
Umakanthan, Ramanan; Leacche, Marzia; Petracek, Michael R; Zhao, David X; Byrne, John G
Combined coronary artery valvular heart disease is a major cause of morbidity and mortality in the adult patient population. The standard treatment for such disease has been open heart surgery in which coronary artery bypass grafting (CABG) is performed concurrently with valve surgery using a median sternotomy and cardiopulmonary bypass. With the increasing complexity of patients referred to surgery, some patients may prove to be poor surgical candidates for combined valve and CABG surgery. In certain selected patients who fall into this category, valve surgery and percutaneous coronary intervention (PCI) have been considered a feasible alternative. Conventionally, valve surgery is performed in the cardiac surgical operating room, whereas PCI is carried out in the cardiac catheterization laboratory. Separation of these two procedural suites has presented a logistic limitation because it impedes the concomitant performance of both procedures in one setting. Hence, PCI and valve surgery usually have been performed as a "two-stage" procedure in two different operative suites, with the procedures being separated by hours, days, or weeks. Technologic advancements have made possible the construction of a "hybrid" procedural suite that combines the facilities of a cardiac surgical operating room with those of a cardiac catheterization laboratory. This design has enabled the concept of "one-stage" or "one-stop" PCI and valve surgery, allowing both procedures to be performed in a hybrid suite in one setting, separated by minutes. The advantages of such a method could prove to be multifold by enabling a less invasive surgical approach and improving logistics, patient satisfaction, and outcomes in selected patients.
Rich, S; Wix, H L; Shapiro, E P
It has been generally accepted that enhanced blood flow during closed-chest CPR is generated from compression of the heart between the sternum and the spine. To visualize the heart during closed-chest massage, we performed two-dimensional echocardiography (2DE) during resuscitation efforts in four patients who had cardiac arrest. 2DE analysis showed that (1) the LV internal dimensions did not change appreciably with chest compression; (2) the mitral and aortic valves were open simultaneously during the compression phase; (3) blood flow into the right heart, as evidenced by saline bubble contrast, occurred during the relaxation phase; and (4) compression of the right ventricle and LA occurred in varying amounts in all patients. We conclude that stroke volume from the heart during CPR does not result from compression of the LV. Rather, CPR-induced improved cardiocirculatory dynamics appear to be principally the result of changes in intrathoracic pressure created by sternal compression.
Zhang, Xing; Xu, Bin; Puperi, Daniel S; Yonezawa, Aline L; Wu, Yan; Tseng, Hubert; Cuchiara, Maude L; West, Jennifer L; Grande-Allen, K Jane
The development of advanced scaffolds that recapitulate the anisotropic mechanical behavior and biological functions of the extracellular matrix in leaflets would be transformative for heart valve tissue engineering. In this study, anisotropic mechanical properties were established in poly(ethylene glycol) (PEG) hydrogels by crosslinking stripes of 3.4 kDa PEG diacrylate (PEGDA) within 20 kDa PEGDA base hydrogels using a photolithographic patterning method. Varying the stripe width and spacing resulted in a tensile elastic modulus parallel to the stripes that was 4.1-6.8 times greater than that in the perpendicular direction, comparable to the degree of anisotropy between the circumferential and radial orientations in native valve leaflets. Biomimetic PEG-peptide hydrogels were prepared by tethering the cell-adhesive peptide RGDS and incorporating the collagenase-degradable peptide PQ (GGGPQG↓IWGQGK) into the polymer network. The specific amounts of RGDS and PEG-PQ within the resulting hydrogels influenced the elongation, de novo extracellular matrix deposition and hydrogel degradation behavior of encapsulated valvular interstitial cells (VICs). In addition, the morphology and activation of VICs grown atop PEG hydrogels could be modulated by controlling the concentration or micro-patterning profile of PEG-RGDS. These results are promising for the fabrication of PEG-based hydrogels using anatomically and biologically inspired scaffold design features for heart valve tissue engineering. Copyright © 2014 Acta Materialia Inc. Published by Elsevier Ltd. All rights reserved.
E-R. Andrinopoulou (Eleni-Rosalina)
markdownabstract__Abstract__ The heart is one of the most important organs in the entire human body. Specifically, it is a pump composed of muscle which pumps blood throughout the blood vessels to various parts of the body by repeated rhythmic contractions. The four heart valves determine the
Kaminsky, R; Morbiducci, U; Rossi, M; Scalise, L; Verdonck, P; Grigioni, M
Prosthetic heart valves (PHVs) have been used to replace diseased native valves for more than five decades. Among these, mechanical PHVs are the most frequently implanted. Unfortunately, these devices still do not achieve ideal behavior and lead to many complications, many of which are related to fluid mechanics. The fluid dynamics of mechanical PHVs are particularly complex and the fine-scale characteristics of such flows call for very accurate experimental techniques. Adequate temporal resolution can be reached by applying time-resolved PIV, a high-resolution dynamic technique which is able to capture detailed chronological changes in the velocity field. The aim of this experimental study is to investigate the evolution of the flow field in a detailed time domain of a commercial bileaflet PHV in a mock-loop mimicking unsteady conditions, by means of time-resolved 2D Particle Image Velocimetry (PIV). The investigated flow field corresponded to the region immediately downstream of the valve plane. Spatial resolution as in "standard" PIV analysis of prosthetic valve fluid dynamics was used. The combination of a Nd:YLF high-repetition-rate double-cavity laser with a high frame rate CMOS camera allowed a detailed, highly temporally resolved acquisition (up to 10000 fps depending on the resolution) of the flow downstream of the PHV. Features that were observed include the non-homogeneity and unsteadiness of the phenomenon and the presence of large-scale vortices within the field, especially in the wake of the valve leaflets. Furthermore, we observed that highly temporally cycle-resolved analysis allowed the different behaviors exhibited by the bileaflet valve at closure to be captured in different acquired cardiac cycles. By accurately capturing hemodynamically relevant time scales of motion, time-resolved PIV characterization can realistically be expected to help designers in improving PHV performance and in furnishing comprehensive validation with experimental data
Topic of this thesis is " Mitral Valvuloplasty". The thesis has been divided into theoretical and practical parts. The theoretical part begins with the classification of heart diseases. Main topic of this thesis is mitral insufficiency - its etiology and pathogenesis, clinical picture, therapy, the possibility of prosthetic valves and post surgery complications. Theoretical part also contains information about the preoperative and post-operative care at cardiac surgery department. The practic...
Lepelaars, E.S.A.M.; Ooijen, van W.D.R.; Tijhuis, A.G.
Investigates the feasibility of detecting defects in certain artificial heart valves by determining the electromagnetic behavior of some simple models with the aid of thin-wire integral equations. The idea is to use the stationary current that occurs at late times after the excitation of a closed
Hillebrand, Julia; Hoffmeier, Andreas; Djie Tiong Tjan, Tonny; Sindermann, Juergen R; Schmidt, Christoph; Martens, Sven; Scherer, Mirela
Left ventricular assist device (LVAD) implantation is a well-established therapy to support patients with end-stage heart failure. However, the operative procedure is associated with severe trauma. Third generation LVADs like the HeartWare assist device (HeartWare, Inc., Framingham, MA, USA) are characterized by enhanced technology despite smaller size. These devices offer new minimally invasive surgical options. Tricuspid regurgitation requiring valve repair is frequent in patients with the need for mechanical circulatory support as it is strongly associated with ischemic and nonischemic cardiomyopathy. We report on HeartWare LVAD implantation and simultaneous tricuspid valve reconstruction through minimally invasive access by partial upper sternotomy to the fifth left intercostal space. Four male patients (mean age 51.72 ± 11.95 years) suffering from chronic heart failure due to dilative (three patients) and ischemic (one patient) cardiomyopathy and also exhibiting concomitant tricuspid valve insufficiency due to annular dilation underwent VAD implantation and tricuspid valve annuloplasty. Extracorporeal circulation was established via the ascending aorta, superior vena cava, and right atrium. In all four cases the LVAD implantation and tricuspid valve repair via partial median sternotomy was successful. During the operative procedure, no conversion to full sternotomy was necessary. One patient needed postoperative re-exploration because of pericardial effusion. No postoperative focal neurologic injury was observed. New generation VADs are advantageous because of the possibility of minimally invasive implantation procedure which can therefore minimize surgical trauma. Concomitant tricuspid valve reconstruction can also be performed simultaneously through partial upper sternotomy. Nevertheless, minimally invasive LVAD implantation is a challenging operative technique. © 2016 International Center for Artificial Organs and Transplantation and Wiley Periodicals
Martins, Carlo de Oliveira; Demarchi, Lea; Ferreira, Frederico Moraes; Pomerantzeff, Pablo Maria Alberto; Brandao, Carlos; Sampaio, Roney Orismar; Spina, Guilherme Sobreira; Kalil, Jorge; Cunha-Neto, Edecio; Guilherme, Luiza
Autoimmune inflammatory reactions leading to rheumatic fever (RF) and rheumatic heart disease (RHD) result from untreated Streptococcus pyogenes throat infections in individuals who exhibit genetic susceptibility. Immune effector mechanisms have been described that lead to heart tissue damage culminating in mitral and aortic valve dysfunctions. In myxomatous valve degeneration (MXD), the mitral valve is also damaged due to non-inflammatory mechanisms. Both diseases are characterized by structural valve disarray and a previous proteomic analysis of them has disclosed a distinct profile of matrix/structural proteins differentially expressed. Given their relevance in organizing valve tissue, we quantitatively evaluated the expression of vimentin, collagen VI, lumican, and vitronectin as well as performed immunohistochemical analysis of their distribution in valve tissue lesions of patients in both diseases. We identified abundant expression of two isoforms of vimentin (45 kDa, 42 kDa) with reduced expression of the full-size protein (54 kDa) in RHD valves. We also found increased vitronectin expression, reduced collagen VI expression and similar lumican expression between RHD and MXD valves. Immunohistochemical analysis indicated disrupted patterns of these proteins in myxomatous degeneration valves and disorganized distribution in rheumatic heart disease valves that correlated with clinical manifestations such as valve regurgitation or stenosis. Confocal microscopy analysis revealed a diverse pattern of distribution of collagen VI and lumican into RHD and MXD valves. Altogether, these results demonstrated distinct patterns of altered valve expression and tissue distribution/organization of structural/matrix proteins that play important pathophysiological roles in both valve diseases.
... working correctly. Most valve replacements involve the aortic Tricuspid valve and mitral valves. The aortic valve separates ... where it shouldn’t. This is called incompetence, insufficiency or regurgitation. • Prolapse — mitral valve flaps don’t ...
... and replacing it with a man-made or biological valve. Biological valves are made from pig, cow, or human ... the valve. Man-made valves last longer than biological valves and usually don’t have to be ...
Theodoridis, Karolina; Müller, Janina; Ramm, Robert; Findeisen, Katja; Andrée, Birgit; Korossis, Sotirios; Haverich, Axel; Hilfiker, Andres
Non-fixed, decellularized allogeneic heart valve scaffolds seem to be the best choice for heart valve replacement, their availability, however, is quite limited. Cryopreservation could prolong their shelf-life, allowing for their ideal match to a recipient. In this study, porcine pulmonary valves were decellularized using detergents, either prior or after cryopreservation, and analyzed. Mechanical integrity was analyzed by uniaxial tensile testing, histoarchitecture by histological staining, and composition by DNA, collagen (hydroxyproline) and GAG (chondroitin sulfate) quantification. Residual sodium dodecyl sulfate (SDS) in the scaffold was quantified by applying a methylene blue activation assay (MBAS). Cryopreserved decellularized scaffolds (DC) and scaffolds that were decellularized after cryopreservation (CD) were compared to fresh valves (F), cryopreserved native valves (C), and decellularized only scaffolds (D). The E-modulus and tensile strength of decellularized (D) tissue showed no significant difference compared to DC and CD. The decellularization resulted in an overall reduction of DNA and GAG, with DC containing the lowest amount of GAGs. The DNA content in the valvular wall of the CD group was higher than in the D and DC groups. CD valves showed slightly more residual SDS than DC valves, which might be harmful to recipient cells. In conclusion, cryopreservation after decellularization was shown to be preferable over cryopreservation before decellularization. However, in vivo testing would be necessary to determine whether these differences are significant in biocompatibility or immunogenicity of the scaffolds. Absence of adverse effects on biomechanical stability of acellular heart valve grafts by cryopreservation, neither before nor after decellularization, allows the identification of best matching patients in a less time pressure dictated process, and therefore to an optimized use of a very limited, but best-suited heart valve prosthesis
Schapkaitz, Elise; Jacobson, Barry Frank; Manga, Pravin; Chitsike, Rufaro Saeed; Benade, Estee; Haas, Sylvia; Buller, Harry R.
The management of pregnant patients with mechanical heart valves remains challenging because there are no large randomised studies to provide guidelines for effective anticoagulant therapy. Both vitamin K antagonists and heparins may be associated with maternal and foetal adverse events. The
Freling, H. G.; van Slooten, Y. J.; van Melle, J. P.; Mulder, B. J. M.; van Dijk, A. P. J.; Hillege, H. L.; Post, M. C.; Sieswerda, G. Tj; Jongbloed, M. R. M.; Willems, T. P.; Pieper, P. G.
Data on long-term complications in adult patients with congenital heart disease (ACHD) and a prosthetic valve are scarce. Moreover, the influence of prosthetic valves on quality of life (QoL) and functional outcome in ACHD patients with prosthetic valves has not been studied. The primary objective
Blot, William J; Signorello, Lisa B; Cohen, Sarah S; Ibrahim, Michel A
Björk-Shiley convexo-concave (BSCC) prosthetic heart valves are believed to have been implanted in over 86,000 patients worldwide. Limited data are available on the prevalence of single leg separations (SLS) of the valves' outlet struts, a potential precursor to complete valve fracture. Data maintained by the manufacturer, including results of examinations for SLS in explanted valves, were merged with available information on the characteristics of the valve. The prevalence of SLS in the examined valves was calculated according to valve angle, size, position, and study. Among 343 examined valves, the overall prevalence of SLS was 8.2%, but this varied significantly by valve size, being three-fold higher among 29+ mm valves than among smaller valves, with statistically non-significantly higher prevalences among mitral than aortic, and among 70 degrees than 60 degrees valves. By applying the size, position and angle-specific SLS prevalences to the worldwide valve distribution, it is estimated that SLS may be present in 6.8% (95% confidence limits 4.1-9.4%) of all BSCC valves. These findings suggest that SLS may affect between 820 and 1,880 of the almost 20,000 BSCC valves among surviving patients worldwide. Such estimates help frame the context for potential patient screenings, should imaging and acoustic techniques to detect SLS become available.
Kovarević, Pavle; Mihajlović, Bogoljub; Velicki, Lazar; Redzek, Aleksandar; Ivanović, Vladimir; Komazec, Nikola
The last decade of the 20th century brought up a significant development in the field of minimally invasive approaches to the valvular heart surgery. Potential benefits of this method are: good esthetic appearance, reduced pain, reduction of postoperative hemorrhage and incidence of surgical site infection, shorter postoperative intensive care units (ICU) period and overall in-hospital period. Partial upper median stemotomy currently presents as a state-of-the art method for minimally invasive surgery of cardiac valves. The aim of this study was to report on initial experience in application of this surgical method in the surgery of mitral and aortic valves. The study was designed and conducted in a prospective manner and included all the patients who underwent minimally invasive cardiac valve surgery through the partial upper median stemotomy during the period November 2008 - August 2009. We analyzed the data on mean age of patients, mean extubation time, mean postoperative drainage, mean duration of hospital stay, as well as on occurance of postoperative complications (postoperative bleeding, surgical site infection and cerebrovascular insult). During the observed period, in the Institute for Cardiovascular Diseases of Vojvodina, Clinic for Cardiovascular Surgery, 17 ministernotomies were performed, with 14 aortic valve replacements (82.35%) and 3 mitral valve replacements (17.65%). Mean age of the patients was 60.78 +/- 12.99 years (64.71% males, 35.29% females). Mean extubation time was 12.53 +/- 8.87 hours with 23.5% of the patients extubated in less than 8 hours. Mean duration of hospital stay was 12.35 +/- 10.17 days (in 29.4% of the patients less than 8 days). Mean postoperative drainage was 547.06 +/- 335.2 mL. Postoperative complications included: bleeding (5.88%) and cerebrovascular insult (5.88%). One patient (5.88%) required conversion to full stemotomy. Partial upper median sternotomy represents the optimal surgical method for the interventions on the
Bae, J. H.; Lee, K. N.; Jeong, W. K.
This study describes the methodology for predicting a required dynamic torque in motor-operated butterfly valves. The results of this methodology have been compared with test data for motor-operated butterfly valves in nuclear power plant. With the close review of test data and torque prediction, it is concluded that the prediction methodology is conservative to predict a required dynamic torque of motor-operated butterfly valves. In addition, the information of correct differential pressure is vital to predict a required dynamic torque of motor-operated butterfly valves
Aggarwal, Sanjeev; Delius, Ralph E; Walters, Henry L; L'Ecuyer, Thomas J
This case report describes a toddler who developed a protein-losing enteropathy (PLE) 4 years after orthotopic heart transplantation (OHT). He was born with a hypoplastic left heart syndrome for which he underwent a successful Norwood procedure, a Hemi-Fontan palliation, and a Fontan palliation at 18 months of age. Fifteen months following the Fontan operation, he developed a PLE and Fontan failure requiring OHT. Four years after OHT, he developed a severe tricuspid regurgitation and a PLE. His PLE improved after tricuspid valve replacement. It is now 2 years since his tricuspid valve replacement and he remains clinically free of ascites and peripheral edema with a normal serum albumin level. His prosthetic tricuspid valve is functioning normally. © 2011 Wiley Periodicals, Inc.
Full Text Available In order to study the influence that dynamic performance of shift control valve has on shifting process of construction machinery, the paper introduces working principle of the shift control valve and sets up the dynamically mathematical model and corresponding simulation model with simulation software LMS Imagine. Lab AMESim. Based on simulation, the paper analyzes the influence of pressure variation characteristics and buffering characteristics acting on vehicle performance during the process of shifting, meanwhile conducting experiments to verify the simulation. The results indicate that the simulation model is accurate and credible; the performance of the valve is satisfactory, which indeed reduces impact during shifting. Furthermore, the valve can meet the demand of other construction machineries in better degree by suitable matching between control spring stiffness and damping holes diameter.
Khalil Ullah; Badsha, S.; Khan, A.; Kiani, M.R.; Ahmed, S.A.
Objective: To examine the prevalence, age, sex and topographical distribution of the rheumatic heart diseases and its morphology. Design: A cross sectional descriptive study. Place and Duration of Study: Pathology Department, Army Medical College, Rawalpindi between 1981-1990. Patients and Methods: Five hundred and twenty six surgically excised cardiac valves and biopsies were studied in the laboratory in the light of clinical data. Results: Carditis constituted 87.4 % of the cardiac valvular disease with 23.5% active and 71% healed rheumatic lesions. About 5.5% had morphological appearances consistent with RHD. The lesions affected mitral valves (37.0%), aortic valve (22.1%), mitral and aortic valves together (21.0%) and atrial appendages (19.0%). Presentation was mostly as mitral stenosis either isolated (49.2% ) or combined (31.0%), aortic stenosis (11.7% ) and aortic incompetence with regurgitation (7.3%). Conclusion: Rheumatic carditis constitutes a significant proportion of cardiac valvular disease and affects comparatively younger age, with slight male preponderance and primarily affects mitral valve. (author)
Lindman, Brian R; Maniar, Hersh S; Jaber, Wael A; Lerakis, Stamatios; Mack, Michael J; Suri, Rakesh M; Thourani, Vinod H; Babaliaros, Vasilis; Kereiakes, Dean J; Whisenant, Brian; Miller, D Craig; Tuzcu, E Murat; Svensson, Lars G; Xu, Ke; Doshi, Darshan; Leon, Martin B; Zajarias, Alan
Tricuspid regurgitation (TR) and right ventricular (RV) dysfunction adversely affect outcomes in patients with heart failure or mitral valve disease, but their impact on outcomes in patients with aortic stenosis treated with transcatheter aortic valve replacement has not been well characterized. Among 542 patients with symptomatic aortic stenosis treated in the Placement of Aortic Transcatheter Valves (PARTNER) II trial (inoperable cohort) with a Sapien or Sapien XT valve via a transfemoral approach, baseline TR severity, right atrial and RV size and RV function were evaluated by echocardiography according to established guidelines. One-year mortality was 16.9%, 17.2%, 32.6%, and 61.1% for patients with no/trace (n=167), mild (n=205), moderate (n=117), and severe (n=18) TR, respectively (Pright atrial and RV enlargement were also associated with increased mortality (Pright atrial and RV enlargement, but not RV dysfunction. There was an interaction between TR and mitral regurgitation severity (P=0.04); the increased hazard of death associated with moderate/severe TR only occurred in those with no/trace/mild mitral regurgitation. In inoperable patients treated with transcatheter aortic valve replacement, moderate or severe TR and right heart enlargement are independently associated with increased 1-year mortality; however, the association between moderate or severe TR and an increased hazard of death was only found in those with minimal mitral regurgitation at baseline. These findings may improve our assessment of anticipated benefit from transcatheter aortic valve replacement and support the need for future studies on TR and the right heart, including whether concomitant treatment of TR in operable but high-risk patients with aortic stenosis is warranted. URL: http://www.clinicaltrials.gov. Unique identifier: NCT01314313. © 2015 American Heart Association, Inc.
Yang, Min; Chen, Chang-Zhi; Wang, Xue-Ning; Zhu, Ya-Bin; Gu, Y. John
Bovine pericardium has been extensively applied as the biomaterial for artificial heart valves and may potentially be used as a scaffold for tissue-engineered heart valves after decellularization. Although various methods of decellularization are currently available, it is unknown which method is
Mohammadi, H; Klassen, R J; Wan, W-K
Pyrolytic carbon mechanical heart valves (MHVs) are widely used to replace dysfunctional and failed heart valves. As the human heart beats around 40 million times per year, fatigue is the prime mechanism of mechanical failure. In this study, a finite element approach is implemented to develop a model for fatigue analysis of MHVs due to the impact force between the leaflet and the stent and cavitation in the aortic position. A two-step method to predict crack propagation in the leaflets of MHVs has been developed. Stress intensity factors (SIFs) are computed at a small initiated crack located on the leaflet edge (the worst case) using the boundary element method (BEM). Static analysis of the crack is performed to analyse the stress distribution around the front crack zone when the crack is opened; this is followed by a dynamic crack analysis to consider crack propagation using the finite element approach. Two factors are taken into account in the calculation of the SIFs: first, the effect of microjet formation due to cavitation in the vicinity of leaflets, resulting in water hammer pressure; second, the effect of the impact force between the leaflet and the stent of the MHVs, both in the closing phase. The critical initial crack length, the SIFs, the water hammer pressure, and the maximum jet velocity due to cavitation have been calculated. With an initial crack length of 35 microm, the fatigue life of the heart valve is greater than 60 years (i.e. about 2.2 x 10(9) cycles) and, with an initial crack length of 170 microm, the fatigue life of the heart valve would be around 2.5 years (i.e. about 9.1 x 10(7) cycles). For an initial crack length greater than 170 microm, there is catastrophic failure and fatigue cracking no longer occurs. A finite element model of fatigue analysis using Patran command language (PCL custom code) in MSC software can be used to evaluate the useful lifespan of MHVs. Similar methodologies can be extended to other medical devices under cyclic
Ferreira, Renata Maria; da Costa, Marise Teresinha Brenner Affonso; Canciglieri Junior, Osiris; Sant'Anna, Ângelo Márcio Oliveira
Human heart valves are used as replacement valves and have satisfactory functional results compared with conventional prostheses. Characterize the profile of effective heart donors from the human valve bank of the santa casa de misericórdia de curitiba and analyze the association between the profile variables. It consists of a retrospective and quantitative study of electronic medical records from heart donors for heart valves. every heart donation made to the bank between january 2004 and december 2014 was studied. 2,149 donations were analyzed, from donors aged 0 to 71 years old, with an average of 34.9 ± 15.03 years old. most donors were male 65.7% (n=1,411) and 34.3% (n=738) were female. among the most frequent causes of the donors' death are trauma at 53% (n=1,139) and cerebral vascular accident at 34.2% (n=735). there was significant statistical association between the analyzed variables. There has been an improvement in brazil's donation rate, being essential that the tissue banks work together with the state and federal district centers for notification, procurement and distribution of organs in order to increase the number of donors.
Full Text Available Since 2000, transcatheter pulmonary valve replacement has steadily advanced. However, the available prosthetic valves are restricted to bioprosthesis which have defects like poor durability. Polymeric heart valve is thought as a promising alternative to bioprosthesis. In this study, we introduced a novel polymeric transcatheter pulmonary valve and evaluated its feasibility and safety in sheep by a hybrid approach.We designed a novel polymeric trileaflet transcatheter pulmonary valve with a balloon-expandable stent, and the valve leaflets were made of 0.1-mm expanded polytetrafluoroethylene (ePTFE coated with phosphorylcholine. We chose glutaraldehyde-treated bovine pericardium valves as control. Pulmonary valve stents were implanted in situ by a hybrid transapical approach in 10 healthy sheep (8 for polymeric valve and 2 for bovine pericardium valve, weighing an average of 22.5±2.0 kg. Angiography and cardiac catheter examination were performed after implantation to assess immediate valvular functionality. After 4-week follow-up, angiography, echocardiography, computed tomography, and cardiac catheter examination were used to assess early valvular function. One randomly selected sheep with polymeric valve was euthanized and the explanted valved stent was analyzed macroscopically and microscopically.Implantation was successful in 9 sheep. Angiography at implantation showed all 9 prosthetic valves demonstrated orthotopic position and normal functionality. All 9 sheep survived at 4-week follow-up. Four-week follow-up revealed no evidence of valve stent dislocation or deformation and normal valvular and cardiac functionality. The cardiac catheter examination showed the peak-peak transvalvular pressure gradient of the polymeric valves was 11.9±5.0 mmHg, while that of two bovine pericardium valves were 11 and 17 mmHg. Gross morphology demonstrated good opening and closure characteristics. No thrombus or calcification was seen macroscopically
Ellis, J T; Healy, T M; Fontaine, A A; Weston, M W; Jarret, C A; Saxena, R; Yoganathan, A P
Fluid stresses occurring in retrograde flow fields during valve closure may play a significant role in thrombogenesis. The squeeze flow and regurgitant jets can cause damage to formed blood elements due to high levels of turbulent shear stress. The aim of this study was to characterize in detail the spatial structure and temporal behavior of the retrograde flow fields of the St. Jude Medical and Medtronic Parallel bileaflet mechanical heart valves. Three-component, coincident laser Doppler anemometry (LDA) velocity measurements were obtained facilitating the determination of the full Reynolds stress tensor and the principal stresses in the valve flow fields. The experiments were performed in the Georgia Tech aortic flow chamber under physiologic pulsatile flow conditions. Data were collected over several hundred cardiac cycles for subsequent phase window averaging and generation of mean velocity and turbulence statistics over 20 ms intervals. A region approximately 8 mm x 10 mm was mapped 1.0 mm upstream of one hinge of each valve with an incremental resolution of 0.13-0.25 mm. Animation of the data allowed the visualization of the flow fields and a quantitative display of mean velocity and turbulent stress values. In the St. Jude Medical squeeze flow, the peak turbulent shear stress was 800 dynes/cm2 and the peak reverse velocity was 0.60 m/s. In the Medtronic Parallel squeeze flow, the peak turbulent shear stress was 1,000 dynes/cm2 and the peak velocity 0.70 m/s. The leakage jet fields of the two valves were very different: in the case of the St. Jude Medical valve, turbulent shear stresses reached 1,800 dynes/cm2 and peak jet velocity was 0.80 m/s; in the case of the Medtronic Parallel valve, turbulent shear stresses reached 3,690 dynes/cm2 and the peak jet velocity was 1.9 m/s. The retrograde flow fields of these two bileaflet mechanical heart valves appear to be design-dependent. The elevated turbulent shear stresses generated by both valve designs may
Allareddy, Veerasathpurush; Elangovan, Satheesh; Rampa, Sankeerth; Shin, Kyungsup; Nalliah, Romesh P; Allareddy, Veerajalandhar
To examine the prevalence and impact of gingivitis and periodontitis in patients having heart valve surgical procedures. Nationwide Inpatient Sample for the years 2004-2010 was used. All patients who had heart valve surgical procedures were selected. Prevalence of gingivitis/periodontitis was examined in these patients. Impact of gingivitis/periodontitis on hospital charges, length of stay, and infectious complications was examined. 596,190 patients had heart valve surgical procedures. Gingivitis/periodontitis was present in 0.2 percent. Outcomes included: median hospital charges ($175,418 with gingivitis/ periodontitis versus $149,353 without gingivitis/periodontitis) and median length of stay (14 days with gingivitis/periodontitis versus 8 days without gingivitis/periodontitis). After adjusting for the effects of patient- and hospital-level confounding factors, hospital charges and length of stay were significantly higher (p gingivitis/periodontitis compared to their counterparts. Further, patients with gingivitis/periodontitis had significantly higher odds for having bacterial infections (OR = 3.41, 95% CI = 2.33-4.98, p gingivitis/periodontitis. Presence of gingivitis and periodontitis is associated with higher risk for bacterial infections and significant hospital resource utilization.
Jana, Soumen; Lerman, Amir
Heart valve tissue engineering could be a possible solution for the limitations of mechanical and biological prostheses, which are commonly used for heart valve replacement. In tissue engineering, cells are seeded into a 3-dimensional platform, termed the scaffold, to make the engineered tissue construct. However, mimicking the mechanical and spatial heterogeneity of a heart valve structure in a fabricated scaffold with uniform cell distribution is daunting when approached conventionally. Bioprinting is an emerging technique that can produce biological products containing matrix and cells, together or separately with morphological, structural and mechanical diversity. This advance increases the possibility of fabricating the structure of a heart valve in vitro and using it as a functional tissue construct for implantation. This review describes the use of bioprinting technology in heart valve tissue engineering. Copyright © 2015 Elsevier Inc. All rights reserved.
Christensen, Thomas D; Attermann, Jørn; Pilegaard, Hans K
.4%–2.9%) for the control group. Conclusion: Self-management of OAT is a feasible and safe concept for selected patients with mechanical heart valve prostheses also on a long-term basis. It provides at least as good and most likely better quality of anticoagulant therapy than conventional management assessed by time within......Objective: Self-management of oral anticoagulant therapy (OAT) has shown good results on a short-term basis. We hypothesize that self-management of OAT provides a better quality of treatment than conventional management also on a long-term basis. The aim of this study was to assess the quality...... of conventionally managed heart valve patients (control group) was used as reference. Results: The median observation time was 1175 days (range: 174–1428 days). The self-managed patients were within therapeutic INR target range for a mean of 78.0% (range: 36.1%–93.9%) of the time compared with 61.0% (range 37...
Full Text Available INTRODUCTION: In vitro hydrodynamic characterization of prosthetic heart valves provides important information regarding their operation, especially if performed by noninvasive techniques of anemometry. Once velocity profiles for each valve are provided, it is possible to compare them in terms of hydrodynamic performance. In this first experimental study using laser doppler anemometry with mechanical valves, the simulations were performed at a steady flow workbench. OBJECTIVE: To compare unidimensional velocity profiles at the central plane of two bi-leaflet aortic prosthesis from St. Jude (AGN 21 - 751 and 21 AJ - 501 models exposed to a steady flow regime, on four distinct sections, three downstream and one upstream. METHODS: To provide similar conditions for the flow through each prosthesis by a steady flow workbench (water, flow rate of 17L/min. and, for the same sections and sweeps, to obtain the velocity profiles of each heart valve by unidimensional measurements. RESULTS: It was found that higher velocities correspond to the prosthesis with smaller inner diameter and instabilities of flow are larger as the section of interest is closer to the valve. Regions of recirculation, stagnation of flow, low pressure, and flow peak velocities were also found. CONCLUSIONS: Considering the hydrodynamic aspect and for every section measured, it could be concluded that the prosthesis model AGN 21 - 751 (RegentTM is superior to the 21 AJ - 501 model (Master Series. Based on the results, future studies can choose to focus on specific regions of the these valves.
Kagramanov, I I; Kokshenev, I V; Dobrova, N B; Kastava, V T; Serov, R A; Zaets, S B
The optimal material for heart valve bioprostheses remains disputable. This investigation was initiated to compare the properties of hepatic Glisson's capsule, clinical experience of which in cardiovascular surgery is minimal, with those of bovine pericardium. Hepatic Glisson's capsule was harvested from bull calves and used to create composite pulmonary arterial monocusp grafts and bioprostheses. Comparison of the strength and elastic properties of Glisson's capsule and bovine pericardium, as well as the hydrodynamic characteristics of valves made from these materials, was performed. Late results of operations using these materials were estimated echocardiographically. Although Glisson's capsule tissue is thinner than the bovine pericardium, its elasticity modulus is greater. However, the hydrodynamic characteristics of heart valves made from either tissue are similar. Moreover, valves made from Glisson's capsule have a lower systolic pressure gradient on the prosthesis and a higher effective orifice area. Composite pulmonary arterial xenopericardial grafts with a monocusp of Glisson's capsule were used in 30 patients during tetralogy of Fallot repair. Glisson's capsule was also used for tricuspid valve reconstruction and as a bioprosthesis in six patients with Ebstein's anomaly. At 1-2 years after surgery, the Glisson's capsule tissue remained thin and flexible, with no calcification. Although the hydrodynamic properties of hepatic Glisson's capsule and the bovine pericardium are similar, the capsule tissue is thinner and has a greater elasticity modulos. Thus, Glisson's capsule may be used for bioprosthesis construction both independently and in combination with bovine pericardium.
Full Text Available Background/Aim. The last decade of the 20th century brought up a significant development in the field of minimally invasive approaches to the valvular heart surgery. Potential benefits of this method are: good esthetic appearance, reduced pain, reduction of postoperative hemorrhage and incidence of surgical site infection, shorter postoperative intensive care units (ICU period and overall in-hospital period. Partial upper median sternotomy currently presents as a state-of-the art method for minimally invasive surgery of cardiac valves. The aim of this study was to report on initial experience in application of this surgical method in the surgery of mitral and aortic valves. Methods. The study was designed and conducted in a prospective manner and included all the patients who underwent minimally invasive cardiac valve surgery through the partial upper median sternotomy during the period November 2008 - August 2009. We analyzed the data on mean age of patients, mean extubation time, mean postoperative drainage, mean duration of hospital stay, as well as on occurance of postoperative complications (postoperative bleeding, surgical site infection and cerebrovascular insult. Results. During the observed period, in the Institute for Cardiovascular Diseases of Vojvodina, Clinic for Cardiovascular Surgery, 17 ministernotomies were performed, with 14 aortic valve replacements (82.35% and 3 mitral valve replacements (17.65%. Mean age of the patients was 60.78 ± 12.99 years (64.71% males, 35.29% females. Mean extubation time was 12.53 ± 8.87 hours with 23.5% of the patients extubated in less than 8 hours. Mean duration of hospital stay was 12.35 ± 10.17 days (in 29.4% of the patients less than 8 days. Mean postoperative drainage was 547.06 ± 335.2 mL. Postoperative complications included: bleeding (5.88% and cerebrovascular insult (5.88%. One patient (5.88% required conversion to full sternotomy. Conclusion. Partial upper median sternotomy represents
Full Text Available Tissue-engineered heart valves are proposed as novel viable replacements granting longer durability and growth potential. However, they require extensive in vitro cell-conditioning in bioreactor before implantation. Here, the propensity of non-preconditioned decellularized heart valves to spontaneous in body self-regeneration was investigated in a large animal model. Decellularized porcine aortic valves were evaluated for right ventricular outflow tract (RVOT reconstruction in Vietnamese Pigs (n = 11 with 6 (n = 5 and 15 (n = 6 follow-up months. Repositioned native valves (n = 2 for each time were considered as control. Tissue and cell components from explanted valves were investigated by histology, immunohistochemistry, electron microscopy, and gene expression. Most substitutes constantly demonstrated in vivo adequate hemodynamic performances and ex vivo progressive repopulation during the 15 implantation months without signs of calcifications, fibrosis and/or thrombosis, as revealed by histological, immunohistochemical, ultrastructural, metabolic and transcriptomic profiles. Colonizing cells displayed native-like phenotypes and actively synthesized novel extracellular matrix elements, as collagen and elastin fibers. New mature blood vessels, i.e. capillaries and vasa vasorum, were identified in repopulated valves especially in the medial and adventitial tunicae of regenerated arterial walls. Such findings correlated to the up-regulated vascular gene transcription. Neoinnervation hallmarks were appreciated at histological and ultrastructural levels. Macrophage populations with reparative M2 phenotype were highly represented in repopulated valves. Indeed, no aspects of adverse/immune reaction were revealed in immunohistochemical and transcriptomic patterns. Among differentiated elements, several cells were identified expressing typical stem cell markers of embryonic, hematopoietic, neural and mesenchymal lineages in significantly
Frese, L.; Sasse, T.; Sanders, B.; Baaijens, F.P.T.; Beer, G.M.; Hoerstrup, S.P.
Tissue-engineered heart valves represent a promising strategy for the growing need for valve replacements in cardiovascular medicine. Recent studies have shown that adipose-derived stem cells (ADSC) are a viable cell source, as they are readily available in both the young and the elderly, show
Wyller, Vegard Bruun; Aaberge, Lars; Thaulow, Erik; Døhlen, Gaute
Percutaneous catheter-based implantation of artificial heart valves is a new technique that may supplement surgery and which may be used more in the future. We here report our first experience with implantation of artificial pulmonary valves in children with congenital heart defects. Eligible patients were those with symptoms of heart failure combined with stenosis and/or insufficiency in an established artificial right ventricular outflow tract. The valve was inserted through a catheter from a vein in the groin or neck. Symptoms, echocardiography, invasive measurements and angiography were assessed for evaluation of treatment effect. Our treatment results are reported for the period April 2007-September 2009. Ten patients (seven men and three women, median age 17 years) were assessed. The procedure reduced pressure in the right ventricle (p = 0.008) and resolved the pulmonary insufficiency in all patients. The median time in hospital was two days. No patients had complications that were directly associated with the implantation procedure. One patient developed a pseudoaneurysm in the femoral artery, another had a short-lasting fever two days after the procedure and one patient experienced a stent fracture that required surgery 9 months after the implantation. After 6 months all patients had a reduced pressure gradient in the right ventricular outflow tract (p = 0.008), the pulmonary insufficiency had improved (p = 0.006) and they all reported improval of symptoms. These results persisted for at least 24 months for the four patients who were monitored until then. Percutaneous catheter-based implantation of artificial pulmonary valves improves hemodynamics in the right ventricle of selected patients with congenital heart defects. A randomized controlled study should be undertaken to provide a stronger evidence-base for usefulness of this procedure.
... valve syndrome . What happens during MVP? Watch an animation of mitral valve prolapse When the heart pumps ( ... our brochures Popular Articles 1 Understanding Blood Pressure Readings 2 Sodium and Salt 3 Heart Attack Symptoms ...
... Taking warfarin (Coumadin) References Otto CM, Bonow RO. Valvular heart disease. In: Mann DL, Zipes DP, Libby P, Bonow ... A.M. Editorial team. Heart Surgery Read more Heart Valve Diseases Read more Mitral Valve Prolapse Read more A. ...
Full Text Available Phonocardiography has shown a great potential for developing low-cost computer-aided diagnosis systems for cardiovascular monitoring. So far, most of the work reported regarding cardiosignal analysis using multifractals is oriented towards heartbeat dynamics. This paper represents a step towards automatic detection of one of the most common pathological syndromes, so-called mitral valve prolapse (MVP, using phonocardiograms and multifractal analysis. Subtle features characteristic for MVP in phonocardiograms may be difficult to detect. The approach for revealing such features should be locally based rather than globally based. Nevertheless, if their appearances are specific and frequent, they can affect a multifractal spectrum. This has been the case in our experiment with the click syndrome. Totally, 117 pediatric phonocardiographic recordings (PCGs, 8 seconds long each, obtained from 117 patients were used for PMV automatic detection. We propose a two-step algorithm to distinguish PCGs that belong to children with healthy hearts and children with prolapsed mitral valves (PMVs. Obtained results show high accuracy of the method. We achieved 96.91% accuracy on the dataset (97 recordings. Additionally, 90% accuracy is achieved for the evaluation dataset (20 recordings. Content of the datasets is confirmed by the echocardiographic screening.
Midha, Prem A; Raghav, Vrishank; Sharma, Rahul; Condado, Jose F; Okafor, Ikechukwu U; Rami, Tanya; Kumar, Gautam; Thourani, Vinod H; Jilaihawi, Hasan; Babaliaros, Vasilis; Makkar, Raj R; Yoganathan, Ajit P
Transcatheter heart valve (THV) thrombosis has been increasingly reported. In these studies, thrombus quantification has been based on a 2-dimensional assessment of a 3-dimensional phenomenon. Postprocedural, 4-dimensional, volume-rendered CT data of patients with CoreValve, Evolut R, and SAPIEN 3 transcatheter aortic valve replacement enrolled in the RESOLVE study (Assessment of Transcatheter and Surgical Aortic Bioprosthetic Valve Dysfunction With Multimodality Imaging and Its Treatment with Anticoagulation) were included in this analysis. Patients on anticoagulation were excluded. SAPIEN 3 and CoreValve/Evolut R patients with and without hypoattenuated leaflet thickening were included to study differences between groups. Patients were classified as having THV thrombosis if there was any evidence of hypoattenuated leaflet thickening. Anatomic and THV deployment geometries were analyzed, and thrombus volumes were computed through manual 3-dimensional reconstruction. We aimed to identify and evaluate risk factors that contribute to THV thrombosis through the combination of retrospective clinical data analysis and in vitro imaging in the space between the native and THV leaflets (neosinus). SAPIEN 3 valves with leaflet thrombosis were on average 10% further expanded (by diameter) than those without (95.5±5.2% versus 85.4±3.9%; P <0.001). However, this relationship was not evident with the CoreValve/Evolut R. In CoreValve/Evolut Rs with thrombosis, the thrombus volume increased linearly with implant depth ( R 2 =0.7, P <0.001). This finding was not seen in the SAPIEN 3. The in vitro analysis showed that a supraannular THV deployment resulted in a nearly 7-fold decrease in stagnation zone size (velocities <0.1 m/s) when compared with an intraannular deployment. In addition, the in vitro model indicated that the size of the stagnation zone increased as cardiac output decreased. Although transcatheter aortic valve replacement thrombosis is a multifactorial process
Bluestein, D; Einav, S
The presence of turbulence in the cardiovascular system is generally an indication of some type of abnormality. Most cardiologists agree that turbulence near a valve indicates either valvular stenosis or regurgitation, depending on the phase of its occurrence during the cardiac cycle. As no satisfying analytical solutions of the stability of turbulent pulsatile flow exist, accurate, unbiased flow stability criteria are needed for the identification of turbulence initiation. The traditional approach uses a stability diagram based upon the stability of a plane Stokes layer where alpha (the Womersley parameter) is defined by the fundamental heart rate. We suggest a modified approach that involves the decomposition of alpha into its frequency components, where alpha is derived from the preferred modes induced on the flow by interaction between flow pulsation and the valve. Transition to turbulence in pulsatile flow through heart values was investigated in a pulse duplicator system using three polymer aortic valve models representing a normal aortic valve, a 65 percent stenosed valve and a 90 percent severely stenosed valve, and two mitral valve models representing a normal mitral valve and a 65 percent stenosed valve. Valve characteristics were closely simulated as to mimic the conditions that alter flow stability and initiate turbulent flow conditions. Valvular velocity waveforms were measured by laser Doppler anemometry (LDA). Spectral analysis was performed on velocity signals at selected spatial and temporal points to produce the power density spectra, in which the preferred frequency modes were identified. The spectra obtained during the rapid closure stage of the valves were found to be governed by the stenosis geometry. A shift toward higher dominant frequencies was correlated with the severity of the stenosis. According to the modified approach, stability of the flow is represented by a cluster of points, each corresponding to a specific dominant mode apparent
The protocol for cryopreservation of allograft heart valves at the Donor Tissue Bank of Victoria was based on validation studies on the viability of the heart valve leaflets at the time of processing. The heart block is removed within 24 hours of death and the aor-tic and pulmonary valves trimmed immediately following retrieval. Following this processing, the valves are incubated in antibiotics at 30 degree C for 6 to 8 hours before being frozen in 10% DMSO at a controlled rate. A sample of tricuspid valve leaflet is placed in Krebs solution at the time of trimfning and is used for viability studies. Leaflet viability studies have been perfon-ned on all heart valves retrieved from 1993 to the present day at the Donor Tissue Bank of Victoria. Viability involves a qualitative assessment of the cellular outgrowth by leaflet fibroblasts, this assessment ranging from '-' for no outgrowth to '++++' for maximum outgrowth. Surgeons do not request valves with any particular viability and will use them whether they are viable or not. This evaluation was to determine the effects of long-term cryopreservation, cause of death, and also time lapse of heart removal following death on the viability of the retrieved leaflets. The aim of investigating the effects of long-term cryopreservation was to determine whether there was any correlation between initial viability and viability following storage for several months to several years. It was also decided to investigate whether there was any correlation between time length between death and heart retrieval and the viability. It was also thought that the cause of death may have had an effect on the viability, for example, did death by carbon monoxide poisoning have an effect on the viability of heart valve cells. Heart valves, which had been cryopreserved but could not be transplanted for various reasons were used to study the effects of cryopreservation in this study. These were thawed according to protocol and a sample of the valve
W. Tanis (Wilco); R.P.J. Budde (Ricardo); I.A.C. van der Bilt (Ivo); B. Delemarre; G. Hoohenker; J.-K. Van Rooden; A.M. Scholtens (Asbjørn M.); J. Habets; S.A.J. Chamuleau (Steven)
textabstractProsthetic heart valve (PHV) dysfunction remains difficult to recognise correctly by two-dimensional (2D) transthoracic and transoesophageal echocardiography (TTE/TEE). ECG-triggered multidetector-row computed tomography (MDCT), 18-fluorine-fluorodesoxyglucose positron emission
Cheng, Lingbo; Sharifi, Mojtaba; Tavakoli, Mahdi
Beating-heart intracardiac surgery promises significant benefits for patients compared with cardiopulmonary bypass based procedures. However, the fast motions of the heart introduce serious challenges for surgeons. In this work, a new impedance-controlled master-slave telerobotic system is developed to help perform anchor deployment for mitral valve annuloplasty under the guidance of live ultrasound images of the heart. The proposed bilateral teleoperation system can both reflect the non-oscillatory portion of slave-heart tissue interaction force on the surgeon's hand as haptic feedback and implement rapid compensation for the beating heart's motion. The surgical task involves performing anchor deployment on a simulated moving heart tissue to evaluate the effectiveness of the proposed strategy for safely interacting with a moving organ. The results obtained show that the telerobotic system increases the success rate of anchor deployment by 100% and reduces the excess force application rate by 70% compared with manual attempts. Copyright © 2018 John Wiley & Sons, Ltd.
