Design of an elliptically bent refocus mirror for the MERLIN beamline at the advanced light source
International Nuclear Information System (INIS)
Kelez, Nicholas; Chuang, Yi-De; Smith-Baumann, Alexis; Franck, Keith; Duarte, R.; Lanzara, A.; Hasan, M.Z.; Dessau, D.S.; Chiang, T.C.; Shen, Z.-X.; Hussain, Zahid
2007-01-01
There is an increasing demand for well-focused beam with high quality imaging of the full field to further explore the potential of novel instruments. For beamlines operating at focal point, mechanical benders have often been used to shape the refocusing mirror into an ideal elliptical cylinder. Unfortunately, the limited number of couplings for these mechanisms requires specific substrate side-shaping, often calculated using beam bending theory [J.H. Underwood, Space Sci. Instrum. 3 (1977) 259; M. Howells et al., Opt. Eng. 39 (2000) 2748; T. Warwick et al., Document available at OSTI website: http://www.osti.gov/energycitations/servlets/purl/842557-JYZCS3/native/], to meet demanding figure requirements. Here, we use finite element analysis (FEA) to validate the side-shaping algorithm and then couple the output with SHADOW to evaluate mirror performance when an ideal ellipse is not achieved
Garcia-Botella, Angel; Fernandez-Balbuena, Antonio Alvarez; Bernabeu, Eusebio
2006-10-10
Nonimaging optics is a field devoted to the design of optical components for applications such as solar concentration or illumination. In this field, many different techniques have been used to produce optical devices, including the use of reflective and refractive components or inverse engineering techniques. However, many of these optical components are based on translational symmetries, rotational symmetries, or free-form surfaces. We study a new family of nonimaging concentrators called elliptical concentrators. This new family of concentrators provides new capabilities and can have different configurations, either homofocal or nonhomofocal. Translational and rotational concentrators can be considered as particular cases of elliptical concentrators.
Secure network for beamline control
International Nuclear Information System (INIS)
Ohata, T.; Fukui, T.; Ishii, M.; Furukawa, Y.; Nakatani, T.; Matsushita, T.; Takeuchi, M.; Tanaka, R.; Ishikawa, T.
2001-01-01
In SPring-8, beamline control system is constructed with a highly available distributed network system. The socket based communication protocol is used for the beamline control mainly. Beamline users can control the equipment by sending simple control commands to a server process, which is running on a beamline-managing computer (Ohata et al., SPring-8 beamline control system, ICALEPCS'99, Trieste, Italy, 1999). At the beginning the network was based on the shared topology at all beamlines. Consequently, it has a risk for misapplication of the user's program to access different machines on the network system cross over beamlines. It is serious problem for the SPring-8 beamline control system, because all beamlines controlled with unified software interfaces. We introduced the switching technology and the firewalls to support network access control. Also the virtual networking (VLAN: IEEE 802.1Q) and the gigabit Ethernet technology (IEEE 802.3ab) are introduced. Thus the network security and the reliability are guaranteed at the higher level in SPring-8 beamline
Mirrors for synchrotron-radiation beamlines
International Nuclear Information System (INIS)
Howells, M.R.
1993-09-01
The authors consider the role of mirrors in synchrotron-radiation beamlines and discuss the optical considerations involved in their design. They discuss toroidal, spherical, elliptical, and paraboloidal mirrors in detail with particular attention to their aberration properties. They give a treatment of the sine condition and describe its role in correcting the coma of axisymmetric systems. They show in detail how coma is inevitable in single-reflection, grazing-incidence systems but correctable in two-reflection systems such as those of the Wolter type. In an appendix, they give the theory of point aberrations of reflectors of a general shape and discuss the question of correct naming of aberrations. In particular, a strict definition of coma is required if attempts at correction are to be based on the sine condition
APS beamline standard components handbook
International Nuclear Information System (INIS)
Kuzay, T.M.
1992-01-01
It is clear that most Advanced Photon Source (APS) Collaborative Access Team (CAT) members would like to concentrate on designing specialized equipment related to their scientific programs rather than on routine or standard beamline components. Thus, an effort is in progress at the APS to identify standard and modular components of APS beamlines. Identifying standard components is a nontrivial task because these components should support diverse beamline objectives. To assist with this effort, the APS has obtained advice and help from a Beamline Standardization and Modularization Committee consisting of experts in beamline design, construction, and operation. The staff of the Experimental Facilities Division identified various components thought to be standard items for beamlines, regardless of the specific scientific objective of a particular beamline. A generic beamline layout formed the basis for this identification. This layout is based on a double-crystal monochromator as the first optical element, with the possibility of other elements to follow. Pre-engineering designs were then made of the identified standard components. The Beamline Standardization and Modularization Committee has reviewed these designs and provided very useful input regarding the specifications of these components. We realize that there will be other configurations that may require special or modified components. This Handbook in its current version (1.1) contains descriptions, specifications, and pre-engineering design drawings of these standard components. In the future, the APS plans to add engineering drawings of identified standard beamline components. Use of standard components should result in major cost reductions for CATs in the areas of beamline design and construction
Ellipticities of Elliptical Galaxies in Different Environments
Chen, Cheng-Yu; Hwang, Chorng-Yuan; Ko, Chung-Ming
2016-10-01
We studied the ellipticity distributions of elliptical galaxies in different environments. From the ninth data release of the Sloan Digital Sky Survey, we selected galaxies with absolute {r}\\prime -band magnitudes between -21 and -22. We used the volume number densities of galaxies as the criterion for selecting the environments of the galaxies. Our samples were divided into three groups with different volume number densities. The ellipticity distributions of the elliptical galaxies differed considerably in these three groups of different density regions. We deprojected the observed 2D ellipticity distributions into intrinsic 3D shape distributions, and the result showed that the shapes of the elliptical galaxies were relatively spherically symmetric in the high density region (HDR) and that relatively more flat galaxies were present in the low density region (LDR). This suggests that the ellipticals in the HDRs and LDRs have different origins or that different mechanisms might be involved. The elliptical galaxies in the LDR are likely to have evolved from mergers in relatively anisotropic structures, such as filaments and webs, and might contain information on the anisotropic spatial distribution of their parent mergers. By contrast, elliptical galaxies in the HDR might be formed in more isotropic structures, such as galaxy clusters, or they might encounter more torqueing effects compared with galaxies in LDRs, thereby becoming rounder.
International Nuclear Information System (INIS)
Nannarone, S.; Pasquali, L.; Selvaggi, G.; Borgatti, F.; DeLuisa, A.; Doyle, B.P.; Gazzadi, G.C.; Giglia, A.; Finetti, P.; Pedio, M.; Mahne, N.; Naletto, G.; Pelizzo, M.G.; Tondello, G.
2004-01-01
The BEAR (Bending Magnet for Emission Absorption and Reflectivity) beamline is installed at the right exit of the 8.1 bending magnet at ELETTRA. The beamline - in operation since January 2003 - delivers linear and circularly polarized radiation in the 5 - 1600 eV energy range. The experimental station is composed of a UHV chamber for reflectivity, absorption, fluorescence and angle resolved photoemission measurements and a UHV chamber for in-situ sample preparation
Radiation protection of a proton beamline at ELI-Beamlines
Czech Academy of Sciences Publication Activity Database
Bechet, Sabrina; Versaci, Roberto; Rollet, S.; Olšovcová, Veronika; Fajstavr, Antonín; Žáková, Martina; Margarone, Daniele
2016-01-01
Roč. 11, č. 12 (2016), 1-5, č. článku C12019. ISSN 1748-0221 R&D Projects: GA MŠk EF15_008/0000162 Grant - others:ELI Beamlines(XE) CZ.02.1.01/0.0/0.0/15_008/0000162 Institutional support: RVO:68378271 Keywords : models * simulations * wake-field acceleration Subject RIV: BL - Plasma and Gas Discharge Physics Impact factor: 1.220, year: 2016
ALS beamline design requirements: A guide for beamline designers
Energy Technology Data Exchange (ETDEWEB)
NONE
1996-06-01
This manual is written as a guide for researchers in designing beamlines and endstations acceptable for use at the ALS. It contains guidelines and policies related to personnel safety and equipment and vacuum protection. All equipment and procedures must ultimately satisfy the safety requirements set aside in the Lawrence Berkeley National Laboratory (LBNL) Health and Safety Manual (PUB-3000) which is available from the ALS User Office or on the World WideWeb from the LBNL Homepage (http:// www.lbl.gov).
Energy Technology Data Exchange (ETDEWEB)
Mineev, Mark [Los Alamos National Laboratory
2008-01-01
The planar elliptic extension of the Laplacian growth is, after a proper parametrization, given in a form of a solution to the equation for areapreserving diffeomorphisms. The infinite set of conservation laws associated with such elliptic growth is interpreted in terms of potential theory, and the relations between two major forms of the elliptic growth are analyzed. The constants of integration for closed form solutions are identified as the singularities of the Schwarz function, which are located both inside and outside the moving contour. Well-posedness of the recovery of the elliptic operator governing the process from the continuum of interfaces parametrized by time is addressed and two examples of exact solutions of elliptic growth are presented.
TARA beamline and injection system
International Nuclear Information System (INIS)
Post, R.S.; Brindza, P.; Coleman, J.W.; Torti, R.P.; Blackfield, D.T.; Goodrich, P.
1983-01-01
The TARA beamline for neutral beam injection will permit one to three sources to fire into each plug (60 degree or optional 90 degree injection with respect to the TARA axis) or into each anchor (90 degree injection only). The sources, pre-aimed on their mounting plate at the NB test stand, may be fired into neutralizer ducts or optionally through a magnesium curtain, and the unneutralized fraction is dumped by the TARA fringing field onto a receiver plate. The beamline is housed in a cylindrical tank with the beam axis along the tank diameter at the midplane. The tank will be sorption pumped using LN + T/sub I/ or N/sub B/ and/or e-beam gettering. The beam burial tank contains sed arrays and a thin foil dump which reaches sufficiently high temperatures during the shot to boil out gas between shots
Optical design of an x-ray absorption spectroscopy beamline at Indus-2 synchrotron radiation source
International Nuclear Information System (INIS)
Das, N.C.; Jha, S.N.; Roy, A.P.
1999-10-01
Details of optical design of EXAFS beamline at Indus-2 SRS, under development at CAT, Indore, have been discussed in this report. This beamline will cover the photon energy range of 5 keV to 20 keV and will use a bent crystal of Si(111) having 2d value equal to 6.2709 A. It will accept a horizontal divergence of 1.5 mrad. The heart of the beamline is the bent crystal polychromator which will disperse and focus the synchrotron beam at the experimental sample position. The transmitted radiation from the sample will be, subsequently, detected by a position sensitive detector (CCD type). The detector length is 25 mm. Assuming a suitable value for the distance between the source and the crystal, we have computed several geometrical parameters of the beamline, such as, Bragg angle, crystal length, crystal radius, crystal to sample distance, sample to detector distance, etc. for three different photon energies, namely, 5 keV, 10 keV, and 20 keV. The band passes around these photon energies are 0.3 keV, 1 keV and 2 keV respectively. It has been found that computed geometrical parameters are well within acceptable limits. An extensive ray tracing work was done using the software program SHADOW to evaluate the imaging properties of the beamline. It was established that the image spot size at the sample position improved substantially when the crystal is changed from spherical cylinder shape to elliptic cylinder shape. From the ray intensity plots, the average resolution of the crystal bender was estimated to be 1 eV per channel. Finally based on the optical layout of the beamline, a schematic mechanical layout of the beamline has been prepared. (author)
SPring-8 beamline control system.
Ohata, T; Konishi, H; Kimura, H; Furukawa, Y; Tamasaku, K; Nakatani, T; Tanabe, T; Matsumoto, N; Ishii, M; Ishikawa, T
1998-05-01
The SPring-8 beamline control system is now taking part in the control of the insertion device (ID), front end, beam transportation channel and all interlock systems of the beamline: it will supply a highly standardized environment of apparatus control for collaborative researchers. In particular, ID operation is very important in a third-generation synchrotron light source facility. It is also very important to consider the security system because the ID is part of the storage ring and is therefore governed by the synchrotron ring control system. The progress of computer networking systems and the technology of security control require the development of a highly flexible control system. An interlock system that is independent of the control system has increased the reliability. For the beamline control system the so-called standard model concept has been adopted. VME-bus (VME) is used as the front-end control system and a UNIX workstation as the operator console. CPU boards of the VME-bus are RISC processor-based board computers operated by a LynxOS-based HP-RT real-time operating system. The workstation and the VME are linked to each other by a network, and form the distributed system. The HP 9000/700 series with HP-UX and the HP 9000/743rt series with HP-RT are used. All the controllable apparatus may be operated from any workstation.
Beamlines on the SPring-8 project
Energy Technology Data Exchange (ETDEWEB)
Ohno, Hideo [Japan Atomic Energy Research Inst., Kamigori, Hyogo (Japan). Kansai Research Establishment
1997-03-01
SPring-8 project is going to construct a ultrahigh-brilliance X-ray synchrotron radiation facility and commissioning of the storage ring is expected in Spring 1977. The facility will be available to scientists and engineers of universities, national laboratories and industries not only from Japan but also from abroad. 20 proposals for public beamlines are submitted to the Beamline Committee and the ten public beamlines are scheduled for completion by the end of 1997. (author)
Elliptic Determinantal Processes and Elliptic Dyson Models
Katori, Makoto
2017-10-01
We introduce seven families of stochastic systems of interacting particles in one-dimension corresponding to the seven families of irreducible reduced affine root systems. We prove that they are determinantal in the sense that all spatio-temporal correlation functions are given by determinants controlled by a single function called the spatio-temporal correlation kernel. For the four families {A}_{N-1}, {B}_N, {C}_N and {D}_N, we identify the systems of stochastic differential equations solved by these determinantal processes, which will be regarded as the elliptic extensions of the Dyson model. Here we use the notion of martingales in probability theory and the elliptic determinant evaluations of the Macdonald denominators of irreducible reduced affine root systems given by Rosengren and Schlosser.
Beamline 9.0.1 - a high-resolution undulator beamline for gas-phase spectroscopy
Energy Technology Data Exchange (ETDEWEB)
Bozek, J.D.; Heimann, P.A.; Mossessian, D. [Ernest Orlando Lawrence Berkeley National Lab., CA (United States)
1997-04-01
Beamline 9.0.1 at the Advanced Light Source is an undulator beamline with a Spherical Grating Monochromator (SGM) which provides very high resolution and flux over the photon energy range 20-320eV. The beamline has been used primarily by the atomic and molecular science community to conduct spectroscopy experiments using electron, ion and fluorescence photon detection. A description of the beamline and its performance will be provided in this abstract.
International Nuclear Information System (INIS)
Kawamoto, Masahide; Hasegawa, Kazuya; Shimizu, Nobutaka; Sakai, Hisanobu; Shimizu, Tetsuya; Nisawa, Atsushi; Yamamoto, Masaki
2007-01-01
SPring-8 has 2 protein crystallography beamlines for public use, BL38B1 (Structural Biology III) and BL41XU (Structural Biology I). The BL38B1 is a bending magnet beamline for routine data collection, and the BL41XU is an undulator beamline specially customized for micro beam and ultra-high resolutional experiment. The designs and the performances of each beamline are presented
An object model for beamline descriptions
International Nuclear Information System (INIS)
Hill, B.W.; Martono, H.; Gillespie, J.S.
1997-01-01
Translation of beamline model descriptions between different accelerator codes presents a unique challenge due to the different representations used for various elements and subsystems. These differences range from simple units conversions to more complex translations involving multiple beamline components. A representation of basic accelerator components is being developed in order to define a meta-structure from which beamline models, in different codes, can be described and to facilitate the translation of models between these codes. Sublines of basic components will be used to represent more complex beamline descriptions and bridge the gap between codes which may represent a beamline element as a single entity, and those which use multiple elements to describe the same physical device. A C++ object model for supporting this beamline description and a grammar for describing beamlines in terms of these components is being developed. The object model will support a common graphic user interface and translation filters for representing native beamline descriptions for a variety of accelerator codes. An overview of our work on the object model for beamline descriptions is presented here. copyright 1997 American Institute of Physics
Energy Technology Data Exchange (ETDEWEB)
Fedurin, M. [Brookhaven National Lab. (BNL), Upton, NY (United States); Jing, Y. [Brookhaven National Lab. (BNL), Upton, NY (United States); Stratakis, D. [Brookhaven National Lab. (BNL), Upton, NY (United States); Swinson, C. [Brookhaven National Lab. (BNL), Upton, NY (United States)
2015-05-03
The Brookhaven National Laboratory. Accelerator Test Facility (BNL ATF) is currently undergoing a major upgrade (ATF-II). Together with a new location and much improved facilities, the ATF will see an upgrade in its major capabilities: electron beam energy and quality and CO_{2} laser power. The electron beam energy will be increased in stages, first to 100-150 MeV followed by a further increase to 500 MeV. Combined with the planned increase in CO_{2} laser power (from 1-100 TW), the ATF-II will be a powerful tool for Advanced Accelerator research. A high-brightness electron beam, produced by a photocathode gun, will be accelerated and optionally delivered to multiple beamlines. Besides the energy range (up to a possible 500 MeV in the final stage) the electron beam can be tailored to each experiment with options such as: small transverse beam size (<10 um), short bunch length (<100 fsec) and, combined short and small bunch options. This report gives a detailed overview of the ATFII capabilities and beamlines configuration.
Diamond Beamline I16 (Materials and Magnetism)
International Nuclear Information System (INIS)
Collins, S. P.; Bombardi, A.; Marshall, A. R.; Williams, J. H.; Barlow, G.; Day, A. G.; Pearson, M. R.; Woolliscroft, R. J.; Walton, R. D.; Beutier, G.; Nisbet, G.
2010-01-01
We describe the key features and performance specifications of a facility for high-resolution single-crystal x-ray diffraction at Diamond Light Source. The scientific emphasis of the beamline is materials- and x-ray-physics, including resonant and magnetic scattering. We highlight some of the more novel aspects of the beamline design.
Beamline for Schools Safety Awareness Day
Photo Service, CERN
2014-01-01
The first two teams to participate in CERN's Beamline for Schools project spent their first day at CERN at the Safety Training Center in Prévessin. They covered amongst others radiation protection, cryogenics and fire-fighting. The teams will spend the rest of the week at the T9 beamline.
Radiation-resistant beamline components at LAMPF
International Nuclear Information System (INIS)
Macek, R.J.; Grisham, D.L.; Lambert, J.e.; Werbeck, R.
1983-01-01
A variety of highly radiation-resistant beamline components have been successfully developed at LAMPF primarily for use in the target cells and beam stop area of the intense proton beamline. Design features and operating experience are reviewed for magnets, instrumentation, targets, vacuum seals, vacuum windows, collimators, and beam stops
Triaxiality in elliptical galaxies
Energy Technology Data Exchange (ETDEWEB)
Benacchio, L; Galletta, G [Padua Univ. (Italy). Ist. di Astronomia
1980-12-01
The existence of a triaxial shape for elliptical galaxies has been considered in recent years to explain the new kinematical and geometrical findings, i.e. (a) the low rotation/velocity dispersion ratio found also in some flat systems, (b) the presence of twisting in the isophotes, (c) the recently found correlation between maximum twisting and maximum flattening, (d) the presence of rotation along the minor axis. A simple geometrical model of elliptical galaxies having shells with different axial ratios c/a, b/a has been produced to interpret three fundamental key-features of elliptical galaxies: (i) the distribution of the maximum flattening observed; (ii) the percentage of ellipticals showing twisting; and (iii) the correlation between maximum twisting and maximum flattening. The model has been compared with observational data for 348 elliptical systems as given by Strom and Strom. It is found that a triaxial ellipsoid with coaxial shells having axial ratios c/a and b/a mutually dependent in a linear way can satisfy the observations.
Limiting effects in double EEX beamline
Ha, G.; Power, J. G.; Conde, M.; Doran, D. S.; Gai, W.
2017-07-01
The double emittance exchange (EEX) beamline is suggested to overcome the large horizontal emittance and transverse jitter issues associated with the single EEX beamline while preserving its powerful phase-space manipulation capability. However, the double EEX beamline also has potential limitations due to coherent synchrotron radiation (CSR) and transverse jitter. The former limitation arises because double EEX uses twice as many bending magnets as single EEX which means stronger CSR effects degrading the beam quality. The latter limitation arises because a longitudinal jitter in front of the first EEX beamline is converted into a transverse jitter in the middle section (between the EEX beamlines) which can cause beam loss or beam degradation. In this paper, we numerically explore the effects of these two limitations on the emittance and beam transport.
Limiting effects in double EEX beamline
International Nuclear Information System (INIS)
Ha, G; Power, J G; Conde, M; Doran, D S; Gai, W
2017-01-01
The double emittance exchange (EEX) beamline is suggested to overcome the large horizontal emittance and transverse jitter issues associated with the single EEX beamline while preserving its powerful phase-space manipulation capability. However, the double EEX beamline also has potential limitations due to coherent synchrotron radiation (CSR) and transverse jitter. The former limitation arises because double EEX uses twice as many bending magnets as single EEX which means stronger CSR effects degrading the beam quality. The latter limitation arises because a longitudinal jitter in front of the first EEX beamline is converted into a transverse jitter in the middle section (between the EEX beamlines) which can cause beam loss or beam degradation. In this paper, we numerically explore the effects of these two limitations on the emittance and beam transport. (paper)
A Superbend X-Ray Microdiffraction Beamline at the Advanced Light Source
Energy Technology Data Exchange (ETDEWEB)
Tamura, N.; Kunz, M.; Chen, K.; Celestre, R.S.; MacDowell, A.A.; Warwick, T.
2009-03-10
Beamline 12.3.2 at the Advanced Light Source is a newly commissioned beamline dedicated to x-ray microdiffraction. It operates in both monochromatic and polychromatic radiation mode. The facility uses a superconducting bending magnet source to deliver an X-ray spectrum ranging from 5 to 22 keV. The beam is focused down to {approx} 1 um size at the sample position using a pair of elliptically bent Kirkpatrick-Baez mirrors enclosed in a vacuum box. The sample placed on high precision stages can be raster-scanned under the microbeam while a diffraction pattern is taken at each step. The arrays of diffraction patterns are then analyzed to derive distribution maps of phases, strain/stress and/or plastic deformation inside the sample.
Simulation of beamline alignment operations
International Nuclear Information System (INIS)
Annese, C; Miller, M G.
1999-01-01
The CORBA-based Simulator was a Laboratory Directed Research and Development (LDRD) project that applied simulation techniques to explore critical questions about distributed control systems. The simulator project used a three-prong approach that studied object-oriented distribution tools, computer network modeling, and simulation of key control system scenarios. The National Ignition Facility's (NIF) optical alignment system was modeled to study control system operations. The alignment of NIF's 192 beamlines is a large complex operation involving more than 100 computer systems and 8000 mechanized devices. The alignment process is defined by a detailed set of procedures; however, many of the steps are deterministic. The alignment steps for a poorly aligned component are similar to that of a nearly aligned component; however, additional operations/iterations are required to complete the process. Thus, the same alignment operations will require variable amounts of time to perform depending on the current alignment condition as well as other factors. Simulation of the alignment process is necessary to understand beamline alignment time requirements and how shared resources such as the Output Sensor and Target Alignment Sensor effect alignment efficiency. The simulation has provided alignment time estimates and other results based on documented alignment procedures and alignment experience gained in the laboratory. Computer communication time, mechanical hardware actuation times, image processing algorithm execution times, etc. have been experimentally determined and incorporated into the model. Previous analysis of alignment operations utilized average implementation times for all alignment operations. Resource sharing becomes rather simple to model when only average values are used. The time required to actually implement the many individual alignment operations will be quite dynamic. The simulation model estimates the time to complete an operation using
Superconducting elliptical cavities
Sekutowicz, J K
2011-01-01
We give a brief overview of the history, state of the art, and future for elliptical superconducting cavities. Principles of the cell shape optimization, criteria for multi-cell structures design, HOM damping schemes and other features are discussed along with examples of superconducting structures for various applications.
Microspectroscopy At Beamline 73 MAX-lab
International Nuclear Information System (INIS)
Engdahl, Anders
2010-01-01
Presentation of some projects at the infrared microspectroscopy experimental station at beamline 73 MAX-lab. Among the subjects are found identification of organic residues in fossil material and examination of the chemistry in an old oak wood wreck.
A Test Beamline on Diamond Light Source
International Nuclear Information System (INIS)
Sawhney, K. J. S.; Dolbnya, I. P.; Tiwari, M. K.; Alianelli, L.; Scott, S. M.; Preece, G. M.; Pedersen, U. K.; Walton, R. D.
2010-01-01
A Test beamline B16 has been built on the 3 GeV Diamond synchrotron radiation source. The beamline covers a wide photon energy range from 2 to 25 keV. The beamline is highly flexible and versatile in terms of the available beam size (a micron to 100 mm) and the range of energy resolution and photon flux; by virtue of its several operational modes, and the different inter-changeable instruments available in the experiments hutch. Diverse experimental configurations can be flexibly configured using a five-circle diffractometer, a versatile optics test bench, and a suite of detectors. Several experimental techniques including reflectivity, diffraction and imaging are routinely available. Details of the beamline and its measured performance are presented.
Doublet III beamline: as-built
International Nuclear Information System (INIS)
Harder, C.R.; Holland, M.M.; Parker, J.W.; Gunn, J.; Resnick, L.
1980-03-01
In order to fully exploit Doublet III capabilities and to study new plasma physics regimes, a Neutral Beam Injector System has been constructed. Initially, a two beamline system will supply 7 MW of heat to the plasma. The system is currently being expanded to inject approx. 20 MW of power (6 beamlines). Each beamline is equipped with two Lawrence Berkeley Laboratory type rectangular ion sources with 10 cm x 40 cm extraction grids. These sources will accelerate hydrogen ions to 80 keV, with extracted beam currents in excess of 80 A per source expected. The first completed source is currently being tested and conditioned on the High Voltage Test Stand at Lawrence Livermore Laboratory. This paper pictorially reviews the as-built Doublet III neutral beamline with emphasis on component relation and configuration relative to spatial and source imposed design constraints
Anisotropic elliptic optical fibers
Kang, Soon Ahm
1991-05-01
The exact characteristic equation for an anisotropic elliptic optical fiber is obtained for odd and even hybrid modes in terms of infinite determinants utilizing Mathieu and modified Mathieu functions. A simplified characteristic equation is obtained by applying the weakly guiding approximation such that the difference in the refractive indices of the core and the cladding is small. The simplified characteristic equation is used to compute the normalized guide wavelength for an elliptical fiber. When the anisotropic parameter is equal to unity, the results are compared with the previous research and they are in close agreement. For a fixed value normalized cross-section area or major axis, the normalized guide wavelength lambda/lambda(sub 0) for an anisotropic elliptic fiber is small for the larger value of anisotropy. This condition indicates that more energy is carried inside of the fiber. However, the geometry and anisotropy of the fiber have a smaller effect when the normalized cross-section area is very small or very large.
The properties of radio ellipticals
International Nuclear Information System (INIS)
Sparks, W.B.; Disney, M.J.; Rodgers, A.W.
1984-01-01
Optical and additional radio data are presented for the bright galaxies of the Disney and Wall survey (1977 Mon. Not. R. Astron. Soc. 179, 235). These data form the basis of a statistical comparison of the properties of radio elliptical galaxies to radio-quiet ellipticals. The correlations may be explained by the depth of the gravitational potential well in which the galaxy resides governing the circumstances under which an elliptical galaxy rids itself of internally produced gas. (author)
Baking controller for synchrotron beamline vacuum systems
International Nuclear Information System (INIS)
Garg, C.K.; Kane, S.R.; Dhamgaye, V.P.
2003-01-01
The 2.5 GeV electron storage ring Indus-2 is a hard X-ray Synchrotron Radiation (SR) Source. Nearly 27 beamlines will be installed on Indus-2 and they will cater to different experiments and applications. Most of the beamlines will be in Ultra High Vacuum (UHV) the only exception being hard X-rays beamlines. However the front ends of all the beamlines will be in UHV. Practicing UHV requires efforts and patience. Evacuating any chamber, volume gases can be removed easily. However, outgassing phenomena like desorption, diffusion and permeation restricts the system to attain UHV. All processes except the volume gas removal are temperature dependent. At ambient temperature, gas pressure decreases so slowly that outgassing limit (i.e. 10 -10 1/s/cm 2 ) can hardly be achieved on a practical time scale. Also there are three orders of magnitude difference in outgassing between baked and unbaked systems. Depending on the vacuum chamber and the components inside it, the thermal outgassing (baking) of system is required and can be done at various temperatures between 150 degC to 450 deg C. For whole baking cycle, constant monitoring and controlling of the systems is required which takes tens of hours. This paper describes the automation for such baking system, which will be used for SR beamlines
Modeling of X-ray beamlines and devices
International Nuclear Information System (INIS)
Ice, G.E.
1996-01-01
X-ray beamlines on synchrotron sources are similar in size and complexity to beamlines at state-of-the-art neutron sources. The design principles, tools, and optimization strategies for synchrotron beamlines are also similar to those of neutron beamlines. The authors describe existing design tools for modeling synchrotron radiation beamlines and describe how these tools have evolved over the last two decades. The development of increasingly powerful modeling tools has been driven by the escalating cost and sophistication of state-of-the-art beamlines and by a world-wide race to exploit advanced synchrotron radiation sources
Uniformization of elliptic curves
Ülkem, Özge; Ulkem, Ozge
2015-01-01
Every elliptic curve E defined over C is analytically isomorphic to C*=qZ for some q ∊ C*. Similarly, Tate has shown that if E is defined over a p-adic field K, then E is analytically isomorphic to K*=qZ for some q ∊ K . Further the isomorphism E(K) ≅ K*/qZ respects the action of the Galois group GK/K, where K is the algebraic closure of K. I will explain the construction of this isomorphism.
Elliptic partial differential equations
Han, Qing
2011-01-01
Elliptic Partial Differential Equations by Qing Han and FangHua Lin is one of the best textbooks I know. It is the perfect introduction to PDE. In 150 pages or so it covers an amazing amount of wonderful and extraordinary useful material. I have used it as a textbook at both graduate and undergraduate levels which is possible since it only requires very little background material yet it covers an enormous amount of material. In my opinion it is a must read for all interested in analysis and geometry, and for all of my own PhD students it is indeed just that. I cannot say enough good things abo
Simulation and optimization of the SIRIUS IPE soft x-ray beamline
Meyer, Bernd C.; Rocha, Tulio C. R.; Luiz, Sergio A. L.; C. Pinto, Artur; Westfahl, Harry
2017-08-01
The soft X-ray beamline IPE is one of the first phase SIRIUS beamlines at the LNLS, Brazil. Divided into two branches, IPE is designed to perform ambient pressure X-ray photo-electron spectroscopy (AP-XPS) and high resolution resonant inelastic X-ray scattering (RIXS) for samples in operando/environmental conditions inside cells and liquid jets. The aim is to maximize the photon flux in the energy range 200-1400 eV generated by an elliptically polarizing undulator source (EPU) and focus it to a 1 μm vertical spot size at the RIXS station and 10 μm at the AP-XPS station. In order to achieve the required resolving power (40.000 at 930 eV) for RIXS both the dispersion properties of the plane grating monochromator (PGM) and the thermal deformation of the optical elements need special attention. The grating parameters were optimized with the REFLEC code to maximize the efficiency at the required resolution. Thermal deformation of the PGM plane mirror limits the possible range of cff parameters depending of the photon energy used. Hence, resolution of the PGM and thermal deformation effects define the boundary conditions of the optical concept and the simulations of the IPE beamline. We compare simulations performed by geometrical ray-tracing (SHADOW) and wave front propagation (SRW) and show that wave front diffraction effects (apertures, optical surface error profiles) has a small effect on the beam spot size and shape.
G4Beamline Program for Radiation Simulations
International Nuclear Information System (INIS)
Beard, Kevin; Roberts, Thomas J.; Degtiarenko, Pavel
2008-01-01
G4beamline, a program that is an interface to the Geant4 toolkit that we have developed to simulate accelerator beamlines, is being extended with a graphical user interface to quickly and efficiently model experimental equipment and its shielding in experimental halls. The program is flexible, user friendly, and requires no programming by users, so that even complex systems can be simulated quickly. This improved user interface is of much wider application than just the shielding simulations that are the focus of this project. As an initial application, G4beamline is being extended to provide the simulations that are needed to determine the radiation sources for the proposed experiments at Jefferson Laboratory so that shielding issues can be evaluated. Since the program already has the capabilities needed to simulate the transport of all known particles, including scattering, attenuation, interactions, and decays, the extension involves implementing a user-friendly graphical user inter
Fast computation of complete elliptic integrals and Jacobian elliptic functions
Fukushima, Toshio
2009-12-01
As a preparation step to compute Jacobian elliptic functions efficiently, we created a fast method to calculate the complete elliptic integral of the first and second kinds, K( m) and E( m), for the standard domain of the elliptic parameter, 0 procedure to compute simultaneously three Jacobian elliptic functions, sn( u| m), cn( u| m), and dn( u| m), by repeated usage of the double argument formulae starting from the Maclaurin series expansions with respect to the elliptic argument, u, after its domain is reduced to the standard range, 0 ≤ u procedure is 25-70% faster than the methods based on the Gauss transformation such as Bulirsch’s algorithm, sncndn, quoted in the Numerical Recipes even if the acceleration of computation of K( m) is not taken into account.
Elliptic curves for applications (Tutorial)
Lange, T.; Bernstein, D.J.; Chatterjee, S.
2011-01-01
More than 25 years ago, elliptic curves over finite fields were suggested as a group in which the Discrete Logarithm Problem (DLP) can be hard. Since then many researchers have scrutinized the security of the DLP on elliptic curves with the result that for suitably chosen curves only exponential
National synchrotron light source user's manual: Guide to the VUV and x-ray beamlines: Third edition
International Nuclear Information System (INIS)
Gmuer, N.F.; Thomlinson, W.; White-DePace, S.
1989-01-01
This report contains information on the following topics: A Word on the Writing of Beamline Descriptions; Beamline Equipment Utilization for General Users; the Vacuum Ultraviolet (VUV) Storage Ring and Beamlines; VUV Beamline Descriptions--An Explanation; VUV Beamline Descriptions; X-Ray Storage Ring and Beamlines; X-Ray Beamline Descriptions--An Explanation; and X-Ray Beamline Descriptions
Status of the ELIMED multidisciplinary and medical beam-line at ELI-Beamlines
International Nuclear Information System (INIS)
Romano, F; Cirrone, G A P; Cuttone, G; Schillaci, F; Scuderi, V; Amico, A; Candiano, G; Larosa, G; Leanza, R; Manna, R; Marchese, V; Milluzzo, G; Petringa, G; Pipek, J; Giordanengo, S; Guarachi, L F; Marchetto, F; Sacchi, R; Korn, G; Margarone, D
2017-01-01
Nowadays, one of the biggest challenges consists in using high intensity laser-target interaction to generate high-energy ions for medical purposes, eventually replacing the old paradigm of acceleration characterized by huge and complex machines. In order to investigate the feasibility of using laser-driven ion beams for multidisciplinary application, a dedicated beam transport line will be installed at the ELI-Beamlines facility in Prague (CZ), as a part of the User-oriented ELIMAIA beam-line dedicated to ion acceleration and their potential applications. The beam-line section dedicated to transport and dosimetric endpoints is called ELIMED (ELI-Beamlines MEDical and multidisciplinary applications) and will be developed by the INFN-LNS. (paper)
EPICS and its role in data acquisition and beamline control
International Nuclear Information System (INIS)
Mooney, T. M.; Arnold, N. D.; Boucher, E.; Cha, B. K.; Goetze, K. A.; Kraimer, M. R.; Rivers, M. L.; Sluiter, R. L.; Sullivan, J. P.; Wallis, D. B.
1999-01-01
Beamline-control and data-acquisition software based on EPICS (a tool kit for building distributed control systems) has been running on many Advanced Photon Source beamlines for several years. EPICS itself, the collaborative software-development effort surrounding it, and EPICS-based beamline software have been described previously in general terms. This talk will review and update that material, focusing on the role EPICS core software plays in beamline applications and on the effects of a few defining characteristics of EPICS on the beamline software we have developed with it
Human factors design for the BMIT biomedical beamlines
International Nuclear Information System (INIS)
Miller, C Denise; Wysokinski, Tomasz W; Belev, George; Chapman, L Dean
2013-01-01
Operation of a biomedical beamline poses a unique set of operational and instrumentation challenges for a synchrotron facility. From proper handling and care of live animals and animal tissues, to a user community drawn primarily from the medical and veterinary realms, the work of a biomedical beamline is unique when compared to other beamlines. At the Biomedical Imaging and Therapy (BMIT) beamlines at Canadian Light Source (CLS), operation of the beamlines is geared towards our user community of medical personnel, in addition to basic science researchers. Human factors considerations have been incorporated wherever possible on BMIT, including in the design of software and hardware, as well as ease-of-use features of beamline control stations and experiment hutches. Feedback from users continues to drive usability improvements to beamline operations.
Wagner, U. H.; Pešić, Z. D.; De Fanis, A.; Rau, C.
2013-03-01
I13L is a 250 m long hard x-ray beamline (6 keV to 35 keV) at the Diamond Light Source. The beamline comprises of two independent experimental endstations: one for imaging in direct space using x-ray microscopy and one for imaging in reciprocal space using coherent diffraction based imaging techniques. In this paper we will discuss the fundamental design concepts of the beamline and explain their implications for the civil engineering of the endstation building and the beamline instrumentation. For the latter this paper will focus on the beamline mirror systems and monochromators.
International Nuclear Information System (INIS)
Wagner, U H; Pešić, Z D; Fanis, A De; Rau, C
2013-01-01
I13L is a 250 m long hard x-ray beamline (6 keV to 35 keV) at the Diamond Light Source. The beamline comprises of two independent experimental endstations: one for imaging in direct space using x-ray microscopy and one for imaging in reciprocal space using coherent diffraction based imaging techniques. In this paper we will discuss the fundamental design concepts of the beamline and explain their implications for the civil engineering of the endstation building and the beamline instrumentation. For the latter this paper will focus on the beamline mirror systems and monochromators.
Long, elliptically bent, active X-ray mirrors with slope errors <200 nrad.
Nistea, Ioana T; Alcock, Simon G; Kristiansen, Paw; Young, Adam
2017-05-01
Actively bent X-ray mirrors are important components of many synchrotron and X-ray free-electron laser beamlines. A high-quality optical surface and good bending performance are essential to ensure that the X-ray beam is accurately focused. Two elliptically bent X-ray mirror systems from FMB Oxford were characterized in the optical metrology laboratory at Diamond Light Source. A comparison of Diamond-NOM slope profilometry and finite-element analysis is presented to investigate how the 900 mm-long mirrors sag under gravity, and how this deformation can be adequately compensated using a single, spring-loaded compensator. It is shown that two independent mechanical actuators can accurately bend the trapezoidal substrates to a range of elliptical profiles. State-of-the-art residual slope errors of <200 nrad r.m.s. are achieved over the entire elliptical bending range. High levels of bending repeatability (ΔR/R = 0.085% and 0.156% r.m.s. for the two bending directions) and stability over 24 h (ΔR/R = 0.07% r.m.s.) provide reliable beamline performance.
1993 CAT workshop on beamline optical designs
International Nuclear Information System (INIS)
1993-11-01
An Advanced Photon Source (APS) Collaborative Access Team (CAT) Workshop on Beamline Optical Designs was held at Argonne National Laboratory on July 26--27, 1993. The goal of this workshop was to bring together experts from various synchrotron sources to provide status reports on crystal, reflecting, and polarizing optics as a baseline for discussions of issues facing optical designers for CAT beamlines at the APS. Speakers from the European Synchrotron Radiation Facility (ESRF), the University of Chicago, the National Synchrotron Light Source, and the University of Manchester (England) described single- and double-crystal monochromators, mirrors, glass capillaries, and polarizing optics. Following these presentations, the 90 participants divided into three working groups: Crystal Optics Design, Reflecting Optics, and Optics for Polarization Studies. This volume contains copies of the presentation materials from all speakers, summaries of the three working groups, and a ''catalog'' of various monochromator designs
1993 CAT workshop on beamline optical designs
Energy Technology Data Exchange (ETDEWEB)
1993-11-01
An Advanced Photon Source (APS) Collaborative Access Team (CAT) Workshop on Beamline Optical Designs was held at Argonne National Laboratory on July 26--27, 1993. The goal of this workshop was to bring together experts from various synchrotron sources to provide status reports on crystal, reflecting, and polarizing optics as a baseline for discussions of issues facing optical designers for CAT beamlines at the APS. Speakers from the European Synchrotron Radiation Facility (ESRF), the University of Chicago, the National Synchrotron Light Source, and the University of Manchester (England) described single- and double-crystal monochromators, mirrors, glass capillaries, and polarizing optics. Following these presentations, the 90 participants divided into three working groups: Crystal Optics Design, Reflecting Optics, and Optics for Polarization Studies. This volume contains copies of the presentation materials from all speakers, summaries of the three working groups, and a ``catalog`` of various monochromator designs.
Thoresen, Freja
2015-01-01
Detailed simulations of the H8 beam line at the North Area, using the G4beamline software were performed in the framework of this study. The conventions used by the program are analysed. Having modelled precisely the beam line, several studies examining the beam transmission and composition were performed. The results were compared with measurements, where a satisfactory agreement was found. The muon production and transport is studied in details throughout the beam line.
How good can our beamlines be?
Energy Technology Data Exchange (ETDEWEB)
Liebschner, Dorothee; Dauter, Miroslawa; Rosenbaum, Gerold, E-mail: rosenbaum@anl.gov; Dauter, Zbigniew, E-mail: rosenbaum@anl.gov [Argonne National Laboratory, Argonne, IL 60439 (United States)
2012-10-01
A repetitive measurement of the same diffraction image allows to judge the performance of a data collection facility. The accuracy of X-ray diffraction data depends on the properties of the crystalline sample and on the performance of the data-collection facility (synchrotron beamline elements, goniostat, detector etc.). However, it is difficult to evaluate the level of performance of the experimental setup from the quality of data sets collected in rotation mode, as various crystal properties such as mosaicity, non-uniformity and radiation damage affect the measured intensities. A multiple-image experiment, in which several analogous diffraction frames are recorded consecutively at the same crystal orientation, allows minimization of the influence of the sample properties. A series of 100 diffraction images of a thaumatin crystal were measured on the SBC beamline 19BM at the APS (Argonne National Laboratory). The obtained data were analyzed in the context of the performance of the data-collection facility. An objective way to estimate the uncertainties of individual reflections was achieved by analyzing the behavior of reflection intensities in the series of analogous diffraction images. The multiple-image experiment is found to be a simple and adequate method to decompose the random errors from the systematic errors in the data, which helps in judging the performance of a data-collection facility. In particular, displaying the intensity as a function of the frame number allows evaluation of the stability of the beam, the beamline elements and the detector with minimal influence of the crystal properties. Such an experiment permits evaluation of the highest possible data quality potentially achievable at the particular beamline.
Muon Colliders: the Ultimate Neutrino Beamlines
International Nuclear Information System (INIS)
King, Bruce J.
1999-01-01
It is shown that muon decays in straight sections of muon collider rings will naturally produce highly collimated neutrino beams that can be several orders of magnitude stronger than the beams at existing accelerators. We discuss possible experimental setups and give a very brief overview of the physics potential from such beamlines. Formulae are given for the neutrino event rates at both short and long baseline neutrino experiments in these beams
Nomenclature of SLC Arc beamline components
International Nuclear Information System (INIS)
Silva, J.; Weng, W.T.
1986-01-01
This note defines I and C formal names for beamline components in the Arc as specified in the TRANSPORT decks ARCN FINAL and ARCS FINAL of June 5, 1985. The formal name consists of three fields: the primary name, the zone and the unit number. The general principles and guidelines are explained in Reference 1. The rationale and the final resolutions of the naming conventions for the Arc are explained
Upgrade of the Proton West secondary beamline
International Nuclear Information System (INIS)
Spiegel, L.
1989-01-01
As originally designed and operated, protons entering PW6 were steered by a series of EPB dipoles into a single interaction length beryllium target, some 43 feet from the enclosure wall. Ensuing secondary beams, either p + /π + or p - /π - , were collected by a string of quadrupoles following the target, steered westward, away from the Proton Center line, through PW6 and PW7, and ultimately focussed on experiment production targets located within the large PW8 hall. Around the Spring of 1988 it was decided to upgrade the existing Proton West secondary beamline to allow for transport of a primary proton beam, anticipated to be either 800 or 900 GeV/c, through PW8. This upgrade project, which is now nearing completion, was largely motivated by the then recent approval of E-771, a hadronic beauty production experiment located in PW8. E-771 represents the third in a series of experiments for the large-acceptance dimuon spectrometer presently located at the end of the Proton West beamline. This Technical Memo is a summary of the upgrade --- an explanation of the underlying strategy and a documentation of the final locations of the secondary beamline elements. 6 refs., 2 figs., 2 tabs
Nuclear resonant scattering beamline at SPring-8
Energy Technology Data Exchange (ETDEWEB)
Harami, Taikan [Japan Atomic Energy Research Inst., Kamigori, Hyogo (Japan). Kansai Research Establishment
1996-04-01
Mainly by Japan Atomic Energy Research Institute, the Institute of Physical and Chemical Research and Japan Synchrotron Radiation Research Institute, the construction of the Super Photon ring-8 GeV (SPring-8) which is the large scale synchrotron radiation facility for a high luminance light source placing emphasis on short wavelength region (shorter than about 1 nm) is in progress at the Harima Science Park City, Hyogo Prefecture. The features of the SPring-8 are the high luminance of light, the good parallelism and directionality of light, the quasi-monochromatic light with variable wavelength, and the possibility of design from straight polarization to circular polarization. The injection system that accelerates electrons up to 8 GeV and the storage ring storing the 8 GeV electrons for long hours, and 61 beamlines are explained. The manufacture of the nuclear resonant scattering beamline as the beamline for joint utilization was begun. Its transport channel and the experiment hutch are shown. By the features of undulator synchrotron radiation, the research on the matters with small recoilless fraction such as biological substances, liquid, gas and others and the research on time-dependent phenomena become feasible anew. The research on the dynamic structural analysis of heme protein is planned. (K.I.)
