Li Xianjie; Qiu Shoukang; Zhou Jianliang; Liu Chunkui; Pan Jialin; Yang Mingli
Calibration facilities for radon fluxmeter with three kinds of different emanation medium have been developed. The stability of radon flux is 5%, 9% (RSD) respectively. The uniformity of radon flux is 4.5%, 8.5% (RSD) respectively. These specifications fulfill the calibration requirement for radon fluxmeter. The determination of radon flux of facility takes full account of eliminating the main error source-attenuation effect (including leakage and back diffusion etc.): not only prevent attenuation and make a relevant correction. Therefore the accuracy of determination is assured. The calibration, intercomparison of radon flux meter and the quantitatively evaluation on the measurement method of radon flux are made to be possible by the successful establishment of this facility. (author)
Gee, G.W.; Ward, A.L.; Caldwell, T.G.; Ritter, J.C.
Unsaturated water flux densities are needed to quantify water and contaminant transfer within the vadose zone. However, water flux densities are seldom measured directly and often are predicted with uncertainties of an order or magnitude or more. A water fluxmeter was designed, constructed, and tested to directly measure drainage fluxes in field soils. The fluxmeter was designed to minimize divergence. It concentrates flow into a narrow sensing region filled with a fiberglass wick. The wick applies suction, proportional to its length, and passively drains the meter. The meter can be installed in an augured borehole at almost any depth below the root zone. Water flux through the meter is measured with a self‐calibrating tipping bucket, with a sensitivity of ∼4 mL tip−1. For our meter this is equivalent to detection limit of ∼0.1 mm. Passive‐wick devices previously have not properly corrected for flow divergence. Laboratory measurements supported predictions of a two‐dimensional (2‐D) numerical model, which showed that control of the collector height H and knowledge of soil hydraulic properties are required for improving divergence control, particularly at fluxes below 1000 mm yr−1. The water fluxmeter is simple in concept, is inexpensive, and has the capability of providing continuous and reliable monitoring of unsaturated water fluxes ranging from less than 1 mm yr−1 to more than 1000 mm yr−1.
Audouin, L.; Hovhanessian, G.
A new device composed of a prefabricated concrete block including temperature sensors and resistance wire that are used to identify wall material properties and thermal flux transmitted to the wall, has been validated in the lab. This prefabricated blocks have to be integrated into the structure to monitor. These device provides real time information about thermal flux in the walls that can be useful for the monitoring of deterioration of wall physical properties (conductivity and diffusivity) due to aging or in case of accidents or during fire
Full Text Available This paper describes the use of modern measuring and data acquisition system for determining characteristics of feromagnetic material. For this purpose data acquisition card NI USB-6009, PC with data acquisition software and fluxmeter Electrical Steel Measuring System MPG 100 D were used. Based on the results obtained by measurements the modeling of hysteresis loop is performed by using appropriate mathematical model.
Billan, J.; Henrichsen, K.N.; Walckiers, L.
The main problems related to magnetic measurements of particle accelerator components are discussed. Measurements of the properties of magnetic materials as well as the measurements of field distribution in the electromagnets for the Large Electron-Positron Collider (LEP) are illustrated. The fluxmeter method is extensively employed in this work. The impact of recent advances in electronic technology on measurement techniques is explained. Magnetic measurements (including the harmonic coil method) can be performed with improved accuracy applying modern technology to the classical methods. New methods for the non-destructive testing of magnetic materials and for the measurement of magnetic geometry are described. (orig.) [de
The combustion heat of excess hydrogen in chloro fluoride compounds ClF, ClF 3 , ClF 5 and oxychloro fluoride compounds O 2 ClF, O 3 ClF, OClF 3 were determined in an original bomb calorimeter. This apparatus which can work at temperature up to 473K and under 10 atmospheres pressure as two compartments and high frequency electric spark ignition. The enthalpies of formation and bond energies are calculated. The temperature and enthalpies of the solid/solid and solid/liquid transformations were determined with a differential micro-calorimeter, of the fluxmeter type (M.C.B.) (sensibility 70μV detector/mW.cell, temperature range 80-1000K, maximal pressure 15 atmospheres) [fr
A transportable 11.7-12.5-GHz flux-density measurement facility for use in the in-orbit testing (IOT) of the FRG TV-Sat direct-broadcast satellites is described. Major components include a 1.2-m-diameter antenna, the fluxmeter, a radiometer to determine atmospheric attenuation, a weather station, and a control and data-processing computer; all of the components are mounted on a 5.10 x 2.35 x 2.70-m trailer. IOT performance parameters include gain/temperature ratio 15.9 dB/K, measurement range -97 to -117 dBW/sq m, measurement accuracy less than 0.5 dB rms, and measurement rate 250-650 msec. Photographs and a block diagram are provided.