Tanis, W; Budde, R P J; van der Bilt, I A C; Delemarre, B; Hoohenkerk, G; van Rooden, J-K; Scholtens, A M; Habets, J; Chamuleau, S
Prosthetic heart valve (PHV) dysfunction remains difficult to recognise correctly by two-dimensional (2D) transthoracic and transoesophageal echocardiography (TTE/TEE). ECG-triggered multidetector-row computed tomography (MDCT), 18-fluorine-fluorodesoxyglucose positron emission tomography including
Li, Chi-Pei; Lu, Po-Chien; Liu, Jia-Shing; Lo, Chi-Wen; Hwang, Ned H
Cavitation occurs during mechanical heart valve closure when the local pressure drops below vapor pressure. The formation of stable gas bubbles may result in gaseous emboli, and secondarily cause transient ischemic attacks or strokes. It is noted that instantaneous valve closure, occluder rebound and high-speed leakage flow generate vortices that promote low-pressure regions in favor of stable bubble formation; however, to date no studies have been conducted for the quantitative measurement and analysis of these vortices. A Björk-Shiley Monostrut (BSM) monoleaflet valve was placed in the mitral position of a pulsatile mock circulatory loop. Particle image velocimetry (PIV) and pico coulomb (PCB) pressure measurements were applied. Flow field measurements were carried out at t = -5, -3, -1, -0.5, 0 (valve closure), 0.3, 0.5, 0.75, 1.19, 1.44, 1.69, 1.94, 2, 2.19, 2.54, 2.79, 3.04, 3.29, 3.54, 5 and 10 ms. The vortices were quantitatively analyzed using the Rankine vortex model. A single counter-clockwise vortex was The instantaneous formation of cavitation bubbles at mechanical heart valve (MHV) closure, which subsequently damage blood cells and valve integrity, is a well-known and widely studied phenomenon (1-4). Contributing factors seem to include the water-hammer, squeeze flow and Venturi effects, all of which are short-lived. Both, Dauzat et al. (5) and Sliwka et al. (6) have detected high-intensity transient signals (HITS) with transcranial Doppler ultrasound in the carotid and cerebral arteries of MHV recipients, while Deklunder (7) observed clinical occurrences of cerebral gas emboli that were not seen with bioprosthetic valves. These detected over the major orifice, while a pair of counter-rotating vortices was found over the minor orifice. Velocity profiles were consistent with Rankine vortices. The vortex strength and magnitude of the pressure drop peaked shortly after initial occluder-housing impact and rapidly decreased after 0.5 ms, indicating viscous
Vandenbroucke, J P; Grobbee, D E
Two lessons are to be learnt from the Björk-Shiley heart valve prosthesis tragedy. In the first place pharmacoepidemiologic studies are seriously hampered by recent privacy legislation. Individual patients carrying such a prosthesis cannot be traced and advised as to their health risks any more, because their legal autonomy has to be respected. This is clearly not to their advantage. In the second place the atmosphere of marketing and litigation and the increasing dependency of researchers on money from sources with conflicting interests is not conducive to a well-informed and balanced judgement of the epidemiological evidence of safety and efficacy of medical treatments.
Elhenawy, Abdelsalam M; Feindel, Christopher M; Ross, Heather; Butany, Jagdish; Yau, Terrence M
A 45-year-old female underwent heart transplantation 17 years ago, with a heart from a 15-year-old donor. Recently, she had developed an aneurysm of the donor aortic root and ascending aorta, with severe aortic insufficiency. Two surgical options were considered; retransplantation versus replacement of the aortic root and ascending aorta. A valve-sparing replacement of the aortic root and ascending aortic aneurysm was performed. The donor aorta showed pathologic changes typical of Marfan syndrome. Nineteen months postoperatively, the patient remains in functional class I, with trivial aortic insufficiency. Copyright © 2012 The Society of Thoracic Surgeons. Published by Elsevier Inc. All rights reserved.
Thyregod, Hans Gustav Hørsted; Holmberg, Fredrik; Gerds, Thomas Alexander
Objectives: After transcatheter aortic valve implantation (TAVI) has been available for high-risk patients with severe aortic valve stenosis (AVS), the decision-making of the Heart Team (HT) has not been examined. Design: All adult patients with severe AVS referred to a large tertiary medical......%), and surgical aortic valve replacement (SAVR) in 392 (81%) of patients. In patients referred to intervention, TAVI compared with SAVR patients were older (OR = 1.17 per year, 95% CI 1.09-1.26; p obesity (OR = 4.69, 1.......51-13.77; p disease (COPD) (OR = 3.66, 1.21-10.75; p = 0.02). MT patients compared with patients referred to any intervention were older, had a higher prevalence of COPD, peripheral arterial disease, previous myocardial infarction, and cerebrovascular disease...
Dvir, Danny; Webb, John G; Bleiziffer, Sabine
for patients with structural valve deterioration; however, a comprehensive evaluation of survival after the procedure has not yet been performed. OBJECTIVE: To determine the survival of patients after transcatheter valve-in-valve implantation inside failed surgical bioprosthetic valves. DESIGN, SETTING......, stroke, and New York Heart Association functional class. RESULTS: Modes of bioprosthesis failure were stenosis (n = 181 [39.4%]), regurgitation (n = 139 [30.3%]), and combined (n = 139 [30.3%]). The stenosis group had a higher percentage of small valves (37% vs 20.9% and 26.6% in the regurgitation...... and combined groups, respectively; P = .005). Within 1 month following valve-in-valve implantation, 35 (7.6%) patients died, 8 (1.7%) had major stroke, and 313 (92.6%) of surviving patients had good functional status (New York Heart Association class I/II). The overall 1-year Kaplan-Meier survival rate was 83...
Vennemann, Bernhard M; Rösgen, Thomas; Carrel, Thierry P; Obrist, Dominik
The Lapeyre-Triflo FURTIVA valve aims at combining the favorable hemodynamics of bioprosthetic heart valves with the durability of mechanical heart valves (MHVs). The pivoting region of MHVs is hemodynamically of special interest as it may be a region of high shear stresses, combined with areas of flow stagnation. Here, platelets can be activated and may form a thrombus which in the most severe case can compromise leaflet mobility. In this study we set up an experiment to replicate the pulsatile flow in the aortic root and to study the flow in the pivoting region under physiological hemodynamic conditions (CO = 4.5 L/min / CO = 3.0 L/min, f = 60 BPM). It was found that the flow velocity in the pivoting region could reach values close to that of the bulk flow during systole. At the onset of diastole the three valve leaflets closed in a very synchronous manner within an average closing time of 55 ms which is much slower than what has been measured for traditional bileaflet MHVs. Hot spots for elevated viscous shear stresses were found at the flanges of the housing and the tips of the leaflet ears. Systolic VSS was maximal during mid-systole and reached levels of up to 40 Pa.
... blood. There are 4 valves in the heart: tricuspid, pulmonary, mitral, and aortic. Two types of problems can disrupt blood flow through the valves: regurgitation or stenosis. Regurgitation is also called insufficiency or incompetence. Regurgitation happens when a valve doesn’ ...
Rubbens, M.P.; Mol, A.; Boerboom, R.A.; Bank, R.A.; Baaijens, F.P.T.; Bouten, C.V.C.
Tissue-engineered heart valves lack sufficient amounts of functionally organized structures and consequently do not meet in vivo mechanical demands. To optimize tissue architecture and hence improve mechanical properties, various in vitro mechanical conditioning protocols have been proposed, of
Kang, Yuan; Hu, Sheng-Yan; Chou, Hsien-Chin; Lee, Hsing-Han
Constant flow valves have been presented in industrial applications or academic studies, which compensate recess pressures of a hydrostatic bearing to resist load fluctuating. The flow rate of constant-flow valves can be constant in spite of the pressure changes in recesses, however the design parameters must be specified. This paper analyzes the dynamic responses of DSI-type constant-flow valves that is designed as double pistons on both ends of a spool with single feedback of working pressure and regulating restriction at inlet. In this study the static analysis presents the specific relationships among design parameters for constant flow rate and the dynamic analyses give the variations around the constant flow rate as the working pressure fluctuates.
Milano, A; Bortolotti, U; Mazzucco, A; Mossuto, E; Testolin, L; Thiene, G; Gallucci, V
aortic valve replacement, 92% +/- 4% after mitral valve replacement, and 89% +/- 3% after aortic valve replacement, with no cases of mechanical fracture. The Sorin valve has shown a satisfactory long-term overall performance, comparable with other mechanical prostheses, and an excellent durability that renders it a reliable heart valve substitute for the mitral and aortic positions.
Rasmussen, Caroline Elisabeth; Falk, Bo Torkel; Zois, Nora Elisabeth
Sciences, University of Copenhagen, Frederiksberg, Denmark, 2Department of Clinical Sciences, Swedish University of Agricultural Sciences, Uppsala, Sweden; and 3Novo Nordic A/S, Maaloev, Denmark. Introduction: Modulation of heart rate by the autonomic nervous system can indirectly be measured by heart rate...... variability (HRV). Reduced HRV is seen in dogs with heart failure secondary to myxomatous mitral valve disease (MMVD). However, HRV is suggested to increase with disease progression in dogs with early stages of MMVD. Comparable results are found in people with primary mitral valve prolapse, a disease...... resembling canine MMVD. Aim: To associate progression of MMVD in dogs with time and frequency domain HRV, analysed from 24-hour electrocardiography. Materials and Methods: Eighty-one Cavalier King Charles Spaniels (CKCS) were examined by echocardiography and 24-hour electrocardiography. CKCS were divided...
Full Text Available The purpose of the present work is the analysis of the fluid dynamic behavior of a high performance internal combustion engine during the intake phase. In particular, a four-valve spark-ignition engine has been characterized at the steady flow rig. Dimensionless discharge coefficients have been used to define the global fluid dynamic efficiency of the intake system, while the Laser Doppler Anemometry (LDA technique has been employed to evaluate the mean flow in the valve curtain area and to characterise the interference phenomena between the two intake valves. The investigation has shown the significant influence of the valve lift on the volumetric efficiency of the intake apparatus. Moreover, the experimental analysis has highlighted that the valve-valve interference phenomena have a relevant impact on the head breathability, on the flow development within the combustion chamber and on the velocity standard deviations.
Oana, Fumiki; Onozuka, Hiroshi; Tsuchioka, Akihiro; Suzuki, Takayuki; Tanaka, Nobuyuki; Kaidoh, Kouichi; Hoyano, Yuji; Hiratochi, Masahiro; Kikuchi, Shinji; Takehana, Yasuo; Shibata, Nobuo
The symptoms of Parkinson's disease are alleviated by dopamine D2 agonists, which are classified as ergot dopamine D2 agonists and non-ergot D2 agonists. Among the former, pergolide has been associated with valvular heart disease, since it has both potent D2 receptor and serotonin 5-HT(2B) receptor agonistic properties. Among the latter, pramipexole has few incidences of heart valve disease onset, since it has an absence of 5-HT(2B) receptor agonism. A [3H]thymidine incorporation assay was performed to monitor function, and microarray global analysis to monitor gene expression, on porcine heart valve interstitial cells (VICs) treated with pergolide or pramipexole. The 5-HT(2B) receptor was abundantly expressed in porcine VICs. The 5-HT(2B) receptor agonist pergolide induced an increase in [3H]thymidine incorporation, accompanied by a decrease in 5-HT(2B) receptor mRNA expression. [3H]thymidine incorporation was blocked by lisuride, a 5-HT(2B) receptor antagonist, and also by LY-294002, a specific inhibitor of PI3K and Akt. Moreover, type 2 iodothyronine deiodinase (Dio2) expression in porcine VICs treated with pergolide was shown, by a global analysis of mRNA, to be markedly increased compared to that induced by pramipexole. Such changes in VICs may correlate with the mechanism of heart valve disease pathogenesis. There were substantial differences (increased [3H]thymidine incorporation, and Dio2 expression) between pergolide and pramipexole, which might correlate with the mechanism of heart valve disease onset.
Kamensky, David; Hsu, Ming-Chen; Schillinger, Dominik; Evans, John A.; Aggarwal, Ankush; Bazilevs, Yuri; Sacks, Michael S.; Hughes, Thomas J. R.
In this paper, we develop a geometrically flexible technique for computational fluid–structure interaction (FSI). The motivating application is the simulation of tri-leaflet bioprosthetic heart valve function over the complete cardiac cycle. Due to the complex motion of the heart valve leaflets, the fluid domain undergoes large deformations, including changes of topology. The proposed method directly analyzes a spline-based surface representation of the structure by immersing it into a non-boundary-fitted discretization of the surrounding fluid domain. This places our method within an emerging class of computational techniques that aim to capture geometry on non-boundary-fitted analysis meshes. We introduce the term “immersogeometric analysis” to identify this paradigm. The framework starts with an augmented Lagrangian formulation for FSI that enforces kinematic constraints with a combination of Lagrange multipliers and penalty forces. For immersed volumetric objects, we formally eliminate the multiplier field by substituting a fluid–structure interface traction, arriving at Nitsche’s method for enforcing Dirichlet boundary conditions on object surfaces. For immersed thin shell structures modeled geometrically as surfaces, the tractions from opposite sides cancel due to the continuity of the background fluid solution space, leaving a penalty method. Application to a bioprosthetic heart valve, where there is a large pressure jump across the leaflets, reveals shortcomings of the penalty approach. To counteract steep pressure gradients through the structure without the conditioning problems that accompany strong penalty forces, we resurrect the Lagrange multiplier field. Further, since the fluid discretization is not tailored to the structure geometry, there is a significant error in the approximation of pressure discontinuities across the shell. This error becomes especially troublesome in residual-based stabilized methods for incompressible flow, leading
van Slooten, Ymkje J.; Freling, Hendrik G.; van Melle, Joost P.; Mulder, Barbara J. M.; Jongbloed, Monique R. M.; Ebels, Tjark; Voors, Adriaan A.; Pieper, Petronella G.
In patients with acquired valvar disease, morbidity and mortality rates after tricuspid valve replacement (TVR) are high. However, in adult patients with congenital heart disease, though data concerning outcome after TVR are scarce, even poorer results are suggested in patients with Ebstein anomaly.
van Slooten, Ymkje J.; Freling, Hendrik G.; van Melle, Joost P.; Mulder, Barbara J. M.; Jongbloed, Monique R. M.; Ebels, Tjark; Voors, Adriaan A.; Pieper, Petronella G.
OBJECTIVES: In patients with acquired valvar disease, morbidity and mortality rates after tricuspid valve replacement (TVR) are high. However, in adult patients with congenital heart disease, though data concerning outcome after TVR are scarce, even poorer results are suggested in patients with
Lamas, Gloria I.; Kairiyama, Eulogia; Navia, Jose
Porcine heart valves glutaraldehyde fixed are implanted in patients with valvular deterioration. Mineralization may be the major factor in the long-term failure of tissue bio prosthesis. Gamma radiation randomly breaks some glutaraldehyde cross-links. As a consequence of irradiation, the polymeric fibers belonging to the valvular tissue are broken too, leading to sites of collagen fiber disorganisation. It is well known that the collagen fibers may act as a passive nucleator of salts where the calcium phosphate salts precipitate. This salt concentration has been described in association with disintegrated sites of protein fiber, which may favour new sites where the calcium salts would be deposit. The irradiation process is a technique used for sterilization of porcine heart valve. The main objective of this work was to study the effect of different doses of gamma radiation on the calcification process of subcutaneously implanted valves in rats. Small pieces from glutaraldehyde fixed valves, irradiated to different doses with a 60 Co sources were implanted subcutaneously in rats. The calcium was measured by X-ray and atomic absorption spectrophotometry. In our experimental conditions and at the radiation doses used in these tests, the calcium measurements on control and irradiated material were not significantly different. We conclude that, at the employed doses, the gamma radiation does not alter the process. (author) [es
Aortic valve stenosis has become the most prevalent valvular heart disease in Europe and North America, and is generally caused by age-related calcification of the aortic valve. For most patients, severe symptomatic aortic stenosis needs effective mechanical relief in the form of valve replacement
Waziri, Farhad; Lyager Nielsen, Sten; Michael Hasenkam, John
Tricuspid regurgitation may be a precursor for heart failure, reduced functional capacity, and poor survival. A human compatible experimental model is required to understand the pathophysiology of the tricuspid valve disease as a basis for validating novel tricuspid valve interventions before clinical use. The study aim was to evaluate and compare the tricuspid valve anatomy of porcine and human hearts. The anatomy of the tricuspid valve and the surrounding structures that affect the valve during a cardiac cycle were examined in detail in 100 fresh and 19 formalin-fixed porcine hearts obtained from Danish Landrace pigs (body weight 80 kg). All valvular dimensions were compared with human data acquired from literature sources. No difference was seen in the tricuspid annulus circumference between porcine and human hearts (13.0 ± 1.2 cm versus 13.5 ± 1.5 cm; p = NS), or in valve area (5.7 ± 1.6 cm2 versus 5.6 ± 1.0 cm2; p = NS). The majority of chordae types exhibited a larger chordal length and thickness in human hearts compared to porcine hearts. In both species, the anterior papillary muscle (PM) was larger than other PMs in the right ventricle, but muscle length varied greatly (range: 5.2-40.3 mm) and was significantly different in pigs and in humans (12.2 ± 3.2 mm versus 19.2 mm; p human hearts.
Sandeep, Nefthi; Punn, Rajesh; Balasubramanian, Sowmya; Smith, Shea N; Reinhartz, Olaf; Zhang, Yulin; Wright, Gail E; Peng, Lynn F; Wise-Faberowski, Lisa; Hanley, Frank L; McElhinney, Doff B
Palliation of hypoplastic left heart syndrome with a standard nonvalved right ventricle to pulmonary artery conduit results in an inefficient circulation in part due to diastolic regurgitation. A composite right ventricle pulmonary artery conduit with a homograft valve has a hypothetical advantage of reducing regurgitation, but may differ in the propensity for stenosis because of valve remodeling. This retrospective cohort study included 130 patients with hypoplastic left heart syndrome who underwent a modified stage 1 procedure with a right ventricle to pulmonary artery conduit from 2002 to 2015. A composite valved conduit (cryopreserved homograft valve anastomosed to a polytetrafluoroethylene tube) was placed in 100 patients (47 aortic, 32 pulmonary, 13 femoral/saphenous vein, 8 unknown), and a nonvalved conduit was used in 30 patients. Echocardiographic functional parameters were evaluated before and after stage 1 palliation and before the bidirectional Glenn procedure, and interstage interventions were assessed. On competing risk analysis, survival over time was better in the valved conduit group (P = .040), but this difference was no longer significant after adjustment for surgical era. There was no significant difference between groups in the cumulative incidence of bidirectional Glenn completion (P = .15). Patients with a valved conduit underwent more interventions for conduit obstruction in the interstage period, but this difference did not reach significance (P = .16). There were no differences between groups in echocardiographic parameters of right ventricle function at baseline or pre-Glenn. In this cohort of patients with hypoplastic left heart syndrome, inclusion of a valved right ventricle to pulmonary artery conduit was not associated with any difference in survival on adjusted analysis and did not confer an identifiable benefit on right ventricle function. Copyright © 2017 The American Association for Thoracic Surgery. Published by Elsevier
Roemer, Daniel Beck; Johansen, Per; Pedersen, Henrik Clemmensen
Switching valves experience opposing fluid forces due to movement of the moving member itself, as the surrounding fluid volume must move to accommodate the movement. This movement-induced fluid force may be divided into three main components; the added mass term, the viscous term and the socalled...... history term. For general valve geometries there are no simple solution to either of these terms. During development and design of such switching valves, it is therefore, common practice to use simple models to describe the opposing fluid forces, neglecting all but the viscous term which is determined...... based on shearing areas and venting channels. For fast acting valves the opposing fluid force may retard the valve performance significantly, if appropriate measures are not taken during the valve design. Unsteady Computational Fluid Dynamics (CFD) simulations are available to simulate the total fluid...
Ritelli, Guido Francesco; Vacca, Andrea
Highlights: • A graphical method to study the operation of counterbalance valves is presented. • The method is suitable to study the impact of valve settings on energy consumption. • Energetic considerations on a hydraulic crane were performed through numerical simulations. • Test were performed to validate the model on steady-state and transient conditions. • Energy consumption reduction potentials in current applications were discussed. - Abstract: Counterbalance valves negatively affect the dynamical behavior of most of the hydraulic systems in which they are utilized, and they introduce additional energy consumption. Various valve architectures have been developed to address the dynamical issues related to the use of these components, and empirical guidelines are also available to determine the best value of their set parameters such as the pilot ratio and the setting pressure. However, the problem of characterizing the portion of energy consumption introduced by counterbalance valves in a hydraulic circuit, and how this consumption is related to the valve setting, has never been addressed according to a systematic approach. This paper addresses this problem and proposes a general approach to study the energy consumption introduced by the counterbalance valves for given operating cycles of the machine in which they are used. In particular, a graphical method is utilized to study the effects of counterbalance valves in the whole hydraulic system. This method is also used to interpret the potentials for energy recovery associated with different settings of the counterbalance valves. The basic considerations which can be obtained using a graphical method are then supported by results provided by a more detailed numerical model of a particular reference machine, a hydraulic crane. The numerical model is validated on the basis of experimental results and permits to derive not only considerations about the energy consumption of counterbalance valves, but
van Gorp, Maarten J; van der Graaf, Yolanda; de Mol, Bas A J M; Bakker, Chris J G; Witkamp, Theo D; Ramos, Lino M P; Mali, Willem P T M
To assess the relationship between heart valve history and susceptibility artifacts at magnetic resonance (MR) imaging of the brain in patients with Björk-Shiley convexoconcave (BSCC) valves. MR images of the brain were obtained in 58 patients with prosthetic heart valves: 20 patients had BSCC valve replacements, and 38 had other types of heart valves. Two experienced neuroradiologists determined the presence or absence of susceptibility artifacts in a consensus reading. Artifacts were defined as characteristic black spots that were visible on T2*-weighted gradient-echo MR images. The statuses of the 20 explanted BSCC valves-specifically, whether they were intact or had an outlet strut fracture (OSF) or a single-leg fracture (SLF)-had been determined earlier. Number of artifacts seen at brain MR imaging was correlated with explanted valve status, and differences were analyzed with nonparametric statistical tests. Significantly more patients with BSCC valves (17 [85%] of 20 patients) than patients with other types of prosthetic valves (18 [47%] of 38 patients) had susceptibility artifacts at MR imaging (P =.005). BSCC valve OSFs were associated with a significantly higher number of artifacts than were intact BSCC valves (P =.01). No significant relationship between SLF and number of artifacts was observed. Susceptibility artifacts at brain MR imaging are not restricted to patients with BSCC valves. These artifacts can be seen on images obtained in patients with various other types of fractured and intact prosthetic heart valves. Copyright RSNA, 2004
Steele, R. Jr.; MacDonald, P.E.; Arendts, J.G.
The Idaho National Engineering Laboratory (INEL) participated in an internationally sponsored seismic research program conducted at the decommissioned Heissdampfreaktor (HDR) located in the Federal Republic of Germany. An existing piping system was modified by installation of an 8-in., naturally aged, motor-operated gate valve from a US nuclear power plant and a piping support system of US design. Six other piping support systems of varying flexibility from stiff to flexible were also installed at various times during the tests. Additional valve loadings included internal hydraulic loads and, during one block of tests, elevated temperature. The operability and integrity of the aged gate valve and the dynamic response of the various piping support system were measured during 25 representative seismic events
Karakoyun, Süleyman; Gürsoy, Ozan Mustafa; Kalçık, Macit; Coban Kökten, Sermin; Ozkan, Mehmet
Pannus formation is an infrequent but serious complication of prosthetic heart valve surgery. The cause of pannus is recognized as a bioreaction to the prostheses; histological investigations have shown that pannus comprises collagen and elastic tissues containing endothelial cells, chronic inflammatory cells, and myofibroblasts. However, the detailed mechanism of its formation has not been fully demonstrated. We aimed to evaluate the potential role of vascular endothelial growth factor (VEGF) and matrix metalloproteinase-2 (MMP-2) in the pathogenesis of pannus formation in three patients with mechanical prosthetic heart valves. Pannus specimens removed from the prostheses were fixed in 10% neutral-buffered formalin for 24 hours after surgical removal and paraffin-embedded using standard procedures. Serial sections were cut at 4 µm for immunohistochemistry analysis. Hematoxylin and eosin (HE) was used in the histological analysis. VEGF and MMP-2 were studied in the immunohistochemistry analysis. Three patients with mechanical prosthetic obstruction due to pannus overgrowth underwent redo valve surgery. In the first and second patients, the mitral prosthesis was explanted along with the pannus overgrowth. The third patient had both aortic and mitral prostheses; the aortic prosthesis was explanted with obstructive pannus formation, whereas the mitral valve was spared with excision of the nonobstructive pannus. The immunohistochemical study demonstrated the expressions of MMP-2 and VEGF in all of the pannus specimens acquired from these cases. VEGF and MMP-2 may play a role in the mechanism of pannus formation as the elements of the chronic active inflammatory process.
Antunes, Manuel J; Rodríguez-Palomares, José; Prendergast, Bernard; De Bonis, Michele; Rosenhek, Raphael; Al-Attar, Nawwar; Barili, Fabio; Casselman, Filip; Folliguet, Thierry; Iung, Bernard; Lancellotti, Patrizio; Muneretto, Claudio; Obadia, Jean-François; Pierard, Luc; Suwalski, Piotr; Zamorano, Pepe
Tricuspid regurgitation (TR) is a very frequent manifestation of valvular heart disease. It may be due to the primary involvement of the valve or secondary to pulmonary hypertension or to the left-sided heart valve disease (most commonly rheumatic and involving the mitral valve). The pathophysiology of secondary TR is complex and is intrinsically connected to the anatomy and function of the right ventricle. A systematic multimodality approach to diagnosis and assessment (based not only on the severity of the TR but also on the assessment of annular size, RV function and degree of pulmonary hypertension) is, therefore, essential. Once considered non-important, treatment of secondary TR is currently viewed as an essential concomitant procedure at the time of mitral (and, less frequently, aortic valve) surgery. Although the indications for surgical management of severe TR are now generally accepted (Class I), controversy persists concerning the role of intervention for moderate TR. However, there is a trend for intervention in this setting, especially at the time of surgery for left-sided heart valve disease and/or in patients with significant tricuspid annular dilatation (Class IIa). Currently, surgery remains the best approach for the interventional treatment of TR. Percutaneous tricuspid valve intervention (both repair and replacement) is still in its infancy but may become a reliable option in future, especially for high-risk patients with isolated primary TR or with secondary TR related to advanced left-sided heart valve disease. © The Author 2017. Published by Oxford University Press on behalf of the European Association for Cardio-Thoracic Surgery. All rights reserved.
Infective endocarditis (IE) has been increasingly diagnosed in patients without previously detected predisposing heart disease, but its clinical features have yet to be fully determined. A recent single-center study including echocardiographic images and surgical findings investigated the incidence of undiagnosed, clinically silent valvular or congenital heart diseases and healthcare-associated infective endocarditis (HAIE). The study confirmed that a large proportion of patients with IE have no previous history of heart disease. Analysis of underlying disease in these patients showed that undetected mitral valve prolapse was the most common disease, followed by an apparently structurally normal valve. The patients who developed IE of apparently structurally normal valves had different clinical characteristics and worse outcomes. IE involving a structurally normal valve was associated with both nosocomial and non-nosocomial HAIE, whereas community-acquired IE was more frequent than HAIE. The pathophysiologic mechanism involving the development of non-HAIE or community-acquired IE due to predominantly staphylococcal infection in an apparently structurally normal valve is not yet clearly understood. Structurally normal valves are not necessarily free of regurgitation or abnormal turbulence and, given the dynamic nature and fluctuating hemodynamic effects of conditions such as poorly controlled hypertension, end-stage renal disease, and sleep apnea, further investigation is necessary to evaluate the potential role of these diseases in the development of IE. An apparently normal-looking valve is associated with IE development in patients without previously recognized predisposing heart disease, warranting repartition of at-risk groups to achieve better clinical outcomes.
Sibilitz, Kirstine L.; Berg, Selina K.; Rasmussen, Trine B.
Objective: The evidence for cardiac rehabilitation after valve surgery remains sparse. Current recommendations are therefore based on patients with ischaemic heart disease. The aim of this randomised clinical trial was to assess the effects of cardiac rehabilitation versus usual care after heart......-educational consultations (intervention) versus usual care without structured physical exercise or psycho-educational consultations (control). Primary outcome was physical capacity measured by VO2 peak and secondary outcome was self-reported mental health measured by Short Form-36. Results: 76% were men, mean age 62 years......, with aortic (62%), mitral (36%) or tricuspid/pulmonary valve surgery (2%). Cardiac rehabilitation compared with control had a beneficial effect on VO2 peak at 4 months (24.8 mL/kg/min vs 22.5 mL/kg/min, p=0.045) but did not affect Short Form-36 Mental Component Scale at 6 months (53.7 vs 55.2 points, p=0...
van Slooten, Ymkje J; van Melle, Joost P; Freling, Hendrik G; Bouma, Berto J; van Dijk, Arie Pj; Jongbloed, Monique Rm; Post, Martijn C; Sieswerda, Gertjan T; Huis In 't Veld, Anna; Ebels, Tjark; Voors, Adriaan A; Pieper, Petronella G
To report the prevalence of aortic valve prosthesis-patient mismatch (PPM) in an adult population with congenital heart disease (CHD) and its impact on exercise capacity. Adults with congenital heart disease (ACHD) with a history of aortic valve replacement may outgrow their prosthesis later in life. However, the prevalence and clinical consequences of aortic PPM in ACHD are presently unknown. From the national Dutch Congenital Corvitia (CONCOR) registry, we identified 207 ACHD with an aortic valve prosthesis for this cross-sectional cohort study. Severe PPM was defined as an indexed effective orifice area ≤0.65 cm2/m2 and moderate PPM as an indexed orifice area ≤0.85 cm2/m2 measured using echocardiography. Exercise capacity was reported as percentage of predicted exercise capacity (PPEC). Of the 207 patients, 68% was male, 71% had a mechanical prosthesis and mean age at inclusion was 43.9 years ±11.4. The prevalence of PPM was 42%, comprising 23% severe PPM and 19% moderate PPM. Prevalence of PPM was higher in patients with mechanical prostheses (pHeart Association (NYHA) class remained stable in most patients. PPM showed no significant effect on death or hospitalisation during follow-up (p=0.218). In this study we report a high prevalence (42%) of PPM in ACHD with an aortic valve prosthesis and an independent association of PPM with diminished exercise capacity. Published by the BMJ Publishing Group Limited. For permission to use (where not already granted under a licence) please go to http://www.bmj.com/company/products-services/rights-and-licensing/
Mitral valve prolapse (MVP) occurs when one of your heart's valves doesn't work properly. The flaps of the valve are "floppy" and ... to run in families. Most of the time, MVP doesn't cause any problems. Rarely, blood can ...
Full Text Available Cardiac valves are dynamic structures, exhibiting a highly specialized architecture consisting of cells and extracellular matrix with a relevant proteoglycan and glycosaminoglycan content, collagen and elastic fibers. Biological valve substitutes are obtained from xenogenic cardiac and pericardial tissues. To overcome the limits of such non viable substitutes, tissue engineering approaches emerged to create cell repopulated decellularized scaffolds. This study was performed to determine the glycosaminoglycans content, distribution, and disaccharides composition in porcine aortic and pulmonary valves and in pericardium before and after a detergent-based decellularization procedure. The fine structural characteristics of galactosaminoglycans chondroitin sulfate and dermatan sulfate were examined by FACE. Furthermore, the mechanical properties of decellularized pericardium and its propensity to be repopulated by in vitro seeded fibroblasts were investigated. Results show that galactosaminoglycans and hyaluronan are differently distributed between pericardium and valves and within heart valves themselves before and after decellularization. The distribution of glycosaminoglycans is also dependent from the vascular district and topographic localization. The decellularization protocol adopted resulted in a relevant but not selective depletion of galactosaminoglycans. As a whole, data suggest that both decellularized porcine heart valves and bovine pericardium represent promising materials bearing the potential for future development of tissue engineered heart valve scaffolds.
Ma, Wei-Guo; Hou, Bin; Abdurusul, Adiljan; Gong, Ding-Xu; Tang, Yue; Chang, Qian; Xu, Jian-Ping
Background Dysfunction of mechanical heart valve prostheses is an unusual but potentially lethal complication after mechanical prosthetic valve replacement. We seek to report our experience with mechanical valve dysfunction regarding etiology, surgical techniques and early outcomes. Methods Clinical data of 48 patients with mechanical valve dysfunction surgically treated between October 1996 and June 2011 were analyzed. Results Mean age was 43.7±10.9 years and 34 were female (70.8%). The median interval from primary valve implantation to dysfunction was 44.5 months (range, 1 hour to 20 years). There were 21 emergent and 27 elective reoperations. The etiology was thrombosis in 19 cases (39.6%), pannus in 12 (25%), thrombosis and pannus in 11 (22.9%), improper disc orientation in 2 (4.1%), missing leaflet in 1 (2.1%), excessively long knot end in 1 (2.1%), endogenous factor in 1 (2.1%) and unidentified in 1 (2.1%). Surgical procedure was mechanical valve replacement in 37 cases (77.1%), bioprosthetic valve replacement in 7 (14.9%), disc rotation in 2 (4.2%) and excision of excessive knot end in 1 (2.1%). Early deaths occurred in 7 patients (14.6%), due to low cardiac output in 3 (6.3%), multi-organ failure in 2 (4.2%) and refractory ventricular fibrillation in 2 (4.2%). Complications occurred in 10 patients (20.8%). Conclusions Surgical management of mechanical valve dysfunction is associated with significant mortality and morbidity. Earlier identification and prompt reoperation are vital to achieving better clinical outcomes. The high incidence of thrombosis in this series highlights the need for adequate anticoagulation and regular follow-up after mechanical valve replacement. PMID:26793354
Kwon, Ki Tae; Han, Hwa Taik
It is intended to develop an algorithm for dynamic simulation of fast-acting solenoid valves. The coupled equations of the electric, magnetic, and mechanical systems should be solved simultaneously in a transient nonlinear manner. The transient nonlinear electromagnetic field is analyzed by the Finite Element Method (FEM), which is coupled with nonlinear electronic circuitry. The dynamic movement of the solenoid valve is analyzed at every time step from the force balances acting on the plunger, which include the electromagnetic force calculated from the finite element analysis as well as the elastic force by a spring and the hydrodynamic pressure force along the flow passage. Dynamic responses of the solenoid valves predicted by this algorithm agree well the experimental results including bouncing effects
Li, Chi-Pei; Lo, Chi-Wen; Lu, Po-Chien
Among the clinical complications of mechanical heart valves (MHVs), hemolysis was previously thought to result from Reynolds stresses in turbulent flows. A more recent hypothesis suggests viscous dissipative stresses at spatial scales similar in size to red blood cells may be related to hemolysis in MHVs, but the resolution of current instrumentation is insufficient to measure the smallest eddy sizes. We studied the St. Jude Medical (SJM) 27 mm valve in the aortic position of a pulsatile circulatory mock loop under physiologic conditions with particle image velocimetry (PIV). Assuming a dynamic equilibrium assumption between the resolved and sub-grid-scale (SGS) energy flux, the SGS energy flux was calculated from the strain rate tensor computed from the resolved velocity fields and the SGS stress was determined by the Smagorinsky model, from which the turbulence dissipation rate and then the viscous dissipative stresses were estimated. Our results showed Reynolds stresses up to 80 N/m2 throughout the cardiac cycle, and viscous dissipative stresses below 12 N/m2. The viscous dissipative stresses remain far below the threshold of red blood cell hemolysis, but could potentially damage platelets, implying the need for further study in the phenomenon of MHV hemolytic complications.
Thyregod, Hans Gustav; Søndergaard, Lars; Ihlemann, Nikolaj
Degenerative aortic valve (AV) stenosis is the most prevalent heart valve disease in the western world. Surgical aortic valve replacement (SAVR) has until recently been the standard of treatment for patients with severe AV stenosis. Whether transcatheter aortic valve implantation (TAVI) can...
Ohta, Y; Kikuta, Y; Shimooka, T; Mitamura, Y; Yuhta, T; Dohi, T
Conventional bileaflet prosthetic mechanical heart valves close passively with backflow. Naturally, the valve has problems associated with closure, such as backflow, water hammer effect, and fracture of the leaflet. On the other hand, in the case of the natural aortic valve, the vortex flow in the sinus of Valsalva pushes the leaflet to close, and the valve starts the closing motion earlier than the prosthetic valve as the forward flow decelerates. This closing mechanism is thought to decrease backflow at valve closure. In this study, we propose a new bileaflet mechanical valve resembling a drawbridge in shape, and the prototype valve was designed so that the leaflet closes with the help of the vortex flow in the sinus. The test valve was made of aluminum alloy, and its closing motion was compared to that of the CarboMedics (CM) valve. Both valves were driven by a computer controlled hydraulic mock circulator and were photographed at 648 frames/s by a high speed charge-coupled device (CCD) camera. Each frame of the valve motion image was analyzed with a personal computer, and the opening angles were measured. The flow rate was set as 5.0 L/min. The system was pulsed with 70 bpm, and the systolic/diastolic ratio was 0.3. Glycerin water was used as the circulation fluid at room temperature, and polystyrene particles were used to visualize the streamline. The model of the sinus of Valsalva was made of transparent silicone rubber. As a result, high speed video analysis showed that the test valve started the closing motion 41 ms earlier than the CM valve, and streamline analysis showed that the test valve had a closing mechanism similar to the natural one with the effect of vortex flow. The structure of the test valve was thought to be effective for soft closure and could solve problems associated with closure.
D'Avenio, Giuseppe; Grigioni, Mauro; Daniele, Carla; Morbiducci, Umberto; Hamilton, Kathrin
Prosthetic heart valves can be associated to mechanical loading of blood, potentially linked to complications (hemolysis and thrombogenicity) which can be clinically relevant. In order to test such devices in pulsatile mode, pulse duplicators (PDs) have been designed and built according to different concepts. This study was carried out to compare anemometric measurements made on the same prosthetic device, with two widely used PDs. The valve (a 27-mm bileaflet valve) was mounted in the aortic section of the PD. The Sheffield University PD and the RWTH Aachen PD were selected as physical models of the circulation. These two PDs differ mainly in the vertical vs horizontal realization, and in the ventricular section, which in the RWTH PD allows for storage of potential energy in the elastic walls of the ventricle. A glassblown aorta, realized according to the geometric data of the same anatomical district in healthy individuals, was positioned downstream of the valve, obtaining 1:1 geometric similarity conditions. A NaI-glycerol-water solution of suitable kinematic viscosity and, at the same time, the proper refractive index, was selected. The flow field downstream of the valve was measured by means of the stereo-PIV (Particle Image Velocimetry) technique, capable of providing the complete 3D velocity field as well as the entire Reynolds stress tensor. The measurements were carried out at the plane intersecting the valve axis. A three-jet profile was clearly found in the plane crossing the leaflets, with both PDs. The extent of the typical recirculation zone in the Valsalva sinus was much larger in the RWTH PD, on account of the different duration of the swirling motion in the ventricular chamber, caused by the elasticity of the ventricle and its geometry. The comparison of the hemodynamical behaviour of the same bileaflet valve tested in two PDs demonstrated the role of the mock loop in affecting the valve performance.
Lancellotti, Patrizio; Rosenhek, Raphael; Pibarot, Philippe; Iung, Bernard; Otto, Catherine M; Tornos, Pilar; Donal, Erwan; Prendergast, Bernard; Magne, Julien; La Canna, Giovanni; Piérard, Luc A; Maurer, Gerald
With an increasing prevalence of patients with valvular heart disease (VHD), a dedicated management approach is needed. The challenges encountered are manifold and include appropriate diagnosis and quantification of valve lesion, organization of adequate follow-up, and making the right management decisions, in particular with regard to the timing and choice of interventions. Data from the Euro Heart Survey have shown a substantial discrepancy between guidelines and clinical practice in the field of VHD and many patients are denied surgery despite having clear indications. The concept of heart valve clinics (HVCs) is increasingly recognized as the way to proceed. At the same time, very few centres have developed such expertise, indicating that specific recommendations for the initial development and subsequent operating requirements of an HVC are needed. The aim of this position paper is to provide insights into the rationale, organization, structure, and expertise needed to establish and operate an HVC. Although the main goal is to improve the clinical management of patients with VHD, the impact of HVCs on education is of particular importance: larger patient volumes foster the required expertise among more senior physicians but are also fundamental for training new cardiologists, medical students, and nurses. Additional benefits arise from research opportunities resulting from such an organized structure and the delivery of standardized care protocols. The growing volume of patients with VHD, their changing characteristics, and the growing technological opportunities of refined diagnosis and treatment in addition to the potential dismal prognosis if overlooked mandate specialized evaluation and care by dedicated physicians working in a specialized environment that is called the HVC.
... and dentist if you have a history of heart valve disease or congenital heart disease before treatment. Some people ... the middle Heart, front view References Carabello BA. Valvular heart disease. In: Goldman L, Schafer AI, eds. Goldman-Cecil ...
Ihlberg, Leo; Nissen, Henrik Hoffmann; Nielsen, Niels Erik
Transcatheter valve-in-valve implantation has emerged as an option, in addition to reoperative surgical aortic valve replacement, to treat failed biologic heart valve substitutes. However, the clinical experience with this approach is still limited. We report the comprehensive experience...
van Gorp, Maarten J.; van der Graaf, Yolanda; de Mol, Bas A. J. M.; Bakker, Chris J. G.; Witkamp, Theo D.; Ramos, Lino M. P.; Mali, Willem P. T. M.
PURPOSE: To assess the relationship between heart valve history and susceptibility artifacts at magnetic resonance (MR) imaging of the brain in patients with Bjork-Shiley convexoconcave (BSCC) valves. MATERIALS AND METHODS: MR images of the brain were obtained in 58 patients with prosthetic heart
Meijer, K.; Kim, Y. -K.; Schulman, S.
Introduction: Vitamin K antagonists are prescribed to prevent thromboembolic complications after heart valve surgery. In our experience, patients often show a progressive decrease in sensitivity to warfarin after surgery making it difficult to reach and maintain a therapeutic International
Lv, Dongwei; Zhang, Jian; Yu, Xinhai
In this paper, a fluid-structure interaction dynamic simulation method of spring-loaded pressure relief valve was established. The dynamic performances of the fluid regions and the stress and strain of the structure regions were calculated at the same time by accurately setting up the contact pairs between the solid parts and the coupling surfaces between the fluid regions and the structure regions. A two way fluid-structure interaction dynamic simulation of a simplified pressure relief valve model was carried out. The influence of vertical sinusoidal seismic waves on the performance of the pressure relief valve was preliminarily investigated by loading sine waves. Under vertical seismic waves, the pressure relief valve will flutter, and the reseating pressure was affected by the amplitude and frequency of the seismic waves. This simulation method of the pressure relief valve under vertical seismic waves can provide effective means for investigating the seismic performances of the valves, and make up for the shortcomings of the experiment.
Full Text Available Abstract Carcinoid heart disease is a rare but important cause of intrinsic right heart valve disorders leading to right heart failure. Occasionally, left-sided heart valves may also be involved. The characteristic cardiac pathological findings of carcinoid heart disease are endocardial thickening as a result of fibrous deposits on the endocardium. Echocardiographic examination and right heart catheterization are very useful for the diagnosis of the lesion. If more cardiac valves are affected, multiple valve replacement should be considered. The management of the pulmonary valve lesion depends on the extent of the diseased valve, either by valvulotomy, valvectomy, or valve replacement. Percutaneous valve implantations in the pulmonary and in the inferior vena cava positions have been advocated for high-risk patients.
Full Text Available Transcatheter valve implantation continues to grow worldwide and has been used principally for the nonsurgical management of native aortic valvular disease-as a potentially less invasive method of valve replacement in high-risk and inoperable patients with severe aortic valve stenosis. Given the burden of valvular heart disease in the general population and the increasing numbers of patients who have had previous valve operations, we are now seeing a growing number of high-risk patients presenting with prosthetic valve stenosis, who are not potential surgical candidates. For this high-risk subset transcatheter valve delivery may be the only option. Here, we present an inoperable patient with severe, prosthetic valve aortic and mitral stenosis who was successfully treated with a trans catheter based approach, with a valve-in-valve implantation procedure of both aortic and mitral valves.
Silaschi, Miriam; Conradi, Lenard; Wendler, Olaf; Schlingloff, Friederike; Kappert, Utz; Rastan, Ardawan J; Baumbach, Hardy; Holzhey, David; Eichinger, Walter; Bader, Ralf; Treede, Hendrik
We present 1-year outcomes of the post-market registry of a next-generation transcatheter heart valve used for aortic regurgitation (AR). Transcatheter aortic valve replacement (TAVR) is routine in high-risk patients with aortic stenosis but is not recommended for AR. The JenaValve™ (JenaValve Technology GmbH, Munich, Germany) overcomes technical challenges in AR patients through a leaflet clipping mechanism. The JenaValve EvalUation of Long Term Performance and Safety In PaTients with SEvere Aortic Stenosis oR Aortic Insufficiency (JUPITER) Registry is a European study to evaluate safety and effectiveness of this THV. From 2012-2015, 30 patients with AR were enrolled. Mean age was 74.4 ± 9.3 years. Procedural success was 96.7% (29/30). One patient was converted to open surgery. No annular rupture or coronary ostia obstruction occurred. Mortality at 30 days was 10.0% (3/30). Combined safety endpoint was met in 13.3% (4/30). Paravalvular regurgitation was not present/trivial in 84.6% (22/26) and mild in 15.4% (4/26). Rate of permanent pacemaker implantation was 3.8% (1/26). One-year Kaplan-Meier survival was 79.9%, one-year combined efficacy was 73.1% (19/30). No further strokes were observed during 1 year of follow-up. The JenaValve overcomes technical challenges of TAVR in AR through a clipping mechanism. We report satisfactory outcomes of a multicenter registry using the JenaValve for predominant AR, as rate of THV embolization, residual AR and permanent pacemaker implantation was low. One-year results using the JenaValve for AR encourage its use for this indication. © 2017 Wiley Periodicals, Inc.
Wang, Chong; Zhang, Guan-xin; Zhang, Hao; Lu, Fang-lin; Li, Bai-ling; Xu, Ji-bin; Han, Lin; Xu, Zhi-yun
The aim of this study was to develop a preoperative risk prediction model and an scorecard for prolonged intensive care unit length of stay (PrlICULOS) in adult patients undergoing heart valve surgery. This is a retrospective observational study of collected data on 3925 consecutive patients older than 18 years, who had undergone heart valve surgery between January 2000 and December 2010. Data were randomly split into a development dataset (n=2401) and a validation dataset (n=1524). A multivariate logistic regression analysis was undertaken using the development dataset to identify independent risk factors for PrlICULOS. Performance of the model was then assessed by observed and expected rates of PrlICULOS on the development and validation dataset. Model calibration and discriminatory ability were analysed by the Hosmer-Lemeshow goodness-of-fit statistic and the area under the receiver operating characteristic (ROC) curve, respectively. There were 491 patients that required PrlICULOS (12.5%). Preoperative independent predictors of PrlICULOS are shown with odds ratio as follows: (1) age, 1.4; (2) chronic obstructive pulmonary disease (COPD), 1.8; (3) atrial fibrillation, 1.4; (4) left bundle branch block, 2.7; (5) ejection fraction, 1.4; (6) left ventricle weight, 1.5; (7) New York Heart Association class III-IV, 1.8; (8) critical preoperative state, 2.0; (9) perivalvular leakage, 6.4; (10) tricuspid valve replacement, 3.8; (11) concurrent CABG, 2.8; and (12) concurrent other cardiac surgery, 1.8. The Hosmer-Lemeshow goodness-of-fit statistic was not statistically significant in both development and validation dataset (P=0.365 vs P=0.310). The ROC curve for the prediction of PrlICULOS in development and validation dataset was 0.717 and 0.700, respectively. We developed and validated a local risk prediction model for PrlICULOS after adult heart valve surgery. This model can be used to calculate patient-specific risk with an equivalent predicted risk at our centre in
Blot, William J; Ibrahim, Michel A; Ivey, Tom D; Acheson, Donald E; Brookmeyer, Ron; Weyman, Arthur; Defauw, Joseph; Smith, J Kermit; Harrison, Donald
The first Björk-Shiley convexoconcave (BSCC) prosthetic heart valves were implanted in 1978. The 25th anniversary provided a stimulus to summarize the research data relevant to BSCC valve fracture, patient management, and current clinical options. Published and unpublished data on the risks of BSCC valve fracture and replacement were compiled, and strategies for identifying candidates for prophylactic valve reoperation were summarized. By December 2003, outlet strut fractures (OSFs), often with fatal outcomes, had been reported in 633 BSCC valves (0.7% of 86,000 valves implanted). Fractures still continue to occur, but average rates of OSFs in 60 degrees valves are now valve characteristics, especially valve angle and size, with weaker effects associated with other manufacturing variables. OSF risks are mildly lower among women than men but decline sharply with advancing age. The risks of valve replacement typically greatly exceed those of OSF. By comparing individualized estimated risks of OSF versus valve replacement, guidelines have been developed to identify the small percentage of BSCC patients (mostly younger men) who would be expected to have a gain in life expectancy should reoperative surgery be performed. Twenty-five years after the initial BSCC valve implants, fractures continue to occur. Continued monitoring of BSCC patients is needed to track and quantify risks and enable periodic updating of guidelines for patients and their physicians.