Beamline instrumentation and experiment automation for the Rossendorf Beamline at ESRF/Grenoble (F)
International Nuclear Information System (INIS)
Oehme, W.; Dienel, S.; Proehl, D.
2000-03-01
The Rossendorf research center finished the work (from 1996 - 1998) for building its own beamline for experiments with synchrotron radiation at the ESRF (European Synchrotron Radiation Facility) at Grenoble/France. The beamline has two measuring systems for alternative use for experiments investigating radioactive samples by means of X-ray absorption spectrometry, and materials structures by means of X-ray diffraction analyses. The report in hand focuses on work performed required for the control equipment for optical systems and performance of the measuring systems with respect to electronics, calculation methods, and software. Hardware components of importance to system instrumentation are explained, as well as the basic computing tasks and general software principles, and are illustrated with some examples. The general setting of the beamline is represented in a tabulated overview. The literature list serves as a guideline for detailed documentation. (orig./CB) [de
Status of Indus-1 and Indus-2 beamlines
Nandedkar, R V
2003-01-01
Indus-1 and Indus-2 are two synchrotron radiation sources that are planned in India. Indus-1 is a 450 MeV electron storage ring for vacuum ultra-violet soft X-ray radiation. This source is operational. Two beamlines, viz. a soft X-ray/vacuum ultra-violet reflectometry beamline and an angle integrated photoelectron spectroscopy beamline are already operational. Angle resolved photoelectron spectroscopy and photophysics beamlines are going to be operational soon. The second Indian synchrotron source is the 2.5 GeV Indus-2 electron storage ring that is under construction and is expected to be ready for commissioning in the year 2003. Of the total 27 beamlines possible on this ring, about 10 beamlines are already planned and are in the design stage.
Status of Indus-1 and Indus-2 beamlines
International Nuclear Information System (INIS)
Nandedkar, R.V.; Sawhney, K.J.S.
2003-01-01
Indus-1 and Indus-2 are two synchrotron radiation sources that are planned in India. Indus-1 is a 450 MeV electron storage ring for vacuum ultra-violet soft X-ray radiation. This source is operational. Two beamlines, viz. a soft X-ray/vacuum ultra-violet reflectometry beamline and an angle integrated photoelectron spectroscopy beamline are already operational. Angle resolved photoelectron spectroscopy and photophysics beamlines are going to be operational soon. The second Indian synchrotron source is the 2.5 GeV Indus-2 electron storage ring that is under construction and is expected to be ready for commissioning in the year 2003. Of the total 27 beamlines possible on this ring, about 10 beamlines are already planned and are in the design stage
Rotary Valve & Beamline Highlights for Fiscal Year 2017
Energy Technology Data Exchange (ETDEWEB)
Fitsos, P [Lawrence Livermore National Lab. (LLNL), Livermore, CA (United States)
2017-09-21
This Fiscal Year (FY) work was divided between continued testing and characterization work of the Rotary Valve (RV) and mechanical engineering support for the beamline hardware stands. This configuration is more like the final setup with the accelerator firing deuterons down the evacuated beamline toward the RV for interaction with the deuterium and neutron production. The beamline cells were part of an experiment to reduce the impact that RV gas would have on the beamline vacuum. This work will be reported separately from this report. Previous testing had been with the beamline at atmospheric pressure and now the goal was to get test results of the RV with it connected to a beamline that’s running at some level of vacuum.
Computer control of the ISX-B neutral injection beamlines
International Nuclear Information System (INIS)
Hanna, P.C.
1982-09-01
A system of controls for the Impurity Study Experiment (ISX-B) neutral injection beamlines at the Oak Ridge National Laboratory is presented. The system uses standard CAMAC equipment interfaced to the actual beamline controls and driven by a PDP-11/34 mini-computer. It is designed to relieve the operator of most of the mundane tasks of beam injection and also to reduce the number of operators needed to monitor multiple beamlines
Rational points on elliptic curves
Silverman, Joseph H
2015-01-01
The theory of elliptic curves involves a pleasing blend of algebra, geometry, analysis, and number theory. This book stresses this interplay as it develops the basic theory, thereby providing an opportunity for advanced undergraduates to appreciate the unity of modern mathematics. At the same time, every effort has been made to use only methods and results commonly included in the undergraduate curriculum. This accessibility, the informal writing style, and a wealth of exercises make Rational Points on Elliptic Curves an ideal introduction for students at all levels who are interested in learning about Diophantine equations and arithmetic geometry. Most concretely, an elliptic curve is the set of zeroes of a cubic polynomial in two variables. If the polynomial has rational coefficients, then one can ask for a description of those zeroes whose coordinates are either integers or rational numbers. It is this number theoretic question that is the main subject of this book. Topics covered include the geometry and ...
Energy and the Elliptical Orbit
Nettles, Bill
2009-03-01
In the January 2007 issue of The Physics Teacher, Prentis, Fulton, Hesse, and Mazzino describe a laboratory exercise in which students use a geometrical analysis inspired by Newton to show that an elliptical orbit and an inverse-square law force go hand in hand. The historical, geometrical, and teamwork aspects of the exercise are useful and important. This paper presents an exercise which uses an energy/angular momentum conservation model for elliptical orbits. This exercise can be done easily by an individual student and on regular notebook-sized paper.
Interstellar matter within elliptical galaxies
Jura, Michael
1988-01-01
Multiwavelength observations of elliptical galaxies are reviewed, with an emphasis on their implications for theoretical models proposed to explain the origin and evolution of the interstellar matter. Particular attention is given to interstellar matter at T less than 100 K (atomic and molecular gas and dust), gas at T = about 10,000 K, and gas at T = 10 to the 6th K or greater. The data are shown to confirm the occurrence of mass loss from evolved stars, significant accretion from companion galaxies, and cooling inflows; no evidence is found for large mass outflow from elliptical galaxies.
Hydrodynamic simulation of elliptic flow
Kolb, P F; Ruuskanen, P V; Heinz, Ulrich W
1999-01-01
We use a hydrodynamic model to study the space-time evolution transverse to the beam direction in ultrarelativistic heavy-ion collisions with nonzero impact parameters. We focus on the influence of early pressure on the development of radial and elliptic flow. We show that at high energies elliptic flow is generated only during the initial stages of the expansion while radial flow continues to grow until freeze-out. Quantitative comparisons with SPS data from semiperipheral Pb+Pb collisions suggest the applicability of hydrodynamical concepts already $\\approx$ 1 fm/c after impact.
Functional description of APS beamline front ends
International Nuclear Information System (INIS)
Kuzay, T.
1993-02-01
Traditional synchrotron sources were designed to produce bending magnet radiation and have proven to be an essential scientific tool. Currently, a new generation of synchrotron sources is being built that will be able to accommodate a large number of insertion device (ID) and high quality bending magnet (BM) sources. One example is the 7-GeV Advanced Photon Source (APS) now under construction at Argonne National Laboratory. The research and development effort at the APS is designed to fully develop the potential of this new generation of synchrotron sources. Of the 40 straight sections in the APS storage ring, 34 will be available for IDs. The remaining six sections are reserved for the storage ring hardware and diagnostics. Although the ring incorporates 80 BMs, only 40 of them can be used to extract radiation. The accelerator hardware shadows five of these 40 bending magnets, so the maximum number of BM sources on the lattice is 35. Generally, a photon beamline consists of four functional sections. The first section is the ID or the BM, which provides the radiation source. The second section, which is immediately outside the storage ring but inside a concrete shielding tunnel, is the front end, which is designed to control, define, and/or confine the x-ray beam. In the case of the APS, the front ends are designed to confine the photon beam. The third section, just outside the concrete shielding tunnel and on the experimental floor, is the first optics enclosure, which contains optics to filter and monochromatize the photon beam. The fourth section of a beamline consists of beam transports, additional optics, and experiment stations to do the scientific investigations. This document describes only the front ends of the APS beamlines
Elliptic net and its cryptographic application
Muslim, Norliana; Said, Mohamad Rushdan Md
2017-11-01
Elliptic net is a generalization of elliptic divisibility sequence and in cryptography field, most cryptographic pairings that are based on elliptic curve such as Tate pairing can be improved by applying elliptic nets algorithm. The elliptic net is constructed by using n dimensional array of values in rational number satisfying nonlinear recurrence relations that arise from elliptic divisibility sequences. The two main properties hold in the recurrence relations are for all positive integers m>n, hm +nhm -n=hm +1hm -1hn2-hn +1hn -1hm2 and hn divides hm whenever n divides m. In this research, we discuss elliptic divisibility sequence associated with elliptic nets based on cryptographic perspective and its possible research direction.
Coercive properties of elliptic-parabolic operator
International Nuclear Information System (INIS)
Duong Min Duc.
1987-06-01
Using a generalized Poincare inequality, we study the coercive properties of a class of elliptic-parabolic partial differential equations, which contains many degenerate elliptic equations considered by the other authors. (author). 16 refs
Newton flows for elliptic functions
Helminck, G.F.; Twilt, F.
2015-01-01
Newton flows are dynamical systems generated by a continuous, desingularized Newton method for mappings from a Euclidean space to itself. We focus on the special case of meromorphic functions on the complex plane. Inspired by the analogy between the rational (complex) and the elliptic (i.e., doubly
Diffeomorphisms of elliptic 3-manifolds
Hong, Sungbok; McCullough, Darryl; Rubinstein, J Hyam
2012-01-01
This work concerns the diffeomorphism groups of 3-manifolds, in particular of elliptic 3-manifolds. These are the closed 3-manifolds that admit a Riemannian metric of constant positive curvature, now known to be exactly the closed 3-manifolds that have a finite fundamental group. The (Generalized) Smale Conjecture asserts that for any elliptic 3-manifold M, the inclusion from the isometry group of M to its diffeomorphism group is a homotopy equivalence. The original Smale Conjecture, for the 3-sphere, was proven by J. Cerf and A. Hatcher, and N. Ivanov proved the generalized conjecture for many of the elliptic 3-manifolds that contain a geometrically incompressible Klein bottle. The main results establish the Smale Conjecture for all elliptic 3-manifolds containing geometrically incompressible Klein bottles, and for all lens spaces L(m,q) with m at least 3. Additional results imply that for a Haken Seifert-fibered 3 manifold V, the space of Seifert fiberings has contractible components, and apart from a small...
Elliptic curves and primality proving
Atkin, A. O. L.; Morain, F.
1993-07-01
The aim of this paper is to describe the theory and implementation of the Elliptic Curve Primality Proving algorithm. Problema, numeros primos a compositis dignoscendi, hosque in factores suos primos resolvendi, ad gravissima ac utilissima totius arithmeticae pertinere, et geometrarum tum veterum tum recentiorum industriam ac sagacitatem occupavisse, tam notum est, ut de hac re copiose loqui superfluum foret.
Color gradients in elliptical galaxies
International Nuclear Information System (INIS)
Franx, M.; Illingworth, G.
1990-01-01
The relationship of the color gradients within ellipticals and the color differences between them are studied. It is found that the local color appears to be strongly related to the escape velocity. This suggests that the local escape velocity is the primary factor that determines the metallicity of the stellar population. Models with and without dark halos give comparable results. 27 refs
Second order degenerate elliptic equations
International Nuclear Information System (INIS)
Duong Minh Duc.
1988-08-01
Using an improved Sobolev inequality we study a class of elliptic operators which is degenerate inside the domain and strongly degenerate near the boundary of the domain. Our results are applicable to the L 2 -boundary value problem and the mixed boundary problem. (author). 18 refs
A Remote and Virtual Synchrotron Beamline
Jackson, J. M.; Alp, E.; Sturhahn, W.
2012-12-01
National facilities offer one-of-a-kind opportunities to apply state-of-the-art experimental techniques to the pressing scientific problems of today. Yet, few students are able to experience research projects at national facilities due to limited accessibility caused in part by limited involvement in the local academic institution, constrained working areas at the experimental stations, and/or travel costs. We present a virtual and remote beam-line for Earth science studies using nuclear resonant and inelastic x-ray scattering methods at Sector 3 of the Advanced Photon Source at Argonne National Laboratory. Off-site students have the capability of controlling their measurements via secure internet connections and webcams. Students can access a 'view only mode' for ease of interaction and safety-control. More experienced users have exclusive control of the experiment and can remotely change variables within the experimental setup. Students may also access the virtual aspects these experiments by simulating certain conditions with our newly developed software. We evaluate such a tool by giving "before" and "after" assignments to students at different levels. These levels include high-school students from the Pasadena and greater Los Angeles area school districts, undergraduate students from Caltech's SURF/MURF program, and graduate students at Caltech. We specifically target underrepresented groups. Our results thus far show that the capabilities offered by our remote and virtual beamline show improved knowledge and understanding of applying experimental-based studies at the synchrotron to solve problems in the Earth sciences.
Diagnostics Beamline for the SRF Gun Project
Kamps, T; Goldammer, K; Krämer, Dietrich; Kuske, P; Kuszynski, J; Lipka, D; Marhauser, F; Quast, T; Richter, R
2005-01-01
A superconducting rf photo electron injector (SRF gun) is currently under construction by a collaboration between BESSY, DESY, FZR and MBI. The project aims at the design and setup of an CW SRF gun including a diagnostics beamline for the ELBE FEL and to address R&D issues on low emittance injectors for future light sources such as the BESSY FEL. Of critical importance for the injector performance is the control of the electron beam parameters. For this reason a compact diagnostics beamline is under development serving a multitude of operation settings ranging from low-charge (77pC), low-emittance (1 pi mm mrad) mode to high-charge (2.5nC) operation of the gun. For these operation modes beam dynamics simulations are resulting in boundary conditions for the beam instrumentation. Proven and mature technology is projected wherever possible, for example for current and beam position monitoring. The layout of the beam profile and emittance measurement systems is described. For the bunch length, which varies be...
Elliptical shape of the coma cluster
International Nuclear Information System (INIS)
Schipper, L.; King, I.R.
1978-01-01
The elliptical shape of the Coma cluster is examined quantitatively. The degree of ellipticity is high and depends to some extent on the radial distance of the sample from the Coma center as well as on the brightness of the sample. The elliptical shape does not appear to be caused by rotation; other possible causes are briefly discussed
A class of strongly degenerate elliptic operators
International Nuclear Information System (INIS)
Duong Minh Duc.
1988-04-01
Using a weighted Poincare inequality, we study (ω 1 ,...,ω n )-elliptic operators. This method is applicable to solve singular elliptic equations with conditions in W 1,2 on the boundary. We also get a result about the regularity of solutions of singular elliptic equations. An application to (ω 1 ,...ω n )-parabolic equations is given. (author). 33 refs
Macromolecular crystallography beamline X25 at the NSLS
Energy Technology Data Exchange (ETDEWEB)
Héroux, Annie; Allaire, Marc; Buono, Richard; Cowan, Matthew L.; Dvorak, Joseph; Flaks, Leon; LaMarra, Steven; Myers, Stuart F.; Orville, Allen M.; Robinson, Howard H.; Roessler, Christian G.; Schneider, Dieter K.; Shea-McCarthy, Grace; Skinner, John M.; Skinner, Michael; Soares, Alexei S.; Sweet, Robert M.; Berman, Lonny E., E-mail: berman@bnl.gov [Brookhaven National Laboratory, PO Box 5000, Upton, NY 11973-5000 (United States)
2014-04-08
A description of the upgraded beamline X25 at the NSLS, operated by the PXRR and the Photon Sciences Directorate serving the Macromolecular Crystallography community, is presented. Beamline X25 at the NSLS is one of the five beamlines dedicated to macromolecular crystallography operated by the Brookhaven National Laboratory Macromolecular Crystallography Research Resource group. This mini-gap insertion-device beamline has seen constant upgrades for the last seven years in order to achieve mini-beam capability down to 20 µm × 20 µm. All major components beginning with the radiation source, and continuing along the beamline and its experimental hutch, have changed to produce a state-of-the-art facility for the scientific community.
Automation in structural biology beamlines of the Photon Factory
International Nuclear Information System (INIS)
Igarashi, Noriyuki; Hiraki, Masahiko; Matsugaki, Naohiro; Yamada, Yusuke; Wakatsuki, Soichi
2007-01-01
The Photon Factory currently operates four synchrotron beamlines for protein crystallography and two more beamlines are scheduled to be constructed in the next years. Over the last years these beamlines have been upgraded and equipped with a fully automated beamline control system based on a robotic sample changer. The current system allows for remote operation, controlled from the user's area, of sample mounting, centering and data collection of pre-frozen crystals mounted in Hampton-type cryo-loops on goniometer head. New intuitive graphical user interfaces have been developed so as to control the complete beamline operation. Furthermore, algorithms for automatic sample centering based on pattern matching and X-ray beam scanning are being developed and combined with newly developed diffraction evaluation programs in order to complete entire automation of the data collection. (author)
Macromolecular crystallography beamline X25 at the NSLS
International Nuclear Information System (INIS)
Héroux, Annie; Allaire, Marc; Buono, Richard; Cowan, Matthew L.; Dvorak, Joseph; Flaks, Leon; LaMarra, Steven; Myers, Stuart F.; Orville, Allen M.; Robinson, Howard H.; Roessler, Christian G.; Schneider, Dieter K.; Shea-McCarthy, Grace; Skinner, John M.; Skinner, Michael; Soares, Alexei S.; Sweet, Robert M.; Berman, Lonny E.
2014-01-01
A description of the upgraded beamline X25 at the NSLS, operated by the PXRR and the Photon Sciences Directorate serving the Macromolecular Crystallography community, is presented. Beamline X25 at the NSLS is one of the five beamlines dedicated to macromolecular crystallography operated by the Brookhaven National Laboratory Macromolecular Crystallography Research Resource group. This mini-gap insertion-device beamline has seen constant upgrades for the last seven years in order to achieve mini-beam capability down to 20 µm × 20 µm. All major components beginning with the radiation source, and continuing along the beamline and its experimental hutch, have changed to produce a state-of-the-art facility for the scientific community
Generic radiation safety design for SSRL synchrotron radiation beamlines
Energy Technology Data Exchange (ETDEWEB)
Liu, James C. [Radiation Protection Department, Stanford Linear Accelerator Center (SLAC), MS 48, P.O. Box 20450, Stanford, CA 94309 (United States)]. E-mail: james@slac.stanford.edu; Fasso, Alberto [Radiation Protection Department, Stanford Linear Accelerator Center (SLAC), MS 48, P.O. Box 20450, Stanford, CA 94309 (United States); Khater, Hesham [Radiation Protection Department, Stanford Linear Accelerator Center (SLAC), MS 48, P.O. Box 20450, Stanford, CA 94309 (United States); Prinz, Alyssa [Radiation Protection Department, Stanford Linear Accelerator Center (SLAC), MS 48, P.O. Box 20450, Stanford, CA 94309 (United States); Rokni, Sayed [Radiation Protection Department, Stanford Linear Accelerator Center (SLAC), MS 48, P.O. Box 20450, Stanford, CA 94309 (United States)
2006-12-15
To allow for a conservative, simple, uniform, consistent, efficient radiation safety design for all SSRL beamlines, a generic approach has been developed, considering both synchrotron radiation (SR) and gas bremsstrahlung (GB) hazards. To develop the methodology and rules needed for generic beamline design, analytic models, the STAC8 code, and the FLUKA Monte Carlo code were used to pre-calculate sets of curves and tables that can be looked up for each beamline safety design. Conservative beam parameters and standard targets and geometries were used in the calculations. This paper presents the SPEAR3 beamline parameters that were considered in the design, the safety design considerations, and the main pre-calculated results that are needed for generic shielding design. In the end, the rules and practices for generic SSRL beamline design are summarized.
Elliptic Diophantine equations a concrete approach via the elliptic logarithm
Tzanakis, Nikos
2013-01-01
This book presents in a unified way the beautiful and deep mathematics, both theoretical and computational, on which the explicit solution of an elliptic Diophantine equation is based. It collects numerous results and methods that are scattered in literature. Some results are even hidden behind a number of routines in software packages, like Magma. This book is suitable for students in mathematics, as well as professional mathematicians.
An ultrahigh vacuum monochromator for photophysics beamline
International Nuclear Information System (INIS)
Meenakshi Raja Rao, P.; Padmanabhan, Saraswathy; Raja Sekhar, B.N.; Shastri, Aparna; Khan, H.A.; Sinha, A.K.
2000-08-01
The photophysics beamline designed for carrying out photoabsorption and fluorescence studies using the 450 MeV Synchrotron Radiation Source (SRS), INDUS-1, uses a 1 metre monochromator as premonochromator for monochromatising the continuum. An ultra high vacuum compatible monochromator in Seya-Namioka mount has been designed and fabricated indigenously. The monochromator was assembled and tested for its performance. Wavelength scanning mechanism was tested for its reproducibility and the monochromator was tested for its resolution using UV and VUV sources. An average spectral resolution of 2.5 A was achieved using a 1200 gr/mm grating. A wavelength repeatability of ± 1A was obtained. An ultra high vacuum of 2 X 10 -8 mbar was also achieved in the monochromator. Details of fabrication, assembly and testing are presented in this report. (author)
MX: A beamline control system toolkit
Lavender, William M.
2000-06-01
The development of experimental and beamline control systems for two Collaborative Access Teams at the Advanced Photon Source has resulted in the creation of a portable data acquisition and control toolkit called MX. MX consists of a set of servers, application programs and libraries that enable the creation of command line and graphical user interface applications that may be easily retargeted to new and different kinds of motor and device controllers. The source code for MX is written in ANSI C and Tcl/Tk with interprocess communication via TCP/IP. MX is available for several versions of Unix, Windows 95/98/NT and DOS. It may be downloaded from the web site http://www.imca.aps.anl.gov/mx/.
EPICS data archiver at SSRF beamlines
International Nuclear Information System (INIS)
Hu Zheng; Mi Qingru; Zheng Lifang; Li Zhong
2014-01-01
The control system of SSRF (Shanghai Synchrotron Radiation Facility) is based on EPICS (Experimental Physics and Industrial Control System). Operation data storage for synchrotron radiation facility is important for its status monitoring and analysis. At SSRF, operation data used to be index files recorded by traditional EPICS Channel Archiver. Nevertheless, index files are not suitable for long-term maintenance and difficult for data analysis. Now, RDB Channel Archiver and MySQL are used for SSRF beamline operation data archiving, so as to promote the data storage reliability and usability. By applying a new uploading mechanism to RDB Channel Archiver, its writing performance is improved. A web-based GUI (Graphics User Interface) is also developed to make it easier to access database. (authors)
Elliptical and lenticular galaxies evolution
International Nuclear Information System (INIS)
Vigroux, L.
1981-01-01
Different evolutionnary models for elliptical and lenticular galaxies are discussed. In the first part, we show that, at least some peculiar early types galaxies exhibit some activity. Then we describe the observationnal constraints: the color-magnitude diagram, the color gradient and the high metallicity of intraclusters gas. Among the different models, only the dissipation collapse followed by a hot wind driven by supernovae explosion explain in a natural way these constraints. Finally, the origin of SO is briefly discussed [fr
Holomorphic bundles over elliptic manifolds
International Nuclear Information System (INIS)
Morgan, J.W.
2000-01-01
In this lecture we shall examine holomorphic bundles over compact elliptically fibered manifolds. We shall examine constructions of such bundles as well as (duality) relations between such bundles and other geometric objects, namely K3-surfaces and del Pezzo surfaces. We shall be dealing throughout with holomorphic principal bundles with structure group GC where G is a compact, simple (usually simply connected) Lie group and GC is the associated complex simple algebraic group. Of course, in the special case G = SU(n) and hence GC = SLn(C), we are considering holomorphic vector bundles with trivial determinant. In the other cases of classical groups, G SO(n) or G = Sympl(2n) we are considering holomorphic vector bundles with trivial determinant equipped with a non-degenerate symmetric, or skew symmetric pairing. In addition to these classical cases there are the finite number of exceptional groups. Amazingly enough, motivated by questions in physics, much interest centres around the group E8 and its subgroups. For these applications it does not suffice to consider only the classical groups. Thus, while often first doing the case of SU(n) or more generally of the classical groups, we shall extend our discussions to the general semi-simple group. Also, we shall spend a good deal of time considering elliptically fibered manifolds of the simplest type, namely, elliptic curves
Beamline for X-ray Free Electron Laser of SACLA
International Nuclear Information System (INIS)
Tono, K; Togashi, T; Ohashi, H; Kimura, H; Takahashi, S; Takeshita, K; Tomizawa, H; Goto, S; Inubushi, Y; Sato, T; Yabashi, M
2013-01-01
A beamline for X-ray free electron laser (XFEL) has been developed at SACLA, SPring-8 Angstrom Compact free electron LAser. The beamline delivers and diagnoses an XFEL beam without degrading the beam quality. The transport optics are applicable in the range of 4–30 keV with a double-crystal monochromator or 4–15 keV with either of two double-mirror systems. A photon diagnostic system of the beamline monitors intensity, photon energy, center-of-mass position, and spatial profile in shot-by-shot and non-destructive manners.
Intrinsic shapes of discy and boxy ellipticals
International Nuclear Information System (INIS)
Fasano, Giovanni
1991-01-01
Statistical tests for intrinsic shapes of elliptical galaxies have given so far inconclusive and sometimes contradictory results. These failures have been often charged to the fact that classical tests consider only the two axisymmetric shapes (oblate versus prolate), while ellipticals are truly triaxial bodies. On the other hand, recent analyses indicate that the class of elliptical galaxies could be a mixture of (at least) two families having different morphology and dynamical behaviour: (i) a family of fast-rotating, disc-like ellipticals (discy); (ii) a family of slow-rotating, box-shaped ellipticals (boxy). In this paper we review the tests for instrinsic shapes of elliptical galaxies using data of better quality (CCD) with respect to previous applications. (author)
Moly99 Production Facility: Report on Beamline Components, Requirements, Costs
Energy Technology Data Exchange (ETDEWEB)
Bishofberger, Kip A. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2015-12-23
In FY14 we completed the design of the beam line for the linear accelerator production design concept. This design included a set of three bending magnets, quadrupole focusing magnets, and octopoles to flatten the beam on target. This design was generic and applicable to multiple different accelerators if necessary. In FY15 we built on that work to create specifications for the individual beam optic elements, including power supply requirements. This report captures the specification of beam line components with initial cost estimates for the NorthStar production facility.This report is organized as follows: The motivation of the beamline design is introduced briefly, along with renderings of the design. After that, a specific list is provided, which accounts for each beamline component, including part numbers and costs, to construct the beamline. After that, this report details the important sections of the beamline and individual components. A final summary and list of follow-on activities completes this report.
The BESSY X-ray microfocus beamline project
International Nuclear Information System (INIS)
Erko, A.; Schaefers, F.; Firsov, A.; Peatman, W.B.; Eberhardt, W.; Signorato, R.
2004-01-01
The design and construction of a beamline dedicated to X-ray absorption spectroscopy, X-ray fluorescence analysis, X-ray diffraction and X-ray small-angle scattering measurements with micron and submicron spatial resolution in the photon energy range of 1.9-30 keV are reported here. The main feature of the beamline is a combination of all these methods in one experimental station. The source will be a BESSY 7-T wavelength shifter, which is already in operation. Such a wavelength shifter with its continuous spectrum is an optimum source for these experiments. Glass capillaries and the combination of a linear Bragg-Fresnel lens and bimorph mirror were chosen as the special optical components in the beamline. This beamline will open for BESSY users the possibility to perform experiments on the same sample applying a microfocused X-ray beam in combination with advanced experimental methods
The monitoring system for macromolecular crystallography beamlines at BSRF
International Nuclear Information System (INIS)
Guo Xian; Chang Guangcai; Gan Quan; Shi Hong; Liu Peng; Sun Gongxing
2012-01-01
The monitoring system for macromolecular crystallography beamlines at BSRF (Beijing Synchrotron Radiation Facility) based on LabVIEW is introduced. In order to guarantee a safe, stable, and reliable running for the beamline devices, the system monitors the state of vacuum, cooling-water, optical components, beam, Liquid nitrogen in the beamlines in real time, detects faults and gives the alarm timely. System underlying uses the driver developed for the field devices for data acquisition, Data of collection is uploaded to the data-sharing platform makes it accessible via a network share. The upper system divides modules according to the actual function, and establishes the main interface of the monitoring system of beamline. To Facilitate data storage, management and inquiry, the system use LabSQL toolkit to achieve the interconnection with MySQL database which data of collection is sent to. (authors)
TREFF: Reflectometer and instrument component test beamline at MLZ
Directory of Open Access Journals (Sweden)
Peter Link
2017-11-01
Full Text Available TREFF is a high resolution polarized neutron reflectometer and instrument component test beamline resulting in a highly modular instrument providing a flexible beam line for various applications.
Overdetermined elliptic problems in topological disks
Mira, Pablo
2018-06-01
We introduce a method, based on the Poincaré-Hopf index theorem, to classify solutions to overdetermined problems for fully nonlinear elliptic equations in domains diffeomorphic to a closed disk. Applications to some well-known nonlinear elliptic PDEs are provided. Our result can be seen as the analogue of Hopf's uniqueness theorem for constant mean curvature spheres, but for the general analytic context of overdetermined elliptic problems.
Flattening and radio emission among elliptical galaxies
International Nuclear Information System (INIS)
Disney, M.J.; Sparks, W.B.; Wall, J.V.
1984-01-01
In a sample of 132 bright elliptical galaxies it is shown that there is a strong correlation between radio activity and flattening in the sense that radio ellipticals are both apparently and inherently rounder than the average elliptical. Both extended and compact sources are subject to the same correlation. No galaxies with axial ratios below 0.65 are found to be radio emitters. (author)
Elliptic hypergeometric functions associated with root systems
Rosengren, Hjalmar; Warnaar, S. Ole
2017-01-01
We give a survey of elliptic hypergeometric functions associated with root systems, comprised of three main parts. The first two form in essence an annotated table of the main evaluation and transformation formulas for elliptic hypergeometric integeral and series on root systems. The third and final part gives an introduction to Rains' elliptic Macdonald-Koornwinder theory (in part also developed by Coskun and Gustafson).
Experimental stations at I13 beamline at Diamond Light Source
Pešić, Z. D.; De Fanis, A.; Wagner, U.; Rau, C.
2013-03-01
The I13 beamline of Diamond Light Source has been operational since December 2011. The beamline encompass two fully independent branches devoted to coherent imaging experiments (coherent x-ray diffraction, coherent diffraction imaging and ptychography) and x-ray imaging (in-line phase contrast imaging, tomography and full-field microscopy). This paper gives an overview of the current status of experimental stations on both branches and outlines planned developments.
Experimental stations at I13 beamline at Diamond Light Source
International Nuclear Information System (INIS)
Pešić, Z D; Fanis, A De; Wagner, U; Rau, C
2013-01-01
The I13 beamline of Diamond Light Source has been operational since December 2011. The beamline encompass two fully independent branches devoted to coherent imaging experiments (coherent x-ray diffraction, coherent diffraction imaging and ptychography) and x-ray imaging (in-line phase contrast imaging, tomography and full-field microscopy). This paper gives an overview of the current status of experimental stations on both branches and outlines planned developments.
Circular dichroism beamline B23 at the Diamond Light Source.
Hussain, Rohanah; Jávorfi, Tamás; Siligardi, Giuliano
2012-01-01
Synchrotron radiation circular dichroism (SRCD) is a well established technique in structural biology. The first UV-VIS beamline, dedicated to circular dichroism, at Diamond Light Source Ltd, a third-generation synchrotron facility in south Oxfordshire, UK, has recently become operational and it is now available for the user community. Herein the main characteristics of the B23 SRCD beamline, the ancillary facilities available for users, and some of the recent advances achieved are summarized.
Current schemes for National Synchrotron Light Source UV beamlines
International Nuclear Information System (INIS)
Williams, G.P.; Howells, M.R.; McKinney, W.R.
1979-01-01
We describe in some detail four beamlines proposed for the National Synchrotron Light Source uv ring at Brookhaven National Laboratory. Three grazing-incidence instruments, one of the plane grating Mijake type and two with toroidal gratings at grazing angles of 2-1/2 0 and 15 0 are described. Two normal incidence instruments, one using the source as entrance slit and accepting 75 milliradians horizontally are also discussed. In each case we have estimated the output fluxes expected from such beamlines
An overview of what is required and when for developing a beamline at the ALS
International Nuclear Information System (INIS)
Hussain, Z.
1995-08-01
This report discusses these topics: reviews required for developing a beamline at the ALS; work items and documentation required for the beamline design review; information to be communicated to the ALS staff before the beamline readiness review; work items and documentation required for the beamline readiness review; contacts for information, technical questions, and sources of additional information; and checklist of what is required and when for developing a beamline
Arbitrarily elliptical-cylindrical invisible cloaking
International Nuclear Information System (INIS)
Jiang Weixiang; Cui Tiejun; Yu Guanxia; Lin Xianqi; Cheng Qiang; Chin, J Y
2008-01-01
Based on the idea of coordinate transformation (Pendry, Schurig and Smith 2006 Science 312 1780), arbitrarily elliptical-cylindrical cloaks are proposed and designed. The elliptical cloak, which is composed of inhomogeneous anisotropic metamaterials in an elliptical-shell region, will deflect incoming electromagnetic (EM) waves and guide them to propagate around the inner elliptical region. Such EM waves will return to their original propagation directions without distorting the waves outside the elliptical cloak. General formulations of the inhomogeneous and anisotropic permittivity and permeability tensors are derived for arbitrarily elliptical axis ratio k, which can also be used for the circular cloak when k = 1. Hence the elliptical cloaks can make a large range of objects invisible, from round objects (when k approaches 1) to long and thin objects (when k is either very large or very small). We also show that the material parameters in elliptical cloaking are singular at only two points, instead of on the whole inner circle for circular cloaking, which are much easier to be realized in actual applications. Full-wave simulations are given to validate the arbitrarily elliptical cloaking
Elliptic-symmetry vector optical fields.
Pan, Yue; Li, Yongnan; Li, Si-Min; Ren, Zhi-Cheng; Kong, Ling-Jun; Tu, Chenghou; Wang, Hui-Tian
2014-08-11
We present in principle and demonstrate experimentally a new kind of vector fields: elliptic-symmetry vector optical fields. This is a significant development in vector fields, as this breaks the cylindrical symmetry and enriches the family of vector fields. Due to the presence of an additional degrees of freedom, which is the interval between the foci in the elliptic coordinate system, the elliptic-symmetry vector fields are more flexible than the cylindrical vector fields for controlling the spatial structure of polarization and for engineering the focusing fields. The elliptic-symmetry vector fields can find many specific applications from optical trapping to optical machining and so on.
Doppler Velocity Signatures of Idealized Elliptical Vortices
Directory of Open Access Journals (Sweden)
Wen-Chau Lee
2006-01-01
Full Text Available Doppler radar observations have revealed a class of atmospheric vortices (tropical cyclones, tornadoes, dust devils that possess elliptical radar reflectivity signatures. One famous example is Typhoon Herb (1996 that maintained its elliptical reflectivity structure over a 40-hour period. Theoretical work and dual-Doppler analyses of observed tropical cyclones have suggested two physical mechanisms that can explain the formation of two types of elliptical vortices observed in nature, namely, the combination of a circular vortex with either a wavenumber two vortex Rossby wave or a deformation field. The characteristics of these two types of elliptical vortices and their corresponding Doppler velocity signatures have not been previously examined.
Dark matter in elliptical galaxies
Carollo, C. M.; Zeeuw, P. T. DE; Marel, R. P. Van Der; Danziger, I. J.; Qian, E. E.
1995-01-01
We present measurements of the shape of the stellar line-of-sight velocity distribution out to two effective radii along the major axes of the four elliptical galaxies NGC 2434, 2663, 3706, and 5018. The velocity dispersion profiles are flat or decline gently with radius. We compare the data to the predictions of f = f(E, L(sub z)) axisymmetric models with and without dark matter. Strong tangential anisotropy is ruled out at large radii. We conclude from our measurements that massive dark halos must be present in three of the four galaxies, while for the fourth galaxy (NGC 2663) the case is inconclusive.
Stellar Populations in Elliptical Galaxies
Angeletti, Lucio; Giannone, Pietro
The R1/n law for the radial surface brightness of elliptical galaxies and the "Best Accretion Model" together with the "Concentration Model" have been combined in order to determine the mass and dynamical structure of largely-populated star systems. Families of models depending on four parameters have been used to fit the observed surface radial profiles of some spectro-photometric indices of a sample of eleven galaxies. We present the best agreements of the spectral index Mg2 with observations for three selected galaxies representative of the full sample. For them we have also computed the spatial distributions of the metal abundances, which are essential to achieve a population synthesis.
Beamline for Schools 2016: How to be a CERN scientist
2016-01-01
Two teams of high-school students from the UK and Poland had the opportunity to conduct their own experiments at a fully equipped CERN beamline. Students from the 2016 Beamline for Schools competition working on their experiment. (Image: Noemí Carabán Gonzalez/CERN) Two teams of high-school students from the UK and Poland had the opportunity to conduct their own experiments at a fully equipped CERN beamline, after winning the Beamline for Schools competition. The teams, ”Pyramid Hunters” from Poland and “Relatively Special” from the United Kingdom, spent 10 days at CERN conducting the experiments they had dreamt up in their winning proposals. The Beamline for Schools competition gives high-school students the chance to run an experiment on a fully equipped CERN beamline, in the same way researchers do at the Large Hadron Collider and other CERN facilities every day. To know more about their stay at CERN and the experiments they&r...
Data Management System at the Photon Factory Macromolecular Crystallography Beamline
International Nuclear Information System (INIS)
Yamada, Y; Matsugaki, N; Chavas, L M G; Hiraki, M; Igarashi, N; Wakatsuki, S
2013-01-01
Macromolecular crystallography is a very powerful tool to investigate three-dimensional structures of macromolecules at the atomic level, and is widely spread among structural biology researchers. Due to recent upgrades of the macromolecular crystallography beamlines at the Photon Factory, beamline throughput has improved, allowing more experiments to be conducted during a user's beam time. Although the number of beamlines has increased, so has the number of beam time applications. Consequently, both the experimental data from users' experiments and data derived from beamline operations have dramatically increased, causing difficulties in organizing these diverse and large amounts of data for the beamline operation staff and users. To overcome this problem, we have developed a data management system by introducing commercial middleware, which consists of a controller, database, and web servers. We have prepared several database projects using this system. Each project is dedicated to a certain aspect such as experimental results, beam time applications, beam time schedule, or beamline operation reports. Then we designed a scheme to link all the database projects.
The INE-Beamline for Actinide Research at ANKA
Brendebach, Boris; Denecke, Melissa A.; Rothe, Jörg; Dardenne, Kathy; Römer, Jürgen
2007-02-01
The INE-Beamline for actinide research at the synchrotron source ANKA is now fully operational. This beamline was designed, built, and commissioned by the Institut für Nukleare Entsorgung (INE) at the Forschungszentrum Karlsruhe (FZK), Germany. It is dedicated to actinide speciation investigations related to nuclear waste disposal as well as applied and basic actinide research. Experiments on radioactive samples with activities up to 106 times the limit of exemption inside a safe and flexible double containment concept are possible. The close proximity of the beamline to INE's active laboratories is unique in Europe. Currently, experiments can be performed in an X-ray energy range from around 2.15 keV (P K edge) to 24.35 keV (Pd K edge). The INE-Beamline design is optimized for spectroscopy with emphasis on surface sensitive techniques. A microfocus option is presently being installed at the INE-Beamline. Access to the INE-Beamline is possible through cooperation with INE, through the ANKA proposal system and via the European Network of Excellence for Actinide Sciences (ACTINET).
Discussions for the shielding materials of synchrotron radiation beamline hutches
International Nuclear Information System (INIS)
Asano, Y.
2006-01-01
Many synchrotron radiation facilities are now under operation such as E.S.R.F., APS, and S.P.ring-8. New facilities with intermediated stored electron energy are also under construction and designing such as D.I.A.M.O.N.D., S.O.L.E.I.L., and S.S.R.F.. At these third generation synchrotron radiation facilities, the beamline shielding as well as the bulk shield is very important for designing radiation safety because of intense and high energy synchrotron radiation beam. Some reasons employ lead shield wall for the synchrotron radiation beamlines. One is narrow space for the construction of many beamlines at the experimental hall, and the other is the necessary of many movable mechanisms at the beamlines, for examples. Some cases are required to shield high energy neutrons due to stored electron beam loss and photoneutrons due to gas Bremsstrahlung. Ordinary concrete and heavy concrete are coming up to shield material of synchrotron radiation beamline hutches. However, few discussions have been performed so far for the shielding materials of the hutches. In this presentation, therefore, we will discuss the characteristics of the shielding conditions including build up effect for the beamline hutches by using the ordinary concrete, heavy concrete, and lead for shielding materials with 3 GeV and 8 GeV class synchrotron radiation source. (author)
Capillary concentrators for synchrotron radiation beamlines
International Nuclear Information System (INIS)
Heald, S.M.; Brewe, D.L.; Kim, K.H.; Brown, F.C.; Barg, B.; Stern, E.A.
1996-01-01
Capillary concentrators condense x-rays by multiple reflections down a gradually tapering capillary. They can provide sub-micron beam spots, and are promising candidates for use in the next generation x-ray microprobe beamlines. The weak energy dependence of their properties make them especially useful for energy scanning applications such as micro-XAFS. This paper examines the potential performance of capillary optics for an x-ray microprobe, as well as some practical issues such as fabrication and alignment. Best performance at third generation sources requires long capillaries, and the authors have been using fiber optics techniques to fabricate capillaries up to one meter in length. The performance of shorter (less than about 0.5 m) capillaries has often been found to agree well with theoretical calculations, indicating the inner surface is a high quality x-ray reflector. These capillaries have been tested at the NSLS for imaging and micro-XAFS down to 2.6 microm resolution with excellent results. On an unfocused bend magnet line flux density approaching 10 6 ph/sec/microm 2 has been achieved. While nearly optimum profiles have been achieved for longer capillaries, the results have been disappointing, and alignment problems are suspected. The dramatic improvement in performance possible at third generation synchrotrons such as the APS is discussed along with improvements possible by using the capillaries in conjunction with coupling optics
Design of the LBNF Beamline Target Station
Energy Technology Data Exchange (ETDEWEB)
Tariq, S. [Fermilab; Ammigan, K. [Fermilab; Anderson, K.;
2016-10-01
The Long Baseline Neutrino Facility (LBNF) project will build a beamline located at Fermilab to create and aim an intense neutrino beam of appropriate energy range toward the DUNE detectors at the SURF facility in Lead, South Dakota. Neutrino production starts in the Target Station, which consists of a solid target, magnetic focusing horns, and the associated sub-systems and shielding infrastructure. Protons hit the target producing mesons which are then focused by the horns into a helium-filled decay pipe where they decay into muons and neutrinos. The target and horns are encased in actively cooled steel and concrete shielding in a chamber called the target chase. The reference design chase is filled with air, but nitrogen and helium are being evaluated as alternatives. A replaceable beam window separates the decay pipe from the target chase. The facility is designed for initial operation at 1.2 MW, with the ability to upgrade to 2.4 MW, and is taking advantage of the experience gained by operating Fermilab’s NuMI facility. We discuss here the design status, associated challenges, and ongoing R&D and physics-driven component optimization of the Target Station.
Diamond beamline I07: a beamline for surface and interface diffraction.
Nicklin, Chris; Arnold, Tom; Rawle, Jonathan; Warne, Adam
2016-09-01
Beamline I07 at Diamond Light Source is dedicated to the study of the structure of surfaces and interfaces for a wide range of sample types, from soft matter to ultrahigh vacuum. The beamline operates in the energy range 8-30 keV and has two endstations. The first houses a 2+3 diffractometer, which acts as a versatile platform for grazing-incidence techniques including surface X-ray diffraction, grazing-incidence small- (and wide-) angle X-ray scattering, X-ray reflectivity and grazing-incidence X-ray diffraction. A method for deflecting the X-rays (a double-crystal deflector) has been designed and incorporated into this endstation, extending the surfaces that can be studied to include structures formed on liquid surfaces or at liquid-liquid interfaces. The second experimental hutch contains a similar diffractometer with a large environmental chamber mounted on it, dedicated to in situ ultrahigh-vacuum studies. It houses a range of complementary surface science equipment including a scanning tunnelling microscope, low-energy electron diffraction and X-ray photoelectron spectroscopy ensuring that correlations between the different techniques can be performed on the same sample, in the same chamber. This endstation allows accurate determination of well ordered structures, measurement of growth behaviour during molecular beam epitaxy and has also been used to measure coherent X-ray diffraction from nanoparticles during alloying.
Convex bodies with many elliptic sections
Arelio, Isaac; Montejano, Luis
2014-01-01
{We show in this paper that two normal elliptic sections through every point of the boundary of a smooth convex body essentially characterize an ellipsoid and furthermore, that four different pairwise non-tangent elliptic sections through every point of the $C^2$-differentiable boundary of a convex body also essentially characterize an ellipsoid.