A 304L mixing tee mock-up is instrumented to assess the fluctuating temperature in the mixing area generated by two fluids (water) at large gap of temperature meet. The turbulent mixing layer impacts the structure wall and creates stresses, which lead to the damages. The case studied in this paper corresponds to the 'swinging streak' within a flow rate ratio of 25 %. The instrumentation is specifically planned to measure the fluctuating temperature in the fluid close to the internal skin and inside the wall too. This experiment is performed using a new sensor 'fluxmeter' which is non intrusive and typically designed to catch the fluctuation without any signal attenuation, within a frequency range 0-25Hz. The facility called 'Fatherino' supplies an available delta T of 70 degree C in water at 4 m/s mixture velocity in a mixing tee mock-up 50 mm in diameter. The flow features generate a large turbulent flow in the mixing layer and favour the heat flux transfer to the wall. By applying an inverse heat conduction method applied to the output data given by the fluxmeter, both the heat flux is deduced and the temperature (mean and fluctuating values) at the internal surface can be accurately determined. In addition, a calculation using the Trio U code (thermal hydraulic code) within the large eddy simulation module is computed to assess the fluid temperature distribution in the mixing area close to the internal surface. The output data in mean and standard deviation are compared with the Fatherino measurements. The comparison consists in analysing the main parameters as the mean and standard deviation in the fluid along the main axis and in a circumferential view. The mixing layer geometry and the frequency of the fluctuation are also analysed. These experiments added to the calculation allow us improving the state of the knowledge in the mixing tees and the thermal load to be used in the industrial mixing tees in operating for the long lifetime assessment or for the
Zaborcwski, H [Commissariat a l' Energie Atomique, Cadarache (France). Centre d' Etudes Nucleaires
A study is made of the general characteristics and the applications of the multisphere technique for flux measurements, dosimetry and the spectrometry of neutron sources ranging from thermal, up to 7 MeV neutrons. We give the equations relating to three fluxmeters (proportional long counters) for the measurements of the flux. Other equations have been derived : 1 - for multi-collision rad and rem dosimetry and first collision rad dosimetry; 2 - for the spectrometry and dosimetry of neutrons in five energy bands using a method similar to that used by threshold detectors. (author) [French] Nous etudions les caracteristiques generales et les applications de la technique multisphere pour les mesures de flux, la dosimetrie et la spectrometrie des sources de neutrons depuis les thermiques jusqu'a 7 MeV. Nous donnons les equations relatives a trois fluxmetres (longs compteurs equivalents) pour les mesures en flux. D'autres equations ont ete derivees: 1 - pour la dosimetrie en rad et en rem de multicollision et en rad de premiere collision; 2 - pour la spectrometrie et la dosimetrie des neutrons en cinq bandes energetiques suivant une methode voisine de celle utilisee par detecteurs a seuil. (auteur)
Rizzante, Fabio A P; Maenosono, Rafael M; Duarte, Marco A H; Furuse, Adilson Y; Palma-Dibb, Regina G; Ishikiriama, Sérgio K
Dentin hypersensitivity treatments are based on the physical obliteration of the dentinal tubules to reduce hydraulic conductance. The aim of the present study is to evaluate the hydraulic conductance of bovine root dentin after irradiation with a 980-nm diode laser, with or without associated fluoride varnish. Sixty bovine root dentin specimens were divided into six groups (n = 10 in each group): G1, G3, and G5 (0.5 W, 0.7 W, and 1 W diode laser, respectively); G2, G4, and G6 (fluoride varnish application + 0.5 W, 0.7 W, and 1 W diode laser, respectively). The dentin hydraulic conductance was evaluated at four time periods with a fluxmeter: 1) with smear layer, 2) after 37% phosphoric acid etching, 3) after the treatments, and 4) after 6% citric acid challenge. After the dentinal fluid flow measurements, specimens were also evaluated for mineral composition using energy dispersive X-ray spectroscopy (EDS). Analysis demonstrated a better result with increased irradiation power (P diode laser irradiation was associated with the application of fluoride varnish (P laser irradiation, the 1 W group was superior when compared with the 0.5 W and 0.7 W irradiated groups immediately after treatment (P laser treatments. Laser irradiation of exposed dentin promoted significant reduction in the dentin hydraulic conductance, mainly with higher energy densities and association with fluoride varnish.