S. A. Kovalev
Full Text Available Aim. To evaluate in-hospital and long-term results of surgical treatment of patients with infective endocarditis of the tricuspid valve, to compare the effectiveness of valve repair and valve replacement techniques, and to identify risk factors of mortality and reoperations. Materials and methods. 31 surgical patients with tricuspid valve infective endocarditis were evaluated. Patients were divided into 2 groups. In Group 1 (n = 14 repairs of the tricuspid valve were performed, in Group 2 (n = 17 patients had undergone tricuspid valve replacements. Epidemiological, clinical, microbiological and echocardiographic data were studied. Methods of comparative analysis, the Kaplan–Meier method, and Cox risk models were applied. Results. The most common complication of in-hospital stay was atrioventricular block (17.7% of cases in Group 2. In Group 1, this type of complication was not found. Hospital mortality was 7.14% in Group 1, and 0% in Group 2. Long-term results have shown the significant reduction of heart failure in general cohort and in both groups. In Group 1 the severity of heart failure in the long term was less than in Group 2. No significant differences in the severity of tricuspid regurgitation were found between the groups. In 7-year follow up no cases of death were registered in Group 1. Cumulative survival rate in Group 2 within 60 months was 67.3 ± 16.2%. No reoperations were performed in patients from Group 1. In Group 2, the freedom from reoperation within 60 months was 70.9 ± 15.3%. Combined intervention was found as predictor of postoperative mortality. Prosthetic valve endocarditis was identified as risk factor for reoperation. Conclusion. Valve repair and valve replacement techniques of surgical treatment of tricuspid valve endocarditis can provide satisfactory hospital and long-term results. Tricuspid valve repair techniques allowed reducing the incidence of postoperative atrioventricular block. In the long-term, patients
Bellofiore, Alessandro; Quinlan, Nathan J
We investigate the potential of prosthetic heart valves to generate abnormal flow and stress patterns, which can contribute to platelet activation and lysis according to blood damage accumulation mechanisms. High-resolution velocity measurements of the unsteady flow field, obtained with a standard particle image velocimetry system and a scaled-up model valve, are used to estimate the shear stresses arising downstream of the valve, accounting for flow features at scales less than one order of magnitude larger than blood cells. Velocity data at effective spatial and temporal resolution of 60 μm and 1.75 kHz, respectively, enabled accurate extraction of Lagrangian trajectories and loading histories experienced by blood cells. Non-physiological stresses up to 10 Pa were detected, while the development of vortex flow in the wake of the valve was observed to significantly increase the exposure time, favouring platelet activation. The loading histories, combined with empirical models for blood damage, reveal that platelet activation and lysis are promoted at different stages of the heart cycle. Shear stress and blood damage estimates are shown to be sensitive to measurement resolution.
Ye, Xinghai; Ding, Jianjun; Zheng, Gang; Jiang, Kunpeng; Chen, Dongdong
This work focuses on static and dynamic characteristics of direct-acting force motor valve. First, we analyzed the structure features and operating principle of the Mitsubishi-Hitachi force motor valve (FMV) and the operating principle of its internal permanent-magnet moving-coil force motor magnetic circuit, determined the transfer function of the FMV force motor system, and established a mathematical model for the system. Secondly, we established a static performance analysis model using the AMESIM software and utilized the model in combination with experimental results to analyze the effects of electro-hydraulic servo valve structural parameters on static characteristics. Lastly, we deduced the trajectory equation of the system, established the relationship between dynamic characteristic indexes and structural parameters, and analyzed the effects of different parameter values on the dynamic characteristics of the system. This research can provide a theoretical guidance for designing and manufacturing the FMV body.
... surgery - discharge Images Heart valves References Carabello BA. Valvular heart disease. In: Goldman L, Schafer AI, eds. Goldman's Cecil ... Saunders; 2016:chap 69. Otto CM, Bownow RO. Valvular heart disease. In: Mann DL, Zipes DP, Libby P, Bonow ...
Tanis, Wilco; Teske, Arco J.; Van Herwerden, Lex A.; Chamuleau, Steven; Meijboom, Folkert; Budde, Ricardo P J; Cramer, MJ
Background Two-dimensional transthoracic and transesophageal echocardiography (2DTTE and 2DTEE) may fail to detect signs of prosthetic heart valve (PHV) endocarditis due to acoustic shadowing. Three-dimensional (3D) TEE may have additional value; however, data are scarce. This study was performed to
Lærum Sibilitz, Kristine; Berg, Selina Kikkenborg; Tang, Lars Hermann
This is the protocol for a review and there is no abstract. The objectives are as follows: To assess the benefits and harms of exercise-based intervention programmes (exercise-based interventions alone or in combination with psycho-educational components), compared to no intervention, or treatment...... as usual, in adults who have had heart valve surgery. In this review we will focus on programmes that include an exercise-based intervention with, or without, another rehabilitation component (such as a psycho-educational component)....
Hansen, Tina B; Berg, Selina K; Sibilitz, Kirstine L
in a cardiac rehabilitation programme, and none have analysed their experiences with it. AIMS: The purpose of this qualitative analysis was to gain insight into patients' experiences in cardiac rehabilitation, the CopenHeartVR trial. This trial specifically assesses patients undergoing isolated heart valve...... to take active personal responsibility for their health. Despite these benefits, participants experienced existential and psychological challenges and musculoskeletal problems. Participants also sought additional advice from healthcare professionals both inside and outside the healthcare system....... CONCLUSIONS: Even though the cardiac rehabilitation programme reduced insecurity and helped participants take active personal responsibility for their health, they experienced existential, psychological and physical challenges during recovery. The cardiac rehabilitation programme had several limitations...
... stenosis. Tricuspid Regurgitation Tricuspid regurgitation is also called tricuspid insufficiency or tricuspid incompetence. It means there is a ... require valve surgery. Tags: heart valves , tricuspid incompetence , ... tricuspid regurgitation , tricuspid stenosis , valve disease Related Links ...
Chhatriwalla, Adnan K; Allen, Keith B; Saxon, John T; Cohen, David J; Aggarwal, Sanjeev; Hart, Anthony J; Baron, Suzanne J; Dvir, Danny; Borkon, A Michael
Valve-in-valve (VIV) transcatheter aortic valve replacement (TAVR) may be less effective in small surgical valves because of patient/prosthesis mismatch. Bioprosthetic valve fracture (BVF) using a high-pressure balloon can be performed to facilitate VIV TAVR. We report data from 20 consecutive clinical cases in which BVF was successfully performed before or after VIV TAVR by inflation of a high-pressure balloon positioned across the valve ring during rapid ventricular pacing. Hemodynamic measurements and calculation of the valve effective orifice area were performed at baseline, immediately after VIV TAVR, and after BVF. BVF was successfully performed in 20 patients undergoing VIV TAVR with balloon-expandable (n=8) or self-expanding (n=12) transcatheter valves in Mitroflow, Carpentier-Edwards Perimount, Magna and Magna Ease, Biocor Epic and Biocor Epic Supra, and Mosaic surgical valves. Successful fracture was noted fluoroscopically when the waist of the balloon released and by a sudden drop in inflation pressure, often accompanied by an audible snap. BVF resulted in a reduction in the mean transvalvular gradient (from 20.5±7.4 to 6.7±3.7 mm Hg, P valve effective orifice area (from 1.0±0.4 to 1.8±0.6 cm 2 , P valves to facilitate VIV TAVR with either balloon-expandable or self-expanding transcatheter valves and results in reduced residual transvalvular gradients and increased valve effective orifice area. © 2017 American Heart Association, Inc.
Full Text Available Xenogenic leaflet material, bovine and porcine pericardium, is widely used for the fabrication of surgically implanted and transcatheter heart valve prostheses. As a biological material, long term durability of pericardium is limited due to calcification, degeneration and homogeneity. Therefore, polymeric materials represent a promising approach for a next generation of artificial heart valve leaflets with improved durability. Within the current study we analyzed the mechanical performance of polymeric structures based on elastomeric materials. Polymeric cast films were prepared and nonwovens were manufactured in an electrospinning process. Analysis of cyclic stress-strain behavior was performed, using a universal testing machine. The uniaxial cyclic tensile experiments of the elastomeric samples yielded a non-linear elastic response due to viscoelastic behavior with hysteresis. Equilibrium of stress-strain curves was found after a specific number of cycles, for cast films and nonwovens, respectively. In conclusion, preconditioning was found obligatory for the evaluation of the mechanical performance of polymeric materials for the use as artificial leaflet material for heart valve prostheses.
Full Text Available We attempted to investigate cardioprotection of electroacupuncture (EA for enhanced recovery after surgery on patients undergoing heart valve replacement with cardiopulmonary bypass. Forty-four patients with acquired heart valve replacement were randomly allocated to the EA group or the control group. Patients in the EA group received EA stimulus at bilateral Neiguan (PC6, Ximen (PC4, Shenting (GV24, and Baihui (GV20 acupoints twenty minutes before anesthesia induction to the end of surgery. The primary end point was cardioprotection effect of electroacupuncture postoperatively and the secondary endpoints were quality of recovery and cognitive functioning postoperatively. The present study demonstrated that electroacupuncture reduced the occurrence of complications and played a role of cardioprotective effect on patients after heart valve replacement surgery with cardiopulmonary bypass, and it benefits patients more comfortable and contributes to recovery after surgery. This trial is registered with ChiCTR-IOC-16009123.
Adrie, Christophe; Parlato, Marianna; Salmi, Lynda; Adib-Conquy, Minou; Bical, Olivier; Deleuze, Philippe; Fitting, Catherine; Cavaillon, Jean Marc; Monchi, Mehran
To determine whether the good safety profile of transarterial aortic valve implantation (TAVI) is related to lower levels of systemic bacterial translocation and systemic inflammation compared with open-heart surgery. Transcatheter aortic valve implantation via the transfemoral approach is increasingly used in very high-risk patients with aortic stenosis. The outcomes seem similar to those after open-heart aortic valve replacement (OHAVR). Each of 26 consecutive high-risk patients (EuroSCORE >20% for risk of operative death) who underwent TAVI (cases) was matched to the first low-risk patient treated next in our department using elective OHAVR without coronary artery bypass (control subjects). We collected severity, outcome, and echocardiography indicators before and after surgery; complications; proinflammatory cytokine levels; and markers for microbial translocation. Despite greater illness severity, the TAVI patients had significantly lower vasopressor agent requirements, lower delirium rates, shorter hospital stays, and better hemodynamic findings compared with OHAVR patients. Vascular complications were more common after TAVI than after OHAVR (12, with seven requiring interventional therapy vs. 0, P = 0.006). Patients who underwent TAVI had lower blood transfusion requirements. Two TAVI patients died: one from iliac artery injury and the other from intracardiac prosthesis migration. Patients who underwent TAVI had lower plasma levels of endotoxin and bacterial peptidoglycan, as well as lower proinflammatory cytokine levels, suggesting less gastrointestinal bacterial translocation compared with OHAVR. Compared with OHAVR, TAVI was associated with decreases in bacterial translocation and inflammation. These differences may explain the lower delirium rate and better hemodynamic stability observed, despite the greater disease severity in TAVI patients.
Ozkan, Mehmet; Gündüz, Sabahattin; Yildiz, Mustafa; Duran, Nilüfer Eksi
Prosthetic heart valve obstruction (PHVO) caused by pannus formation is an uncommon but serious complication. Although two-dimensional transesophageal echocardiography (2D-TEE) is the method of choice in the evaluation of PHVO, visualization of pannus is almost impossible with 2D-TEE. While demonstrating the precise aetiology of PHVO is essential for guiding the therapy, either thrombolysis for valve thrombosis or surgery for pannus formation, more sophisticated imaging techniques are needed in patients with suspected pannus formation. We present real-time 3D-TEE imaging in a patient with mechanical mitral PHVO, clearly demonstrating pannus overgrowth.
Mirkhani, S.H.; Golestani, M.G.; Hosini, M.; Kazemian, A.
There are some reasons for malfunction of aortic and pulmonary valve like fibrosis, calcification, and atheroma. Although, in some papers fenestration were known as a pathologic sign, but it is not generally accepted, while this matter is important in choosing suitable Homograft Heart Valve. In this paper fenestrations and its size, numbers and situation effect was studied. We collected 98 hearts, the donors died because of accident, we excluded valves with atheroma, calcification, fibrosis and unequal cusps, 91 aortic and 93 pulmonary valves were given further consideration. We classified valves according to situation, number and size of fenestration. Each valve was tested with 104 cm of non-nal saline column pressure which is equal to 76 mm Hg. Valve efficacy was detected by fluid flow assay. With study of 184 valves, 95 had no fenestration, 64 had less than 2 fenestration and 25 had more than 2 fenestration. Valve efficacy in condition of less than 2 fenestration was more than others (p <0.01). Malfunction effects of fenestration increased in larger valve and it will be decreased if their situation would be marginal (free margin of cusp). In the comparison of aortic and pulmonary valve we saw that malfunction effect of fenestration in pulmonary valve was more than aortic valve. Our experience in Immam Khomeini Homograft Valve Bank has shown that a great deal of valves is fenestrated. It seems that fenestration must be considered as a quality criterion in homograft valve preparation, especially in pulmonary and large aortic valves; but complementary studies is necessary
Mart, Christopher Robin; Eckhauser, Aaron Wesley; Murri, Michael; Su, Jason Thomas
With surgical palliation of hypoplastic left heart syndrome (HLHS), the tricuspid valve (TV) becomes the systemic atrioventricular valve and moderate/severe TV insufficiency (TVI), an adverse risk factor for survival to Fontan, has been reported in up to 35% of patients prior to stage I palliation. Precise echocardiographic identification of the mechanism of TVI cannot be determined by two-dimensional echocardiography. Three-dimensional echocardiography (3DE) can provide significant insight i...
В. Г. Постнов
Full Text Available Reviewed in this paper are modern approaches in the intensive therapy of acute hypoxic encephalopathy developing in a number of occasions after the heart valve replacement surgery. The study is based on the results of neurological, neuropsychological and neurophysiological (EEG examinations of 240 patients who underwent heart valve replacement surgery under cardiopulmonary bypass conditions complicated later by the development of hypoxic encephalopathies of varying severity and who received complex intensive care. Relying on many years of experience in the treatment of heart surgery patients in whom manifestations of encephalopathy developed in the early postoperative period, or were delayed, we have formulated the following algorithms of therapy. (1 Maintenance of normal blood gas: Hb>100 g/L, pH 7.45, PaCO2 35 mmHg. (2 Maintenance of hemodynamics: ABPsystolic>90 mmHg. (3 Supplying fluids and electrolytes: isoosmolar infusion solutions, adding of KCl and MgSO4 to the infusion. (4 Antiedemic therapy: 15% mannitol or 40% glycerol solution. (5 If necessary (in case of psychomotor agitation, seizures, short-acting barbiturates (sodium thiopental, neuroleptics (haloperidol, propofol. No benzodiazepines in case of psychoses (6 Cerebral metabolism stimulation (not earlier than 48 hours after surgery with cholinomimetics, nootropics, cerebral blood flow protectors. Cholinomimetics are allowed on the first day after surgery. This algorithm and the above-mentioned groups of drugs, especially central cholinomimetics, allow for correcting the neurocognitive impairment in the discussed group of patients quickly and effectively.
Sibilitz, Kirstine Laerum; Berg, Selina Kikkenborg; Hansen, Tina Birgitte
, either valve replacement or repair, remains the treatment of choice. However, post-surgery, the transition to daily living may become a physical, mental and social challenge. We hypothesize that a comprehensive cardiac rehabilitation program can improve physical capacity and self-assessed mental health...... and reduce hospitalization and healthcare costs after heart valve surgery. METHODS: This randomized clinical trial, CopenHeartVR, aims to investigate whether cardiac rehabilitation in addition to usual care is superior to treatment as usual after heart valve surgery. The trial will randomly allocate 210...... patients 1:1 to an intervention or a control group, using central randomization, and blinded outcome assessment and statistical analyses. The intervention consists of 12 weeks of physical exercise and a psycho-educational intervention comprising five consultations. The primary outcome is peak oxygen uptake...
Qian, Jin-yuan; Wei, Lin; Jin, Zhi-jiang; Wang, Jian-kai; Zhang, Han; Lu, An-le
Highlights: • PCGV utilizes pressure difference to control the action of the valve core. • Three different opening processes with the same spring stiffness are analyzed. • Valve core’s displacements with different spring stiffness are analyzed. • The best design point of spring stiffness and inlet pressure is obtained. • The selection formula for the design of PCGV is generalized. - Abstract: The pilot-control globe valve (PCGV) is a new kind valve with simple structures and low driving energy consumption. It can utilize the pressure difference before and after the valve to control the action of the valve core. However, systematic theoretical research and numerical analysis are deficient at present. In this paper, the mathematical model of PCGV is established and Computational Fluid Dynamics (CFD) method is employed to numerically simulate its dynamic characteristics. Through the analysis of the internal flow field distribution, its working principle is verified. Then three different opening processes with the same spring stiffness are analyzed under different static inlet pressures, and the best design point is obtained by studying the characteristic curves of the valve core’s displacement. The relationship of static inlet pressure and the valve core’s displacement is summarized and the selection formula for the valve design is generalized which can reduce the various design work for further optimization and engineering applications of PCGV
Full Text Available The aim – to determine factors that may influence the global longitudinal strain (GLS and its dynamics after replacement aortic valve (RAV in patients with severe aortic stenosis (AS. Materials and methods. In a one-center prospective study, 30 patients with severe AS were sequentially screened for RAV surgery. Among them there were 16 (53,3 % men and 14 (46,7 % women. The median age was 63 (range from 46 to 77, quartiles 57–69 years. All patients underwent clinical and instrumental study, including transthoracic echocardiography (TTE and coronary angiography. In the early postoperative period and 6–12 months after surgery, all patients performed a repeat TTE according to a standard protocol that included an assessment of the GLS. GLS data and their dynamics was performed in the groups of patients with preserved (n = 18 and low (n = 12 left ventricular ejection fraction (LV EF. Results. In patients with preserved LV EF, the median of the preoperative GLS was –12 % (in the quartiles from –15 to –11 %, postoperative GLS –11 % (quartiles from –11 to –9 %. In patients with reduced LV EF, the median of the preoperative GLS was –6 % (from –8.5 to –4 % for quartiles, –8.5 % for postoperative GLS (from –10 to –6 % for quartiles (p < 0.001 for comparison of baseline and p < 0.05 relative to differences in dynamics of GLS in comparable groups. In the examined patients, significant correlations were found between baseline GLS with functional class of heart failure, concomitant tricuspid insufficiency, atrial fibrillation, duration of QRS complex, LV EF, end-diastolic, endsystolic volume index (EDV, ESV and volume left atrium (LA, left ventricular myocardial mass index, aortic valve orifice area index, E/A ratio, and Thei index. In turn, the positive dynamics of GLS significantly depended on the initial functional class of heart failure, the stage of arterial hypertension, the degree of tricuspid insufficiency, LV EF, the index
Ahmed N. U.
Full Text Available In this paper we discuss some problems arising in mathematical modeling of artificial hearts. The hydrodynamics of blood flow in an artificial heart chamber is governed by the Navier-Stokes equation, coupled with an equation of hyperbolic type subject to moving boundary conditions. The flow is induced by the motion of a diaphragm (membrane inside the heart chamber attached to a part of the boundary and driven by a compressor (pusher plate. On one side of the diaphragm is the blood and on the other side is the compressor fluid. For a complete mathematical model it is necessary to write the equation of motion of the diaphragm and all the dynamic couplings that exist between its position, velocity and the blood flow in the heart chamber. This gives rise to a system of coupled nonlinear partial differential equations; the Navier-Stokes equation being of parabolic type and the equation for the membrane being of hyperbolic type. The system is completed by introducing all the necessary static and dynamic boundary conditions. The ultimate objective is to control the flow pattern so as to minimize hemolysis (damage to red blood cells by optimal choice of geometry, and by optimal control of the membrane for a given geometry. The other clinical problems, such as compatibility of the material used in the construction of the heart chamber, and the membrane, are not considered in this paper. Also the dynamics of the valve is not considered here, though it is also an important element in the overall design of an artificial heart. We hope to model the valve dynamics in later paper.
Geemen, van D.; Driessen - Mol, A.; Grootzwagers, L.G.M.; Soekhradj - Soechit, R.S.; Riem Vis, P.W.; Baaijens, F.P.T.; Bouten, C.V.C.
AIM: Clinical application of tissue engineered heart valves requires precise control of the tissue culture process to predict tissue composition and mechanical properties prior to implantation, and to understand the variation in tissue outcome. To this end we investigated cellular phenotype and
Full Text Available This study is based on the findings of a single surgeon’s practice of mitral valve replacement of 167 patients from April 2005 to June 2017 who developed symptomatic mitral restenosis after closed or open mitral commisurotomy. Both clinical and color doppler echocardiographic data of peri-operative and six months follow-up period were evaluated and compared to assess the early outcome of the redo mitral valve surgery. With male-female ratio of 1: 2.2 and after a duration of 6 to 22 years symptom free interval between the redo procedures, the selected patients with mitral valve restenosis undergone valve replacement with either mechanical valve in 62% cases and also tissue valve in 38% cases. Particular emphasis was given to separate the adhered pericardium from the heart completely to ameliorate base to apex and global contraction of the heart. Besides favorable post-operative clinical outcome, the echocardiographic findings were also encouraging as there was statistically significant increase in the mitral valve area and ejection fraction with significant decrease in the left atrial diameter, pressure gradient across the mitral valve and pulmonary artery systolic pressure. Therefore, in case of inevitable mitral restenosis after closed or open commisurotomy, mitral valve replacement is a promising treatment modality.
Signorello, L B; Kennedy, J A; Richmond, R A; Sieu, K L; Blot, W J; Harrison, D C
The risk of fracture of Björk-Shiley convexo-concave (BSCC) prosthetic heart valves has resulted in consideration of prophylactic explantation and replacement for patients with high-risk valves. Little information exists on perceived quality of life, health status, and serious morbidity among BSCC patients, including those who have undergone explantation. Self-administered questionnaires were completed by a cohort of 585 BSCC patients who participated in an X-ray imaging study to detect precursors to valve fracture up to seven years (average 3.9 years) previously. Responses from 31 explant patients were contrasted with those from 554 BSCC patients in whom explant surgery was not attempted. Perceived quality of life and health status and risk of hospitalization after participating in the imaging study varied considerably among patients, but on average tended not to differ significantly between those with and without explants. A slightly greater proportion of explantees tended to report both improved health status and high rates of heart attack and pacemaker implantation. The health status of these patients was, in general, considerably worse than previously reported among valve implant patients. Over half the cohort were hospitalized during follow up, and half were unable to walk up more than one flight of stairs without shortness of breath. The less than optimal health status of most BSCC patients and relatively high rates of morbidity should be taken into account when considering potential explantation of the valves.
Winroth, P. M.; Ford, C. L.; Alfredsson, P. H.
Simplified flow models are commonly used to design and optimize internal combustion engine systems. The exhaust valves and ports are modelled as straight pipe flows with a corresponding discharge coefficient. The discharge coefficient is usually determined from steady-flow experiments at low pressure ratios and at fixed valve lifts. The inherent assumptions are that the flow through the valve is insensitive to the pressure ratio and may be considered as quasi-steady. The present study challenges these two assumptions through experiments at varying pressure ratios and by comparing measurements of the discharge coefficient obtained under steady and dynamic conditions. Steady flow experiments were performed in a flow bench, whereas the dynamic measurements were performed on a pressurized, 2 l, fixed volume cylinder with one or two moving valves. In the latter experiments an initial pressure (in the range 300-500 kPa) was established whereafter the valve(s) was opened with a lift profile corresponding to different equivalent engine speeds (in the range 800-1350 rpm). The experiments were only concerned with the blowdown phase, i.e. the initial part of the exhaustion process since no piston was simulated. The results show that the process is neither pressure-ratio independent nor quasi-steady. A measure of the "steadiness" has been defined, relating the relative change in the open flow area of the valve to the relative change of flow conditions in the cylinder, a measure that indicates if the process can be regarded as quasi-steady or not.
Martínez, G J; Seco, M; Jaijee, S K; Adams, M R; Cartwright, B L; Forrest, P; Celermajer, D S; Vallely, M P; Wilson, M K; Ng, M K C
Transcatheter aortic valve implantation (TAVI) has been developed to treat symptomatic aortic stenosis in patients deemed too high risk for open-heart surgery. To address this complex population, an interdisciplinary heart team approach was proposed. Present the short- and mid-term outcomes of the first 100 patients in the Royal Prince Alfred Hospital multidisciplinary TAVI programme. Single-centre registry. Baseline and procedural data were prospectively recorded. Outcomes were recorded according to Valve Academic Research Consortium - version 2 guidelines. All patients underwent a comprehensive interdisciplinary pre-procedural evaluation. Sixty-eight transfemoral and 32 transapical implantations were performed. Mean age was 82 (±8.9) years old with an average logistic EuroSCORE of 33. Although 13 procedures had major complications, there was no intraprocedural mortality. During the first month, 9% of patients were re-admitted due to heart failure and 13% had a permanent pacemaker implanted. A 3% 30-day and 8% follow-up (mean 17 months) mortalities were recorded. While no significant differences in the rate of complications were found between the first and second half of the experience, all cases of mortality within 30 days (n = 3) occurred in the initial half. Sustained haemodynamic results were obtained with TAVI (immediate mean aortic valve gradient reduction from 47 to 9 mmHg; 1-year echocardiographic gradient 9.9 mmHg, with no moderate or severe aortic regurgitation). Excellent results can be achieved with TAVI in very high-risk patients at an Australian institution. A comprehensive evaluation based on a heart team can overcome most of the difficulties imposed by this challenging population. © 2014 The Authors; Internal Medicine Journal © 2014 Royal Australasian College of Physicians.
Full Text Available Abstract Background Valve dysfunction is a common cardiovascular pathology. Despite significant clinical research, there is little formal study of how valve dysfunction affects overall circulatory dynamics. Validated models would offer the ability to better understand these dynamics and thus optimize diagnosis, as well as surgical and other interventions. Methods A cardiovascular and circulatory system (CVS model has already been validated in silico, and in several animal model studies. It accounts for valve dynamics using Heaviside functions to simulate a physiologically accurate "open on pressure, close on flow" law. However, it does not consider real-time valve opening dynamics and therefore does not fully capture valve dysfunction, particularly where the dysfunction involves partial closure. This research describes an updated version of this previous closed-loop CVS model that includes the progressive opening of the mitral valve, and is defined over the full cardiac cycle. Results Simulations of the cardiovascular system with healthy mitral valve are performed, and, the global hemodynamic behaviour is studied compared with previously validated results. The error between resulting pressure-volume (PV loops of already validated CVS model and the new CVS model that includes the progressive opening of the mitral valve is assessed and remains within typical measurement error and variability. Simulations of ischemic mitral insufficiency are also performed. Pressure-Volume loops, transmitral flow evolution and mitral valve aperture area evolution follow reported measurements in shape, amplitude and trends. Conclusions The resulting cardiovascular system model including mitral valve dynamics provides a foundation for clinical validation and the study of valvular dysfunction in vivo. The overall models and results could readily be generalised to other cardiac valves.
Said, Sameh M; Dearani, Joseph A; Burkhart, Harold M; Connolly, Heidi M; Eidem, Ben; Stensrud, Paul E; Schaff, Hartzell V
Tricuspid valve (TV) regurgitation in congenital heart disease includes a heterogeneous group of lesions, and few series have documented the outcomes. We reviewed the records of 553 patients with congenital heart disease who had undergone TV surgery for tricuspid regurgitation from January 1993 to December 2010. Patients with Ebstein malformation were excluded. Their mean age was 32 ± 21 years, and 300 were female (54%). The most common diagnoses were conotruncal anomaly in 216 patients (39%), previous ventricular septal defect closure in 83 (15%), atrioventricular septal defect in 77 (14%), and pulmonary atresia with an intact ventricular septum in 11 (2%). Preoperative right-sided heart failure was present in 124 patients (22%), and 55 patients (10%) had pulmonary hypertension. TV repair was performed in 442 (80%) and TV replacement in 111 (20%) patients. Repeat sternotomy was performed in 415 patients (75%). Previous TV repair was present in 44 patients (8%); of these, 17 (38.6%) underwent repeat TV repair. The overall early mortality was 3.1% (17 patients) and was 2.5% for TV repair and 5.4% for TV replacement (P = .001). The mean follow-up period was 4.5 ± 4.1 years (maximum, 18). The overall survival at 1, 5, and 10 years was 97%, 93%, and 85%, respectively. Survival was better for patients with repair than with replacement. TV repair was an independent predictor of better survival (P = .001). Important tricuspid regurgitation can occur with a variety of congenital diagnoses. Early mortality is low and late survival is superior with tricuspid repair than with valve replacement. Surgical treatment of tricuspid regurgitation in congenital heart disease should be performed before the onset of heart failure. Copyright © 2014 The American Association for Thoracic Surgery. Published by Mosby, Inc. All rights reserved.
Full Text Available A 25-year-old man presented with a history of breathlessness for the past 2 years. He had a history of operation for Tetralogy of Fallot at the age of 5 years and history suggestive of Rheumatic fever at the age of 7 years. On echocardiographic examination, all his heart valves were severely regurgitating. Morphologically, all the valves were irreparable. The ejection fraction was 35%. He underwent quadruple valve replacement. The aortic and mitral valves were replaced by metallic valve and the tricuspid and pulmonary by tissue valve.
Determinants and outcomes of acute transcatheter valve-in-valve therapy or embolization: a study of multiple valve implants in the U.S. PARTNER trial (Placement of AoRTic TraNscathetER Valve Trial Edwards SAPIEN Transcatheter Heart Valve).
Makkar, Raj R; Jilaihawi, Hasan; Chakravarty, Tarun; Fontana, Gregory P; Kapadia, Samir; Babaliaros, Vasilis; Cheng, Wen; Thourani, Vinod H; Bavaria, Joseph; Svensson, Lars; Kodali, Susheel; Shiota, Takahiro; Siegel, Robert; Tuzcu, E Murat; Xu, Ke; Hahn, Rebecca T; Herrmann, Howard C; Reisman, Mark; Whisenant, Brian; Lim, Scott; Beohar, Nirat; Mack, Michael; Teirstein, Paul; Rihal, Charanjit; Douglas, Pamela S; Blackstone, Eugene; Pichard, Augusto; Webb, John G; Leon, Martin B
This study investigated the determinants and outcomes of acute insertion of a second transcatheter prosthetic valve (TV) within the first (TV-in-TV) or transcatheter valve embolization (TVE) after transcatheter aortic valve replacement (TAVR). TAVR failure can occur with both TV-in-TV and TVE as a consequence of TAVR malpositioning. Only case reports and limited series pertaining to these complications have been reported to date. Patients undergoing TAVR in the PARTNER (Placement of AoRTic TraNscathetER Valve Trial Edwards SAPIEN Transcatheter Heart Valve) randomized trial (cohorts A and B) and accompanying registries were studied. Data were dichotomized for those with and without TV-in-TV or TVE, respectively. From a total of 2,554 consecutive patients, 63 (2.47%) underwent TV-in-TV and 26 (1.01%) TVE. The indication for TV-in-TV was significant aortic regurgitation in most patients, often due not only to malpositioning but also to leaflet dysfunction. Despite similar aortic valve function on follow-up echoes, TV-in-TV was an independent predictor of 1-year cardiovascular mortality (hazard ratio [HR]: 1.86, 95% confidence interval [CI]: 1.03 to 3.38, p = 0.041), with a nonsignificant trend toward greater all-cause mortality (HR: 1.43, 95% CI: 0.88 to 2.33, p = 0.15). Technical and anatomical reasons accounted for most cases of TVE. A multivariable analysis found TVE to be an independent predictor of 1-year mortality (HR: 2.68, 95% CI: 1.34 to 5.36, p = 0.0055) but not cardiovascular mortality (HR: 1.30, 95% CI: 0.48 to 3.52, p = 0.60). Acute TV-in-TV and TVE are serious sequelae of TAVR, often resulting in multiple valve implants. They carry an excess of mortality and are caused by anatomic and technical factors, which may be avoidable with judicious procedural planning. Copyright © 2013 American College of Cardiology Foundation. Published by Elsevier Inc. All rights reserved.
Zhang, Penglei; Zhou, Dehai; Shi, Wenxing; Li, Xianting; Wang, Baolong
A hybrid air conditioner combining a thermosyphon cycle with a vapor compression refrigeration cycle has a large energy saving potential compared with a common air conditioner for spaces requiring year-round cooling. The performance of the switch between the vapor compression mode and the thermosyphon mode largely impacts the safety and reliability of hybrid air conditioners. Therefore, a self-operated three-way valve is proposed. A thermodynamic model and a kinetic model are developed in this paper to evaluate the dynamic performance of the switch valve. The effects of the spring force constant, compressor discharging volume, fit clearance and piston length on the dynamic performance of the switch valve are analyzed. In conclusion, the proposed self-operated three-way valve can realize the switch operation accurately. - Highlights: •A self-operated three-way valve is proposed for hybrid air conditioners. •The thermodynamic model and kinetic model of the self-operated three-way valve are developed. •The validity of models is verified by experiments. •Effects of four main design parameters on the operating performance of the valve are researched
... gov/ency/article/007408.htm Aortic valve surgery - open To use the sharing features on this page, ... separates the heart and aorta. The aortic valve opens so blood can flow out. It then closes ...
... each time the left ventricle contracts. Watch an animation of mitral valve regurgitation A leaking mitral valve ... Not Alone Popular Articles 1 Understanding Blood Pressure Readings 2 Sodium and Salt 3 Heart Attack Symptoms ...
Horstkotte, D; Körfer, R; Budde, T; Haerten, K; Schulte, H D; Bircks, W; Loogen, F
Valve-related complications after Björk-Shiley mitral (n = 475), aortic (n = 424), or mitral-aortic implantation (n = 119) were compared to complications after St. Jude mitral (n = 173), aortic (n = 152), and St. Jude mitral and aortic (n = 63) replacements. The 1,018 consecutive patients with Björk-Shiley valves had been operated upon between 1974 and 1982, those with St. Jude valves between 1978 and 1982. All patients were placed on anticoagulant therapy with phenprocoumon early after operation and no significant intergroup differences in the effectiveness of the anticoagulant therapy were found. At a comparable follow-up time of approximately 23 months, 24 major thromboembolic episodes were observed after Björk-Shiley mitral (BSM) and 3 after St. Jude mitral valve implantation (SJM), corresponding to a thromboembolic rate of 2.82/100 patient years with BSM and 0.93/100 patient years with SJM. After aortic valve replacements, 1.93 events in 100 patient years occurred after Björk-Shiley aortic (BSA) and 0.73 after St. Jude aortic implantation (SJA). In patients with double valve replacements, these rates were 3.2 (BSM + BSA) and 0.88 (SJM + SJA), respectively. The cerebral vessels were involved in 52% and the arteries of the extremities in 22% of these major events. Six Björk-Shiley prostheses had to be replaced because of valve thrombosis. The overall incidence of severe hemorrhagic complications was 2.94/100 patient years in BSM and 1.79 in SJM. After aortic valve replacement, we found rates of 1.80/100 patient years (BSA) and 2.57/100 patient years (SJA), respectively. Intravascular hemolysis no longer seems to be a significant clinical problem. However, indications of red cell damage after heart valve replacement were significantly greater in patients with perivalvular leakage, valve thrombosis, or dysfunction than in those with normally functioning prostheses. Reoperations were necessary because of valve thrombosis (0.46%), perivalvular leakage (2
Full Text Available Background. Carcinoid heart disease is a rare heart disease which affects endocard and heart valves on the right side of heart. It affects only patients with manifested carcinoid syndrome, which is thought to be the consequence of secretory active metastases of carcinoid tumour. The carcinoid endocardial plaques cause structural changes of tricuspid and pulmonic valve and later on their stenosis and/or insufficiency.Patients and methods. In this article we introduce a carcinoid valve heart disease (CVHD scoring system for easier end exact echocardiographic diagnostics. Four echocardiographic parameters are beeing estimated: structural changes of tricuspid valve, tricuspid valve regurgitation, stenosis of pulmonic valve and pulmonic valve regurgitation.Conclusions. The scoring system allows us to make an early diagnosis and evaluation of progression of carcinoid heart disease, which is very important for planning the treatment process. Our experiences confirm the usefulness of this scoring system in echocardiographic follow–up of patients with carcinoid syndrome.
Lantz, Jonas; Gupta, Vikas; Henriksson, Lilian; Karlsson, Matts; Persson, Ander; Carhall, Carljohan; Ebbers, Tino
In this study, cardiac blood flow was simulated using Computational Fluid Dynamics and compared to in vivo flow measurements by 4D Flow MRI. In total, nine patients with various heart diseases were studied. Geometry and heart wall motion for the simulations were obtained from clinical CT measurements, with 0.3x0.3x0.3 mm spatial resolution and 20 time frames covering one heartbeat. The CFD simulations included pulmonary veins, left atrium and ventricle, mitral and aortic valve, and ascending aorta. Mesh sizes were on the order of 6-16 million cells, depending on the size of the heart, in order to resolve both papillary muscles and trabeculae. The computed flow field agreed visually very well with 4D Flow MRI, with characteristic vortices and flow structures seen in both techniques. Regression analysis showed that peak flow rate as well as stroke volume had an excellent agreement for the two techniques. We demonstrated the feasibility, and more importantly, fidelity of cardiac flow simulations by comparing CFD results to in vivo measurements. Both qualitative and quantitative results agreed well with the 4D Flow MRI measurements. Also, the developed simulation methodology enables ``what if'' scenarios, such as optimization of valve replacement and other surgical procedures. Funded by the Wallenberg Foundation.
Di Mauro, Michele; Dato, Guglielmo Mario Actis; Barili, Fabio; Gelsomino, Sandro; Santè, Pasquale; Corte, Alessandro Della; Carrozza, Antonio; Ratta, Ester Della; Cugola, Diego; Galletti, Lorenzo; Devotini, Roger; Casabona, Riccardo; Santini, Francesco; Salsano, Antonio; Scrofani, Roberto; Antona, Carlo; Botta, Luca; Russo, Claudio; Mancuso, Samuel; Rinaldi, Mauro; De Vincentiis, Carlo; Biondi, Andrea; Beghi, Cesare; Cappabianca, Giangiuseppe; Tarzia, Vincenzo; Gerosa, Gino; De Bonis, Michele; Pozzoli, Alberto; Nicolini, Francesco; Benassi, Filippo; Rosato, Francesco; Grasso, Elena; Livi, Ugolino; Sponga, Sandro; Pacini, Davide; Di Bartolomeo, Roberto; De Martino, Andrea; Bortolotti, Uberto; Onorati, Francesco; Faggian, Giuseppe; Lorusso, Roberto; Vizzardi, Enrico; Di Giammarco, Gabriele; Marinelli, Daniele; Villa, Emmanuel; Troise, Giovanni; Picichè, Marco; Musumeci, Francesco; Paparella, Domenico; Margari, Vito; Tritto, Francesco; Damiani, Girolamo; Scrascia, Giuseppe; Zaccaria, Salvatore; Renzulli, Attilio; Serraino, Giuseppe; Mariscalco, Giovanni; Maselli, Daniele; Foschi, Massimiliano; Parolari, Alessandro; Nappi, Giannantonio
The aim of this large retrospective study was to provide a logistic risk model along an additive score to predict early mortality after surgical treatment of patients with heart valve or prosthesis infective endocarditis (IE). From 2000 to 2015, 2715 patients with native valve endocarditis (NVE) or prosthesis valve endocarditis (PVE) were operated on in 26 Italian Cardiac Surgery Centers. The relationship between early mortality and covariates was evaluated with logistic mixed effect models. Fixed effects are parameters associated with the entire population or with certain repeatable levels of experimental factors, while random effects are associated with individual experimental units (centers). Early mortality was 11.0% (298/2715); At mixed effect logistic regression the following variables were found associated with early mortality: age class, female gender, LVEF, preoperative shock, COPD, creatinine value above 2mg/dl, presence of abscess, number of treated valve/prosthesis (with respect to one treated valve/prosthesis) and the isolation of Staphylococcus aureus, Fungus spp., Pseudomonas Aeruginosa and other micro-organisms, while Streptococcus spp., Enterococcus spp. and other Staphylococci did not affect early mortality, as well as no micro-organisms isolation. LVEF was found linearly associated with outcomes while non-linear association between mortality and age was tested and the best model was found with a categorization into four classes (AUC=0.851). The following study provides a logistic risk model to predict early mortality in patients with heart valve or prosthesis infective endocarditis undergoing surgical treatment, called "The EndoSCORE". Copyright © 2017. Published by Elsevier B.V.
Blot, W J; Omar, R Z; Kallewaard, M; Morton, L S; Fryzek, J P; Ibrahim, M A; Acheson, D; Taylor, K M; van der Graaf, Y
Approximately 82,000 Björk-Shiley convexo-concave (BSCC) 60 degree prosthetic heart valves were implanted in patients worldwide between 1979 and 1986. Outlet strut fractures (OSF) of some of the valves were first reported shortly after their introduction. Here, the determinants of OSF are examined, and the between-country variation and long-term risk are assessed. Cohorts of patients in the UK, Netherlands and USA with 15,770 BSCC 60 degree heart valves were followed up to 18 years for the occurrence of OSF. Crude rates of OSF were highest in the UK (0.18% per year), intermediate in the Netherlands (0.13%), and lowest in the USA (0.06%), although risk factor adjustment reduced the inter-country differences. Furthermore, in the UK and Netherlands, OSF rates (particularly for mitral valves) declined with time since implantation, and between-country differences were considerably diminished 10 or more years post implantation. The risk of OSF decreased steadily with advancing patient age. Fracture rates were lower among women than men, and also varied significantly with valve size and position and OSF status of other valves in the same shoporder. This long-term follow up of BSCC 60 degree heart valve patients indicates that risk factors for valve fracture are generally similar in the UK, Netherlands and USA. It also identifies a strong association between fracture risk and age, newly reveals gender-related differences, and shows that the risk of valve fracture persisted, albeit at a reduced rate, into the 1990s.
di Labbio, Giuseppe; Vetel, Jerome; Kadem, Lyes
Aortic valve regurgitation is a rather prevalent condition where the aortic valve improperly closes, allowing filling of the left ventricle of the heart to occur partly from backflow through the aortic valve. Although studies of intraventricular flow are rapidly gaining popularity in the fluid dynamics research community, much attention has been given to the left ventricular vortex and its potential for early detection of disease, particularly in the case of dilated cardiomyopathy. Notably, the subsequent flow in the left ventricle in the presence of aortic valve regurgitation ought to be appreciably disturbed and has yet to be described. Aortic valve regurgitation was simulated in vitro in a double-activation left heart duplicator and the ensuing flow was captured using two-dimensional time-resolved particle image velocimetry. Further insight into the regurgitant flow is obtained by computing attracting and repelling Lagrangian coherent structures. An interesting interplay between the two inflowing jets and their shear layer roll-up is observed for various grades of regurgitation. This study highlights flow features which may find use in further assessing regurgitation severity.
Willemink, Martin J; Takx, Richard A P; Išgum, Ivana; de Koning, Harry J; Oudkerk, Matthijs; Mali, Willem P Th M; Budde, Ricardo P J; Leiner, Tim; Vliegenthart, Rozemarijn; de Jong, Pim A
To assess the prognostic value of aortic valve and mitral valve/annulus calcifications for cardiovascular events in heavily smoking men without a history of cardiovascular disease. Heavily smoking men without a cardiovascular disease history who underwent non-contrast-enhanced low-radiation-dose chest CT for lung cancer screening were included. Non-imaging predictors (age, smoking status and pack-years) were collected and imaging-predictors (calcium volume of the coronary arteries, aorta, aortic valve and mitral valve/annulus) were obtained. The outcome was the occurrence of cardiovascular events. Multivariable Cox proportional-hazards regression was used to calculate hazard-ratios (HRs) with 95% confidence interval (CI). Subsequently, concordance-statistics were calculated. In total 3111 individuals were included, of whom 186 (6.0%) developed a cardiovascular event during a follow-up of 2.9 (Q1-Q3, 2.7-3.3) years. If aortic (n = 657) or mitral (n = 85) annulus/valve calcifications were present, cardiovascular event incidence increased to 9.0% (n = 59) or 12.9% (n = 11), respectively. HRs of aortic and mitral valve/annulus calcium volume for cardiovascular events were 1.46 (95% CI, 1.09-1.84) and 2.74 (95% CI, 0.92-4.56) per 500 mm(3). The c-statistic of a basic model including age, pack-years, current smoking status, coronary and aorta calcium volume was 0.68 (95% CI, 0.63-0.72), which did not change after adding heart valve calcium volume. Aortic valve calcifications are predictors of future cardiovascular events. However, there was no added prognostic value beyond age, number of pack-years, current smoking status, coronary and aorta calcium volume for short term cardiovascular events.
Silaschi, Miriam; Treede, Hendrik; Rastan, Ardawan J; Baumbach, Hardy; Beyersdorf, Friedhelm; Kappert, Utz; Eichinger, Walter; Rüter, Florian; de Kroon, Thomas L; Lange, Rüdiger; Ensminger, Stephan; Wendler, Olaf
Transcatheter aortic valve replacement (TAVR) is an established therapy for patients with aortic stenosis (AS) at high surgical risk. The JenaValve™ is a second-generation, self-expanding transcatheter heart valve (THV), implanted through transapical access (TA). During stent deployment, a specific 'clipping-mechanism' engages native aortic valve cusps for fixation. We present 1-year outcomes of the JUPITER registry, a post-market registry of the JenaValve for TA-TAVR. The JUPITER registry is a prospective, multicentre, uncontrolled and observational European study to evaluate the long-term safety and effectiveness of the Conformité Européenne-marked JenaValve THV. A total of 180 patients with AS were enrolled between 2012 and 2014. End-points were adjudicated in accordance with the valve academic research consortium document no. 1 definitions. The mean age was 80.4 ± 5.9 years and the mean logistic European system for cardiac operative risk evaluation I 21.2 ± 14.7%. The procedure was successful in 95.0% (171/180), implantation of a second THV (valve-in-valve) was performed in 2.2% (4/180) and conversion to surgical aortic valve replacement (SAVR) was necessary in 2.8% (5/180). No annular rupture or coronary ostia obstruction occurred. Two patients required SAVR after the day of index procedure (1.1%). All-cause mortality at 30 days was 11.1% (20/180), being cardiovascular in 7.2% (13/180). A major stroke occurred in 1.1% (2/180) at 30 days, no additional major strokes were observed during 1 year. All-cause mortality after 30 days was 13.1% (21/160) and combined efficacy at 1 year was 80.8% (122/151). At 1-year follow-up, no patient presented with more than moderate paravalvular leakage, while 2 patients (3.2%) showed moderate, 12 (19.0%) mild and 49 (82.4%) trace/none paravalvular regurgitation. In a high-risk cohort of patients undergoing TA-TAVR for AS, the use of the JenaValve THV is safe and effective. In patients at higher risk for coronary ostia
Sucha, Dominika; Mali, Willem P.T.M.; Habets, Jesse [University Medical Center Utrecht, Department of Radiology, Utrecht (Netherlands); Chamuleau, Steven A.J. [University Medical Center Utrecht, Department of Cardiology, Utrecht (Netherlands); Symersky, Petr [VU Medical Center, Department of Cardiothoracic Surgery, Amsterdam (Netherlands); Meijs, Matthijs F.L. [Thoraxcentrum Twente, Medisch Spectrum Twente, Department of Cardiology, Enschede (Netherlands); Brink, Renee B.A. van den [Academic Medical Center, Department of Cardiology, Amsterdam (Netherlands); Mol, Bas A.J.M. de [Academic Medical Center, Department of Cardiothoracic Surgery, Amsterdam (Netherlands); Herwerden, Lex A. van [University Medical Center Utrecht, Department of Cardiothoracic Surgery, Utrecht (Netherlands); Budde, Ricardo P.J. [University Medical Center Utrecht, Department of Radiology, Utrecht (Netherlands); Erasmus University Medical Center, Department of Radiology, Rotterdam (Netherlands)
Recent studies have proposed additional multidetector-row CT (MDCT) for prosthetic heart valve (PHV) dysfunction. References to discriminate physiological from pathological conditions early after implantation are lacking. We present baseline MDCT findings of PHVs 6 weeks post implantation. Patients were prospectively enrolled and TTE was performed according to clinical guidelines. 256-MDCT images were systematically assessed for leaflet excursions, image quality, valve-related artefacts, and pathological and additional findings. Forty-six patients were included comprising 33 mechanical and 16 biological PHVs. Overall, MDCT image quality was good and relevant regions remained reliably assessable despite mild-moderate PHV-artefacts. MDCT detected three unexpected valve-related pathology cases: (1) prominent subprosthetic tissue, (2) pseudoaneurysm and (3) extensive pseudoaneurysms and valve dehiscence. The latter patient required valve surgery to be redone. TTE only showed trace periprosthetic regurgitation, and no abnormalities in the other cases. Additional findings were: tilted aortic PHV position (n = 3), pericardial haematoma (n = 3) and pericardial effusion (n = 3). Periaortic induration was present in 33/40 (83 %) aortic valve patients. MDCT allowed evaluation of relevant PHV regions in all valves, revealed baseline postsurgical findings and, despite normal TTE findings, detected three cases of unexpected, clinically relevant pathology. (orig.)