Kinematically Decoupled Cores in Dwarf (Elliptical) Galaxies
Toloba, E.; Peletier, R. F.; Guhathakurta, P.; van de Ven, G.; Boissier, S.; Boselli, A.; Brok, M. d.; Falcón-Barroso, J.; Hensler, G.; Janz, J.; Laurikainen, E.; Lisker, T.; Paudel, S.; Ryś, A.; Salo, H.
An overview is given of what we know about the frequency of kinematically decoupled cores in dwarf elliptical galaxies. New observations show that kinematically decoupled cores happen just as often in dwarf elliptical as in ordinary early-type galaxies. This has important consequences for the
Drinfeld currents of dynamical elliptic algebra
International Nuclear Information System (INIS)
Hou Boyu; Fan Heng; Yang Wenli; Cao Junpeng
2000-01-01
From the generalized Yang-Baxter relations RLL=LLR*, where R and R* are the dynamical R-matrix of A n-1 (1) type face model with the elliptic module shifted by the center of the algebra, using the Ding-Frenkel correspondence, the authors obtain the Drinfeld currents of dynamical elliptic algebra
The elliptic genus and Hidden symmetry
International Nuclear Information System (INIS)
Jaffe, A.
2001-01-01
We study the elliptic genus (a partition function) in certain interacting, twist quantum field theories. Without twists, these theories have N=2 supersymmetry. The twists provide a regularization, and also partially break the supersymmetry. In spite of the regularization, one can establish a homotopy of the elliptic genus in a coupling parameter. Our construction relies on a priori estimates and other methods from constructive quantum field theory; this mathematical underpinning allows us to justify evaluating the elliptic genus at one endpoint of the homotopy. We obtain a version of Witten's proposed formula for the elliptic genus in terms of classical theta functions. As a consequence, the elliptic genus has a hidden SL(2,Z) symmetry characteristic of conformal theory, even though the underlying theory is not conformal. (orig.)
Multicolor surface photometry of 17 ellipticals
International Nuclear Information System (INIS)
Franx, M.; Illingworth, G.; Heckman, T.
1989-01-01
Multicolor two-dimensional surface photometry was used to obtain radial profiles for surface brightness, color, ellipticity, position angle, and the residuals from the fitted ellipses described by the cos(n phi) and sin(n phi) terms (where n = 3 and 4) for 17 elliptical galaxies. It is found that at radii as large as five times the seeing FWHM, seeing can affect the ellipticity at the 10 percent level and introduce uncertainty in the position angles of several degrees, particularly for very round ellipticals. The present profiles are found to agree well with previous data, with rms differences of 0.02 in ellipticity and 2 deg in position angle. The observed color gradients are consistent with a decrease in the metallicity by a factor of about 2 per decade in radius. 61 refs
Elliptical excisions: variations and the eccentric parallelogram.
Goldberg, Leonard H; Alam, Murad
2004-02-01
The elliptical (fusiform) excision is a basic tool of cutaneous surgery. To assess the design, functionality, ease of construction, and aesthetic outcomes of the ellipse. A systematic review of elliptical designs and their site-specific benefits and limitations. In particular, we consider the (1). context of prevailing relaxed skin tension lines and tissue laxity; and (2). removal of the smallest possible amount of tissue around the lesion and in the "dog-ears." Attention is focused on intuitive methods that can be reproducibly planned and executed. Elliptical variations are easily designed and can be adapted to many situations. The eccentric parallelogram excision is offered as a new technique that minimizes notching and focal tension in the center of an elliptical closure. Conclusion The elliptical (fusiform) excision is an efficient, elegant, and versatile technique that will remain a mainstay of the cutaneous surgical armamentarium.
Optimization of a coherent soft x-ray beamline for coherent scattering experiments at NSLS-II
Energy Technology Data Exchange (ETDEWEB)
Shapiro D.; Chubar, O.; Kaznatcheev, K.; Reininger, R.; Sanchez-Hanke, C.; Wang, S.
2011-08-21
The coherent soft x-ray and full polarization control (CSX) beamline at the National Synchrotron Light Source - II (NSLS-II) will deliver 1013 coherent photons per second in the energy range of 0.2-2 keV with a resolving power of 2000. The source, a dual elliptically polarizing undulator (EPU), and beamline optics should be optimized to deliver the highest possible coherent flux in a 10-30 {micro}m spot for use in coherent scattering experiments. Using the computer code Synchrotron Radiation Workshop (SRW), we simulate the photon source and focusing optics in order to investigate the conditions which provide the highest usable coherent intensity on the sample. In particular, we find that an intermediate phasing magnet is needed to correct for the relative phase between the two EPUs and that the optimum phase setting produces a spectrum in which the desired wavelength is slightly red-shifted thus requiring a larger aperture than originally anticipated. This setting is distinct from that which produces an on-axis spectrum similar to a single long undulator. Furthermore, partial coherence calculations, utilizing a multiple electron approach, indicate that a high degree of spatial coherence is still obtained at the sample location when such an aperture is used. The aperture size which maximizes the signal-to-noise ratio of a double-slit experiment is explored. This combination of high coherence and intensity is ideally suited for x-ray ptychography experiments which reconstruct the scattering density from micro-diffraction patterns. This technique is briefly reviewed and the effects on the image quality of proximity to the beamline focus are explored.
Emittance Measurement for Beamline Extension at the PET Cyclotron
Directory of Open Access Journals (Sweden)
Sae-Hoon Park
2016-01-01
Full Text Available Particle-induced X-ray emission is used for determining the elemental composition of materials. This method uses low-energy protons (of several MeV, which can be obtained from high-energy (of tens MeV accelerators. Instead of manufacturing an accelerator for generating the MeV protons, the use of a PET cyclotron has been suggested for designing the beamline for multipurpose applications, especially for the PIXE experiment, which has a dedicated high-energy (of tens MeV accelerator. The beam properties of the cyclotron were determined at this experimental facility by using an external beamline before transferring the ion beam to the experimental chamber. We measured the beam profile and calculated the emittance using the pepper-pot method. The beam profile was measured as the beam current using a wire scanner, and the emittance was measured as the beam distribution at the beam dump using a radiochromic film. We analyzed the measurement results and are planning to use the results obtained in the simulations of external beamline and aligned beamline components. We will consider energy degradation after computing the beamline simulation. The experimental study focused on measuring the emittance from the cyclotron, and the results of this study are presented in this paper.
A beamline for macromolecular crystallography at the Advanced Light Source
International Nuclear Information System (INIS)
Padmore, H.A.; Earnest, T.; Kim, S.H.; Thompson, A.C.; Robinson, A.L.
1994-08-01
A beamline for macromolecular crystallography has been designed for the ALS. The source will be a 37-pole wiggler with a, 2-T on-axis peak field. The wiggler will illuminate three beamlines, each accepting 3 mrad of horizontal aperture. The central beamline will primarily be used for multiple-wavelength anomalous dispersion measurements in the wavelength range from 4 to 0.9 angstrom. The beamline optics will comprise a double-crystal monochromator with a collimating pre-mirror and a double-focusing mirror after the monochromator. The two side stations will be used for fixed-wavelength experiments within the wavelength range from 1.5 to 0.95 angstrom. The optics will consist of a conventional vertically focusing cylindrical mirror followed by an asymmetrically cut curved-crystal monochromator. This paper presents details of the optimization of the wiggler source for crystallography, gives a description of the beamline configuration, and discusses the reasons for the choices made
A hard X-ray nanoprobe beamline for nanoscale microscopy
Energy Technology Data Exchange (ETDEWEB)
Winarski, Robert P., E-mail: winarski@anl.gov; Holt, Martin V. [Argonne National Laboratory, 9700 South Cass Avenue, Argonne, IL 60441 (United States); Rose, Volker [Argonne National Laboratory, 9700 South Cass Avenue, Argonne, IL 60441 (United States); Fuesz, Peter; Carbaugh, Dean; Benson, Christa; Shu, Deming; Kline, David; Stephenson, G. Brian; McNulty, Ian [Argonne National Laboratory, 9700 South Cass Avenue, Argonne, IL 60441 (United States); Maser, Jörg [Argonne National Laboratory, 9700 South Cass Avenue, Argonne, IL 60441 (United States)
2012-11-01
The Hard X-ray Nanoprobe Beamline is a precision platform for scanning probe and full-field microscopy with 3–30 keV X-rays. A combination of high-stability X-ray optics and precision motion sensing and control enables detailed studies of the internal features of samples with resolutions approaching 30 nm. The Hard X-ray Nanoprobe Beamline (or Nanoprobe Beamline) is an X-ray microscopy facility incorporating diffraction, fluorescence and full-field imaging capabilities designed and operated by the Center for Nanoscale Materials and the Advanced Photon Source at Sector 26 of the Advanced Photon Source at Argonne National Laboratory. This facility was constructed to probe the nanoscale structure of biological, environmental and material sciences samples. The beamline provides intense focused X-rays to the Hard X-ray Nanoprobe (or Nanoprobe) which incorporates Fresnel zone plate optics and a precision laser sensing and control system. The beamline operates over X-ray energies from 3 to 30 keV, enabling studies of most elements in the periodic table, with a particular emphasis on imaging transition metals.
The Scanning Nanoprobe Beamline Nanoscopium at Synchrotron Soleil
Somogyi, A.; Kewish, C. M.; Polack, F.; Moreno, T.
2011-09-01
The Nanoscopium beamline at Synchrotron Soleil will offer advanced scanning-based hard x-ray imaging techniques in the 5- to 20-keV energy range, for user communities working in the earth, environmental, and life sciences. Two dedicated end stations will exploit x-ray coherence to produce images in which contrast is based on a range of physical processes. In the first experiment hutch, coherent scatter imaging techniques will produce images in which contrast arises from spatial variations in the complex refractive index, and orientation in the nanostructure of samples. In the second experiment hutch, elemental mapping will be carried out at the trace (ppm) level by scanning x-ray fluorescence, speciation mapping by XANES, and phase gradient mapping by scanning differential phase contrast imaging. The beamline aims to reach sub-micrometric, down to 30 nm, spatial resolution. This ˜155-meter-long beamline will share the straight section with a future tomography beamline by using canted undulators having 6.5-mrad separation angle. The optical design of Nanoscopium aims to reduce the effect of instabilities on the probing nanobeam by utilizing an all-horizontal geometry for the reflections of the primary beamline mirrors, which focus onto a slit, creating an over-filled secondary source. Kirkpatrick-Baez mirrors and Fresnel zone plates will be used as focusing devices in the experiment hutches. Nanoscopium is expected to commence user operation in 2013.
Upgrade of Spring-8 Beamline Network with Vlan Technology Over Gigabit Ethernet
Ishii, M.; Fukui, T.; Furukawa, Y.; Nakatani, T.; Ohata, T.; Tanaka, R.
2001-01-01
The beamline network system at SPring-8 consists of three LANs; a BL-LAN for beamline component control, a BL-USER-LAN for beamline experimental users and an OA-LAN for the information services. These LANs are interconnected by a firewall system. Since the network traffic and the number of beamlines have increased, we upgraded the backbone of BL-USER-LAN from Fast Ethernet to Gigabit Ethernet. And then, to establish the independency of a beamline and to raise flexibility of every beamline, we...
Modeling groundwater flow to elliptical lakes and through multi-aquifer elliptical inhomogeneities
Bakker, Mark
2004-05-01
Two new analytic element solutions are presented for steady flow problems with elliptical boundaries. The first solution concerns groundwater flow to shallow elliptical lakes with leaky lake beds in a single-aquifer. The second solution concerns groundwater flow through elliptical cylinder inhomogeneities in a multi-aquifer system. Both the transmissivity of each aquifer and the resistance of each leaky layer may differ between the inside and the outside of an inhomogeneity. The elliptical inhomogeneity may be bounded on top by a shallow elliptical lake with a leaky lake bed. Analytic element solutions are obtained for both problems through separation of variables of the Laplace and modified-Helmholtz differential equations in elliptical coordinates. The resulting equations for the discharge potential consist of infinite sums of products of exponentials, trigonometric functions, and modified-Mathieu functions. The series are truncated but still fulfill the differential equation exactly; boundary conditions are met approximately, but up to machine accuracy provided enough terms are used. The head and flow may be computed analytically at any point in the aquifer. Examples are given of uniform flow through an elliptical lake, a well pumping near two elliptical lakes, and uniform flow through three elliptical inhomogeneities in a multi-aquifer system. Mathieu functions may be applied in a similar fashion to solve other groundwater flow problems in semi-confined aquifers and leaky aquifer systems with elliptical internal or external boundaries.
Excursion Processes Associated with Elliptic Combinatorics
Baba, Hiroya; Katori, Makoto
2018-06-01
Researching elliptic analogues for equalities and formulas is a new trend in enumerative combinatorics which has followed the previous trend of studying q-analogues. Recently Schlosser proposed a lattice path model in the square lattice with a family of totally elliptic weight-functions including several complex parameters and discussed an elliptic extension of the binomial theorem. In the present paper, we introduce a family of discrete-time excursion processes on Z starting from the origin and returning to the origin in a given time duration 2 T associated with Schlosser's elliptic combinatorics. The processes are inhomogeneous both in space and time and hence expected to provide new models in non-equilibrium statistical mechanics. By numerical calculation we show that the maximum likelihood trajectories on the spatio-temporal plane of the elliptic excursion processes and of their reduced trigonometric versions are not straight lines in general but are nontrivially curved depending on parameters. We analyze asymptotic probability laws in the long-term limit T → ∞ for a simplified trigonometric version of excursion process. Emergence of nontrivial curves of trajectories in a large scale of space and time from the elementary elliptic weight-functions exhibits a new aspect of elliptic combinatorics.
The microspectroscopy beamline for the Australian synchrotron project
International Nuclear Information System (INIS)
Boldeman, J.W.; Ryan, C.; Cohen, D.D.
2005-01-01
A new multi-million dollar synchrotron facility is currently being built in Clayton, Victoria. This is a 3GeV electron machine, it will be a world class machine and contain state of the art beamline facilities for both Australian and overseas scientists. It is due for completion in mid-2007. This specialised beamline will provide sub-micron spatial resolution with the highest flux possible. It will combine 2D mapping with micro X-ray fluorescence (μ-XRF), micro X-ray absorption near edge spectroscopy (μ-XANES) and micro X-ray absorption fine structure spectroscopy (μ-XAFS) for elemental and chemical analysis to solve scientific problems that can only be understood using X-ray beams with sub-micron resolutions. In this paper we describe some key beamline components and give details about their performance specifications. 7 refs., 4 figs.; 1 tab
Simulating the Beam-line at CERN's ISOLDE Experiment
McGrath, Casey
2013-01-01
Maximizing the optical matching along portions of the ISOLDE beam-line and automating this procedure will make it easier for scientists to determine what the strengths of the electrical elds of each beam-line element should be in order to reduce particle loss. Simulations are run using a program called MAD-X, however, certain issues were discovered that hindered an immediate success of the simulations. Specifically, the transfer matrices for electrostatic components like the switchyards, kickers, and electric quadrupoles were missing from the original coding. The primary aim of this project was to design these components using AutoCAD and then extract the transfer matrices using SIMION. Future work will then implement these transfer matrices into the MAD-X code to make the simulations of the beam-line more accurate.
The Materials Science beamline upgrade at the Swiss Light Source
Energy Technology Data Exchange (ETDEWEB)
Willmott, P. R., E-mail: philip.willmott@psi.ch; Meister, D.; Leake, S. J.; Lange, M.; Bergamaschi, A. [Swiss Light Source, Paul Scherrer Institut, CH-5232 Villigen (Switzerland); and others
2013-07-16
The wiggler X-ray source of the Materials Science beamline at the Swiss Light Source has been replaced with a 14 mm-period cryogenically cooled in-vacuum undulator. In order to best exploit the increased brilliance of this new source, the entire front-end and optics have been redesigned. The Materials Science beamline at the Swiss Light Source has been operational since 2001. In late 2010, the original wiggler source was replaced with a novel insertion device, which allows unprecedented access to high photon energies from an undulator installed in a medium-energy storage ring. In order to best exploit the increased brilliance of this new source, the entire front-end and optics had to be redesigned. In this work, the upgrade of the beamline is described in detail. The tone is didactic, from which it is hoped the reader can adapt the concepts and ideas to his or her needs.
Long-pulse beamlines for the mirror fusion test facility
International Nuclear Information System (INIS)
Stone, R.R.; Goldner, A.I.; Poulsen, P.
1984-01-01
We have recently obtained test results indicating that a beam of pure full-energy deuterium particles can be delivered to the plasma targets in MFTF-B. We used a close-coupled separator magnet with the ion source to separate the impurities from the full-energy deuterium particles. Our completed studies show that the usual iron-core sweep magnet and shielding used in neutral beamlines can be eliminated and the gas flow out of the beamline decreased. This design also reduces beam losses. We will use smooth-bore OFHC tube arrays brazed to manifolds for the active heat transfer surfaces. Tests indicate that both burnout and life requirements are met by this design. In this paper, we present test results and discuss the MFTF-B long-pulse beamline configuration
Physical optics simulations with PHASE for SwissFEL beamlines
Energy Technology Data Exchange (ETDEWEB)
Flechsig, U.; Follath, R.; Reiche, S. [Paul Scherrer Institut, Swiss Light Source, 5232 Villigen PSI (Switzerland); Bahrdt, J. [Helmholtz Zentrum Berlin (Germany)
2016-07-27
PHASE is a software tool for physical optics simulation based on the stationary phase approximation method. The code is under continuous development since about 20 years and has been used for instance for fundamental studies and ray tracing of various beamlines at the Swiss Light Source. Along with the planning for SwissFEL a new hard X-ray free electron laser under construction, new features have been added to permit practical performance predictions including diffraction effects which emerge with the fully coherent source. We present the application of the package on the example of the ARAMIS 1 beamline at SwissFEL. The X-ray pulse calculated with GENESIS and given as an electrical field distribution has been propagated through the beamline to the sample position. We demonstrate the new features of PHASE like the treatment of measured figure errors, apertures and coatings of the mirrors and the application of Fourier optics propagators for free space propagation.
Elliptic hypergeometric functions and the representation theory
International Nuclear Information System (INIS)
Spiridonov, V.P.
2011-01-01
Full text: (author)Elliptic hypergeometric functions were discovered around ten years ago. They represent the top level known generalization of the Euler beta integral and Euler-Gauss 2 F 1 hypergeometric function. In general form they are defined by contour integrals involving elliptic gamma functions. We outline the structure of the simplest examples of such functions and discuss their relations to the representation theory of the classical Lie groups and their various deformations. In one of the constructions elliptic hypergeometric integrals describe purely group-theoretical objects having the physical meaning of superconformal indices of four-dimensional supersymmetric gauge field theories
International Workshop on Elliptic and Parabolic Equations
Schrohe, Elmar; Seiler, Jörg; Walker, Christoph
2015-01-01
This volume covers the latest research on elliptic and parabolic equations and originates from the international Workshop on Elliptic and Parabolic Equations, held September 10-12, 2013 at the Leibniz Universität Hannover. It represents a collection of refereed research papers and survey articles written by eminent scientist on advances in different fields of elliptic and parabolic partial differential equations, including singular Riemannian manifolds, spectral analysis on manifolds, nonlinear dispersive equations, Brownian motion and kernel estimates, Euler equations, porous medium type equations, pseudodifferential calculus, free boundary problems, and bifurcation analysis.
Kinematic mounting systems for NSLS beamlines and experiments
International Nuclear Information System (INIS)
Oversluizen, T.; Stoeber, W.; Johnson, E.D.
1991-01-01
Methods for kinematically mounting equipment are well established, but applications at synchrotron radiation facilities are subject to constraints not always encountered in more traditional laboratory settings. Independent position adjustment of beamline components can have significant benefits in terms of minimizing time spent aligning, and maximizing time spent acquiring data. In this paper, we use examples taken from beamlines at the NSLS to demonstrate approaches for optimization of the reproducibility, stability, excursion, and set-up time for various situations. From our experience, we extract general principles which we hope will be useful for workers at other synchrotron radiation facilities. 7 refs., 4 figs
Infrared Spectroscopy Beamline Based on a Tabletop Storage Ring
Haque, Md. Monirul; Moon, Ahsa; Yamada, Hironari
2012-01-01
An optical beamline dedicated to the infrared (IR) spectroscopy has been constructed at MIRRORCLE, a tabletop storage ring. The beamline has been designed for the use of infrared synchrotron radiation (IRSR) emitted from a bending magnet of 156 mm bending radius with the acceptance angle of 355(H) × 138(V) mrad to obtain high flux. The IR emission is forced by an exactly circular optics, named photon storage ring (PhSR), placed around the electron orbit and is collected by a “magic mirror” as...
High Power Primary Slits For The ESRF Beamlines
International Nuclear Information System (INIS)
Marion, Philippe; Zhang Lin
2004-01-01
At the ESRF, in order to face the increased heatload on the beamlines, new Primary slits have been developed. The main features of the new slits are: beam size = 0 to 4mm; accuracy = 10μm; acceptable power density: 600 W/mm2; acceptable total power: 7kW; total length: 740mm. The adopted design offers a wide water cooled area; the shapes of the heated parts have been optimized to reduce the local thermal stresses. The maximum calculated temperature and stress are 353C and 606MPa. Six of these new slits are now in operation on ESRF Beamlines
A compact cost-effective beamline for a PET Cyclotron
International Nuclear Information System (INIS)
Dehnel, M.P.; Jackle, P.; Roeder, M.; Stewart, T.; Theroux, J.; Brasile, J.P.; Sirot, P.; Buckley, K.R.; Bedue, M.
2007-01-01
Most commercial PET Cyclotrons have targets mounted on or near the main cyclotron vacuum chamber. There is often little or no system capability for centering or focusing the extracted beam on target to achieve maximum production. This paper describes the ion-optics, design and development of a compact cost-effective beamline comprised of low activation and radiation resistant materials. The beamline, complete with suitable diagnostic devices, permits the extracted proton beam to be centered (X-Y steering magnet), and focused (quadrupole doublet) on target eliminating unnecessary beamspill and ensuring high production
APS beamline standard components handbook, Version 1.3
International Nuclear Information System (INIS)
Hahn, U.; Shu, D.; Kuzay, T.M.
1993-02-01
This Handbook in its current version (1.3) contains descriptions, specifications, and preliminary engineering design drawings for many of the standard components. The design status and schedules have been provided wherever possible. In the near future, the APS plans to update engineering drawings of identified standard beamline components and complete the Handbook. The completed version of this Handbook will become available to both the CATs and potential vendors. Use of standard components should result in major cost reductions for CATs in the areas of beamline design and construction
Status of the Nanoscopium scanning nanoprobe beamline of Synchrotron Soleil
Somogyi, A.; Medjoubi, K.; Kewish, C. M.; Leroux, V.; Ribbens, M.; Baranton, G.; Polack, F.; Samama, J. P.
2013-09-01
The Nanoscopium 155 m-long scanning nanoprobe beamline of Synchrotron Soleil (St Aubin, France) is dedicated to quantitative multi-modal imaging. Dedicated experimental stations, working in consecutive operation mode, will provide coherent scatter imaging and spectro-microscopy techniques in the 5-20 keV energy range for various user communities. Next to fast scanning, cryogenic cooling will reduce the radiation damage of sensitive samples during the measurements. Nanoscopium is in the construction phase, the first user experiments are expected in 2014. The main characteristics of the beamline and an overview of its status are given in this contribution.
Partial differential operators of elliptic type
Shimakura, Norio
1992-01-01
This book, which originally appeared in Japanese, was written for use in an undergraduate course or first year graduate course in partial differential equations and is likely to be of interest to researchers as well. This book presents a comprehensive study of the theory of elliptic partial differential operators. Beginning with the definitions of ellipticity for higher order operators, Shimakura discusses the Laplacian in Euclidean spaces, elementary solutions, smoothness of solutions, Vishik-Sobolev problems, the Schauder theory, and degenerate elliptic operators. The appendix covers such preliminaries as ordinary differential equations, Sobolev spaces, and maximum principles. Because elliptic operators arise in many areas, readers will appreciate this book for the way it brings together a variety of techniques that have arisen in different branches of mathematics.
On mod 2 and higher elliptic genera
International Nuclear Information System (INIS)
Liu Kefeng
1992-01-01
In the first part of this paper, we construct mod 2 elliptic genera on manifolds of dimensions 8k+1, 8k+2 by mod 2 index formulas of Dirac operators. They are given by mod 2 modular forms or mod 2 automorphic functions. We also obtain an integral formula for the mod 2 index of the Dirac operator. As a by-product we find topological obstructions to group actions. In the second part, we construct higher elliptic genera and prove some of their rigidity properties under group actions. In the third part we write down characteristic series for all Witten genera by Jacobi theta-functions. The modular property and transformation formulas of elliptic genera then follow easily. We shall also prove that Krichever's genera, which come from integrable systems, can be written as indices of twisted Dirac operators for SU-manifolds. Some general discussions about elliptic genera are given. (orig.)
Distributed control of protein crystallography beamline 5.0 using CORBA
International Nuclear Information System (INIS)
Timossi, Chris
1999-01-01
The Protein Crystallography Beamline at Berkeley Lab's Advanced Light Source is a facility that is being used to solve the structure of proteins. The software that is being used to control this beamline uses Java for user interface applications which communicate via CORBA with workstations that control the beamline hardware. We describe the software architecture for the beamline and our experiences after two years of operation
The Pharmaceutical Industry Beamline of Pharmaceutical Consortium for Protein Structure Analysis
International Nuclear Information System (INIS)
Nishijima, Kazumi; Katsuya, Yoshio
2002-01-01
The Pharmaceutical Industry Beamline was constructed by the Pharmaceutical Consortium for Protein Structure Analysis which was established in April 2001. The consortium is composed of 22 pharmaceutical companies affiliating with the Japan Pharmaceutical Manufacturers Association. The beamline is the first exclusive on that is owned by pharmaceutical enterprises at SPring-8. The specification and equipments of the Pharmaceutical Industry Beamline is almost same as that of RIKEN Structural Genomics Beamline I and II. (author)
Constructing elliptic curves from Galois representations
Snowden, Andrew; Tsimerman, Jacob
2017-01-01
Given a non-isotrivial elliptic curve over an arithmetic surface, one obtains a lisse $\\ell$-adic sheaf of rank two over the surface. This lisse sheaf has a number of straightforward properties: cyclotomic determinant, finite ramification, rational traces of Frobenius, and somewhere not potentially good reduction. We prove that any lisse sheaf of rank two possessing these properties comes from an elliptic curve.
Elliptic and parabolic equations for measures
Energy Technology Data Exchange (ETDEWEB)
Bogachev, Vladimir I [M. V. Lomonosov Moscow State University, Moscow (Russian Federation); Krylov, Nikolai V [University of Minnesota, Minneapolis, MN (United States); Roeckner, Michael [Universitat Bielefeld, Bielefeld (Germany)
2009-12-31
This article gives a detailed account of recent investigations of weak elliptic and parabolic equations for measures with unbounded and possibly singular coefficients. The existence and differentiability of densities are studied, and lower and upper bounds for them are discussed. Semigroups associated with second-order elliptic operators acting in L{sup p}-spaces with respect to infinitesimally invariant measures are investigated. Bibliography: 181 titles.
Description and calibration beamline SEM/Ion Chamber Current Digitizer
International Nuclear Information System (INIS)
Schoo, D.
1994-05-01
This report discusses the following on beamline SEM/ion chamber current digitizers: Module description; testing and calibration; common setup procedures; summary of fault indications and associated causes; summary of input and output connections; SEM conversion constant table; ion chamber conversion constant table; hexadecimal to decimal conversion table; and schematic diagram
The Nanoscience Beamline (I06) at Diamond Light Source
International Nuclear Information System (INIS)
Dhesi, S. S.; Cavill, S. A.; Potenza, A.; Marchetto, H.; Mott, R. A.; Steadman, P.; Peach, A.; Shepherd, E. L.; Ren, X.; Wagner, U. H.; Reininger, R.
2010-01-01
The Nanoscience beamline (I06) is one of seven Diamond Phase-I beamlines which has been operational since January 2007 delivering polarised soft x-rays, for a PhotoEmission Electron Microscope (PEEM) and branchline, in the energy range 80-2100 eV. The beamline is based on a collimated plane grating monochromator with sagittal focusing elements, utilising two APPLE II helical undulator sources, and has been designed for high flux density at the PEEM sample position. A ∼5 μm (σ) diameter beam is focussed onto the sample in the PEEM allowing a range of experiments using x-ray absorption spectroscopy (XAS), x-ray magnetic circular dichroism (XMCD) and x-ray magnetic linear dichroism (XMLD) as contrast mechanisms. The beamline is also equipped with a branchline housing a 6T superconducting magnet for XMCD and XMLD experiments. The magnet is designed to move on and off the branchline which allows a diverse range of experiments.
Optical design study of the PEARL beamline at SLS
Czech Academy of Sciences Publication Activity Database
Oberta, Peter; Flechsig, U.; Muntwiler, M.; Quitmann, C.
2011-01-01
Roč. 635, č. 1 (2011), s. 116-120 ISSN 0168-9002 Institutional research plan: CEZ:AV0Z10100522 Keywords : PGM * beamline design * photo-emision Subject RIV: BH - Optics, Masers, Lasers Impact factor: 1.207, year: 2011
Aberration analysis calculations for synchrotron radiation beamline design
International Nuclear Information System (INIS)
McKinney, W.R.; Howells, M.; Padmore, H.A.
1997-09-01
The application of ray deviation calculations based on aberration coefficients for a single optical surface for the design of beamline optical systems is reviewed. A systematic development is presented which allows insight into which aberration may be causing the rays to deviate from perfect focus. A new development allowing analytical calculation of line shape is presented
Remote access and automation of SPring-8 MX beamlines
Energy Technology Data Exchange (ETDEWEB)
Ueno, Go, E-mail: ueno@spring8.or.jp; Hikima, Takaaki; Yamashita, Keitaro; Hirata, Kunio; Yamamoto, Masaki [RIKEN SPring-8 Center, 1-1-1 Koto, Sayo-cho, Sayo-gun, Hyogo 679-5148 JAPAN (Japan); Hasegawa, Kazuya; Murakami, Hironori; Furukawa, Yukito; Mizuno, Nobuhiro; Kumasaka, Takashi [SPring-8/JASRI, 1-1-1 Koto, Sayo-cho, Sayo-gun, Hyogo 679-5198 JAPAN (Japan)
2016-07-27
At SPring-8 MX beamlines, a remote access system has been developed and started user operation in 2010. The system has been developed based on an automated data collection and data management architecture utilized for the confirmed scheme of SPring-8 mail-in data collection. Currently, further improvement to the remote access and automation which covers data processing and analysis are being developed.
Beamline standard component designs for the Advanced Photon Source
International Nuclear Information System (INIS)
Shu, D.; Barraza, J.; Brite, C.; Chang, J.; Sanchez, T.; Tcheskidov, V.; Kuzay, T.M.
1994-01-01
The Advanced Photon Source (APS) has initiated a design standardization and modularization activity for the APS synchrotron radiation beamline components. These standard components are included in components library, sub-components library and experimental station library. This paper briefly describes these standard components using both technical specifications and side view drawings
Trichromatic concept at the SPring-8 RIKEN beamline I
International Nuclear Information System (INIS)
Yamamoto, Masaki; Kumasaka, Takashi; Ueki, Tatzuo
1998-01-01
At the SPring-8, RIKEN beamline I has been designed and developed for structural biology research by the Institute of Physical and Chemical Research (RIKEN). RIKEN beamline I consists of two experimental stations, protein crystallography (PX) and small-angle X-ray scattering (SAXS). Both experiments can be carried out simultaneously, with dichromatic synchrotron radiation emitted from two coaxial undulators with vertical polarization. The branched beams are generated by a transparent diamond crystal. With synchrotron radiation, the multi-wavelength anomalous diffraction (MAD) method, which gives phases from a single anomalous scatterer, has been developed. Anomalous scattering contributes a small portion of diffraction intensity so that the accuracy of intensity data is definitely important. The PX branch of RIKEN beamline I has been designed based on a 'trichromatic concept' to optimize for the MAD data collection. This concept is that three kinds of intensity data sets with three different wavelengths are taken quasi-simultaneously for the single protein crystal without changing any setting by 'trichromator'. The main feature of this concept is to minimize systematic errors in the measurements of anomalous diffraction for the MAD method. The construction of RIKEN beamline I had been progressed satisfactorily until June 1997. The initial commissioning successfully provided the three different monochromatized undulator beams were successfully observed on the phosphor screen, which located at the near end of the trichromator. (author)
The High Energy Materials Science Beamline (HEMS) at PETRA III
International Nuclear Information System (INIS)
Schell, Norbert; King, Andrew; Beckmann, Felix; Ruhnau, Hans-Ulrich; Kirchhof, Rene; Kiehn, Ruediger; Mueller, Martin; Schreyer, Andreas
2010-01-01
The HEMS Beamline at the German high-brilliance synchrotron radiation storage ring PETRA III is fully tunable between 30 and 250 keV and optimized for sub-micrometer focusing. Approximately 70 % of the beamtime will be dedicated to Materials Research. Fundamental research will encompass metallurgy, physics and chemistry with first experiments planned for the investigation of the relationship between macroscopic and micro-structural properties of polycrystalline materials, grain-grain-interactions, and the development of smart materials or processes. For this purpose a 3D-microsctructure-mapper has been designed. Applied research for manufacturing process optimization will benefit from high flux in combination with ultra-fast detector systems allowing complex and highly dynamic in-situ studies of micro-structural transformations, e.g. during welding processes. The beamline infrastructure allows accommodation of large and heavy user provided equipment. Experiments targeting the industrial user community will be based on well established techniques with standardized evaluation, allowing full service measurements, e.g. for tomography and texture determination. The beamline consists of a five meter in-vacuum undulator, a general optics hutch, an in-house test facility and three independent experimental hutches working alternately, plus additional set-up and storage space for long-term experiments. HEMS is under commissioning as one of the first beamlines running at PETRA III.
High pressure XAFS experiments at the XAFS beamline, INDUS-2
International Nuclear Information System (INIS)
Ramanan, Nitya; Lahiri, Debdutta; Garg, Nandini; Sharma, Surinder M.; Bhattacharyya, D.; Jha, S.N.; Sahoo, N.K.
2011-01-01
The dispersive EXAFS beamline at the INDUS-2 synchrotron source, RRCAT, Indore uses a bent Si (111) crystal as a dispersive-cum-focusing element and a position sensitive CCD detector to enable instantaneous measurement of the whole EXAFS spectrum around the absorption edge of a particular atom. One of the proposed activities with this beamline is the characterization of amorphous materials under high pressure. Polychromator-based beamline is ideal for high pressure studies using Diamond Anvil Cell (DAC) with ∼ 50 μm sample size. Larger spot size would give rise to unwanted diffraction peaks from diamond, superimposed on the XAFS data. Micro-focusing by polychromator crystal and absence of its mechanical movement (unlike monochromator-based scanning-mode beamlines), during data collection, lead to required focal spot stability for DAC experiments. Currently, the theoretically determined spot size (Horizontal x vertical) varies between 17 x 137 μm and 37 x 142 μm for the X-ray energy range 5 keV-20 keV. To reduce the vertical spot size to <50 μm, we have designed an additional focusing mirror between the polychromator and sample position. The mirror, fabricated at SESO, France will be installed shortly. Meanwhile, we have carried out preliminary XAFS experiments on Sr-compounds at ∼16 keV, under ambient conditions and inside diamond anvil cell, in order to assess the signal intensity and quality. We have obtained reasonably good signal. (author)
LNLS soft x-ray spectroscopy (SXS) beamline
International Nuclear Information System (INIS)
Tolentino, Helio; Rocha, Milton C.; Tamura, Edilson; Cezar, Julio C.; Vicentin, Flavio C.; Giles, Carlos; Compagnon-Cailhol, Valerie; Abbate, Miguel; Cruz, Daniela Z.N.; Mocellin, Alexandra
1996-01-01
The Soft X-ray Spectroscopy beamline will be dedicated to the study of structural, electronic and magnetic properties of materials by using photoabsorption and photoemission techniques, X-ray dischroism will be used to study magnetism of transition metals and rare earths compounds. This beamline is one of the first seven beamlines which were decided to start operation along with the storage ring. Part of the beamline - mostly importations - has been granted by fundings from the state of Sao Paulo (Fapesp). The electron energy analyser came through EEC from a cooperation with a French group at LURE. All components of the beamline are either constructed or bougth and being mounted at the storage ring. The monochromator has already been commissioned under UHV, attaining the specification of 5x10 -9 Torr. To cover the whole energy range, from 800 eV up to 4000 eV, many crystals have been bought, cut and tested. The mirror has been specified in order to focus the source in both directions. Simulations using the Shadow code (source simulation and ray tracing technique) were performed in order to optimize the performance of the optics. We expert to focus 10 mrad down to a spot of 3.0x1.5 mm 2 . The mirror chamber has already been constructed and commissioned under UHV conditions (pressure -9 Torr). The mechanics (mechanical feedthroughs, stability, etc..) has been tested using an X-ray source and has been approved. The experimental chamber has already been used for photoemission experiments using a conventional AL/Mg X-ray source. Many results have been obtained and two master thesis have been performed using this set-up. (author)
The INE-Beamline for actinide science at ANKA
Rothe, J.; Butorin, S.; Dardenne, K.; Denecke, M. A.; Kienzler, B.; Löble, M.; Metz, V.; Seibert, A.; Steppert, M.; Vitova, T.; Walther, C.; Geckeis, H.
2012-04-01
Since its inauguration in 2005, the INE-Beamline for actinide research at the synchrotron source ANKA (KIT North Campus) provides dedicated instrumentation for x-ray spectroscopic characterization of actinide samples and other radioactive materials. R&D work at the beamline focuses on various aspects of nuclear waste disposal within INE's mission to provide the scientific basis for assessing long-term safety of a final nuclear waste repository. The INE-Beamline is accessible for the actinide and radiochemistry community through the ANKA proposal system and the European Union Integrated Infrastructure Initiative ACTINET-I3. Experiments with activities up to 1 × 10+6 times the European exemption limit are feasible within a safe but flexible containment concept. Measurements with monochromatic radiation are performed at photon energies varying between ˜2.1 keV (P K-edge) and ˜25 keV (Pd K-edge), including the lanthanide L-edges and the actinide M- and L3-edges up to Cf. The close proximity of the INE-Beamline to INE controlled area labs offers infrastructure unique in Europe for the spectroscopic and microscopic characterization of actinide samples. The modular beamline design enables sufficient flexibility to adapt sample environments and detection systems to many scientific questions. The well-established bulk techniques x-ray absorption fine structure (XAFS) spectroscopy in transmission and fluorescence mode have been augmented by advanced methods using a microfocused beam, including (confocal) XAFS/x-ray fluorescence detection and a combination of (micro-)XAFS and (micro-)x-ray diffraction. Additional instrumentation for high energy-resolution x-ray emission spectroscopy has been successfully developed and tested.
The INE-Beamline for actinide science at ANKA
International Nuclear Information System (INIS)
Rothe, J.; Dardenne, K.; Denecke, M. A.; Kienzler, B.; Loeble, M.; Metz, V.; Steppert, M.; Vitova, T.; Geckeis, H.; Butorin, S.; Seibert, A.; Walther, C.
2012-01-01
Since its inauguration in 2005, the INE-Beamline for actinide research at the synchrotron source ANKA (KIT North Campus) provides dedicated instrumentation for x-ray spectroscopic characterization of actinide samples and other radioactive materials. R and D work at the beamline focuses on various aspects of nuclear waste disposal within INE's mission to provide the scientific basis for assessing long-term safety of a final nuclear waste repository. The INE-Beamline is accessible for the actinide and radiochemistry community through the ANKA proposal system and the European Union Integrated Infrastructure Initiative ACTINET-I3. Experiments with activities up to 1 x 10 +6 times the European exemption limit are feasible within a safe but flexible containment concept. Measurements with monochromatic radiation are performed at photon energies varying between ∼2.1 keV (P K-edge) and ∼25 keV (Pd K-edge), including the lanthanide L-edges and the actinide M- and L3-edges up to Cf. The close proximity of the INE-Beamline to INE controlled area labs offers infrastructure unique in Europe for the spectroscopic and microscopic characterization of actinide samples. The modular beamline design enables sufficient flexibility to adapt sample environments and detection systems to many scientific questions. The well-established bulk techniques x-ray absorption fine structure (XAFS) spectroscopy in transmission and fluorescence mode have been augmented by advanced methods using a microfocused beam, including (confocal) XAFS/x-ray fluorescence detection and a combination of (micro-)XAFS and (micro-)x-ray diffraction. Additional instrumentation for high energy-resolution x-ray emission spectroscopy has been successfully developed and tested.
Fabrication of elliptical SRF cavities
Singer, W.
2017-03-01
The technological and metallurgical requirements of material for high-gradient superconducting cavities are described. High-purity niobium, as the preferred metal for the fabrication of superconducting accelerating cavities, should meet exact specifications. The content of interstitial impurities such as oxygen, nitrogen, and carbon must be below 10 μg g-1. The hydrogen content should be kept below 2 μg g-1 to prevent degradation of the quality factor (Q-value) under certain cool-down conditions. The material should be free of flaws (foreign material inclusions or cracks and laminations) that can initiate a thermal breakdown. Traditional and alternative cavity mechanical fabrication methods are reviewed. Conventionally, niobium cavities are fabricated from sheet niobium by the formation of half-cells by deep drawing, followed by trim machining and electron beam welding. The welding of half-cells is a delicate procedure, requiring intermediate cleaning steps and a careful choice of weld parameters to achieve full penetration of the joints. A challenge for a welded construction is the tight mechanical and electrical tolerances. These can be maintained by a combination of mechanical and radio-frequency measurements on half-cells and by careful tracking of weld shrinkage. The main aspects of quality assurance and quality management are mentioned. The experiences of 800 cavities produced for the European XFEL are presented. Another cavity fabrication approach is slicing discs from the ingot and producing cavities by deep drawing and electron beam welding. Accelerating gradients at the level of 35-45 MV m-1 can be achieved by applying electrochemical polishing treatment. The single-crystal option (grain boundary free) is discussed. It seems that in this case, high performance can be achieved by a simplified treatment procedure. Fabrication of the elliptical resonators from a seamless pipe as an alternative is briefly described. This technology has yielded good
Beamline 9.3.2 - a high-resolution, bend-magnet beamline with circular polarization capability
Energy Technology Data Exchange (ETDEWEB)
Moler, E.J.; Hussain, Z.; Howells, M.R. [Lawrence Berkeley National Lab., CA (United States)] [and others
1997-04-01
Beamline 9.3.2 is a high resolution, SGM beamline on an ALS bending magnet with access to photon energies from 30-1500 eV. Features include circular polarization capability, a rotating chamber platform that allows switching between experiments without breaking vacuum, an active feedback system that keeps the beam centered on the entrance slit of the monochromator, and a bendable refocusing mirror. The beamline optics consist of horizontally and vertically focussing mirrors, a Spherical Grating Monochromator (SGM) with movable entrance and exit slits, and a bendable refocussing mirror. In addition, a movable aperature has been installed just upstream of the vertically focussing mirror which can select the x-rays above or below the plane of the synchrotron storage ring, allowing the user to select circularly or linearly polarized light. Circularly polarized x-rays are used to study the magnetic properties of materials. Beamline 9.3.2 can supply left and right circularly polarized x-rays by a computer controlled aperture which may be placed above or below the plane of the synchrotron storage ring. The degree of linear and circular polarization has been measured and calibrated.
The Project for the High Energy Materials Science Beamline at Petra III
International Nuclear Information System (INIS)
Martins, R. V.; Lippmann, T.; Beckmann, F.; Schreyer, A.
2007-01-01
The high energy materials science beamline will be among the first fourteen beamlines planned to be operational in 2009 at the new third generation synchrotron light source Petra III at DESY, Germany. The operation and funding of this beamline is assured by GKSS. 70% of the beamline will be dedicated to materials science. The remaining 30% are reserved for physics and are covered by DESY. The materials science activities will be concentrating on three intersecting topics which are industrial, applied, and fundamental research. The beamline will combine three main features: Firstly, the high flux, fast data acquisition systems, and the beamline infrastructure will allow carrying out complex and highly dynamic in-situ experiments. Secondly, a high flexibility in beam shaping will be available, fully exploiting the high brilliance of the source. Thirdly, the beamline will provide the possibility to merge in one experiment different analytical techniques such as diffraction and tomography
Ultraluminous Infrared Mergers: Elliptical Galaxies in Formation?
Genzel, R.; Tacconi, L. J.; Rigopoulou, D.; Lutz, D.; Tecza, M.
2001-12-01
We report high-quality near-IR spectroscopy of 12 ultraluminous infrared galaxy mergers (ULIRGs). Our new VLT and Keck data provide ~0.5" resolution, stellar and gas kinematics of these galaxies, most of which are compact systems in the last merger stages. We confirm that ULIRG mergers are ``ellipticals in formation.'' Random motions dominate their stellar dynamics, but significant rotation is common. Gasdynamics and stellar dynamics are decoupled in most systems. ULIRGs fall on or near the fundamental plane of hot stellar systems, and especially on its less evolution-sensitive, reff-σ projection. The ULIRG velocity dispersion distribution, their location in the fundamental plane, and their distribution of vrotsini/σ closely resemble those of intermediate-mass (~L*), elliptical galaxies with moderate rotation. As a group ULIRGs do not resemble giant ellipticals with large cores and little rotation. Our results are in good agreement with other recent studies indicating that disky ellipticals with compact cores or cusps can form through dissipative mergers of gas-rich disk galaxies while giant ellipticals with large cores have a different formation history. Based on observations at the European Southern Observatory, Chile (ESO 65.N-0266, 65.N-0289), and on observations at the W. M. Keck Observatory, which is operated as a scientific partnership among the California Institute of Technology, The University of California, and the National Aeronautics and Space Administration. The Keck Observatory was made possible by the general financial support by the W. M. Keck Foundation.