Soltani, Mohamed Larbi; Touares, Abdelhay; Aboki, Tiburce A. M.; Gasser, Jean-Georges
In the present work, we study the influence of thermal treatments on the magnetic properties of as-quenched and pre-crystallized Fe78Si9B13 after stress relaxation. The crystallization behavior of amorphous and treated Fe78Si9B13 ribbons was revisited. The measurements were carried out by means of Differential Scanning Calorimetry, by X-ray diffraction and by Vibrating Sample Magnetometer, Susceptometer and fluxmeter. Relaxed samples were heated in the resistivity device up to 700°C and annealed near the onset temperature about 420°C for respectively 1, 3, 5, 8 hours. In as-quenched samples, two transition points occur at about 505°C and 564°C but in relaxed sample, the transition points have been found about 552°C and 568°C. Kinetics of crystallization was deduced for all studied samples. Annealing of the as-purchased ribbon shows the occurrence of α-Fe and tetragonal Fe3B resulting from the crystallization of the remaining amorphous phase. The effects on magnetic properties were pointed out by relating the structural evolution of the samples. The magnetic measurements show that annealing change the saturation magnetization and the coercive magnetic field values, hence destroying the good magnetic properties of the material. The heat treatment shows that the crystallization has greatly altered the shape of the cycles and moved the magnetic saturation point of the samples. The effect of treatment on the magneto-crystalline anisotropy is also demonstrated.
Campbell, Timothy J; Hatfield, Kirk; Klammler, Harald; Annable, Michael D; Rao, P S C
A new configuration of the passive fluxmeter (PFM) is presented that provides for simultaneous measurements of both the magnitude and the direction of ambient groundwater specific discharge qo and Cr(VI) mass flux J(Cr). The PFM is configured as a cylindrical unit with an interior divided into a center section and three outer sectors, each packed with a granular anion exchange resin having high sorption capacity for the Cr(VI) oxyanions CrO4(2-) and HCrO4-. The sorbent in the center section is preloaded with benzoate as the "resident" tracer. Laboratory experiments were conducted in which PFMs were placed in porous packed bed columns, through which was passed a measured volume of synthetic groundwater containing Cr(VI). During the deployment period, some of the resident tracer is depleted while the Cr(VI) is sorbed. The resin was then removed from the four sectors separately and extracted to determine the "captured" mass of Cr(VI) and the residual mass of the resident tracer in each. Cumulative specific discharge, q0t, values were assessed using the residual mass of benzoate retained in the center section. The direction of this discharge theta was ascertained from the mass distribution of benzoate intercepted and retained in the outer three sections of the PFM. Cumulative chromium fluxes, J(Cr)t, were quantified using the total Cr(VI) mass intercepted and retained on the PFM. Experiments produced an average measurement error for direction theta of 3 degrees +/- 14 degrees, while the average measurement errors for q0 and J(Cr) were, respectively, -8% +/- 15% and -12% +/- 23%. Results demonstrate the potential utility of the new PFM configuration for characterizing groundwater and contaminant fluxes.
Clothier, B. E.; van der Velde, M.; Green, S. R.; Gee, G. W.; Manu, V.; Menoniti, V.; Vanclooster, M.
Intensification of agriculture on the raised coral atolls of the Tongan archipelago, notably through squash-pumpkin production, has lead to increased use of agrichemicals. Agrichemicals, both fertilisers and pesticides, pose a risk to these fragile environments. Sustainable land-management practices are needed for small-island developing states. On Tongatapu, solutes leaving the rootzone of the squash can rapidly find their way to the underlying freshwater lenses. These lenses are hydraulically linked to the internal lagoon, and the fringing reefs. We have used buried, non-suction fluxmeters to monitor both the quantity and quality of drainage leaving the rootzone of squash. Fertiliser is traditionally applied at planting. During establishment of the squash in 2003, some 350 mm of rain fell, with 70 % of this leaving the rootzone of this permeable soil as drainage. The concentration of nitrate-N in the drainage water was measured at around 50 mg-N/L. All of the initial fertiliser dressing had been lost, along with N mineralised from the plowed-in grass. Pesticides are needed in humid tropical environments to control weeds, pests and diseases. These chemicals can leach though the rootzone to contaminate receiving waters. We modeled the transport and fate of the presticides used in squash production, and we developed a Decision Support Tool (DST). Our DST can be used to select the best pesticides for local conditions, to tailor practices for minimising leaching losses below the rootzone, and to avoid the build-up of residues in the soil. This project, funded by the European Union and NZAID, took a multi-disciplinary approach through measurement and modeling protocols. Our DST enabled us to engage the wider community and stakeholders. There has been increased awareness of the impacts and risks associated with productive land management in the fragile hydrological environments of this small-island developing state.