Zhao, Li; Chen, Chunxia; Li, Bei; Dong, Li; Guo, Yingqiang; Xiao, Xijun; Zhang, Eryong; Qin, Li
To study the performance of pharmacogenetics-based warfarin dosing algorithms in the initial and the stable warfarin treatment phases in a cohort of Han-Chinese patients undertaking mechanic heart valve replacement. We searched PubMed, Chinese National Knowledge Infrastructure and Wanfang databases for selecting pharmacogenetics-based warfarin dosing models. Patients with mechanic heart valve replacement were consecutively recruited between March 2012 and July 2012. The predicted warfarin dose of each patient was calculated and compared with the observed initial and stable warfarin doses. The percentage of patients whose predicted dose fell within 20% of their actual therapeutic dose (percentage within 20%), and the mean absolute error (MAE) were utilized to evaluate the predictive accuracy of all the selected algorithms. A total of 8 algorithms including Du, Huang, Miao, Wei, Zhang, Lou, Gage, and International Warfarin Pharmacogenetics Consortium (IWPC) model, were tested in 181 patients. The MAE of the Gage, IWPC and 6 Han-Chinese pharmacogenetics-based warfarin dosing algorithms was less than 0.6 mg/day in accuracy and the percentage within 20% exceeded 45% in all of the selected models in both the initial and the stable treatment stages. When patients were stratified according to the warfarin dose range, all of the equations demonstrated better performance in the ideal-dose range (1.88-4.38 mg/day) than the low-dose range (pharmacogenetics-based warfarin dosing regimens performed similarly in our cohort. However, the algorithms of Wei, Huang, and Miao showed a better potential for warfarin prediction in the initial and the stable treatment phases in Han-Chinese patients undertaking mechanic heart valve replacement.
Cabrera, M.S.; Oomens, C.W.J.; Baaijens, F.P.T.
A proper interpretation of the forces developed during stent crimping and deployment is of paramount importance for a better understanding of the requirements for successful heart valve replacement. The present study combines experimental and computational methods to assess the performance of a
Cabrera, María Sol; Oomens, Cees W J; Baaijens, Frank P T
A proper interpretation of the forces developed during stent crimping and deployment is of paramount importance for a better understanding of the requirements for successful heart valve replacement. The present study combines experimental and computational methods to assess the performance of a nitinol stent for tissue-engineered heart valve implantation. To validate the stent model, the mechanical response to parallel plate compression and radial crimping was evaluated experimentally. Finite element simulations showed good agreement with the experimental findings. The computational models were further used to determine the hoop force on the stent and radial force on a rigid tool during crimping and self-expansion. In addition, stent deployment against ovine and human pulmonary arteries was simulated to determine the hoop force on the stent-artery system and the equilibrium diameter for different degrees of oversizing. Copyright © 2017 The Authors. Published by Elsevier Ltd.. All rights reserved.
Sechtem, U.; Welslau, R.; Hilger, H.H.; Theissen, P.; Jungehuelsing, M.; Schicha, H.
In spite of the great number of imaging procedures available, it is still difficult to quantify the severity of valvular heart disease. Dynamic MRI offers new approaches to visualize the turbulent blood flow through stenosed or insufficient heart valves. In addition, it is possible to assess the severity of valvular insufficiency based on precise measurements of rigth- and left-ventricular stroke volumes or of systemic and pulmonary flow. Valvular stenoses are difficult to quantify by MRI because flow velocity measurements based on phase analysis can only be made at low flow rates at present. Some progress may be achieved by further shortening of echo times. In patients with cardiac valve replacement MRI is often superior to other imaging procedures because variable imaging planes facilitate differentiation between transvalvular and paravalvular leaks. Additionally, the severity of valvular incompetence can be assessed in such cases in the same way as in patients with insufficiency of the native heart valve. (orig.) [de
Alskaf, Ebraham; McConkey, Hannah; Laskar, Nabila; Kardos, Attila
The Medtronic ATS Open Pivot mechanical valve has been successfully used in heart valve surgery for more than two decades. We present the case of a patient who, 19 years following a tricuspid valve replacement with an ATS prosthesis as part of a triple valve operation following infective endocarditis, developed severe tricuspid regurgitation due to pannus formation.
Tulke, K.D.; Stoppler, W.; Stern, G.
Thermal tensile tests were performed at a temperature of 270 C, with two new original insulation valve conical screws M30-Tx92,5 mm (material: 21 CrMo V 5 7)and two prestrained ones during the event on 27.12.92. In order to assure the results obtained with regard to the dynamic load on the insulation valve during ''quick opening'', in addition tensile impact tests were performed at 270 C with six original insulation valve conical screws. Impact velocity reached 13,5 m/s at four screws and 6 m/s at two screws. Test conditions regarding collision damping and mass distribution were adapted, by means of parameter studies, to the situation of the insulation valve. During thermal tensile tests, strength and deformation values, such as stress at flow start, tensile strength, fracture prolongation and strain, necking at fracture as well as energy absorption up to maximum force and up to rupture, were determined. During tensile impact tests, deformation values, such as elongation, strain and necking, and energy absorption by the screw, were determined. (orig.) [de
Buelow, Matthew W; Earing, Michael G; Hill, Garick D; Cohen, Scott B; Bartz, Peter J; Tweddell, James S; Ginde, Salil
The impact of obesity on surgical morbidity in adults with congenital heart disease is currently unknown. The aim of our study was to investigate the impact of obesity on postoperative outcomes in adults with congenital heart disease undergoing reoperation for pulmonary valve replacement. A retrospective analysis was performed assessing the influence of obesity on surgical outcomes. Obesity was defined as a body mass index ≥30 kg/m2. The mean body mass index of the cohort was 25.9 ± 6.9 kg/m2 . The cohort included 71 patients with 17 patients (24%) being obese. There was no postoperative mortality. Obese patients had a longer hospital length of stay (6.6 vs. 4.7 days; P obesity was independently associated with hospital length of stay >5 days (odds ratio [OR] = 5.2; 95% confidence interval [CI]: 1.5-18.2, P = .01) and with increased postoperative arrhythmias (OR = 4.2; 95% CI: 1.7-40, P Obesity is associated with increased morbidity in adults with congenital heart disease undergoing pulmonary valve replacement, including longer hospitalization and higher risk for postoperative arrhythmias. © 2015 Wiley Periodicals, Inc.
Zhang, L; Zheng, X; Long, Y; Wu, M; Chen, Y; Yang, J; Liu, Z; Zhang, Z
Essentials Low anticoagulation intensity reduces bleeding but increases thrombosis during warfarin therapy. Elevated D-dimer level is associated with increased thrombosis events. D-dimer can be used to find potential thrombosis in those receiving low intensity therapy. D-dimer-guided therapy may be the optimal strategy for those with mechanical heart valve replacement. Background Controversies remain regarding the optimal anticoagulation intensity for Chinese patients after mechanical heart valve replacement despite guidelines having recommended a standard anticoagulation intensity. Objectives To investigate whether D-dimer could be used to determine the optimal anticoagulation intensity in Chinese patients after mechanical heart valve replacement. Patients/Methods This was a prospective, randomized controlled clinical study. A total of 748 patients following mechanical heart valve replacement in Wuhan Asia Heart Hospital were randomized to three groups at a ratio of 1 : 1 : 1. Patients in two control groups received warfarin therapy based on constant standard intensity (international normalized ratio [INR], 2.5-3.5; n = 250) and low intensity (INR, 1.8-2.6; n = 248), respectively. In the experimental group (n = 250), warfarin therapy was initiated at low intensity, then those with elevated D-dimer levels were adjusted to standard intensity. All patients were followed-up for 24 months until the occurrence of endpoints, including bleeding events, thrombotic events and all-cause mortality. Results A total of 718 patients were included in the analysis. Fifty-three events occurred during follow-up. There was less hemorrhage (3/240 vs. 16/241; hazard ratio [HR], 0.18; 95% confidence interval [CI], 0.07-0.45) and all-cause mortality (4/240 vs. 12/241; HR, 0.33; 95% CI, 0.12-0.87) observed in the D-dimer-guided group than in the standard-intensity group. A lower incidence of thrombotic events was also observed in the D-dimer-guided group when compared with the
Saxon, John T; Allen, Keith B; Cohen, David J; Chhatriwalla, Adnan K
Valve-in-valve (VIV) transcatheter aortic valve replacement (TAVR) has been established as a safe and effective means of treating failed surgical bioprosthetic valves (BPVs) in patients at high risk for complications related to reoperation. Patients who undergo VIV TAVR are at risk of patient-prosthesis mismatch, as the transcatheter heart valve (THV) is implanted within the ring of the existing BPV, limiting full expansion and reducing the maximum achievable effective orifice area of the THV. Importantly, patient-prosthesis mismatch and high residual transvalvular gradients are associated with reduced survival following VIV TAVR. Bioprosthetic valve fracture (BVF) is as a novel technique to address this problem. During BPV, a non-compliant valvuloplasty balloon is positioned within the BPV frame, and a highpressure balloon inflation is performed to fracture the surgical sewing ring of the BPV. This allows for further expansion of the BPV as well as the implanted THV, thus increasing the maximum effective orifice area that can be achieved after VIV TAVR. This review focuses on the current evidence base for BVF to facilitate VIV TAVR, including initial bench testing, procedural technique, clinical experience and future directions.
E. G. Nurtdinova
Full Text Available Aim. To compare effects of monotherapy with magnesium and combined therapy with magnesium and β-blocker on primary mitral valve prolapse (MVP with heart rhythm disorders.Material and methods. 71 patients with primary MVP 1-2 degree and heart rhythm disorders were involved in the study. The patients were split into three groups. Group I (25 persons received monotherapy with magnesium orotate at a dose of 1-3 g per day; group II (28 persons received combined therapy with magnesium orotate and betaxolol. The control group (18 persons received no therapy. Initially and after 12 weeks of observation all the patients underwent electrocardiography (ECG, ECG-Holter monitoring, echocardiography and autonomic balance assessment by A.M. Vein’s questionnaire.Results. In 12 weeks of treatment groups I and II showed positive dynamics in the MVP manifestations, including significant reduction in severity of the autonomic dysfunction syndrome, ECG positive dynamics, antiarrhythmic effect, decrease in the degree of prolapse, diminution of mitral regurgitation and left auricle volumes. More substantial hemodynamic effects were found in the group of patients who received combination therapy.Conclusion. Combined therapy has proven advantages in comparison with magnesium monotherapy in terms of daily quantity of extrasystoles, reduction in heart rate, decrease in autonomic disfunction and normalization of intracardiac hemodynamics.
E. G. Nurtdinova
Full Text Available Aim. To compare effects of monotherapy with magnesium and combined therapy with magnesium and β-blocker on primary mitral valve prolapse (MVP with heart rhythm disorders.Material and methods. 71 patients with primary MVP 1-2 degree and heart rhythm disorders were involved in the study. The patients were split into three groups. Group I (25 persons received monotherapy with magnesium orotate at a dose of 1-3 g per day; group II (28 persons received combined therapy with magnesium orotate and betaxolol. The control group (18 persons received no therapy. Initially and after 12 weeks of observation all the patients underwent electrocardiography (ECG, ECG-Holter monitoring, echocardiography and autonomic balance assessment by A.M. Vein’s questionnaire.Results. In 12 weeks of treatment groups I and II showed positive dynamics in the MVP manifestations, including significant reduction in severity of the autonomic dysfunction syndrome, ECG positive dynamics, antiarrhythmic effect, decrease in the degree of prolapse, diminution of mitral regurgitation and left auricle volumes. More substantial hemodynamic effects were found in the group of patients who received combination therapy.Conclusion. Combined therapy has proven advantages in comparison with magnesium monotherapy in terms of daily quantity of extrasystoles, reduction in heart rate, decrease in autonomic disfunction and normalization of intracardiac hemodynamics.
Symersky, P.; Habets, J.; Westers, P.; Mol, de B.A.J.M.; Prokop, M.; Budde, R.P.J.
Objectives Multidetector computed tomography (MDCT) has diagnostic value for the evaluation of prosthetic heart valve (PHV) dysfunction but it is hampered by artefacts. We hypothesised that image acquisition using prospective triggering instead of retrospective gating would reduce artefacts related
Subirana, M Teresa; Barón-Esquivias, Gonzalo; Manito, Nicolás; Oliver, José M; Ripoll, Tomás; Lambert, Jose Luis; Zunzunegui, José L; Bover, Ramon; García-Pinilla, José Manuel
This article presents the most relevant developments in 2013 in 3 key areas of cardiology: congenital heart disease, clinical cardiology, and heart failure and transplant. Within the area of congenital heart disease, we reviewed contributions related to sudden death in adult congenital heart disease, the importance of specific echocardiographic parameters in assessing the systemic right ventricle, problems in patients with repaired tetralogy of Fallot and indication for pulmonary valve replacement, and confirmation of the role of specific factors in the selection of candidates for Fontan surgery. The most recent publications in clinical cardiology include a study by a European working group on correct diagnostic work-up in cardiomyopathies, studies on the cost-effectiveness of percutaneous aortic valve implantation, a consensus document on the management of type B aortic dissection, and guidelines on aortic valve and ascending aortic disease. The most noteworthy developments in heart failure and transplantation include new American guidelines on heart failure, therapeutic advances in acute heart failure (serelaxin), the management of comorbidities such as iron deficiency, risk assessment using new biomarkers, and advances in ventricular assist devices. Copyright © 2013 Sociedad Española de Cardiología. Published by Elsevier Espana. All rights reserved.
Full Text Available Ji Jun, Ji Shang-Yi, Yang Jian-An, He Xia, Yang Xiao-Han, Ling Wen-Ping, Chen Xiao-LingDepartment of Pathology and Cardiovascular Surgery, Shenzhen Sun Yat-Sen Cardiovascular Hospital, Shenzhen, Guangdong, People's Republic of ChinaBackground: There are more than 300,000 prosthetic heart valve replacements each year worldwide. These patients are faced with a higher risk of thromboembolic events after heart valve surgery and long-term or even life-long anticoagulative and antiplatelet therapies are necessary. Some severe complications such as hemorrhaging or rebound thrombosis can occur when the therapy ceases. Tissue-type plasminogen activator (t-PA is a thrombolytic agent. One of the best strategies is gene therapy, which offers a local high expression of t-PA over a prolonged time period to avoid both systemic hemorrhaging and local rebound thrombosis. There are some issues with t-PA that need to be addressed: currently, there is no up-to-date report on how the t-PA gene targets the heart in vivo and the gene vector for t-PA needs to be determined.Aims: To fabricate an albumin nano-t-PA gene ultrasound-targeted agent and investigate its targeting effect on prevention of thrombosis after heart mechanic valve replacement under therapeutic ultrasound.Methods: A dog model of mechanical tricuspid valve replacement was constructed. A highly expressive t-PA gene plasmid was constructed and packaged by nanoparticles prepared with bovine serum albumin. This nanopackaged t-PA gene plasmid was further cross-linked to ultrasonic microbubbles prepared with sucrose and bovine serum albumin to form the ultrasonic-targeted agent for t-PA gene transfection. The agent was given intravenously followed by a therapeutic ultrasound treatment (1 MHz, 1.5 w/cm2, 10 minutes of the heart soon after valve replacement had been performed. The expression of t-PA in myocardium was detected with multiclonal antibodies to t-PA by the indirect immunohistochemical method
Lifschultz, B D; Donoghue, E R
Strut fracture can be a life-threatening adverse effect of mechanical prosthetic heart valves. This complication has occurred in the DeBakey, the Beall, the Cooley-Cutter and, most recently, the Bjork-Shiley valves. We report the case of a 35-year-old man who died suddenly 16 months after a 60 degree Bjork-Shiley Convexo-Concave heart valve prosthesis was inserted in the aortic position. At autopsy, the two welded attachments of the valve's outlet strut had fractured. The valve's tilting disc was found in his abdominal aorta.
Abbas, S. S.; Nasif, M. S.; Said, M. A. M.; Kadhim, S. K.
Structural stresses developed in an artificial bileaflet mechanical heart valve (BMHV) due to pulsed blood flow may cause valve failure due to yielding. In this paper, von-Mises stresses are computed and compared for BMHV placed in two types of aortic root geometries that are aortic root with axisymmetric sinuses and with axisymmetric bulb, at different physiological blood flow rates. With BMHV placed in an aortic root with axisymmetric sinuses, the von-Mises stresses developed in the valve were found to be up to 47% higher than BMHV placed in aortic root with axisymmetric bulb under similar physiological conditions. High velocity vectors and therefore high von-Mises stresses have been observed for BMHV placed in aortic root with axisymmetric sinuses, that can lead to valve failure.
Ma Yuan; He Zhilong; Peng Xueyuan; Xing Ziwen
The self-acting valve has a significant influence on the efficiency and reliability of the reciprocating compressor. In the trans-critical CO 2 cycle, the large density and high pressure difference across the valve cause serious bending and impact stresses in the valve, offering great challenges for successful valve design. Experimental investigation of the valve dynamics is required in order to design a self-acting valve with a high efficiency and long life span for the trans-critical CO 2 compressor. A semi-hermetic reciprocating compressor was developed for application in CO 2 refrigeration, and a test system was incorporated into the compressor performance test rig, with a focus on investigating the dynamics of the discharge valves. With the experimental results, the movement of the valve was discussed in detail for the trans-critical CO 2 compressor, allowing for the study of the thermodynamic performance of the compressor. While varying design parameters such as pressure ratio, valve lift, spring stiffness and compressor speed, the movement of the discharge valve in the reciprocating CO 2 compressor was measured in order to investigate the major factors that influence the valve dynamics. The average valve speed increased from 0.71 m/s to 0.81 m/s as the discharge pressure changed from 7.8 MPa to 12 MPa. The experimental methods and results discussed in this paper could provide useful information for both valve testing and the optimization of their reliability in trans-critical CO 2 compressors.
Gujral, Dorothy M; Lloyd, Guy; Bhattacharyya, Sanjeev
Radiation to the mediastinum is a key component of treatment with curative intent for a range of cancers including Hodgkin's lymphoma and breast cancer. Exposure to radiation is associated with a risk of radiation-induced heart valve damage characterised by valve fibrosis and calcification. There is a latent interval of 10-20 years between radiation exposure and development of clinically significant heart valve disease. Risk is related to radiation dose received, interval from exposure and use of concomitant chemotherapy. Long-term outlook and the risk of valve surgery are related to the effects of radiation on mediastinal structures including pulmonary fibrosis and pericardial constriction. Dose prediction models to predict the risk of heart valve disease in the future and newer radiation techniques to reduce the radiation dose to the heart are being developed. Surveillance strategies for this cohort of cancer survivors at risk of developing significant heart valve complications are required. Published by the BMJ Publishing Group Limited. For permission to use (where not already granted under a licence) please go to http://www.bmj.com/company/products-services/rights-and-licensing/
Zhao, Li; Chen, Chunxia; Li, Bei; Dong, Li; Guo, Yingqiang; Xiao, Xijun; Zhang, Eryong; Qin, Li
Objective To study the performance of pharmacogenetics-based warfarin dosing algorithms in the initial and the stable warfarin treatment phases in a cohort of Han-Chinese patients undertaking mechanic heart valve replacement. Methods We searched PubMed, Chinese National Knowledge Infrastructure and Wanfang databases for selecting pharmacogenetics-based warfarin dosing models. Patients with mechanic heart valve replacement were consecutively recruited between March 2012 and July 2012. The predicted warfarin dose of each patient was calculated and compared with the observed initial and stable warfarin doses. The percentage of patients whose predicted dose fell within 20% of their actual therapeutic dose (percentage within 20%), and the mean absolute error (MAE) were utilized to evaluate the predictive accuracy of all the selected algorithms. Results A total of 8 algorithms including Du, Huang, Miao, Wei, Zhang, Lou, Gage, and International Warfarin Pharmacogenetics Consortium (IWPC) model, were tested in 181 patients. The MAE of the Gage, IWPC and 6 Han-Chinese pharmacogenetics-based warfarin dosing algorithms was less than 0.6 mg/day in accuracy and the percentage within 20% exceeded 45% in all of the selected models in both the initial and the stable treatment stages. When patients were stratified according to the warfarin dose range, all of the equations demonstrated better performance in the ideal-dose range (1.88–4.38 mg/day) than the low-dose range (warfarin dose prediction and in the low-dose and the ideal-dose ranges. Conclusions All of the selected pharmacogenetics-based warfarin dosing regimens performed similarly in our cohort. However, the algorithms of Wei, Huang, and Miao showed a better potential for warfarin prediction in the initial and the stable treatment phases in Han-Chinese patients undertaking mechanic heart valve replacement. PMID:24728385
Ducci, Andrea; Tzamtzis, Spyridon; Mullen, Michael J; Burriesci, Gaetano
The study aim was to assess, in vitro, the hemodynamic modifications produced by transcatheter valves in the Valsalva sinuses, by mean of phase-resolved particle image velocimetry (PIV) measurements. Flow measurements were performed on a glass mock aortic root that included three polymeric valve leaflets, before and after the implantation of a Medtronic CoreValve device and of an Edwards SAPIEN valve. All experiments were carried out in a hydro-mechanical cardiovascular pulse duplicator system (Vivitro Superpump System SP3891) that reproduced physiologically equivalent pressures and flow rates conforming to the requirements of the standard ISO 5840:2005. The flow dynamics, before and after implantation of the two prosthetic devices, was characterized on the basis of phase-resolved velocity field and viscous shear rate measurements. Direct comparison indicated that both transcatheter valves determined a significant variation of flow during the early stages of valve opening and during valve closure. In general, the presence of the two valve implants significantly reduced the flow activity in the Valsalva sinuses, promoting regions of stagnation at their base. The reduction in flow in the Valsalva sinuses could be associated with the higher incidence of ischemic events reported after transcatheter heart valve implantation.
Endocarditis has been reported in patients with valvular heart disease who have undergone acupuncture treatment, although most have been associated with the use of semi-permanent needles. This has led reviewers to suggest that acupuncture may not only be contraindicated in such patients but that prophylactic antibiotics should be given. This study investigated the use of acupuncture treatment in patients with proven valvular heart disease and observed whether endocarditis developed in such patients. All patients in a single-handed GP practice with proven valvular heart disease, including those with prosthetic valves, were identified over a ten-year period. Those who had undergone acupuncture treatment underwent a clinical examination and diagnostic tests, which focused on the signs, symptoms and laboratory criteria for the diagnosis of endocarditis and included a transthoracic echocardiogram. Autopsy findings were reviewed in any patient who died. Based on these clinical and laboratory data, using the modified Duke's criteria for the diagnosis of endocarditis, patients were identified as having definite or possible endocarditis, or the diagnosis was rejected. All patients underwent brief acupuncture with no skin disinfectant and no prophylactic antibiotics were given. Semi-permanent needles were avoided. Thirty-six patients with valvular heart disease underwent a total of 479 acupuncture treatments over a ten-year period. The median number of treatments was 9 (range 1-72), with a follow-up after treatment of 5.75 years (range 0.5-10 years). Definite endocarditis was not found in any patient, but two patients had possible endocarditis, eventually discounted by both negative blood cultures and echocardiography. In conclusion, brief acupuncture was safe in this small cohort of valvular heart disease patients and no case of endocarditis was detected over a ten-year period.
Fontaine, A A; Heinrich, R S; Walker, P G; Pedersen, E M; Scheidegger, M B; Boesiger, P; Walton, S P; Yoganathan, A P
The non-invasive, in-vivo assessment of prosthetic valve function is compromised by the lack of accurate measurements of the transvalvular flow fields or hemodynamics by current techniques. Short echo time magnetic resonance imaging (MRI) may provide a method for the non-invasive, in vivo assessment of prosthetic valve function by accurately measuring changes in the transvalvular flow fields associated with normal and dysfunctional prosthetic valves. The objectives of these in vitro experiments were to investigate the potential for using MRI as a tool to measure the complex flow fields distal to replacement heart valves, and to assess the accuracy of MRI velocity measurements by comparison with Laser Doppler Anemometry (LDA), a gold standard. The velocity fields downstream of tilting disc, bileaflet, ball and cage, and pericardial tissue valves were measured using both three-component LDA and MRI phase velocity encoding under a steady flow rate of 22.8 l/min, simulating peak systolic flow. The valves were tested under normal and stenotic conditions to assess the MRI capabilities under a wide range of local flow conditions, velocities and turbulence levels. A new short echo time MRI technique (FAcE), which allowed velocity measurements in stenotic jets with high turbulence, was tested. Good overall agreement was obtained between the MRI velocity measurements and the LDA data. The MRI velocity measurements adequately reproduced the spatial structure of the flow fields. In most cases peak velocities were accurately measured to within 15%. The results indicate that the FAcE MRI method has the potential to be used as a diagnostic tool to assess prosthetic valve function.
Liu, Xianbao; He, Yuxin; Zhu, Qifeng; Gao, Feng; He, Wei; Yu, Lei; Zhou, Qijing; Kong, Minjian; Wang, Jian'an
To explore assessment of supra-annular structure for self-expanding transcatheter heart valve (THV) size selection in patients with bicuspid aortic stenosis (AS). Annulus-based device selection from CT measurement is the standard sizing strategy for tricuspid aortic valve before transcatheter aortic valve replacement (TAVR). Because of supra-annular deformity, device selection for bicuspid AS has not been systemically studied. Twelve patients with bicuspid AS who underwent TAVR with self-expanding THVs were included in this study. To assess supra-annular structure, sequential balloon aortic valvuloplasty was performed in every 2 mm increments until waist sign occurred with less than mild regurgitation. Procedural results and 30 day follow-up outcomes were analyzed. Seven patients (58.3%) with 18 mm; three patients (25%) with sequential 18 mm, 20 mm; and only two patients (16.7%) with sequential 18 mm, 20 mm, and 22 mm balloon sizing were performed, respectively. According to the results of supra-annular assessment, a smaller device size (91.7%) was selected in all but one patient compared with annulus based sizing strategy, and the outcomes were satisfactory with 100% procedural success. No mortality and 1 minor stroke were observed at 30 d follow-up. The percentage of NYHA III/IV decreased from 83.3% (9/12) to 16.7% (2/12). No new permanent pacemaker implantation and no moderate or severe paravalvular leakage were found. A supra-annular structure based sizing strategy is feasible for TAVR in patients with bicuspid AS. © 2018 The Authors Catheterization and Cardiovascular Interventions Published by Wiley Periodicals, Inc.
Chest sounds - murmurs; Heart sounds - abnormal; Murmur - innocent; Innocent murmur; Systolic heart murmur; Diastolic heart murmur ... The heart has 4 chambers: Two upper chambers (atria) Two lower chambers (ventricles) The heart has valves that close ...
Giersiepen, M; Krause, U; Knott, E; Reul, H; Rau, G
Ten mechanical valves (TAD 27 mm): Starr-Edwards Silastic Ball, Björk-Shiley Standard, Björk-Shiley Concave-Convex, Björk-Shiley Monostrut, Hall-Kaster (Medtronic-Hall), OmniCarbon, Bicer Val, Sorin, Saint-Jude Medical and Hemex (Duromedics) are investigated in a comparative in vitro study. The velocity and turbulent shear stress profiles of the valves were determined by Laser Doppler anemometry in two different downstream axes within a model aortic root. Depending on the individual valve design, velocity peaks up to 1.5 m/s and turbulent shear stress peaks up to 150 N/m2 were measured during the systolic phase. These shear stress peaks mainly occurred in areas of flow separation and intense momentum exchange. Directly downstream of the valves (measuring axis 0.55.dAorta) turbulent shear stress peaks occurred at peak systole and during the deceleration phase, while in the second measuring axis (1.5.dAorta) turbulence levels were lower. Shear stress levels were high at the borders of the fluid jets. The results are discussed from a fluid-dynamic point of view.
Hatoum, Hoda; Dasi, Lakshmi Prasad
Understanding blood flow related adverse complications such as leaflet thrombosis post-transcatheter aortic valve implantation (TAVI) requires a deeper understanding of how patient-specific anatomic and hemodynamic factors, and relative valve positioning dictate sinus vortex flow and stasis regions. High resolution time-resolved particle image velocimetry measurements were conducted in compliant and transparent 3D printed patient-specific models of stenotic bicuspid and tricuspid aortic valve roots from patients who underwent TAVI. Using Lagrangian particle tracking analysis of sinus vortex flows and probability distributions of residence time and blood damage indices we show that (a) patient specific modeling provides a more realistic assessment of TAVI flows, (b) TAVI deployment alters sinus flow patterns by significantly decreasing sinus velocity and vorticity, and (c) relative valve positioning can control critical vortex structures that may explain preferential leaflet thrombosis corresponding to separated flow recirculation, secondary to valve jet vectoring relative to the aorta axis. This work provides new methods and understanding of the spatio-temporal aortic sinus vortex dynamics in post TAVI pathology. This study was supported by the Ohio State University DHLRI Trifit Challenge award.
LaHaye, Stephanie; Lincoln, Joy
Valvular heart disease is associated with significant morbidity and mortality and often the result of congenital malformations. However, the prevalence is increasing in adults not only because of the growing aging population, but also because of improvements in the medical and surgical care of children with congenital heart valve defects. The success of the Human Genome Project and major advances in genetic technologies, in combination with our increased understanding of heart valve development, has led to the discovery of numerous genetic contributors to heart valve disease. These have been uncovered using a variety of approaches including the examination of familial valve disease and genome-wide association studies to investigate sporadic cases. This review will discuss these findings and their implications in the treatment of valvular heart disease. PMID:24743897
Dalgliesh, Ailsa J; Liu, Zhi Zhao; Griffiths, Leigh G
Current heart valve prostheses are associated with significant complications, including aggressive immune response, limited valve life expectancy, and inability to grow in juvenile patients. Animal derived "tissue" valves undergo glutaraldehyde fixation to mask tissue antigenicity; however, chronic immunological responses and associated calcification still commonly occur. A heart valve formed from an unfixed bovine pericardium (BP) extracellular matrix (ECM) scaffold, in which antigenic burden has been eliminated or significantly reduced, has potential to overcome deficiencies of current bioprostheses. Decellularization and antigen removal methods frequently use sequential solutions extrapolated from analytical chemistry approaches to promote solubility and removal of tissue components from resultant ECM scaffolds. However, the extent to which such prefractionation strategies may inhibit removal of antigenic tissue components has not been explored. We hypothesize that presence of magnesium in prefractionation steps causes DNA precipitation and reduces removal of nuclear-associated antigenic proteins. Keeping all variables consistent bar the addition or absence of magnesium (2 mM magnesium chloride hexahydrate), residual BP ECM scaffold antigenicity and removed antigenicity were assessed, along with residual and removed DNA content, ECM morphology, scaffold composition, and recellularization potential. Furthermore, we used proteomic methods to determine the mechanism by which magnesium presence or absence affects scaffold residual antigenicity. This study demonstrates that absence of magnesium from antigen removal solutions enhances solubility and subsequent removal of antigenic nuclear-associated proteins from BP. We therefore conclude that the primary mechanism of action for magnesium removal during antigen removal processes is avoidance of DNA precipitation, facilitating solubilization and removal of nuclear-associated antigenic proteins. Future studies are
Bapat, Vinnie; Pirone, Francesco; Kapetanakis, Stam; Rajani, Ronak; Niederer, Steven
To determine the factors influencing left ventricular outflow tract (LVOT) area reduction after a mitral valve-in-valve (VIV) or a valve-in-ring (VIR) procedure. Transcatheter heart valves (THVs) are increasingly used in performing a VIV or a VIR procedure in high-risk patients. Although less invasive, a potential complication is LVOT obstruction. However, the factors predisposing to LVOT obstruction are ill defined. To understand the effects of the various factors, the study was carried out in three parts: To understand the effect of VIV and VIR on reduction in LVOT area with special attention to different surgical heart valve (SHV) orientations and depth of THV implant. This was carried out in porcine and cadaver hearts. To quantify aorto-mitral-annular (AMA) angle in 20 patients with or without mitral disease and to derive a static computational model to predict LVOT obstruction. To study the effect of SHV design on LVOT obstruction after VIV. This was carried out as a bench test. LVOT area reduction was similar after VIV irrespective of orientation of the mitral SHV implantation as it pinned open the SHV leaflets. Similar effect was seen after VIR. The degree of LVOT obstruction was partly determined by AMAangle and was inversely proportional. SHV design, ring design, and depth of SPAIEN XT implantation also had effect on LVOT obstruction. A possibility of LVOT obstruction should be considered when performing a VIV and VIR procedure. Type of SHV, flexible ring, less obtuse AMA angle, and depth of SAPIEN XT implant can influence the risk. © 2015 Wiley Periodicals, Inc.
Head, Stuart J; Çelik, Mevlüt; Kappetein, A Pieter
Mechanical valves used for aortic valve replacement (AVR) continue to be associated with bleeding risks because of anticoagulation therapy, while bioprosthetic valves are at risk of structural valve deterioration requiring reoperation. This risk/benefit ratio of mechanical and bioprosthetic valves has led American and European guidelines on valvular heart disease to be consistent in recommending the use of mechanical prostheses in patients younger than 60 years of age. Despite these recommendations, the use of bioprosthetic valves has significantly increased over the last decades in all age groups. A systematic review of manuscripts applying propensity-matching or multivariable analysis to compare the usage of mechanical vs. bioprosthetic valves found either similar outcomes between the two types of valves or favourable outcomes with mechanical prostheses, particularly in younger patients. The risk/benefit ratio and choice of valves will be impacted by developments in valve designs, anticoagulation therapy, reducing the required international normalized ratio, and transcatheter and minimally invasive procedures. However, there is currently no evidence to support lowering the age threshold for implanting a bioprosthesis. Physicians in the Heart Team and patients should be cautious in pursuing more bioprosthetic valve use until its benefit is clearly proven in middle-aged patients. Published on behalf of the European Society of Cardiology. All rights reserved. © The Author 2017. For permissions, please email: firstname.lastname@example.org.
Han, Moon Hee; Im, Chung Ki; Im, Dong Ran; Han, Man Chung; Lee, Young Woo; Seo, Jung Don
Prolapse of mitral valve is characterized by its unique auscultatory, echocardiographic and angiographic findings and may be associated with various disease entities such as congenital heart disease, coronary heart disease and Marfan's syndrome etc. Authors report recent experience of 3 cases of prolapsed mitral valve, 2 cases associated with A.S.D. and 1 case with Marfan's syndrome.
Geemen, van D.; Riem Vis, P.W.; Soekhradj - Soechit, R.S.; Sluijter, J.P.G.; Liefde - van Beest, de M.; Kluin, J.; Bouten, C.V.C.
In autologous heart valve tissue engineering, there is an ongoing search for alternatives of fetal bovine serum (FBS). Human platelet-lysate (PL) might be a promising substitute. In the present article, we aimed to examine the tissue formation, functionality, and mechanical properties of engineered
Kim, Woojin; Choi, Haecheon; Kweon, Jihoon; Yang, Dong Hyun; Kim, Namkug; Kim, Young-Hak
A pannus, an abnormal layer of fibrovascular tissue observed on a bileaflet mechanical heart valve (BMHV), induces dysfunctions of BMHV such as the time delay and incomplete valve closing. We numerically simulate the flows around an intra-annular type BMHV model with and without pannus formation, respectively, and investigate the flow and bileaflet-movement modifications due to the pannus formation. Simulations are conducted at a physiological condition (mean flow rate of 5 l/min, cycle duration of 866 ms, and the Reynolds number of 7200 based on the inflow peak bulk velocity and inflow diameter). We model the pannus as an annulus with fixed outer radius and vary the inner radius of the pannus. Our preliminary results indicate that the flow field changes significantly and the bileaflet does not close properly due to the pannus formation. The detailed results will be given at the final presentation. Supported by the NRF Programs (NRF-2011-0028032, NRF-2012M2A8A4055647).
Hasler, David; Obrist, Dominik
The flow field past a prosthetic aortic valve comprises many details that indicate whether the prosthesis is functioning well or not. It is, however, not yet fully understood how an optimal flow scenario would look, i.e. which subtleties of the fluid dynamics in place are essential regarding the durability and compatibility of a prosthetic valve. In this study, we measured and analyzed the 3D flow field in the vicinity of a bio-prosthetic heart valve in function of the aortic root size. The m...
Wu, Changfu; Hwang, Ned H C; Lin, Yu-Kweng M
Cavitation damage has been observed on mechanical heart valves (MHVs) undergoing accelerated testing. Cavitation itself can be modeled as a stochastic process, as it varies from beat to beat of the testing machine. This in-vitro study was undertaken to investigate the statistical characteristics of MHV cavitation. A 25-mm St. Jude Medical bileaflet MHV (SJM 25) was tested in an accelerated tester at various pulse rates, ranging from 300 to 1,000 bpm, with stepwise increments of 100 bpm. A miniature pressure transducer was placed near a leaflet tip on the inflow side of the valve, to monitor regional transient pressure fluctuations at instants of valve closure. The pressure trace associated with each beat was passed through a 70 kHz high-pass digital filter to extract the high-frequency oscillation (HFO) components resulting from the collapse of cavitation bubbles. Three intensity-related measures were calculated for each HFO burst: its time span; its local root-mean-square (LRMS) value; and the area enveloped by the absolute value of the HFO pressure trace and the time axis, referred to as cavitation impulse. These were treated as stochastic processes, of which the first-order probability density functions (PDFs) were estimated for each test rate. Both the LRMS value and cavitation impulse were log-normal distributed, and the time span was normal distributed. These distribution laws were consistent at different test rates. The present investigation was directed at understanding MHV cavitation as a stochastic process. The results provide a basis for establishing further the statistical relationship between cavitation intensity and time-evolving cavitation damage on MHV surfaces. These data are required to assess and compare the performance of MHVs of different designs.
Full Text Available It is obvious that the main function of the nozzle valve is to shut off the stream of fluid in the piping system. The response rate of the valve to the decreasing or reversing flow in the system will then depend on the valve properties and equally on the properties of the piping system. The interaction of these two elements is also important for the origin of pressure pulsations in the system. While the pressure pulsations were the cause for design of this particular valve it should be noted that the general design of the valve for any pipeline system is not possible. The valve cannot properly work under all circumstances and operating conditions. With respect to this, the dynamic properties of the valve will be assessed on the basis of the valve equation of motion and the pipeline model. An adequate response of the whole system can be obtained by combining both approaches. The valve equations of motion are also complemented by CFD simulations, which enable to capture the movement of the valve disc with respect to flow rate.
Gibbons, J. R. P.; Alladine, M. F. [London Chest Hospital, London (United Kingdom)
Starr-Edwards valves have normally been sterilized by exposure to ethylene oxide or by autoclaving. Patients having a prosthetic valve replacement are known to have a higher incidence of endocarditis in comparison with patients in which no prosthesis has been used. Ethylene oxide will only sterilize the surface of the valve and autoclaving has caused distortion of the polytetrafluorethylene ring. Work has been done on the effect of gamma radiation on the components of these valve prostheses and is given in detail. The bacteriological efficiency, at a total absorbed dose of 2. 5 Mrad, has been established. Thirty valves treated by this method have now been inserted and twelve patients have been examined post-operatively for a period of one to two years. All valves are working normally and there has been no evidence of blood-borne infection or malfunction of the valve. (author)
Pires, Luis F.G.; Barreto, Claudio V.; Sarmento, Renata C. [Pontificia Univ. Catolica do Rio de Janeiro, RJ (Brazil). Dept. de Engenharia Mecanica. Nucleo de Simulacao Termohidraulica de Dutos (SIMDUT)
As usual a shipping terminal contains a header and its diameter is smaller than the main pipeline diameter. This diameter reduction amplifies the effects caused by hydraulic transients. It was noticed during simulations that check valves without dynamic modeling may introduce some error in the maximum pressure results without a dynamic model approach. The current paper uses commercial pipeline simulation software to model the dynamic behavior of the check valves. It was studied the header diameter influence and how the check valve model type may change the maximum pressure in the pipeline. It was proved that even a pipeline that works with low pressure and small elevation drop needs a detailed valve modeling to prevent the calculation of unreal pressure values in the region with diameter reduction. (author)
Li, Feng P; Rajchl, Martin; Moore, John; Peters, Terry M
porcine data, the authors compared the tracked MVA to a manually segmented MVA. The overall accuracy is 2.37 ± 1.67 mm for single plane images and 2.35 ± 1.55 mm for biplane images. The interoperator variation in manual segmentation was 2.32 ± 1.24 mm for single plane images and 1.73 ± 1.18 mm for biplane images. The computational efficiency of the algorithm on a desktop computer with an Intel(®) Xeon(®) CPU @3.47 GHz and an NVIDIA GeForce 690 graphic card is such that the time required for registering four MVA points was about 60 ms. The authors developed a rapid MVA tracking algorithm for use in the guidance of off-pump beating heart transapical mitral valve repair. This approach uses 2D biplane TEE images and was tested on a dynamic heart phantom and interventional porcine image data. Results regarding the accuracy and efficiency of the authors' MVA tracking algorithm are promising, and fulfill the requirements for surgical navigation.
Hedayat, Mohammadali; Borazjani, Iman
Bileaflet mechanical heart valves increase the risk of thrombus formation in patients which is believed to be initiated by platelet activation. Platelets can be activated by the elevated shear stresses in the bulk flow during the systole phase or the flow through the hinge during the diastole. However, the importance of platelet activation by the bulk flow vs the hinge in MHVs has yet to be studied. Here, we investigate the contribution of each of the above mechanisms to the activation of platelets in MHs by performing simulation of the flow through a 25mm St. Jude Medical valve placed in a straight aorta. Two different gap sizes (250 and 150 micrometer) are used in this study. The simulations are done using a sharp interface curvilinear immersed boundary method along with a strong-coupling algorithm for FSI solver on overset grids. The platelet activation through the hinge for different gap sizes is compared to the activation in the bulk flow using two platelet activation models to ensure the consistency of the results. Our results for all gap sizes using different activation models show that the integration of platelet activation caused by the bulk flow is several times higher in comparison to the activation through the hinge. This work is supported by the American Heart Association Grant 13SDG17220022, and the computational resources were partly provided by Center for Computational Research (CCR) at University at Buffalo.
Failed surgical mitral valve repair using an annuloplasty ring has traditionally been treated with surgical valve replacement or repair1. For patients at high risk for repeat open heart surgery, placement of a trans-catheter aortic valve (i.e., TAVI valve) within the mitral ring (i.e., Mitral-Valve-in-Ring, MViR) has emerged as a novel alternative treatment strategy2-5 . We describe our experience of a failed mitral valve repair that was successfully treated with a TAVI valve delivered via the trans-septal approach, and summarise the data relating to this emerging treatment strategy.
Han, Moon Hee; Im, Chung Ki; Im, Dong Ran; Han, Man Chung; Lee, Young Woo; Seo, Jung Don [Seoul National University College of Medicine, Seoul (Korea, Republic of)
Prolapse of mitral valve is characterized by its unique auscultatory, echocardiographic and angiographic findings and may be associated with various disease entities such as congenital heart disease, coronary heart disease and Marfan's syndrome etc. Authors report recent experience of 3 cases of prolapsed mitral valve, 2 cases associated with A.S.D. and 1 case with Marfan's syndrome.
Tsai, I.Chen [Taichung Veterans General Hospital, Department of Radiology, Taichung (China); Institute of Clinical Medicine and Faculty of Medicine, National Yang-Ming University, Taipei (China); Lin, Yung-Kai; Chang, Yen; Wang, Chung-Chi; Hsieh, Shih-Rong; Wei, Hao-Ji; Tsai, Hung-Wen [Taichung Veterans General Hospital, Section of Cardiovascular Surgery, Cardiovascular Center, Taichung (China); Fu, Yun-Ching; Jan, Sheng-Ling [Institute of Clinical Medicine and Faculty of Medicine, National Yang-Ming University, Taipei (China); Taichung Veterans General Hospital, Section of Pediatric Cardiology, Department of Pediatrics, Taichung (China); Wang, Kuo-Yang [Taichung Veterans General Hospital, Section of General Cardiology, Cardiovascular Center, Taichung (China); Chung-Shan Medical University, Department of Medicine, Taichung (China); Chen, Min-Chi; Chen, Clayton Chi-Chang [Taichung Veterans General Hospital, Department of Radiology, Taichung (China); Central Taiwan University of Science and Technology, Department of Radiological Technology, Taichung (China)
The purpose was to compare the findings of multi-detector computed tomography (MDCT) in prosthetic valve disorders using the operative findings as a gold standard. In a 3-year period, we prospectively enrolled 25 patients with 31 prosthetic heart valves. MDCT and transthoracic echocardiography (TTE) were done to evaluate pannus formation, prosthetic valve dysfunction, suture loosening (paravalvular leak) and pseudoaneurysm formation. Patients indicated for surgery received an operation within 1 week. The MDCT findings were compared with the operative findings. One patient with a Bjoerk-Shiley valve could not be evaluated by MDCT due to a severe beam-hardening artifact; thus, the exclusion rate for MDCT was 3.2% (1/31). Prosthetic valve disorders were suspected in 12 patients by either MDCT or TTE. Six patients received an operation that included three redo aortic valve replacements, two redo mitral replacements and one Amplatzer ductal occluder occlusion of a mitral paravalvular leak. The concordance of MDCT for diagnosing and localizing prosthetic valve disorders and the surgical findings was 100%. Except for images impaired by severe beam-hardening artifacts, MDCT provides excellent delineation of prosthetic valve disorders. (orig.)