Hot interstellar matter in elliptical galaxies
Kim, Dong-Woo
2012-01-01
Based on a number of new discoveries resulting from 10 years of Chandra and XMM-Newton observations and corresponding theoretical works, this is the first book to address significant progress in the research of the Hot Interstellar Matter in Elliptical Galaxies. A fundamental understanding of the physical properties of the hot ISM in elliptical galaxies is critical, because they are directly related to the formation and evolution of elliptical galaxies via star formation episodes, environmental effects such as stripping, infall, and mergers, and the growth of super-massive black holes. Thanks to the outstanding spatial resolution of Chandra and the large collecting area of XMM-Newton, various fine structures of the hot gas have been imaged in detail and key physical quantities have been accurately measured, allowing theoretical interpretations/predictions to be compared and tested against observational results. This book will bring all readers up-to-date on this essential field of research.
Nonlinear elliptic equations of the second order
Han, Qing
2016-01-01
Nonlinear elliptic differential equations are a diverse subject with important applications to the physical and social sciences and engineering. They also arise naturally in geometry. In particular, much of the progress in the area in the twentieth century was driven by geometric applications, from the Bernstein problem to the existence of Kähler-Einstein metrics. This book, designed as a textbook, provides a detailed discussion of the Dirichlet problems for quasilinear and fully nonlinear elliptic differential equations of the second order with an emphasis on mean curvature equations and on Monge-Ampère equations. It gives a user-friendly introduction to the theory of nonlinear elliptic equations with special attention given to basic results and the most important techniques. Rather than presenting the topics in their full generality, the book aims at providing self-contained, clear, and "elementary" proofs for results in important special cases. This book will serve as a valuable resource for graduate stu...
Elliptical cross section fuel rod study II
International Nuclear Information System (INIS)
Taboada, H.; Marajofsky, A.
1996-01-01
In this paper it is continued the behavior analysis and comparison between cylindrical fuel rods of circular and elliptical cross sections. Taking into account the accepted models in the literature, the fission gas swelling and release were studied. An analytical comparison between both kinds of rod reveals a sensible gas release reduction in the elliptical case, a 50% swelling reduction due to intragranular bubble coalescence mechanism and an important swelling increase due to migration bubble mechanism. From the safety operation point of view, for the same linear power, an elliptical cross section rod is favored by lower central temperatures, lower gas release rates, greater gas store in ceramic matrix and lower stored energy rates. (author). 6 refs., 8 figs., 1 tab
MONO: A program to calculate synchrotron beamline monochromator throughputs
International Nuclear Information System (INIS)
Chapman, D.
1989-01-01
A set of Fortran programs have been developed to calculate the expected throughput of x-ray monochromators with a filtered synchrotron source and is applicable to bending magnet and wiggler beamlines. These programs calculate the normalized throughput and filtered synchrotron spectrum passed by multiple element, flat un- focussed monochromator crystals of the Bragg or Laue type as a function of incident beam divergence, energy and polarization. The reflected and transmitted beam of each crystal is calculated using the dynamical theory of diffraction. Multiple crystal arrangements in the dispersive and non-dispersive mode are allowed as well as crystal asymmetry and energy or angle offsets. Filters or windows of arbitrary elemental composition may be used to filter the incident synchrotron beam. This program should be useful to predict the intensities available from many beamline configurations as well as assist in the design of new monochromator and analyzer systems. 6 refs., 3 figs
Indus-I beamlines for condensed matter physics
International Nuclear Information System (INIS)
Nandedkar, R.V.
2001-01-01
Full text: A 450 MeV electron storage ring Indus-I is now operational. This storage ring gives synchrotron radiation in soft x-ray vacuum ultra violet (VUV) and to visible region. On this storage ring six beamlines are now being set up for atomic and molecular spectroscopy experiments, solid state spectroscopy experiments and soft and VUV reflectivity experiments. In this talk, present status of beamlines which condense matter physicists will be interested in will be given along with some commissioning experiments. These beam lines are based on a toroidal grating monochromators in the range 40 - 1000 A with moderate energy resolution. Some experiments which can be conducted using these beam lines will be discussed
Intensity interferometry at the X13A undulator beamline
International Nuclear Information System (INIS)
Gluskin, E.; McNulty, I.; Yang, L.; Randall, K.J.; Johnson, E.D.
1993-01-01
We are constructing a soft x-ray intensity interferometer and an undulator based beamline to demonstrate intensity interferometry in the x-ray region. The 10-period soft x-ray undulator at the NSLS provides the necessary coherent flux; the X13A beamline is designed to preserve the spatial coherence of the bright x-ray beam and provide sufficient temporal coherence using a horizontally deflecting spherical grating monochromator. Using the interferometer, which consists of an array of small slits, a wedge-shaped beamsplitter and two fast microchannel plate detectors, we expect to measure the spatial coherence of the undulator beam and therefore the size of the source in the vertical plane. Details of the bean-dine design and the interferometer experiment are discussed
Magnetic shielding tests for MFTF-B neutral beamlines
International Nuclear Information System (INIS)
Kerns, J.; Fabyan, J.; Wood, R.; Koger, P.
1983-01-01
A test program to determine the effectiveness of various magnetic shielding designs for MFTF-B beamlines was established at Lawrence Livermore National Laboratory (LLNL). The proposed one-tenth-scale shielding-design models were tested in a uniform field produced by a Helmholtz coil pair. A similar technique was used for the MFTF source-injector assemblies, and the model test results were confirmed during the Technology Demonstration in 1982. The results of these tests on shielding designs for MFTF-B had an impact on the beamline design for MFTF-B. The iron-core magnet and finger assembly originally proposed were replaced by a simple, air-core, race-track-coil, bending magnet. Only the source injector needs to be magnetically shielded from the fields of approximately 400 gauss
Electromagnetic Invisibility of Elliptic Cylinder Cloaks
International Nuclear Information System (INIS)
Kan, Yao; Chao, Li; Fang, Li
2008-01-01
Structures with unique electromagnetic properties are designed based on the approach of spatial coordinate transformations of Maxwell's equations. This approach is applied to scheme out invisible elliptic cylinder cloaks, which provide more feasibility for cloaking arbitrarily shaped objects. The transformation expressions for the anisotropic material parameters and the field distribution are derived. The cloaking performances of ideal and lossy elliptic cylinder cloaks are investigated by finite element simulations. It is found that the cloaking performance will degrade in the forward direction with increasing loss. (fundamental areas of phenomenology (including applications))
Quantum W-algebras and elliptic algebras
International Nuclear Information System (INIS)
Feigin, B.; Kyoto Univ.; Frenkel, E.
1996-01-01
We define a quantum W-algebra associated to sl N as an associative algebra depending on two parameters. For special values of the parameters, this algebra becomes the ordinary W-algebra of sl N , or the q-deformed classical W-algebra of sl N . We construct free field realizations of the quantum W-algebras and the screening currents. We also point out some interesting elliptic structures arising in these algebras. In particular, we show that the screening currents satisfy elliptic analogues of the Drinfeld relations in U q (n). (orig.)
Nonlinear elliptic partial differential equations an introduction
Le Dret, Hervé
2018-01-01
This textbook presents the essential parts of the modern theory of nonlinear partial differential equations, including the calculus of variations. After a short review of results in real and functional analysis, the author introduces the main mathematical techniques for solving both semilinear and quasilinear elliptic PDEs, and the associated boundary value problems. Key topics include infinite dimensional fixed point methods, the Galerkin method, the maximum principle, elliptic regularity, and the calculus of variations. Aimed at graduate students and researchers, this textbook contains numerous examples and exercises and provides several comments and suggestions for further study.
Elliptic Tales Curves, Counting, and Number Theory
Ash, Avner
2012-01-01
Elliptic Tales describes the latest developments in number theory by looking at one of the most exciting unsolved problems in contemporary mathematics--the Birch and Swinnerton-Dyer Conjecture. The Clay Mathematics Institute is offering a prize of 1 million to anyone who can discover a general solution to the problem. In this book, Avner Ash and Robert Gross guide readers through the mathematics they need to understand this captivating problem. The key to the conjecture lies in elliptic curves, which are cubic equations in two variables. These equations may appear simple, yet they arise from
Abundance ratios in dwarf elliptical galaxies
Şen, Ş.; Peletier, R. F.; Boselli, A.; den Brok, M.; Falcón-Barroso, J.; Hensler, G.; Janz, J.; Laurikainen, E.; Lisker, T.; Mentz, J. J.; Paudel, S.; Salo, H.; Sybilska, A.; Toloba, E.; van de Ven, G.; Vazdekis, A.; Yesilyaprak, C.
2018-04-01
We determine abundance ratios of 37 dwarf ellipticals (dEs) in the nearby Virgo cluster. This sample is representative of the early-type population of galaxies in the absolute magnitude range -19.0 originate from late-type dwarfs or small spirals. Na-yields appear to be very metal-dependent, in agreement with studies of giant ellipticals, probably due to the large dependence on the neutron-excess in stars. We conclude that dEs have undergone a considerable amount of chemical evolution, they are therefore not uniformly old, but have extended SFH, similar to many of the Local Group galaxies.
First transmission of electrons and ions through the KATRIN beamline
Czech Academy of Sciences Publication Activity Database
Arenz, M.; Dragoun, Otokar; Kovalík, Alojz; Lebeda, Ondřej; Ryšavý, Miloš; Sentkerestiová, Jana; Suchopár, Martin; Vénos, Drahoslav
2018-01-01
Roč. 13, č. 4 (2018), č. článku P04020. ISSN 1748-0221 R&D Projects: GA MŠk LM2015056; GA MŠk LTT18021 Institutional support: RVO:61389005 Keywords : ion sources * electron beam * detector control systems * beam-line instrumentation * spectrometers Subject RIV: BG - Nuclear, Atomic and Molecular Physics, Colliders OBOR OECD: Nuclear physics Impact factor: 1.220, year: 2016
Measurement of intensity distribution of CSR in LEBRA PXR beamline
International Nuclear Information System (INIS)
Nakao, Keisuke; Sakai, Takeshi; Hayakawa, Ken; Tanaka, Toshinari; Hayakawa, Yasushi; Nogami, Kyoko; Inagaki, Manabu; Sei, Norihiro
2014-01-01
Last year, the intensity of Coherent Synchrotron Radiation (CSR) in LEBRA PXR beamline was measured. As a result, it turned out that the intensity of CSR was stronger than anticipation. It is suggested that Coherent Edge Radiation (CER) is mixed with CSR. Then, in order to confirm whether CER is contained, the intensity distribution of CSR was measured. The result of the experiment is reported in this paper. (author)
Surface science station of the infrared beamline at SPring-8
International Nuclear Information System (INIS)
Sakurai, M.; Moriwaki, T.; Kimura, H.; Nishida, S.; Nanba, T.
2001-01-01
An experimental station for surface science has been constructed at the infrared beamline (BL43IR) of SPring-8, Japan. The station utilizes synchrotron radiation in the energy range of 100-20000 cm -1 to perform infrared reflection absorption spectroscopy (IRAS) of surfaces. It consists of an experimental section, a preparation chamber, gas handling equipment and a pair of focusing optics. In situ observation of vibrational spectra is possible using both IRAS and high-resolution electron energy loss spectroscopy
Successful test of SPS-to-LHC beamline
Maximilien Brice
2004-01-01
On 23 October there was great excitement in the Prevessin control room when, on the first attempt, a beam passed over 2.5 km down the new SPS-to-LHC transfer line, TI8, to within a few metres of the LHC tunnel. Members of the AB, AT and TS departments involved in the beamline and its test, celebrate their success with the Director General, Robert Aymar, and the LHC Project Leader, Lyn Evans.
Successful test of SPS-to-LHC beamline
2004-01-01
On 23 October there was great excitement in the Prevessin control room when, on the first attempt, a beam passed over 2.5 km down the new SPS-to-LHC transfer line, TI8, to within a few metres of the LHC tunnel. Above: members of the AB, AT and TS departments involved in the beamline and its test, celebrate their success with the Director General, Robert Aymar, and the LHC Project Leader, Lyn Evans.
Beamlines on Indus-1 and Indus-2: present status
International Nuclear Information System (INIS)
Lodha, G.S.; Deb, S.K.
2010-01-01
Indus-1 (450 MeV) is an efficient synchrotron radiation (SR) source in the soft X-ray/vacuum ultra violet region of the electromagnetic spectrum. For Indus-1 the higher order energy contamination in soft X-ray region, heat load and radiation safety problems are also significantly low. At present five beamlines are operational. This SR source is a national science facility being used by various research group across the country. Strong efforts are underway to increase the user base of Indus-1. The talk presents some of the recent studies carried out using Indus-1. Indus-2 (2.5 GeV) synchrotron source is one of the most important accelerator based science facilities being setup in India. A few beamlines have been commissioned and are being used by researchers from different institutes. This talk gives present status of the various beamlines and experimental stations on Indus-2. It is envisaged that the atomic and molecular science community can actively participate in planning experiments on Indus-1 and Indus-2 and setup experimental stations on Indus-2. (author)
Pulsed beam tests at the SANAEM RFQ beamline
Turemen, G.; Akgun, Y.; Alacakir, A.; Kilic, I.; Yasatekin, B.; Ergenlik, E.; Ogur, S.; Sunar, E.; Yildiz, V.; Ahiska, F.; Cicek, E.; Unel, G.
2017-07-01
A proton beamline consisting of an inductively coupled plasma (ICP) source, two solenoid magnets, two steerer magnets and a radio frequency quadrupole (RFQ) is developed at the Turkish Atomic Energy Authority’s (TAEA) Saraykoy Nuclear Research and Training Center (SNRTC-SANAEM) in Ankara. In Q4 of 2016, the RFQ was installed in the beamline. The high power tests of the RF power supply and the RF transmission line were done successfully. The high power RF conditioning of the RFQ was performed recently. The 13.56 MHz ICP source was tested in two different conditions, CW and pulsed. The characterization of the proton beam was done with ACCTs, Faraday cups and a pepper-pot emittance meter. Beam transverse emittance was measured in between the two solenoids of the LEBT. The measured beam is then reconstructed at the entrance of the RFQ by using computer simulations to determine the optimum solenoid currents for acceptance matching of the beam. This paper will introduce the pulsed beam test results at the SANAEM RFQ beamline. In addition, the high power RF conditioning of the RFQ will be discussed.
Performance of a novel VUV bending magnet beamline
Song, Y F; Hsieh, T F; Huang, L R; Chung, S C; Cheng, N F; Hsiung, G Y; Wang, D J; Chen, C T; Tsang, K L
2001-01-01
A novel high resolution, high flux bending magnet beamline with an energy range from 5 to 40 eV has been constructed at SRRC. This Dragon-like beamline, which horizontally collects 50 mrad of synchrotron radiation from a bending magnet source, uses four cylindrical gratings with an included angle of 140 deg. and a movable curved exit slit. The average photon flux with an energy resolving power of 1000 is about 2x10 sup 1 sup 2 photons/s, which is among the highest of all existing VUV bending magnet beamlines. An energy resolving power of 24,000 at 6.8 eV has been obtained from the Schumann-Runge bands (B sup 3 limit construction operator in a limit construction/sum L: summation operator operator End lower limit of a limit construction u lower limit End limit End sup - /leftarrow/gets A: =leftward arrow X sup 3 limit construction operator in a limit construction/sum L: summation operator operator End lower limit of a limit construction g lower limit End limit End sup -) absorption spectra of O sub 2 gas. A pho...
ADLIB: A simple database framework for beamline codes
International Nuclear Information System (INIS)
Mottershead, C.T.
1993-01-01
There are many well developed codes available for beamline design and analysis. A significant fraction of each of these codes is devoted to processing its own unique input language for describing the problem. None of these large, complex, and powerful codes does everything. Adding a new bit of specialized physics can be a difficult task whose successful completion makes the code even larger and more complex. This paper describes an attempt to move in the opposite direction, toward a family of small, simple, single purpose physics and utility modules, linked by an open, portable, public domain database framework. These small specialized physics codes begin with the beamline parameters already loaded in the database, and accessible via the handful of subroutines that constitute ADLIB. Such codes are easier to write, and inherently organized in a manner suitable for incorporation in model based control system algorithms. Examples include programs for analyzing beamline misalignment sensitivities, for simulating and fitting beam steering data, and for translating among MARYLIE, TRANSPORT, and TRACE3D formats
The actinide beamline - A new AMS facility at ANTARES
International Nuclear Information System (INIS)
Hotchkis, M.A.C.; Lee, P.J.; Mino, N.
1998-01-01
At the ANTARES accelerator a new beamline has been commissioned, incorporating new magnetic and electrostatic analysers, to optimise the efficiency for Actinides detection by Accelerator Mass Spectrometry. The detection of Actinides, particularly the isotopic ratios of uranium and plutonium, provide unique signatures for nuclear safeguards purposes. We are currently engaged in a project to evaluate the application of AMS to the measurement of Actinides in environmental samples for nuclear safeguards. Measurement of 236 U is of particular interest as a means of tracing the anthropogenic component of uranium. 236 U is expected to be present in natural samples at an extremely low level ( 236 U: 238 U ratio ∼10 -10 ). It has recently been demonstrated that AMS has sufficient sensitivity to detect 236 U at this level. The principal components of the new beamline include: an electrostatic quadrupole, a 12 deg electrostatic deflector, a 90 deg electrostatic analyser, a multi-isotope detection system including ion counters and Faraday cups. The beamline was completed in September 1998 and initial tests have been performed with iodine samples
The vacuum interlock system for the PETRA III beamlines
Energy Technology Data Exchange (ETDEWEB)
Degenhardt, Markus; Hahn, Ulrich; Hesse, Mathias; Schulte-Schrepping, Horst [Deutsches Elektronen-Synchrotron (DESY), Hamburg (Germany)
2008-07-01
The storage ring PETRA at DESY in Hamburg is being reconstructed into the third generation source for synchrotron radiation, PETRA III. The up to 100 m long beamlines are large UHV-systems that guide the synchrotron radiation from the storage ring to the experiments. Each beamline will be equipped with a vacuum interlock system to ensure the safe operation of the vacuum components. In particular the task of the vacuum interlock is to prevent faulty operations that can cause a ventilation of the vacuum system or a damage of vacuum components by the high power synchrotron radiation beam. The interlock system will be implemented as a PLC that is connected to a distributed input/output layer via a field bus system. As a specialty, the PLC will be realised as a soft-PLC running on a PC with a real time windows operating system. Another specialty is the visualisation and remote control of the vacuum interlock system by means of a website. At the beamline the interlock will be operated via a touch panel that displays the visualisation website. Additionally, the interlock can be remotely operated from any location by opening the visualisation website with a browser. The interlock is protected against unauthorised operation by a login page. All relevant interlock data will be fed into the existing network-based archive system.
Ultrabroadband terahertz source and beamline based on coherent transition radiation
Directory of Open Access Journals (Sweden)
S. Casalbuoni
2009-03-01
Full Text Available Coherent transition radiation (CTR in the THz regime is an important diagnostic tool for analyzing the temporal structure of the ultrashort electron bunches needed in ultraviolet and x-ray free-electron lasers. It is also a powerful source of such radiation, covering an exceptionally broad frequency range from about 200 GHz to 100 THz. At the soft x-ray free-electron laser FLASH we have installed a beam transport channel for transition radiation (TR with the intention to guide a large fraction of the radiation to a laboratory outside the accelerator tunnel. The radiation is produced on a screen inside the ultrahigh vacuum beam pipe of the linac, coupled out through a diamond window and transported to the laboratory through an evacuated tube equipped with five focusing and four plane mirrors. The design of the beamline has been based on a thorough analysis of the generation of TR on metallic screens of limited size. The optical propagation of the radiation has been computed taking into account the effects of near-field (Fresnel diffraction. The theoretical description of the TR source is presented in the first part of the paper, while the design principles and the technical layout of the beamline are described in the second part. First experimental results demonstrate that the CTR beamline covers the specified frequency range and preserves the narrow time structure of CTR pulses emitted by short electron bunches.
Design and simulation of the nuSTORM pion beamline
Energy Technology Data Exchange (ETDEWEB)
Liu, A., E-mail: aoliu@fnal.gov; Neuffer, D.; Bross, A.
2015-11-21
The nuSTORM (neutrinos from STORed Muons) proposal presents a detailed design for a neutrino facility based on a muon storage ring, with muon decay in the production straight section of the ring providing well defined neutrino beams. The facility includes a primary high-energy proton beam line, a target station with pion production and collection, and a pion beamline for pion transportation and injection into a muon decay ring. The nuSTORM design uses “stochastic injection”, in which pions are directed by a chicane, referred to as the Orbit Combination Section (OCS), into the production straight section of the storage ring. Pions that decay within that straight section provide muons within the circulating acceptance of the ring. The design enables injection without kickers or a separate pion decay transport line. The beam line that the pions traverse before being extracted from the decay ring is referred to as the pion beamline. This paper describes the design and simulation of the pion beamline, and includes full beam dynamics simulations of the system.
Nanotomography endstation at the P05 beamline: Status and perspectives
Greving, I.; Ogurreck, M.; Marschall, F.; Last, A.; Wilde, F.; Dose, T.; Burmester, H.; Lottermoser, L.; Müller, M.; David, C.; Beckmann, F.
2017-06-01
The Imaging Beamline IBL/P05 at the DESY storage ring PETRA III, operated by the Helmholtz-Zentrum Geesthacht, has two dedicated endstations optimized for micro- and nanotomography experiments [1-3]. Here we present the status of the nanotomography endstation, highlight the latest instrumentation upgrades and present first experimental results. In particular in materials science, where structures with ceramics or metallic materials are of interest, X-ray energies of 15 keV and above are required even for sample sizes of several 10 μm in diameter. The P05 imaging beamline is dedicated to materials science and is designed to allow for imaging applications with X-ray energies of 10 to 50 keV. In addition to the full field X-ray microscopy setup, the layout of the nanotomography endstation allows switching to cone-beam configuration. Kinematics for X-ray optics like compound refractive lenses (CRLs), Fresnel zone plates (FZP) or beam-shaping optics are implemented and the installation of a Kirkpatrick Baez-mirror (KB mirror) system is foreseen at a later stage of the beamline development. Altogether this leads to a high flexibility of the nanotomography setup such that the instrument can be tailored to the specific experimental requirements of a range of sample systems.
Index profile measurement of asymmetrical elliptical preforms or fibers
Blitterswijk, van W.; Smit, M.K.
1987-01-01
An extension of the beam-deflection method to the case of elliptical preforms with eccentric core (asymmetrical elliptical preforms) is presented, which can be easily implemented on automatic measurement equipment
Conceptual design of NBI beamline for VEST plasma heating
Energy Technology Data Exchange (ETDEWEB)
Kim, T.S., E-mail: tskim@kaeri.re.kr; In, S.R.; Jeong, S.H.; Park, M.; Chang, D.H.; Jung, B.K.; Lee, K.W.
2016-11-01
Highlights: • VEST NBI injector is conceptually designed to support further VEST plasma experiment. • VEST NBI injector composed of 2 sets of 20 keV/25A magnetic cusp type bucket ion source, neutralizer ducts, electrostatic ion dumps, NB vessel with cryopump, and rotating calorimerter. • The vacuum vessel of the beamline is divided into two parts for high injection efficiency and different direction (co- and counter-current) of neutral beam injection. • An ion source for the VEST NBI system was also designed to deliver neutral hydrogen beams with a power of 0.3 MW. The plasma generator of the VEST NB ion source has modified TFTR bucket multi-cusp chamber. The plasma generator has twelve hair-pin shaped tungsten filaments used as a cathode and an arc chamber including a bucket and an electron dump which serve as anode. The accelerator system consists of three grids, each having extraction area of 100 mm × 320 mm and 64 shaped slits of 3 mm spacing. • The preliminary structure design and the layout of the main components of the injector have been completed. Simulation and calculation for optimization of the NB beamline design results prove that the parameters of ion source, neutralization efficiency (76%:95% equilibrium neutralization efficiency), and beam power transmission efficiency (higher than 90%) are in agreement with design targets of the VEST NB beamline. • This VEST NBI system will provide a neutral beam of ∼0.6 MW for both heating and current drive in torus plasma. - Abstract: A 10 m s-pulsed NBI (Neutral Beam Injection) system for VEST (Versatile Experiment Spherical Torus) plasma heating is designed to provide a beam power of more than 0.6 MW with 20 keV H° neutrals. The VEST NBI injector is composed of 2 sets of 20 keV/25A magnetic cusp type bucket ion source, neutralizer ducts, residual ion dump, NB vessel with a cryopump, and rotating calorimeter. The position and size of these beamline components are roughly determined with geometric
Conceptual design of NBI beamline for VEST plasma heating
International Nuclear Information System (INIS)
Kim, T.S.; In, S.R.; Jeong, S.H.; Park, M.; Chang, D.H.; Jung, B.K.; Lee, K.W.
2016-01-01
Highlights: • VEST NBI injector is conceptually designed to support further VEST plasma experiment. • VEST NBI injector composed of 2 sets of 20 keV/25A magnetic cusp type bucket ion source, neutralizer ducts, electrostatic ion dumps, NB vessel with cryopump, and rotating calorimerter. • The vacuum vessel of the beamline is divided into two parts for high injection efficiency and different direction (co- and counter-current) of neutral beam injection. • An ion source for the VEST NBI system was also designed to deliver neutral hydrogen beams with a power of 0.3 MW. The plasma generator of the VEST NB ion source has modified TFTR bucket multi-cusp chamber. The plasma generator has twelve hair-pin shaped tungsten filaments used as a cathode and an arc chamber including a bucket and an electron dump which serve as anode. The accelerator system consists of three grids, each having extraction area of 100 mm × 320 mm and 64 shaped slits of 3 mm spacing. • The preliminary structure design and the layout of the main components of the injector have been completed. Simulation and calculation for optimization of the NB beamline design results prove that the parameters of ion source, neutralization efficiency (76%:95% equilibrium neutralization efficiency), and beam power transmission efficiency (higher than 90%) are in agreement with design targets of the VEST NB beamline. • This VEST NBI system will provide a neutral beam of ∼0.6 MW for both heating and current drive in torus plasma. - Abstract: A 10 m s-pulsed NBI (Neutral Beam Injection) system for VEST (Versatile Experiment Spherical Torus) plasma heating is designed to provide a beam power of more than 0.6 MW with 20 keV H° neutrals. The VEST NBI injector is composed of 2 sets of 20 keV/25A magnetic cusp type bucket ion source, neutralizer ducts, residual ion dump, NB vessel with a cryopump, and rotating calorimeter. The position and size of these beamline components are roughly determined with geometric
MARS, a new beamline for radioactive matter studies at SOLEIL
International Nuclear Information System (INIS)
Solari, Pier Lorenzo; Schlutig, Sandrine; Hermange, Herve; Sitaud, Bruno
2009-01-01
MARS (Multi Analyses on Radioactive Samples) beamline is the hard X-ray bending magnet beamline dedicated to the study of radioactive matter of the new French synchrotron SOLEIL. The beamline, which has been built thanks to a close partnership and support by the CEA, has been designed to provide X-rays in the energy range of 3.5 keV to 35 keV. This allows to encompass M and L absorption edges of actinides, as well as K edges of transition metals (that are present in alloys and fuel claddings) up to heavy halogens, rare gases and alkalis (fission products in nuclear fuels). The MARS project aims to extend the possibilities of synchrotron based X-ray characterizations towards a wider variety of radioactive elements and a wider variety of techniques than what is currently available at other facilities. Thus, its specific and innovative infrastructure has been optimized in order to carry out analyses on materials with activities up to 18.5 GBq per sample for α and β emitters and 2 GBq for γ and n emitters. So, today, more than 70 different elements and more than 350 different isotopes have been proposed for studies on the beamline by the involved user community. The arrangement of the different elements in the optics hutch is based on an original scheme which permits to have two alternative optical configurations (monochromatic or dispersive) depending on the nature of experiments to be performed. At least three main techniques are progressively being proposed on the three complementary end-stations located in the experimental hutch: transmission and high resolution powder diffraction (TXRD and HRXRD), standard and dispersive X-ray absorption spectroscopy (XAS and EDXAS) and X-ray fluorescence (XRF). In addition, by using the KB optics, a micro-focused beam will be available on the second station of the monochromatic branch. The beamline is currently under commissioning. The first two experimental stations, using the monochromatic branch, are scheduled to be
Elliptic genus of singular algebraic varieties and quotients
Libgober, Anatoly
2018-02-01
This paper discusses the basic properties of various versions of the two-variable elliptic genus with special attention to the equivariant elliptic genus. The main applications are to the elliptic genera attached to non-compact GITs, including the theories regarding the elliptic genera of phases on N = 2 introduced in Witten (1993 Nucl. Phys. B 403 159-222).
Impedances in lossy elliptical vacuum chambers
International Nuclear Information System (INIS)
Piwinski, A.
1994-04-01
The wake fields of a bunched beam caused by the resistivity of the chamber walls are investigated for a vacuum chamber with elliptical cross section. The longitudinal and transverse impedances are calculated for arbitrary energies and for an arbitrary position of the beam in the chamber. (orig.)
Quasilinear infiltration from an elliptical cavity
Kuhlman, Kristopher L.; Warrick, Arthur W.
2008-08-01
We develop analytic solutions to the linearized steady-state Richards equation for head and total flowrate due to an elliptic cylinder cavity with a specified pressure head boundary condition. They are generalizations of the circular cylinder cavity solutions of Philip [Philip JR. Steady infiltration from circular cylindrical cavities. Soil Sci Soc Am J 1984;48:270-8]. The circular and strip sources are limiting cases of the elliptical cylinder solution, derived for both horizontally- and vertically-aligned ellipses. We give approximate rational polynomial expressions for total flowrate from an elliptical cylinder over a range of sizes and shapes. The exact elliptical solution is in terms of Mathieu functions, which themselves are generalizations of and computed from trigonometric and Bessel functions. The required Mathieu functions are computed from a matrix eigenvector problem, a modern approach that is straightforward to implement using available linear algebra libraries. Although less efficient and potentially less accurate than the iterative continued fraction approach, the matrix approach is simpler to understand and implement and is valid over a wider parameter range.
Carleman estimates for some elliptic systems
International Nuclear Information System (INIS)
Eller, M
2008-01-01
A Carleman estimate for a certain first order elliptic system is proved. The proof is elementary and does not rely on pseudo-differential calculus. This estimate is used to prove Carleman estimates for the isotropic Lame system as well as for the isotropic Maxwell system with C 1 coefficients
Spatial scan statistics using elliptic windows
DEFF Research Database (Denmark)
Christiansen, Lasse Engbo; Andersen, Jens Strodl; Wegener, Henrik Caspar
2006-01-01
The spatial scan statistic is widely used to search for clusters. This article shows that the usually applied elimination of secondary clusters as implemented in SatScan is sensitive to smooth changes in the shape of the clusters. We present an algorithm for generation of a set of confocal elliptic...
Nonlinear Elliptic Differential Equations with Multivalued Nonlinearities
Indian Academy of Sciences (India)
In this paper we study nonlinear elliptic boundary value problems with monotone and nonmonotone multivalued nonlinearities. First we consider the case of monotone nonlinearities. In the first result we assume that the multivalued nonlinearity is defined on all R R . Assuming the existence of an upper and of a lower ...
Vortex precession in thin elliptical ferromagnetic nanodisks
Energy Technology Data Exchange (ETDEWEB)
Zaspel, C.E., E-mail: craig.zaspel@umwestern.edu
2017-07-01
Highlights: • A general form for the magnetostatic energy is calculated for the vortex state in a ferromagnetic ellipse. • The ellipse magnetostatic energy is minimized by conformal mapping the circular disk onto the ellipse. • The gyrotropic precession frequency is obtained in general for a range of ellipticities. - Abstract: The magnetostatic energy is calculated for a magnetic vortex in a noncircular elliptical nanodisk. It is well-known that the energy of a vortex in the circular disk is minimized though an ansatz that eliminates the magnetostatic charge at the disk edge. Beginning with this ansatz for the circular disk, a conformal mapping of a circle interior onto the interior of an ellipse results in the magnetization of the elliptical disk. This magnetization in the interior of an ellipse also has no magnetostatic charge at the disk edge also minimizing the magnetostatic energy. As expected the energy has a quadratic dependence on the displacement of the vortex core from the ellipse center, but reflecting the lower symmetry of the ellipse. Through numerical integration of the magnetostatic integral a general expression for the energy is obtained for ellipticity values from 1.0 to about 0.3. Finally a general expression for the gyrotropic frequency as described by the Thiele equation is obtained.
Abundance Ratios in Dwarf Elliptical Galaxies
Sen, Seyda; Peletier, Reynier F.; Toloba, Elisa; Mentz, Jaco J.
The aim of this study is to determine abundance ratios and star formation histories (SFH) of dwarf ellipticals in the nearby Virgo cluster. We perform a stellar population analysis of 39 dEs and study them using index-index and scaling relations. We find an unusual behaviour where [Na/Fe] is
Elastic plastic buckling of elliptical vessel heads
International Nuclear Information System (INIS)
Alix, M.; Roche, R.L.
1981-08-01
The risks of buckling of dished vessel head increase when the vessel is thin walled. This paper gives the last results on experimental tests of 3 elliptical heads and compares all the results with some empirical formula dealing with elastic and plastic buckling
Equivalent operator preconditioning for elliptic problems
Czech Academy of Sciences Publication Activity Database
Axelsson, Owe; Karátson, J.
2009-01-01
Roč. 50, č. 3 (2009), s. 297-380 ISSN 1017-1398 Institutional research plan: CEZ:AV0Z30860518 Keywords : Elliptic problem * Conjugate gradient method * preconditioning * equivalent operators * compact operators Subject RIV: BA - General Mathematics Impact factor: 0.716, year: 2009 http://en.scientificcommons.org/42514649
Hyper-and-elliptic-curve cryptography
Bernstein, D.J.; Lange, T.
2014-01-01
This paper introduces ‘hyper-and-elliptic-curve cryptography’, in which a single high-security group supports fast genus-2-hyperelliptic-curve formulas for variable-base-point single-scalar multiplication (for example, Diffie–Hellman shared-secret computation) and at the same time supports fast
Local identities involving Jacobi elliptic functions
Indian Academy of Sciences (India)
systematize the local identities by deriving four local 'master identities' analogous to the ... involving Jacobi elliptic functions can be explicitly evaluated and a number of .... most of these integrals do not seem to be known in the literature. In §6 ...
Elliptic genera from multi-centers
Energy Technology Data Exchange (ETDEWEB)
Gaddam, Nava [Institute for Theoretical Physics and Center for Extreme Matter and Emergent Phenomena,Utrecht University, 3508 TD Utrecht (Netherlands)
2016-05-13
I show how elliptic genera for various Calabi-Yau threefolds may be understood from supergravity localization using the quantization of the phase space of certain multi-center configurations. I present a simple procedure that allows for the enumeration of all multi-center configurations contributing to the polar sector of the elliptic genera — explicitly verifying this in the cases of the quintic in ℙ{sup 4}, the sextic in Wℙ{sub (2,1,1,1,1)}, the octic in Wℙ{sub (4,1,1,1,1)} and the dectic in Wℙ{sub (5,2,1,1,1)}. With an input of the corresponding ‘single-center’ indices (Donaldson-Thomas invariants), the polar terms have been known to determine the elliptic genera completely. I argue that this multi-center approach to the low-lying spectrum of the elliptic genera is a stepping stone towards an understanding of the exact microscopic states that contribute to supersymmetric single center black hole entropy in N=2 supergravity.
Broedel, Johannes; Duhr, Claude; Dulat, Falko; Tancredi, Lorenzo
2018-06-01
We introduce a class of iterated integrals that generalize multiple polylogarithms to elliptic curves. These elliptic multiple polylogarithms are closely related to similar functions defined in pure mathematics and string theory. We then focus on the equal-mass and non-equal-mass sunrise integrals, and we develop a formalism that enables us to compute these Feynman integrals in terms of our iterated integrals on elliptic curves. The key idea is to use integration-by-parts identities to identify a set of integral kernels, whose precise form is determined by the branch points of the integral in question. These kernels allow us to express all iterated integrals on an elliptic curve in terms of them. The flexibility of our approach leads us to expect that it will be applicable to a large variety of integrals in high-energy physics.
Elliptic Flow, Initial Eccentricity and Elliptic Flow Fluctuations in Heavy Ion Collisions at RHIC
Nouicer, Rachid; Alver, B.; Back, B. B.; Baker, M. D.; Ballintijn, M.; Barton, D. S.; Betts, R. R.; Bickley, A. A.; Bindel, R.; Busza, W.; Carroll, A.; Chai, Z.; Decowski, M. P.; García, E.; Gburek, T.; George, N.; Gulbrandsen, K.; Halliwell, C.; Hamblen, J.; Hauer, M.; Henderson, C.; Hofman, D. J.; Hollis, R. S.; Holzman, B.; Iordanova, A.; Kane, J. L.; Khan, N.; Kulinich, P.; Kuo, C. M.; Li, W.; Lin, W. T.; Loizides, C.; Manly, S.; Mignerey, A. C.; Nouicer, R.; Olszewski, A.; Pak, R.; Reed, C.; Roland, C.; Roland, G.; Sagerer, J.; Seals, H.; Sedykh, I.; Smith, C. E.; Stankiewicz, M. A.; Steinberg, P.; Stephans, G. S. F.; Sukhanov, A.; Tonjes, M. B.; Trzupek, A.; Vale, C.; van Nieuwenhuizen, G. J.; Vaurynovich, S. S.; Verdier, R.; Veres, G. I.; Walters, P.; Wenger, E.; Wolfs, F. L. H.; Wosiek, B.; Woźniak, K.; Wysłouch, B.
2008-12-01
We present measurements of elliptic flow and event-by-event fluctuations established by the PHOBOS experiment. Elliptic flow scaled by participant eccentricity is found to be similar for both systems when collisions with the same number of participants or the same particle area density are compared. The agreement of elliptic flow between Au+Au and Cu+Cu collisions provides evidence that the matter is created in the initial stage of relativistic heavy ion collisions with transverse granularity similar to that of the participant nucleons. The event-by-event fluctuation results reveal that the initial collision geometry is translated into the final state azimuthal particle distribution, leading to an event-by-event proportionality between the observed elliptic flow and initial eccentricity.
Centrality dependence of directed and elliptic flow at the SPS
International Nuclear Information System (INIS)
Poskanzer, A.M.; Voloshin, S.A.; Baechler, J.; Barna, D.; Barnby, L.S.; Bartke, J.; Barton, R.A.; Betev, L.; Bialkowska, H.; Billmeier, A.; Blume, C.; Blyth, C.O.; Boimska, B.; Bracinik, J.; Brady, F.P.; Brockmann, R.; Brun, R.; Buncic, P.; Carr, L.; Cebra, D.; Cooper, G.E.; Cramer, J.G.; Csato, P.; Eckardt, V.; Eckhardt, F.; Ferenc, D.; Fischer, H.G.; Fodor, Z.; Foka, P.; Freund, P.; Friese, V.; Ftacnik, J.; Gal, J.; Ganz, R.; Gazdzicki, M.; Gladysz, E.; Grebieszkow, J.; Harris, J.W.; Hegyi, S.; Hlinka, V.; Hoehne, C.; Igo, G.; Ivanov, M.; Jacobs, P.; Janik, R.; Jones, P.G.; Kadija, K.; Kolesnikov, V.I.; Kowalski, M.; Lasiuk, B.; Levai, P.; Malakhov, A.I.; Margetis, S.; Markert, C.; Mayes, B.W.; Melkumov, G.L.; Molnar, J.; Nelson, J.M.; Odyniec, G.; Oldenburg, M.D.; Palla, G.; Panagiotou, A.D.; Petridis, A.; Pikna, M.; Pinsky, L.; Poskanzer, A.M.; Prindle, D.J.; Puehlhofer, F.; Reid, J.G.; Renfordt, R.; Retyk, W.; Ritter, H.G.; Roehrich, D.; Roland, C.; Roland, G.; Rybicki, A.; Sammer, T.; Sandoval, A.; Sann, H.; Semenov, A.Yu.; Schaefer, E.; Schmitz, N.; Seyboth, P.; Sikler, F.; Sitar, B.; Skrzypczak, E.; Snellings, R.; Squier, G.T.A.; Stock, R.; Strmen, P.; Stroebele, H.; Susa, T.; Szarka, I.; Szentpetery, I.; Sziklai, J.; Toy, M.; Trainor, T.A.; Trentalange, S.; Ullrich, T.; Varga, D.; Vassiliou, M.; Veres, G.I.; Vesztergombi, G.; Voloshin, S.; Vranic, D.; Wang, F.; Weerasundara, D.D.; Wenig, S.; Whitten, C.; Xu, N.; Yates, T.A.; Yoo, I.K.; Zimanyi, J.
1999-01-01
New data with a minimum bias trigger for 158 GeV/nucleon Pb + Pb have been analyzed. Directed and elliptic flow as a function of rapidity of the particles and centrality of the collision are presented. The centrality dependence of the ratio of elliptic flow to the initial space elliptic anisotropy is compared to models
Systematics of elliptic flow in heavy-ion collisions
Indian Academy of Sciences (India)
We analyze elliptic ﬂow from SIS to RHIC energies systematically in a realistic dynamical cascade model. We compare our results with the recent data from STAR and PHOBOS collaborations on elliptic ﬂow of charged particles at midrapidity in Au + Au collisions at RHIC. In the analysis of elliptic ﬂow at RHIC energy, we ﬁnd ...
I20; the Versatile X-ray Absorption spectroscopy beamline at Diamond Light Source
International Nuclear Information System (INIS)
Diaz-Moreno, S; Hayama, S; Amboage, M; Freeman, A; Sutter, J; Duller, G
2009-01-01
The Versatile Spectroscopy beamline at Diamond Light Source, I20, is currently under construction and aims to begin operation in late 2009 and early 2010. The beamline aims to cover applications from physics, chemistry and biology through materials, environmental and geological science. Three very distinctive modes of operation will be offered at the beamline: scanning X-ray Absorption spectroscopy (XAS), XAS in dispersive mode, and X-ray emission spectroscopy (XES). To achieve this, the beamline has been designed around two independent experimental end-stations operating from a pair of canted wigglers located in a 5m diamond straight section. One branch of the beamline will deliver monochromatic x-ray radiation of high spectral purity to one of the experimental hutches, whilst the other branch will constitute an energy dispersive spectrometer. The novel design of the beamline allows both branches to operate simultaneously.
Advanced light source vacuum policy and vacuum guidelines for beamlines and experiment endstations
International Nuclear Information System (INIS)
Hussain, Z.
1995-08-01
The purpose of this document is to: (1) Explain the ALS vacuum policy and specifications for beamlines and experiment endstations. (2) Provide guidelines related to ALS vacuum policy to assist in designing beamlines which are in accordance with ALS vacuum policy. This document supersedes LSBL-116. The Advanced Light Source is a third generation synchrotron radiation source whose beam lifetime depends on the quality of the vacuum in the storage ring and the connecting beamlines. The storage ring and most of the beamlines share a common vacuum and are operated under ultra-high-vacuum (UHV) conditions. All endstations and beamline equipment must be operated so as to avoid contamination of beamline components, and must include proper safeguards to protect the storage ring vacuum from an accidental break in the beamline or endstation vacuum systems. The primary gas load during operation is due to thermal desorption and electron/photon induced desorption of contaminants from the interior of the vacuum vessel and its components. The desorption rates are considerably higher for hydrocarbon contamination, thus considerable emphasis is placed on eliminating these sources of contaminants. All vacuum components in a beamline and endstation must meet the ALS vacuum specifications. The vacuum design of both beamlines and endstations must be approved by the ALS Beamline Review Committee (BRC) before vacuum connections to the storage ring are made. The vacuum design is first checked during the Beamline Design Review (BDR) held before construction of the beamline equipment begins. Any deviation from the ALS vacuum specifications must be approved by the BRC prior to installation of the equipment on the ALS floor. Any modification that is incorporated into a vacuum assembly without the written approval of the BRC is done at the user's risk and may lead to rejection of the whole assembly
Low cost ESR based X-ray beamline for lithography experimentation
Energy Technology Data Exchange (ETDEWEB)
Kovacs, S.; Doumas, A.; Truncale, M. (Grumman Corp., Bethpage, NY (United States). Space and Electronics Div.)
1992-08-01
Any application of the electron storage ring (ESR) based X-ray lithography technology requires an X-ray radiation transport system to transfer the synchrotron radiation into a spectrum defined by the lithography process requirements. Structure of this transport system (i.e. the beamline) depends on the nature of the application. In this paper a beamline conceptual design will be discussed. The beamline is intended for the developmment of X-ray lithography technology. (orig.).
10 years of protein crystallography at AR-NW12A beamline
Chavas, L. M. G.; Yamada, Y.; Hiraki, M.; Igarashi, N.; Matsugaki, N.; Wakatsuki, S.
2013-03-01
The exponential growth of protein crystallography can be observed in the continuously increasing demand for synchrotron beam time, both from academic and industrial users. Nowadays, the screening of a profusion of sample crystals for more and more projects is being implemented by taking advantage of fully automated procedures at every level of the experiments. The insertion device AR-NW12A beamline is one of the five macromolecular crystallography (MX) beamlines at the Photon Factory (PF). Currently the oldest MX beamline operational at the High Energy Accelerator Research Organization (KEK), the end-station was launched in 2001 as part of an upgrade of the PF Advanced Ring. Since its commissioning, AR-NW12A has been operating as a high-throughput beamline, slowly evolving to a multipurpose end-station for MX experiments. The development of the beamline took place about a decade ago, in parallel with a drastic development of protein crystallography and more general synchrotron technology. To keep the beamline up-to-date and competitive with other MX stations in Japan and worldwide, new features have been constantly added, with the goal of user friendliness of the various beamline optics and other instruments. Here we describe the evolution of AR-NW12A for its tenth anniversary. We also discuss the plans for upgrades for AR-NW12A, the future objectives in terms of the beamline developments, and especially the strong desire to open the beamline to a larger user community.