Khaled, Mahmoud; Harambat, Fabien; Peerhossaini, Hassan
The present paper reports an experimental study of the aerothermal phenomena in the vehicle underhood compartment as investigated by measuring temperature, convective heat flux, and radiative heat flux. Measurements are carried out on a passenger vehicle in wind tunnel S4 of Saint-Cyr-France. The underhood space is instrumented by 120 surface and air thermocouples and 20 fluxmeters. Measurements are performed for three thermal functioning conditions while the engine is in operation and the front wheels are positioned on the test facility with power-absorption-controlled rollers. In the thermal analysis, particular attention is given to measuring absorbed convective heat fluxes at component surfaces. It is shown that, in some components, the outside air entering the engine compartment (for cooling certain components) can in fact heat other components. This problem arises from the underhood architecture, specifically the positioning of some components downstream of warmer components in the same airflow. Optimized thermal management suggests placing these components further upstream or isolating them from the hot stream by deflectors. Given style constraints, however, the use of air deflectors is more suitable than underhood architectural changes. Much of the present paper is devoted to heat flux analysis of the specific thermal behaviours in the underhood compartment (especially the absorption of convective heat fluxes) and to a description of a new control approach exploiting air deflectors to optimize underhood aerothermal management. - Research highlights: → We present a physical analysis of particular underhood aerothermal behaviors. → In this analysis, convective heat flux absorption should be noted. → A new optimization procedure based on this physical analysis is proposed. → It entails airflow redistribution in the underhood through deflectors. → The new procedures are simple and easy to implement in the car underhood.
Mroos, Katja; Baroni, Gabriele; Er-Raki, Salah; Francke, Till; Khabba, Said; Jarlan, Lionel; Hanich, Lahoucine; Oswald, Sascha E.
Irrigation water requirement plays a crucial role in many agricultural areas and especially in arid and semi-arid landscapes. Improvements in the water management and the performance of the irrigation systems require a correct evaluation of the hydrological processes involved. However, some difficulties can arise due to the heterogeneity of the soil-plant system and of the irrigation scheme. To overcome these limitations, in this study, the soil water balance is analyzed by the combination of the Eddy Covariance technique (EC) and Cosmic Ray neutron Sensing (CRS). EC provides the measurement of the actual evapotranspiration over the area as it was presented in many field conditions. Moreover CRS showed to be a valuable approach to measure the root zone soil moisture integrated in a footprint of ~30 ha. In this way, the combination of the two methodologies should provide a better analysis of the soil water balance at field scale, as opposed to point observations, e.g. by TDR, evaporimeter and fluxmeter. Then, this could increase the capability to assess the irrigation efficiency and the agricultural water management. The study is conducted in a citrus orchard situated in a semi-arid region, 30 km southwest of Marrakesh (Morocco). The site is flat and planted with trees of same age growing in parallel rows with drip irrigation lines and application of fertilizer and pesticides. The original soil seems modified on the surface by the agricultural use, creating differences between trees, rows and lines. In addition, the drip irrigation creates also a spatial variability of the water flux distribution in the field, making this site an interesting area to test the methodology. Particular attention is given to the adaptation of the standard soil sampling campaign used for the calibration of the CRS and the introduction of a weighing function. Data were collected from June to December 2013, which corresponds to the high plant transpiration. Despite the intention of the
Chu, F.; Haines, P.; Hudson, M.; Kress, B.; Freidel, R.; Kanekal, S.