Molnar, Adrian; Muresan, Ioan; Trifan, Catalin; Pop, Dana; Sacui, Diana
The introduction of Duke's criteria and the improvement of imaging methods has lead to an earlier and a more accurate diagnosis of infectious endocarditis (IE). The options for the best therapeutic approach and the timing of surgery are still a matter of debate and require a close colaboration between the cardiologist, the infectionist and the cardiac surgeon. We undertook a retrospective, descriptive study, spanning over a period of five years (from January 1st, 2007 to December 31st, 2012), on 100 patients who underwent surgery for native valve infectious endocarditis in our unit. The patients' age varied between 13 and 77 years (with a mean of 54 years), of which 85 were males (85%). The main microorganisms responsible for IE were: Streptococcus Spp. (21 cases - 21%), Staphylococcus Spp. (15 cases - 15%), and Enterococcus Spp. (9 cases - 9%). The potential source of infection was identified in 26 patients (26%), with most cases being in the dental area (16 cases - 16%). The lesions caused by IE were situated in the left heart in 96 patients (96%), mostly on the aortic valve (50 cases - 50%). In most cases (82%) we found preexisting endocardial lesions which predisposed to the development of IE, most of them being degenerative valvular lesions (38 cases - 38%). We performed the following surgical procedures: surgery on a single valve - aortic valve replacement (40 cases), mitral valve replacement (19 cases), mitral valve repair (1 case), surgery on more than one valve - mitral and aortic valve replacement (20 cases), aortic and tricuspid valve replacement (1 case), aortic valve replacement with a mechanical valve associated with mitral valve repair (5 cases), aortic valve replacement with a biological valve associated with mitral valve repair (2 cases), and mitral valve replacement with a mechanical valve combined with De Vega procedure on the tricuspid valve (1 case). In 5 patients (5%) the bacteriological examination of valve pieces excised during surgery was
Scanlan, Adam B; Nguyen, Alex V; Ilina, Anna; Lasso, Andras; Cripe, Linnea; Jegatheeswaran, Anusha; Silvestro, Elizabeth; McGowan, Francis X; Mascio, Christopher E; Fuller, Stephanie; Spray, Thomas L; Cohen, Meryl S; Fichtinger, Gabor; Jolley, Matthew A
Mastering the technical skills required to perform pediatric cardiac valve surgery is challenging in part due to limited opportunity for practice. Transformation of 3D echocardiographic (echo) images of congenitally abnormal heart valves to realistic physical models could allow patient-specific simulation of surgical valve repair. We compared materials, processes, and costs for 3D printing and molding of patient-specific models for visualization and surgical simulation of congenitally abnormal heart valves. Pediatric atrioventricular valves (mitral, tricuspid, and common atrioventricular valve) were modeled from transthoracic 3D echo images using semi-automated methods implemented as custom modules in 3D Slicer. Valve models were then both 3D printed in soft materials and molded in silicone using 3D printed "negative" molds. Using pre-defined assessment criteria, valve models were evaluated by congenital cardiac surgeons to determine suitability for simulation. Surgeon assessment indicated that the molded valves had superior material properties for the purposes of simulation compared to directly printed valves (p 3D echo-derived molded valves are a step toward realistic simulation of complex valve repairs but require more time and labor to create than directly printed models. Patient-specific simulation of valve repair in children using such models may be useful for surgical training and simulation of complex congenital cases.
Giuseppe MC Rosano
Full Text Available Structural valvular heart disease may be the cause of heart failure or may worsen the clinical status of patients with heart failure. Heart failure may also develop in patients treated with valve surgery. Patients with heart failure with valvular heart disease are at increased risk of events including sudden cardiac death. Before considering intervention (surgical or percutaneous all patients should receive appropriate medical and device therapy taking into account that vasodilators must be used with caution in patients with severe aortic stenosis. Numerous percutaneous and/or hybrid procedures have been introduced in the past few years and they are changing the management of valvular heart disease. In patients with heart failure and valvular heart disease, either primary or functional, the whole process of decision-making should be staged through a comprehensive evaluation of the risk– benefit ratio of different treatment strategies and should be made by a multidisciplinary ‘heart team’ with a particular expertise in valvular heart disease. The heart team should include heart failure cardiologists, cardiac surgeons/structural valve interventionists, imaging specialists, anaesthetists, geriatricians and intensive care specialists. This article will review recent developments and distill practical guidance in the management of this important heart failure co-morbidity.
Waziri, Farhad; Lyager Nielsen, Sten; Hasenkam, J. Michael
before clinical use. The study aim was to evaluate and compare the tricuspid valve anatomy of porcine and human hearts. METHODS: The anatomy of the tricuspid valve and the surrounding structures that affect the valve during a cardiac cycle were examined in detail in 100 fresh and 19 formalin...
Lomivorotov, Vladimir V; Efremov, Sergey M; Boboshko, Vladimir A; Nikolaev, Dmitry A; Vedernikov, Pavel E; Shilova, Anna N; Lomivorotov, Vladimir N; Karaskov, Alexander M
The study aim was to detect the most sensitive nutritional screening tool and to assess its prognostic value with regards to an adverse clinical course in patients with heart valve disease undergoing cardiopulmonary bypass (CPB). This prospective cohort study included 441 adult patients who were screened using four nutritional screening tools: Nutritional Risk Screening 2002 (NRS-2002); Malnutrition Universal Screening Tool (MUST); Mini Nutritional Assessment (MNA); and Short Nutritional Assessment Questionnaire (SNAQ). Nutritional assessment was performed using a Subjective Global Assessment (SGA). In-hospital mortality, postoperative complications, and duration of hospital stay were each analyzed. With regards to the detection of malnutrition, the sensitivities of MUST, SNAQ, MNA and NRS-2002 were 100%, 92%, 84.6% and 43.6%, respectively. Malnutrition identified by MUST and MNA were associated with postoperative complications (OR 1.63, p = 0.033 and OR 1.6, p = 0.035) and prolonged hospitalization (OR 1.57, p = 0.048 and OR 1.7, p = 0.02). According to multivariate logistic regression analysis, along with well-known age and duration of CPB, malnutrition identified by MUST and MNA was associated with a risk of development of complications (OR 1.6, p = 0.049 and OR 1.6, p = 0.04, respectively). The sensitivities of SNAQ, MUST, NRS-2002 and MNA with regards to postoperative complications were 26.8%, 28.8%, 10%, and 31.6%, respectively. The MUST tool is preferable with regards to the detection of malnutrition. Both, MUST and MNA independently predicted postoperative complications. SNAQ and NRS-2002 proved insensitive with regards to the postoperative course among patients with heart valve disease who were scheduled for cardiothoracic surgery.
Sponga, Sandro; Barbera, Mila Della; Pavoni, Daisy; Lechiancole, Andrea; Mazzaro, Enzo; Valente, Marialuisa; Nucifora, Gaetano; Thiene, Gaetano; Livi, Ugolino
Freedom Solo (FS) is a pericardial stentless heart valve showing excellent haemodynamic performance at mid-term. The aim of this study was to evaluate the long-term performance of such bioprostheses. Between December 2004 and November 2009, 109 patients (31 men; mean age 76 ± 6 years) underwent aortic valve replacement with FS. Preoperatively, the mean NYHA class was 2.5 ± 0.7, the mean EuroSCORE II, 2.8 ± 2.5. Mean prosthesis size was 22.7 ± 1.9 mm; concomitant procedures were performed in 65 patients. Structural valve deterioration (SVD) was diagnosed according to the Valve Academic Research Consortium-2 definition. Two patients (1.8%) died within 30 days. Follow-up (72 ± 36 months) was 100% completed. The 1-, 5- and 10-year actuarial survival rates were 89, 73 and 42%, respectively, with 8 valve-related deaths; the actuarial freedom from SVD was 99, 93 and 76%. During 61 ± 39 months of follow-up, echocardiographic findings worsened progressively: At discharge, 3-5 and 7-9 years, the mean gradient was 8 ± 4, 12 ± 11 and 19 ± 19 mmHg ( P < 0.01); the indexed effective orifice area was 1.0 ± 0.2, 0.9 ± 0.2 and 0.8 ± 0.3 cm 2 /m 2 ( P < 0.01). Of the 13 patients who developed SVD, it was due to aortic stenosis in 11. SVD was a predictor of cardiovascular mortality at univariate analysis (HR 2.87, 1.12-7.29); 2 explanted prostheses showed massive calcium deposits with mean calcium and phosphorus contents of 234 ± 16 and 116 ± 7 mg/g dry weight, respectively. The FS bioprosthesis shows excellent mid-term clinical and haemodynamic results and offers an alternative to other valves, particularly in the case of a small aortic annulus. Worsening of FS performance was observed at late follow-up because of progressive SVD with stenosis, questioning whether it should be used in patients with a long life expectancy. © The Author 2017. Published by Oxford University Press on behalf of the
Altunbas, Gokhan; Yuce, Murat; Ozer, Hasan O; Davutoglu, Vedat; Ercan, Suleyman; Kizilkan, Nese; Bilici, Muhammet
BACKGROUND AND AIM OF STUDY: Rheumatic valvular heart disease, which remains a common health problem in developing countries, has numerous consequences on the heart chambers and circulation. The study aim was to investigate the effects of chronic rheumatic valve disease on the diameters of the descending aorta (DA) and inferior vena cava (IVC). METHODS: A total of 88 patients with echocardiographically documented rheumatic valvular heart disease and 112 healthy controls were enrolled into the study. All patients underwent detailed echocardiographic examinations, while their height and body weight were recorded and adjusted to their body surface area. RESULTS: The most common involvement was mitral valve disease, followed by aortic valve disease and tricuspid valve disease. The mean diameter of the DA (indexed to BSA) was 1.79 ± 0.49 cm for patients and 1.53 ± 0.41 for controls (p Rheumatic valve disease, especially mitral stenosis, was closely related to remodeling of the great vessels.
Heart surgery - pediatric; Heart surgery for children; Acquired heart disease; Heart valve surgery - children ... There are many kinds of heart defects. Some are minor, and others are more serious. Defects can occur inside the heart or in the large blood vessels ...
Hof, Alexander; Raschke, Silja; Baier, Karina; Nehrenheim, Laura; Selig, Jessica Isabel; Schomaker, Markus; Lichtenberg, Artur; Meyer, Heiko; Akhyari, Payam
Biological heart valve prostheses are characterized by a limited durability due to the degenerative processes after implantation. Tissue engineering may provide new approaches in the development of optimized valvular grafts. While re-endothelialization of decellularized heart valves has already been successfully implemented, interstitial repopulation still remains an unaccomplished objective although it is essential for valvular functionality and regeneration potential. The aim of this study was to compare different concepts for an improved in vitro interstitial repopulation of decellularized heart valves. A novel 3D heart valve model has been developed to investigate the cell behaviour of valvular interstitial cells (VIC) in their physiological environment and to evaluate the potential of in vitro repopulation of acellular heart valves. Ovine aortic heart valves were decellularized by detergent solutions and additionally treated with trypsin or laser perforation. Subsequently, the decellularized extracellular matrices (dECM) were reseeded with ovine VIC using reseeding devices to provide a repopulation of the matrix on a defined area under controlled conditions. After an initial attachment of the VIC, reseeded dECM were transferred into a transwell system to improve the nutrient supply inside the valvular matrix. Cell migration and expression of cell markers were analysed histologically. The results were compared with VIC cultivation in a biological scaffold. VIC did not migrate into the matrix of untreated dECM and reseeding in laser perforated dECM showed inconsistent results. However, trypsinization increased the susceptibility of the valvular cusps to VIC penetration and repopulation of superficial areas. Additionally, the cultivation of reseeded dECM in a transwell system significantly increased the total number of cells repopulating the valvular matrix and their mean migration distance, representing the best repopulation results. Immunohistological analysis
Full Text Available This review covers the role of three-dimensional (3D echocardiography in the diagnosis of heart valve disease. Several factors have contributed to the evolution of this technique, which is currently a simple and routine method: rapid evolution in probe and computer technologies, demonstration that 3D data sets allowed more complete and accurate evaluation of cardiac structures, emerging clinical experience indicating the strong potential particularly in valve diseases, volume and function of the two ventricle measurements and several other fields. This report will review current and future applications of 3D echocardiography in mitral, aortic and tricuspid valve diseases underlying both qualitative (morphologic and quantitative advantages of this technique. (Heart International 2007; 3: 35-41
Park, Jong Woon
In Korean 3 Loop plants a water loop seal pipe is installed containing condensed water upstream of a pressurizer safety valve to protect the valve disk from the hot steam environment. The loop seal water purge time is a key parameter in safety analyses for overpressure transients, because it delays valve opening. The loop seal purge time is uncertain to measure by test and thus 3-dimensional realistic computational fluid dynamics (CFD) model is developed in this paper to predict the seal water purge time before full opening of the valve which is driven by steam after water purge. The CFD model for a typical pressurizer safety valve with a loop seal pipe is developed using the computer code of ANSYS CFX 11. Steady-state simulations are performed for full discharge of steam at the valve full opening. Transient simulations are performed for the loop seal dynamics and to estimate the loop seal purge time. A sudden pressure drop higher than 2,000 psia at the tip of the upper nozzle ring is expected from the steady-state calculation. Through the transient simulation, almost loop seal water is discharged within 1.2 second through the narrow opening between the disk and the nozzle of the valve. It can be expected that the valve fully opens at least before 1.2 second because constant valve opening is assumed in this CFX simulation, which is conservative because the valve opens fully before the loop seal water is completely discharged. The predicted loop seal purge time is compared with previous correlation. (orig.)
Park, Jong Woon [Dongguk Univ., Gyeongju (Korea, Republic of). Nuclear and Energy Engineering Dept.
In Korean 3 Loop plants a water loop seal pipe is installed containing condensed water upstream of a pressurizer safety valve to protect the valve disk from the hot steam environment. The loop seal water purge time is a key parameter in safety analyses for overpressure transients, because it delays valve opening. The loop seal purge time is uncertain to measure by test and thus 3-dimensional realistic computational fluid dynamics (CFD) model is developed in this paper to predict the seal water purge time before full opening of the valve which is driven by steam after water purge. The CFD model for a typical pressurizer safety valve with a loop seal pipe is developed using the computer code of ANSYS CFX 11. Steady-state simulations are performed for full discharge of steam at the valve full opening. Transient simulations are performed for the loop seal dynamics and to estimate the loop seal purge time. A sudden pressure drop higher than 2,000 psia at the tip of the upper nozzle ring is expected from the steady-state calculation. Through the transient simulation, almost loop seal water is discharged within 1.2 second through the narrow opening between the disk and the nozzle of the valve. It can be expected that the valve fully opens at least before 1.2 second because constant valve opening is assumed in this CFX simulation, which is conservative because the valve opens fully before the loop seal water is completely discharged. The predicted loop seal purge time is compared with previous correlation. (orig.)
Rasmussen, Vibeke Guldbrand; Poulsen, Steen Hvitfeldt; Dupont, E
regurgitation (n = 5) was found in 22 EDDA patients (25.9%). Two patients had coexistent moderate mitral and tricuspid valvular regurgitation. Two non-EDDA patients had moderate valve insufficiency (3.8%, P insufficiency in the EDDA patients was 7....... Interventions. Patients were screened for valvular heart disease by clinical means and by examiner-blinded echocardiography. Main outcome measure was valvular regurgitation revealed by echocardiography. RESULTS: Severe aortic regurgitation (n = 4) or moderate aortic (n = 12), mitral (n = 3) or tricuspidal valve...
MOLNAR, ADRIAN; MURESAN, IOAN; TRIFAN, CATALIN; POP, DANA; SACUI, DIANA
Background and aims The introduction of Duke’s criteria and the improvement of imaging methods has lead to an earlier and a more accurate diagnosis of infectious endocarditis (IE). The options for the best therapeutic approach and the timing of surgery are still a matter of debate and require a close colaboration between the cardiologist, the infectionist and the cardiac surgeon. Methods We undertook a retrospective, descriptive study, spanning over a period of five years (from January 1st, 2007 to December 31st, 2012), on 100 patients who underwent surgery for native valve infectious endocarditis in our unit. Results The patients’ age varied between 13 and 77 years (with a mean of 54 years), of which 85 were males (85%). The main microorganisms responsible for IE were: Streptococcus Spp. (21 cases – 21%), Staphylococcus Spp. (15 cases – 15%), and Enterococcus Spp. (9 cases – 9%). The potential source of infection was identified in 26 patients (26%), with most cases being in the dental area (16 cases – 16%). The lesions caused by IE were situated in the left heart in 96 patients (96%), mostly on the aortic valve (50 cases – 50%). In most cases (82%) we found preexisting endocardial lesions which predisposed to the development of IE, most of them being degenerative valvular lesions (38 cases – 38%). We performed the following surgical procedures: surgery on a single valve - aortic valve replacement (40 cases), mitral valve replacement (19 cases), mitral valve repair (1 case), surgery on more than one valve – mitral and aortic valve replacement (20 cases), aortic and tricuspid valve replacement (1 case), aortic valve replacement with a mechanical valve associated with mitral valve repair (5 cases), aortic valve replacement with a biological valve associated with mitral valve repair (2 cases), and mitral valve replacement with a mechanical valve combined with De Vega procedure on the tricuspid valve (1 case). In 5 patients (5%) the bacteriological
Qu, Zhilin; Hu, Gang; Garfinkel, Alan; Weiss, James N.
In a normal human life span, the heart beats about 2 to 3 billion times. Under diseased conditions, a heart may lose its normal rhythm and degenerate suddenly into much faster and irregular rhythms, called arrhythmias, which may lead to sudden death. The transition from a normal rhythm to an arrhythmia is a transition from regular electrical wave conduction to irregular or turbulent wave conduction in the heart, and thus this medical problem is also a problem of physics and mathematics. In the last century, clinical, experimental, and theoretical studies have shown that dynamical theories play fundamental roles in understanding the mechanisms of the genesis of the normal heart rhythm as well as lethal arrhythmias. In this article, we summarize in detail the nonlinear and stochastic dynamics occurring in the heart and their links to normal cardiac functions and arrhythmias, providing a holistic view through integrating dynamics from the molecular (microscopic) scale, to the organelle (mesoscopic) scale, to the cellular, tissue, and organ (macroscopic) scales. We discuss what existing problems and challenges are waiting to be solved and how multi-scale mathematical modeling and nonlinear dynamics may be helpful for solving these problems. PMID:25267872
Qu, Zhilin, E-mail: email@example.com [Department of Medicine (Cardiology), David Geffen School of Medicine, University of California, Los Angeles, CA 90095 (United States); Hu, Gang [Department of Physics, Beijing Normal University, Beijing 100875 (China); Garfinkel, Alan [Department of Medicine (Cardiology), David Geffen School of Medicine, University of California, Los Angeles, CA 90095 (United States); Department of Integrative Biology and Physiology, University of California, Los Angeles, CA 90095 (United States); Weiss, James N. [Department of Medicine (Cardiology), David Geffen School of Medicine, University of California, Los Angeles, CA 90095 (United States); Department of Physiology, David Geffen School of Medicine, University of California, Los Angeles, CA 90095 (United States)
In a normal human life span, the heart beats about 2–3 billion times. Under diseased conditions, a heart may lose its normal rhythm and degenerate suddenly into much faster and irregular rhythms, called arrhythmias, which may lead to sudden death. The transition from a normal rhythm to an arrhythmia is a transition from regular electrical wave conduction to irregular or turbulent wave conduction in the heart, and thus this medical problem is also a problem of physics and mathematics. In the last century, clinical, experimental, and theoretical studies have shown that dynamical theories play fundamental roles in understanding the mechanisms of the genesis of the normal heart rhythm as well as lethal arrhythmias. In this article, we summarize in detail the nonlinear and stochastic dynamics occurring in the heart and their links to normal cardiac functions and arrhythmias, providing a holistic view through integrating dynamics from the molecular (microscopic) scale, to the organelle (mesoscopic) scale, to the cellular, tissue, and organ (macroscopic) scales. We discuss what existing problems and challenges are waiting to be solved and how multi-scale mathematical modeling and nonlinear dynamics may be helpful for solving these problems.
Qu, Zhilin; Hu, Gang; Garfinkel, Alan; Weiss, James N.
In a normal human life span, the heart beats about 2–3 billion times. Under diseased conditions, a heart may lose its normal rhythm and degenerate suddenly into much faster and irregular rhythms, called arrhythmias, which may lead to sudden death. The transition from a normal rhythm to an arrhythmia is a transition from regular electrical wave conduction to irregular or turbulent wave conduction in the heart, and thus this medical problem is also a problem of physics and mathematics. In the last century, clinical, experimental, and theoretical studies have shown that dynamical theories play fundamental roles in understanding the mechanisms of the genesis of the normal heart rhythm as well as lethal arrhythmias. In this article, we summarize in detail the nonlinear and stochastic dynamics occurring in the heart and their links to normal cardiac functions and arrhythmias, providing a holistic view through integrating dynamics from the molecular (microscopic) scale, to the organelle (mesoscopic) scale, to the cellular, tissue, and organ (macroscopic) scales. We discuss what existing problems and challenges are waiting to be solved and how multi-scale mathematical modeling and nonlinear dynamics may be helpful for solving these problems
Three-dimensional (3D) echocardiography has been conceived as one of the most promising methods for the diagnosis of valvular heart disease, and recently has become an integral clinical tool thanks to the development of high quality real-time transesophageal echocardiography (TEE). In particular, for mitral valve diseases, this new approach has proven to be the most unique, powerful, and convincing method for understanding the complicated anatomy of the mitral valve and its dynamism. The method has been useful for surgical management, including robotic mitral valve repair. Moreover, this method has become indispensable for nonsurgical mitral procedures such as edge to edge mitral repair and transcatheter closure of paravaluvular leaks. In addition, color Doppler 3D echo has been valuable to identify the location of the regurgitant orifice and the severity of the mitral regurgitation. For aortic and tricuspid valve diseases, this method may not be quite as valuable as for the mitral valve. However, the necessity of 3D echo is recognized for certain situations even for these valves, such as for evaluating the aortic annulus for transcatheter aortic valve implantation. It is now clear that this method, especially with the continued development of real-time 3D TEE technology, will enhance the diagnosis and management of patients with these valvular heart diseases. PMID:25378966
Ghanbari, Hossein; Kidane, Asmeret G; Burriesci, Gaetano; Ramesh, Bala; Darbyshire, Arnold; Seifalian, Alexander M
Calcification currently represents a major cause of failure of biological tissue heart valves. It is a complex phenomenon influenced by a number of biochemical and mechanical factors. Recent advances in material science offer new polymers with improved properties, potentially suitable for synthetic leaflets heart valves manufacturing. In this study, the calcification-resistance efficacy and mechanical and surface properties of a new nanocomposite polymeric material (polyhedral oligomeric silsesquioxane-poly(carbonate-urea)urethane; POSS-PCU) which has been developed by our group are assessed by means of in vitro testing. In particular, thin sheets of nanocomposite, glutaraldehyde-fixed bovine pericardium (BP) and polyurethane (PU) were exposed to a calcium solution into a specially designed in vitro accelerated physiological pulsatile pressure system for a period of 31days and a total of 4×10(7) cycles. The samples were investigated for signs of calcification after exposure to calcium solution by means of X-ray, microscopic and chemical inspections. Mechanical and surface properties were also studied using stress-strain behaviour and surface morphology and hydrophobicity. Comparison shows that, in the experimental conditions, the level of calcification for the nanocomposite is considerably lower than for the fixed BP (p=0.008) and PU samples (p=0.015). Also, mechanical properties were unchanged in POSS-PCU, while there was a significant deterioration in PU samples (pnanocomposite remained more hydrophobic than the PU sample (pnanocomposite in synthetic leaflets heart valves may lead to potential advantages in terms of long-term performances and durability. Copyright © 2010. Published by Elsevier Ltd.
Hatt, Charles R.; Wagner, Martin; Raval, Amish N.; Speidel, Michael A.
Transcatheter aortic valve replacement (TAVR) requires navigation and deployment of a prosthetic valve within the aortic annulus under fluoroscopic guidance. To support improved device visualization in this procedure, this study investigates the feasibility of frame-by-frame 3D reconstruction of a moving and expanding prosthetic valve structure from simultaneous bi-plane x-ray views. In the proposed method, a dynamic 3D model of the valve is used in a 2D/3D registration framework to obtain a reconstruction of the valve. For each frame, valve model parameters describing position, orientation, expansion state, and deformation are iteratively adjusted until forward projections of the model match both bi-plane views. Simulated bi-plane imaging of a valve at different signal-difference-to-noise ratio (SDNR) levels was performed to test the approach. 20 image sequences with 50 frames of valve deployment were simulated at each SDNR. The simulation achieved a target registration error (TRE) of the estimated valve model of 0.93 +/- 2.6 mm (mean +/- S.D.) for the lowest SDNR of 2. For higher SDNRs (5 to 50) a TRE of 0.04 mm +/- 0.23 mm was achieved. A tabletop phantom study was then conducted using a TAVR valve. The dynamic 3D model was constructed from high resolution CT scans and a simple expansion model. TRE was 1.22 +/- 0.35 mm for expansion states varying from undeployed to fully deployed, and for moderate amounts of inter-frame motion. Results indicate that it is feasible to use bi-plane imaging to recover the 3D structure of deformable catheter devices.
David Šuran; Vojko Kanič; Tatjana Golob Gulič; Husam Franjo Naji; Robert Lipovec
Background Prosthetic heart valve thrombosis (PHVT) represents a dangerous postoperative complication following prosthetic heart valve replacement. Incidence varies according to different data from 0.5–4 % per year following mitral or aortic valve replacement in spite of adequate oral anticoagulation with coumarins. Case report We are presenting a case of prosthetic mitral valve thrombosis as a result of 6-month lowmolecular-weight heparin (LMWH) (nadroparine) treatment failure. Our pat...
Karimi, Ashkan; Pourafshar, Negiin; Dibu, George; Beaver, Thomas M.; Bavry, Anthony A.
A 55-year-old male with a history of two prior cardiac surgeries presented with decompensated heart failure due to severe bioprosthetic aortic valve insufficiency. A third operation was viewed prohibitively high risk and valve-in-valve trans-catheter aortic valve replacement was considered. There were however several high-risk features and technically challenging aspects including low coronary ostia height, poor visualization of the aortic sinuses, and difficulty in identification of the copl...
Jeannette H. Spühler
Full Text Available Due to advances in medical imaging, computational fluid dynamics algorithms and high performance computing, computer simulation is developing into an important tool for understanding the relationship between cardiovascular diseases and intraventricular blood flow. The field of cardiac flow simulation is challenging and highly interdisciplinary. We apply a computational framework for automated solutions of partial differential equations using Finite Element Methods where any mathematical description directly can be translated to code. This allows us to develop a cardiac model where specific properties of the heart such as fluid-structure interaction of the aortic valve can be added in a modular way without extensive efforts. In previous work, we simulated the blood flow in the left ventricle of the heart. In this paper, we extend this model by placing prototypes of both a native and a mechanical aortic valve in the outflow region of the left ventricle. Numerical simulation of the blood flow in the vicinity of the valve offers the possibility to improve the treatment of aortic valve diseases as aortic stenosis (narrowing of the valve opening or regurgitation (leaking and to optimize the design of prosthetic heart valves in a controlled and specific way. The fluid-structure interaction and contact problem are formulated in a unified continuum model using the conservation laws for mass and momentum and a phase function. The discretization is based on an Arbitrary Lagrangian-Eulerian space-time finite element method with streamline diffusion stabilization, and it is implemented in the open source software Unicorn which shows near optimal scaling up to thousands of cores. Computational results are presented to demonstrate the capability of our framework.
Spühler, Jeannette H; Jansson, Johan; Jansson, Niclas; Hoffman, Johan
Due to advances in medical imaging, computational fluid dynamics algorithms and high performance computing, computer simulation is developing into an important tool for understanding the relationship between cardiovascular diseases and intraventricular blood flow. The field of cardiac flow simulation is challenging and highly interdisciplinary. We apply a computational framework for automated solutions of partial differential equations using Finite Element Methods where any mathematical description directly can be translated to code. This allows us to develop a cardiac model where specific properties of the heart such as fluid-structure interaction of the aortic valve can be added in a modular way without extensive efforts. In previous work, we simulated the blood flow in the left ventricle of the heart. In this paper, we extend this model by placing prototypes of both a native and a mechanical aortic valve in the outflow region of the left ventricle. Numerical simulation of the blood flow in the vicinity of the valve offers the possibility to improve the treatment of aortic valve diseases as aortic stenosis (narrowing of the valve opening) or regurgitation (leaking) and to optimize the design of prosthetic heart valves in a controlled and specific way. The fluid-structure interaction and contact problem are formulated in a unified continuum model using the conservation laws for mass and momentum and a phase function. The discretization is based on an Arbitrary Lagrangian-Eulerian space-time finite element method with streamline diffusion stabilization, and it is implemented in the open source software Unicorn which shows near optimal scaling up to thousands of cores. Computational results are presented to demonstrate the capability of our framework.
Du, Juan; Zhu, Tonghe; Yu, Haiyan; Zhu, Jingjing; Sun, Changbing; Wang, Jincheng; Chen, Sihao; Wang, Jihu; Guo, Xuran
Tissue engineering heart valves (TEHV) are thought to have many advantages in low immunogenicity, good histocompatibility, excellent mechanical properties. In this paper, we reported the fabrication and characterization of a novel composite nanofibrous scaffold consisting of silk fibroin (SF) and poly(ester-urethane) urea (LDI-PEUU) by using electrospinning. Chemical and physical properties of scaffolds were evaluated using scanning electron microscopy, attenuated total reflectance Fourier transform infrared, X-ray diffraction, contact angle measurement, thermogravimetric analysis, biodegradation test and tensile strength analysis. We determined that the composite scaffolds supported the growth of human umbilical vein endothelial cell (HUVEC). The results of cell proliferation and cell morphology indicate that SF/LDI-PEUU nanofibers promoted cell viability, which supporting the application in tissue engineering. All results clarified that SF/LDI-PEUU (40:60) nanofibrous scaffolds meet the required specifications for tissue engineering and could be used as a promising construct for heart valve tissue engineering.
Ritchie, R O; Dauskardt, R H; Pennisi, F J
A scanning electron microscopy study is reported of the nature and morphology of fracture surfaces in pyrocarbons commonly used for the manufacture of mechanical heart-valve prostheses. Specifically, silicon-alloyed low-temperature-isotropic (LTI)-pyrolytic carbon is examined, both as a coating on graphite and as a monolithic material, following overload, stress corrosion (static fatigue), and cyclic fatigue failures in a simulated physiological environment of 37 degrees C Ringer's solution. It is found that, in contrast to most metallic materials yet in keeping with many ceramics, there are no distinct fracture morphologies in pyro-carbons which are characteristic of a specific mode of loading; fracture surfaces appear to be identical for both catastrophic and subcritical crack growth under either sustained or cyclic loading. We conclude that caution should be used in assigning the likely cause of failure of pyrolytic carbon heart-valve components using fractographic examination.
Full Text Available Mechanical prosthetic valves are predisposed to bleeding, thrombosis & thromboembolic complications. Overall incidence of thromboembolic complications is 1% per year who are on oral anticoagulants, whereas bleeding complications incidence is 0.5% to 6.6% per year. 1, 2 Minimization of Scylla of thromboembolic & Charybdis of bleeding complication needs a balancing act of optimal antithrombotic therapy. We are reporting a case of middle aged male patient with prosthetic mitral valve presenting in heart failure. Patient had discontinued anticoagulants, as he had subdural hematoma in the past. He presented to our institute with a giant prosthetic valve thrombus.
Geidel, Stephan; Wohlmuth, Peter; Schmoeckel, Michael
The objective of this study was to analyze the results of open heart mitral valve operations for survival prediction in patients with previously unsuccessful MitraClip procedures. Thirty-three consecutive patients who underwent mitral valve surgery in our institution were studied. At a median of 41 days, they had previously undergone one to five futile MitraClip implantations. At the time of their operations, patients were 72.6 ± 10.3 years old, and the calculated risk, using the European System for Cardiac Operative Risk Evaluation (EuroSCORE) II, was a median of 26.5%. Individual outcomes were recorded, and all patients were monitored postoperatively. Thirty-day mortality was 9.1%, and the overall survival at 2.2 years was 60.6%. Seven cardiac-related and six noncardiac deaths occurred. Univariate survival regression models demonstrated a significant influence of the following variables on survival: EuroSCORE II (p = 0.0022), preoperative left ventricular end-diastolic dimension (p = 0.0052), left ventricular ejection fraction (p = 0.0249), coronary artery disease (p = 0.0385), and severe pulmonary hypertension (p = 0.0431). Survivors showed considerable improvements in their New York Heart Association class (p < 0.0001), left ventricular ejection fraction (p = 0.0080), grade of mitral regurgitation (p = 0.0350), and mitral valve area (p = 0.0486). Survival after mitral repair was not superior to survival after replacement. Indications for surgery after failed MitraClip procedures must be considered with the greatest of care. Variables predicting postoperative survival should be taken into account regarding the difficult decision as to whether to operate or not. Our data suggest that replacement of the pretreated mitral valve is probably the more reasonable concept rather than complex repairs. When the EuroSCORE II at the time of surgery exceeds 30%, conservative therapy is advisable. Copyright © 2016 The Society of Thoracic Surgeons. Published by Elsevier Inc
Zhang, Ruihang; Zhang, Yan
Flow past aortic valve represents a complex fluid-structure interaction phenomenon that involves pulsatile, vortical, and turbulent conditions. The flow characteristics immediately downstream of the valve, such as the variation of pulsatile flow velocity, formation of vortices, distribution of shear stresses, are of particular interest to further elucidate the role of hemodynamics in various aortic diseases. However, the fluid dynamics of a realistic aortic valve is not fully understood. Particularly, it is unclear how the flow fields downstream of the aortic valve would change under varied pulsatile inlet boundary conditions. In this study, an in vitro experiment has been conducted to investigate the flow fields downstream of a silicone aortic valve model within a cardiovascular flow simulator. Phased-locked Particle Image Velocimetry measurements were performed to map the velocity fields and Reynolds normal and shear stresses at different phases in a cardiac cycle. Temporal variations of pressure across the valve model were measured using high frequency transducers. Results have been compared for different pulsatile inlet conditions, including varied frequencies (heart rates), magnitudes (stroke volumes), and cardiac contractile functions (shapes of waveforms).
Marciniak, Anna; Glover, Keli; Sharma, Rajan
The aim of this study was to evaluate the proportion of suspected heart failure patients with significant valvular heart disease. Early diagnosis of valve disease is essential as delay can limit treatment and negatively affect prognosis for undiagnosed patients. The prevalence of unsuspected valve disease in the community is uncertain. We prospectively evaluated 79 043 patients, between 2001 and 2011, who were referred to a community open access echocardiography service for suspected heart failure. All patients underwent a standard transthoracic echocardiogram according to British Society of Echocardiography guidelines. Of the total number, 29 682 patients (37.5%) were diagnosed with mild valve disease, 8983 patients (11.3%) had moderate valve disease and 2134 (2.7%) had severe valve disease. Of the total number of patients scanned, the prevalence of aortic stenosis, aortic regurgitation, mitral stenosis, mitral regurgitation was 10%, 8.4%, 1%, and 12.5% respectively. 18% had tricuspid regurgitation. 5% had disease involving one or more valves. Of patients with suspected heart failure in the primary care setting, a significant proportion have important valvular heart disease. These patients are at high risk of future cardiac events and will require onward referral for further evaluation. We recommend that readily available community echocardiography services should be provided for general practitioners as this will result in early detection of valve disease. Published by the BMJ Publishing Group Limited. For permission to use (where not already granted under a licence) please go to http://www.bmj.com/company/products-services/rights-and-licensing/.
Voigt, Ingmar; Vitanovski, Dime; Ionasec, Razvan I.; Tsymal, Alexey; Georgescu, Bogdan; Zhou, Shaohua K.; Huber, Martin; Navab, Nassir; Hornegger, Joachim; Comaniciu, Dorin
Disorders of the heart valves constitute a considerable health problem and often require surgical intervention. Recently various approaches were published seeking to overcome the shortcomings of current clinical practice,that still relies on manually performed measurements for performance assessment. Clinical decisions are still based on generic information from clinical guidelines and publications and personal experience of clinicians. We present a framework for retrieval and decision support using learning based discriminative distance functions and visualization of patient similarity with relative neighborhood graphsbased on shape and derived features. We considered two learning based techniques, namely learning from equivalence constraints and the intrinsic Random Forest distance. The generic approach enables for learning arbitrary user-defined concepts of similarity depending on the application. This is demonstrated with the proposed applications, including automated diagnosis and interventional suitability classification, where classification rates of up to 88.9% and 85.9% could be observed on a set of valve models from 288 and 102 patients respectively.
Rabbah, Jean-Pierre; Saikrishnan, Neelakantan; Yoganathan, Ajit P
Numerical models of the mitral valve have been used to elucidate mitral valve function and mechanics. These models have evolved from simple two-dimensional approximations to complex three-dimensional fully coupled fluid structure interaction models. However, to date these models lack direct one-to-one experimental validation. As computational solvers vary considerably, experimental benchmark data are critically important to ensure model accuracy. In this study, a novel left heart simulator was designed specifically for the validation of numerical mitral valve models. Several distinct experimental techniques were collectively performed to resolve mitral valve geometry and hemodynamics. In particular, micro-computed tomography was used to obtain accurate and high-resolution (39 μm voxel) native valvular anatomy, which included the mitral leaflets, chordae tendinae, and papillary muscles. Three-dimensional echocardiography was used to obtain systolic leaflet geometry. Stereoscopic digital particle image velocimetry provided all three components of fluid velocity through the mitral valve, resolved every 25 ms in the cardiac cycle. A strong central filling jet (V ~ 0.6 m/s) was observed during peak systole with minimal out-of-plane velocities. In addition, physiologic hemodynamic boundary conditions were defined and all data were synchronously acquired through a central trigger. Finally, the simulator is a precisely controlled environment, in which flow conditions and geometry can be systematically prescribed and resultant valvular function and hemodynamics assessed. Thus, this work represents the first comprehensive database of high fidelity experimental data, critical for extensive validation of mitral valve fluid structure interaction simulations.
Alexeev, Alexander; Ballard, Matthew; Nepiyushchikh, Zhanna; Dixon, Brandon
The lymphatic vasculature is present in nearly all invertebrate tissue, and is essential in the transport of fluid and particles such as immune cells, antigens, proteins and lipids from the tissue to lymph nodes and to the venous circulation. Lymphatic vessels are made of up a series of contractile units that work together in harmony as "micro hearts" to pump fluid against a pressure gradient. Lymphatic valves are critical to this functionality, as they open and close with the oscillating pressure gradients from contractions, thus allowing flow in only one direction and leading to a net pumping effect. We use a hybrid lattice-Boltzmann lattice spring model which captures fluid-solid interactions through two-way coupling between a viscous fluid and lymphatic valves in a section of a lymphatic vessel to study the dynamics of lymphatic valves and their effect on fluid transport. Further, we investigate the effect of variations in valve geometry and material properties on fluid pumping. This work helps to increase our understanding of the mechanisms of lymphatic fluid transport, which has implications in a variety of pathologies, including cancer metastasis, autoimmunity, atherosclerosis and obesity. Support from NSF CMMI 1635133 is gratefully acknowledged.
Vidmar, J; Brilej, D; Voga, G; Kovacic, N; Smrkolj, V
Lesions of the heart valve caused by blunt chest trauma is rare, but when it does occur it can significantly injure the patient. On the basis of autopsy studies, research shows that heart valves are injured in less than 5% of patients who have died due to impact thoracic trauma. Among the heart valves, the aortic valve is the most often lacerated, which has been proved by relevant autopsy and clinical studies. Aortic valve lesions can be the only injury, but it is possible that additional heart or large vessel injuries are also present (myocardial contusion, rupture of the atrial septum, aortic rupture, rupture of the left common carotid artery). The force that causes such an injury is often great and often causes injuries to other organs and organ systems. In a multiple trauma patient, it is very important to specifically look for heart-related injuries because it is possible that they may be overlooked or missed by the surgeon, because of other obvious injuries. We describe the case of a 41-year-old man with multiple trauma who was diagnosed with aortic valve insufficiency due to rupture of the left coronary cusp 6 weeks after a road accident. Valvuloplasty was performed. Seven years later the patient is free of symptoms and is in good physical condition. Echocardiography showed normal dimensions of the heart chambers, a normal thickness of the heart walls, and normal systolic and diastolic function of the left ventricle. Heart valves are morphologically normal, and only an unimportant aortic insufficiency was noticed by echocardiography.
Tanis, W; Budde, R P J; van der Bilt, I A C; Delemarre, B; Hoohenkerk, G; van Rooden, J-K; Scholtens, A M; Habets, J; Chamuleau, S
Prosthetic heart valve (PHV) dysfunction remains difficult to recognise correctly by two-dimensional (2D) transthoracic and transoesophageal echocardiography (TTE/TEE). ECG-triggered multidetector-row computed tomography (MDCT), 18-fluorine-fluorodesoxyglucose positron emission tomography including low-dose CT (FDG-PET) and three-dimensional transoesophageal echocardiography (3D-TEE) may have additional value. This paper reviews the role of these novel imaging tools in the field of PHV obstruction and endocarditis.For acquired PHV obstruction, MDCT is of additional value in mechanical PHVs to differentiate pannus from thrombus as well as to dynamically study leaflet motion and opening/closing angles. For biological PHV obstruction, additional imaging is not beneficial as it does not change patient management. When performed on top of 2D-TTE/TEE, MDCT has additional value for the detection of both vegetations and pseudoaneurysms/abscesses in PHV endocarditis. FDG-PET has no complementary value for the detection of vegetations; however, it appears more sensitive in the early detection of pseudoaneurysms/abscesses. Furthermore, FDG-PET enables the detection of metastatic and primary extra-cardiac infections. Evidence for the additional value of 3D-TEE is scarce.As clinical implications are major, clinicians should have a low threshold to perform additional MDCT in acquired mechanical PHV obstruction. For suspected PHV endocarditis, both FDG-PET and MDCT have complementary value.
Ferrari, Enrico; Franciosi, Giorgio; Clivio, Sara; Faletra, Francesco; Moccetti, Marco; Moccetti, Tiziano; Pedrazzini, Giovanni; Demertzis, Stefanos
The goal was to show the technical details, feasibility and clinical results of balloon-expandable stent valve implantation in the aortic position during conventional redo open-heart surgery in selected obese patients with a small aortic prosthesis and severe patient-prosthesis mismatch. Two symptomatic overweight patients (body mass index of 31 and 38), each with a small aortic prosthesis (a 4-year-old, 21-mm Hancock II biological valve and a 29-year-old, 23-mm Duromedic mechanical valve), increased transvalvular gradients (59/31 and 74/44 mmHg) and a reduced indexed effective orifice area (0.50 and 0.43 cm 2 /m 2 ) underwent implantation of two 26-mm balloon-expandable Sapien 3 valves during standard on-pump redo valve surgery. Using full re-sternotomy, cardiopulmonary bypass and cardioplegic arrest, the two balloon-expandable stent valves were implanted under direct view using a standard aortotomy, after prosthesis removal and without annulus enlargement. Aortic cross-clamp times were 162 and 126 min; cardiopulmonary bypass times were 178 and 180 min; total surgical times were 360 and 318 min. At discharge, echocardiograms showed transvalvular peak and mean gradients of 13/9 and 23/13 mmHg and indexed effective orifice areas of 0.64 and 1.08 cm 2 /m 2 . The 3-month echocardiographic follow-up showed transvalvular peak and mean gradients of 18/9 and 19/11 mmHg and indexed effective orifice areas of 0.78 cm 2 /m 2 and 0.84 cm 2 /m 2 , with improved symptoms (New York Heart Association class 1). Implantation of a balloon-expandable stent valve during redo aortic valve surgery is feasible in selected cases and prevents patient-prosthesis mismatch in obese patients without need for aortic annulus enlargement. Moreover, in the case of stent valve degeneration, this approach permits additional valve-in-valve procedures with large stent valves and prevents re-redo surgery. © The Author 2016. Published by Oxford University Press on behalf of the
Bongert, Markus; Geller, Marius; Pennekamp, Werner; Nicolas, Volkmar
Diseases of the cardiovascular system account for nearly 42% of all deaths in the European Union. In Germany, approximately 12,000 patients receive surgical replacement of the aortic valve due to heart valve disease alone each year. A three-dimensional (3D) numerical model based on patient-specific anatomy derived from four-dimensional (4D) magnetic resonance imaging (MRI) data was developed to investigate preoperatively the flow-induced impact of mounting positions of aortic prosthetic valves to select the best orientation for individual patients. Systematic steady-state analysis of blood flow for different rotational mounting positions of the valve is only possible using a virtual patient model. A maximum velocity of 1 m/s was used as an inlet boundary condition, because the opening angle of the valve is at its largest at this velocity. For a comparative serial examination, it is important to define the standardised general requirements to avoid impacts other than the rotated implantation of the prosthetic aortic valve. In this study, a uniform velocity profile at the inlet for the inflow of the aortic valve and the real aortic anatomy were chosen for all simulations. An iterative process, with the weighted parameters flow resistance (1), shear stress (2) and velocity (3), was necessary to determine the best rotated orientation. Blood flow was optimal at a 45° rotation from the standard implantation orientation, which will offer a supply to the coronary arteries.
Pedersen, Thais A L; Wierup, Per; Pedersen, Lia Mendes
INTRODUCTION: Tricuspid valve regurgitation (TVR) is often secondary to left-sided or congenital heart disease (CHD). Surgical correction of TVR is indicated when the primary abnormalities require operation. MATERIAL AND METHODS: Retrospective analysis of all 50 patients (mean age: 65 years (rang...
Akutsu, Toshinosuke; Matsumoto, Akira
The current design of the bileaflet valve, the leaflets of which open outside first, differs significantly from the natural valve whose leaflets open center first. This difference generates a completely different flow field in the bileaflet valve compared to that in the natural heart valve. In a previous study, it was demonstrated that the valve design greatly affects the aortic flow field as well as the circulatory flow inside sinuses of Valsalva, using saline solution as a working fluid. A limited discussion on the turbulence flow field that could be generated by the valve was provided. In this continuation of that study, therefore, a dynamic PIV study was conducted to analyze the influence of the heart valve design on the aortic flow field, and particularly on the turbulent profile. This study also aimed to determine the influence of the viscosity of the testing fluid. Three bileaflet prostheses-the St. Jude Medical (SJM), the On-X, and the MIRA valves-were tested under pulsatile flow conditions. Flow through the central orifice of the SJM valve was slower than that through the newer designs. The newer designs tend to show strong flow through all orifices. The On-X valve generates simple jet-type flow while the MIRA valve with circumferentially curved leaflets generates a strong but three-dimensionally diffuse flow, resulting in a more complex flow field downstream of the aortic valve with higher turbulence. A 180° orientation that is more popular clinically seems to provide a less diffuse flow than a 90° orientation. The effect of increasing the viscosity was found to be an increase in the flow velocity through the central orifice and a more organized flow field for all of the valves tested.
... surgery; HF - surgery; Intra-aortic balloon pumps - heart failure; IABP - heart failure; Catheter based assist devices - heart failure ... problem may cause heart failure or make heart failure worse. Heart valve surgery may be needed to repair or ...
Biermann, Anna C; Marzi, Julia; Brauchle, Eva; Schneider, Maria; Kornberger, Angela; Abdelaziz, Sherif; Wichmann, Julian L; Arendt, Christophe T; Nagel, Eike; Brockbank, Kelvin G M; Seifert, Martina; Schenke-Layland, Katja; Stock, Ulrich A
Allogeneic frozen cryopreserved heart valves (allografts or homografts) are commonly used in clinical practice. A major obstacle for their application is the limited availability in particular for paediatrics. Allogeneic large animal studies revealed that alternative ice-free cryopreservation (IFC) results in better matrix preservation and reduced immunogenicity. The objective of this study was to evaluate xenogeneic (porcine) compared with allogeneic (ovine) IFC heart valves in a large animal study. IFC xenografts and allografts were transplanted in 12 juvenile merino sheep for 1-12 weeks. Immunohistochemistry, ex vivo computed tomography scans and transforming growth factor-β release profiles were analysed to evaluate postimplantation immunopathology. In addition, near-infrared multiphoton imaging and Raman spectroscopy were employed to evaluate matrix integrity of the leaflets. Acellular leaflets were observed in both groups 1 week after implantation. Allogeneic leaflets remained acellular throughout the entire study. In contrast, xenogeneic valves were infiltrated with abundant T-cells and severely thickened over time. No collagen or elastin changes could be detected in either group using multiphoton imaging. Raman spectroscopy with principal component analysis focusing on matrix-specific peaks confirmed no significant differences for explanted allografts. However, xenografts demonstrated clear matrix changes, enabling detection of distinct inflammatory-driven changes but without variations in the level of transforming growth factor-β. Despite short-term success, mid-term failure of xenogeneic IFC grafts due to a T-cell-mediated extracellular matrix-triggered immune response was shown.