Wavefront propagation through the beamline designed for seeding the DESY XUV FEL
Reininger, R; Gürtler, P; Bahrdt, J
2001-01-01
A beamline designed to reduce the spectral bandwidth of the DESY XUV FEL is described. The beamline is intended to cover the wavelength range from 6.4 to 50 nm with three variable line spacing gratings. A plane mirror in front of the grating is used to maintain constant magnification in the dispersion direction. The electric field generated by the first undulator at three wavelengths, 6.4, 13, and 25 nm, is propagated through the beamline. The results show that the beamline has the resolution and imaging properties required for seeding the second undulator at these wavelengths.
Construction of Bending Magnet Beamline at the APS for Environmental Studies
International Nuclear Information System (INIS)
Stern, E.A.
1999-01-01
The objective of this research was to design and construct a bending magnet beamline at the Advanced Photon Source. The beamline is to be optimized for x-ray absorption spectroscopy (XAS) studies with a major focus on environmental issues. The beamline will share the experimental facilities under development at the neighboring undulator-based insertion device beamline. It will utilize these facilities for XAS of both bulk and surface samples, with spatial and elemental imaging, on toxic and radioactive samples. It will help meet the rapidly growing need for the application of these techniques to environmental problems
Status of the Nanoscopium Scanning Hard X-ray Nanoprobe Beamline of Synchrotron Soleil
Somogyi, A.; Kewish, C. M.; Ribbens, M.; Moreno, T.; Polack, F.; Baranton, G.; Desjardins, K.; Samama, J. P.
2013-10-01
The Nanoscopium 155 m-long scanning hard X-ray nanoprobe beamline of Synchrotron Soleil (St Aubin, France) is dedicated to quantitative multi-modal 2D/3D imaging. The beamline aims to reach down to 30 nm spatial resolution in the 5-20 keV energy range. Two experimental stations working in consecutive operation mode will be dedicated to coherent diffractive imaging and scanning X-ray nanoprobe techniques. The beamline is in the construction phase, the first user experiments are expected in 2014. The main characteristics of the beamline and an overview of its status are given in this paper.
Status of the Nanoscopium Scanning Hard X-ray Nanoprobe Beamline of Synchrotron Soleil
International Nuclear Information System (INIS)
Somogyi, A; Kewish, C M; Ribbens, M; Moreno, T; Polack, F; Baranton, G; Desjardins, K; Samama, J P
2013-01-01
The Nanoscopium 155 m-long scanning hard X-ray nanoprobe beamline of Synchrotron Soleil (St Aubin, France) is dedicated to quantitative multi-modal 2D/3D imaging. The beamline aims to reach down to 30 nm spatial resolution in the 5–20 keV energy range. Two experimental stations working in consecutive operation mode will be dedicated to coherent diffractive imaging and scanning X-ray nanoprobe techniques. The beamline is in the construction phase, the first user experiments are expected in 2014. The main characteristics of the beamline and an overview of its status are given in this paper
Heterodyne detector for measuring the characteristic of elliptically polarized microwaves
DEFF Research Database (Denmark)
Leipold, Frank; Nielsen, Stefan Kragh; Michelsen, Susanne
2008-01-01
In the present paper, a device is introduced, which is capable of determining the three characteristic parameters of elliptically polarized light (ellipticity, angle of ellipticity, and direction of rotation) for microwave radiation at a frequency of 110 GHz. The device consists of two perpendicu......In the present paper, a device is introduced, which is capable of determining the three characteristic parameters of elliptically polarized light (ellipticity, angle of ellipticity, and direction of rotation) for microwave radiation at a frequency of 110 GHz. The device consists of two...... be calculated. Results from measured and calculated wave characteristics of an elliptically polarized 110 GHz microwave beam for plasma heating launched into the TEXTOR-tokamak experiment are presented. Measurement and calculation are in good agreement. ©2008 American Institute of Physics...
Developing a composite based elliptic spring for automotive applications
International Nuclear Information System (INIS)
Talib, Abdul Rahim Abu; Ali, Aidy; Goudah, G.; Lah, Nur Azida Che; Golestaneh, A.F.
2010-01-01
An automotive suspension system is designed to provide both safety and comfort for the vehicle occupants. In this study, finite element models were developed to optimize the material and geometry of the composite elliptical spring based on the spring rate, log life and shear stress parameters. The influence of the ellipticity ratio on the performance of woven roving-wrapped composite elliptical springs was investigated both experimentally and numerically. The study demonstrated that composite elliptical springs can be used for light and heavy trucks with substantial weight reduction. The results showed that the ellipticity ratio significantly influenced the design parameters. Composite elliptic springs with ellipticity ratios of a/b = 2 had the optimum spring parameters.
The design of an automatically-tuned beamline
International Nuclear Information System (INIS)
Ball, M.S.; Ellison, T.J.P.; Hamilton, B.J.; Jones, W.P.
1994-01-01
A new 30 m beamline (BL1C) is being assembled to connect the new High Intensity Polarized Ion Source (HIPIOS) to the IUCF cyclotrons. This line is being instrumented for complete automatic optimization of all transverse and longitudinal ion optical elements by providing a unique feedback signal for each controllable device. Transversely, steerers and 4-quadrant electrostatic pickups are located approximately 90 degree apart in betatron phase advance along the beamline. Each pickup is instrumented with a single-board, 4-layer op-amp circuit (BPM system) which measures the beam intensity, horizontal (H) and vertical (V) position, and H and V 10 Hz position modulation. The transverse beam ellipse parameters are first automatically determined at the entrance to the beamline by measuring the beam size using a wire scanner as a function of the strength of a quadrupole. The computer then programs the amplitude and phase of four 10 Hz modulators which vary the current in 4 steerers to move the beam centroid around this (reduced area) ellipse in 4-dimensional phase space. The BPM system then outputs voltages proportional to the beam intensity, centroid location, and envelope. Computer algorithms will then set the steerers and quadrupoles to correct the beam position, dispersion, and envelope. Longitudinally, hardware feedback loops, with a bandwidth adjustable from 10 Hz to 30 kHz, will phase-lock the beam to the two bunching systems; another hardware system will automatically vary the buncher amplitudes to compensate for the significant and varying space charge defocusing as the beam current fluctuates. The bunchers' quiescent phases and amplitudes will be optimized using software ''synchronous detectors.''
Subatomic and frontier physics with ELI Beamlines. Reality and dreams
International Nuclear Information System (INIS)
Drska, L.
2010-01-01
Complete text of publication follows. This contribution attempts to review some results of thinking about prospects for meaningful and procreative research by exploitation the potential of the unique laser technology to be available within the future ELI Beamlines Facility - an ultrashort-pulse, multi-petawatt, multi-beam, high-repetition-rate system. The presentation may be (hopefully) regarded as a specific contribution with the emphasis on two concrete areas to general ELI reports. Two sets of potential studies will be discussed: (1) Realistic experiments. (2) 'Dream' research. The first set (maybe realizable in the first phase of the laboratory work) includes the following topics: (1) Laser-driven electronuclear processes. (2) Unconventional / NLTE fusion reactions. (3) Laser positron / antimatter physics. Most detailed analysis will be presented for the subject. Some concrete themes planned for this part of the talk are: Challenges for laser positronium physics. Nuclear excitation in positron annihilation. Positrons and the laboratory astrophysics. Schemes of some experiments exploiting the ELI Beamlines possibilities will be displayed. The second ('dream') set to be outlined (under consideration as potential one for the second phase) should initiate a brainstorming discussion in these areas: (1) High-Z ion physics studies. (2) Exploring of high-gamma systems. (3) Search for hypothetic particles. Again, the highest attention will be paid to the topic. Key themes in this part: The search for hidden-sector lightweights. Challenges and opportunities in photon regeneration experiments. Potential of ELI Beamlines for the research in this area? The final section of the contribution will include some comments on technical issues related to the proposed research themes: (1) Novel targets and particle traps. (2) Diagnostics challenges and solutions. (3) Simulation / Evaluation problems. Some new approaches will be considered. Acknowledgements. This research has been
Considerations for a soft x-ray spectromicroscopy beamline
International Nuclear Information System (INIS)
Winn, B.; Hao, X.; Jacobsen, C.
1996-01-01
The X-1A soft x-ray undulator at the NSLS is the source for the experimental programs in spectromicroscopy. The authors require both spatial and temporal coherence. Due to the relatively large horizontal divergence of the electron beam in the low β straight section of the x-ray storage ring, it has been possible to split the beam using a scraping mirror into two branches: X-1A used by the authors' program and X-1B used for high resolution spectroscopy. They are now rebuilding the X-1A beamline to provide improved resolving power and essentially linear trade-off between photon rate at the zone plate and resolving power for the soft x-ray spectromicroscopy experiments. This new beamline will exploit both additional floorspace due to the NSLS building expansion and increases in the brightness of the x-ray ring. The beam will be further split into two separate beamlines, both of which will use toroidal mirrors to focus the source on the monochromator entrance slits horizontally and to focus on the monochromator exit slits vertically. This separation comes at no loss of coherent flux and permits low thermal loading on the optics, since the authors need little more than the coherent fraction of the beam at the Fresnel zone plate for microfocusing. Because of the small angular acceptance for spatially coherent illumination of the zone plates and the use of an approximately satisfied Rowland condition, the monochromators have sufficient resolving power with fixed exit arms. Experiments can then be placed near the exit slits, with spatial coherence established by the exit slit size. Resolving power will be controlled by adjusting the entrance slit alone with no change of spatial coherence. The zone plates will be overfilled to be less sensitive to beam vibration and drift
Toyota beamline (BL33XU) at SPring-8
Energy Technology Data Exchange (ETDEWEB)
Nonaka, T., E-mail: nonaka@mosk.tytlabs.co.jp; Dohmae, K.; Hayashi, Y.; Yamaguchi, S.; Nagai, Y.; Hirose, Y. [Toyota Central R& D Labs., Inc., 41-1 Nagakute Aichi 480-1192 Japan (Japan); Araki, T. [Diamond Light Source, Harwell Science and Innovation Campus, Didcot, Oxfordshire OX11 0DE UK (United Kingdom); Tanaka, T.; Kitamura, H. [RIKEN Harima Institute 1-1-1 Koto Sayo, Hyogo 679-5148 (Japan); Uruga, T.; Yamazaki, H.; Yumoto, H.; Ohashi, H.; Goto, S. [JASRI/SPring-8 1-1-1 Koto Sayo, Hyogo 679-5148 (Japan)
2016-07-27
The Toyota beamline (BL33XU) at SPring-8 is an undulator beamline developed to assist in the study of various automotive-related materials. The light source is a tapered in-vacuum undulator that provides a variable energy band width as well as a high brilliance X-ray beam. Two different optical arrangements are available: Optics 1 and Optics 2. Optics 1 is dedicated to time-resolved X-ray absorption spectroscopy (XAFS), and consists of two channel-cut crystal monochromators and four water-cooled flat Si mirrors. The Si(111) and Si(220) monochromator crystals cover an energy range of 4.0–46.0 keV and are driven by high-speed AC servo motors. These monochromators, in conjunction with the tapered undulator, enable high-quality XAFS data acquisition with a temporal resolution of 10 ms. Optics 2 is optimized for X-ray diffraction, scattering and imaging and includes a recently installed double crystal monochromator, two water-cooled flat Si mirrors and Kirkpatrick-Baez (KB) focusing mirrors. The monochromator incorporates parallel mounted Si(111) and Si(311) crystals and covers an energy range of 4.5–70 keV. The beamline provides two experimental stations: Exp. Hutch 2 and Exp. Hutch 3. The gas supply system and mass spectrometers installed in Exp. Hutch 2 allow in-operando measurements under various atmospheres. The scanning three-dimensional X-ray diffraction (scanning 3DXRD) microscopy instrumentation developed and installed in Exp. Hutch 3 enables non-destructive orientation and stress mapping of 1 mm-thick steel specimens using a high energy microbeam.
Toyota beamline (BL33XU) at SPring-8
International Nuclear Information System (INIS)
Nonaka, T.; Dohmae, K.; Hayashi, Y.; Yamaguchi, S.; Nagai, Y.; Hirose, Y.; Araki, T.; Tanaka, T.; Kitamura, H.; Uruga, T.; Yamazaki, H.; Yumoto, H.; Ohashi, H.; Goto, S.
2016-01-01
The Toyota beamline (BL33XU) at SPring-8 is an undulator beamline developed to assist in the study of various automotive-related materials. The light source is a tapered in-vacuum undulator that provides a variable energy band width as well as a high brilliance X-ray beam. Two different optical arrangements are available: Optics 1 and Optics 2. Optics 1 is dedicated to time-resolved X-ray absorption spectroscopy (XAFS), and consists of two channel-cut crystal monochromators and four water-cooled flat Si mirrors. The Si(111) and Si(220) monochromator crystals cover an energy range of 4.0–46.0 keV and are driven by high-speed AC servo motors. These monochromators, in conjunction with the tapered undulator, enable high-quality XAFS data acquisition with a temporal resolution of 10 ms. Optics 2 is optimized for X-ray diffraction, scattering and imaging and includes a recently installed double crystal monochromator, two water-cooled flat Si mirrors and Kirkpatrick-Baez (KB) focusing mirrors. The monochromator incorporates parallel mounted Si(111) and Si(311) crystals and covers an energy range of 4.5–70 keV. The beamline provides two experimental stations: Exp. Hutch 2 and Exp. Hutch 3. The gas supply system and mass spectrometers installed in Exp. Hutch 2 allow in-operando measurements under various atmospheres. The scanning three-dimensional X-ray diffraction (scanning 3DXRD) microscopy instrumentation developed and installed in Exp. Hutch 3 enables non-destructive orientation and stress mapping of 1 mm-thick steel specimens using a high energy microbeam.
ELIMED, MEDical and multidisciplinary applications at ELI-Beamlines
International Nuclear Information System (INIS)
Schillaci, F; Anzalone, A; Cirrone, G A P; Cuttone, G; Musumarra, A; Pisciotta, P; Romano, F; Romano, F P; Carpinelli, M; Cutroneo, M; De Martinis, C; Giove, D; Korn, G; Maggiore, M; Margarone, D; Manti, L; Perozziello, F M; Petrovic, I; Ristic-Fira, A; Renis, M
2014-01-01
ELI-Beamlines is one of the pillars of the pan-European project ELI (Extreme Light Infrastructure). It will be an ultra high-intensity, high repetition-rate, femtosecond laser facility whose main goal is generation and applications of high-brightness X-ray sources and accelerated charged particles in different fields. Particular care will be devoted to the potential applicability of laser-driven ion beams for medical treatments of tumors. Indeed, such kind of beams show very interesting peculiarities and, moreover, laser-driven based accelerators can really represent a competitive alternative to conventional machines since they are expected to be more compact in size and less expensive. The ELIMED project was launched thanks to a collaboration established between FZU-ASCR (ELI-Beamlines) and INFN-LNS researchers. Several European institutes have already shown a great interest in the project aiming to explore the possibility to use laser-driven ion (mostly proton) beams for several applications with a particular regard for medical ones. To reach the project goal several tasks need to be fulfilled, starting from the optimization of laser-target interaction to dosimetric studies at the irradiation point at the end of a proper designed transport beam-line. Researchers from LNS have already developed and successfully tested a high-dispersive power Thomson Parabola Spectrometer, which is the first prototype of a more performing device to be used within the ELIMED project. Also a Magnetic Selection System able to produce a small pencil beam out of a wide energy distribution of ions produced in laser-target interaction has been realized and some preliminary work for its testing and characterization is in progress. In this contribution the status of the project will be reported together with a short description of the of the features of device recently developed.
International Nuclear Information System (INIS)
Cowieson, Nathan Philip; Aragao, David; Clift, Mark; Ericsson, Daniel J.; Gee, Christine; Harrop, Stephen J.; Mudie, Nathan; Panjikar, Santosh; Price, Jason R.; Riboldi-Tunnicliffe, Alan; Williamson, Rachel; Caradoc-Davies, Tom
2015-01-01
The macromolecular crystallography beamline MX1 at the Australian Synchrotron is described. MX1 is a bending-magnet crystallography beamline at the 3 GeV Australian Synchrotron. The beamline delivers hard X-rays in the energy range from 8 to 18 keV to a focal spot at the sample position of 120 µm FWHM. The beamline endstation and ancillary equipment facilitate local and remote access for both chemical and biological macromolecular crystallography. Here, the design of the beamline and endstation are discussed. The beamline has enjoyed a full user program for the last seven years and scientific highlights from the user program are also presented
Tomono, Dai; Fukuda, Mitsuhiro; Hatanaka, Kichiji; Higemoto, Wataru; Kawashima, Yoshitaka; Kojima, Kenji M.; Kuno, Yoshitaka; Matsuda, Yugo; Matsuzaki, Teiichiro; Miyake, Yasuhiro; Miyamoto, Koichiro; Morita, Yasuyuki; Motoishi, Takahiro; Nakazawa, Yu; Ninomiya, Kazuhiko; Nishikawa, Ryo; Ohta, Saki; Sato, Akira; Shimomura, Koichiro; Takahisa, Keiji; Weichao, Yao; Wong, Ming L.
At the new DC muon beamline MuSIC at Research Center for Nuclear Physics (RCNP), Osaka University, the beamline construction from the solenoid system of the muon production to the experimental port was completed. A beamline commissioning and a feasibility study for μSR are now in progress. With newly refurbished spectrometer installed at the experimental port, we succeeded in observing μSR spectra and μ-e decay asymmetry in a simple setup down to 4 K. We are still under development of other μSR appratuses.
Status of the ELIMED Beamline at the ELIMAIA facility
Czech Academy of Sciences Publication Activity Database
Schillaci, F.; Cirrone, G.A.P.; Cuttone, G.; Romano, F.; Scuderi, Valentina; Allegra, L.; Amato, A.; Andó, L.; Costa, M.; Gallo, G.; Leanza, R.; Maggiore, M.; Milluzzo, G.; Petringa, G.; Pipek, J.; Russo, A.D.; Korn, Georg; Margarone, Daniele; Leray, M.J.; Tasset-Maye, O.; Antoine, S.; Jehanno, P.
2016-01-01
Roč. 11, Dec (2016), 1-5, č. článku C12052. ISSN 1748-0221 R&D Projects: GA MŠk EF15_008/0000162; GA MŠk LQ1606 Grant - others:ELI Beamlines(XE) CZ.02.1.01/0.0/0.0/15_008/0000162 Institutional support: RVO:68378271 Keywords : accelerator subsystems and technologies * accelerator applications * beam dynamics * beam optics Subject RIV: BL - Plasma and Gas Discharge Physics OBOR OECD: Fluids and plasma physics (including surface physics) Impact factor: 1.220, year: 2016
Monte Carlo simulation of the ELIMED beamline using Geant4
Czech Academy of Sciences Publication Activity Database
Pipek, J.; Romano, F.; Milluzzo, G.; Cirrone, G.A.P.; Cuttone, G.; Amico, A.G.; Margarone, Daniele; Larosa, G.; Leanza, R.; Petringa, G.; Schillaci, Francesco; Scuderi, Valentina
2017-01-01
Roč. 12, Mar (2017), s. 1-5, č. článku C03027. ISSN 1748-0221 R&D Projects: GA MŠk EF15_008/0000162; GA MŠk LQ1606 Grant - others:ELI Beamlines(XE) CZ.02.1.01/0.0/0.0/15_008/0000162 Institutional support: RVO:68378271 Keywords : models and simulation s * accelerator applications * beam dynamics * software architectures * event data models * frameworks and databases Subject RIV: BL - Plasma and Gas Discharge Physics OBOR OECD: Fluids and plasma physics (including surface physics) Impact factor: 1.220, year: 2016
Performance of an elliptically tapered neutron guide
International Nuclear Information System (INIS)
Muehlbauer, Sebastian; Stadlbauer, Martin; Boeni, Peter; Schanzer, Christan; Stahn, Jochen; Filges, Uwe
2006-01-01
Supermirror coated neutron guides are used at all modern neutron sources for transporting neutrons over large distances. In order to reduce the transmission losses due to multiple internal reflection of neutrons, ballistic neutron guides with linear tapering have been proposed and realized. However, these systems suffer from an inhomogeneous illumination of the sample. Moreover, the flux decreases significantly with increasing distance from the exit of the neutron guide. We propose using elliptically tapered guides that provide a more homogeneous phase space at the sample position as well as a focusing at the sample. Moreover, the design of the guide system is simplified because ellipses are simply defined by their long and short axes. In order to prove the concept we have manufactured a doubly focusing guide and investigated its properties with neutrons. The experiments show that the predicted gains using the program package McStas are realized. We discuss several applications of elliptic guides in various fields of neutron physics
A holomorphic anomaly in the elliptic genus
International Nuclear Information System (INIS)
Murthy, Sameer
2014-01-01
We consider a class of gauged linear sigma models (GLSMs) in two dimensions that flow to non-compact (2,2) superconformal field theories in the infra-red, a prototype of which is the SL(2,ℝ)/U(1) (cigar) coset. We compute the elliptic genus of the GLSMs as a path-integral on the torus using supersymmetric localization. We find that the result is a Jacobi-like form that is non-holomorphic in the modular parameter τ of the torus, with mock modular behavior. This agrees with a previously-computed expression in the cigar coset. We show that the lack of holomorphicity of the elliptic genus arises from the contributions of a compact boson carrying momentum and winding excitations. This boson has an axionic shift symmetry and plays the role of a compensator field that is needed to cancel the chiral anomaly in the rest of the theory.
Elliptic differential equations theory and numerical treatment
Hackbusch, Wolfgang
2017-01-01
This book simultaneously presents the theory and the numerical treatment of elliptic boundary value problems, since an understanding of the theory is necessary for the numerical analysis of the discretisation. It first discusses the Laplace equation and its finite difference discretisation before addressing the general linear differential equation of second order. The variational formulation together with the necessary background from functional analysis provides the basis for the Galerkin and finite-element methods, which are explored in detail. A more advanced chapter leads the reader to the theory of regularity. Individual chapters are devoted to singularly perturbed as well as to elliptic eigenvalue problems. The book also presents the Stokes problem and its discretisation as an example of a saddle-point problem taking into account its relevance to applications in fluid dynamics.
Nonlinear elliptic equations and nonassociative algebras
Nadirashvili, Nikolai; Vlăduţ, Serge
2014-01-01
This book presents applications of noncommutative and nonassociative algebras to constructing unusual (nonclassical and singular) solutions to fully nonlinear elliptic partial differential equations of second order. The methods described in the book are used to solve a longstanding problem of the existence of truly weak, nonsmooth viscosity solutions. Moreover, the authors provide an almost complete description of homogeneous solutions to fully nonlinear elliptic equations. It is shown that even in the very restricted setting of "Hessian equations", depending only on the eigenvalues of the Hessian, these equations admit homogeneous solutions of all orders compatible with known regularity for viscosity solutions provided the space dimension is five or larger. To the contrary, in dimension four or less the situation is completely different, and our results suggest strongly that there are no nonclassical homogeneous solutions at all in dimensions three and four. Thus this book gives a complete list of dimensions...
On rotational solutions for elliptically excited pendulum
International Nuclear Information System (INIS)
Belyakov, Anton O.
2011-01-01
The author considers the planar rotational motion of the mathematical pendulum with its pivot oscillating both vertically and horizontally, so the trajectory of the pivot is an ellipse close to a circle. The analysis is based on the exact rotational solutions in the case of circular pivot trajectory and zero gravity. The conditions for existence and stability of such solutions are derived. Assuming that the amplitudes of excitations are not small while the pivot trajectory has small ellipticity the approximate solutions are found both for high and small linear dampings. Comparison between approximate and numerical solutions is made for different values of the damping parameter. -- Highlights: → We study rotations of the mathematical pendulum when its pivot moves along an ellipse. → There are stable exact solutions for a circular pivot trajectory and zero gravity. → Asymptotic solutions are found for an elliptical pivot trajectory
A FUNDAMENTAL LINE FOR ELLIPTICAL GALAXIES
International Nuclear Information System (INIS)
Nair, Preethi; Van den Bergh, Sidney; Abraham, Roberto G.
2011-01-01
Recent studies have shown that massive galaxies in the distant universe are surprisingly compact, with typical sizes about a factor of three smaller than equally massive galaxies in the nearby universe. It has been suggested that these massive galaxies grow into systems resembling nearby galaxies through a series of minor mergers. In this model the size growth of galaxies is an inherently stochastic process, and the resulting size-luminosity relationship is expected to have considerable environmentally dependent scatter. To test whether minor mergers can explain the size growth in massive galaxies, we have closely examined the scatter in the size-luminosity relation of nearby elliptical galaxies using a large new database of accurate visual galaxy classifications. We demonstrate that this scatter is much smaller than has been previously assumed, and may even be so small as to challenge the plausibility of the merger-driven hierarchical models for the formation of massive ellipticals.
Type A Jacobi Elliptic One-Monopole
International Nuclear Information System (INIS)
Teh, Rosy; Wong, Khai-Ming
2010-01-01
We present new classical generalized one-monopole solution of the SU(2) Yang-Mills-Higgs theory with the Higgs field in the adjoint representation. We show that this generalized solution with Θ-winding number m = 1 and φ-winding number n = 1 is an axially symmetric Jacobi elliptic generalization of the 't Hooft-Polyakov one-monopole. We construct this axially symmetric one-monopole solution by generalizing the large distance asymptotic solution of the 't Hooft-Polyakov one-monopole to the Jacobi elliptic functions and solving the second order equations of motion numerically when the Higgs potential is vanishing. This solution is a regular non-BPS finite energy solution.
International Nuclear Information System (INIS)
Kobayashi, Hideki; Sasaki, Shigemi; Shimada, Taihei; Takao, Masaru; Yokoya, Akinori; Miyahara, Yoshikazu
1996-03-01
This paper describes the design of a variably polarizing undulator (APPLE-type) to be installed in soft X-ray beamline in the SPring-8 facility. The magnetic field distribution and radiation spectrum expected from this undulator were calculated. The magnetic field strength is varied by changing the gap distance of upper and lower jaws, so it changes the photon energy in soft X-ray range. By moving the relative position of pairs of magnet rows (phase shift), the polarization of radiation is varied circularly, elliptically and linearly in the horizontal and vertical direction. We expect that right and left handed circular polarizations are obtained alternately at a rate of 1 Hz by high speed phase shifting. The repulsive and attractive magnetic force working on the magnet rows were calculated which interfere in phase shifting at high speed. The magnetic force changes with gap distance and phase shift position, and the magnetic force working on a row in the direction of phase shift becomes up to 500 kgf. The construction of this undulator is started in 1996, that will be inserted in the storage ring in 1997. (author)
Integrable mappings via rational elliptic surfaces
International Nuclear Information System (INIS)
Tsuda, Teruhisa
2004-01-01
We present a geometric description of the QRT map (which is an integrable mapping introduced by Quispel, Roberts and Thompson) in terms of the addition formula of a rational elliptic surface. By this formulation, we classify all the cases when the QRT map is periodic; and show that its period is 2, 3, 4, 5 or 6. A generalization of the QRT map which acts birationally on a pencil of K3 surfaces, or Calabi-Yau manifolds, is also presented
On a fourth order superlinear elliptic problem
Directory of Open Access Journals (Sweden)
M. Ramos
2001-01-01
Full Text Available We prove the existence of a nonzero solution for the fourth order elliptic equation $$Delta^2u= mu u +a(xg(u$$ with boundary conditions $u=Delta u=0$. Here, $mu$ is a real parameter, $g$ is superlinear both at zero and infinity and $a(x$ changes sign in $Omega$. The proof uses a variational argument based on the argument by Bahri-Lions cite{BL}.
A Jacobian elliptic single-field inflation
Energy Technology Data Exchange (ETDEWEB)
Villanueva, J.R. [Universidad de Valparaiso, Instituto de Fisica y Astronomia, Valparaiso (Chile); Centro de Astrofisica de Valparaiso, Valparaiso (Chile); Gallo, Emanuel [FaMAF, Universidad Nacional de Cordoba, Cordoba (Argentina); Instituto de Fisica Enrique Gaviola (IFEG), CONICET, Cordoba (Argentina)
2015-06-15
In the scenario of single-field inflation, this field is described in terms of Jacobian elliptic functions. This approach provides, when constrained to particular cases, analytic solutions already known in the past, generalizing them to a bigger family of analytical solutions. The emergent cosmology is analyzed using the Hamilton-Jacobi approach and then the main results are contrasted with the recent measurements obtained from the Planck 2015 data. (orig.)
The elliptic model for communication fluxes
International Nuclear Information System (INIS)
Herrera-Yagüe, C; Schneider, C M; González, M C; Smoreda, Z; Couronné, T; Zufiria, P J
2014-01-01
In this paper, a model (called the elliptic model) is proposed to estimate the number of social ties between two locations using population data in a similar manner to how transportation research deals with trips. To overcome the asymmetry of transportation models, the new model considers that the number of relationships between two locations is inversely proportional to the population in the ellipse whose foci are in these two locations. The elliptic model is evaluated by considering the anonymous communications patterns of 25 million users from three different countries, where a location has been assigned to each user based on their most used phone tower or billing zip code. With this information, spatial social networks are built at three levels of resolution: tower, city and region for each of the three countries. The elliptic model achieves a similar performance when predicting communication fluxes as transportation models do when predicting trips. This shows that human relationships are influenced at least as much by geography as is human mobility. (paper)
Can elliptical galaxies be equilibrium systems
Energy Technology Data Exchange (ETDEWEB)
Caimmi, R [Padua Univ. (Italy). Ist. di Astronomia
1980-08-01
This paper deals with the question of whether elliptical galaxies can be considered as equilibrium systems (i.e., the gravitational + centrifugal potential is constant on the external surface). We find that equilibrium models such as Emden-Chandrasekhar polytropes and Roche polytropes with n = 0 can account for the main part of observations relative to the ratio of maximum rotational velocity to central velocity dispersion in elliptical systems. More complex models involving, for example, massive halos could lead to a more complete agreement. Models that are a good fit to the observed data are characterized by an inner component (where most of the mass is concentrated) and a low-density outer component. A comparison is performed between some theoretical density distributions and the density distribution observed by Young et al. (1978) in NGC 4473, but a number of limitations must be adopted. Alternative models, such as triaxial oblate non-equilibrium configurations with coaxial shells, involve a number of problems which are briefly discussed. We conclude that spheroidal oblate models describing elliptical galaxies cannot be ruled out until new analyses relative to more refined theoretical equilibrium models (involving, for example, massive halos) and more detailed observations are performed.
The Diamond Beamline I13L for Imaging and Coherence
International Nuclear Information System (INIS)
Rau, C.; Wagner, U.; Peach, A.; Singh, B.; Wilkin, G.; Jones, C.; Robinson, I. K.
2010-01-01
I13L is the first long beamline at Diamond dedicated to imaging and coherence. Two independent branches will operate in the energy range of 6-30 keV with spatial resolution on the micro- to nano-lengthscale. The Imaging branch is dedicated to imaging and tomography with In-line phase contrast and full-field microscopy on the micron to nano-length scale. Ultimate resolution will be achieved on the Coherence branch at I13L with imaging techniques in the reciprocal space. The experimental stations will be located about 250 m from the source, taking advantage of the coherence properties of the source. The beamline has some outstanding features such as the mini-beta layout of the storage ring's straight section. The optical layout is optimized for beam stability and high optical quality to preserve the coherent radiation. In the experimental stations several methods will be available, starting for the first user with in-line phase contrast imaging on the imaging branch and Coherent X-ray Diffraction (CXRD) on the coherence branch.
In situ beamline analysis and correction of active optics.
Sutter, John; Alcock, Simon; Sawhney, Kawal
2012-11-01
At the Diamond Light Source, pencil-beam measurements have enabled long-wavelength slope errors on X-ray mirror surfaces to be examined under ultra-high vacuum and beamline mounting without the need to remove the mirror from the beamline. For an active mirror an automated procedure has been implemented to calculate the actuator settings that optimize its figure. More recently, this in situ pencil-beam method has been applied to additional uses for which ex situ measurements would be inconvenient or simply impossible. First, it has been used to check the stability of the slope errors of several bimorph mirrors at intervals of several weeks or months. Then, it also proved useful for the adjustment of bender and sag compensation actuators on mechanically bent mirrors. Fits to the bending of ideal beams have been performed on the slope errors of a mechanically bent mirror in order to distinguish curvatures introduced by the bending actuators from gravitational distortion. Application of the optimization procedure to another mechanically bent mirror led to an improvement of its sag compensation mechanism.
Material failures observed in Doublet III neutral beamlines
International Nuclear Information System (INIS)
Bailey, E.W.; Colleraine, A.; Doll, D.; Grunloh, H.; Kim, J.; Langhorn, A.; Thurgood, B.
1983-12-01
The Doublet III neutral beam injectors consist of three separable spools two meters in diameter by four meters long overall when assembled. Contained within these spools are the neutralizers, ion dumps, deflecting magnet, calorimeter dumps, cryogenic panels and beam scraping collimators 3,7. To date three beamlines are in operation on Doublet III, and the beams have accumulated operating time of approximately 32 months, with the oldest having been in operation for 18 months. During this time operation of DIII with the neutral beam sources has demonstrated the following: 7.8 MW injected neutrals from three beamlines (6 sources), high β (4.5%), and non-circular plasma shape. The sources have also exhibited a very reliable injected shot history 4, 5, 6, 8. Material failures encountered during the operation of DIII N.B. injectors and the solutions to these failures are described. Failures include cracking of the neutralizer exit collimator due to heating cycles, failure of cyropanel support rods due to cooling cycles, failure of the sliding drive of the moveable calorimeter due to friction
Material failures observed in the Doublet III neutral beamlines
International Nuclear Information System (INIS)
Bailey, E.W.; Colleraine, A.; Doll, D.; Grunloh, H.; Kim, J.; Langhorn, A.; Thurgood, B.
1983-01-01
The Doublet III neutral beam injectors consist of three separable spools two meters in diameter by four meters long overall when assembled. Contained within these spools are the neutralizers, ion dumps, deflecting magnet, calorimeter dumps, cryogenic panels and beam scraping collimators. To date three beamlines are in operation on Doublet III, and the beams have accumulated operating time of approximately 32 months, with the oldest having been in operation for 18 months. During this time operation of DIII with the neutral beam sources has demonstrated the following: 7.8 MW injected neutrals from three beamlines (6 sources), high β (4.5%), and non-circular plasma shape. The sources have also exibited a very reliable injected shot history. Material failures encountered during the operation of DIII N.B. injectors and the solutions to these failures are described. Failures include cracking of the neutralizer exit collimator due to heating cycles, failure of cyropanel support rods due to cooling cycles, failure of the sliding drive of the moveable calorimeter due to friction
CERN announces the fourth annual Beamline for Schools competition
BL4S team
2016-01-01
CERN is pleased to announce the fourth annual Beamline for Schools (BL4S) competition. Once again, in 2017, a fully equipped beamline will be made available at CERN for students. As in previous years, two teams will be invited to the Laboratory to execute the experiments they proposed in their applications. The 2017 competition is being made possible thanks to support from the Alcoa Foundation for the second consecutive year. The competition is open to teams of high-school students aged 16 or older who, if they win, are invited (with two supervisors) to CERN to carry out their experiment. Teams must have at least five students but there is no upper limit to a team’s size (although just nine students per winning team will be invited to CERN). Teams may be composed of pupils from a single school, or from a number of schools working together. As science-loving mega-celebrity Will.I.Am told us: “If you’re interested in science, technology, engineering or ...
Dedicated Beamline Facilities for Catalytic Research. Synchrotron Catalysis Consortium (SCC)
Energy Technology Data Exchange (ETDEWEB)
Chen, Jingguang [Columbia Univ., New York, NY; Frenkel, Anatoly [Yeshiva Univ., New York, NY (United States); Rodriguez, Jose [Brookhaven National Lab. (BNL), Upton, NY (United States); Adzic, Radoslav [Brookhaven National Lab. (BNL), Upton, NY (United States); Bare, Simon R. [UOP LLC, Des Plaines, IL (United States); Hulbert, Steve L. [Brookhaven National Lab. (BNL), Upton, NY (United States); Karim, Ayman [Pacific Northwest National Lab. (PNNL), Richland, WA (United States); Mullins, David R. [Oak Ridge National Lab. (ORNL), Oak Ridge, TN (United States); Overbury, Steve [Oak Ridge National Lab. (ORNL), Oak Ridge, TN (United States)
2015-03-04
Synchrotron spectroscopies offer unique advantages over conventional techniques, including higher detection sensitivity and molecular specificity, faster detection rate, and more in-depth information regarding the structural, electronic and catalytic properties under in-situ reaction conditions. Despite these advantages, synchrotron techniques are often underutilized or unexplored by the catalysis community due to various perceived and real barriers, which will be addressed in the current proposal. Since its establishment in 2005, the Synchrotron Catalysis Consortium (SCC) has coordinated significant efforts to promote the utilization of cutting-edge catalytic research under in-situ conditions. The purpose of the current renewal proposal is aimed to provide assistance, and to develop new sciences/techniques, for the catalysis community through the following concerted efforts: Coordinating the implementation of a suite of beamlines for catalysis studies at the new NSLS-II synchrotron source; Providing assistance and coordination for catalysis users at an SSRL catalysis beamline during the initial period of NSLS to NSLS II transition; Designing in-situ reactors for a variety of catalytic and electrocatalytic studies; Assisting experimental set-up and data analysis by a dedicated research scientist; Offering training courses and help sessions by the PIs and co-PIs.
Energy-dispersive X-ray diffraction beamline at Indus-2 synchrotron ...
Indian Academy of Sciences (India)
An energy-dispersive X-ray diffraction beamline has been designed, developed and commissioned at BL-11 bending magnet port of the Indian synchrotron source, Indus-2. The performance of this beamline has been benchmarked by measuring diffraction patterns from various elemental metals and standard inorganic ...
PLASTIQUE: A synchrotron radiation beamline for time resolved fluorescence in the frequency domain
International Nuclear Information System (INIS)
De Stasio, G.; Zema, N.; Antonangeli, F.; Parasassi, T.; Rosato, N.
1991-01-01
PLASTIQUE is the only synchrotron radiation beamline in the world that performs time resolved fluorescence experiments in the frequency domain. These experiments are extremely valuable sources of informations on the structure and dynamics of molecules. The beamline and some examples of initial data are described
Plastique: A synchrotron radiation beamline for time resolved fluorescence in the frequency domain
De Stasio, Gelsomina; Zema, N.; Antonangeli, F.; Savoia, A.; Parasassi, T.; Rosato, N.
1991-06-01
PLASTIQUE is the only synchrotron radiation beamline in the world that performs time resolved fluorescence experiments in frequency domain. These experiments are extremely valuable sources of information on the structure and dynamics of molecules. We describe the beamline and some initial data.
Elliptical Galaxies: Rotationally Distorted, After All
Directory of Open Access Journals (Sweden)
Caimmi, R.
2009-12-01
Full Text Available On the basis of earlier investigations onhomeoidally striated Mac Laurin spheroids and Jacobi ellipsoids (Caimmi and Marmo2005, Caimmi 2006a, 2007, different sequences of configurations are defined and represented in the ellipticity-rotation plane, $({sf O}hat{e}chi_v^2$. The rotation parameter, $chi_v^2$, is defined as the ratio, $E_mathrm{rot}/E_mathrm{res}$, of kinetic energy related to the mean tangential equatorial velocity component, $M(overline{v_phi}^2/2$, to kineticenergy related to tangential equatorial component velocity dispersion, $Msigma_{phiphi}^2/2$, andresidual motions, $M(sigma_{ww}^2+sigma_{33}^2/2$.Without loss of generality (above a thresholdin ellipticity values, the analysis is restricted to systems with isotropic stress tensor, whichmay be considered as adjoint configurationsto any assigned homeoidally striated density profile with anisotropic stress tensor, different angular momentum, and equal remaining parameters.The description of configurations in the$({sf O}hat{e}chi_v^2$ plane is extendedin two respects, namely (a from equilibriumto nonequilibrium figures, where the virialequations hold with additional kinetic energy,and (b from real to imaginary rotation, wherethe effect is elongating instead of flattening,with respect to the rotation axis.An application is made toa subsample $(N=16$ of elliptical galaxies extracted from richer samples $(N=25,~N=48$of early type galaxies investigated within theSAURON project (Cappellari et al. 2006, 2007.Sample objects are idealized as homeoidallystriated MacLaurinspheroids and Jacobi ellipsoids, and theirposition in the $({sf O}hat{e}chi_v^2$plane is inferred from observations followinga procedure outlined in an earlier paper(Caimmi 2009b. The position of related adjoint configurations with isotropic stresstensor is also determined. With a singleexception (NGC 3379, slow rotators arecharacterized by low ellipticities $(0lehat{e}<0.2$, low anisotropy parameters$(0ledelta<0
Lectures on Selected Topics in Mathematical Physics: Elliptic Functions and Elliptic Integrals
Schwalm, William A.
2015-12-01
This volume is a basic introduction to certain aspects of elliptic functions and elliptic integrals. Primarily, the elliptic functions stand out as closed solutions to a class of physical and geometrical problems giving rise to nonlinear differential equations. While these nonlinear equations may not be the types of greatest interest currently, the fact that they are solvable exactly in terms of functions about which much is known makes up for this. The elliptic functions of Jacobi, or equivalently the Weierstrass elliptic functions, inhabit the literature on current problems in condensed matter and statistical physics, on solitons and conformal representations, and all sorts of famous problems in classical mechanics. The lectures on elliptic functions have evolved as part of the first semester of a course on theoretical and mathematical methods given to first- and second-year graduate students in physics and chemistry at the University of North Dakota. They are for graduate students or for researchers who want an elementary introduction to the subject that nevertheless leaves them with enough of the details to address real problems. The style is supposed to be informal. The intention is to introduce the subject as a moderate extension of ordinary trigonometry in which the reference circle is replaced by an ellipse. This entre depends upon fewer tools and has seemed less intimidating that other typical introductions to the subject that depend on some knowledge of complex variables. The first three lectures assume only calculus, including the chain rule and elementary knowledge of differential equations. In the later lectures, the complex analytic properties are introduced naturally so that a more complete study becomes possible.
Performance of beamline 9.3.1 at the ALS: Flux and resolution measurements
Energy Technology Data Exchange (ETDEWEB)
Uehara, Y. [Univ. of Nevada, Las Vegas, NV (United States); Fischer, G.; Kring, J.; Perera, R.C.C. [Ernest Orlando Lawrence Berkeley National Lab., CA (United States)
1997-04-01
Beamline 9.3.1 at the ALS is a windowless beamline, covering the 1-6 keV photon-energy range. This beamline is the first monochromatic hard x-ray beamline in the ALS, and designed to achieve the goals of high energy resolution, and preservation of the high brightness from the ALS. It consists of a new {open_quotes}Cowan type{close_quotes} double-crystal monochromator and two toroidal mirrors which are positioned before and after the monochromator. The construction of the beamline was completed in December of 1995, with imperfect mirrors. In this report, the authors describe the experimental results of absolute flux measurements and x-ray absorption measurements of gases and solid samples using the present set of mirrors.
Insertion devices and beamlines for the proposed Australian synchrotron light source
International Nuclear Information System (INIS)
Garrett, R.F.; Boldeman, J.W.
1999-01-01
Full text: The proposed Australian synchrotron light source, Boomerang, is a third generation 3 GeV storage ring which is designed to provide for the great majority of Australian requirements for synchrotron radiation well into the next century. The storage ring could accommodate up to 60 experimental stations, including beamlines from 9 insertion devices, which far exceeds the projected Australian requirements over the life of the facility. Undulator radiation will be available up to 20 keV. The first phase construction of Boomerang includes funding for 9 beamlines, comprising 5 bending magnet and 4 insertion device beamlines. The beamline complement has been chosen to cater for approximately 95% of the current and projected Australian demand for synchrotron radiation over the first 5 years operation of the facility. Details will be shown of the performance of the proposed insertion devices, and the initial beamline complement will be presented
Phase II and III the next generation of CLS beamline control and data acquisition systems
International Nuclear Information System (INIS)
Matias, E.; Beauregard, D.; Berg, R.; Black, G.; Boots, M.J.; Dolton, W.; Hunter, D.; Igarashi, R.; Liu, D.; Maxwell, D.; Miller, C.D.; Wilson, T.; Wright, G.