Work is underway by several groups to quantify diffusive radial transport of radiation belt electrons, including a model for pitch angle scattering losses to the atmosphere. The radial diffusion model conserves the first and second adiabatic invariants and breaks the third invariant. We have developed a radial diffusion code which uses the Crank Nicholson method with a variable outer boundary condition. For the radial diffusion coefficient, DLL, we have several choices, including the Brautigam and Albert (JGR, 2000) diffusion coefficient parameterized by Kp, which provides an ad hoc measure of the power level at ULF wave frequencies in the range of electron drift (mHz), breaking the third invariant. Other diffusion coefficient models are Kp-independent, fixed in time but explicitly dependent on the first invariant, or energy at a fixed L, such as calculated by Elkington et al. (JGR, 2003) and Perry et al. (JGR, 2006) based on ULF wave model fields. We analyzed three periods of electron flux and phase space density (PSD) enhancements inside of geosynchronous orbit: March 31 - May 31, 1991, and July 2004 and Nov 2004 storm intervals. The radial diffusion calculation is initialized with a computed phase space density profile for the 1991 interval using differential flux values from the CRRES High Energy Electron Fluxmeter instrument, covering 0.65 - 7.5 MeV. To calculate the initial phase space density, we convert Roederer L* to McIlwain's L- parameter using the ONERA-DESP program. A time averaged model developed by Vampola1 from the entire 14 month CRRES data set is applied to the July 2004 and Nov 2004 storms. The online CRESS data for specific orbits and the Vampola-model flux are both expressed in McIlwain L-shell, while conversion to L* conserves phase space density in a distorted non-dipolar magnetic field model. A Tsyganenko (T04) magnetic field model is used for conversion between L* and L. The outer boundary PSD is updated using LANL GEO satellite fluxes
Reynard-Carette, C.; Carette, M.; Aguir, K.; Bendahan, M.; Fiorido, T. [Aix Marseille Universite, CNRS, Universite de Toulon, IM2NP UMR 7334, 13397, Marseille (France); Lyoussi, A.; Fourmentel, D.; Villard, J.F. [CEA, DEN, DER, Instrumentation Sensors and Dosimetry Laboratory, Cadarache, F-13108 (France); Barthes, M.; Lanzetta, F.; Layes, G.; Vives, S. [FEMTO-ST, UMR 6174, Departement ENERGIE, Universite de Franche-Comte, 90000, Belfort (France)
source by the Joule Effect inside each calorimetric cell. The second one is a zero method consisting in cancelling the difference in cell responses with an additional energy into the reference cell. The last measurement method is based on current additions in the two calorimetric cells. However, one drawback of the existing differential calorimeter is the size of the sensor: a great length equal to 220 mm and a diameter equal to 18 mm. This current size leads to measurement limitations. This paper will begin with a presentation of these measurement limitations from a bibliographic state. Each limitation will be detailed and in particular in the case of a high nuclear heating level expected, for instance, inside the JHR's core at its highest nominal power. The second part of the paper will develop the scientific skills of each partner in heat sciences, micro technology and nuclear physics necessary to design a new calorimetric micro-system: the advantages of studied microelements such as micro-thermocouples, micro- fluxmeters and micro-heaters will be presented. The last part will discuss preliminary designs. (authors)
Mazzini, Adriano; Lusi Lab Team
inaccessible environments. The Lusi drone allowed us to successfully complete video/photo surveys as well as fluids/mud sampling from the crater including spot measurements. In order to estimate the amount of gas that is being released around the Lusi crater area (~7 km2), we conducted two surveys including over 350 stations (CO2 and CH4 flux measurements) using a closed-chamber flux-meter system, measured radon emissions, and collected more than 60 gas samples to analyze the composition of the seeps and the crater plume. We also investigated microbial processes and thriving communities conducting several sampling campaigns to collect samples of fresh mud from the erupting crater using the remote controlled drone. In addition we completed a transect in the mud flood zone to sample older, weathered flows for comparison. The results of the microbial colonies incubation reveal the widespread presence of active microbial colonies, even at high temperatures, opening new questions regarding life in the deep biosphere. Since more than a year we have operating a network of 31 seismic stations distributed around the Arjuno-Welirang volcanic arc, along the Waukosek fault and around Lusi. The purpose of this long term monitoring is to observe how local seismicity and/or the frequent seismic activity ongoing in the subduction zone in southern Java affects the activity of the magma chamber, the Watukosek fault system and the Lusi activity. In addition we also deployed temporary stations inside the embankment area to observe the activity of the pulsating behavior of Lusi and its geysering bursts. This study is coupled with video observations. A comprehensive combined electrical resistivity and self-potential (SP) survey was performed in the 7 km2 area inside the Lusi embankment. The goal of the geophysical survey was to map the near-surface occurrence of the Watukosek fault system, and provide useful data for numerical modelling. The large amount of data collected allows us to test several