Tien, W-H; Chen, H Y; Berwick, Z C; Krieger, J; Chambers, S; Dabiri, D; Kassab, G S
Chronic venous insufficiency (CVI) of the lower extremities is a common clinical problem. Although bioprosthetic valves have been proposed to treat severe reflux, clinical success has been limited due to thrombosis and neointima overgrowth of the leaflets that is, in part, related to the hemodynamics of the valve. A bioprosthetic valve that mimics native valve hemodynamics is essential. A computational model of the prosthetic valve based on realistic geometry and mechanical properties was developed to simulate the interaction of valve structure (fluid-structure interaction, FSI) with the surrounding flow. The simulation results were validated by experiments of a bioprosthetic bicuspid venous valve using particle image velocimetry (PIV) with high spatial and temporal resolution in a pulse duplicator (PD). Flow velocity fields surrounding the valve leaflets were calculated from PIV measurements and comparisons to the FSI simulation results were made. Both the spatial and temporal results of the simulations and experiments were in agreement. The FSI prediction of the transition point from equilibrium phase to valve-closing phase had a 7% delay compared to the PD measurements, while the PIV measurements matched the PD exactly. FSI predictions of reversed flow were within 10% compared to PD measurements. Stagnation or stasis regions were observed in both simulations and experiments. The pressure differential across the valve and associated forces on the leaflets from simulations showed the valve mechanism to be pressure driven. The flow velocity simulations were highly consistent with the experimental results. The FSI simulation and force analysis showed that the valve closure mechanism is pressure driven under the test conditions. FSI simulation and PIV measurements demonstrated that the flow behind the leaflet was mostly stagnant and a potential source for thrombosis. The validated FSI simulations should enable future valve design optimizations that are needed for
Felner, J.; Pollack, S.; McMillan, S.; Yoganathan, A.; Knopf, W.; Craver, J.
Color Doppler flow mapping (CDFM) was performed in 47 normal women ages 18-41 mean (m) 28 to determine the normal flow phenomena across each of the 4 heart valves. The group included: 15 elite marathon runners average run (ave.) 70 miles/week, heart rate (HR) 35-54 (m 45); 14 joggers ave. 40 mi/wkm HR 40-69 (m 53); and 17 controls 0 mi/wk, HR 49-93 (m 77). 2-D echo and CDFM defined the motion of each valve, including the presence of mitral valve prolapse (MVP), the ventricular inflow and outflow patterns and the presence of regurgitation
Baikoussis Nikolaos G
Full Text Available Abstract Mitral valve regurgitation is a relatively common and important heart valve lesion in clinical practice and adequate assessment is fundamental to decision on management, repair or replacement. Disease localised to the posterior mitral valve leaflet or focal involvement of the anterior mitral valve leaflet is most amenable to mitral valve repair, whereas patients with extensive involvement of the anterior leaflet or incomplete closure of the valve are more suitable for valve replacement. Echocardiography is the recognized investigation of choice for heart valve disease evaluation and assessment. However, the technique is depended on operator experience and on patient's hemodynamic profile, and may not always give optimal diagnostic views of mitral valve dysfunction. Cardiac catheterization is related to common complications of an interventional procedure and needs a hemodynamic laboratory. Cardiac magnetic resonance (MRI seems to be a useful tool which gives details about mitral valve anatomy, precise point of valve damage, as well as the quantity of regurgitation. Finally, despite of its higher cost, cardiac MRI using cine images with optimized spatial and temporal resolution can also resolve mitral valve leaflet structural motion, and can reliably estimate the grade of regurgitation.
Full Text Available Severe aortic valve stenosis (AVS can cause an exercise-induced reflex syncope (RS. The precise mechanism of this syncope is not known. The changes in hemodynamics are variable, including arrhythmias and myocardial ischemia, and one of the few consistent changes is a sudden fall in systemic and pulmonary arterial pressures (suggesting a reduced vascular resistance followed by a decline in heart rate. The contribution of the cardioinhibitory and vasodepressor components of the RS to hemodynamics was evaluated by a computer model. This lumped-parameter computer simulation was based on equivalent electronic circuits (EECs that reflect the hemodynamic conditions of a heart with severe AVS and a concomitantly decreased contractility as a long-term detrimental consequence of compensatory left ventricular hypertrophy. In addition, the EECs model simulated the resetting of the sympathetic nervous tone in the heart and systemic circuit during exercise and exercise-induced syncope, the fluctuating intra-thoracic pressure during respiration, and the passive relaxation of ventricle during diastole. The results of this simulation were consistent with the published case reports of exertional syncope in patients with AVS. The value of the EEC model is its ability to quantify the effect of a selective and gradable change in heart rate, ventricular contractility, or systemic vascular resistance on the hemodynamics during an exertional syncope in patients with severe AVS.
Full Text Available Pressure regulators in the field of pneumatic mechanisms can be used as valves for saving compressed air. For example it can be used to reduce the pressure when the piston rod is retracting unloaded and thus it is possible to save some energy. However the problem is that saving valve can significantly affect the dynamics of the pneumatic system. The lower pressure in the piston rod chamber causes extension of time for retraction of the piston rod. This article compare the air consumption experimentally determined and calculated, measured curves of pressure in cylinder chambers and piston speed when saving valve is set up differently.
Dvořák, Lukáš; Fojtášek, Kamil
Pressure regulators in the field of pneumatic mechanisms can be used as valves for saving compressed air. For example it can be used to reduce the pressure when the piston rod is retracting unloaded and thus it is possible to save some energy. However the problem is that saving valve can significantly affect the dynamics of the pneumatic system. The lower pressure in the piston rod chamber causes extension of time for retraction of the piston rod. This article compare the air consumption experimentally determined and calculated, measured curves of pressure in cylinder chambers and piston speed when saving valve is set up differently.
Khamooshian, Arash; Amador, Yannis; Hai, Ting; Jeganathan, Jelliffe; Saraf, Maria; Mahmood, Eitezaz; Matyal, Robina; Khabbaz, Kamal R.; Mariani, Massimo; Mahmood, Feroze
OBJECTIVE: To provide (1) an overview of the aortic valve (AV) apparatus anatomy and nomenclature, and (2) data regarding the normal AV apparatus geometry and dynamism during the cardiac cycle obtained from three-dimensional transesophageal echocardiography (3D TEE). DESIGN: Retrospective
Vaesken, Antoine; Heim, Frederic; Chakfe, Nabil
Transcatheter aortic valve replacement (TAVR) has become today a largely considered alternative technique to surgical valve replacement in patients who are not operable or patients with high risk for open chest surgery. However, the biological valve tissue used in the devices implanted clinically appears to be fragile material when folded for low diameter catheter insertion purpose and released in calcified environment with irregular geometry. Textile polyester material is characterized by outstanding folding and strength properties combined with proven biocompatibility. It could thereof be considered to replace biological valve leaflets in the TAVR procedure. The textile construction parameters must however be tuned to obtain a material compatible with the valve requested durability. In that context, one issue to be addressed is the friction effect that occurs between filaments and between yarns within a fabric under flexure loading. This phenomenon could be critical for the resistance of the material on the long term. The purpose of the present work is to assess the fatigue performances of textile valve prototypes made from different fabric constructions (monofilament, multifilament, calendered mutifilament) under accelerated cyclic loading. The goal is to identify, which construction is the best suited to long term fatigue stress. Results show that calendered multifilament and monofilament fabric constructions undergo strong ruptures already from 40 Mio cycles, while non calendered multifilament appears more durable. The rupture patterns observed point out that durability is directly related to the flexure stiffness level of the fibrous elements in the construction. Copyright © 2014 Elsevier Ltd. All rights reserved.
... discharge; Heart valve surgery - children - discharge; Heart surgery - pediatric - discharge; Heart transplant - pediatric - discharge ... Geme JW, Schor NF, eds. Nelson Textbook of Pediatrics . 20th ed. Philadelphia, PA: Elsevier; 2016:chap 434. ...
Sonmez, Yusuf; Kahraman, H. Tolga; Dosoglu, M. Kenan; Guvenc, Ugur; Duman, Serhat
In this study, symbiotic organisms search (SOS) algorithm is proposed to solve the dynamic economic dispatch with valve-point effects problem, which is one of the most important problems of the modern power system. Some practical constraints like valve-point effects, ramp rate limits and prohibited operating zones have been considered as solutions. Proposed algorithm was tested on five different test cases in 5 units, 10 units and 13 units systems. The obtained results have been compared with other well-known metaheuristic methods reported before. Results show that proposed algorithm has a good convergence and produces better results than other methods.
Piazza, Nicolo; Bleiziffer, Sabine; Brockmann, Gernot; Hendrick, Ruge; Deutsch, Marcus-André; Opitz, Anke; Mazzitelli, Domenico; Tassani-Prell, Peter; Schreiber, Christian; Lange, Rüdiger
This study sought to review the acute procedural outcomes of patients who underwent transcatheter aortic valve (TAV)-in-surgical aortic valve (SAV) implantation at the German Heart Center, Munich, and to summarize the existing literature on TAV-in-SAV implantation (n = 47). There are several case reports and small case series describing transcatheter aortic valve implantation for a failing surgical aortic valve bioprosthesis (TAV-in-SAV implantation). From January 2007 to March 2011, 20 out of 556 patients underwent a TAV-in-SAV implantation at the German Heart Center Munich. Baseline characteristics and clinical outcome data were prospectively entered into a dedicated database. The mean patient age was 75 ± 13 years, and the mean logistic European System for Cardiac Operative Risk Evaluation and Society of Thoracic Surgeons' Risk Model scores were 27 ± 13% and 7 ± 4%, respectively. Of the 20 patients, 14 had stented and 6 had stentless surgical bioprostheses. Most cases (12 of 20) were performed via the transapical route using a 23-mm Edwards Sapien prosthesis (Edwards Lifesciences, Irvine, California). Successful implantation of a TAV in a SAV with the patient leaving the catheterization laboratory alive was achieved in 18 of 20 patients. The mean transaortic valve gradient was 20.0 ± 7.5 mm Hg. None-to-trivial, mild, and mild-to-moderate paravalvular aortic regurgitation was observed in 10, 6, and 2 patients, respectively. We experienced 1 intraprocedural death following pre-implant balloon aortic valvuloplasty ("stone heart") and 2 further in-hospital deaths due to myocardial infarction. TAV-in-SAV implantation is a safe and feasible treatment for high-risk patients with failing aortic bioprosthetic valves and should be considered as part of the armamentarium in the treatment of aortic bioprosthetic valve failure. Copyright © 2011 American College of Cardiology Foundation. Published by Elsevier Inc. All rights reserved.
Karimi, Ashkan; Pourafshar, Negiin; Dibu, George; Beaver, Thomas M; Bavry, Anthony A
A 55-year-old male with a history of two prior cardiac surgeries presented with decompensated heart failure due to severe bioprosthetic aortic valve insufficiency. A third operation was viewed prohibitively high risk and valve-in-valve trans-catheter aortic valve replacement was considered. There were however several high-risk features and technically challenging aspects including low coronary ostia height, poor visualization of the aortic sinuses, and difficulty in identification of the coplanar view due to severe aortic insufficiency, and a highly mobile aortic valve mass. After meticulous peri-procedural planning, trans-catheter aortic valve replacement was carried out with a SAPIEN 3 balloon-expandable valve without any complication. Strategies undertaken to navigate the technically challenging aspects of the case are discussed.
Tanis, Wilco; Habets, Jesse; van den Brink, Renee B. A.; Symersky, Petr; Budde, Ricardo P. J.; Chamuleau, Steven A. J.
For acquired mechanical prosthetic heart valve (PHV) obstruction and suspicion on thrombosis, recently updated European Society of Cardiology guidelines advocate the confirmation of thrombus by transthoracic echocardiography, transesophageal echocardiography (TEE), and fluoroscopy. However, no
Mellwig, Klaus Peter; van Buuren, Frank; Gohlke-Baerwolf, Christa; Bjørnstad, Hans Halvor
Physical check-ups among athletes with valvular heart disease are of significant relevance. In athletes with mitral valve stenosis the extent of allowed physical activity is dependant on the size of the left atrium and the severity of the valve defect. Patients with mild-to-moderate mitral valve regurgitation can participate in all types of sport associated with low and moderate isometric stress and moderate dynamic stress. Patients under anticoagulation should not participate in any type of contact sport. Asymptomatic athletes with mild aortic valve stenosis can take part in all types of sport, as long as left ventricular function and size are normal, a normal response to exercise at the level performed during athletic activities is present and there are no arrhythmias. Asymptomatic athletes with moderate aortic valve stenosis should only take part in sports with low dynamic and static stress. Aortic valve regurgitation is often present due to connective tissue disease of a bicuspid valve. Athletes with mild aortic valve regurgitation, with normal end diastolic left ventricular size and systolic function can participate in all types of sport. A mitral valve prolapse is often associated with structural diseases of the myocardium and endocardium. In patients with mitral valve prolapse Holter-ECG monitoring should also be performed to detect significant arrhythmias. All athletes with known valvular heart disease, a previous history of infective endocarditis and valve surgery should receive endocarditis prophylaxis before dental, oral, respiratory, intestinal and genitourinary procedures associated with bacteraemia. Sport activities have to be avoided during active infection with fever.
Fukunaga, Naoto; Yuzaki, Mitsuru; Hamakawa, Hiroshi; Nasu, Michihiro; Takahashi, Yutaka; Okada, Yukikatsu
Cardiovascular surgery in the setting of chest wall deformities is a clinical challenge. Pectus excavatum, for example, can cause heart displacement to the left thoracic cavity, following the poor operative field. This report highlights a case in which a successful aortic valve-sparing operation via conventional median sternotomy after correction of the heart displacement due to pectus excavatum using Nuss procedure in Marfan syndrome. This technique can be one surgical option in Marfan syndrome patients with pectus excavatum and thoracic aortic aneurysm under close follow up.
Amirante, R.; Del Vescovo, G.; Lippolis, A.
The aim of the present paper is the evaluation of the driving forces acting on a 4/3 hydraulic open center directional control valve spool by means of a complete numerical analysis. In a previous paper by the same authors, the valve was inserted in a closed hydraulic circuit and was tested with different pump flow rate values to obtain experimental results about the driving forces. The experimental results are used in this paper to evaluate and validate the numerical analysis of the valve. The obtained numerical results show important differences between an open center valve and a closed center one, the latter being extensively analyzed in the literature. The numerical analysis is performed by using the commercial code 'Fluent', and the numerical results show the complete flow field inside the valve. The aim of this analysis is to evaluate the valve fluid dynamic performance, exploiting computational fluid dynamics (CFD) techniques, in order to give the reliable indications needed to define the valve design criteria and avoid expensive experimental tests
Treede, Hendrik; Schirmer, Johannes; Rudolph, Volker
Surgical mitral valve repair carries an elevated perioperative risk in the presence of severely reduced ventricular function and relevant comorbidities. We sought to assess the feasibility of catheter-based mitral valve repair using a clip-based percutaneous edge-to-edge repair system in selected...
Funder, Jonas Amstrup; Frost, Markus Winther; Wierup, Per; Klaaborg, Kaj-Erik; Hjortdal, Vibeke; Nygaard, Hans; Hasenkam, J Michael
Regions of turbulence downstream of bioprosthetic heart valves may cause damage to blood components, vessel wall as well as to aortic valve leaflets. Stentless aortic heart valves are known to posses several hemodynamic benefits such as larger effective orifice areas, lower aortic transvalvular pressure difference and faster left ventricular mass regression compared with their stented counterpart. Whether this is reflected by diminished turbulence formation, remains to be shown. We implanted either stented pericardial valve prostheses (Mitroflow), stentless valve prostheses (Solo or Toronto SPV) in pigs or they preserved their native valves. Following surgery, blood velocity was measured in the cross sectional area downstream of the valves using 10MHz ultrasonic probes connected to a dedicated pulsed Doppler equipment. As a measure of turbulence, Reynolds normal stress (RNS) was calculated at two different blood pressures (baseline and 50% increase). We found no difference in maximum RNS measurements between any of the investigated valve groups. The native valve had significantly lower mean RNS values than the Mitroflow (p=0.004), Toronto SPV (p=0.008) and Solo valve (p=0.02). There were no statistically significant differences between the artificial valve groups (p=0.3). The mean RNS was significantly larger when increasing blood pressure (p=0.0006). We, thus, found no advantages for the stentless aortic valves compared with stented prosthesis in terms of lower maximum or mean RNS values. Native valves have a significantly lower mean RNS value than all investigated bioprostheses. Copyright © 2011 Elsevier Ltd. All rights reserved.
Successful thrombolysis of aortic prosthetic valve thrombosis during first trimester of pregnancy. A Shukla, AP Raval, R Shah. Abstract. Prosthetic heart valve thrombosis during pregnancy is life-threatening. Standard surgical treatment using cardiopulmonary bypass carries high maternal and fetal complications. Here we ...
... may be able to replace it with an artificial (man-made) valve. Cardiac size reduction . During this procedure, your doctor removes a piece of the heart muscle from an enlarged heart. This makes your heart ...
Arnáiz-García, María Elena; González-Santos, Jose María; Bueno-Codoñer, María E; López-Rodríguez, Javier; Dalmau-Sorlí, María José; Arévalo-Abascal, Adolfo; Arribas-Jiménez, Antonio; Diego-Nieto, Alejandro; Rodríguez-Collado, Javier; Rodríguez-López, Jose María
A 78-year-old woman was admitted to our institution with progressive dyspnea. She had previously been diagnosed with rheumatic heart disease and had undergone cardiac surgery for mechanical mitral valve replacement ten years previously. Transesophageal echocardiography revealed blockage of the mechanical prosthesis and the patient was scheduled for surgery, in which a thrombus was removed from the left atrial appendage. A partial thrombosis of the mechanical prosthesis and circumferential pannus overgrowth were concomitantly detected. Prosthetic heart valve blockage is a rare but life-threatening complication, the main causes of which are thrombosis and pannus formation. The two conditions are different but both are usually misdiagnosed. Two concurrent mechanisms of prosthesis blockage were found in this patient. Copyright © 2014 Sociedade Portuguesa de Cardiologia. Published by Elsevier España. All rights reserved.
Stępak, Bogusz; Dzienny, Paulina; Franke, Volker; Kunicki, Piotr; Gotszalk, Teodor; Antończak, Arkadiusz
Laser-induced periodic surface structures (LIPSS) are highly periodic wavy surface features which are frequently smaller than incident light wavelength that bring possibility of nanostructuring of many materials. In this paper the possibility of using them to homogeneously structure the surface of artificial heart valve made of PyC was examined. By changing laser irradiation parameters such like energy density and pulse separation the most suitable conditions were established for 1030 nm wavelength. A wide spectrum of periodicities and geometries was obtained. Interesting side effects like creating a thin shell-like layer were observed. Modified surfaces were examined using EDX and Raman spectroscopy to determine change in elemental composition of surface.
Zhuravlyova, L.; Lopina, N.; Kuznetsov, I.; Lopin, D.; Kramarenko, I.; Sumanova, I.
The article describes the etiologic role of various factors such as chronic rheumatic heart disease, bacterial endocarditis, aortic valve stenosis, hypertension, coronary heart disease, including acute myocardial infarction, degenerative disease of the valves, connective tissue diseases, chest trauma, systemic lupus erythematosus, con-genital heart disease which leading to the mitral valve chords rupture, gives the current classification of the nosology and also features of manifestation and ...
Rabus, Murat; Demirbağ, Recep; Sezen, Yusuf; Konukoğlu, Oğuz; Yildiz, Ali; Erel, Ozcan; Zeybek, Rahmi; Yakut, Cevat
We investigated whether patients with rheumatic and degenerative heart valve disease (HVD) differed with regard to plasma and tissue oxidative stress index (OSI). The study included 56 patients who underwent valve replacement due to rheumatic (n=32; 15 males; mean age 47+/-10 years) and degenerative (n=24; 13 males; mean age 55+/-12 years) HVD. Plasma and tissue total oxidative status (TOS) and total antioxidative capacity (TAC) levels were measured and OSI was calculated. Patients with degenerative HVD had significantly higher age, increased interventricular septum thickness, and higher frequency of aortic stenosis, whereas the incidence of mitral stenosis was higher in patients with rheumatic HVD (p0.05). Tissue TAC was significantly lower in patients with rheumatic HVD (p=0.027), whereas tissue TOS and OSI were similar between the two HVD groups (p>0.05). In bivariate analysis, plasma OSI did not show any correlation with clinical, laboratory, and echocardiographic variables (p>0.05). Our data show that plasma and tissue OSI levels are similar in patients with rheumatic and degenerative HVD.
Estudo comparativo das características fluidodinâmicas de próteses valvulares biológicas de pericárdio bovino de perfil alto e baixo Comparative study of fluid dynamic characteristics in high and low biological prosthetic heart valves from bovine pericardium
Aron J. P Andrade
Full Text Available O comportamento fluidodinâmico e a vida útil média de uma prótese valvular cardíaca são características muito importantes na escolha do modelo de prótese a ser implantada em um determinado paciente. Por esse motivo, foi realizado um estudo comparativo entre as válvulas biológicas de perfil alto e baixo, para se verificarem possíveis alterações na vida média, gradiente de pressão e refluxo, causadas pela diminuição do perfil dessas próteses. Com o auxílio de um duplicador de pulso, foram analisadas cinco próteses de perfil alto e cinco de perfil baixo, todas tamanho 29, fabricadas no Instituto Dante Pazzanese de Cardiologia (IDPC. Mantendo-se a freqüência de batimentos em 100 bpm e a diferença de pressão em 1000 mmHg, verificou-se que a vida útil média da válvula de perfil baixo permaneceu equivalente à de perfil alto. Em outro equipamento, denominado simulador cardíaco, as próteses foram submetidas a condições semelhantes às condições fisiológicas de funcionamento para uma freqüência cardíaca de 75 bpm. Os resultados mostraram que o volume de retorno necessário para fechamento (refluxo, nas válvulas de perfil baixo, é menor do que nas de perfil alto. Isso traz um melhor aproveitamento do efeito de bombeamento do coração. As perdas de carga das válvulas foram comparadas através dos gradientes máximos de pressão visibilizados com a sobreposição dos sinais de pressão. Observou-se que as válvulas de perfil baixo oferecem menos resistência ao fluxo; assim, quando são implantadas, possibilitam uma passagem mais livre e desobstruída do sangue.The fluid dynamic performance and lifetime of a prosthetic heart valve are one of the most important characteristics to determine the choice of what prosthesis must be applied to a patient. Because of this, two different biological prosthetic heart valves profiles (one high profile and other low profile were studied, to verify possible alterations of working
Podesser, B K; Khuenl-Brady, G; Eigenbauer, E; Roedler, S; Schmiedberger, A; Wolner, E; Moritz, A
The Edwards Duromedics valve (Baxter Healthcare Corp., Edwards Division, Santa Ana, Calif.) was designed with a self-irrigating hinge mechanism to reduce thromboembolic complications. After good initial clinical results, distribution was suspended in 1988 after reports of valve fracture after 20,000 valves had been implanted. The manufacturer conducted extensive studies to improve the Edwards Duromedics and reintroduced a modified version, which is available as Edwards Tekna. The purpose of the study was the evaluation of long-term results of the original Edwards Duromedics that might be important for the current version, the Edwards Tekna valve. A prospective clinical 10-year follow-up was performed of 508 patients who underwent valve replacement with the Edwards Duromedics valve in the aortic (n = 268), mitral (n = 183), and aortic and mitral (n = 56) position. The perioperative mortality rate was 6.9%; follow-up was 98% complete, comprising 3648 patient-years for a mean follow-up of 86 months (range: 33 to 144 months). The actuarial freedom from complications at the 10-year follow-up and the incidence rate (percent per patient-year) were as follows: late mortality rate, 69.2% +/- 2.4% (3.5% per patient-year); thromboembolism, 90.7% +/- 1.6% (0.96% per patient-year); anticoagulation-related hemorrhage, 87.7% +/- 1.7% (1.34% per patient-year); prosthetic valve endocarditis, 96.7% +/- 0.09% (0.38% per patient-year); valve-related mortality rate, 89.3% +/- 1.6% (1.21% per patient-year); valve failure, 86.2% +/- 1.85% (1.54% per patient-year); and valve-related morbidity and mortality rate, 71.1% +/- 2.3% (3.2% per patient-year). Three leaflet escapes were observed (one lethal, two successful reoperations; 99.1% +/- 0.05% freedom, 0.08% per patient-year). All patients functionally improved (86% in New York Heart Association classes I and II), and incidence of anemia was insignificant. These results confirm that the Edwards Duromedics valve shows excellent performance
Marzec, M.; Lasek, W.; Serafin, Z.; Laskowska, K.; Swiatkiewicz, I.; Kubica, J.
Bicuspid aortic valve (BAV) is one of the most common congenital heart defects and often coexists with other congenital abnormalities of the heart and great vessels. A standard diagnostic technique for valve imaging is echocardiography. However, suboptimal images may cause a diagnostic problem, especially in obese patients, women and in case of significant valve calcifications. The paper reports a case of a 27-year-old male patient with recurrent chest pain, appearing regardless of exertion. Routine physical examination, laboratory tests and echocardiography were negative. Multislice computed tomography (MSCT) showed normal coronary arteries and bicuspid aortic valve. The diagnostics of bicuspid aortic valve and the diagnostic value of MSCT are discussed. MSCT seems to be an important point in cardiac diagnostic algorithms, being an alternative modality for MR in BAV imaging. The presented case emphasizes the potential of MSCT for the diagnostics of cardiac valve congenital diseases. (author)
Valente, M; Minarini, M; Maizza, A F; Bortolotti, U; Thiene, G
Long-term experience with first generation porcine valve xenografts enabled identification of the major limitations to their durability: (1) prosthetic-ventricular mismatch due to the high profile of the stent in patients with mitral stenosis and a small left ventricle; (2) high-pressure fixation with loss of natural collagen crimping in the fibrosa, and wash-out of proteoglycans in the spongiosa; (3) xenograft tissue autolysis, due to the long interval between animal slaughter and aortic valve removal fixation; (4) muscle shelf in the right coronary cusp, which created a gradient and could undergo accelerated calcification and/or spontaneous perforation with time; (5) a flexible polypropylene stent, which could creep or even fracture with consequent inward bending of the stent; (6) progressive time-related dystrophic calcification; (7) host fibrous tissue ingrowth. An awareness of these limitations stimulated technical modifications, which frequently brought about distinct improvements: (1) the reduction of the stent profile eliminated the problem of mismatch, but resulted in a higher tendency towards cusp prolapse and earlier commissural tearing; (2) natural collagen waviness, proteoglycans and cusp extensibility were preserved by employing low or even zero pressure during the fixation process; (3) earlier valve fixation enabled preservation of cell integrity; (4) a new orifice for small valves was designed by replacing the right muscular cusp, thus achieving less gradient and avoiding muscle-shelf-related complications; (5) polypropylene was replaced by Delrin as stent material; (6) calcium-retarding agents like T6 and toluidine blue were applied during commercial processing and storage in order to mitigate tissue mineralization.(ABSTRACT TRUNCATED AT 250 WORDS)
Straka, František; Schorník, David; Mašín, J.; Filová, Elena; Miřejovský, T.; Burdíková, Z.; Švindrych, Z.; Chlup, H.; Horný, L.; Veselý, J.; Pirk, J.; Bačáková, Lucie
Roč. 7, č. 7 (2017), s. 527-543 ISSN 2157-9083 R&D Projects: GA MZd(CZ) NV15-29153A; GA MZd(CZ) NT11270 Institutional support: RVO:67985823 Keywords : autologous human pericardium * pericardial interstitial cells * heart valve * 3D mechanotranduction * bioreactor Subject RIV: FA - Cardiovascular Diseases incl. Cardiotharic Surgery OBOR OECD: Cardiac and Cardiovascular systems Impact factor: 1.383, year: 2016
Mart, Christopher Robin; Eckhauser, Aaron Wesley; Murri, Michael; Su, Jason Thomas
With surgical palliation of hypoplastic left heart syndrome (HLHS), the tricuspid valve (TV) becomes the systemic atrioventricular valve and moderate/severe TV insufficiency (TVI), an adverse risk factor for survival to Fontan, has been reported in up to 35% of patients prior to stage I palliation. Precise echocardiographic identification of the mechanism of TVI cannot be determined by two-dimensional echocardiography. Three-dimensional echocardiography (3DE) can provide significant insight into the mechanisms of TVI. It is the intent of this report to propose a systematic method on how to evaluate and display 3DE images of the TV in HLHS which has not been done previously. TV anatomy, function, and the known mechanisms of insufficiency are reviewed. We defined three regions of the TV (anterior, posterior, septal) that can help define valve "leaflets" that incorporates the many variations of TV anatomy. To determine how the surgeon views the TV, a picture of a pathologic specimen of the TV was placed on a computer screen and rotated until it was oriented as it appears during surgery, the "surgeons view." We have proposed a systematic method for evaluating and displaying the TV using 3DE which can provide significant insight into the mechanisms causing TVI in HLHS. This has the potential to improve both the surgical approach to repairing the valve and, ultimately, patient outcomes.
... and Live Our Interactive Cardiovascular Library has detailed animations and illustrations to help you learn about conditions, treatments and procedures related to heart disease and stroke. Popular Articles 1 Understanding Blood Pressure Readings 2 Sodium and Salt 3 Heart Attack Symptoms ...
Zou, Liang-jian; Xu, Zhi-yun; Wang, Zhi-nong; Lang, Xi-long; Han, Lin; Lu, Fang-lin; Xu, Ji-bin; Tang, Hao; Ji, Guang-yu; Wang, Er-song; Wang, Jun; Qu, Yi
To review the experience of reoperative valve replacement for 104 patients. From January 2002 to December 2009, 104 patients underwent heart valve replacement in reoperations, accounting for 2.92% of the total patient population (3557 cases) who had valve replacement during this period. In this group, 53 male and 51 female patients were included with a median age of 46 years (ranged from 13 to 72 years). The reasons of reoperation included 28 cases suffered from another valve lesion after valve replacement, 10 cases suffered from valve lesion after mitral valvuloplasty, 19 cases suffered from perivalvular leakage after valve replacement, 18 cases suffered from valve lesion after previous correction of congenital heart defect, 7 cases suffered from bioprosthetic valve decline, 10 cases suffered from prosthetic valve endocarditis, 9 cases suffered from dysfunction of machine valve, and 3 cases suffered from other causes. The re-operations were mitral and aortic valve replacement in 2 cases, mitral valve replacement in 59 cases, aortic valve replacement in 24 cases, tricuspid valve replacement in 16 cases, and Bentall's operation in 3 cases. The interval from first operation to next operation was 1 month-19 years. There were 8 early deaths from heart failure, renal failure and multiple organ failure (early mortality 7.69%). Major complications were intraoperative hemorrhage in 2 cases, re-exploration for mediastinal bleeding in 2 cases and sternotomy surgical site infection in 1 case. Complete follow-up (3 months-7 years and 2 months) was available for all patients. Two patients died, one patient died of intracranial hemorrhage, and another cause was unknown. Satisfactory short-term and long-term results can be obtained in reoperative valve replacement with appropriate timing of operation control, satisfactory myocardial protection, accurate surgical procedure and suitable perioperative treatment.
Full Text Available Endocardial to mesenchymal transformation (EMT is a fundamental cellular process required for heart valve formation. Notch, Wnt and Bmp pathways are known to regulate this process. To further address how these pathways coordinate in the process, we specifically disrupted Notch1 or Jagged1 in the endocardium of mouse embryonic hearts and showed that Jagged1-Notch1 signaling in the endocardium is essential for EMT and early valvular cushion formation. qPCR and RNA in situ hybridization assays reveal that endocardial Jagged1-Notch1 signaling regulates Wnt4 expression in the atrioventricular canal (AVC endocardium and Bmp2 in the AVC myocardium. Whole embryo cultures treated with Wnt4 or Wnt inhibitory factor 1 (Wif1 show that Bmp2 expression in the AVC myocardium is dependent on Wnt activity; Wnt4 also reinstates Bmp2 expression in the AVC myocardium of endocardial Notch1 null embryos. Furthermore, while both Wnt4 and Bmp2 rescue the defective EMT resulting from Notch inhibition, Wnt4 requires Bmp for its action. These results demonstrate that Jagged1-Notch1 signaling in endocardial cells induces the expression of Wnt4, which subsequently acts as a paracrine factor to upregulate Bmp2 expression in the adjacent AVC myocardium to signal EMT.
Full Text Available BACKGROUND Diseases of heart valves constitute a major cause of cardiovascular morbidity and mortality worldwide. In developing countries, Rheumatic Heart Disease (RHD continues to be the predominant form of valvular heart disease. The current study was undertaken at a Tertiary Care Institute with an objective of establishing distribution and different patterns of valvular heart diseases by echocardiography. MATERIALS AND METHODS 17,625 consecutive first time Echocardiograms performed between January 2016 and December 2016 were analysed. Echo was performed by consultant cardiologists using Philips HD11XE and Aloka SSD4000 machine following ASE guidelines. Applying exclusion criteria of trivial and functional regurgitant lesions yielded a total of 632 cases of organic valvular heart diseases. RESULTS In our study 632 patients were diagnosed with valvular heart disease, out of which 428 patients (67.7% were diagnosed with Rheumatic Heart Disease. Mitral valve was the most commonly affected followed by aortic and tricuspid valves. The least commonly affected valve was pulmonary valve. In Rheumatic heart disease, most common isolated lesion reported was MS with MR, most commonly reported in females between 21 - 40 years’ age group. CONCLUSION In non-RHD group, mitral valve prolapse (21.3% was the commonest lesion reported followed by calcific degenerative aortic valve (6.17% and congenital bicuspid aortic valve (3.4%; 118 patients were reported with multivalvular lesion. MS + MR + AR was the commonest multivalvular lesion found in 65 patients (55.08%.
Harky, Amer; Garner, Megan; Popa, Miruna; Shipolini, Alex
Infective endocarditis is a rare disease associated with high morbidity and mortality. As a result, early diagnosis and prompt antibiotic treatment with or without surgical intervention is crucial in the management of such condition.We report a case of missed infective endocarditis of the aortic valve. The patient underwent mechanical aortic valve replacement, with the native valve being sent for histopathological examination. On re-admission 16 months later, he presented with syncope, shortness of breathing and complete heart block. On review of the histopathology of native aortic valve, endocarditis was identified which had not been acted on. The patient underwent redo aortic valve replacement for severe aortic regurgitation.We highlight the importance of following up histopathological results as well as the need for multidisciplinary treatment of endocarditis with a combination of surgical and antibiotic therapy. © BMJ Publishing Group Ltd (unless otherwise stated in the text of the article) 2017. All rights reserved. No commercial use is permitted unless otherwise expressly granted.
Voigt, Ingmar; Ionasec, Razvan Ioan; Georgescu, Bogdan; Houle, Helene; Huber, Martin; Hornegger, Joachim; Comaniciu, Dorin
Disorders of the mitral valve are second most frequent, cumulating 14 percent of total number of deaths caused by Valvular Heart Disease each year in the United States and require elaborate clinical management. Visual and quantitative evaluation of the valve is an important step in the clinical workflow according to experts as knowledge about mitral morphology and dynamics is crucial for interventional planning. Traditionally this involves examination and metric analysis of 2D images comprising potential errors being intrinsic to the method. Recent commercial solutions are limited to specific anatomic components, pathologies and a single phase of cardiac 4D acquisitions only. This paper introduces a novel approach for morphological and functional quantification of the mitral valve based on a 4D model estimated from ultrasound data. A physiological model of the mitral valve, covering the complete anatomy and eventual shape variations, is generated utilizing parametric spline surfaces constrained by topological and geometrical prior knowledge. The 4D model's parameters are estimated for each patient using the latest discriminative learning and incremental searching techniques. Precise evaluation of the anatomy using model-based dynamic measurements and advanced visualization are enabled through the proposed approach in a reliable, repeatable and reproducible manner. The efficiency and accuracy of the method is demonstrated through experiments and an initial validation based on clinical research results. To the best of our knowledge this is the first time such a patient specific 4D mitral valve model is proposed, covering all of the relevant anatomies and enabling to model the common pathologies at once.
Madalena D. Costa
Full Text Available Background: We recently introduced the concept of heart rate fragmentation along with a set of metrics for its quantification. The term was coined to refer to an increase in the percentage of changes in heart rate acceleration sign, a dynamical marker of a type of anomalous variability. The effort was motivated by the observation that fragmentation, which is consistent with the breakdown of the neuroautonomic-electrophysiologic control system of the sino-atrial node, could confound traditional short-term analysis of heart rate variability.Objective: The objectives of this study were to: (1 introduce a symbolic dynamical approach to the problem of quantifying heart rate fragmentation; (2 evaluate how the distribution of the different dynamical patterns (“words” varied with the participants' age in a group of healthy subjects and patients with coronary artery disease (CAD; and (3 quantify the differences in the fragmentation patterns between the two sample populations.Methods: The symbolic dynamical method employed here was based on a ternary map of the increment NN interval time series and on the analysis of the relative frequency of symbolic sequences (words with a pre-defined set of features. We analyzed annotated, open-access Holter databases of healthy subjects and patients with CAD, provided by the University of Rochester Telemetric and Holter ECG Warehouse (THEW.Results: The degree of fragmentation was significantly higher in older individuals than in their younger counterparts. However, the fragmentation patterns were different in the two sample populations. In healthy subjects, older age was significantly associated with a higher percentage of transitions from acceleration/deceleration to zero acceleration and vice versa (termed “soft” inflection points. In patients with CAD, older age was also significantly associated with higher percentages of frank reversals in heart rate acceleration (transitions from acceleration to
Alkashkari, Wail; Alsubei, Amani; Hijazi, Ziyad M
The past couple of decades have brought tremendous advances to the field of pediatric and adult congenital heart disease (CHD). Percutaneous valve interventions are now a cornerstone of not just the congenital cardiologist treating patients with congenital heart disease, but also-and numerically more importantly-for adult interventional cardiologists treating patients with acquired heart valve disease. Transcatheter pulmonary valve replacement (tPVR) is one of the most exciting recent developments in the treatment of CHD and has evolved to become an attractive alternative to surgery in patients with right ventricular outflow tract (RVOT) dysfunction. This review aims to summarize (1) the current state of the art for tPVR, (2) the expanding indications, and (3) the technological obstacles to optimizing tPVR. Since its introduction in 2000, more than ten thousands tPVR procedures have been performed worldwide. Although the indications for tPVR have been adapted earlier from those accepted for surgical intervention, they remain incompletely defined. The new imaging modalities give better assessment of cardiac anatomy and function and determine candidacy for the procedure. The procedure has been shown to be feasible and safe when performed in patients who received pulmonary conduit and or bioprosthetic valves between the right ventricle and the pulmonary artery. Fewer selected patients post trans-annular patch repair for tetralogy of Fallot may also be candidates for this technology. Size restrictions of the currently available valves limit deployment in the majority of patients post trans-annular patch repair. Newer valves and techniques are being developed that may help such patients. Refinements and further developments of this procedure hold promise for the extension of this technology to other patient populations.
Arun Kumar Agnihotri
patients with valvular heart disease). Endorsed by the Society of Cardiovascular. Anesthesiologists, Society for Cardiovascular. Angiography and Interventions, and Society of. Thoracic Surgeons. J Am Coll Cardiol. 2008;52(13):e1-142. 5. Elkayam U, Bitar F. Valvular heart disease and pregnancy. Part II: prosthetic valves.
Full Text Available Nitric oxide synthase-3 (NOS3 has recently been shown to promote endothelial-to-mesenchymal transition (EndMT in the developing atrioventricular (AV canal. The present study was aimed to investigate the role of NOS3 in embryonic development of AV valves. We hypothesized that NOS3 promotes embryonic development of AV valves via EndMT. To test this hypothesis, morphological and functional analysis of AV valves were performed in wild-type (WT and NOS3(-/- mice at postnatal day 0. Our data show that the overall size and length of mitral and tricuspid valves were decreased in NOS3(-/- compared with WT mice. Echocardiographic assessment showed significant regurgitation of mitral and tricuspid valves during systole in NOS3(-/- mice. These phenotypes were all rescued by cardiac specific NOS3 overexpression. To assess EndMT, immunostaining of Snail1 was performed in the embryonic heart. Both total mesenchymal and Snail1(+ cells in the AV cushion were decreased in NOS3(-/- compared with WT mice at E10.5 and E12.5, which was completely restored by cardiac specific NOS3 overexpression. In cultured embryonic hearts, NOS3 promoted transforming growth factor (TGFβ, bone morphogenetic protein (BMP2 and Snail1expression through cGMP. Furthermore, mesenchymal cell formation and migration from cultured AV cushion explants were decreased in the NOS3(-/- compared with WT mice. We conclude that NOS3 promotes AV valve formation during embryonic heart development and deficiency in NOS3 results in AV valve insufficiency.
Yamamura, Mitsuhiro; Miyamoto, Yuji; Mitsuno, Masataka; Tanaka, Hiroe; Ryomoto, Masaaki; Fukui, Shinya; Tsujiya, Noriko; Kajiyama, Tetsuya; Nojima, Michio
to evaluate the strategy for open heart surgery after renal transplantation performed in a single institution in Japan. we reviewed 6 open heart surgeries after renal transplantation in 5 patients, performed between January 1992 and December 2012. The patients were 3 men and 2 women with a mean age of 60 ± 11 years (range 46-68 years). They had old myocardial infarction and unstable angina, aortic and mitral stenosis, left arterial myxoma, aortic stenosis, and native valve endocarditis followed by prosthetic valve endocarditis. Operative procedures included coronary artery bypass grafting, double-valve replacement, resection of left arterial myxoma, 2 aortic valve replacements, and a double-valve replacement. Renal protection consisted of steroid cover (hydrocortisone 100-500 mg or methylprednisolone 1000 mg) and intravenous immunosuppressant infusion (cyclosporine 30-40 mg day(-1) or tacrolimus 1.0 mg day(-1)). 5 cases were uneventful and good renal graft function was maintained at discharge (serum creatinine 2.1 ± 0.5 mg dL(-1)). There was one operative death after emergency double-valve replacement for methicillin-resistant Staphylococcus aureus-associated prosthetic valve endocarditis. Although the endocarditis improved after valve replacement, the patient died of postoperative pneumonia on postoperative day 45. careful perioperative management can allow successful open heart surgery after renal transplantation. However, severe complications, especially methicillin-resistant Staphylococcus aureus infection, may cause renal graft loss. © The Author(s) 2013 Reprints and permissions: sagepub.co.uk/journalsPermissions.nav.
Hsi, David H.; Ryan, Gerald F.; Taft, Janice; Arnone, Thomas J.
An 81-year-old woman was evaluated for prosthetic mitral valve function. She had received a Harken disk mitral valve 29 years earlier due to severe mitral valve disease. This particular valve prosthesis is known for premature disk edge wear and erosion. The patient's 2-dimensional Doppler echocardiogram showed the distinctive appearance of a disk mitral valve prosthesis. Color Doppler in diastole showed a unique crown appearance, with initial flow acceleration around the disk followed by convergence to laminar flow in the left ventricle. Cineradiographic imaging revealed normal valve function and minimal disk erosion. We believe this to be the longest reported follow-up of a surviving patient with a rare Harken disk valve. We present images with unique echocardiographic and cineangiographic features. (Tex Heart Inst J 2003;30:319–21) PMID:14677746
Geidel, W.; Leimbach, K.R.
The sudden closing of a valve in a piping system causes a build-up of pressure which, in turn, causes severe vibrations of the structural system. The licensing procedure calls for on-site tests to determine the dynamic effects of such closing of valves, and to check the stresses and displacements against the allowable ones. The measurements include time histories of displacements, accelerations and internal pressure. The computer program KWUROHR for the static and dynamic analysis of piping systems has been used by KWU and several subcontractors during the past four vears. This program has been extended by adding a subroutine package which computes time histories of displacements, accelerations and stresses resulting from the input of measured time histories of internal pressures at selected locations. The computer algorithm establishes the topological connectivity between the internal pressure measuring locations, to set up a logic for linear pressure interpolation between these points and pressure steps at reducers and valves. A minimum number of input points is required to give realistic results. (orig.)
Carev, Mladen; Bulat, Cristijan; Karanović, Nenad; Lojpur, Mihajlo; Jercić, Antonio; Nenadić, Denis; Marovih, Zlatko; Husedzinović, Ino; Letica, Dalibor
Secondary pulmonary hypertension is a frequent condition after heart valve surgery. It may significantly complicate the perioperative management and increase patients' morbidity and mortality. The treatment has not been yet completely defined principally because of lack of the selectivity of drugs for the pulmonary vasculature. The usage of inhaled milrinone could be the possible therapeutic option. Inodilator milrinone is commonly used intravenously for patients with pulmonary hypertension and ventricular dysfunction in cardiac surgery. The decrease in systemic vascular resistance frequently necessitates concomitant use of norepinephrine. Pulmonary vasodilators might be more effective and also devoid of potentially dangerous systemic side effects if applied by inhalation, thus acting predominantly on pulmonary circulation. There are only few reports of inhaled milrinone usage in adult post cardiac surgical patients. We reported 2 patients with severe pulmonary hypertension after valve surgery. Because of desperate clinical situation, we decided to use the combination of inhaled and intravenous milrinone. Inhaled milrinone was delivered by means of pneumatic medication nebulizer dissolved with saline in final concentration of 0.5 mg/ml. The nebulizer was attached to the inspiratory limb of the ventilator circuit, just before the Y-piece. We obtained satisfactory reduction in mean pulmonary artery pressure in both patients, and they were successfully extubated and discharged. Although it is a very small sample of patients, we conclude that the combination of inhaled and intravenous milrinone could be an effective treatment of secondary pulmonary hypertension in high-risk cardiac valve surgery patient. The exact indications for inhaled milrinone usage, optimal concentrations for this route, and the beginning and duration of treatment are yet to be determined.
Kersting-Sommerhoff, B.A.; Diethelm, L.; Teitel, D.F.; Sommerhoff, C.P.; Higgins, C.B.
Sensitivity and specificity of MR imaging for the evaluation of congenital heart disease was assessed in 51 patients (31 male and 20 female, aged 3-69 years), with a total of 115 congenital heart lesions. The true diagnosis was established by angiocardiography, catheterization, or surgery. Sensitivity at a specificity level of 90% was determined by means of receiver operating characteristic curves for great vessel relationships (100%), thoracic aorta anomalies (94%), atrial (91%) and ventricular (100%) septal defects, visceroatrial situs (100%), loop (100%), right ventricular outflow obstructions (95%), aortic valve (52%), mitral valve (62%), and tricuspid valve (76%). Spin-echo MR imaging is a reliable method for the noninvasive evaluation of congenital heart disease but is limited in the assessment of some valvular anomalies
Falk, Bo Torkel; Jönsson, Lennart; Olsen, Lisbeth Høier
Background: The occurrence of small vessel arteriosclerosis in the myocardium, kidney, and lung in dogs with naturally occurring myxomatous mitral valve disease has not been previously investigated systematically. Methods: Twenty-one dogs with naturally occurring congestive heart failure and 21 age......-matched, sex-matched, and weight-matched control dogs underwent extensive pathological and histopathological examination. Morphometry and scoring of tissue sections were used to measure arterial narrowing and fibrosis in the myocardium, kidney, and lung; and intimal thickness and plaque formation in the aorta...... and pulmonary artery. Results: Dogs with congestive heart failure had significantly more arterial narrowing in the left ventricle (Pdogs. However...
Delling, Francesca N; Rong, Jian; Larson, Martin G; Lehman, Birgitta; Fuller, Deborah; Osypiuk, Ewa; Stantchev, Plamen; Hackman, Brianne; Manning, Warren J; Benjamin, Emelia J; Levine, Robert A; Vasan, Ramachandran S
Longitudinal studies of mitral valve prolapse (MVP) progression among unselected individuals in the community, including those with nondiagnostic MVP morphologies (NDMs), are lacking. We measured longitudinal changes in annular diameter, leaflet displacement, thickness, anterior/posterior leaflet projections onto the annulus, coaptation height, and mitral regurgitation jet height in 261 Framingham Offspring participants at examination 5 who had available follow-up imaging 3 to 16 years later. Study participants included MVP (n=63); NDMs, minimal systolic displacement (n=50) and the abnormal anterior coaptation phenotype (n=10, with coaptation height >40% of the annulus similar to posterior MVP); plus 138 healthy referents without MVP or NDMs. At follow-up, individuals with MVP (52% women, 57±11 years) had greater increases of leaflet displacement, thickness, and jet height than referents (all PMVP (17%) had moderate or more severe mitral regurgitation (jet height ≥5 mm) and 5 others (8%) underwent mitral valve repair. Of the individuals with NDM, 8 (80%) participants with abnormal anterior coaptation progressed to posterior MVP; 17 (34%) subjects with minimal systolic displacement were reclassified as either posterior MVP (12) or abnormal anterior coaptation (5). In comparison with the 33 participants with minimal systolic displacement who did not progress, the 17 who progressed had greater leaflet displacement, thickness, coaptation height, and mitral regurgitation jet height (all PMVP, highlighting the clinical significance of mild MVP expression. MVP progresses to significant mitral regurgitation over a period of 3 to 16 years in one-fourth of individuals in the community. Changes in mitral leaflet morphology are associated with both NDM and MVP progression. © 2016 American Heart Association, Inc.