2012-01-01
The Canadian Light Source (CLS) is nearing the completion of its suite of Phase II Beamlines and in detailed design of its Phase III Beamlines. The paper presents an overview of the overall approach adopted by CLS in the development of beamline control and data acquisition systems. Building on the experience of our first phase of beamlines the CLS has continued to make extensive use of EPICS with EDM and QT based user interfaces. Increasing interpretive languages such as Python are finding a place in the beamline control systems. Web based environment such as ScienceStudio have also found a prominent place in the control system architecture as we move to tighter integration between data acquisition, visualization and data analysis. (authors)
Energy Technology Data Exchange (ETDEWEB)
Poswal, A. K., E-mail: poswalashwini@gmail.com; Agrawal, A., E-mail: poswalashwini@gmail.com; Yadav, A. K., E-mail: poswalashwini@gmail.com; Nayak, C., E-mail: poswalashwini@gmail.com; Basu, S., E-mail: poswalashwini@gmail.com; Bhattachryya, D.; Jha, S. N.; Sahoo, N. K. [Atomic and Molecular Physics Division, Bhabha Atomic Research Centre, Mumbai -400085 (India); Kane, S. R.; Garg, C. K. [Indus Synchrotrons Utilization Division, Raja Ramanna Centre for Advanced Technology, Indore- 452013 (India)
2014-04-24
An Energy Scanning X-ray Absorption Fine Structure spectroscopy beamline has recently been installed and commissioned at BL-09 bending magnet port of INDUS-2 synchrotron source, Indore. The beamline uses an UHV compatible fixed exit double crystal monochromator (DCM) with two Si (111) crystals. Two grazing incidence cylindrical mirrors are also used in this beamline; the pre-mirror is used as a collimating mirror while the post mirror is used for vertical focusing and higher harmonic rejection. In this beamline it is possible to carry out EXAFS measurements both in transmission and fluorescence mode on various types of samples, using Ionization chamber detectors and solid state drift detector respectively. In this paper, results from first experiments of the Energy Scanning EXAFS beamline are presented.
Control and data acquisition system for the macromolecular crystallography beamline of SSRF
International Nuclear Information System (INIS)
Wang Qisheng; Huang Sheng; Sun Bo; Tang Lin; He Jianhua
2012-01-01
The macromolecular crystallography beamline BL17U1 of Shanghai Synchrotron Radiation Facility (SSRF) is an important platform for structure biological science. High performance of the beamline would benefit the users greatly in their experiment and data acquisition. To take full advantage of the state-of-the-art mechanical and physical design of the beamline, we have made a series of efforts to develop a robust control and data acquisition system, with user-friendly GUI. These were done by adopting EPICS and Blu-Ice systems on the BL17U1 beamline, with considerations on easy accommodation of new beeline components. In this paper, we report the integration of EPICS and Blu-Ice systems. By using the EPICS gateway interface and several new DHS, Blu-Ice was successfully established for the BL17U1 beamline. As a result, the experiment control and data acquisition system is reliable and functional for users. (authors)
The protein micro-crystallography beamlines for targeted protein research program
International Nuclear Information System (INIS)
Hirata, Kunio; Yamamoto, Masaki; Matsugaki, Naohiro; Wakatsuki, Soichi
2010-01-01
In order to collect proper diffraction data from outstanding micro-crystals, a brand-new data collection system should be designed to provide high signal-to noise ratio in diffraction images. SPring-8 and KEK-PF are currently developing two micro-beam beamlines for Targeted Proteins Research Program by MEXT of Japan. The program aims to reveal the structure and function of proteins that are difficult to solve but have great importance in both academic research and industrial application. At SPring-8, a new 1-micron beam beamline for protein micro-crystallography, RIKEN Targeted Proteins Beamline (BL32XU), is developed. At KEK-PF a new low energy micro-beam beamline, BL-1A, is dedicated for SAD micro-crystallography. The two beamlines will start operation in the end of 2010. The present status of the research and development for protein micro-crystallography will be presented. (author)
WIFIP: a web-based user interface for automated synchrotron beamlines.
Sallaz-Damaz, Yoann; Ferrer, Jean Luc
2017-09-01
The beamline control software, through the associated graphical user interface (GUI), is the user access point to the experiment, interacting with synchrotron beamline components and providing automated routines. FIP, the French beamline for the Investigation of Proteins, is a highly automatized macromolecular crystallography (MX) beamline at the European Synchrotron Radiation Facility. On such a beamline, a significant number of users choose to control their experiment remotely. This is often performed with a limited bandwidth and from a large choice of computers and operating systems. Furthermore, this has to be possible in a rapidly evolving experimental environment, where new developments have to be easily integrated. To face these challenges, a light, platform-independent, control software and associated GUI are required. Here, WIFIP, a web-based user interface developed at FIP, is described. Further than being the present FIP control interface, WIFIP is also a proof of concept for future MX control software.
Energy Technology Data Exchange (ETDEWEB)
Djidel, S.; Bouamar, M.; Khedrouche, D., E-mail: dkhedrouche@yahoo.com [LASS (Laboratoired’Analyse des Signaux et Systèmes), Department of Electronics, University of M’sila BP.166, Route Ichebilia, M’sila, 28000 Algeria (Algeria)
2016-04-21
This paper presents a performances study of UWB monopole antenna using half-elliptic radiator conformed on elliptical surface. The proposed antenna, simulated using microwave studio computer CST and High frequency simulator structure HFSS, is designed to operate in frequency interval over 3.1 to 40 GHz. Good return loss and radiation pattern characteristics are obtained in the frequency band of interest. The proposed antenna structure is suitable for ultra-wideband applications, which is, required for many wearable electronics applications.
International Nuclear Information System (INIS)
Matsugaki, Naohiro; Yamada, Yusuke; Igarashi, Noriyuki; Wakatsuki, Soichi
2007-01-01
A private network, physically separated from the facility network, was designed and constructed which covered all the four protein crystallography beamlines at the Photon Factory (PF) and Structural Biology Research Center (SBRC). Connecting all the beamlines in the same network allows for simple authentication and a common working environment for a user who uses multiple beamlines. Giga-bit Ethernet wire-speed was achieved for the communication among the beamlines and SBRC buildings
Sound Attenuation in Elliptic Mufflers Using a Regular Perturbation Method
Banerjee, Subhabrata; Jacobi, Anthony M.
2012-01-01
The study of sound attenuation in an elliptical chamber involves the solution of the Helmholtz equation in elliptic coordinate systems. The Eigen solutions for such problems involve the Mathieu and the modified Mathieu functions. The computation of such functions poses considerable challenge. An alternative method to solve such problems had been proposed in this paper. The elliptical cross-section of the muffler has been treated as a perturbed circle, enabling the use of a regular perturbatio...
Elliptic genus derivation of 4d holomorphic blocks
Poggi, Matteo
2018-03-01
We study elliptic vortices on ℂ × T 2 by considering the 2d quiver gauge theory describing their moduli spaces. The elliptic genus of these moduli spaces is the elliptic version of vortex partition function of the 4d theory. We focus on two examples: the first is a N = 1, U( N ) gauge theory with fundamental and anti-fundamental matter; the second is a N = 2, U( N ) gauge theory with matter in the fundamental representation. The results are instances of 4d "holomorphic blocks" into which partition functions on more complicated surfaces factorize. They can also be interpreted as free-field representations of elliptic Virasoro algebrae.
Note on twisted elliptic genus of K3 surface
Energy Technology Data Exchange (ETDEWEB)
Eguchi, Tohru, E-mail: eguchi@yukawa.kyoto-u.ac.j [Yukawa Institute for Theoretical Physics, Kyoto University, Kyoto 606-8502 (Japan); Hikami, Kazuhiro, E-mail: KHikami@gmail.co [Department of Mathematics, Naruto University of Education, Tokushima 772-8502 (Japan)
2011-01-03
We discuss the possibility of Mathieu group M{sub 24} acting as symmetry group on the K3 elliptic genus as proposed recently by Ooguri, Tachikawa and one of the present authors. One way of testing this proposal is to derive the twisted elliptic genera for all conjugacy classes of M{sub 24} so that we can determine the unique decomposition of expansion coefficients of K3 elliptic genus into irreducible representations of M{sub 24}. In this Letter we obtain all the hitherto unknown twisted elliptic genera and find a strong evidence of Mathieu moonshine.
Note on twisted elliptic genus of K3 surface
International Nuclear Information System (INIS)
Eguchi, Tohru; Hikami, Kazuhiro
2011-01-01
We discuss the possibility of Mathieu group M 24 acting as symmetry group on the K3 elliptic genus as proposed recently by Ooguri, Tachikawa and one of the present authors. One way of testing this proposal is to derive the twisted elliptic genera for all conjugacy classes of M 24 so that we can determine the unique decomposition of expansion coefficients of K3 elliptic genus into irreducible representations of M 24 . In this Letter we obtain all the hitherto unknown twisted elliptic genera and find a strong evidence of Mathieu moonshine.
Statistics about elliptic curves over finite prime fields
Gekeler, Ernst-Ulrich
2006-01-01
We derive formulas for the probabilities of various properties (cyclicity, squarefreeness, generation by random points) of the point groups of randomly chosen elliptic curves over random prime fields.
COLORS OF ELLIPTICALS FROM GALEX TO SPITZER
Energy Technology Data Exchange (ETDEWEB)
Schombert, James M., E-mail: jschombe@uoregon.edu [Department of Physics, University of Oregon, Eugene, OR 97403 (United States)
2016-12-01
Multi-color photometry is presented for a large sample of local ellipticals selected by morphology and isolation. The sample uses data from the Galaxy Evolution Explorer ( GALEX ), Sloan Digital Sky Survey (SDSS), Two Micron All-Sky Survey (2MASS), and Spitzer to cover the filters NUV , ugri , JHK and 3.6 μ m. Various two-color diagrams, using the half-light aperture defined in the 2MASS J filter, are very coherent from color to color, meaning that galaxies defined to be red in one color are always red in other colors. Comparison to globular cluster colors demonstrates that ellipticals are not composed of a single age, single metallicity (e.g., [Fe/H]) stellar population, but require a multi-metallicity model using a chemical enrichment scenario. Such a model is sufficient to explain two-color diagrams and the color–magnitude relations for all colors using only metallicity as a variable on a solely 12 Gyr stellar population with no evidence of stars younger than 10 Gyr. The [Fe/H] values that match galaxy colors range from −0.5 to +0.4, much higher (and older) than population characteristics deduced from Lick/IDS line-strength system studies, indicating an inconsistency between galaxy colors and line indices values for reasons unknown. The NUV colors have unusual behavior, signaling the rise and fall of the UV upturn with elliptical luminosity. Models with blue horizontal branch tracks can reproduce this behavior, indicating the UV upturn is strictly a metallicity effect.
Constraints on stellar populations in elliptical galaxies
International Nuclear Information System (INIS)
Rose, J.A.
1985-01-01
Photographic image-tube spectra in the wavelength interval 3400--4500 A have been obtained for 12 elliptical galaxy nuclei and for a number of Galactic globular and open clusters in integrated light. The spectra have a wavelength resolution of 2.5 A and a high signal-to-noise ratio. A new quantitative three-dimensional spectral-classification system that has been calibrated on a sample of approx.200 individual stars (Rose 1984) is used to analyze the integrated spectra of the ellipical galaxy nuclei and to compare them with those of the globular clusters. This system is based on spectral indices that are formed by comparing neighborhood spectral features and is unaffected by reddening. The following results have been found: (1) Hot stars (i.e., spectral types A and B) contribute only 2% to the integrated spectra of elliptical galaxies at approx.4000 A, except in the nucleus of NGC 205, where the hot component dominates. This finding is based on a spectral index formed from the relative central intensities in the Ca II H+Hepsilon and Ca II K lines, which is shown to be constant for late-type (i.e., F, G, and K) stars, but changes drastically at earlier types. The observed Ca II H+Hepsilon/Ca II K indices in ellipticals can be reproduced by the inclusion of a small metal-poor population (as in the globular cluster M5) that contributes approx.8% of the light at 4000 A. Such a contribution is qualitatively consistent with the amount of
COLORS OF ELLIPTICALS FROM GALEX TO SPITZER
International Nuclear Information System (INIS)
Schombert, James M.
2016-01-01
Multi-color photometry is presented for a large sample of local ellipticals selected by morphology and isolation. The sample uses data from the Galaxy Evolution Explorer ( GALEX ), Sloan Digital Sky Survey (SDSS), Two Micron All-Sky Survey (2MASS), and Spitzer to cover the filters NUV , ugri , JHK and 3.6 μ m. Various two-color diagrams, using the half-light aperture defined in the 2MASS J filter, are very coherent from color to color, meaning that galaxies defined to be red in one color are always red in other colors. Comparison to globular cluster colors demonstrates that ellipticals are not composed of a single age, single metallicity (e.g., [Fe/H]) stellar population, but require a multi-metallicity model using a chemical enrichment scenario. Such a model is sufficient to explain two-color diagrams and the color–magnitude relations for all colors using only metallicity as a variable on a solely 12 Gyr stellar population with no evidence of stars younger than 10 Gyr. The [Fe/H] values that match galaxy colors range from −0.5 to +0.4, much higher (and older) than population characteristics deduced from Lick/IDS line-strength system studies, indicating an inconsistency between galaxy colors and line indices values for reasons unknown. The NUV colors have unusual behavior, signaling the rise and fall of the UV upturn with elliptical luminosity. Models with blue horizontal branch tracks can reproduce this behavior, indicating the UV upturn is strictly a metallicity effect.
Guided modes of elliptical metamaterial waveguides
International Nuclear Information System (INIS)
Halterman, Klaus; Feng, Simin; Overfelt, P. L.
2007-01-01
The propagation of guided electromagnetic waves in open elliptical metamaterial waveguide structures is investigated. The waveguide contains a negative-index media core, where the permittivity ε and permeability μ are negative over a given bandwidth. The allowed mode spectrum for these structures is numerically calculated by solving a dispersion relation that is expressed in terms of Mathieu functions. By probing certain regions of parameter space, we find the possibility exists to have extremely localized waves that transmit along the surface of the waveguide
Evolution of Hot Gas in Elliptical Galaxies
Mathews, William G.
2004-01-01
This theory grant was awarded to study the curious nature, origin and evolution of hot gas in elliptical galaxies and their surrounding groups. Understanding the properties of this X-ray emitting gas has profound implications over the broad landscape of modern astrophysics: cosmology, galaxy formation, star formation, cosmic metal enrichment, galactic structure and dynamics, and the physics of hot gases containing dust and magnetic fields. One of our principal specific objectives was to interpret the marvelous new observations from the XMM and Chandru satellite X-ray telescopes.
Neutral hydrogen in elliptical and IO galaxies
International Nuclear Information System (INIS)
Bottinelli, L.; Gouguenheim, L.
1979-01-01
New HI detections have been obtained using the Nancay radiotelescope for NGC 2974 and 3962. These results and the large scale distribution obtained for NGC 3962 indicate that the HI-rich elliptical galaxies exhibit common properties which are not easily explained by accretion of an intergalactic cloud. The field aroud NGC 1052 has been mapped and there is an HI connection with the neighbouring galaxies. The HI content of several IO galaxies indicates that the galaxies which are members of groups are relatively HI-rich; this could be produced by additional HI coming from companion galaxies [fr
Beam property studies in the PLS diagnostic beamline
Ko, I S; Seon, D K; Kim, C B; Lee, T Y
1999-01-01
A diagnostic beamline has been operated in the Pohang Light Source (PLS) storage ring for the diagnostics of electron and photon beam properties. It consists of two 1:1 imaging systems: a visible-light imaging system and a soft X-ray imaging system. We have measured the transverse and the longitudinal structures of beams by using a streak camera to obtain a visible image. Accurate transverse beam size have been measured to be 186 mu m horizontally and 43.1 mu m vertically by using soft X-ray images with minimum diffraction errors. The corresponding emittances are 11.7 nm-rad horizontally and 0.59 nm-rad vertically. By comparing the measured data with the design values, we confirmed that the PLS storage ring has reached its designed performance within an error of 3.3 % in the transverse direction.
SR TXRF: performances and perspectives of a dedicated synchrotron beamline
International Nuclear Information System (INIS)
Comin, F.; Apostolo, G.
2000-01-01
In principle the brilliance of synchrotron radiation x-ray beams combined with a high degree of linear polarization allows to reach at the same time low LLDs, mapping of the impurity distribution and chemical identification for elements as light as Na. The TXRF facility at the European synchrotron radiation facility is installed along a beamline dedicated to industry and is designed to reach ultimate detection limits of 6 x 10 7 at/cm 2 in selected areas, or to map the concentration of contaminants with LLD in the scale 10 9 at/cm 2 . In the present configuration the facility works in vacuum with a single element detector. Loading, unloading and pump down of wafers is completely automatic. Typical DDLs are of few 10 9 at/cm 2 for Na and Al and 10 8 at/cm 2 for transition metals. Absorption spectra (XANES and EXAFS) of TM help in the defining the chemistry of the contaminant. (author)
A compact double crystal monochromator for electrochemistry beamline at PLS
Rah, S; Kim, G H
2001-01-01
A compact double crystal monochromator based on 16.5'' CF flange has been designed, fabricated and installed for electrochemistry beamline at Pohang light source. The Bragg angle range of the monochromator is 7-75 deg. The mechanical design is modified from typical Boomerang design [J.A. Golovchenko et al., Rev. Sci. Instrum. 52 (1981) 509; J.P. Kirkland, Nucl. Instr. and Meth. A291 (1990) 185] to have fixed beam offset and single driving axis for spectroscopy experiments. The parallelism error of the crystals is minimized to less than 6 mu rad for the range, by using a precision single axis linear guide, Also, the number of mechanical parts in the vacuum is minimized and 1.8x10 sup - sup 9 Torr of vacuum is achieved without baking.
Shielding and activation studies for the ELI-beamlines project
International Nuclear Information System (INIS)
Fasso, Alberto; Korn, Georg; Versaci, Roberto; Ferrari, Anna
2015-01-01
ELI-beamlines is one of the four pillars of the Extreme Light Infrastructure, a European ESFRI Project, for the next generation of high-energy and high-intensity lasers. It aims at the development of high-brightness sources of X-rays and the acceleration of proton, electron, and ion beams, to be used both for pure research and practical applications. Aiming at a proper radiation protection assessment, for both shielding and activation, extensive FLUKA simulations have been performed, taking into account the laser high repetition rates. The present work, which is the continuation of the calculations presented at SATIF-10, is the first one based on the design of the facility being constructed and on the updated experimental set-up. (authors)
Laser-driven acceleration at ELI Beamlines - radioprotection aspects
International Nuclear Information System (INIS)
Olsovcova, V.; Fasso, A; Versaci, R.
2014-01-01
The international research centre ELI Beamlines, which is under construction in the village of Dolni Brezany near Prague, will exploit high power lasers of PW class to generate and accelerate beams of charged particles (up to tens of GeVs in energy). The beams will be used for both fundamental and applied research by experts from various scientific fields, including biology, medicine, plasma physics but also dosimetry and radiation protection. As laboratories operating lasers do not belong among the traditional 'radiation workplaces', there are no suitable specialized recommendations or standards available. Therefore, it is necessary to newly implement the existing general recommendations. Further, the generated mixed fields possess unique properties due to their production methods. As a result, the routinely used detection methods are not reliable or fail completely. (authors)
The initial scientific program at the NSLS infrared beamline
International Nuclear Information System (INIS)
Williams, G.P.
1989-01-01
Unique extraction optics (90 x 90 mrads) at the NSLS U4IR line offer high brightness beams at up to mm wavelengths with a ∼1ns pulse structure. Radiation from this port has now been carefully characterized and agrees well with calculations, making it 100--1000 times brighter than conventional sources in the middle and far infrared regions. Using rapid scan Michelson interferometers with liquid He cooled bolometer detectors we have been able for the first time to measure molecule substrate vibrations in surface science. We have also made the first measurements of the transmission of a film of the high Tc material YBaCuO in the BCS gap region. These initial experiments have demonstrated the advantages of the superior signal to noise available from this infrared beamline. 19 refs., 6 figs
J-PARC accelerator and neutrino beamline upgrade programme
Friend, M.
2017-09-01
The 30 GeV proton beam from the J-PARC Main Ring (MR) accelerator is used to produce a world-class conventional neutrino beam - the neutrino source for the J-PARC long-baseline neutrino programme, including the current T2K experiment and proposed future experiments. Planned upgrades to increase the beam power of the MR from the current ˜400 kW to the design power of 750 kW and beyond, to 1.3+ MW, are underway. These include hardware modifications, such as upgrades of the MR magnet power supplies, RF systems, and feedback systems, as well as a change of the MR beam betatron tune point. Upgrades to the neutrino beamline, such as to the proton beam monitoring, horns, and radioactive material handling, will also be required to accommodate the increased proton beam power. An overview of planned J-PARC MR and neutrino facility upgrades is given.
New features at the LURE-D22 SAXS beamline
International Nuclear Information System (INIS)
Lesieur, P.; Lombardo, D.; Beauchet, L; Creof, C.; Decamps, T.; Dubuisson, J.M.; Perilhous, G.
1999-01-01
The D22 beamline of the DCI storage ring at LURE is dedicated to the study of structural properties in the field of material science by way of the small-angle X ray scattering (SAXS) technique. The D2 bending magnet of the DCI ring offers a stable source of limited brilliance but long decay time (200 hours) so that the beam can be used up to 110 hours after an injection of the positrons in the ring. Two different settings respectively dedicated to metallurgy and soft matter share the beam time. The latter which is considered here mainly deals with ill condensed matter: non ideal solutions of amphiphiles or polymers, liquid crystals, colloids, gels, xerogels, aerogels. (author)
The Optical Design of the PEP-II Injection Beamlines
Fieguth, T
1996-01-01
The optical design of the PEP-II electron and positron Injection Beamlines is described. Use of the existing high power, low emittance beams available from the SLC damping rings require that pulsed extraction of 9.0 GeV electrons and 3.1 GeV positrons for injection into the PEP-II rings occur in the early sectors of the accelerator. More than 5 kilometers of new beam transport lines have been designed and are being constructed to bring these beams to their respective rings. The optical design maximizes the tolerance to errors especially to those contributing to beam size and position jitter. Secondly, the design minimizes costs by utilizing existing components or component designs and minimizing the number required. Here we discuss important attributes including choice of lattice, specification of error tolerances, including errors in construction, alignment, field errors, power supply stability, and orbit correction.
The Optical Design of the PEP-II Injection Beamlines
Energy Technology Data Exchange (ETDEWEB)
Fieguth, Ted
2003-05-23
The optical design of the PEP-II electron and positron Injection Beamlines is described. Use of the existing high power, low emittance beams available from the SLC damping rings require that pulsed extraction of 9.0 GeV electrons and 3.1 GeV positrons for injection into the PEP-II rings occur in the early sectors of the accelerator. More than 5 kilometers of new beam transport lines have been designed and are being constructed to bring these beams to their respective rings. The optical design maximizes the tolerance to errors especially to those contributing to beam size and position jitter. Secondly, the design minimizes costs by utilizing existing components or component designs and minimizing the number required. Here we discuss important attributes including choice of lattice, specification of error tolerances, including errors in construction, alignment, field errors, power supply stability, and orbit correction.
Design optimization of ultra-precise elliptical mirrors for hard x-ray nanofocusing at Nanoscopium
Kewish, Cameron M.; Polack, François; Signorato, Riccardo; Somogyi, Andrea
2013-09-01
The design and implementation of a pair of 100 mm-long grazing-incidence total-reflection mirrors for the hard X-ray beamline Nanoscopium at Synchrotron Soleil is presented. A vertically and horizontally nanofocusing mirror pair, oriented in Kirkpatrick-Baez geometry, has been designed and fabricated with the aim of creating a diffraction-limited high-intensity 5 - 20 keV beam with a focal spot size as small as 50 nm. We describe the design considerations, including wave-optical calculations of figures-of-merit that are relevant for spectromicroscopy, such as the focal spot size, depth of field and integrated intensity. The mechanical positioning tolerance in the pitch angle that is required to avoid introducing high-intensity features in the neighborhood of the focal spot is demonstrated with simulations to be of the order of microradians, becoming tighter for shorter focal lengths and therefore directly affecting all nanoprobe mirror systems. Metrology results for the completed mirrors are presented, showing that better than 1.5 °A-rms figure error has been achieved over the full mirror lengths with respect to the designed elliptical surfaces, with less than 60 nrad-rms slope errors.
Ellipticity dependence of the near-threshold harmonics of H2 in an elliptical strong laser field.
Yang, Hua; Liu, Peng; Li, Ruxin; Xu, Zhizhan
2013-11-18
We study the ellipticity dependence of the near-threshold (NT) harmonics of pre-aligned H2 molecules using the time-dependent density functional theory. The anomalous maximum appearing at a non-zero ellipticity for the generated NT harmonics can be attributed to multiphoton effects of the orthogonally polarized component of the elliptical driving laser field. Our calculation also shows that the structure of the bound-state, such as molecular alignment and bond length, can be sensitively reflected on the ellipticity dependence of the near-threshold harmonics.
The ELIMED transport and dosimetry beamline for laser-driven ion beams
Energy Technology Data Exchange (ETDEWEB)
Romano, F., E-mail: francesco.romano@lns.infn.it [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Schillaci, F.; Cirrone, G.A.P.; Cuttone, G. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Scuderi, V. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); ELI-Beamlines Project, Institute of Physics ASCR, v.v.i. (FZU), 182 21 Prague (Czech Republic); Allegra, L.; Amato, A.; Amico, A.; Candiano, G.; De Luca, G.; Gallo, G. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Giordanengo, S.; Guarachi, L. Fanola [Istituto Nazionale di Fisica Nucleare, Sezione di Torino, Via P. Giuria 1, Torino (Italy); Universita' di Torino, Dipartimento di Fisica, Via P. Giuria 1, Torino (Italy); Korn, G. [ELI-Beamlines Project, Institute of Physics ASCR, v.v.i. (FZU), 182 21 Prague (Czech Republic); Larosa, G. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Leanza, R. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Universita' di Catania, Dipartimento di Fisica e Astronomia, Via S. Sofia 64, Catania (Italy); Manna, R.; Marchese, V. [Istituto Nazionale di Fisica Nucleare, Laboratori Nazionali del Sud, Via Santa Sofia 62, Catania (Italy); Marchetto, F. [Istituto Nazionale di Fisica Nucleare, Sezione di Torino, Via P. Giuria 1, Torino (Italy); Margarone, D. [ELI-Beamlines Project, Institute of Physics ASCR, v.v.i. (FZU), 182 21 Prague (Czech Republic); and others
2016-09-01
A growing interest of the scientific community towards multidisciplinary applications of laser-driven beams has led to the development of several projects aiming to demonstrate the possible use of these beams for therapeutic purposes. Nevertheless, laser-accelerated particles differ from the conventional beams typically used for multiscipilinary and medical applications, due to the wide energy spread, the angular divergence and the extremely intense pulses. The peculiarities of optically accelerated beams led to develop new strategies and advanced techniques for transport, diagnostics and dosimetry of the accelerated particles. In this framework, the realization of the ELIMED (ELI-Beamlines MEDical and multidisciplinary applications) beamline, developed by INFN-LNS (Catania, Italy) and that will be installed in 2017 as a part of the ELIMAIA beamline at the ELI-Beamlines (Extreme Light Infrastructure Beamlines) facility in Prague, has the aim to investigate the feasibility of using laser-driven ion beams for multidisciplinary applications. In this contribution, an overview of the beamline along with a detailed description of the main transport elements as well as the detectors composing the final section of the beamline will be presented.
International Nuclear Information System (INIS)
Gabadinho, José; Beteva, Antonia; Guijarro, Matias; Rey-Bakaikoa, Vicente; Spruce, Darren
2010-01-01
MxCuBE is a beamline control environment optimized for the needs of macromolecular crystallography. This paper describes the design of the software and the features that MxCuBE currently provides. The design and features of a beamline control software system for macromolecular crystallography (MX) experiments developed at the European Synchrotron Radiation Facility (ESRF) are described. This system, MxCuBE, allows users to easily and simply interact with beamline hardware components and provides automated routines for common tasks in the operation of a synchrotron beamline dedicated to experiments in MX. Additional functionality is provided through intuitive interfaces that enable the assessment of the diffraction characteristics of samples, experiment planning, automatic data collection and the on-line collection and analysis of X-ray emission spectra. The software can be run in a tandem client-server mode that allows for remote control and relevant experimental parameters and results are automatically logged in a relational database, ISPyB. MxCuBE is modular, flexible and extensible and is currently deployed on eight macromolecular crystallography beamlines at the ESRF. Additionally, the software is installed at MAX-lab beamline I911-3 and at BESSY beamline BL14.1
Can mergers make slowly rotating elliptical galaxies
International Nuclear Information System (INIS)
White, S.D.M.
1979-01-01
The results of numerical experiments are used to guide an analytic discussion of hyperbolic mergers among an uncorrelated galaxy population. The expected merger rate is derived as a function of progenitor mass and relative angular momentum, and is used to predict the distribution of the parameter V/sub c//sigma 0 for merger products where V/sub c/ is the maximum observed rotation velocity in a galaxy and sigma 0 is its central velocity dispersion. The median value of this parameter for mergers between comparable galaxies is estimated to be 0.65 and is higher than the observed value in any of the 14 galaxies for which data are available. It seems unlikely that most elliptical galaxies are the result of single or multiple mergers between initially unbound stellar systems; further observational and theoretical work is suggested which should lead to a conclusive test of this picture. The present arguments cannot, however, exclude formation from low angular momentum elliptical orbits
Thermodynamics of Inozemtsev's elliptic spin chain
International Nuclear Information System (INIS)
Klabbers, Rob
2016-01-01
We study the thermodynamic behaviour of Inozemtsev's long-range elliptic spin chain using the Bethe ansatz equations describing the spectrum of the model in the infinite-length limit. We classify all solutions of these equations in that limit and argue which of these solutions determine the spectrum in the thermodynamic limit. Interestingly, some of the solutions are not selfconjugate, which puts the model in sharp contrast to one of the model's limiting cases, the Heisenberg XXX spin chain. Invoking the string hypothesis we derive the thermodynamic Bethe ansatz equations (TBA-equations) from which we determine the Helmholtz free energy in thermodynamic equilibrium and derive the associated Y-system. We corroborate our results by comparing numerical solutions of the TBA-equations to a direct computation of the free energy for the finite-length hamiltonian. In addition we confirm numerically the interesting conjecture put forward by Finkel and González-López that the original and supersymmetric versions of Inozemtsev's elliptic spin chain are equivalent in the thermodynamic limit.
Development of a superconducting elliptically polarized undulator
International Nuclear Information System (INIS)
Chen, S D; Liang, K S; Jan, J C; Hwang, C S
2010-01-01
A superconducting, elliptically polarized undulator (SEPU24) with a period of length 24 mm was developed to provide first-harmonic photons from a 0.8 GeV storage ring for extreme-ultraviolet (EUV) lithography experiment. In SEPU24, two layers of a magnet array structure - with and without rotated magnet arrays - are combined to generate a helical field that provides radiation with wavelength 13.5 nm in the in-band energy. The arrays of iron and aluminium poles were wound with a racetrack coil vertically as for the magnet pole array. The elliptical field is created when the up and down magnet-pole arrays pass excitation currents in alternate directions. SEPU24 is designed with a magnet of gap 6.8 mm, yielding magnetic flux density B x =B z =0.61 T of the helical field. A prototype magnet was fabricated with a diode for quench protection, and assembled in a test dewar to test the magnet performance. A cryogenic Hall-probe system with a precise linear stage was used to measure the distribution of the magnetic field. We describe the design concept and algorithm, the engineering design, the calculation of the magnetic field, the construction and testing of the 10-pole prototype magnet and related issues.
First results from the high-brightness x-ray spectroscopy beamline at ALS
Energy Technology Data Exchange (ETDEWEB)
Perera, R.C.C.; Ng, W.; Jones, G. [Lawrence Berkeley National Lab., CA (United States)] [and others
1997-04-01
Beamline 9.3.1 at the Advanced Light Source (ALS) is a windowless beamline, covering the 1-6 keV photon-energy range, designed to achieve the goal of high brightness at the sample for use in the X-ray Atomic and Molecular Spectroscopy (XAMS) science, surface and interface science, biology and x-ray optical development programs at ALS. X-ray absorption and time of flight photo emission measurements in 2 - 5 keV photon energy in argon along with the flux, resolution, spot size and stability of the beamline will be discussed. Prospects for future XAMS measurements will also be presented.
Characterization of γ-ray background at IMAT beamline of ISIS Spallation Neutron Source
Festa, G.; Andreani, C.; Arcidiacono, L.; Burca, G.; Kockelmann, W.; Minniti, T.; Senesi, R.
2017-08-01
The environmental γ -ray background on the IMAT beamline at ISIS Spallation Neutron Source, Target Station 2, is characterized via γ spectroscopy. The measurements include gamma exposure at the imaging detector position, along with the gamma background inside the beamline. Present results are discussed and compared with previous measurements recorded at INES and VESUVIO beamlines operating at Target Station 1. They provide new outcome for expanding and optimizing the PGAA experimental capability at the ISIS neutron source for the investigation of materials, engineering components and cultural heritage objects at the ISIS neutron source.
Imaging in real and reciprocal space at the Diamond beamline I13
International Nuclear Information System (INIS)
Rau, C.; Wagner, U. H.; Vila-Comamala, J.; Bodey, A.; Parson, A.; García-Fernández, M.; Pešić, Z.; De Fanis, A.
2016-01-01
The Diamond Imaging and Coherence beamline I13 consists of two independent branchlines for imaging in real and reciprocal space. Different microscopies are available providing a range of spatial resolution from 5µm to potentially 5nm. The beamline operates in the energy range of 6-35keV covering different scientific areas such as biomedicine, materials science and geophysics. Several original devices have been developed at the beamline, such as the EXCALIBUR photon counting detector and the combined robot arms for coherent X-ray diffraction
VUV-soft x-ray beamline for spectroscopy and calibration
International Nuclear Information System (INIS)
Bartlett, R.J.; Trela, W.J.; Michaud, F.D.; Southworth, S.H.; Rothe, R.; Alkire, R.W.
1986-01-01
The authors describe the design and performance of the Los Alamos VUV synchrotron radiation beamline, U3C, on the VUV ring of the National Synchrotron Light Source at Brookhaven National Laboratory. The beamline uses separate function optics to collect and focus the horizontally and vertically diverging beam. The monochromator is a grazing incidence Roland circle instrument of the extended grasshopper design (ERG). A post monochromator refocusing mirror is used to focus or collimate the diverging beam from the monochromator. The beamline control and diagnostics systems are also discussed
An unique synchrotron beamline for fine X ray characterizations of nuclear fuel cycle materials
International Nuclear Information System (INIS)
Sitaud, B.; Lequien, S.
2004-01-01
A beamline dedicated to the study of highly radioactive samples up to 18.5 GBq will be constructed on the new third generation synchrotron SOLEIL. Based on the use of X ray beam of very high flux, this beamline named MARS will give true opportunities for new studies of chemistry and physics on fuel cycle materials with the respect of safety conditions. Complementary investigations should be carried out on different experimental stations. The three main techniques will be the micro fluorescence, the micro absorption and the high resolution diffraction. The MARS beamline should be up and working for the international community by the beginning of 2007. (authors)
International Nuclear Information System (INIS)
Khounsary, A.M.; Phillips, W.
1992-01-01
Recent advances in chemical vapor deposition (CVD) technology have made it possible to produce thin free-standing diamond foils that can be used as the window material in high heat load, synchrotron beamlines. Numerical simulations suggest that these windows can offer an attractive and at times the only altemative to beryllium windows for use in third generation x-ray synchrotron radiation beamlines. Utilization, design, and fabrication aspects of diamond windows for high heat load x-ray beamlines are discussed, as are the microstructure characteristics bearing on diamond's performance in this role. Analytic and numerical results are also presented to provide a basis for the design and testing of such windows
An unique synchrotron beamline for fine X ray characterizations of nuclear fuel cycle materials
Energy Technology Data Exchange (ETDEWEB)
Sitaud, B.; Lequien, S
2004-07-01
A beamline dedicated to the study of highly radioactive samples up to 18.5 GBq will be constructed on the new third generation synchrotron SOLEIL. Based on the use of X ray beam of very high flux, this beamline named MARS will give true opportunities for new studies of chemistry and physics on fuel cycle materials with the respect of safety conditions. Complementary investigations should be carried out on different experimental stations. The three main techniques will be the micro fluorescence, the micro absorption and the high resolution diffraction. The MARS beamline should be up and working for the international community by the beginning of 2007. (authors)
CHARACTERIZATION OF THE NEW NSLS INFARED MICROSPECTROSCOPY BEAMLINE U10B.
Energy Technology Data Exchange (ETDEWEB)
CARR,G.L.
1999-07-19
The first of several new infrared beamlines, built on a modified bending magnet port of the NSLS VUV ring, is now operational for mid-infrared microspectroscopy. The port simultaneously delivers 40 mrad by 40 mrad to two separate beamlines and spectrometer endstations designated U10A and U10B. The latter is equipped with a scanning infrared microspectrometer. The combination of this instrument and high brightness synchrotron radiation makes diffraction-limited microspectroscopy practical. This paper describes the beamline's performance and presents quantitative information on the diffraction-limited resolution.
Characterization of the new NSLS infrared microspectroscopy beamline U10B
Energy Technology Data Exchange (ETDEWEB)
Carr, G.L.
1999-07-19
The first of several new infrared beamlines, built on a modified bending magnet port of the NSLS VUV ring, is now operational for mid-infrared microspectroscopy. The port simultaneously delivers 40 mrad by 40 mrad to two separate beamlines and spectrometer endstations designated U10A and U10B. The latter is equipped with a scanning infrared microspectrometer. The combination of this instrument and high brightness synchrotron radiation makes diffraction-limited microspectroscopy practical. This paper describes the beamline's performance and presents quantitative information on the diffraction-limited resolution.
Characterization of γ-ray background at IMAT beamline of ISIS Spallation Neutron Source
International Nuclear Information System (INIS)
Festa, G.; Andreani, C.; Arcidiacono, L.; Senesi, R.; Burca, G.; Kockelmann, W.; Minniti, T.
2017-01-01
The environmental γ -ray background on the IMAT beamline at ISIS Spallation Neutron Source, Target Station 2, is characterized via γ spectroscopy. The measurements include gamma exposure at the imaging detector position, along with the gamma background inside the beamline. Present results are discussed and compared with previous measurements recorded at INES and VESUVIO beamlines operating at Target Station 1. They provide new outcome for expanding and optimizing the PGAA experimental capability at the ISIS neutron source for the investigation of materials, engineering components and cultural heritage objects at the ISIS neutron source.
Imaging in real and reciprocal space at the Diamond beamline I13
Energy Technology Data Exchange (ETDEWEB)
Rau, C., E-mail: Christoph.rau@diamond.ac.uk [Diamond Light Source Ltd., Harwell Science and Innovation Campus, Didcot, OX 11 0DE (United Kingdom); University of Manchester, School of Materials Grosvenor St., Manchester, M1 7HS (United Kingdom); Northwestern University School of Medicine, 303 E. Chicago Avenue, Chicago, IL 60611-3008 (United States); Wagner, U. H.; Vila-Comamala, J.; Bodey, A.; Parson, A.; García-Fernández, M.; Pešić, Z. [Diamond Light Source Ltd., Harwell Science and Innovation Campus, Didcot, OX 11 0DE (United Kingdom); De Fanis, A. [Diamond Light Source Ltd., Harwell Science and Innovation Campus, Didcot, OX 11 0DE (United Kingdom); European XFEL GmbH, Notkestraße 85, 22607 Hamburg (Germany)
2016-01-28
The Diamond Imaging and Coherence beamline I13 consists of two independent branchlines for imaging in real and reciprocal space. Different microscopies are available providing a range of spatial resolution from 5µm to potentially 5nm. The beamline operates in the energy range of 6-35keV covering different scientific areas such as biomedicine, materials science and geophysics. Several original devices have been developed at the beamline, such as the EXCALIBUR photon counting detector and the combined robot arms for coherent X-ray diffraction.
Ronchi test for characterization of X-ray nanofocusing optics and beamlines.
Uhlén, Fredrik; Rahomäki, Jussi; Nilsson, Daniel; Seiboth, Frank; Sanz, Claude; Wagner, Ulrich; Rau, Christoph; Schroer, Christian G; Vogt, Ulrich
2014-09-01
A Ronchi interferometer for hard X-rays is reported in order to characterize the performance of the nanofocusing optics as well as the beamline stability. Characteristic interference fringes yield qualitative data on present aberrations in the optics. Moreover, the visibility of the fringes on the detector gives information on the degree of spatial coherence in the beamline. This enables the possibility to detect sources of instabilities in the beamline like vibrations of components or temperature drift. Examples are shown for two different nanofocusing hard X-ray optics: a compound refractive lens and a zone plate.
Picone-type inequalities for nonlinear elliptic equations and their applications
Directory of Open Access Journals (Sweden)
Takaŝi Kusano
2001-01-01
Full Text Available Picone-type inequalities are derived for nonlinear elliptic equations, and Sturmian comparison theorems are established as applications. Oscillation theorems for forced super-linear elliptic equations and superlinear-sublinear elliptic equations are also obtained.
Recent Major Improvements to the ALS Sector 5 Macromolecular Crystallography Beamlines
International Nuclear Information System (INIS)
Morton, Simon A.; Glossinger, James; Smith-Baumann, Alexis; McKean, John P.; Trame, Christine; Dickert, Jeff; Rozales, Anthony; Dauz, Azer; Taylor, John; Zwart, Petrus; Duarte, Robert; Padmore, Howard; McDermott, Gerry; Adams, Paul
2007-01-01
Although the Advanced Light Source (ALS) was initially conceived primarily as a low energy (1.9GeV) 3rd generation source of VUV and soft x-ray radiation it was realized very early in the development of the facility that a multipole wiggler source coupled with high quality, (brightness preserving), optics would result in a beamline whose performance across the optimal energy range (5-15keV) for macromolecular crystallography (MX) would be comparable to, or even exceed, that of many existing crystallography beamlines at higher energy facilities. Hence, starting in 1996, a suite of three beamlines, branching off a single wiggler source, was constructed, which together formed the ALS Macromolecular Crystallography Facility. From the outset this facility was designed to cater equally to the needs of both academic and industrial users with a heavy emphasis placed on the development and introduction of high throughput crystallographic tools, techniques, and facilities--such as large area CCD detectors, robotic sample handling and automounting facilities, a service crystallography program, and a tightly integrated, centralized, and highly automated beamline control environment for users. This facility was immediately successful, with the primary Multiwavelength Anomalous Diffraction beamline (5.0.2) in particular rapidly becoming one of the foremost crystallographic facilities in the US--responsible for structures such as the 70S ribosome. This success in-turn triggered enormous growth of the ALS macromolecular crystallography community and spurred the development of five additional ALS MX beamlines all utilizing the newly developed superconducting bending magnets ('superbends') as sources. However in the years since the original Sector 5.0 beamlines were built the performance demands of macromolecular crystallography users have become ever more exacting; with growing emphasis placed on studying larger complexes, more difficult structures, weakly diffracting or smaller
Two-dimensional steady unsaturated flow through embedded elliptical layers
Bakker, Mark; Nieber, John L.
2004-12-01
New analytic element solutions are presented for unsaturated, two-dimensional steady flow in vertical planes that include nonoverlapping impermeable elliptical layers and elliptical inhomogeneities. The hydraulic conductivity, which is represented by an exponential function of the pressure head, differs between the inside and outside of an elliptical inhomogeneity; both the saturated hydraulic conductivity and water retention parameters are allowed to differ between the inside and outside. The Richards equation is transformed, through the Kirchhoff transformation and a second standard transformation, into the modified Helmholtz equation. Analytic element solutions are obtained through separation of variables in elliptical coordinates. The resulting equations for the Kirchhoff potential consist of infinite sums of products of exponentials and modified Mathieu functions. In practical applications the series are truncated but still fulfill the differential equation exactly; boundary conditions are met approximately but up to machine accuracy, provided that enough terms are used. The pressure head, saturation, and flow may be computed analytically at any point in the vadose zone. Examples are given of the shadowing effect of an impermeable elliptical layer in a uniform flow field and funnel-type flow between two elliptical inhomogeneities. The presented solutions may be applied to study transport processes in vadose zones containing many impermeable elliptical layers or elliptical inhomogeneities.
Elliptical Orbit [arrow right] 1/r[superscript 2] Force
Prentis, Jeffrey; Fulton, Bryan; Hesse, Carol; Mazzino, Laura
2007-01-01
Newton's proof of the connection between elliptical orbits and inverse-square forces ranks among the "top ten" calculations in the history of science. This time-honored calculation is a highlight in an upper-level mechanics course. It would be worthwhile if students in introductory physics could prove the relation "elliptical orbit" [arrow right]…
Newton flows for elliptic functions: A pilot study
Twilt, F.; Helminck, G.F.; Snuverink, M.; van den Brug, L.
2008-01-01
Elliptic Newton flows are generated by a continuous, desingularized Newton method for doubly periodic meromorphic functions on the complex plane. In the special case, where the functions underlying these elliptic Newton flows are of second-order, we introduce various, closely related, concepts of
Multiple solutions for a quasilinear (p,q-elliptic system
Directory of Open Access Journals (Sweden)
Seyyed Mohsen Khalkhali
2013-06-01
Full Text Available In this article we show the existence of three weak solutions of a Dirichlet quasilinear elliptic system of differential equations which involves a general (p,q-elliptic operator in divergence, with $1
three solutions for a semilinear elliptic boundary value problem
Indian Academy of Sciences (India)
69
Keywords: The Laplacian operator, elliptic problem, Nehari man- ifold, three critical points, weak solution. 1. Introduction. Let Ω be a smooth bounded domain in RN , N ≥ 3 . In this work, we show the existence of at least three solutions for the semilinear elliptic boundary- value problem: (Pλ).. −∆u = f(x)|u(x)|p−2u(x) + ...
Rotational magnetization of anisotropic media: Lag angle, ellipticity and accommodation
International Nuclear Information System (INIS)
Kahler, G.R.; Della Torre, E.