For clinical evaluation and decision-making in patients with valvular heart disease, the diagnostic armamentarium expands steadily. This evolution makes it difficult to choose the most appropriate approach for a specific valvular lesion. It may also reflect our uncertainty of what are the findings that best predict clinical outcome of patients, e. g. after surgery. Accordingly, for each type of valvular lesion, some pathophysiological considerations are stated in order to derive the most important measures that would allow optimal guidance of patients. Based on these considerations the value of an MR study is discussed for each valvular lesion. Newest advances in MR technology allow for highly accurate measurements of regurgitant volumes and hence, MR may be the method of choice for a quantitative evaluation of regurgitant valves. For assessment of stenosis severity, measurement of transvalvular pressure gradient is an appropriate measure and MR may not confer benefits over echocardiography, provided the ultrasound window is adequate (and stroke volume is in the normal range). With respect to surgical treatment, valvular morphology is of pivotal importance, particularly for the mitral valve, and echocardiography still appears to be the first line method. Little data relate lesion severity and/or morphology to clinical outcome. Conversely, the extent of cardiac adaptation to pressure- and/or volume-overload, i. e. ventricular remodeling, is a strong predictor of outcome, and is therefore most important for final judgement of the patient. For assessment of left and right ventricular remodeling, echocardiography typically provides all the necessary information. However, in special cases with discrepant findings, with inadequate ultrasound window, or in the preoperative work-up, MR may provide important information regarding cardiac adaptation to valvular lesion. (orig.) [de
Simon, Hélène A; Ge, Liang; Sotiropoulos, Fotis; Yoganathan, Ajit P
Thromboembolic complications (TECs) of bileaflet mechanical heart valves (BMHVs) are believed to be due to the nonphysiologic mechanical stresses imposed on blood elements by the hinge flows. Relating hinge flow features to design features is, therefore, essential to ultimately design BMHVs with lower TEC rates. This study aims at simulating the pulsatile three-dimensional hinge flows of three BMHVs and estimating the TEC potential associated with each hinge design. Hinge geometries are constructed from micro-computed tomography scans of BMHVs. Simulations are conducted using a Cartesian sharp-interface immersed-boundary methodology combined with a second-order accurate fractional-step method. Leaflet motion and flow boundary conditions are extracted from fluid-structure-interaction simulations of BMHV bulk flow. The numerical results are analyzed using a particle-tracking approach coupled with existing blood damage models. The gap width and, more importantly, the shape of the recess and leaflet are found to impact the flow distribution and TEC potential. Smooth, streamlined surfaces appear to be more favorable than sharp corners or sudden shape transitions. The developed framework will enable pragmatic and cost-efficient preclinical evaluation of BMHV prototypes prior to valve manufacturing. Application to a wide range of hinges with varying design parameters will eventually help in determining the optimal hinge design.
Seiffert, Moritz; Conradi, Lenard; Baldus, Stephan; Knap, Malgorzata; Schirmer, Johannes; Franzen, Olaf; Koschyk, Dietmar; Meinertz, Thomas; Reichenspurner, Hermann; Treede, Hendrik
Transcatheter valve-in-valve implantation is evolving as an alternative to reoperative valve replacement in high-risk patients with degenerated bioprostheses. Nevertheless, hemodynamic performance is limited by the previously implanted xenograft. We report our experience with patient-prosthesis mismatch (PPM) after valve-in-valve implantation in the aortic position. Eleven patients (aged 79.3 ± 6.1 years) received transapical implantation of a balloon-expandable pericardial heart valve into a degenerated bioprosthesis (size, 23.9 ± 1.6 mm; range, 21-27 mm) in the aortic position. All patients were considered high risk for surgical valve replacement (logistic European System for Cardiac Operative Risk Evaluation, 31.8% ± 24.1%). Severe PPM was defined as an indexed effective orifice area less than 0.65 cm(2)/m(2), determined by discharge echocardiography. Severe PPM was evident in 5 patients (group 1) and absent in 6 patients (group 2). Mean transvalvular gradients decreased from 29.2 ± 15.4 mm Hg before implantation to 21.2 ± 9.7 mm Hg at discharge (group 1) and from 28.2 ± 9.0 mm Hg before implantation to 15.2 ± 6.5 mm Hg at discharge (group 2). Indexed effective orifice area increased from 0.5 ± 0.1 cm(2)/m(2) to 0.6 ± 0.1 cm(2)/m(2) and from 0.6 ± 0.3 cm(2)/m(2) to 0.8 ± 0.3 cm(2)/m(2). Aortic regurgitation decreased from grade 2.0 ± 1.1 to 0.4 ± 0.5 overall. No differences in New York Heart Association class improvement or survival during follow-up were observed. One patient required reoperation for symptomatic PPM 426 days after implantation. Valve-in-valve implantation can be performed in high-risk surgical patients to avoid reoperation. However, PPM frequently occurs, making adequate patient selection crucial. Small bioprostheses (body surface area less than 1.8 m(2). Larger prostheses seem to carry a lower risk for PPM. Although no delay in clinical improvement was seen at short-term, 1 PPM-related surgical intervention raises concern regarding
Blaich, A; Fasel, D; Kaech, C; Frei, R
Erysipelothrix rhusiopathiae is the causative agent of swine erysipelas. Systemic infections caused by E. rhusiopathiae are rare, but often (90%) associated with endocarditis. In about 60% of cases endocarditis develops on normal heart valves, and despite appropriate antibiotic therapy about one-third of the patients requires valve replacement. We report the case of a housewife, who developed a mitral valve endocarditis due to E. rhusiopathiae after preparing meat for the Turkish "Festival of Sacrifice".
Young Eun Chon
Full Text Available Liver congestion due to heart failure in patients with valvular heart disease (VHD can result in an overestimate of the liver stiffness (LS as assessed by transient elastography (TE. This prospective pilot study investigated the dynamics of LS during the perioperative valve operation period.Thirty-two patients who underwent a valve operation (case and 12 who underwent a varicose vein operation (control were prospectively enrolled. LS and cardiologic parameters at three time points [baseline, post-operative day (POD7, and POD90] were collected.LS at three time points were consistently higher in the case group than those in the control group, although all participants did not show evidence of underlying chronic liver disease (all P<0.05. In the case group, LS at POD7 increased slightly from the LS at baseline (median 7.9→8.4 kPa, P = 0.816, whereas LS at POD90 decreased significantly from the LS at POD7 (median 8.4→6.0 kPa; P = 0.026. LS was significantly correlated with N-terminal-pro brain natriuretic peptide (NT-proBNP (ρ = 0.412, left ventricular ejection fraction (ρ = -0.494, and central venous pressure during the operation (ρ = 0.555 at baseline (all P<0.05. LS was significantly correlated with NT-proBNP (ρ = 0.526 and right ventricular pressure (ρ = 0.572 at POD7, whereas LS was significantly correlated with NT-proBNP (ρ = 0.590 at POD90 (all P<0.05.LS can be overestimated in patients with VHD due to hepatic congestion. However, LS can be dynamically reversed during the perioperative period reflecting the restoration of cardiac function after a successful operation.
Izumo, Masaki; Akashi, Yoshihiro J
Since the introduction of transcatheter structural heart intervention, the term "structural heart disease" has been widely used in the field of cardiology. Structural heart disease refers to congenital heart disease, valvular heart disease, and cardiomyopathy. In structural heart disease, valvular heart disease is frequently identified in the elderly. Of note, the number of patients who suffer from aortic stenosis (AS) and mitral regurgitation (MR) is increasing in developed countries because of the aging of the populations. Transcatheter aortic valve replacement and percutaneous mitral valve repair has been widely used for AS and MR, individually. Echocardiography is the gold standard modality for initial diagnosis and subsequent evaluation of AS and MR, although the difficulties in assessing patients with these diseases still remain. Here, we review the clinical usefulness and prognostic impact of exercise echocardiography on structural heart disease, particularly on AS and MR.
Pedersen Torben B
Full Text Available Abstract Background Glutaraldehyde-treated bioprosthetic heart valves are commonly used for replacement of diseased heart valves. However, calcification and wear limit their durability, and the development of new and improved bioprosthetic valve designs is needed and must be evaluated in a reliable animal model. We studied glutaraldehyde-treated valves 6 months after implantation to evaluate bioprosthetic valve complications in the mitral position in juvenile pigs. Materials The study material comprised eight, 5-month old, 60-kg pigs. All pigs received a size 27, glutaraldehyde-treated, stented, Carpentier-Edwards S.A.V. mitral valve prosthesis. After six months, echocardiography was performed, and the valves explanted for gross examination, high resolution X-ray, and histological evaluation. Results Five pigs survived the follow-up period. Preexplant echocardiography revealed a median peak and mean velocity of 1.61 m/s (range: 1.17-2.00 and 1.20 (SD = ±0.25, respectively, and a median peak and mean pressure difference of 10.42 mmHg (range: 5.83-16.55 and 6.51 mmHg (SD = ±2.57, respectively. Gross examination showed minor thrombotic depositions at two commissures in two valves and at all three commissures in three valves. High resolution X-ray imaging revealed different degrees of calcification in all explanted valves, primarily in the commissural and belly areas. In all valves, histological evaluation demonstrated various degrees of fibrous sheath formation, limited immunological infiltration, and no overgrowth of host endothelium. Conclusions Bioprosthetic glutaraldehyde-treated mitral valves can be implanted into the mitral position in pigs and function after 6 months. Echocardiographic data, calcification, and histological examinations were comparable to results obtained in sheep models and human demonstrating the suitability of the porcine model.
Kiss, Daniel; Anwaruddin, Saif
With widespread adoption of transcatheter aortic valve replacement, there has been a change in the approach to management of valvular heart disease. New interest has taken hold in transcatheter therapies for valvular heart disease, as well as research into pathophysiology and progression of disease. Additionally, several key trials have further refined our understanding of surgical management of valvular heart disease. This review will elucidate recent clinical trial data leading to changes in practice. There have been several landmark trials expanding the indications for transcatheter aortic valve replacement. Additionally, although still early, trials are beginning to demonstrate the feasibility and safety of transcatheter mitral valves. Options for transcatheter management of right-sided valvular disease continue to evolve, and these are areas of active investigation. The emergence of novel therapies for valvular heart disease has expanded the management options available, allowing physicians to better individualize treatment of patients with valvular heart disease. This review will focus on the recent (within 2 years) trials in this field of interest.
Steger, Christina Maria
Alkaptonuric ochronosis is a heritable disorder of tyrosine metabolism, with various systemic abnormalities related to pigment deposition and degeneration of collagen and other tissues, including the heart and aorta. A 65-year-old woman with alkaptonuric ochronosis and a history of four joint replacements required aortic valve replacement for severe aortic stenosis. Operative findings included ochronosis of a partly calcified aortic valve and the aortic intima. The aortic valve was removed at surgery and histologically investigated. Light microscopic examination of the aortic valve revealed intracellular and extracellular deposits of ochronotic pigment and a chronic inflammatory infiltrate. Beside the case representation, the disease history, aetiology, pathogenesis, clinical presentation and treatment of aortic valve ochronosis are reviewed.
Nägele, Reto; Kaufmann, Beat A
Patients with valvular heart disease or with a prosthetic heart valve replacement are seen with increasing frequency in clinical practice. The medical care and evaluation of patients with valvular heart disease before valve surgery, but also the post-operative treatment is complex and managed by general practitioners, cardiologists and cardiac surgeons. In this mini-review we will first discuss the preoperative assessment of the two most common valvulopathies, aortic stenosis and mitral regurgitation. Then we will discuss the post-operative care, which includes the management of anticoagulation, serial follow up and as well as the diagnostic assessment of complications such as thromboembolism, hemolysis, endocarditis and valve dysfunction.
Full Text Available Valvular dysfunction as the prominent reason of heart failure may causes morbidity and mortality around the world. The inability of human body to regenerate the defected heart valves necessitates the development of the artificial prosthesis to be replaced. Besides, the lack of capacity to grow, repair or remodel of an artificial valves and biological difficulty such as infection or inflammation make the development of tissue engineering heart valve (TEHV concept. This research presented the use of compound of poly-l-lactic acid (PLLA, thermoplastic polyurethane (TPU and maghemite nanoparticle (γ-Fe2O3 as the potential biomaterials to develop three-dimensional (3D aortic heart valve scaffold. Electrospinning was used for fabricating the 3D scaffold. The steepest ascent followed by the response surface methodology was used to optimize the electrospinning parameters involved in terms of elastic modulus. The structural and porosity properties of fabricated scaffold were characterized using FE-SEM and liquid displacement technique, respectively. The 3D scaffold was then seeded with aortic smooth muscle cells (AOSMCs and biological behavior in terms of cell attachment and proliferation during 34 days of incubation was characterized using MTT (3-(4,5-dimethylthiazol-2-yl-2,5-diphenyltetrazolium bromide assay and confocal laser microscopy. Furthermore, the mechanical properties in terms of elastic modulus and stiffness were investigated after cell seeding through macro-indentation test. The analysis indicated the formation of ultrafine quality of nanofibers with diameter distribution of 178 ± 45 nm and 90.72% porosity. In terms of cell proliferation, the results exhibited desirable proliferation (109.32 ± 3.22% compared to the control of cells over the 3D scaffold in 34 days of incubation. The elastic modulus and stiffness index after cell seeding were founded to be 22.78 ± 2.12 MPa and 1490.9 ± 12 Nmm2, respectively. Overall, the fabricated 3D
... Check Recipe Certification Program Nutrition Requirements Heart-Check Professional Resources Contact the Heart-Check Certification Program Simple Cooking and Recipes Dining Out Choosing a Restaurant Deciphering ...
Kremer, Jamila; Farag, Mina; Arif, Rawa; Brcic, Andreas; Sabashnikov, Anton; Schmack, Bastian; Popov, Aron-Frederik; Karck, Matthias; Dohmen, Pascal M; Ruhparwar, Arjang; Weymann, Alexander
BACKGROUND Total artificial heart (TAH) implantation in patients with aggressive tumor infiltration of the heart can be challenging. CASE REPORT We report on a patient with a rare primary undifferentiated high-grade spindle cell sarcoma of the mitral valve and in the left atrium, first diagnosed in 2014. The referring center did a first resection in 2014. In the course of 17 months, computer tomography (CT) scan again showed massive invasion of the mitral valve and left atrium. Partial resection and mitral valve replacement was not an option. We did a subtotal heart excision with total artificial heart implantation. In this report we discuss complications, risk factors, and perioperative management of this patient. CONCLUSIONS Patients with aggressive tumors of the heart can be considered for TAH implantation.
Huenges, Katharina; Panholzer, Bernd; Cremer, Jochen; Haneya, Assad
Left ventricular assist device (LVAD) is nowadays a routine therapy for patients with advanced heart failure. We present the case of a 74-year-old male patient who was admitted to our center with terminal heart failure in dilated cardiomyopathy and ascending aortic aneurysm with aortic valve regurgitation. The LVAD implantation with simultaneous aortic valve and supracoronary ascending aortic replacement was successfully performed.
Vaesken, Antoine; Pelle, Anne; Pavon-Djavid, Graciela; Rancic, Jeanne; Chakfe, Nabil; Heim, Frederic
Transcatheter aortic valve implantation (TAVI) has become a popular alternative technique to surgical valve replacement for critical patients. Biological valve tissue has been used in TAVI procedures for over a decade, with over 150,000 implantations to date. However, with only 6 years of follow up, little is known about the long-term durability of biological tissue. Moreover, the high cost of tissue harvesting and chemical treatment procedures favor the development of alternative synthetic valve leaflet materials. In that context, textile polyester [polyethylene terephthalate (PET)] could be considered as an interesting candidate to replace the biological valve leaflets in TAVI procedures. However, no result is available in the literature about the behavior of textile once in contact with biological tissue in the valve position. The interaction of synthetic textile material with living tissues should be comparable to biological tissue. The purpose of this preliminary work is to compare the in vivo performances of various woven textile PET valves over a 6-month period in order to identify favorable textile construction features. In vivo results indicate that fibrosis as well as calcium deposit can be limited with an appropriate material design.
Galmes Belmonte, F.B.
This work tries to justify the hydrodynamic butterfly valves performance, using the EPRI tests, results carried out in laboratory and in situ. This justification will be possible if: - The valves to study are similar - Their performance is calculated using EPRI's methodology Looking for this objective, the elements of the present work are: 1. Brief EPRI butterfly valve description it wild provide the factors which are necessary to define the butterfly valves similarity. 2. EPRI tests description and range of validation against test data definition. 3. Description of the spanish butterfly analyzed valves, and comparison with the EPRI performance results, to prove that this valves are similar to the EPRI test valves. In this way, it will not be necessary to carry out particular dynamic tests on the spanish valves to describe their hydrodynamic performance. (Author)
Gregersen, Niels; Oliver, James; Hjorth, Poul G.
This problem deals with modelling the flow through a typical Danfoss thermostatic radiator valve.Danfoss is able to employ Computational Fluid Dynamics (CFD) in calculations of the capacity of valves, but an experienced engineer can often by rules of thumb "guess" the capacity, with a precision...
Anjos, Daniela Brianne Martins Dos; Rodrigues, Roberta Cunha Matheus; Padilha, Kátia Melissa; Pedrosa, Rafaela Batista Dos Santos; Gallani, Maria Cecília Bueno Jayme
evaluate the practicality, acceptability and the floor and ceiling effects, estimate the reliability and verify the convergent construct's validity with the instrument called the Heart Valve Disease Impact on daily life (IDCV) of the valve disease in patients with mitral and or aortic heart valve disease. data was obtained from 86 heart valve disease patients through 3 phases: a face to face interview for a socio-demographic and clinic characterization and then other two done through phone calls of the interviewed patients for application of the instrument (test and repeat test). as for the practicality and acceptability, the instrument was applied with an average time of 9,9 minutes and with 110% of responses, respectively. Ceiling and floor effects observed for all domains, especially floor effect. Reliability was tested using the test - repeating pattern to give evidence of temporal stability of the measurement. Significant negative correlations with moderate to strong magnitude were found between the score of the generic question about the impact of the disease and the scores of IDCV, which points to the validity of the instrument convergent construct. the instrument to measure the impact of valve heart disease on the patient's daily life showed evidence of reliability and validity when applied to patients with heart valve disease. avaliar a praticabilidade, aceitabilidade e os efeitos teto e chão, estimar a confiabilidade e verificar a validade de construto convergente do Instrumento para Mensuração do Impacto da Doença no Cotidiano do Valvopata em pacientes com valvopatia mitral e/ou aórtica. dados foram obtidos junto a 86 pacientes valvopatas por meio de três etapas: uma entrevista presencial para caracterização sociodemográfica e clínica e as duas outras realizadas por meio de contato telefônico para as aplicações do instrumento (Teste e teste de repetição). quanto à praticabilidade e aceitabilidade, o instrumento foi aplicado com tempo m
Whelton, Seamus P; Mauer, Andreas C; Pencina, Karol M; Massaro, Joseph M; D'Agostino, Ralph B; Fox, Caroline S; Hoffmann, Udo; Michos, Erin D; Peloso, Gina M; Dufresne, Line; Engert, James C; Kathiresan, Sekar; Budoff, Matthew; Post, Wendy S; Thanassoulis, George; O'Donnell, Christopher J
It is unknown if lifelong exposure to increased hemodynamic stress from an elevated resting heart rate (HR) may contribute to aortic valve calcium (AVC). We performed multivariate regression analyses using data from 1,266 Framingham Heart Study (FHS) Offspring cohort participants and 6,764 Multi-Ethnic Study of Atherosclerosis (MESA) participants. We constructed a genetic risk score (GRS) for HR using summary-level data in the Cohorts for Heart and Aging Research in Genomic Epidemiology (CHARGE) AVC Consortium to investigate if there was evidence in favor of a causal relation. AVC was present in 39% of FHS Offspring cohort participants and in 13% of MESA cohort participants. In multivariate adjusted models, participants in the highest resting HR quartiles had significantly greater prevalence of AVC, with a prevalence ratio of 1.19 (95% confidence interval [CI] 0.99 to 1.44) for the FHS Offspring cohort and 1.32 (95% CI 1.12 to 1.63) for the MESA cohort, compared with those in the lowest quartile. There was a similar increase in the prevalence of AVC per standard deviation increase in resting HR in both FHS Offspring (prevalence ratio 1.08, 95% CI 1.01 to 1.15) and MESA (1.10, 95% CI 1.03 to 1.17). In contrast with these observational findings, a HR associated GRS was not significantly associated with AVC. Although our observational analysis indicates that a higher resting HR is associated with AVC, our genetic results do not support a causal relation. Unmeasured environmental and/or lifestyle factors associated with both increased resting HR and AVC that are not fully explained by covariates in our observational models may account for the association between resting HR and AVC. Copyright © 2018. Published by Elsevier Inc.
Background: Rheumatic heart disease (RHD) is the commonest acquired heart disease in children worldwide but in Uganda, data is scarce regarding its morbidity and mortality. The disease has a progressive course and patients usually require valve repair/replacement in the future. Objectives:To describe the frequency of ...
Xenopoulos, N P; Braden, G A; Applegate, R J
This brief report presents a patient with isolated right heart failure and two rare underlying causes, hyperthyroidism and dysplastic tricuspid valve. Repair of the tricuspid valve and treatment of the hyperthyroidism were both essential for successful treatment of the right heart failure. Most important, recrudescence of hyperthyroidism in this patient was associated with reappearance of florid right heart failure. This report provides further information about a potential linkage of hyperthyroidism and severe right heart failure.
Wang, Zhi; Chen, Qiang; Guo, Hao; Li, Zhishan; Zhang, Jinfeng; Lv, Lei; Guo, Yongqing
This study investigated the effects of dexmedetomidine on heart-type fatty acid binding protein (H-FABP), creatine kinase isoenzymes (CK-MB), and troponin I (cTnI) levels, neurological function and near-term prognosis in patients undergoing heart valve replacement. Patients undergoing heart valve replacement were randomly allocated to remifentanil anesthesia (control group, n=48) or dexmedetomidine anesthesia (observation group, n=48). Hemodynamic parameters were measured before anesthesia induction (T1), 1 min after intubation (T2), 10 min after start of surgery (T3), and on completion of surgery (T4). Levels of plasma H-FABP, CK-MB and cTnI were measured 10 min before anesthesia induction (C1), 10 min after start of surgery (C2), on completion of surgery (C3), 6 h after surgery (C4), and 24 h after surgery (C5). S100β protein and serum neuron-specific enolase (NSE) were detected 10 min before anesthesia induction (C1), and 24 h after surgery (C5). Neurological and cardiac function was evaluated 24 h after surgery. Incidence of cardiovascular adverse events was recorded for 1 year of follow-up. There were no significant differences in the average heart rate between the two groups during the perioperative period. The mean arterial pressure in the observation group was significantly lower than control group (PH-FABP, CK-MB and cTnI at C2, C3, C4 and C5, were significantly higher than C1, but significantly lower in the observation versus control group (P<0.05). Twenty-four hours after surgery, levels of S100β and NSE in both groups were higher than those before induction (P<0.05), but significantly lower in the observation versus control group (P<0.05). Twenty-four hours after surgery, neurological function scores were better, and myocardial contractility and arrhythmia scores significantly lower in the observation versus control group (P<0.05 for all). After follow-up for 1 year, incidence of cardiovascular adverse events was significantly lower in the observation
Cruz Gispert, Albert
The Ocean Grazer, a novel wave energy converter, has been proposed by the University of Groningen. The system can collect and store multiple forms of ocean energy, with a pistontype hydraulic pump as its core technology. In this work, the dynamical behavior of a piston valve for use in the piston pump system is studied. In order to gain insight into the dynamical behavior of the piston-type hydraulic pump, a simulation model is developed to describe the movement of the piston v...
Hassan, Saamir A; Banchs, Jose; Iliescu, Cezar; Dasari, Arvind; Lopez-Mattei, Juan; Yusuf, Syed Wamique
Rare neuroendocrine tumours (NETs) that most commonly arise in the gastrointestinal tract can lead to carcinoid syndrome and carcinoid heart disease. Patients with carcinoid syndrome present with vasomotor changes, hypermotility of the gastrointestinal system, hypotension and bronchospasm. Medical therapy for carcinoid syndrome, typically with somatostatin analogues, can help control symptoms, inhibit tumour progression and prolong survival. Carcinoid heart disease occurs in more than 50% of these patients and is the initial presentation of carcinoid syndrome in up to 20% of patients. Carcinoid heart disease has characteristic findings of plaque-like deposits composed of smooth muscle cells, myofibroblasts, extracellular matrix and an overlying endothelial layer which can lead to valve dysfunction. Valvular dysfunction can lead to oedema, ascites and right-sided heart failure. Medical therapy of carcinoid heart disease is limited to symptom control and palliation. Valve surgery for carcinoid heart disease should be considered for symptomatic patients with controlled metastatic carcinoid syndrome. A multidisciplinary approach is needed to guide optimal management. © Article author(s) (or their employer(s) unless otherwise stated in the text of the article) 2017. All rights reserved. No commercial use is permitted unless otherwise expressly granted.
Engborg, Jonathan; Riechel-Sarup, Casper; Gerke, Oke
OBJECTIVES: Transcatheter aortic valve implantation (TAVI) is an established treatment for high-grade aortic valve stenosis in patients found unfit for open heart surgery. The method may cause cardiac conduction disorders requiring permanent pacemaker (PPM) implantation, and the long-term effect...
Clinical implications of hypothermic ventricular fibrillation versus beating-heart technique during cardiopulmonary bypass for pulmonary valve replacement in patients with repaired tetralogy of Fallot.
Lee, Ji-Hyun; Lee, Ji-Eun; Shin, Jungho; Song, In-Kyung; Kim, Hee-Soo; Kim, Chong-Sung; Kim, Woong-Han; Kim, Jin-Tae
This study aimed to compare the effects of hypothermic ventricular fibrillation and beating-heart techniques during cardiopulmonary bypass (CPB) on postoperative outcomes after simple pulmonary valve replacement in patients with repaired tetralogy of Fallot (TOF). We retrospectively reviewed the data of 47 patients with repaired tetralogy of Fallot at a single institution, who received pulmonary valve replacement under the ventricular fibrillation or beating-heart technique without cardioplegic cardiac arrest during CPB between January 2005 and April 2015. The patients were divided into fibrillation (n = 32) and beating-heart (n = 15) groups. On comparing these groups, the fibrillation group had a larger sinotubular junction (27.1 ± 4.6 vs 22.1 ± 2.4 mm), had a longer operation duration (396 ± 108 vs 345 ± 57 min), required more postoperative transfusions (2.1 ± 2.6 vs 5.0 ± 6.3 units) and had a higher vasoactive-inotropic score at intensive care unit admission (8.0 vs 10, all P tetralogy of Fallot. © The Author 2017. Published by Oxford University Press on behalf of the European Association for Cardio-Thoracic Surgery. All rights reserved.
Wagner, Martin G; Hatt, Charles R; Dunkerley, David A P; Bodart, Lindsay E; Raval, Amish N; Speidel, Michael A
Transcatheter aortic valve replacement (TAVR) is a minimally invasive procedure in which a prosthetic heart valve is placed and expanded within a defective aortic valve. The device placement is commonly performed using two-dimensional (2D) fluoroscopic imaging. Within this work, we propose a novel technique to track the motion and deformation of the prosthetic valve in three dimensions based on biplane fluoroscopic image sequences. The tracking approach uses a parameterized point cloud model of the valve stent which can undergo rigid three-dimensional (3D) transformation and different modes of expansion. Rigid elements of the model are individually rotated and translated in three dimensions to approximate the motions of the stent. Tracking is performed using an iterative 2D-3D registration procedure which estimates the model parameters by minimizing the mean-squared image values at the positions of the forward-projected model points. Additionally, an initialization technique is proposed, which locates clusters of salient features to determine the initial position and orientation of the model. The proposed algorithms were evaluated based on simulations using a digital 4D CT phantom as well as experimentally acquired images of a prosthetic valve inside a chest phantom with anatomical background features. The target registration error was 0.12 ± 0.04 mm in the simulations and 0.64 ± 0.09 mm in the experimental data. The proposed algorithm could be used to generate 3D visualization of the prosthetic valve from two projections. In combination with soft-tissue sensitive-imaging techniques like transesophageal echocardiography, this technique could enable 3D image guidance during TAVR procedures. © 2018 American Association of Physicists in Medicine.
Alavi, S Hamed; Sinha, Aditi; Steward, Earl; Milliken, Jeffrey C; Kheradvar, Arash
The extracellular matrix of the atrioventricular (AV) valves' leaflets has a key role in the ability of these valves to properly remodel in response to constantly varying physiological loads. While the loading on mitral and tricuspid valves is significantly different, no information is available on how collagen fibers change their orientation in response to these loads. This study delineates the effect of physiological loading on AV valves' leaflets microstructures using Second Harmonic Generation (SHG) microscopy. Fresh natural porcine tricuspid and mitral valves' leaflets (n = 12/valve type) were cut and prepared for the experiments. Histology and immunohistochemistry were performed to compare the microstructural differences between the valves. The specimens were imaged live during the relaxed, loading, and unloading phases using SHG microscopy. The images were analyzed with Fourier decomposition to mathematically seek changes in collagen fiber orientation. Despite the similarities in both AV valves as seen in the histology and immunohistochemistry data, the microstructural arrangement, especially the collagen fiber distribution and orientation in the stress-free condition, were found to be different. Uniaxial loading was dependent on the arrangement of the fibers in their relaxed mode, which led the fibers to reorient in-line with the load throughout the depth of the mitral leaflet but only to reorient in-line with the load in deeper layers of the tricuspid leaflet. Biaxial loading arranged the fibers in between the two principal axes of the stresses independently from their relaxed states. Unlike previous findings, this study concludes that the AV valves' three-dimensional extracellular fiber arrangement is significantly different in their stress-free and uniaxially loaded states; however, fiber rearrangement in response to the biaxial loading remains similar. Copyright © 2015 the American Physiological Society.
Mvondo, Charles Mve; Pugliese, Marta; Giamberti, Alessandro; Chelo, David; Kuate, Liliane Mfeukeu; Boombhi, Jerome; Dailor, Ellen Marie
Rheumatic valve disease, a consequence of acute rheumatic fever, remains endemic in developing countries in the sub-Saharan region where it is the leading cause of heart failure and cardiovascular death, involving predominantly a young population. The involvement of the mitral valve is pathognomonic and mitral surgery has become the lone therapeutic option for the majority of these patients. However, controversies exist on the choice between valve repair or prosthetic valve replacement. Although the advantages of mitral valve repair over prosthetic valve replacement in degenerative mitral disease are well established, this has not been the case for rheumatic lesions, where the use of prosthetic valves, specifically mechanical devices, even in poorly compliant populations remains very common. These patients deserve more accurate evaluation in the choice of the surgical strategy which strongly impacts the post-operative outcomes. This report discusses the factors supporting mitral repair surgery in rheumatic disease, according to the patients' characteristics and the effectiveness of the current repair techniques compared to prosthetic valve replacement in developing countries.
Paul, Abhijit; Das, Sucharita
Valvular heart disease presents as mixed spectrum lesion in healthcare settings in the third-world and developing countries. Rheumatic heart disease still forms the bulk of the aetiopathology of valve lesions. Mitral and aortic valve lesions top the list of valvular pathology. A thorough understanding of the pathophysiology of valvular heart disease is essential while planning anaesthesia and perioperative care for such patients. Meticulous use of optimal fluids, close monitoring of the changing haemodynamics and avoidance of situations that can cause major reduction of cardiac output and fluid shifts are mandatory to achieve good clinical outcome. We searched MEDLINE using combinations of the following: anaesthesia, aortic, mitral, regurgitation, stenosis and valvular heart disease. We also hand searched textbooks and articles on valvular heart disease and anaesthesia. This article mainly focuses on the understanding the pathophysiology of valvular heart disease in patients presenting for non-cardiac surgeries in secondary and tertiary care setting.
Tagliasacchi, Isabella; Martinelli, Luigi; Bardaro, Leopoldo; Chierchia, Sergio
Osteogenesis imperfecta is a disorder of the connective tissue that affects several structures including heart valves. However, cardiac surgery is associated with high mortality and morbidity rates. In a 48-year-old man with osteogenesis imperfecta and mitral valve prolapse, we performed the first successful mitral valve repair by right anterior mini-thoracotomy. At the 1-year follow-up, he was asymptomatic and echocardiography confirmed the initial success. © The Author 2017. Published by Oxford University Press on behalf of the European Association for Cardio-Thoracic Surgery. All rights reserved.
... Ebstein's Anomaly l-Transposition of the Great Arteries Patent Ductus Arteriosus (PDA) Pulmonary Valve Stenosis and Regurgitation ... Sodium and Salt 3 Heart Attack Symptoms in Women 4 Warning Signs of a Heart Attack 5 ...
Bartoletti, Stefano; Santangeli, Pasquale; DI Biase, Luigi; Natale, Andrea
Patients with mechanical "hardware" in the heart, such as those with mechanical cardiac valves or atrial septal closure devices, represent a population at high risk of developing AF. Catheter ablation of AF in these subjects might represent a challenge, due to the perceived higher risk of complications associated with the presence of intracardiac mechanical devices. Accordingly, such patients were excluded or poorly represented in major trials proving the benefit of catheter ablation for the rhythm-control of AF. However, recent evidence supports the concept that catheter ablation procedures might be equally effective in these patients, without a significant increase in the risk of procedural complications. This review will summarize the current state-of-the-art on catheter ablation of AF in patients with mechanical "hardware" in the heart.
Di Leo, Giovanni; Spadavecchia, Chiara; Zanardo, Moreno; Secchi, Francesco; Veronese, Ivan; Cantone, Marie Claire; Sardanelli, Francesco
To estimate the impact of endoaortic stents/mechanical heart valves on the output of an automatic exposure control (AEC) system and CT radiation dose. In this phantom study, seven stents and two valves were scanned with varying tube voltage (80/100/120 kVp), AEC activation (enabled/disabled) and prosthesis (present/absent), for a total of 540 scans. For each prosthesis, the dose-length product (DLP) was compared between scans with the AEC enabled and disabled. Percentage confidence levels for differences due to the prosthesis were calculated. Differences between results with the AEC enabled and disabled were not statistically significant (p ≥ 0.059). In the comparison with and without the prosthesis, DLP was unchanged at 80 kVp and 100 kVp, while a slight increase was observed at 120 kVp. The radiation dose varied from 1.8 mGy to 2.4 mGy without the prosthesis and from 1.8 mGy to 2.5 mGy with the prosthesis (confidence level 37-100%). The effect of the prosthesis on the AEC system was negligible and not clinically relevant. Therefore, disabling the AEC system when scanning these patients is not likely to provide a benefit. • CT-AEC system is not impaired in patients with endoaortic prostheses/heart valves. • Negligible differences may be observed only at 120 kVp. • Disabling the AEC system in these patients is not recommended.
Full Text Available Prosthetic Valve Obstruction (PVO is a serious complication which is associated with increased morbidity and mortality. This could result from thrombus formation, development of pannus, or a combination of both. Patients with this complication often present with symptoms and signs of heart failure, systemic embolism, acute cardiovascular collapse, and sudden death. Transesophageal echocardiography and cine fluoroscopy play a vital role in diagnosis of this potentially lethal condition. Herein, we reported a 56-year-old male patient who presented with severe heart failure and was found to have obstructed ATS27 bileaflet mitral prosthetic valve. Thrombolysis and redo surgery are two important options for treating this condition although guidelines for choosing between the two are not very definite.
Kemp, I.; Dellimore, K.; Rodriguez, R.; Scheffer, C.; Blaine, D.; Weich, H.; Doubell, A.
Experiments performed on a 19 mm diameter bioprosthetic valve were used to successfully validate the fluid–structure interaction (FSI) simulation of an aortic valve at 72 bpm. The FSI simulation was initialized via a novel approach utilizing a Doppler sonogram of the experimentally tested valve. Using this approach very close quantitative agreement (≤12.5 %) between the numerical predictions and experimental values for several key valve performance parameters, including the peak systolic transvalvular pressure gradient, rapid valve opening time and rapid valve closing time, was obtained. The predicted valve leaflet kinematics during opening and closing were also in good agreement with the experimental measurements.
Gonçalves, Luís F; Lee, Wesley; Chaiworapongsa, Tinnakorn; Espinoza, Jimmy; Schoen, Mary Lou; Falkensammer, Peter; Treadwell, Marjorie; Romero, Roberto
This study was undertaken to describe a new technique for the examination of the fetal heart using four-dimensional ultrasonography with spatiotemporal image correlation (STIC). Volume data sets of the fetal heart were acquired with a new cardiac gating technique (STIC), which uses automated transverse and longitudinal sweeps of the anterior chest wall. These volumes were obtained from 69 fetuses: 35 normal, 16 with congenital anomalies not affecting the cardiovascular system, and 18 with cardiac abnormalities. Dynamic multiplanar slicing and surface rendering of cardiac structures were performed. To illustrate the STIC technique, two representative volumes from a normal fetus were compared with volumes obtained from fetuses with the following congenital heart anomalies: atrioventricular septal defect, tricuspid stenosis, tricuspid atresia, and interrupted inferior vena cava with abnormal venous drainage. Volume datasets obtained with a transverse sweep were utilized to demonstrate the cardiac chambers, moderator band, interatrial and interventricular septae, atrioventricular valves, pulmonary veins, and outflow tracts. With the use of a reference dot to navigate the four-chamber view, intracardiac structures could be simultaneously studied in three orthogonal planes. The same volume dataset was used for surface rendering of the atrioventricular valves. The aortic and ductal arches were best visualized when the original plane of acquisition was sagittal. Volumes could be interactively manipulated to simultaneously visualize both outflow tracts, in addition to the aortic and ductal arches. Novel views of specific structures were generated. For example, the location and extent of a ventricular septal defect was imaged in a sagittal view of the interventricular septum. Furthermore, surface-rendered images of the atrioventricular valves were employed to distinguish between normal and pathologic conditions. Representative video clips were posted on the Journal's Web
Douglas, Pamela S; Leon, Martin B; Mack, Michael J; Svensson, Lars G; Webb, John G; Hahn, Rebecca T; Pibarot, Philippe; Weissman, Neil J; Miller, D Craig; Kapadia, Samir; Herrmann, Howard C; Kodali, Susheel K; Makkar, Raj R; Thourani, Vinod H; Lerakis, Stamatios; Lowry, Ashley M; Rajeswaran, Jeevanantham; Finn, Matthew T; Alu, Maria C; Smith, Craig R; Blackstone, Eugene H
Use of transcatheter aortic valve replacement (TAVR) for severe aortic stenosis is growing rapidly. However, to our knowledge, the durability of these prostheses is incompletely defined. To determine the midterm hemodynamic performance of balloon-expandable transcatheter heart valves. In this study, we analyzed core laboratory-generated data from echocardiograms of all patients enrolled in the Placement of Aortic Transcatheter Valves (PARTNER) 1 Trial with successful TAVR or surgical AVR (SAVR) obtained preimplantation and at 7 days, 1 and 6 months, and 1, 2, 3, 4, and 5 years postimplantation. Patients from continued access observational studies were included for comparison. Successful implantation after randomization to TAVR vs SAVR (PARTNER 1A; TAVR, n = 321; SAVR, n = 313), TAVR vs medical treatment (PARTNER 1B; TAVR, n = 165), and continued access (TAVR, n = 1996). Five-year echocardiogram data were available for 424 patients after TAVR and 49 after SAVR. Death or reintervention for aortic valve structural indications, measured using aortic valve mean gradient, effective orifice area, Doppler velocity index, and evidence of hemodynamic deterioration by reintervention, adverse hemodynamics, or transvalvular regurgitation. Of 2795 included patients, the mean (SD) age was 84.5 (7.1) years, and 1313 (47.0%) were female. Population hemodynamic trends derived from nonlinear mixed-effects models showed small early favorable changes in the first few months post-TAVR, with a decrease of -2.9 mm Hg in aortic valve mean gradient, an increase of 0.028 in Doppler velocity index, and an increase of 0.09 cm2 in effective orifice area. There was relative stability at a median follow-up of 3.1 (maximum, 5) years. Moderate/severe transvalvular regurgitation was noted in 89 patients (3.7%) after TAVR and increased over time. Patients with SAVR showed no significant changes. In TAVR, death/reintervention was associated with lower ejection fraction, stroke volume
Full Text Available We report a newborn with an unusual combination of aneurysmally dilated thin-walled right ventricle with hypertrophy of the apical muscles of the right ventricle. There was narrow pulmonary annulus, pulmonary regurgitation, and hypoplasia of the mitral valve and left ventricle. We propose that this heart represents a partial form of Uhl`s anomaly.
Utz, J.A.; Herfkens, R.J.; Heinsimer, J.; Glover, G.H.; Pelc, N.J.; Shimakawa, A.
Dynamic MR images were obtained utilizing a repetition time of 21 msec, a 30-degree flip angle, and 12-msec gradient refocused echoes. Images were ECG-triggered and constructed at 16 images per cardiac cycle. To assess the sensitivity of dynamic MR imaging of valvular regurgitation, 17 patients were so examined and results were compared with results of echocardiography or cardiac catheterization. Mitral and tricuspid valves were well visualized in all patients. Regurgitation was demonstrated in 14 cases as an area of decreased signal extending from the valve into the regurgitant chamber, corresponding in time to ventricular systole for mitral and tricuspid regurgitation and diastole for aortic insufficiency. There were no false-negative MR imaging studies
Kim, Darae; Hun, Sin Sang; Cho, In-Jeong; Shim, Chi-Young; Ha, Jong-Won; Chung, Namsik; Ju, Hyun Chul; Sohn, Jang Won
Leaflet escape of prosthetic valve is rare but potentially life threatening. It is essential to make timely diagnosis in order to avoid mortality. Transesophageal echocardiography and cinefluoroscopy is usually diagnostic and the location of the missing leaflet can be identified by computed tomography (CT). Emergent surgical correction is mandatory. We report a case of fractured escape of Edward-Duromedics mitral valve 27 years after the surgery. The patient presented with symptoms of acute decompensated heart failure and cardiogenic shock. She was instantly intubated and mechanically ventilated. After prompt evaluation including transthoracic echocardiography and CT, the escape of the leaflet was confirmed. The patient underwent emergent surgery for replacement of the damaged prosthetic valves immediately. Eleven days after the surgery, the dislodged leaflet in iliac artery was removed safely and the patient recovered well. PMID:23837121
Obletter, N.; Picker, D.; Stirner, H.; Schmitt, R.; Helmberger, T.
The article surveys the most important indications for MRT of the heart, the great vessels and the mediastinum. MRT is clearly indicated with search or clarification of aneurysms of the heart wall, atrial or ventriucular septal defects with shunting or other congenital vitia in children or in the newborn, aortic aneurysms as well as connatal defects. MRT is superior to other screening methods when clarifying mediastinal neoplasms because of its multiplanar presentation and its strong quality in contrasting soft tissue. With MRT impressive dynamic studies can be obtained through cine modes and gradient echo sequences, which are very successfully used for aortic and mitral valve stenoses, for aneurysms of the heart wall, for aortic aneurysms and atrial and septal shunts. (orig./MG) [de
Hansen, Tina B; Berg, Selina K; Sibilitz, Kirstine L
variation. The overall uptake rate was 52%. Simultaneous CABG was associated with a higher probability of referral to CR (OR 2.02 (95%CI 1.12-3.65)); being unmarried (0.44 (0.27-0.72)) and having TAVI with a lower probability (0.26; 0.13-0.52). The referral pattern varied across administrative regions...... were linked to nationwide registries. We analysed the provision and content of programmes using descriptive statistics, and factors associated with referral and participation using logistic regression analysis. RESULTS: Coverage of CR programmes for these patients was high, with national programme......- and patient-level influenced uptake and included significant regional variation in referral pattern. Further research into the effect and organization of CR post heart valve surgery is needed....
Full Text Available Abstract As a dreadful complication after the mechanical heart valve replacement, prosthetic valve obstruction caused by pannus formation occurs increasingly with time. The authors here present a case of 42-year-old woman who was urgently admitted to hospital with acute heart failure symptoms due to the mechanical mitral valve failure only 3 months after surgery. Transthoracic and transesophageal echocardiography demonstrated that the bileaflet of the mitral prosthesis were completely immobilized with only a small transvalvular jet remained. During the reoperation, the reason of the prosthetic valve obstruction was attributed to the noncircular pannus ingrowth extending from the atrioventricular side. For a better understanding of the prosthetic valve dysfunction caused by pannus formation, the authors then compile a literature review to briefly discuss the status quo of the clinical characteristics of this uncommon complication.
De Backer, Tine; Petrovic, Mirko; Audenaert, Kurt; Coeman, Mathieu; De Bacquer, Dirk
The objective was to investigate the risk of valvular heart disease in humans in relation to the use of selective serotonin reuptake inhibitors (SSRIs) and serotonin and noradrenaline reuptake inhibitors. A case-control study. We conducted a case-control study within this cohort in which patients with newly diagnosed cardiac valve regurgitation were age-matched to controls. Patient demographics, their cardiovascular risk factors and medication use were extracted in both series. Use of SSRIs, serotonin and noradreline reuptake inhibitors (SNRIs) and other pro-serotonergic agents, their dose and treatment duration were recorded. Logistic regression analysis was used to establish the strength of the association between SSRI/SNRI use and valvular heart disease. Outpatient clinic of the cardiology department at the Ghent University Hospital, East-Flanders in Belgium. Total of 2911 persons 21-58 years of age who had undergone an echocardiogram in the period 2006-2010 and had no known cardiovascular disease or previous cardiac intervention. Two hundred and six echocardiographically proven cases of valvular regurgitation and 195 matched controls. Odd ratio of valvular disease associated with intake of serotonergic drugs. Of the 206 patients with newly diagnosed cardiac valve regurgitation, 11.6% were exposed to serotonergic agents compared to 4.1% in the 195 control patients, leading to an odds ratio of 3.08 (95% confidence interval [CI] 1.35-7.04). The analysis of doses and treatment durations revealed a dose-relationship pattern between SSRI/SNRI use and prevalent valvular heart disease. In this study, use of serotonergic antidepressants was associated with an increased rate of valvular regurgitation in humans.
Full Text Available It is known that a dynamic left ventricular outflow tract (LVOT obstruction exists in patients, following aortic valve replacement (AVR and is usually considered to be benign. We present a patient with dynamic LVOT obstruction following AVR, who developed refractory cardiogenic shock and expired inspite of various treatment strategies. This phenomenon must be diagnosed early and should be considered as a serious and potentially fatal complication following AVR. The possible mechanisms and treatment options are reviewed.
Full Text Available Hanna A Jensen, Lillian L Tsai, Vinod H Thourani Division of Cardiothoracic Surgery, Joseph B Whitehead Department of Surgery, Structural Heart and Valve Center, Emory University School of Medicine, Atlanta, GA, USA Abstract: Severe aortic stenosis (AS is associated with considerable morbidity and mortality and is increasing in prevalence as the global population increases. Since AS primarily affects the elderly, many of these patients have comorbidities that make them poor candidates for the gold standard treatment for AS, surgical aortic valve replacement. Transcatheter aortic valve replacement has emerged as a novel technology for the management of AS in higher risk patients over the past decade. Randomized trials have established the safety and efficacy of transcatheter aortic valve replacement, and the medical community has rallied to identify the patients who are most suitable for this transformative treatment. This review focuses on outlining the key procedural differences, describing the unique challenges of both operations, and finally assessing and comparing outcomes both on a general level and in challenging patient subgroups. Keywords: aortic valve replacement, transcatheter aortic valve replacement, surgical aortic valve replacement
Lembcke, Alexander [Department of Radiology, Charite-Universitaetsmedizin Berlin, Freie Universitaet Berlin and Humboldt-Universitaet zu Berlin, Berlin (Germany)]. E-mail: firstname.lastname@example.org; Hein, Patrick A. [Department of Radiology, Charite-Universitaetsmedizin Berlin, Freie Universitaet Berlin and Humboldt-Universitaet zu Berlin, Berlin (Germany); Enzweiler, Christian N.H. [Department of Radiology, Charite-Universitaetsmedizin Berlin, Freie Universitaet Berlin and Humboldt-Universitaet zu Berlin, Berlin (Germany); Hoffmann, Udo [Department of Radiology, Massachusetts General Hospital and Harvard Medical School, Boston, MA (United States); Klessen, Christian [Department of Radiology, Charite-Universitaetsmedizin Berlin, Freie Universitaet Berlin and Humboldt-Universitaet zu Berlin, Berlin (Germany); Dohmen, Pascal M. [Department of Cardiovascular Surgery, Charite-Universitaetsmedizin Berlin, Freie Universitaet Berlin and Humboldt-Universitaet zu Berlin, Berlin (Germany)
We describe the case of a 72-year-old man presenting with endocarditis and clinical signs of acute myocardial ischemia after biological aortic valve replacement. A comprehensive cardiac dynamic multislice spiral computed tomography demonstrated: (1) an endocarditic vegetation of the aortic valve; (2) a subvalvular leakage feeding a paravalvular pseudoaneurysm based on an aortic root abscess with subsequent compromise of the systolic blood flow in the left main coronary artery and the resulting myocardial perfusion deficit.
Lembcke, Alexander; Hein, Patrick A.; Enzweiler, Christian N.H.; Hoffmann, Udo; Klessen, Christian; Dohmen, Pascal M.
We describe the case of a 72-year-old man presenting with endocarditis and clinical signs of acute myocardial ischemia after biological aortic valve replacement. A comprehensive cardiac dynamic multislice spiral computed tomography demonstrated: (1) an endocarditic vegetation of the aortic valve; (2) a subvalvular leakage feeding a paravalvular pseudoaneurysm based on an aortic root abscess with subsequent compromise of the systolic blood flow in the left main coronary artery and the resulting myocardial perfusion deficit
Full Text Available Infective endocarditis (IE of the left heart is the most frequent type of IE in chronic hemodialysis (CHD (in 90% of cases whereas involvement of the right heart is rare. The aim of this study was to determine the clinical, biological, and echocardiographic characteristics, as well as the prognosis of IE in CHD. This is a retrospective study conducted at the Center of Nephrology and Hemodialysis in Oujda, Morocco. Over a period of 56 months, we compiled data on a series of 11 CHD patients with IE. Their mean age was 40.5 ± 14 years, 72% were male and 27.3% had diabetes. All patients had native valve. All patients had bacteremia preceding the episode of IE. The tricuspid valve was the site of IE in 45% of the cases. Cardiac complications were observed in 72% of the patients and mortality was observed in 72% of cases. The period from IE diagnosis to death was 9 ± 6 days. In our study, the tricuspid valve was the most affected valve of IE in CHD.
Okhotnikov, Ivan; Noroozi, Siamak; Sewell, Philip; Godfrey, Philip
Highlights: • A novel design of a rotary flow control valve driven by a stepper motor is proposed. • The intended use of the valve in the high flow rate independent metering hydraulic system is suggested. • Pressure drops, steady flow torques of the valve for various flow rates and orifice openings are studied by means of computational fluid dynamics. • The discharge coefficient and flow jet angles dependencies on the orifice opening are obtained. • A design method to decrease the flow forces without reducing the flow rate in single-staged valves is demonstrated. - Abstract: In this paper, a novel design of a rotary hydraulic flow control valve has been presented for high flow rate fluid power systems. High flow rates in these systems account for substantial flow forces acting on the throttling elements of the valves and cause the application of mechanically sophisticated multi-staged servo valves for flow regulation. The suggested design enables utilisation of single-stage valves in power hydraulics operating at high flow rates regimes. A spool driver and auxiliary mechanisms of the proposed valve design were discussed and selection criteria were suggested. Analytical expressions for metering characteristics as well as steady flow torques have been derived. Computational fluid dynamics (CFD) analysis of steady state flow regimes was conducted to evaluate the hydraulic behaviour of the proposed valve. This study represents a special case of an independent metering concept applied to the design of power hydraulic systems with direct proportional valve control operating at flow rates above 150 litres per minute. The result gained using parametric CFD simulations predicted the induced torque and the pressure drops due to a steady flow. Magnitudes of these values prove that by minimising the number of spool's mobile metering surfaces it is possible to reduce the flow-generated forces in the new generation of hydraulic valves proposed in this study
Infectious endocarditis (IE), a complication that is both cardiac and infectious, occurs frequently and is associated with a heavy burden of morbidity and mortality in chronic hemodialysis patients (CHD). About 2-6% of chronic hemodialysis patients develop IE and the incidence is 50-60 times higher among CHD patients than in the general population. The left heart is the most frequent location of IE in CHD and the different published series report a prevalence of left valve involvement varying from 80% to 100%. Valvular and perivalvular abnormalities, alteration of the immune system, and bacteremia associated with repeated manipulation of the vascular access, particularly central venous catheters, comprise the main factors explaining the left heart IE in CHD patients. While left-sided IE develops in altered valves in a high-pressure system, right-sided IE on the contrary, generally develops in healthy valves in a low-pressure system. Right-sided IE is rare, with its incidence varying from 0% to 26% depending on the study, and the tricuspid valve is the main location. Might the massive influx of pathogenic and virulent germs via the central venous catheter to the right heart, with the tricuspid being the first contact valve, have a role in the physiopathology of IE in CHD, thus facilitating bacterial adhesion? While the physiopathology of left-sided IE entails multiple and convincing mechanisms, it is not the case for right-sided IE, for which the physiopathological mechanism is only partially understood and remains shrouded in mystery.
Pecini, Redi; Dalsgaard, Morten; Møller, Daniel Vega
Mitral regurgitation (MR) secondary to ischemic heart disease (IHD) increases during exercise. We tested the hypothesis that the same is also true for MR due to mitral valve prolapse (MVP).......Mitral regurgitation (MR) secondary to ischemic heart disease (IHD) increases during exercise. We tested the hypothesis that the same is also true for MR due to mitral valve prolapse (MVP)....
After 5 years the European Society of Cardiology and the European Association for Cardio-Thoracic Surgery have released an update on the guidelines for the management of valvular heart diseases. In recent years published results of randomized trials in patients with aortic valve stenosis have resulted in updated recommendations for catheter-based prosthesis implantation (TAVI), which is now extended to patients presenting without a low risk for conventional surgical valve replacement. In mitral or tricuspid valvular disease, the recommendations for catheter-based therapies are less strong because of a lack of supportive scientific data. A special focus of these updated guidelines is on concomitant antithrombotic therapy in valvular heart disease and in the context of a combination with coronary artery disease and/or accompanying arrhythmia. Special emphasis was again put on the multidisciplinary heart team for the diagnostics and treatment of patients with valvular heart disease. In order to support the quality of treatment for patients with valvular heart disease, it is suggested that heart valve centers of excellence should be established, which have to fulfil complex personnel, structural and technological prerequisites.
Möllmann, Helge; Linke, Axel; Holzhey, David M
OBJECTIVES: The aim of this study was to evaluate the short-term safety and performance of the full range of valve sizes offered within the Portico transcatheter aortic valve replacement system. BACKGROUND: The Portico transcatheter aortic heart valve is a fully resheathable, repositionable.......8% of patients improved by ≥1 New York Heart Association functional class at 30 days. The rate of moderate paravalvular leak was 5.7%, with no severe paravalvular leak reported. No differences in paravalvular leak incidence and severity were observed among valve sizes (p = 0.24). CONCLUSIONS: Across all valve...... sizes, use of the repositionable Portico transcatheter aortic valve replacement system resulted in safe and effective treatment of aortic stenosis in high-risk patients....
Gondrie, Martijn J. A.; van der Graaf, Yolanda; Jacobs, Peter C.; Oen, Ay L.; Mali, Willem P. Th. M.
Objectives This study aims to investigate the prognostic value of incidental aortic valve calcification (AVC), mitral valve calcification (MVC) and mitral annular calcification (MAC) for cardiovascular events and non-rheumatic valve disease in particular on routine diagnostic chest CT. Methods The study followed a case-cohort design. 10410 patients undergoing chest CT were followed for a median period of 17 months. Patients referred for cardiovascular disease were excluded. A random sample of...
Arvidsson, Per M; Kovács, Sándor J; Töger, Johannes; Borgquist, Rasmus; Heiberg, Einar; Carlsson, Marcus; Arheden, Håkan
The laws of fluid dynamics govern vortex ring formation and precede cardiac development by billions of years, suggesting that diastolic vortex ring formation is instrumental in defining the shape of the heart. Using novel and validated magnetic resonance imaging measurements, we show that the healthy left ventricle moves in tandem with the expanding vortex ring, indicating that cardiac form and function is epigenetically optimized to accommodate vortex ring formation for volume pumping. Healthy hearts demonstrate a strong coupling between vortex and cardiac volumes (R(2) = 0.83), but this optimized phenotype is lost in heart failure, suggesting restoration of normal vortex ring dynamics as a new, and possibly important consideration for individualized heart failure treatment. Vortex ring volume was unrelated to early rapid filling (E-wave) velocity in patients and controls. Characteristics of vortex-wall interaction provide unique physiologic and mechanistic information about cardiac diastolic function that may be applied to guide the design and implantation of prosthetic valves, and have potential clinical utility as therapeutic targets for tailored medicine or measures of cardiac health.
Rambod, Edmond; Beizaie, Masoud; Shusser, Michael; Gharib, Morteza
The growth mechanism of microbubbles at mitral MHV closure has been experimentally studied. In the heart, some of the tiny bubbles grow explosively and form larger and persistent bubbles. An experimental set-up was designed to allow the passage of micron-size bubbles through an 80 micron-wide slot, simulating a typical gap between the housing ring and the occluders in MHV. The bubbles were generated using an air-liquid dispenser and were delivered to the system via a 250 micron-diameter hypedermic needle positioned vertically near the slot. A solenoid valve was used to deliver a 10cc volume of liquid in 25ms time through the slot. High-speed imaging was used to study the impact of flow through the slot on bubble growth. The velocity of liquid through the slot was assessed to be in the range of 12-15 m/s. Our observations confirmed the rapid and drastic growth of microbubbles following their passage through the narrow slot, due to pressure drop. Vortices, which were induced by flow separation on the downstream of the slot, caused the grown bubbles to shatter and form more stable bubbles.
Stortecky, Stefan; Wenaweser, Peter; Windecker, Stephan
Transcatheter aortic valve implantation (TAVI) is an evidence-based treatment alternative for selected high-risk patients with symptomatic severe aortic stenosis as acknowledged in the most recent edition of the ESC Guidelines on Valvular Heart Disease 2012. However, periprocedural complications and in particular cerebrovascular accidents remain a matter of concern. While transcatheter heart valve technology continuously improves and the development of novel and even less invasive implantation techniques is on-going, cerebrovascular events complicating TAVI may abrogate the usual improvement in terms of prognosis and quality of life. This article describes the incidence of cerebrovascular events after cardiovascular procedures, provides an overview of the pathophysiological mechanisms as well as the impact on outcomes and provides some insights into preventive strategies as well as the acute management of these events.
Tomas A Salerno
Full Text Available OBJECTIVE: Mitral valve surgery can be performed through the trans-atrial or the trans-septal approach. Although the trans-atrial is the preferred method, the trans-septal approach has also been used recently and has a particular value in beating-heart mitral valve surgery. Herein we report our experience with beating-heart mitral valve surgery via trans-septal approach, and discuss its advantages and pitfalls. METHODS: Between 2000 and 2007, 214 consecutive patients were operated upon utilizing beating heart technique for mitral valve surgery. The operation was performed via transseptal approach with the aorta unclamped, the heart beating, with normal electrocardiogram and in sinus rhythm. RESULTS: Mean age was 56.03 ± 13.93 years (range: 19-86 years; median: 56 years. There were 131 (61.2% males and 83 (38.8% females. Of the prostheses used, 108 (50.5% were biological, and 39 (18.2% were mechanical. Mitral repairs were performed in 67 (31.3% patients. Mean hospital stay was 17.4 ± 20.0 days (range: 3-135 days; median: 11 days. Intra-aortic balloon pump (IABP utilization was required in 12 (5.6% of 214 patients. One-month mortality was 7.4%, and re-operation for bleeding was needed in 15 (7% patients. CONCLUSIONS: Beating-heart mitral valve surgery is an option for myocardial protection in patients undergoing mitral valve surgery. This technique is facilitated by the trans-septal approach due to reduced aortic insufficiency and improved visualization of the mitral apparatus.OBJETIVO: A cirurgia da valva mitral pode ser feita via transatrial ou transeptal. Embora a transatrial seja a preferida, a via transeptal tem sido utilizada mais recentemente e tido um grande valor nas operações com o coração batendo. Mostramos a nossa experiência na cirurgia da valva mitral via transeptal com coração batendo e discutimos seus benefícios e problemas. MÉTODOS: Entre 2000 e 2007, 214 pacientes consecutivos foram operados com o coração batendo. A
Aicher, Diana; Bewarder, Moritz; Kindermann, Michael; Abdul-Khalique, Hashim; Schäfers, Hans-Joachim
Unicuspid aortic valve (UAV) anatomy leads to dysfunction of the valve in young individuals. We introduced a reconstructive technique of bicuspidizing the UAV. Initially we copied the typical asymmetry of a normal bicuspid aortic valve (BAV) (I), later we created a symmetric BAV (II). This study compared the hemodynamic function of the two designs of a bicuspidized UAV. Aortic valve function was studied at rest and during exercise in 28 patients after repair of UAV (group I, n = 8; group II, n = 20). There were no differences among the groups I and II with respect to gender, age, body size, or weight. All patients were in New York Heart Association class I. Six healthy adults served as control individuals. All patients were studied with transthoracic echocardiography between 4 and 65 months postoperatively. Systolic gradients were assessed by continuous wave Doppler while patients were at rest and exercising on a bicycle ergometer. Aortic regurgitation was grade I or less in all patients. Resting gradients were significantly elevated in group I compared with group II and control individuals (group I, peak 33.8 ± 7.8 mm Hg; mean 19.1 ± 5.4 mm Hg; group II, peak 15.8 ± 5.4, mean 8.2 ± 2.8 mm Hg; control individuals, peak 6.0 ± 1.6, mean 3.2 ± 0.8 mm Hg; p competence. A symmetric repair design leads to improved systolic aortic valve function at rest and during exercise. Copyright © 2013 The Society of Thoracic Surgeons. Published by Elsevier Inc. All rights reserved.
Amri, Amna; Laroche, Gaetan; Chakfe, Nabil; Heim, Frederic
With over 150,000 implantations performed over the world, transcatheter aortic valve replacement (TAVR) has become a surgical technique, which largely competes with open surgery valve replacement for an increasing number of patients. The success of the procedure favors the research toward synthetic valve leaflet materials as an alternative to biological tissues, whose durability remains unknown. In particular, fibrous constructions have recently proven to be durable in vivo over a 6-month period of time in animal sheep models. Exaggerated fibrotic tissue formation remains, however, a critical issue to be addressed. This work investigates the design of a composite fibrous construction combining a woven polyethylene terephthalate (PET) layer and a non-woven PET mat, which are expected to provide, respectively, strength and appropriate topography toward limited fibrotic tissue ingrowth. For this purpose, a specific equipment has been developed to produce non-woven PET mats made from fibers with small diameter. These mats were assembled with woven PET substrates using various assembling techniques in order to obtain hybrid fibrous constructions. The physical and mechanical properties of the obtained materials were assessed and valve samples were manufactured to be tested in vitro for hydrodynamic performances. The results show that the composite fibrous construction is characterized by properties suitable for the valve leaflet function, but the durability of the assembling is however limited under accelerated cyclic loading.
Delate, Thomas; Meisinger, Stephanie M; Witt, Daniel M; Jenkins, Daniel; Douketis, James D; Clark, Nathan P
Bridge therapy is associated with an increased risk of major bleeding in patients with atrial fibrillation and venous thromboembolism (TE) without a corresponding reduction in TE. The benefits of bridge therapy in patients with mechanical heart valve (MHV) prostheses interrupting warfarin for invasive procedures are not well described. A retrospective cohort study was conducted at an integrated health-care delivery system. Anticoagulated patients with MHV interrupting warfarin for invasive diagnostic or surgical procedures between January 1, 2006, and March 31, 2012, were identified. Patients were categorized according to exposure to bridge therapy during the periprocedural period and TE risk (low, medium, and high). Outcomes validated via manual chart review included clinically relevant bleeding, TE, and all-cause mortality in the 30 days following the procedure. There were 547 procedures in 355 patients meeting inclusion criteria. Mean cohort age was 65.2 years, and 38% were female. Bridge therapy was utilized in 466 (85.2%) procedures (95.2%, 77.3%, and 65.8% of high, medium, and low TE risk category procedures, respectively). The 30-day rate of clinically relevant bleeding was numerically higher in bridged (5.8%; 95% confidence interval [CI], 3.9%-8.3%) versus not bridged procedures (1.2%; 95% CI, bridge therapy is common among patients with MHV and may be associated with increased bleeding risk. Further research is needed to determine whether bridge therapy reduces TE in patients with MHV interrupting warfarin for invasive procedures.
Matsuzaki, Yuichi; Tomioka, Hideyuki; Saso, Masaki; Azuma, Takashi; Saito, Satoshi; Aomi, Shigeyuki; Yamazaki, Kenji
Hereditary spherocytosis is a genetic, frequently familial hemolytic blood disease characterized by varying degrees of hemolytic anemia, splenomegaly, and jaundice. There are few reports on adult open-heart surgery for patients with hereditary spherocytosis. We report a rare case of an adult open-heart surgery associated with hereditary spherocytosis. A 63-year-old man was admitted for congestive heart failure due to bicuspid aortic valve, aortic valve regurgitation, and sinus of subaortic aneurysm. The family history, the microscopic findings of the blood smear, and the characteristic osmotic fragility confirmed the diagnosis of hereditary spherocytosis. Furthermore, splenectomy had not been undertaken preoperatively. The patient underwent a successful operation by means of a centrifugal pump. Haptoglobin was used during the cardiopulmonary bypass, and a biological valve was selected to prevent hemolysis. No significant hemolysis occurred intraoperatively or postoperatively. There are no previous reports of patients with hereditary spherocytosis, and bicuspid aortic valve. We have successfully performed an adult open-heart surgery using a centrifugal pump in an adult patient suffering from hereditary spherocytosis and bicuspid aortic valve.
Hutchison, Chris; Sullivan, Pierre; Ethier, C Ross
Computational modeling of bileaflet mechanical heart valve (BiMHV) flow requires experimentally validated datasets and improved knowledge of BiMHV fluid mechanics. In this study, flow was studied downstream of a model BiMHV in an axisymmetric aortic sinus using stereoscopic particle image velocimetry. The inlet flow was steady and the Reynolds number based on the aortic diameter was 7600. Results showed the out-of-plane velocity was of similar magnitude as the transverse velocity. Although additional studies are needed for confirmation, analysis of the out-of-plane velocity showed the possible presence of a four-cell streamwise vortex structure in the mean velocity field. Spatial data for all six Reynolds stress components were obtained. Reynolds normal stress profiles revealed similarities between the central jet and free jets. These findings are important to BiMHV flow modeling, though clinical relevance is limited due to the idealized conditions chosen. To this end, the dataset is publicly available for CFD validation purposes.
Marchisio, Daniele L; Galan, Miquel; Barresi, Antonello A
This manuscript shows how computational models, mainly based on Computational Fluid Dynamics (CFD), can be used to simulate different parts of an industrial freeze-drying equipment and to properly design them; in particular in this part the duct connecting the chamber with the condenser, with its valves, is considered, while the chamber design and its effect on drying kinetics have been investigated in Part 1. Such an approach allows a much deeper process understanding and assessment of the critical aspects of lyophilisation. This methodology will be demonstrated on freeze-drying equipment of different sizes, investigating influence of valve type (butterfly and mushroom) and shape on duct conductance and critical flow conditions. The role of the inlet and boundary conditions considered has been assessed, also by modelling the whole apparatus including chamber and condenser, and the influence of the duct diameter has been discussed; the results show a little dependence of the relationship between critical mass flux and chamber pressure on the duct size. Results concerning the fluid dynamics of a simple disk valve, a profiled butterfly valve and a mushroom valve installed in a medium size horizontal condenser are presented. Also in these cases the maximum allowable flow when sonic flow conditions are reached can be described by a correlation similar to that found valid for empty ducts; for the mushroom valve the parameters are dependent on the valve opening length. The possibility to use the equivalent length concept, and to extend the validity of the results obtained for empty ducts will be also discussed. Finally the presence of the inert gas modifies the conductance of the duct, reducing the maximum flow rate of water that can be removed through it before the flow is choked; this also requires a proper over-sizing of the duct (or duct-butterfly valve system). Copyright © 2018. Published by Elsevier B.V.
Gündüz, Sabahattin; Özkan, Mehmet; Kalçik, Macit; Gürsoy, Ozan Mustafa; Astarcioğlu, Mehmet Ali; Karakoyun, Süleyman; Aykan, Ahmet Çağri; Biteker, Murat; Gökdeniz, Tayyar; Kaya, Hasan; Yesin, Mahmut; Duran, Nilüfer Ekşi; Sevinç, Deniz; Güneysu, Tahsin
Distinguishing pannus and thrombus in patients with prosthetic valve dysfunction is essential for the selection of proper treatment. We have investigated the utility of 64-slice multidetector computed tomography (MDCT) in distinguishing between pannus and thrombus, the latter amenable to thrombolysis. Sixty-two (23 men, mean age 44±14 years) patients with suspected mechanical prosthetic valve dysfunction assessed by transesophageal echocardiography were included in this prospective observational trial. Subsequently, MDCT was performed before any treatment was started. Periprosthetic masses were detected by MDCT in 46 patients, and their attenuation values were measured as Hounsfield Units (HU). Patients underwent thrombolysis unless contraindicated, and those with a contraindication or failed thrombolysis underwent surgery. A mass which was completely lysed or surgically detected as a clot was classified as thrombus, whereas a mass which was surgically detected as tissue overgrowth was classified as pannus. A definitive diagnosis could be achieved in 37 patients with 39 MDCT masses (22 thrombus and 17 pannus). The mean attenuation value of 22 thrombotic masses was significantly lower than that in 17 pannus (87±59 versus 322±122; Ppannus from thrombus. Complete lysis was more common for masses with HUpannus overgrowth, whereas a lower value is associated with thrombus formation. A higher attenuation (HU>90) is associated with reduced lysis rates. © 2015 American Heart Association, Inc.
Caprari, Patrizia; Tarzia, Anna; Mojoli, Giorgio; Cianciulli, Paolo; Mannella, Emilio; Martorana, Maria Cristina
The implantation of a prosthetic heart valve (HVP) in patients with hereditary spherocytosis (HS) and hereditary elliptocytosis (HE) is rare, and the changes in the structure and deformability of erythrocytes that follow implantation in these patients have been poorly described. In the present study, the erythrocytes in HS and HE patients with mechanical HVP were compared to the erythrocytes in patients with only congenital membrane defects, in terms of biochemical modifications and rheological behaviour. Integral and cytoskeletal erythrocyte membrane proteins were studied, and blood viscosity (shear rate/shear stress ratio), aggregation ratio [eta(1 s(-1))/eta(200 s(-1))], and red cell visco-elasticity were determined. Valve replacement with a mechanical prosthesis worsened anaemia and resulted in a change in haemolysis, from sub-clinical to evident. The rheological investigation of erythrocytes from HS patients confirmed the characteristic increased viscosity and aggregation ratio and the decreased deformability. The rheological behaviour of erythrocytes from patients with HVP showed a decrease in viscosity and an increase in elastic modulus. In these patients, the prosthesis seems to have induced traumatic damage to the erythrocyte membrane, leading to fragmentation and lysis, which in turn modified rheological parameters. The biochemical and rheological investigation allowed us to understand the clinical and haematological pictures of the patients and to describe the role played by different factors in haemolytic anaemia.
The cardiac valves are targets of both congenital and acquired diseases. The formation of valves during embryogenesis (i.e., valvulogenesis) originates from endocardial cells lining the myocardium. These cells undergo an endothelial-mesenchymal transition, proliferate and migrate within an extracellular matrix. This leads to the formation of bilateral cardiac cushions in both the atrioventricular canal and the outflow tract. The embryonic origin of both the endocardium and prospective valve cells is still elusive. Endocardial and myocardial lineages are segregated early during embryogenesis and such a cell fate decision can be recapitulated in vitro by embryonic stem cells (ESC). Besides genetically modified mice and ex vivo heart explants, ESCs provide a cellular model to study the early steps of valve development and might constitute a human therapeutic cell source for decellularized tissue-engineered valves. This article is part of a Special Issue entitled: Cardiomyocyte Biology: Cardiac Pathways of Differentiation, Metabolism and Contraction. Copyright © 2012 Elsevier B.V. All rights reserved.
Windram, Jonathan D; Colman, Jack M; Wald, Rachel M; Udell, Jacob A; Siu, Samuel C; Silversides, Candice K
In women with valvular heart disease, pregnancy-associated cardiovascular changes can contribute to maternal, foetal and neonatal complications. Ideally, a woman with valvular heart disease should receive preconception assessment and counselling from a cardiologist with expertise in pregnancy. For women with moderate- and high-risk valve lesions, appropriate risk stratification and management during pregnancy will optimise outcomes. Pregnancy in women with high-risk lesions, such as severe aortic stenosis, severe mitral stenosis and those with mechanical valves, requires careful planning and coordination of antenatal care by a multidisciplinary team. The purpose of this overview is to describe the expected haemodynamic changes in pregnancy, review pregnancy risks for women with valvular heart disease and discuss strategies for management. Copyright © 2014. Published by Elsevier Ltd.
Keshavarz-Motamed, Zahra; Garcia, Julio; Gaillard, Emmanuel; Maftoon, Nima; Di Labbio, Giuseppe; Cloutier, Guy; Kadem, Lyes
Blood flow in the aorta has been of particular interest from both fluid dynamics and physiology perspectives. Coarctation of the aorta (COA) is a congenital heart disease corresponding to a severe narrowing in the aortic arch. Up to 85 % of patients with COA have a pathological aortic valve, leading to a narrowing at the valve level. The aim of the present work was to advance the state of understanding of flow through a COA to investigate how narrowing in the aorta (COA) affects the characteristics of the velocity field and, in particular, turbulence development. For this purpose, particle image velocimetry measurements were conducted at physiological flow and pressure conditions, with three different aorta configurations: (1) normal case: normal aorta + normal aortic valve; (2) isolated COA: COA (with 75 % reduction in aortic cross-sectional area) + normal aortic valve and (3) complex COA: COA (with 75 % reduction in aortic cross-sectional area) + pathological aortic valve. Viscous shear stress (VSS), representing the physical shear stress, Reynolds shear stress (RSS), representing the turbulent shear stress, and turbulent kinetic energy (TKE), representing the intensity of fluctuations in the fluid flow environment, were calculated for all cases. Results show that, compared with a healthy aorta, the instantaneous velocity streamlines and vortices were deeply changed in the presence of the COA. The normal aorta did not display any regions of elevated VSS, RSS and TKE at any moment of the cardiac cycle. The magnitudes of these parameters were elevated for both isolated COA and complex COA, with their maximum values mainly being located inside the eccentric jet downstream of the COA. However, the presence of a pathologic aortic valve, in complex COA, amplifies VSS (e.g., average absolute peak value in the entire aorta for a total flow of 5 L/min: complex COA: = 36 N/m2; isolated COA = 19 N/m2), RSS (e.g., average peak value in the entire aorta for a total flow of 5
Balocchia, R.; Varanini, M.; Paoletti, G.; Mecacci, G.; Santarcangelo, E.L.
The present study evaluated the heart-rate dynamics of subjects reporting decreased (responders) or paradoxically increased relaxation (nonresponders) at the end of a threatening movie. Heart-rate dynamics were characterized by indices extracted through recurrence quantification analysis (RQA) and
Sawaya, Fadi J; Spaziano, Marco; Lefèvre, Thierry; Roy, Andrew; Garot, Phillippe; Hovasse, Thomas; Neylon, Antoinette; Benamer, Hakim; Romano, Mauro; Unterseeh, Thierry; Morice, Marie-Claude; Chevalier, Bernard
To investigate the clinical outcomes of transcatheter aortic valve implantation (TAVI) with the SAPIEN 3 transcatheter heart valve (S3-THV) vs the SAPIEN XT valve (XT-THV). We retrospectively analyzed 507 patients that underwent TAVI with the XT-THV and 283 patients that received the S3-THV at our institution between March 2010 and December 2015. Thirty-day mortality (3.5% vs 8.7%; OR = 0.44, P = 0.21) and 1-year mortality (25.7% vs 20.1%, P = 0.55) were similar in the S3-THV and the XT-THV groups. The rates of both major vascular complication and paravalvular regurgitation (PVR) > 1 were almost 4 times lower in the S3-THV group than the XT-THV group (major vascular complication: 2.8% vs 9.9%, P 1: 2.4% vs 9.7%, P < 0.0001). However, the rate of new pacemaker implantation was almost twice as high in the S3-THV group (17.3% vs 9.8%, P = 0.03). In the S3 group, independent predictors of new permanent pacemaker were pre-procedural RBBB (OR = 4.9; P = 0.001), pre-procedural PR duration (OR = 1.14, P = 0.05) and device lack of coaxiality (OR = 1.13; P = 0.05) during deployment. The S3-THV is associated to lower rates of major vascular complications and PVR but higher rates of new pacemaker compared to the XT-THV. Sub-optimal visualization of the S3-THV in relation to the aortic valvular complex during deployment is a predictor of new permanent pacemaker.
Aleksandra Domanjko Petrič
Full Text Available Myxomatous mitral valve disease is a common cause of congestive heart failure in geriatric dogs. Many studies have been done in terms of epidemiology, pathology, associated neurohormonal changes in the disease progression, prognostic factors, and survival and treatment modalities. The presented paper presents a review of some of the studies in the mitral valve disease story.
Richard, M.C.; Chiou, D.
Utilities operating nuclear power plants recognize that the correct functioning of all motor-operated valves, and particularly those in safety-related systems, is of paramount importance. The U.S. Nuclear Regulatory Commission has issued Generic Letter 89-10 relative to this concern. Operability must be demonstrated under design-basis conditions. In order to demonstrate operability of motor-operated butterfly valves, the valve stem torque must be determined. The valve stem torque is a function of seat material, stem packing, stem bearing friction, and hydrodynamic lift and drag. The total valve operating hydrodynamic torque can be predicted using the valve manufacturer's data and the differential pressure. In order to validate the valve manufacturer's data, the actual total valve hydrodynamic torque is measured using strain gages mounted directly on the valve stem. This paper presents the results of comparing the predicted total valve operating hydrodynamic torque with the actual total valve operating hydrodynamic torque for six POSI-SEAL Class 150 high performance butterfly valves
Full Text Available We report a 42-year-old lady on combined oral contraceptive pills with deep venous thrombosis and mild pulmonary embolism, who developed shock, while being treated with heparin. On auscultation, there was an unexpected loud, superficial, squeaky, triphasic friction rub over the left lower parasternal area. Simultaneous echocardiogram revealed a large serpiginous, freely floating mass in right atrium, prolapsing freely to right ventricular inlet across the tricuspid valve. The endocardial friction rub persisted for about 24 h as long as the thrombus was intracardiac and disappeared when the thrombus disappeared from right heart, occluding the proximal left pulmonary artery. She died due to shock awaiting surgery. We highlight that such a rub in a patient with deep venous thrombosis, though rare, may be an important clue to impending pulmonary embolism.
Merryman, W David
The following hypothetical case study was developed for bioengineering students and is concerned with choosing between two devices used for development of a pediatric tissue engineered heart valve (TEHV). This case is intended to elicit assessment of the devices, possible future outcomes, and ramifications of the decision making. It is framed in light of two predominant ethical theories: utilitarianism and rights of persons. After the case was presented to bioengineering graduate students, they voted on which device should be released. The results revealed that these bioengineering students preferred the more reliable (and substantially more expensive) design, though this choice precludes the majority of the world from having access to this technology. This case is intended to examine and explore where the balance lies between design, cost, and adequate distribution of biomedical devices.
Roorda, O. [SMX International, Toronto, ON (Canada)
Check valves perform an important function in preventing reverse flow and protecting plant and mechanical equipment. However, the variety of different types of valves and extreme differences in performance even within one type can change maintenance requirements and life cycle costs, amounting to millions of dollars over the typical 15-year design life of piping components. A next-generation non-slam nozzle check valve which prevents return flow has greatly reduced operating costs by protecting the mechanical equipment in a piping system. This article described the check valve varieties such as the swing check valve, a dual-plate check valve, and nozzle check valves. Advancements in optimized design of a non-slam nozzle check valve were also discussed, with particular reference to computer flow modelling such as computational fluid dynamics; computer stress modelling such as finite element analysis; and flow testing (using rapid prototype development and flow loop testing), both to improve dynamic performance and reduce hydraulic losses. The benefits of maximized dynamic performance and minimized pressure loss from the new designed valve were also outlined. It was concluded that this latest non-slam nozzle check valve design has potential applications in natural gas, liquefied natural gas, and oil pipelines, including subsea applications, as well as refineries, and petrochemical plants among others, and is suitable for horizontal and vertical installation. The result of this next-generation nozzle check valve design is not only superior performance, and effective protection of mechanical equipment but also minimized life cycle costs. 1 fig.
Eldiwany, B.H.; Sharma, V.; Kalsi, M.S.; Wolfe, K.
As part of the Motor-Operated Valve (MOV) Performance Prediction Program, the Electric Power Research Institute has sponsored the development of methodologies for predicting thrust and torque requirements of gate, globe, and butterfly MOVs. This paper presents the methodology that will be used by utilities to calculate the dynamic torque requirements for butterfly valves. The total dynamic torque at any disc position is the sum of the hydrodynamic torque, bearing torque (which is induced by the hydrodynamic force), as well as other small torque components (such as packing torque). The hydrodynamic torque on the valve disc, caused by the fluid flow through the valve, depends on the disc angle, flow velocity, upstream flow disturbances, disc shape, and the disc aspect ratio. The butterfly valve model provides sets of nondimensional flow and torque coefficients that can be used to predict flow rate and hydrodynamic torque throughout the disc stroke and to calculate the required actuation torque and the maximum transmitted torque throughout the opening and closing stroke. The scope of the model includes symmetric and nonsymmetric discs of different shapes and aspects ratios in compressible and incompressible fluid applications under both choked and nonchoked flow conditions. The model features were validated against test data from a comprehensive flowloop and in situ test program. These tests were designed to systematically address the effect of the following parameters on the required torque: valve size, disc shapes and disc aspect ratios, upstream elbow orientation and its proximity, and flow conditions. The applicability of the nondimensional coefficients to valves of different sizes was validated by performing tests on 42-in. valve and a precisely scaled 6-in. model. The butterfly valve model torque predictions were found to bound test data from the flow-loop and in situ testing, as shown in the examples provided in this paper
Nakashima, Kouki; Itatani, Keiichi; Kitamura, Tadashi; Oka, Norihiko; Horai, Tetsuya; Miyazaki, Shohei; Nie, Masaki; Miyaji, Kagami
Mitral valve morphology after mitral valve surgery affects postoperative intraventricular flow patterns and long-term cardiac performance. We visualized ventricular flow by echocardiography vector flow mapping (VFM) to reveal the impact of different mitral valve procedures. Eleven cases of mechanical mitral valve replacement (nine in the anti-anatomical and two in the anatomical position), three bioprosthetic mitral valve replacements, and four mitral valve repairs were evaluated. The mean age at the procedure was 57.4 ± 17.8 year, and the echocardiography VFM in the apical long-axis view was performed 119.9 ± 126.7 months later. Flow energy loss (EL), kinetic pressure (KP), and the flow energy efficiency ratio (EL/KP) were measured. The cases with MVR in the anatomical position and with valve repair had normal vortex directionality ("Clockwise"; N = 6), whereas those with MVR in the anti-anatomical position and with a bioprosthetic mitral valve had the vortex in the opposite direction ("Counterclockwise"; N = 12). During diastole, vortex direction had no effect on EL ("Clockwise": 0.080 ± 0.025 W/m; "Counterclockwise": 0.083 ± 0.048 W/m; P = 0.31) or KP ("Clockwise": 0.117 ± 0.021 N; "Counterclockwise": 0.099 ± 0.057 N; P = 0.023). However, during systole, the EL/KP ratio was significantly higher in the "Counterclockwise" vortex than that in the "Clockwise" vortex (1.056 ± 0.463 vs. 0.617 ± 0.158; P = 0.009). MVP and MVR with a mechanical valve in the anatomical position preserve the physiological vortex, whereas MVR with a mechanical valve in the anti-anatomical position and a bioprosthetic mitral valve generate inefficient vortex flow patterns, resulting in a potential increase in excessive cardiac workload.
Horstkotte, D; Körfer, R; Seipel, L; Bircks, W; Loogen, F
Valve-related complications after Björk-Shiley mitral valve implantation (n = 475), aortic valve implantation (n = 424), or mitral-aortic valve implantation (n = 119) were compared with those after St. Jude Medical mitral valve replacement (n = 173), aortic valve replacement (n = 152), or mitral-aortic valve replacement (n = 69). All patients were placed on anticoagulant therapy with phenprocoumon early after operation. All patients had a comparable follow-up time of approximately 23 months, which showed that cumulative thromboembolic rates were significantly higher after St. Jude valve implantation than after Björk-Shiley valve implantation. Reoperations were necessary because of valve thrombosis (0.46%), perivalvular leakage (2.2%), or prosthetic valve endocarditis with perivalvular regurgitation (0.46%). One Björk-Shiley mitral valve prosthesis had to be replaced because of fracture of the outlet strut. Without significant intergroup differences, hemorrhage due to anticoagulant treatment was the most frequent complication. Thromboembolic complications were significantly more frequent after Björk-Shiley mitral, aortic, and double valve replacements than after St. Jude valve implantation. This may lead to consideration of changes in the prophylaxis of thrombus formations in the St. Jude valve, especially in aortic valve replacements, in patients with sinus rhythm.
He Dakuo; Dong Gang; Wang Fuli; Mao Zhizhong
A chaotic sequence based differential evolution (DE) approach for solving the dynamic economic dispatch problem (DEDP) with valve-point effect is presented in this paper. The proposed method combines the DE algorithm with the local search technique to improve the performance of the algorithm. DE is the main optimizer, while an approximated model for local search is applied to fine tune in the solution of the DE run. To accelerate convergence of DE, a series of constraints handling rules are adopted. An initial population obtained by using chaotic sequence exerts optimal performance of the proposed algorithm. The combined algorithm is validated for two test systems consisting of 10 and 13 thermal units whose incremental fuel cost function takes into account the valve-point loading effects. The proposed combined method outperforms other algorithms reported in literatures for DEDP considering valve-point effects.
Schwitter, J. [Div. of Cardiology and Cardiovascular, MR Center, Univ. Hospital Zurich (Switzerland)
For clinical evaluation and decision-making in patients with valvular heart disease, the diagnostic armamentarium expands steadily. This evolution makes it difficult to choose the most appropriate approach for a specific valvular lesion. It may also reflect our uncertainty of what are the findings that best predict clinical outcome of patients, e. g. after surgery. Accordingly, for each type of valvular lesion, some pathophysiological considerations are stated in order to derive the most important measures that would allow optimal guidance of patients. Based on these considerations the value of an MR study is discussed for each valvular lesion. Newest advances in MR technology allow for highly accurate measurements of regurgitant volumes and hence, MR may be the method of choice for a quantitative evaluation of regurgitant valves. For assessment of stenosis severity, measurement of transvalvular pressure gradient is an appropriate measure and MR may not confer benefits over echocardiography, provided the ultrasound window is adequate (and stroke volume is in the normal range). With respect to surgical treatment, valvular morphology is of pivotal importance, particularly for the mitral valve, and echocardiography still appears to be the first line method. Little data relate lesion severity and/or morphology to clinical outcome. Conversely, the extent of cardiac adaptation to pressure- and/or volume-overload, i. e. ventricular remodeling, is a strong predictor of outcome, and is therefore most important for final judgement of the patient. For assessment of left and right ventricular remodeling, echocardiography typically provides all the necessary information. However, in special cases with discrepant findings, with inadequate ultrasound window, or in the preoperative work-up, MR may provide important information regarding cardiac adaptation to valvular lesion. (orig.) [German] Das diagnostische Repertoire zur Evaluation von Klappenvitien ist in den letzten Jahren
Sharma, Ashutosh; Cote, Anita T; Hosking, Martin C K; Harris, Kevin C
The aim of this study was to systematically evaluate the incidence of infective endocarditis (IE) in right ventricle-to-pulmonary artery conduits and valves, comparing bovine jugular vein (BJV) valves with all others. Recent evidence suggests that the incidence of IE is higher in patients with congenital heart disease who have undergone implantation of BJV valves in the pulmonary position compared with other valves. Systematic searches of published research were conducted using electronic databases (MEDLINE, Embase, and CINAHL) and citations cross-referenced current to April 2016. Included studies met the following criteria: patients had undergone right ventricle-to-pulmonary artery conduit or percutaneous pulmonary valve implantation, and investigators reported on the type of conduit or valve implanted, method of intervention (surgery or catheter based), IE incidence, and follow-up time. Fifty studies (Levels of Evidence: 2 to 4) were identified involving 7,063 patients. The median cumulative incidence of IE was higher for BJV compared with other valves (5.4% vs. 1.2%; p endocarditis with BJV valves than other types of right ventricle-to-pulmonary artery conduits. There was no difference in the incidence of endocarditis between catheter-based bovine valves and surgically implanted bovine valves, suggesting that the substrate for future infection is related to the tissue rather than the method of implantation. Copyright © 2017 American College of Cardiology Foundation. Published by Elsevier Inc. All rights reserved.
Full Text Available AIM AND OBJECTIVE: The aim of this study was to determine the effectiveness and results of repair of Organic Tricuspid Valve disease. INTRODUCTION : since tricuspid valve disease most often found in association with other valve disease. Isolated tricuspid valve disease is ra re. Pattern of involvement of tricuspid valve disease shows functional (75% and primary (organic in (25%. Surgical repair of organic tricuspid valve disease often fails because of abnormal valve. This usually leads to limited options. This study examine s our experience of tricuspid valve repair with autologous pericardium for organic tricuspid valve disease. MATERIAL AND METHODS : From Jan 2014 to May 2015, 22 patients underwent repairs for organic tricuspid valve disease. The patient aged 15 to 65 years and all were in New York Heart Association (NYHA class of III or IV. All patients presented with severe tricuspid disease coexisting with other cardiac pathology, usually left - sided heart valve disease. Repair techniques included Commisurotomy, division o f secondary chordae, Glutaraldehyde treated autologous pericardial patch augmentation of tricuspid valve leaflets, anterior papillary muscle advancement etc with or without ring/suture annuloplasty. Follow - up duration was 3 to 18 months. RESULTS : No deaths or late reoperations occurred. All patients demonstrated clinical improvements on follow up. Echocardiographic studies before hospital discharge showed less than mild tricuspid regurgitation in all patients except one. CONCLUSIONS : Large majorit y of organic tricuspid valve regurgitation is repairable with acceptable early results. Tricuspid stenosis and mixed tricuspid valve disease are more challenging. In the latter group, it is a judgment call whether to accept a suboptimal result or replace t he valve
Gürsoy, Mustafa Ozan; Kalçık, Macit; Yesin, Mahmut; Karakoyun, Süleyman; Bayam, Emrah; Gündüz, Sabahattin; Özkan, Mehmet
Prosthetic valve thrombosis is one of the major causes of primary valve failure, which can be life-threatening. Multimodality imaging is necessary for determination of leaflet immobilization, cause of underlying pathology (thrombus versus pannus or both), and whether thrombolytic therapy attempt in the patient would be successful or surgery is needed. Current guidelines for the management of prosthetic valve thrombosis lack definitive class I recommendations due to lack of randomized controlled trials, and usually leave the choice of treatment to the clinician’s experience. In this review, we aimed to summarize the pathogenesis, diagnosis, and management of mechanical prosthetic valve thrombosis. PMID:28005024
Zeng, Y I; Sun, Rongrong; Li, Xianchi; Liu, Min; Chen, Shuang; Zhang, Peiying
Valvular heart disease (VHD) is caused by either damage or defect in one of the four heart valves, aortic, mitral, tricuspid or pulmonary. Defects in these valves can be congenital or acquired. Age, gender, tobacco use, hypercholesterolemia, hypertension, and type II diabetes contribute to the risk of disease. VHD is an escalating health issue with a prevalence of 2.5% in the United States alone. Considering the likely increase of the aging population worldwide, the incidence of acquired VHD is expected to increase. Technological advances are instrumental in identifying congenital heart defects in infants, thereby adding to the growing VHD population. Almost one-third of elderly individuals have echocardiographic or radiological evidence of calcific aortic valve (CAV) sclerosis, an early and subclinical form of CAV disease (CAVD). Of individuals ages >60, ~2% suffer from disease progression to its most severe form, calcific aortic stenosis. Surgical intervention is therefore required in these patients as no effective pharmacotherapies exist. Valvular calcium load and valve biomineralization are orchestrated by the concerted action of diverse cell-dependent mechanisms. Signaling pathways important in skeletal morphogenesis are also involved in the regulation of cardiac valve morphogenesis, CAVD and the pathobiology of cardiovascular calcification. CAVD usually occurs without any obvious symptoms in early stages over a long period of time and symptoms are identified at advanced stages of the disease, leading to a high rate of mortality. Aortic valve replacement is the only primary treatment of choice. Biomarkers such as asymmetric dimethylarginine, fetuin-A, calcium phosphate product, natriuretic peptides and osteopontin have been useful in improving outcomes among various disease states. This review, highlights the current understanding of the biology of VHD, with particular reference to molecular and cellular aspects of its regulation. Current clinical questions
Helgadottir, Anna; Thorleifsson, Gudmar; Gretarsdottir, Solveig
Aortic valve stenosis (AS) is the most common valvular heart disease, and valve replacement is the only definitive treatment. Here we report a large genome-wide association (GWA) study of 2,457 Icelandic AS cases and 349,342 controls with a follow-up in up to 4,850 cases and 451,731 controls...
Full Text Available Renal transplant recipients are more prone to developing infections. We report a 37-year old renal transplant recipient who developed infective endocarditis of the aortic valve, heart failure and renal allograft dysfunction. He underwent aortic valve replacement which was followed by improvement in cardiac as well as allograft function.
Jahnavi, S; Saravanan, U; Arthi, N; Bhuvaneshwar, G S; Kumary, T V; Rajan, S; Verma, R S
Major challenge in heart valve tissue engineering for paediatric patients is the development of an autologous valve with regenerative capacity. Hybrid tissue engineering approach is recently gaining popularity to design scaffolds with desired biological and mechanical properties that can remodel post implantation. In this study, we fabricated aligned nanofibrous Bio-Hybrid scaffold made of decellularized bovine pericardium: polycaprolactone-chitosan with optimized polymer thickness to yield the desired biological and mechanical properties. CD44 + , αSMA + , Vimentin + and CD105 - human valve interstitial cells were isolated and seeded on these Bio-Hybrid scaffolds. Subsequent biological evaluation revealed interstitial cell proliferation with dense extra cellular matrix deposition that indicated the viability for growth and proliferation of seeded cells on the scaffolds. Uniaxial mechanical tests along axial direction showed that the Bio-Hybrid scaffolds has at least 20 times the strength of the native valves and its stiffness is nearly 3 times more than that of native valves. Biaxial and uniaxial mechanical studies on valve interstitial cells cultured Bio-Hybrid scaffolds revealed that the response along the axial and circumferential direction was different, similar to native valves. Overall, our findings suggest that Bio-Hybrid scaf