2006-01-01
This paper discusses the change in the ellipticity of two-dimensional magnetization trajectories as the applied field rotates from the easy axis to the hard axis of a material. Furthermore, the impact that the reversible magnetization has on the ellipticity is discussed, including the relationship between the magnetization squareness and the reversible component of the magnetization
Weierstrass Elliptic Function Solutions to Nonlinear Evolution Equations
International Nuclear Information System (INIS)
Yu Jianping; Sun Yongli
2008-01-01
This paper is based on the relations between projection Riccati equations and Weierstrass elliptic equation, combined with the Groebner bases in the symbolic computation. Then the novel method for constructing the Weierstrass elliptic solutions to the nonlinear evolution equations is given by using the above relations
Near-infrared photometry of bright elliptical galaxies
Peletier, R. F.; Valentijn, E. A.; Jameson, R. F.
High-quality visual-infrared color profiles have been determined for elliptical galaxies for the first time. Surface photometry in J and K is presented for 12 bright elliptical galaxies, and the results have been combined with CCD data in visual passbands. It is shown that the galaxies become bluer
On the Behavior of Eisenstein Series Through Elliptic Degeneration
Garbin, D.; Pippich, A.-M. V.
2009-12-01
Let Γ be a Fuchsian group of the first kind acting on the hyperbolic upper half plane {mathbb{H}}, and let {M = Γbackslash mathbb{H}} be the associated finite volume hyperbolic Riemann surface. If γ is a primitive parabolic, hyperbolic, resp. elliptic element of Γ, there is an associated parabolic, hyperbolic, resp. elliptic Eisenstein series. In this article, we study the limiting behavior of these Eisenstein series on an elliptically degenerating family of finite volume hyperbolic Riemann surfaces. In particular, we prove the following result. The elliptic Eisenstein series associated to a degenerating elliptic element converges up to a factor to the parabolic Eisenstein series associated to the parabolic element which fixes the newly developed cusp on the limit surface.
Type-2 fuzzy elliptic membership functions for modeling uncertainty
DEFF Research Database (Denmark)
Kayacan, Erdal; Sarabakha, Andriy; Coupland, Simon
2018-01-01
Whereas type-1 and type-2 membership functions (MFs) are the core of any fuzzy logic system, there are no performance criteria available to evaluate the goodness or correctness of the fuzzy MFs. In this paper, we make extensive analysis in terms of the capability of type-2 elliptic fuzzy MFs...... in modeling uncertainty. Having decoupled parameters for its support and width, elliptic MFs are unique amongst existing type-2 fuzzy MFs. In this investigation, the uncertainty distribution along the elliptic MF support is studied, and a detailed analysis is given to compare and contrast its performance...... advantages mentioned above, elliptic MFs have comparable prediction results when compared to Gaussian and triangular MFs. Finally, in order to test the performance of fuzzy logic controller with elliptic interval type-2 MFs, extensive real-time experiments are conducted for the 3D trajectory tracking problem...
Ellipticity of near-threshold harmonics from stretched molecules.
Li, Weiyan; Dong, Fulong; Yu, Shujuan; Wang, Shang; Yang, Shiping; Chen, Yanjun
2015-11-30
We study the ellipticity of near-threshold harmonics (NTH) from aligned molecules with large internuclear distances numerically and analytically. The calculated harmonic spectra show a broad plateau for NTH which is several orders of magnitude higher than that for high-order harmonics. In particular, the NTH plateau shows high ellipticity at small and intermediate orientation angles. Our analyses reveal that the main contributions to the NTH plateau come from the transition of the electron from continuum states to these two lowest bound states of the system, which are strongly coupled together by the laser field. Besides continuum states, higher excited states also play a role in the NTH plateau, resulting in a large phase difference between parallel and perpendicular harmonics and accordingly high ellipticity of the NTH plateau. The NTH plateau with high intensity and large ellipticity provides a promising manner for generating strong elliptically-polarized extreme-ultraviolet (EUV) pulses.
Multipacting studies in elliptic SRF cavities
Prakash, Ram; Jana, Arup Ratan; Kumar, Vinit
2017-09-01
Multipacting is a resonant process, where the number of unwanted electrons resulting from a parasitic discharge rapidly grows to a larger value at some specific locations in a radio-frequency cavity. This results in a degradation of the cavity performance indicators (e.g. the quality factor Q and the maximum achievable accelerating gradient Eacc), and in the case of a superconducting radiofrequency (SRF) cavity, it leads to a quenching of superconductivity. Numerical simulations are essential to pre-empt the possibility of multipacting in SRF cavities, such that its design can be suitably refined to avoid this performance limiting phenomenon. Readily available computer codes (e.g.FishPact, MultiPac,CST-PICetc.) are widely used to simulate the phenomenon of multipacting in such cases. Most of the contemporary two dimensional (2D) codes such as FishPact, MultiPacetc. are unable to detect the multipacting in elliptic cavities because they use a simplistic secondary emission model, where it is assumed that all the secondary electrons are emitted with same energy. Some three-dimensional (3D) codes such as CST-PIC, which use a more realistic secondary emission model (Furman model) by following a probability distribution for the emission energy of secondary electrons, are able to correctly predict the occurrence of multipacting. These 3D codes however require large data handling and are slower than the 2D codes. In this paper, we report a detailed analysis of the multipacting phenomenon in elliptic SRF cavities and development of a 2D code to numerically simulate this phenomenon by employing the Furman model to simulate the secondary emission process. Since our code is 2D, it is faster than the 3D codes. It is however as accurate as the contemporary 3D codes since it uses the Furman model for secondary emission. We have also explored the possibility to further simplify the Furman model, which enables us to quickly estimate the growth rate of multipacting without
The ELSA laser beamline for electron polarization measurements via Compton backscattering
Energy Technology Data Exchange (ETDEWEB)
Switka, Michael; Hinterkeuser, Florian; Koop, Rebecca; Hillert, Wolfgang [Electron Stretcher Facility ELSA, Physics Institute of Bonn University (Germany)
2016-07-01
The Electron Stretcher Facility ELSA provides a spin polarized electron beam with energies of 0.5 - 3.2 GeV for double polarization hadron physics experiments. As of 2015, the laser beamline of the polarimeter based on Compton backscattering restarted operation. It consists of a cw disk laser with design total beam power of 40 W and features two polarized 515 nm photon beams colliding head-on with the stored electron beam in ELSA. The polarization measurement is based on the vertical profile asymmetry of the back-scattered photons, which is dependent on the polarization degree of the stored electron beam. After recent laser repairs, beamline and detector modifications, the properties of the beamline have been determined and first measurements of the electron polarization degree were conducted. The beamline performance and first measurements are presented.
I18--the microfocus spectroscopy beamline at the Diamond Light Source.
Mosselmans, J Frederick W; Quinn, Paul D; Dent, Andrew J; Cavill, Stuart A; Moreno, Sofia Diaz; Peach, Andrew; Leicester, Peter J; Keylock, Stephen J; Gregory, Simon R; Atkinson, Kirk D; Rosell, Josep Roque
2009-11-01
The design and performance of the microfocus spectroscopy beamline at the Diamond Light Source are described. The beamline is based on a 27 mm-period undulator to give an operable energy range between 2 and 20.7 keV, enabling it to cover the K-edges of the elements from P to Mo and the L(3)-edges from Sr to Pu. Micro-X-ray fluorescence, micro-EXAFS and micro-X-ray diffraction have all been achieved on the beamline with a spot size of approximately 3 microm. The principal optical elements of the beamline consist of a toroid mirror, a liquid-nitrogen-cooled double-crystal monochromator and a pair of bimorph Kirkpatrick-Baez mirrors. The performance of the optics is compared with theoretical values and a few of the early experimental results are summarized.
I19, the small-molecule single-crystal diffraction beamline at Diamond Light Source.
Nowell, Harriott; Barnett, Sarah A; Christensen, Kirsten E; Teat, Simon J; Allan, David R
2012-05-01
The dedicated small-molecule single-crystal X-ray diffraction beamline (I19) at Diamond Light Source has been operational and supporting users for over three years. I19 is a high-flux tunable-wavelength beamline and its key details are described in this article. Much of the work performed on the beamline involves structure determination from small and weakly diffracting crystals. Other experiments that have been supported to date include structural studies at high pressure, studies of metastable species, variable-temperature crystallography, studies involving gas exchange in porous materials and structural characterizations that require analysis of the diffuse scattering between Bragg reflections. A range of sample environments to facilitate crystallographic studies under non-ambient conditions are available as well as a number of options for automation. An indication of the scope of the science carried out on the beamline is provided by the range of highlights selected for this paper.
National Synchrotron Light Source user's manual: Guide to the VUV and x-ray beamlines
International Nuclear Information System (INIS)
Gmuer, N.F.
1993-04-01
The success of the National Synchrotron Light Source is based, in large part, on the size of the user community and the diversity of the scientific and technical disciplines represented by these users. As evidence of this success, the VUV Ring has just celebrated its 10th anniversary and the X-ray Ring will do the same in 1995. In order to enhance this success, the NSLS User's Manual: Guide to the VUV and X-Ray Beamlines - Fifth Edition, is being published. This Manual presents to the scientific community-at-large the current and projected architecture, capabilities and research programs of the various VUV and X-ray beamlines. Also detailed is the research and computer equipment a General User can expect to find and use at each beamline when working at the NSLS. The Manual is updated periodically in order to keep pace with the constant changes on these beamlines
Czech Academy of Sciences Publication Activity Database
Frank, Luděk
2008-01-01
Roč. 15, č. 1 (2008), s. 111-112 ISSN 1210-8529 Institutional research plan: CEZ:AV0Z20650511 Keywords : CESLAB * beamline * LEEM/PEEM Subject RIV: JA - Electronics ; Optoelectronics, Electrical Engineering
Safety upgrades for NSRRC beamlines in the upcoming top-up operation
International Nuclear Information System (INIS)
Liu, Joseph C.; Sheu, R.-J.; Wang, J.-P.; Chen, C.-R.; Chang, F.-D.; Kao, S.-P.
2006-01-01
The original beamline shielding of NSRRC was designed for the decay mode operation that safety shutter was closed during injection. The proposed top-up operation that opens safety shutter during top-up injection will introduce additional beam loss scenarios and radiation sources, especially when the injection efficiency needs to be improved. Careful comparison was made to differentiate the radiation doses into beamlines for both operation modes. Detailed evaluation was made to identify the possible inadequacies of the old beamline shielding and safety control procedures. Remedy actions and safety upgrades for each individual beamline were issued to ensure that dose limit of 2 mSv/yr for users can be fulfilled when running top-up operation
The bio-crystallography beamline (BL41XU) at SPring-8
Kawamoto, M; Kamiya, N
2001-01-01
The bio-crystallography beamline (BL41XU), one of two pilot beamlines at SPring-8, was constructed using a standard in-vacuum-type undulator and opened for general users from domestic and overseas countries. Many tests and improvements were carried out on beamline elements and equipment for macromolecular crystallography, especially on the so-called 'pin-post' water cooling crystal of rotated-inclined double crystal monochromator. The maximum brilliance at sample position reached to 4x10 sup 1 sup 5 photons/s/mm sup 2 /mrad sup 2 at an X-ray energy of 11 keV. Commercially available X-ray detectors of CCD and imaging plate were installed in the experimental station. A beamline control software system for beam tracking and an on-line reader for large-format imaging plate were newly developed.
PF-AR NW14, a new time-resolved diffraction/scattering beamline
International Nuclear Information System (INIS)
Nozawa, Shunsuke; Adachi, Shin-ichi; Tazaki, Ryoko; Takahashi, Jun-ichi; Itatani, Jiro; Daimon, Masahiro; Mori, Takeharu; Sawa, Hiroshi; Kawata, Hiroshi; Koshihara, Shin-ya
2005-01-01
NW14 is a new insertion device beamline at the Photon Factory Advanced Ring (PF-AR), which is a unique ring with full-time single-bunched operation, aiming for timeresolved x-ray diffraction/scattering and XAFS experiments. The primary scientific goal of this beamline is to observe the ultrafast dynamics of condensed matter systems such as organic and inorganic crystals, biological systems and liquids triggered by optical pulses. With the large photon fluxes derived from the undulator, it should become possible to take a snapshoot an atomic-scale image of the electron density distribution. By combining a series of images it is possible to produce a movie of the photo-induced dynamics with 50-ps resolution. The construction of the beamline is being funded by the ERATO Koshihara Non-equilibrium Dynamics Project of the Japan Science and Technology Agency (JST), and the beamline will be operational from autumn 2005
PF-AR NW14, a new time-resolved diffraction/scattering beamline
Nozawa, Shunsuke; Adachi, Shin-ichi; Tazaki, Ryoko; Takahashi, Jun-ichi; Itatani, Jiro; Daimon, Masahiro; Mori, Takeharu; Sawa, Hiroshi; Kawata, Hiroshi; Koshihara, Shin-ya
2005-01-01
NW14 is a new insertion device beamline at the Photon Factory Advanced Ring (PF-AR), which is a unique ring with full-time single-bunched operation, aiming for timeresolved x-ray diffraction/scattering and XAFS experiments. The primary scientific goal of this beamline is to observe the ultrafast dynamics of condensed matter systems such as organic and inorganic crystals, biological systems and liquids triggered by optical pulses. With the large photon fluxes derived from the undulator, it should become possible to take a snapshoot an atomic-scale image of the electron density distribution. By combining a series of images it is possible to produce a movie of the photo-induced dynamics with 50-ps resolution. The construction of the beamline is being funded by the ERATO Koshihara Non-equilibrium Dynamics Project of the Japan Science and Technology Agency (JST), and the beamline will be operational from autumn 2005.
Safety Analysis Report: X17B2 beamline Synchrotron Medical Research Facility
International Nuclear Information System (INIS)
Gmuer, N.F.; Thomlinson, W.
1990-02-01
This report contains a safety analysis for the X17B2 beamline synchrotron medical research facility. Health hazards, risk assessment and building systems are discussed. Reference is made to transvenous coronary angiography
Evaluation of beam-line components for use in a large neutral-beam injector
International Nuclear Information System (INIS)
Fink, J.H.
1977-01-01
A conceptual model of a neutral-beam injector was used to examine the effect of beam-line components on reactor performance. Criteria were established to optimize a reactor's reliability and minimize its cost
Magnetic elliptical polarization of Schumann resonances
International Nuclear Information System (INIS)
Sentman, D.D.
1987-01-01
Measurements of orthogonal, horizontal components of the magnetic field in the ELF range obtained during September 1985 show that the Schumann resonance eigenfrequencies determined separately for the north-south and east-west magnetic components differ by as much as 0.5 Hz, suggesting that the underlying magnetic signal is not linearly polarized at such times. The high degree of magnetic ellipticity found suggests that the side multiplets of the Schumann resonances corresponding to azimuthally inhomogeneous normal modes are strongly excited in the highly asymmetric earth-ionosphere cavity. The dominant sense of polarization over the measurement passband is found to be right-handed during local daylight hours, and to be left-handed during local nighttime hours. 16 references
Elliptic flow and incomplete equilibration at RHIC
Bhalerao, R S; Borghini, N; Ollitrault, Jean Yves
2005-01-01
We argue that RHIC data, in particular those on the anisotropic flow coefficients v_2 and v_4, suggest that the matter produced in the early stages of nucleus-nucleus collisions is incompletely thermalized. We interpret the parameter (1/S)(dN/dy), where S is the transverse area of the collision zone and dN/dy the multiplicity density, as an indicator of the number of collisions per particle at the time when elliptic flow is established, and hence as a measure of the degree of equilibration. This number serves as a control parameter which can be varied experimentally by changing the system size, the centrality or the beam energy. We provide predictions for Cu-Cu collisions at RHIC as well as for Pb-Pb collisions at the LHC.
International Nuclear Information System (INIS)
Yamada, Yusuke; Hiraki, Masahiko; Matsugaki, Naohiro; Chavas, Leonard M.G.; Igarashi, Noriyuki; Wakatsuki, Soichi
2012-01-01
Fully automated data collection and processing system has been developed on macromolecular crystallography beamlines at the Photon Factory. In this system, the sample exchange, centering and data collection are sequentially performed for all samples stored in the sample exchange system at a beamline without any manual operations. Data processing of collected data sets is also performed automatically. These results are stored into the database system, and users can monitor the progress and results of automated experiment via a Web browser. (author)
The design of the beamline for magnetic circular dichroism study at HNSRL
ShengWeiFan; Li Da Shi; Yan Yong Lia
2000-01-01
A synchrotron radiation beamline for MCD study is being developed at Hefei National Synchrotron Radiation Laboratory (HNSRL) in the P. R. China. As a key component on the beamline, the varied line-spacing plane grating (VLSPG) monochromator is designed to cover a broad photon energy range from 100 to 1000 eV with a medium energy resolving power of 2000. The design is analyzed and optimized to meet the high requirements of MCD studying.
Inelastic X-ray Scattering Beamline Collaborative Development Team Final Report
International Nuclear Information System (INIS)
Burns, Clement
2008-01-01
This is the final report for the project to create a beam line for inelastic x-ray scattering at the Advanced Photon Source. The facility is complete and operating well, with spectrometers for both high resolution and medium resolution measurements. With the advent of third generation synchrotron sources, inelastic x-ray scattering (IXS) has become a valuable technique to probe the electronic and vibrational states of a wide variety of systems of interest in physics, chemistry, and biology. IXS is a weak probe, and experimental setups are complex and require well-optimized spectrometers which need a dedicated beamline to function efficiently. This project was the result of a proposal to provide a world-class, user friendly beamline for IXS at the Advanced Photon Source. The IXS Collaborative Development Team (IXS-CDT) was formed from groups at the national laboratories and a number of different universities. The beamline was designed from the front end to the experimental stations. Two different experimental stations were provided, one for medium resolution inelastic x-ray scattering (MERIX) and a spectrometer for high resolution inelastic x-ray scattering (HERIX). Funding for this project came from several sources as well as the DOE. The beamline is complete with both spectrometers operating well. The facility is now open to the general user community and there has been a tremendous demand to take advantage of the beamline's capabilities. A large number of different experiments have already been carried out on the beamline. A detailed description of the beamline has been given in the final design report (FDR) for the beamline from which much of the material in this report came. The first part of this report contains a general overview of the project with more technical details given later.
Construction and characterization of a laser-driven proton beamline at GSI
Busold, Simon
2014-01-01
The thesis includes the first experiments with the new 100 TW laser beamline of the PHELIX laser facility at GSI Darmstadt to drive a TNSA (Target Normal Sheath Acceleration) proton source at GSI's Z6 experimental area. At consecutive stages a pulsed solenoid has been applied for beam transport and energy selection via chromatic focusing, as well as a radiofrequency cavity for energy compression of the bunch. This novel laser-driven proton beamline, representing a central experiment of the...
eBooking of beam-time over internet for beamlines of Indus synchrotron radiation sources
International Nuclear Information System (INIS)
Jain, Alok; Verma, Rajesh; Rajan, Alpana; Modi, M.H.; Rawat, Anil
2015-01-01
Users from various research labs and academic institutes carry out experiments on beamlines of two Synchrotron Radiation Sources Indus-1 and Indus-2 available at RRCAT, Indore. To carry out experimental work on beamlines of both synchrotron radiation sources, beam-time is booked over Internet by the users of beamlines using user portal designed, developed and deployed over Internet. This portal has made the process of beamtime booking fast, hassle free and paperless as manual booking of beam-time for carrying out experiment on a particular beamline is cumbersome. The portal facilitates in-charge of Indus-1 and Indus-2 beamlines to keep track of users' records, work progress and other activities linked to experiments carried on beamlines. It is important to keep record and provide statistics about the usage of the beam lines from time-to-time. The user portal for e-booking of beam-time has been developed in-house using open source software development tools. Multi-step activities of users and beamline administrators are workflow based with seamless flow of information across various modules and fully authenticated using role based mechanism for different roles of software usage. The software is in regular use since November 2013 and has helped beamline in- charges in efficiently managing various activities related to user registration, booking of beam-time, booking of Guest House, Generation of Security permits, User feedback etc. Design concept, role based authentication mechanism and features provided by the web portal are discussed in detail in this paper. (author)
General design of the layout for new undulator-only beamline front ends
International Nuclear Information System (INIS)
Shu Deming; Ramanathan, Mohan; Kuzay, Tuncer M.
2001-01-01
A great majority of the Advanced Photon Source (APS) users have chosen an undulator as the only source for their insertion device beamline. Compared with a wiggler source, the undulator source has a much smaller horizontal divergence, providing us with an opportunity to optimize the beamline front-end design further. In this paper, the particular designs and specifications, as well as the optical and bremsstrahlung ray-tracing analysis of the new APS front ends for undulator-only operation are presented
Protein crystallography beamline (PX-BL21); its utilization and research highlights
International Nuclear Information System (INIS)
Kumar, Ashwani; Ghosh, Biplab; Singh, Rahul; Makde, Ravindra; Sharma, Surinder M.
2016-01-01
The protein crystallography beamline (PX-BL21) is sourced on 1.5 T bending magnet of 2.5 GeV Indus-2 synchrotron. This beamline has been designed to perform monochromatic and anomalous diffraction experiments on single crystals of biological macromolecules such as protein, DNA and their complexes. PX beamline also has a state-of-art ancillary biochemical laboratory to prepare single crystals of biological macromolecules. Since the commissioning of the beamline, it has been utilized by more than 70% of research groups working in the area of protein crystallography in India. About 30 crystal structures of proteins, determined using this beamline, have been deposited in Protein Data Bank (PDB). Some of these structures have been determined using experimental phasing, such as the single wavelength anomalous diffraction (SAD) experiments. The energy tunability of the synchrotron have been exploited to carry our various SAD experiments: Selenium-SAD, Zinc-SAD and Manganese-SAD and Sulphar-SAD. In the present talk, the key results from the PX-BL21 beamline will be discussed. (author)
International Nuclear Information System (INIS)
Yang, C.Y.; Penner-Hahn, J.E.; Stefan, P.M.
1989-01-01
Characterization of the X-19A beamline at the National Synchrotron Light Source (NSLS) is described. The beamline is designed for high resolution x-ray absorption spectroscopy over a wide energy range. All of the beamline optical components are compatible with ultrahigh vacuum (UHV) operation. This permits measurements to be made in a window-less mode, thereby facilitating lower energy (<4 KeV) studies. To upgrade the beamline performance, several possible improvements in instrumentation and practice are discussed to increase photon statistics with an optimum energy resolution, while decreasing the harmonic contamination and noise level. A special effort has been made to improve the stability and UHV compatibility of the monochromator system. Initial x-ray absorption results demonstrate the capabilities of this beamline for x-ray absorption studies of low Z elements (e.g. S) in highly dilute systems. The future use of this beamline for carrying out various x-ray absorption experiments is presented. 10 refs., 4 figs
EMA beamline at SIRIUS: extreme condition X-ray methods of analysis
International Nuclear Information System (INIS)
Souza Neto, Narcizo
2016-01-01
Full text: The EMA beamline (Extreme condition X-ray Methods of Analysis) is one of the hard x-ray undulator beamlines within the first phase of the new synchrotron source in Brazil (Sirius project). This beamline is thought to make a difference where a high brilliance (high flux of up to 2 x 10 14 photons/sec with beam size down to 0.5 x 0.5 μm 2 ) is essential, which is the case for extreme pressures that require small focus and time-resolved that require high photon flux. With that in mind we propose the beamline to have two experimental hutches to cover most of the extreme condition techniques today employed at synchrotron laboratories worldwide. These two stations are thought to provide the general infrastructure for magnets and lasers experiments, which may evolve as new scientific problems appear. In addition to the hutches, support laboratories will be strongly linked and supportive to the experiments at the beamline, covering high pressure instrumentations using diamond anvil cells and pump-and-probe requirements for ultrafast and high power lasers. Along these lines, we will describe the following techniques covered at this beamline: magnetic spectroscopy (XMCD) and scattering (XRMS) under high pressure and very low temperature in order to fully probe both ferromagnetic and antiferromagnetic materials and the dependence with pressure; extreme pressure and temperature XRD and XAS experiments using very small diamond culet anvils and high power lasers. (author)
EMA beamline at SIRIUS: extreme condition X-ray methods of analysis
Energy Technology Data Exchange (ETDEWEB)
Souza Neto, Narcizo, E-mail: narcizo.souza@lnls.br [Centro Nacional de Pesquisa em Energia e Materiais (CNPEM), Campinas, SP (Brazil)
2016-07-01
Full text: The EMA beamline (Extreme condition X-ray Methods of Analysis) is one of the hard x-ray undulator beamlines within the first phase of the new synchrotron source in Brazil (Sirius project). This beamline is thought to make a difference where a high brilliance (high flux of up to 2 x 10{sup 14} photons/sec with beam size down to 0.5 x 0.5 μm{sup 2}) is essential, which is the case for extreme pressures that require small focus and time-resolved that require high photon flux. With that in mind we propose the beamline to have two experimental hutches to cover most of the extreme condition techniques today employed at synchrotron laboratories worldwide. These two stations are thought to provide the general infrastructure for magnets and lasers experiments, which may evolve as new scientific problems appear. In addition to the hutches, support laboratories will be strongly linked and supportive to the experiments at the beamline, covering high pressure instrumentations using diamond anvil cells and pump-and-probe requirements for ultrafast and high power lasers. Along these lines, we will describe the following techniques covered at this beamline: magnetic spectroscopy (XMCD) and scattering (XRMS) under high pressure and very low temperature in order to fully probe both ferromagnetic and antiferromagnetic materials and the dependence with pressure; extreme pressure and temperature XRD and XAS experiments using very small diamond culet anvils and high power lasers. (author)
Realtime control system for microprobe beamline at PLS
Energy Technology Data Exchange (ETDEWEB)
Yoon, J.C.; Lee, J.W.; Kim, K.H.; Ko, I.S. [Pohang Accelerator Laboratory, POSTECH, Pohang (Korea)
1998-11-01
The microprobe beamline of the Pohang Light Source (PLS) consists of main and second slits, a microprobe system, two ion chambers, a video-microscope, and a Si(Li) detector. These machine components must be controlled remodely through the computer system to make user's experiments precise and speedy. A real-time computer control system was developed to control and monitor these components. A VMEbus computer with an OS-9 real-time operating system was used for the low-level data acquisition and control. VME I/O modules were used for the step motor control and the scalar control. The software has a modular structure for the maximum performance and the easy maintenance. We developed the database, the I/O driver, and the control software. We used PC/Windows 95 for the data logging and the operator interface. Visual C{sup ++} was used for the graphical user interface programming. RS232C was used for the communication between the VME and the PC. (author)
Mirror boxes and mirror mounts for photophysics beamline
International Nuclear Information System (INIS)
Raja Rao, P.M.; Raja Sekhar, B.N.; Das, N.C.; Khan, H.A.; Bhattacharya, S.S.; Roy, A.P.
1996-01-01
Photophysics beamline makes use of one metre Seya-Namioka monochromator and two toroidal mirrors in its fore optics. The first toroidal mirror (pre mirror) focuses light originating from the tangent point of the storage ring onto the entrance slit of the monochromator and second toroidal mirror (post mirror) collects light from the exit slit of the monochromator and focuses light onto the sample placed at a distance of about one metre away from the 2nd mirror. To steer light through monochromator and to focus it on the sample of 1mm x 1mm size require precision rotational and translational motion of the mirrors and this has been achieved with the help of precision mirror mounts. Since Indus-1 operates at pressures less than 10 -9 m.bar, the mirror mounts should be manipulated under similar ultra high vacuum conditions. Considering these requirements, two mirror boxes and two mirror mounts have been designed and fabricated. The coarse movements to the mirrors are imparted from outside the mirror chamber with the help of x-y tables and precision movements to the mirrors are achieved with the help of mirror mounts. The UHV compatibility and performance of the mirror mounts connected to mirror boxes under ultra high vacuum condition is evaluated. The details of the design, fabrication and performance evaluation are discussed in this report. 5 refs., 9 figs., 1 tab
The EMIL project at BESSY II: Beamline design and performance
Energy Technology Data Exchange (ETDEWEB)
Hendel, Stefan, E-mail: stefan.hendel@helmholtz-berlin.de; Schäfers, Franz; Reichardt, Gerd; Scheer, Michael; Bahrdt, Johannes; Lips, Klaus [Helmholtz-Zentrum Berlin für Materialien und Energie GmbH, Albert-Einstein-Straße 15, 12489 Berlin (Germany); Hävecker, Michael [Helmholtz-Zentrum Berlin für Materialien und Energie GmbH, Albert-Einstein-Straße 15, 12489 Berlin (Germany); MPI for Chemical Energy Conversion, Stiftstrasse 34-36, 45470 Mülheim an der Ruhr (Germany)
2016-07-27
The Energy Materials In-Situ Laboratory Berlin (EMIL) at BESSY-II is currently under construction. Two canted undulators for soft- and hard X-rays will be installed into the BESSY II storage ring in one straight section, complex beamlines with more than twenty optical elements will be set up and a new laboratory building attached to BESSY II will host three endstations and a large UHV-transfer system connecting various HV- and UHV-deposition systems. The undulators, UE48 and U17, provide a broad energy spectrum of 80 - 10000 eV, of which the harder radiation (>700 eV) is provided by a cryogenic in-vacuum device. Three monochromators (two plane grating monochromators (PGM) and one LN{sub 2}-cooled double crystal monochromator (DCM)) disperse the radiation into separate pathways of 65 m length, while downstream of the monochromators split-mirror chambers distribute the photon beam to one (or simultaneously to two) of five upcoming endstations. Three of these endstations are designed for the full energy range with spatial overlap of the soft and hard foci, whereas one endstation (PEEM) uses only the soft and another one (PINK) only the hard branch, respectively.
Neutrino Flux Prediction for the NuMI Beamline
Energy Technology Data Exchange (ETDEWEB)
Aliaga Soplin, Leonidas [William-Mary Coll.
2016-01-01
The determination of the neutrino flux in any conventional neutrino beam presents a challenge for the current and future short and long baseline neutrino experiments. The uncertainties associated with the production and attenuation of the hadrons in the beamline materials along with those associated with the beam optics have a big effect in the flux spectrum knowledge. For experiments like MINERvA, understanding the flux is crucial since it enters directly into every neutrino-nucleus cross-sections measurements. The foundation of this work is predicting the neutrino flux at MINERvA using dedicated measurements of hadron production in hadron-nucleus collisions and incorporating in-situ MINERvA data that can provide additional constraints. This work also includes the prospect for predicting the flux at other detectors like the NOvA Near detector. The procedure and conclusions of this thesis will have a big impact on future hadron production experiments and on determining the fl ux for the upcoming DUNE experiment.
Neutrino Flux Prediction for the NuMI Beamline
Energy Technology Data Exchange (ETDEWEB)
Soplin, Leonidas Aliaga [Coll. William and Mary
2016-01-01
The determination of the neutrino flux in any conventional neutrino beam presents a challenge for the current and future short and long baseline neutrino experiments. The uncertainties associated with the production and attenuation of the hadrons in the beamline materials along with those associated with the beam optics have a big effect in the flux spectrum knowledge. For experiments like MINERvA, understanding the flux is crucial since it enters directly into every neutrino-nucleus cross-sections measurements. The foundation of this work is predicting the neutrino flux at MINERvA using dedicated measurements of hadron production in hadron-nucleus collisions and incorporating in-situ MINERvA data that can provide additional constraints. This work also includes the prospect for predicting the flux at other detectors like the NOvA Near detector. The procedure and conclusions of this thesis will have a big impact on future hadron production experiments and on determining the flux for the upcoming DUNE experiment.
An x-ray beamline for the LNLS
International Nuclear Information System (INIS)
Cusatis, C.; Giles, C.
1990-01-01
In this paper an x-ray optics and high energy-resolution beamline for the XUV ring at LNLS is described. It consists of a monochromator serving a station for x-ray optics and another for high energy-resolution experiments. The constant offset, vacuum compatible double crystal monochromator is to be mounted in a bending magnet or wiggler port and will be located as near as possible of the ring. One rotation movement of both mechanically linked crystals and one short elastic translation of each crystal covers the energy range of 1 to 10 KeV. For θ B ∼ 90 on the first monochromator crystal the monochromatized beam is sent back through the electron ring and is received on the opposite side of the line at the high energy-resolution station. The main instrument at the x-ray optics station will be a triple axis diffractometer. This station, located as far as possible of the ring and, possibly, outside of main hall, will be used for x-ray interferometry, perfect crystal diffractometry, etc
Optimized baffle and aperture placement in neutral beamlines
International Nuclear Information System (INIS)
Stone, R.; Duffy, T.; Vetrovec, J.
1983-01-01
Most neutral beamlines contain an iron-core ion-bending magnet that requires shielding between the end of the neutralizer and this magnet. This shielding allows the gas pressure to drop prior to the beam entering the magnet and therefore reduces beam losses in this drift region. We have found that the beam losses can be reduced even further by eliminating the iron-core magnet and the magnetic shielding altogether. The required bending field can be supplied by current coils without the iron poles. In addition, placement of the baffles and apertures can affect the cold gas entering the plasma region and the losses in the neutral beam due to re-ionization. In our study we varied the placement of the baffles, which determine the amount of pumping in each chamber, and the apertures, which determine the beam loss. Our results indicate that a baffle/aperture configuration can be set for either minimum cold gas into the plasma region or minimum beam losses, but not both
Optimization of High-Energy Implanter Beamline Pumping
International Nuclear Information System (INIS)
LaFontaine, Marvin; Pharand, Michel; Huang Yongzhang; Pokidov, Ilya; Ferrara, Joseph
2006-01-01
A high-energy implanter process chamber and its pumping configuration were designed to minimize the residual gas density in the endstation. A modified Nastran trade mark sign finite-element analysis (FEA) code was used to calculate the pressure distribution and gas flow within the process chamber. The modified FE method was readily applied to the internal geometry of the scan chamber, the corrector magnet waveguide, and the process chamber, which included the scan arm assembly, 300mm wafer, and plasma electron flood gun (PEF). Using the modified Nastran code, the gas flow and pressure distribution within the beamline geometry were calculated. The gas load consisted of H2, which is generated by photoresist (PR) outgassing from the 300mm wafer, and Xe from the plasma electron flood gun. Several pumping configurations were assessed, with each consisting of various locations and pumping capacities of vacuum pumps. The pressure distribution results for each configuration are presented, along with pumping efficiency results which are helpful in selecting the optimum pump configuration. The analysis results were compared to measured data, indicating a good correlation between the two
Emittance measurements in Grumman 1 MeV beamline
International Nuclear Information System (INIS)
Debiak, T.; Gammel, G.; Melnychuk, S.
1992-01-01
The emittance of a 30 keV H - beam has been measured with an Allison type electrostatic analyser at two positions separated by 85 cm along the Grumman 1 MeV beamline LEBT at low currents (about 4 mA, no Cs 2 O additive in the source) and at higher currents (10-15 mA, with Cs 2 O additive in the source). No emittance growth was observed between the two positions, but, at the higher current level, the emittance was about 60% higher than at the low current level (Σ n ,rms = .0045 π cm-mrad vs. 0070 π cm-mrad). Argon was then introduced up to a partial pressure of 4x10 -5 torr, and the emittance decreased back to a range corresponding to that found at the lower currents. However, beam noise was observed at the downstream position, and there is evidence for a small amount of emittance growth (<20%) between the two positions
Effective Elliptic Models for Efficient Wavefield Extrapolation in Anisotropic Media
Waheed, Umair bin
2014-05-01
Wavefield extrapolation operator for elliptically anisotropic media offers significant cost reduction compared to that of transversely isotropic media (TI), especially when the medium exhibits tilt in the symmetry axis (TTI). However, elliptical anisotropy does not provide accurate focusing for TI media. Therefore, we develop effective elliptically anisotropic models that correctly capture the kinematic behavior of the TTI wavefield. Specifically, we use an iterative elliptically anisotropic eikonal solver that provides the accurate traveltimes for a TI model. The resultant coefficients of the elliptical eikonal provide the effective models. These effective models allow us to use the cheaper wavefield extrapolation operator for elliptic media to obtain approximate wavefield solutions for TTI media. Despite the fact that the effective elliptic models are obtained by kinematic matching using high-frequency asymptotic, the resulting wavefield contains most of the critical wavefield components, including the frequency dependency and caustics, if present, with reasonable accuracy. The methodology developed here offers a much better cost versus accuracy tradeoff for wavefield computations in TTI media, considering the cost prohibitive nature of the problem. We demonstrate the applicability of the proposed approach on the BP TTI model.
Effective Elliptic Models for Efficient Wavefield Extrapolation in Anisotropic Media
Waheed, Umair bin; Alkhalifah, Tariq Ali
2014-01-01
Wavefield extrapolation operator for elliptically anisotropic media offers significant cost reduction compared to that of transversely isotropic media (TI), especially when the medium exhibits tilt in the symmetry axis (TTI). However, elliptical anisotropy does not provide accurate focusing for TI media. Therefore, we develop effective elliptically anisotropic models that correctly capture the kinematic behavior of the TTI wavefield. Specifically, we use an iterative elliptically anisotropic eikonal solver that provides the accurate traveltimes for a TI model. The resultant coefficients of the elliptical eikonal provide the effective models. These effective models allow us to use the cheaper wavefield extrapolation operator for elliptic media to obtain approximate wavefield solutions for TTI media. Despite the fact that the effective elliptic models are obtained by kinematic matching using high-frequency asymptotic, the resulting wavefield contains most of the critical wavefield components, including the frequency dependency and caustics, if present, with reasonable accuracy. The methodology developed here offers a much better cost versus accuracy tradeoff for wavefield computations in TTI media, considering the cost prohibitive nature of the problem. We demonstrate the applicability of the proposed approach on the BP TTI model.
Investigation on computation of elliptical microwave plasma cavity
Liao, Xiaoli; Liu, Hua; Zhang, Kai
2008-12-01
In recent years, the advance of the elliptical resonant cavity and focus cavity is known by many people. There are homogeneous and multipatternal virtues in the focus dimensional microwave field of the elliptical resonant cavity. It is very suitable for applying the low power microwave biological effect equipment. However, when designing the elliptical resonant cavity may meet the problems of complex and huge computation need to be solved. This paper proposed the simple way of approximate processing the Mathieu function. It can greatly simplify the difficulty and decrease the scale of computation. This method can satisfy the requirements of research and development within project permitted precision.
Multilevel quadrature of elliptic PDEs with log-normal diffusion
Harbrecht, Helmut
2015-01-07
We apply multilevel quadrature methods for the moment computation of the solution of elliptic PDEs with lognormally distributed diffusion coefficients. The computation of the moments is a difficult task since they appear as high dimensional Bochner integrals over an unbounded domain. Each function evaluation corresponds to a deterministic elliptic boundary value problem which can be solved by finite elements on an appropriate level of refinement. The complexity is thus given by the number of quadrature points times the complexity for a single elliptic PDE solve. The multilevel idea is to reduce this complexity by combining quadrature methods with different accuracies with several spatial discretization levels in a sparse grid like fashion.
Convergence criteria for systems of nonlinear elliptic partial differential equations
International Nuclear Information System (INIS)
Sharma, R.K.
1986-01-01
This thesis deals with convergence criteria for a special system of nonlinear elliptic partial differential equations. A fixed-point algorithm is used, which iteratively solves one linearized elliptic partial differential equation at a time. Conditions are established that help foresee the convergence of the algorithm. Under reasonable hypotheses it is proved that the algorithm converges for such nonlinear elliptic systems. Extensive experimental results are reported and they show the algorithm converges in a wide variety of cases and the convergence is well correlated with the theoretical conditions introduced in this thesis
Dusty Feedback from Massive Black Holes in Two Elliptical Galaxies
Temi, P.; Brighenti, F.; Mathews, W. G.; Amblard, A.; Riguccini, L.
2013-01-01
Far-infrared dust emission from elliptical galaxies informs us about galaxy mergers, feedback energy outbursts from supermassive black holes and the age of galactic stars. We report on the role of AGN feedback observationally by looking for its signatures in elliptical galaxies at recent epochs in the nearby universe. We present Herschel observations of two elliptical galaxies with strong and spatially extended FIR emission from colder grains 5-10 kpc distant from the galaxy cores. Extended excess cold dust emission is interpreted as evidence of recent feedback-generated AGN energy outbursts in these galaxies, visible only in the FIR, from buoyant gaseous outflows from the galaxy cores.
UHV mirror mounts for photophysics beamline at Indus-I
International Nuclear Information System (INIS)
Meenakshi Raja Rao, P.; Bhattacharya, S.S.; Das, N.C.; Rajasekhar, B.N.; Roy, A.P.
1995-01-01
Photophysics beamline makes use of a combination of two toroidal mirrors and one meter Seya-Namioka Monochromator in its fore optics. The fore optics monochromatises and steers the synchrotron radiation source (SRS) beam from its tangent point to the sample situated at a distance of about five meters. Slit widths of the monochromator are of the order of 100μ and the sample size is one mm 2 . Hence it is essential to impart precision rotational and translational movements of the same order of magnitude to the mirrors with the use of appropriate mirror mounts. Since Indus-1 operates at a pressure -9 mbar, the mirror mounts should be UHV compatible and the movements should be actuated under UHV. The mirrors along with the mirror mounts are enclosed in UHV chambers. The mirror chambers have been fabricated at Centre for Advanced Technology (CAT) workshops and tested up to a pressure of 10 -9 mbar. The mirror mounts are designed, fabricated and leak checked (He leak rate -10 std cc/s) The precision movements are achieved with the help of bellow sealed shaft mechanism and adjustable screws provided with the kinematic mount of the mirror frame. The performance of the mirror mount was tested at atmospheric pressure by using a laser beam and found to be good. The minimum displacement of the laser beam at slit and sample positions is ∼ 70μ which is quite adequate for optical alignment. The performance of the mirror mount under UHV conditions is being evaluated. (author). 4 refs., 3 figs
RBF Multiscale Collocation for Second Order Elliptic Boundary Value Problems
Farrell, Patricio; Wendland, Holger
2013-01-01
In this paper, we discuss multiscale radial basis function collocation methods for solving elliptic partial differential equations on bounded domains. The approximate solution is constructed in a multilevel fashion, each level using compactly
Multilevel quadrature of elliptic PDEs with log-normal diffusion
Harbrecht, Helmut; Peters, Michael; Siebenmorgen, Markus
2015-01-01
Each function evaluation corresponds to a deterministic elliptic boundary value problem which can be solved by finite elements on an appropriate level of refinement. The complexity is thus given by the number
C1,1 regularity for degenerate elliptic obstacle problems
Daskalopoulos, Panagiota; Feehan, Paul M. N.
2016-03-01
The Heston stochastic volatility process is a degenerate diffusion process where the degeneracy in the diffusion coefficient is proportional to the square root of the distance to the boundary of the half-plane. The generator of this process with killing, called the elliptic Heston operator, is a second-order, degenerate-elliptic partial differential operator, where the degeneracy in the operator symbol is proportional to the distance to the boundary of the half-plane. In mathematical finance, solutions to the obstacle problem for the elliptic Heston operator correspond to value functions for perpetual American-style options on the underlying asset. With the aid of weighted Sobolev spaces and weighted Hölder spaces, we establish the optimal C 1 , 1 regularity (up to the boundary of the half-plane) for solutions to obstacle problems for the elliptic Heston operator when the obstacle functions are sufficiently smooth.
Electromagnetic fields and Green functions in elliptical vacuum chambers
AUTHOR|(CDS)2084216; Biancacci, Nicolo; Migliorati, Mauro; Palumbo, Luigi; Vaccaro, Vittorio; CERN. Geneva. ATS Department
2017-01-01
In this paper, we discuss the electromagnetic interaction between a point charge travelling inside a waveguide of elliptical cross section, and the waveguide itself. By using a convenient expansion of the Mathieu functions, useful in particular for treating a variety of problems in applied mathematics and physics with elliptic geometry, we first obtain the longitudinal electromagnetic field of a point charge (Green function) in free space in terms of elliptical coordinates. This expression allows, then, to calculate the scattered field due to the boundary conditions in our geometry. By summing the contribution of the direct or primary field and the indirect field scattered by the boundary, after a careful choice of some expansion expressions, we derive a novel formula of the longitudinal electric field, in any transverse position of the elliptical cross section, generated by the charge moving along the longitudinal axis of the waveguide. The obtained expression is represented in a closed form, it can be diffe...
Vertical elliptic operator for efficient wave propagation in TTI media
Waheed, Umair bin; Alkhalifah, Tariq Ali
2015-01-01
Elliptic wave extrapolation operators require significantly less computational cost than the ones for transversely isotropic (TI) media. However, it does not provide accurate wavefield representation or imaging for the prevalent TI media. We propose a new vertical elliptically anisotropic (VEA) wave equation by decomposing the acoustic TI pseudo-differential wave equation. The decomposition results in a vertical elliptic differential equation and a scalar operator. The new VEA-like wave equation shares the same dispersion relation as that of the original acoustic TI wave equation. Therefore, the kinematic contents are correctly matched to the original equation. Moreover, the proposed decomposition yields better amplitude properties than the isotropic decomposition without increasing the computational load. Therefore, it exhibits better cost versus accuracy tradeoff compared to the isotropic or the tilted elliptic decompositions. We demonstrate with numerical examples that the proposed methodology is numerically stable for complex models and is free from shear-wave artifacts.
Topology of the elliptical billiard with the Hooke's potential
Directory of Open Access Journals (Sweden)
Radnović Milena
2015-01-01
Full Text Available Using Fomenko graphs, we present a topological description of the elliptical billiard with Hooke's potential. [Projekat Ministarstva nauke Republike Srbije, br. 174020: Geometry and Topology of Manifolds and Integrable Dynamical Systems
Electron energy spectrum in core-shell elliptic quantum wire
Directory of Open Access Journals (Sweden)
V.Holovatsky
2007-01-01
Full Text Available The electron energy spectrum in core-shell elliptic quantum wire and elliptic semiconductor nanotubes are investigated within the effective mass approximation. The solution of Schrodinger equation based on the Mathieu functions is obtained in elliptic coordinates. The dependencies of the electron size quantization spectrum on the size and shape of the core-shell nanowire and nanotube are calculated. It is shown that the ellipticity of a quantum wire leads to break of degeneration of quasiparticle energy spectrum. The dependences of the energy of odd and even electron states on the ratio between semiaxes are of a nonmonotonous character. The anticrosing effects are observed at the dependencies of electron energy spectrum on the transversal size of the core-shell nanowire.
Vertical elliptic operator for efficient wave propagation in TTI media
Waheed, Umair bin
2015-08-19
Elliptic wave extrapolation operators require significantly less computational cost than the ones for transversely isotropic (TI) media. However, it does not provide accurate wavefield representation or imaging for the prevalent TI media. We propose a new vertical elliptically anisotropic (VEA) wave equation by decomposing the acoustic TI pseudo-differential wave equation. The decomposition results in a vertical elliptic differential equation and a scalar operator. The new VEA-like wave equation shares the same dispersion relation as that of the original acoustic TI wave equation. Therefore, the kinematic contents are correctly matched to the original equation. Moreover, the proposed decomposition yields better amplitude properties than the isotropic decomposition without increasing the computational load. Therefore, it exhibits better cost versus accuracy tradeoff compared to the isotropic or the tilted elliptic decompositions. We demonstrate with numerical examples that the proposed methodology is numerically stable for complex models and is free from shear-wave artifacts.
Orbits in general relativity: the Jacobian elliptic function
Energy Technology Data Exchange (ETDEWEB)
Miro Rodriguez, C
1987-03-11
The Jacobian elliptic functions are applied to the motion of nonzero-rest-mass particles in the Schwarzschild geometry. The bound and unbound trajectories are analysed together with their classical and special-relativity limits.
Jacobian elliptic function expansion solutions of nonlinear stochastic equations
International Nuclear Information System (INIS)
Wei Caimin; Xia Zunquan; Tian Naishuo
2005-01-01
Jacobian elliptic function expansion method is extended and applied to construct the exact solutions of the nonlinear Wick-type stochastic partial differential equations (SPDEs) and some new exact solutions are obtained via this method and Hermite transformation
Mergers of elliptical galaxies and the fundamental plane
Gonzalez-Garcia, AC; van Albada, TS; AvilaReese,; Firmani, C; Frenk, CS; Allen, YC
2003-01-01
N-body simulations have been carried out in order to explore the final state of elliptical galaxies after encounters and more expecifically whether the Fundamental Plane (FP hereafter) relation is affected by merging.
Optimal Rendezvous and Docking Simulator for Elliptical Orbits, Phase I
National Aeronautics and Space Administration — It is proposed to develop and implement a simulation of spacecraft rendezvous and docking guidance, navigation, and control in elliptical orbit. The foundation of...
Reduction of Elliptic Curves in Equal Characteristic 3 (and 2)
Miyamoto, Roland; Top, Jakob
2005-01-01
We determine conductor exponent, minimal discriminant and fibre type for elliptic curves over discrete valued fields of equal characteristic 3. Along the same lines, partial results are obtained in equal characteristic 2.
An imbedding theorem and its applications in degenerate elliptic equations
International Nuclear Information System (INIS)
Duong Minh Duc.
1988-06-01
We improve the Rellich-Kondrachov theorem and apply it to study strongly degenerate and singular elliptic equations. We obtain the maximum principle, Harnacks's inequality and global regularity for solutions of those equations. (author). 11 refs
Handbook of elliptic and hyperelliptic curve cryptography
Cohen, Henri; Avanzi, Roberto; Doche, Christophe; Lange, Tanja; Nguyen, Kim; Vercauteren, Frederik
2005-01-01
… very comprehensive coverage of this vast subject area … a useful and essential treatise for anyone involved in elliptic curve algorithms … this book offers the opportunity to grasp the ECC technology with a diversified and comprehensive perspective. … This book will remain on my shelf for a long time and will land on my desk on many occasions, if only because the coverage of the issues common to factoring and discrete log cryptosystems is excellent.-IACR Book Reviews, June 2011… the book is designed for people who are working in the area and want to learn more about a specific issue. The chapters are written to be relatively independent so that readers can focus on the part of interest for them. Such readers will be grateful for the excellent index and extensive bibliography. … the handbook covers a wide range of topics and will be a valuable reference for researchers in curve-based cryptography. -Steven D. Galbraith, Mathematical Reviews, Issue 2007f.
ON ELLIPTICALLY POLARIZED ANTENNAS IN THE PRESENCE OF GROUND
The effect of ground reflections upon the far field of an elliptically polarized antenna of ar itrary orientation with r spect to ground is...investigated. The equation of the polarization ellipse produced by an elliptically polarized antenna in the presence of ground is derived, AND SEVERAL...EXAMPLES ILLUSTRATE THE VARIATION IN THE AXIS RATIO OF THE POLARIZATION ELLIPSE AS A FUNCTION OF THE GEOMETRY OF THE MEASURING SETUP. A method is presented
Kerr ellipticity effect in a birefringent optical fiber
International Nuclear Information System (INIS)
Ishiekwene, G.C.; Mensah, S.Y.; Brown, C.S.
2004-09-01
An intensity-dependent change in the ellipticity of an input light beam leads to a characteristic shift in polarization instability. Dichroism gives rise to a self-induced ellipticity effect in the polarization state of an intense input light oriented along the fast axis of a birefringent optical fiber. The critical power at which the fiber effective beat length becomes infinite is reduced considerably in the presence of dichroism. (author)
Elliptic flow based on a relativistic hydrodynamic model
Energy Technology Data Exchange (ETDEWEB)
Hirano, Tetsufumi [Department of Physics, Waseda Univ., Tokyo (Japan)
1999-08-01
Based on the (3+1)-dimensional hydrodynamic model, the space-time evolution of hot and dense nuclear matter produced in non-central relativistic heavy-ion collisions is discussed. The elliptic flow parameter v{sub 2} is obtained by Fourier analysis of the azimuthal distribution of pions and protons which are emitted from the freeze-out hypersurface. As a function of rapidity, the pion and proton elliptic flow parameters both have a peak at midrapidity. (author)
Central $L$-values of elliptic curves and local polynomials
Ehlen, Stephan; Guerzhoy, Pavel; Kane, Ben; Rolen, Larry
2018-01-01
Here we study the recently introduced notion of a locally harmonic Maass form and its applications to the theory of $L$-functions. In particular, we find finite formulas for certain twisted central $L$-values of a family of elliptic curves in terms of finite sums over canonical binary quadratic forms. This yields vastly simpler formulas related to work of Birch and Swinnerton-Dyer for such $L$-values, and extends beyond their framework to special non-CM elliptic curves.
Radial, sideward and elliptic flow at AGS energies
Indian Academy of Sciences (India)
the sideward flow, the elliptic flow and the radial transverse mass distribution of protons data at. AGS energies. In order to ... data on both sideward and elliptic flow, NL3 model is better at 2 A¡GeV, while NL23 model is at 4–8. A¡GeV. ... port approach RBUU which is based on a coupled set of covariant transport equations for.
Design, fabrication and testing of elliptical crystal bender for the ...
Indian Academy of Sciences (India)
R. Narasimhan (Krishtel eMaging) 1461 1996 Oct 15 13:05:22
An extended X-ray absorption fine structure (EXAFS) beam-line for. X-ray absorption .... mal expansion coefficient and high thermal conductivity. The source to Si .... The co-ordinate system convention assumed here is shown in figure 5. As the ...
Thickness shear mode quartz crystal resonators with optimized elliptical electrodes
International Nuclear Information System (INIS)
Ma Ting-Feng; Feng Guan-Ping; Zhang Chao; Jiang Xiao-Ning
2011-01-01
Quartz crystal resonators (QCRs) with circular electrodes have been widely used for various liquid and gas sensing applications. In this work, quartz crystal resonators with elliptical electrodes were studied and tested for liquid property measurement. Mindlin's theory was used to optimize the dimension and geometry of the electrodes and a 5-MHz QCR with minimum series resistance and without any spurious modes was obtained. A series of AT-cut QCRs with elliptical electrodes of different sizes were fabricated and their sensing performances were compared to devices with circular electrodes. The experimental result shows that the device with elliptical electrodes can obtain lower resonance impedance and a higher Q factor, which results in a better loading capability. Even though the sensitivities of devices with elliptical and circular electrodes are found to be similar, the sensor with elliptical electrodes has much higher resolution due to a better frequency stability. The study indicates that the performance of QCRs with elliptical electrodes is superior to that of traditional QCRs with circular electrodes. (condensed matter: electronic structure, electrical, magnetic, and optical properties)
ELLIPT2D: A Flexible Finite Element Code Written Python
International Nuclear Information System (INIS)
Pletzer, A.; Mollis, J.C.
2001-01-01
The use of the Python scripting language for scientific applications and in particular to solve partial differential equations is explored. It is shown that Python's rich data structure and object-oriented features can be exploited to write programs that are not only significantly more concise than their counter parts written in Fortran, C or C++, but are also numerically efficient. To illustrate this, a two-dimensional finite element code (ELLIPT2D) has been written. ELLIPT2D provides a flexible and easy-to-use framework for solving a large class of second-order elliptic problems. The program allows for structured or unstructured meshes. All functions defining the elliptic operator are user supplied and so are the boundary conditions, which can be of Dirichlet, Neumann or Robbins type. ELLIPT2D makes extensive use of dictionaries (hash tables) as a way to represent sparse matrices.Other key features of the Python language that have been widely used include: operator over loading, error handling, array slicing, and the Tkinter module for building graphical use interfaces. As an example of the utility of ELLIPT2D, a nonlinear solution of the Grad-Shafranov equation is computed using a Newton iterative scheme. A second application focuses on a solution of the toroidal Laplace equation coupled to a magnetohydrodynamic stability code, a problem arising in the context of magnetic fusion research
A second beam-diagnostic beamline for the advanced light source
International Nuclear Information System (INIS)
Sannibale, Fernando; Baum, Dennis; Kelez, Nicholas; Scarvie, Tom; Holldack, Karsten
2003-01-01
A second beamline, BL 7.2, completely dedicated to beam diagnostics is being installed at the Advanced Light Source (ALS). The design has been optimized for the measurement of the momentum spread and emittance of the stored beam in combination with the existing diagnostic beamline, BL 3.1. A detailed analysis of the experimental error has allowed the definition of the system parameters. The obtained requirements found a good matching with a simple and reliable system based on the detection of X-ray synchrotron radiation (SR) through a pinhole system. The actual beamline, which also includes a port for visible and infrared SR as well as an X-ray beam position monitor (BPM), is mainly based on the design of two similar diagnostic beamlines at BESSY II. This approach allowed a significant saving in time, cost and engineering effort. The design criteria, including a summary of the experimental error analysis, as well as a brief description of the beamline are presented
CAT Guide and Beamline Directory. A key to APS Collaborative Access Teams
International Nuclear Information System (INIS)
1999-01-01
The Advanced Photon Source (APS), a national user facility for synchrotrons radiation research, is located at Argonne National Laboratory, approximately 25 miles southwest of Chicago, Illinois. The APS is considered a third-generation synchrotrons radiation facility (specifically designed to accommodate insertion devices to serve as radiation sources) and is one of three such facilities in the world. Currently, it is the most brilliant source in the United States for research in such diverse fields as biology, medicine, materials science, chemistry, geology, agriculture and soil science, physics, and manufacturing technology. Researchers use the APS either as members of Collaborative Access Teams (CATS) or as Independent Investigators (IIs). CATS are responsible for designing, building, and operating beamlines in one or more sectors, each sector consisting of an insertion-device (ID) beamline and a bending-magnet (BM) beamline. Each beamline is designed to accommodate a specific type of research program(s) and is optimized accordingly. CAT members are entitled to use 75% of the available beam time to pursue CAT research goals. The remaining 25% of the available beam time must be made available to IIs. This document was written to help prospective IIs determine which beamlines are suitable for their specific experiments
Solutions to mitigate heat loads due to electrons on sensitive components of ITER HNB beamlines
Energy Technology Data Exchange (ETDEWEB)
Sartori, Emanuele, E-mail: emanuele.sartori@gmail.com [Consorzio RFX (CNR, ENEA, INFN, Università di Padova, Acciaierie Venete SpA), C.so Stati Uniti 4, 35127 Padova (Italy); Veltri, Pierluigi; Dalla Palma, Mauro; Agostinetti, Piero [Consorzio RFX (CNR, ENEA, INFN, Università di Padova, Acciaierie Venete SpA), C.so Stati Uniti 4, 35127 Padova (Italy); Hemsworth, Ronald; Singh, Mahendrajit [ITER Organization, Route de Vinon sur Verdon, 13115 Saint Paul Lez Durance (France); Serianni, Gianluigi [Consorzio RFX (CNR, ENEA, INFN, Università di Padova, Acciaierie Venete SpA), C.so Stati Uniti 4, 35127 Padova (Italy)
2016-11-01
Highlights: • Energetic electrons leaking out of the ITER HNB accelerator are simulated. • Electrons generated along the ITER HNB beamline are simulated. • Heat loads and heat load maps on cryopumps are calculated for ITER HNB and test facility. • Protection solutions that will be installed are presented and their effect discussed. - Abstract: The operation of neutral beam injectors for plasma heating and current drive in a fusion device provides challenges in the thermal management of beamline components. Sensitive components such as the cryogenic pumps at beamline periphery shall be protected from the heat flux due to stray electrons. These are emitted by the negative ion accelerator or generated along the beamline by interaction of fast electrons, ions or atoms with background gas and surfaces. In this article the case of the ITER Heating Neutral Beam (HNB) and its test facility MITICA is discussed, for which the beam parameters and the required pulse length of one hour is a major leap forward with respect to the present experience with neutral beam systems. The engineering solutions adopted for effective cryopump protection against the heat load from electrons are described. The use of three-dimensional numerical simulations of particle trajectories in the complex geometry of the beamline was needed for the quantitative estimations of the heat loads. The presented solutions were optimized to minimize the impact on gas pumping and on the functionality of other components.
New beamline dedicated to solution scattering from biological macromolecules at the ESRF
International Nuclear Information System (INIS)
Pernot, P; Theveneau, P; Giraud, T; Fernandes, R Nogueira; Nurizzo, D; Spruce, D; Surr, J; McSweeney, S; Round, A; Felisaz, F; Foedinger, L; Gobbo, A; Huet, J; Villard, C; Cipriani, F
2010-01-01
The new bio-SAXS beamline (ID14-3 at the ESRF, Grenoble, France) is dedicated exclusively to small-angle scattering experiments of biological macromolecules in solution and has been in user operation since November 2008. Originally a protein crystallography beamline, ID14-3 was refurbished, still as a part of the ESRF Structural Biology group, with the main aim to provide a facility with 'quick and easy' access to satisfy rapidly growing demands from crystallographers, biochemists and structural biologists. The beamline allows manual and automatic sample loading/unloading, data collection, processing (conversion of a 2D image to a normalized 1D X-ray scattering profile) and analysis. The users obtain on-line standard data concerning the size (radius of gyration, maximum dimension and volume) and molecular weight of samples which allow on-the fly ab-inito shape reconstruction in order to provide feedback enabling the data collection strategies to be optimized. Automation of sample loading is incorporated on the beamline using a device constructed in collaboration between the EMBL (Grenoble and Hamburg outstations) and the ESRF. Semi/automated data analysis is implemented following the model of the SAXS facility at X33, EMBL Hamburg. This paper describes the bio-SAXS beamline and set-up characteristics together with the examples of user data obtained.
Design of a High-Throughput Biological Crystallography Beamline for Superconducting Wiggler
International Nuclear Information System (INIS)
Tseng, P.C.; Chang, C.H.; Fung, H.S.; Ma, C.I.; Huang, L.J.; Jean, Y.C.; Song, Y.F.; Huang, Y.S.; Tsang, K.L.; Chen, C.T.
2004-01-01
We are constructing a high-throughput biological crystallography beamline BL13B, which utilizes the radiation generated from a 3.2 Tesla, 32-pole superconducting multipole wiggler, for multi-wavelength anomalous diffraction (MAD), single-wavelength anomalous diffraction (SAD), and other related experiments. This beamline is a standard double crystal monochromator (DCM) x-ray beamline equipped with a collimating mirror (CM) and a focusing mirror (FM). Both the CM and FM are one meter long and made of Si substrate, and the CM is side-cooled by water. Based on detailed thermal analysis, liquid nitrogen (LN2) cooling for both crystals of the DCM has been adopted to optimize the energy resolution and photon beam throughput. This beamline will deliver, through a 100 μm diameter pinhole, photon flux of greater than 1011 photons/sec in the energy range from 6.5 keV to 19 keV, which is comparable to existing protein crystallography beamlines from bending magnet source at high energy storage rings
Energy Technology Data Exchange (ETDEWEB)
Hulbert, S.L.; Sharma, S.
1987-01-01
At present, grazing incidence mirrors are used almost exclusively as the first optical element in VUV and soft x-ray synchrotron radiation beamlines. The performance of these mirrors is determined by thermal and mechanical stress-induced figure errors as well as by figure errors remaining from the grinding and polishing process. With the advent of VUV and soft x-ray undulators and wigglers has come a new set of thermal stress problems related to both the magnitude and the spatial distribution of power from these devices. In many cases the power load on the entrance slits and gratings in these beamlines is no longer negligible. The dependence of thermally-induced front-end mirror figure errors on various storage ring and insertion device parameters (especially those at the National Synchrotron Light Source) and the effects of these figure errors on two classes of soft x-ray beamlines are presented.
Neutron imaging options at the BOA beamline at Paul Scherrer Institut
International Nuclear Information System (INIS)
Morgano, M.; Peetermans, S.; Lehmann, E.H.; Panzner, T.; Filges, U.
2014-01-01
The BOA beamline at the Swiss spallation neutron source SINQ at Paul Scherrer Institut is a flexible instrument used mainly for testing novel techniques and devices for neutron scattering and optics, but, due to the large and relatively homogeneous field of view, it can be successfully used for experiments in the field of neutron imaging. The beamline allows also for the exploitation of advanced imaging concepts such as polarized neutron imaging and diffractive neutron imaging. In this paper we present the characterization of the BOA beamline in the light of its neutron imaging capabilities. We show also the different techniques that can be employed there as user-friendly plugins for non-standard neutron imaging experiments
Soft x-ray spectroscopy undulator beamline at the Advanced Photon Source
Energy Technology Data Exchange (ETDEWEB)
Randall, K.J.; Xu, Z.; Moore, J.F.; Gluskin, E.
1997-09-01
Construction of the high-resolution soft x ray spectroscopy undulator beamline, 2ID-C, at the Advanced Photon Source (APS) has been completed. The beamline, one of two soft x ray beamlines at the APS, will cover the photon energy range from 500 to 3,000 eV, with a maximum resolving power between 7,000 and 14,000. The optical design is based on a spherical grating monochromator (SGM) giving both high resolution and high flux throughput. Photon flux is calculated to be approximately 10{sup 12}--10{sup 13} photons per second with a beam size of approximately 1 x 1 mm{sup 2} at the sample.
Beamline for low-energy transport of highly charged ions at HITRAP
International Nuclear Information System (INIS)
Andelkovic, Z.; Herfurth, F.; Kotovskiy, N.; König, K.; Maaß, B.; Murböck, T.; Neidherr, D.; Schmidt, S.; Steinmann, J.; Vogel, M.; Vorobjev, G.
2015-01-01
A beamline for transport of highly charged ions with energies as low as a few keV/charge has been constructed and commissioned at GSI. Complementary to the existing infrastructure of the HITRAP facility for deceleration of highly charged ions from the GSI accelerator, the new beamline connects the HITRAP ion decelerator and an EBIT with the associated experimental setups. Therefore, the facility can now transport the decelerated heavy highly charged ions to the experiments or supply them offline with medium-heavy highly charged ions from the EBIT, both at energies as low as a few keV/charge. Here we present the design of the 20 m long beamline with the corresponding beam instrumentation, as well as its performance in terms of energy and transport efficiency
The status of the first infrared beamline at Shanghai Synchrotron Radiation Facility
International Nuclear Information System (INIS)
Ji, Te; Tong, Yajun; Zhu, Huachun; Zhang, Zengyan; Peng, Weiwei; Chen, Min; Xiao, Tiqiao; Xu, Hongjie
2015-01-01
Construction of the first infrared beamline BL01B1 at Shanghai Synchrotron Radiation Facility (SSRF) was completed at the end of 2013. The IR beamline collects both edge radiation (ER) and bending magnet radiation (BMR) from a port, providing a solid angle of 40 mrad and 20 mrad in the horizontal and vertical directions, respectively. The optical layout of the infrared beamline and the design of the extraction mirror are described in this paper. A calculation of the beam propagation has been used to optimize the parameters of the optical components. The photon flux and spatial resolution have been measured at the end-station, and the experimental results are in good agreement with the theoretical calculation
The status of the first infrared beamline at Shanghai Synchrotron Radiation Facility
Energy Technology Data Exchange (ETDEWEB)
Ji, Te; Tong, Yajun; Zhu, Huachun; Zhang, Zengyan; Peng, Weiwei; Chen, Min, E-mail: chenmin@sinap.ac.cn; Xiao, Tiqiao; Xu, Hongjie
2015-07-11
Construction of the first infrared beamline BL01B1 at Shanghai Synchrotron Radiation Facility (SSRF) was completed at the end of 2013. The IR beamline collects both edge radiation (ER) and bending magnet radiation (BMR) from a port, providing a solid angle of 40 mrad and 20 mrad in the horizontal and vertical directions, respectively. The optical layout of the infrared beamline and the design of the extraction mirror are described in this paper. A calculation of the beam propagation has been used to optimize the parameters of the optical components. The photon flux and spatial resolution have been measured at the end-station, and the experimental results are in good agreement with the theoretical calculation.
The status of the first infrared beamline at Shanghai Synchrotron Radiation Facility
Ji, Te; Tong, Yajun; Zhu, Huachun; Zhang, Zengyan; Peng, Weiwei; Chen, Min; Xiao, Tiqiao; Xu, Hongjie
2015-07-01
Construction of the first infrared beamline BL01B1 at Shanghai Synchrotron Radiation Facility (SSRF) was completed at the end of 2013. The IR beamline collects both edge radiation (ER) and bending magnet radiation (BMR) from a port, providing a solid angle of 40 mrad and 20 mrad in the horizontal and vertical directions, respectively. The optical layout of the infrared beamline and the design of the extraction mirror are described in this paper. A calculation of the beam propagation has been used to optimize the parameters of the optical components. The photon flux and spatial resolution have been measured at the end-station, and the experimental results are in good agreement with the theoretical calculation.
Using the particle beam optics lab. (PBO LABtm) for beamline design and analysis
International Nuclear Information System (INIS)
Gillespie, G.H.; Hill, B.W.; Martono, H.; Moore, J.M.; Lampel, M.C.; Brown, N.A.
1999-01-01
The Particle Beam Optics Interactive Computer Laboratory (PBO Lab) represents a new approach to providing software for particle beam optics modeling. The PBO Lab includes four key elements: a graphic user interface shell; a graphic beamline construction kit for users to interactively and visually construct optical beam lines; a knowledge database on the physics and technology of optical elements, and various charged particle optics computational engines. A first-order matrix code, including a space charge model, can be used to produce scaled images of beamlines together with overlays of single trajectories and beam envelopes. The qualitative results of graphically sliding beamline components, or adjusting bend angles, can be explored interactively. Quantitative computational engines currently include the third-order TRANSPORT code and the multi-particle ray tracing program TURTLE. The use of the PBO Lab for designing and analyzing a second order achromatic bend is illustrated with the Windows 95/NT version of the software. (authors)
Soft x-ray beamline BL7A at the UVSOR
International Nuclear Information System (INIS)
Murata, T.; Matsukawa, T.; Naoe, S.; Horigome, T.; Matsudo, O.; Watanabe, M.
1992-01-01
A vacuum-compatible double-crystal monochromator with constant exit-beam height has been installed and operated for several years at beamline BL7A at the UVSOR facility at the Institute for Molecular Science, Okazaki, Japan, The beamline is used both for the photons from a normal bending-magnet section and those from a 4 T superconducting three-pole horizontal wiggler. Various pairs of monochromator crystals are being used with sufficient intensity and signal to noise ratio of the output signal. Basic structure of the beamline, the mechanism of the monochromator, and some typical spectra of materials with absorption edges between 850 eV and 4 keV are reported
International Nuclear Information System (INIS)
Martin, M.C.; McKinney, W.R.
1998-05-01
A set of new infrared (IR) beamlines on the 1.4 bending magnet port at the Advanced Light Source, LBNL, are described. Using a synchrotron as an IR source provides considerable brightness advantages, which manifests itself most beneficially when performing spectroscopy on a microscopic length scale. Beamline (BL) 1.4.3 is a dedicated microspectroscopy beamline, where the much smaller focused spot size using the synchrotron source is utilized. This enables an entirely new set of experiments to be performed where spectroscopy on a truly microscopic scale is now possible. BL 1.4.2 consists of a vacuum FTIR bench with a wide spectral range and step-scan capabilities. The fast timing is demonstrated by observing the synchrotron electron storage pattern at the ALS
Wojdyla, Justyna Aleksandra; Kaminski, Jakub W; Panepucci, Ezequiel; Ebner, Simon; Wang, Xiaoqiang; Gabadinho, Jose; Wang, Meitian
2018-01-01
Data acquisition software is an essential component of modern macromolecular crystallography (MX) beamlines, enabling efficient use of beam time at synchrotron facilities. Developed at the Paul Scherrer Institute, the DA+ data acquisition software is implemented at all three Swiss Light Source (SLS) MX beamlines. DA+ consists of distributed services and components written in Python and Java, which communicate via messaging and streaming technologies. The major components of DA+ are the user interface, acquisition engine, online processing and database. Immediate data quality feedback is achieved with distributed automatic data analysis routines. The software architecture enables exploration of the full potential of the latest instrumentation at the SLS MX beamlines, such as the SmarGon goniometer and the EIGER X 16M detector, and development of new data collection methods.
BEAMLINE-CONTROLLED STEERING OF SOURCE-POINT ANGLE AT THE ADVANCED PHOTON SOURCE
Energy Technology Data Exchange (ETDEWEB)
Emery, L.; Fystro, G.; Shang, H.; Smith, M.
2017-06-25
An EPICS-based steering software system has been implemented for beamline personnel to directly steer the angle of the synchrotron radiation sources at the Advanced Photon Source. A script running on a workstation monitors "start steering" beamline EPICS records, and effects a steering given by the value of the "angle request" EPICS record. The new system makes the steering process much faster than before, although the older steering protocols can still be used. The robustness features of the original steering remain. Feedback messages are provided to the beamlines and the accelerator operators. Underpinning this new steering protocol is the recent refinement of the global orbit feedback process whereby feedforward of dipole corrector set points and orbit set points are used to create a local steering bump in a rapid and seamless way.
High-brightness beamline for x-ray spectroscopy at the ALS
Energy Technology Data Exchange (ETDEWEB)
Perera, R.C.C.; Jones, G. [Ernest Orlando Lawrence Berkeley National Lab., CA (United States); Lindle, D.W. [Univ. of Nevada, Las Vegas, NV (United States)
1997-04-01
Beamline 9.3.1 at the Advanced Light Source (ALS) is a windowless beamline, covering the 1-6 keV photon-energy range, designed to achieve the goals of high energy resolution, high flux, and high brightness at the sample. When completed later this year, it will be the first ALS monochromatic hard x-ray beamline, and its brightness will be an order of magnitude higher than presently available in this energy range. In addition, it will provide flux and resolution comparable to any other beamline now in operation. To achieve these goals, two technical improvements, relative to existing x-ray beamlines, were incorporated. First, a somewhat novel optical design for x-rays, in which matched toroidal mirrors are positioned before and after the double-crystal monochromator, was adopted. This configuration allows for high resolution by passing a collimated beam through the monochromator, and for high brightness by focusing the ALS source on the sample with unit magnification. Second, a new {open_quotes}Cowan type{close_quotes} double-crystal monochromator based on the design used at NSLS beamline X-24A was developed. The measured mechanical precision of this new monochromator shows significant improvement over existing designs, without using positional feedback available with piezoelectric devices. Such precision is essential because of the high brightness of the radiation and the long distance (12 m) from the source (sample) to the collimating (focusing) mirror. This combination of features will provide a bright, high resolution, and stable x-ray beam for use in the x-ray spectroscopy program at the ALS.
International Nuclear Information System (INIS)
Job, P.K.; Haeffner, D.R.; Shu, D.
1994-12-01
Bremsstrahlung is produced in the APS storage ring by the interaction of positrons with the residual gas molecules in the vacuum chamber of the storage ring. The bremsstrahlung production causes a serious challenge in shielding the insertion-device beamlines because the entire straight section (15 meters) is in the line of sight of the beamline. The radiation emerges in a narrow cone tangential to the beam path with the characteristic emission angle 1/γ, where γ is E/mc 2 which is the ratio of the kinetic energy to the rest mass for the positrons. This high-energy gamma radiation has an approximate 1/E spectrum with the maximum energy extending up to the particle energy (7 GeV for the APS). Bremsstrahlung, being high-energy photons, produces an electromagnetic shower when it encounters the beamline elements. A beamline element not thick enough to fully contain an electromagnetic shower can cause considerable scatter of the high-energy bremsstrahlung radiation. The low-energy component of the bremsstrahlung can also be scattered and create high dose rates in the first-optical and white-beam enclosures. The fully developed electromagnetic shower will have a photon spectrum almost independent of the material. The electromagnetic showers in the high-Z materials can also produce photoneutrons. This note reports the summary of EGS4 calculations performed on bremsstrahlung scattering from different beamline components in a typical APS insertion-device beamline. The related recommendations for shielding are also given. The shielding criterion adopted is a total dose rate of 2.5μSv/h (0.25 mrem/h) at 30 cm from the shield
Design of a dedicated beamline for THz coherent synchrotron radiation at UVSOR-III
International Nuclear Information System (INIS)
Kimura, Shin-ichi; Nakamura, Eiken; Imura, Keiichiro; Katoh, Masahiro; Hosaka, Masahito; Takahashi, Toshiharu
2012-01-01
We report the design of a THz beamline for coherent synchrotron radiation (CSR) at the UVSOR-III very-low-emittance synchrotron radiation light source. The emitted THz-CSR is collected by a three-dimensional 'magic mirror', which is a perfect collecting mirror for bending-magnet radiation with an acceptance angle of 288 mrad (H) × 80 mrad (V). A quasi-monochromatic THz-CSR with an average flux of 104 μW/0.1 % b.w. and a peak power of 120 nJ/pulse/0.1 % b.w. is expected at the beamline.
Motion control system of MAX IV Laboratory soft x-ray beamlines
International Nuclear Information System (INIS)
Sjöblom, Peter; Lindberg, Mirjam; Forsberg, Johan; Persson, Andreas G.; Urpelainen, Samuli; Såthe, Conny
2016-01-01
At the MAX IV Laboratory, five new soft x-ray beamlines are under development. The first is Species and it will be used to develop and set the standard of the control system, which will be common across the facility. All motion axes at MAX IV will be motorized using stepper motors steered by the IcePAP motion controller and a mixture of absolute and incremental encoders following a predefined coordinate system. The control system software is built in Tango and uses the Python-based Sardana framework. The user controls the entire beamline through a synoptic overview and Sardana is used to run the scans.
The X-ray microscopy beamline UE46-PGM2 at BESSY
International Nuclear Information System (INIS)
Follath, R.; Schmidt, J. S.; Weigand, M.; Fauth, K.
2010-01-01
The Max Planck Institute for Metal Physics in Stuttgart and the Helmholtz Center Berlin operate a soft X-ray microscopy beamline at the storage ring BESSY II. A collimated PGM serves as monochromator for a scanning X-ray microscope and a full field X-ray microscope at the helical undulator UE46. The selection between both instruments is accomplished via two switchable focusing mirrors. The scanning microscope (SM) is based on the ALS STXM microscope and fabricated by the ACCEL company. The full field microscope (FFM) is currently in operation at the U41-SGM beamline and will be relocated to its final location this year.
Gas phase THz spectroscopy of toxic agent simulant compounds using the AILES synchrotron beamline
Cuisset, A.; Smirnova, I.; Bocquet, R.; Hindle, F.; Mouret, G.; Yang, C.; Pirali, O.; Roy, P.
2010-02-01
A new study is currently underway aiming at recording and assigning the gas phase rovibrational spectra of several organophosphorus and organosulphur compounds in the THz frequency domain. Thanks to the exceptional properties of flux, brilliance and spectral range of the AILES beamline coupled to the FTIR spectrometer, the gas phase vibrational spectra of low volatility organophosphorous compounds have been recorded across the entire THz frequency range. High resolution FTIR spectroscopy was used to record the pure rotational and the low-frequency rovibrational spectrum of DMSO. A comparison between the spectra measured with the AILES beamline and the spectra obtained with optoelectronic THz sources is possible.
Design and development of PEEM/ARPES beamline for Indus-2 synchrotron radiation source
International Nuclear Information System (INIS)
Goutam, U.K.; Sharma, R.K.; Jagannath; Gadkari, S.C.; Yakhmi, J.V.; Sahni, V.C.
2008-06-01
A high resolution beamline having two branches dedicated to Photo Emission Electron Microscopy (PEEM) and Angle Resolved Photo Electron Spectroscopy (ARPES) is planned for Indus-2 synchrotron radiation source. These two techniques open a wide field of new applications in materials research and have proven to be powerful tools to investigate topological, elemental, chemical state, electronic and magnetic properties of surfaces, thin films, and multilayers at high resolutions.The beamline will cover a large energy range from 10 to 4000 eV and is expected to deliver a flux of the order of ∼10 13 ph/s/0.1%B.W. with an energy resolution of ∼10 -4 . This report describes the optical design, beamline layout, effects of heat load on various components and the expected performance of the beamline. This beamline would have a collimating mirror for vertical collimation of the beam, plane grating/double crystal monochromator to make the white synchrotron beam monochromatic in entire energy range, toroidal mirror for splitting the beam as well as for intermediate focusing and a Kirkpatrick-Baez (K-B) mirror system for focusing the beam both in vertical and horizontal directions at the final sample location. Total beamline will be 36m long. Optical design has been carried out involving various computer codes such as XOP2.1, SHADOWVUI, SPECTRA 8.0 etc. Head load calculations have been performed using ANSYS, a finite element analysis code. Using this code, temperature distribution, thermal deformation and slope error values for collimating mirror, grating monochromator and double crystal monochromator using several possible cooling arrangements have been calculated and depending on these parameters, best options for different components have been selected for the beamline. Experimental stations of this beamline consist of ultra-high vacuum compatible chambers in which various probes, analyzers, detectors and other facilities are housed. A toroidal electron energy analyzer will
Layout and first results of the nanotomography endstation at the P05 beamline at PETRA III
Energy Technology Data Exchange (ETDEWEB)
Ogurreck, M.; Greving, I.; Beckmann, F.; Wilde, F.; Müller, M. [Institute of Materials Research, Helmholtz–Zentrum Geesthacht (Germany); Marschall, F.; Vogt, H.; Last, A. [Institute of Microstructure Technology, Karlsruhe Institute of Technology (Germany); Rosario, J. J. do [Institute of Advanced Ceramics, Technical University Hamburg–Harburg (Germany); Leib, E. W. [Institute of Physical Chemistry, University of Hamburg (Germany)
2016-01-28
The Helmholtz-Zentrum Geesthacht operates the P05 Imaging Beamline at the DESY storage ring PETRA III. This beamline is dedicated to micro- and nanotomography with two endstations. This paper will present the nanotomography endstation layout and first results obtained from commissioning and test experiments. First tests have been performed with CRLs as X-ray objectives and newly developed rolled X-ray prism lenses as condenser optics. This setup allows a resolution of 100 nm half period with an effective detector pixel size of 15nm. A first tomograph of a photonic glass sample was measured in early 2014.
A pixel detector for the protein crystallography beamline at the SLS
Brönnimann, C; Eikenberry, E F; Fischer, P; Florin, S; Horisberger, R P; Lindner, Manfred; Schmitt, B; Schulze, C
2002-01-01
At the Paul Scherrer Institute a new synchrotron light source is currently under construction, the Swiss Light Source (SLS), which will be operational in summer 2001. Among the first beamlines is a high brightness, micro-focusing protein crystallography beamline. It will be equipped with a pixel detector, which has several features of interest for the next generation of protein crystallography detectors. The point spread function and the effect of charge sharing was measured with a prototype detector in a test experiment at the European Synchrotron Radiation Facility in Grenoble. The concepts of the SLS pixel detector is presented as well as test results from radiation hard prototype chips.
Motion control system of MAX IV Laboratory soft x-ray beamlines
Energy Technology Data Exchange (ETDEWEB)
Sjöblom, Peter, E-mail: peter.sjoblom@maxlab.lu.se; Lindberg, Mirjam, E-mail: mirjam.lindberg@maxlab.lu.se; Forsberg, Johan, E-mail: johan.forsberg@maxlab.lu.se; Persson, Andreas G., E-mail: andreas-g.persson@maxlab.lu.se; Urpelainen, Samuli, E-mail: samuli.urpelainen@maxlab.lu.se; Såthe, Conny, E-mail: conny.sathe@maxlab.lu.se [MAX IV Laboratory, Photongatan 2, 225 92 Lund (Sweden)
2016-07-27
At the MAX IV Laboratory, five new soft x-ray beamlines are under development. The first is Species and it will be used to develop and set the standard of the control system, which will be common across the facility. All motion axes at MAX IV will be motorized using stepper motors steered by the IcePAP motion controller and a mixture of absolute and incremental encoders following a predefined coordinate system. The control system software is built in Tango and uses the Python-based Sardana framework. The user controls the entire beamline through a synoptic overview and Sardana is used to run the scans.
International Nuclear Information System (INIS)
Singh, M.R.; Mukund, R.; Sahni, V.C.
1999-01-01
The influence of geometrical shape errors and surface errors on the characteristics and performance of grazing incidence optics used in the design of beamlines at synchrotron radiation facilities is considered. The methodology adopted for the simulation of slope errors is described and results presented for the ellipsoidal focussing mirror used in the design of PGM beamline at Indus-1. (author)
Advanced topics in the arithmetic of elliptic curves
Silverman, Joseph H
1994-01-01
In the introduction to the first volume of The Arithmetic of Elliptic Curves (Springer-Verlag, 1986), I observed that "the theory of elliptic curves is rich, varied, and amazingly vast," and as a consequence, "many important topics had to be omitted." I included a brief introduction to ten additional topics as an appendix to the first volume, with the tacit understanding that eventually there might be a second volume containing the details. You are now holding that second volume. it turned out that even those ten topics would not fit Unfortunately, into a single book, so I was forced to make some choices. The following material is covered in this book: I. Elliptic and modular functions for the full modular group. II. Elliptic curves with complex multiplication. III. Elliptic surfaces and specialization theorems. IV. Neron models, Kodaira-Neron classification of special fibers, Tate's algorithm, and Ogg's conductor-discriminant formula. V. Tate's theory of q-curves over p-adic fields. VI. Neron's theory of can...
The two-loop sunrise integral and elliptic polylogarithms
Energy Technology Data Exchange (ETDEWEB)
Adams, Luise; Weinzierl, Stefan [Institut fuer Physik, Johannes Gutenberg-Universitaet Mainz (Germany); Bogner, Christian [Institut fuer Physik, Humboldt-Universitaet zu Berlin (Germany)
2016-07-01
In this talk, we present a solution for the two-loop sunrise integral with arbitrary masses around two and four space-time dimensions in terms of a generalised elliptic version of the multiple polylogarithms. Furthermore we investigate the elliptic polylogarithms appearing in higher orders in the dimensional regularisation ε of the two-dimensional equal mass solution. Around two space-time dimensions the solution consists of a sum of three elliptic dilogarithms where the arguments have a nice geometric interpretation as intersection points of the integration region and an elliptic curve associated to the sunrise integral. Around four space-time dimensions the sunrise integral can be expressed with the ε{sup 0}- and ε{sup 1}-solution around two dimensions, mass derivatives thereof and simpler terms. Considering higher orders of the two-dimensional equal mass solution we find certain generalisations of the elliptic polylogarithms appearing in the ε{sup 0}- and ε{sup 1}-solutions around two and four space-time dimensions. We show that these higher order-solutions can be found by iterative integration within this class of functions.
Canestrari, N.; Bisogni, V.; Walter, A.; Zhu, Y.; Dvorak, J.; Vescovo, E.; Chubar, O.
2014-09-01
A "source-to-sample" wavefront propagation analysis of the Electron Spectro-Microscopy (ESM) UV / soft X-ray beamline, which is under construction at the National Synchrotron Light Source II (NSLS-II) in the Brookhaven National Laboratory, has been conducted. All elements of the beamline - insertion device, mirrors, variable-line-spacing gratings and slits - are included in the simulations. Radiation intensity distributions at the sample position are displayed for representative photon energies in the UV range (20 - 100 eV) where diffraction effects are strong. The finite acceptance of the refocusing mirrors is the dominating factor limiting the spatial resolution at the sample (by ~3 μm at 20 eV). Absolute estimates of the radiation flux and energy resolution at the sample are also obtained from the electromagnetic calculations. The analysis of the propagated UV range undulator radiation at different deflection parameter values demonstrates that within the beamline angular acceptance a slightly "red-shifted" radiation provides higher flux at the sample and better energy resolution compared to the on-axis resonant radiation of the fundamental harmonic.
International Nuclear Information System (INIS)
Orville, A.M.; Buono, R.; Cowan, M.; Heroux, A.; Shea-McCarthy, G.; Schneider, D.K.; Skinner, J.M.; Skinner, M.J.; Stoner-Ma, D.; Sweet, R.M.
2011-01-01
The research philosophy and new capabilities installed at NSLS beamline X26-C to support electronic absorption and Raman spectroscopies coupled with X-ray diffraction are reviewed. This beamline is dedicated full time to multidisciplinary studies with goals that include revealing the relationship between the electronic and atomic structures in macromolecules. The beamline instrumentation has been fully integrated such that optical absorption spectra and X-ray diffraction images are interlaced. Therefore, optical changes induced by X-ray exposure can be correlated with X-ray diffraction data collection. The installation of Raman spectroscopy into the beamline is also briefly reviewed. Data are now routinely generated almost simultaneously from three complementary types of experiments from the same sample. The beamline is available now to the NSLS general user population.
Elliptic flow from Coulomb interaction and low density elastic scattering
Sun, Yuliang; Li, Qingfeng; Wang, Fuqiang
2018-04-01
In high energy heavy ion collisions and interacting cold atom systems, large elliptic flow anisotropies have been observed. For the large opacity (ρ σ L ˜103 ) of the latter hydrodynamics is a natural consequence, but for the small opacity (ρ σ L ˜1 ) of the former the hydrodynamic description is questionable. To shed light onto the situation, we simulate the expansion of a low density argon ion (or atom) system, initially trapped in an elliptical region, under the Coulomb interaction (or elastic scattering). Significant elliptic anisotropy is found in both cases, and the anisotropy depends on the initial spatial eccentricity and the density of the system. The results may provide insights into the physics of anisotropic flow in high energy heavy ion collisions and its role in the study of quantum chromodynamics.
Iron abundance evolution in spiral and elliptical galaxies
International Nuclear Information System (INIS)
Matteucci, F.
1987-01-01
Chemical evolution models for the Galaxy and ellipticals, which take into account the most recent developments on theories of nucleosynthesis and supernova progenitors, are presented. The evolution of the abundance of iron in these systems, under the assumption that this element is mainly produced by type I SNe, originating from white dwarfs in binary systems, has been computed and the results have been compared with the observations. Overabundances of O, Si, Ne and Mg with respect to iron have been predicted for halo stars in their Galaxy. The existence of an Fe - total mass relation with a slope steeper than the corresponding relations for Mg and O has been predicted for ellipticals. The masses of Fe ejected by ellipticals of various masses into the intergalactic medium have also been computed in detail. The general agreement obtained between these theoretical models and the observations for galaxies of different morphological type supports the assumptions made about the origin of iron
Dirac Particles Emission from An Elliptical Black Hole
Directory of Open Access Journals (Sweden)
Yuant Tiandho
2017-03-01
Full Text Available According to the general theory of relativiy, a black hole is defined as a region of spacetime with super-strong gravitational effects and there is nothing can escape from it. So in the classical theory of relativity, it is safe to say that black hole is a "dead" thermodynamical object. However, by using quantum mechanics theory, Hawking has shown that a black hole may emit particles. In this paper, calculation of temperature of an elliptical black hole when emitting the Dirac particles was presented. By using the complexpath method, radiation can be described as emission process in the tunneling pictures. According to relationship between probability of outgoing particle with the spectrum of black body radiation for fermion particles, temperature of the elliptical black hole can be obtained and it depend on the azimuthal angle. This result also showed that condition on the surface of elliptical black hole is not in thermal equilibrium.
Modern cryptography and elliptic curves a beginner's guide
Shemanske, Thomas R
2017-01-01
This book offers the beginning undergraduate student some of the vista of modern mathematics by developing and presenting the tools needed to gain an understanding of the arithmetic of elliptic curves over finite fields and their applications to modern cryptography. This gradual introduction also makes a significant effort to teach students how to produce or discover a proof by presenting mathematics as an exploration, and at the same time, it provides the necessary mathematical underpinnings to investigate the practical and implementation side of elliptic curve cryptography (ECC). Elements of abstract algebra, number theory, and affine and projective geometry are introduced and developed, and their interplay is exploited. Algebra and geometry combine to characterize congruent numbers via rational points on the unit circle, and group law for the set of points on an elliptic curve arises from geometric intuition provided by Bézout's theorem as well as the construction of projective space. The structure of the...
Applications of elliptic Carleman inequalities to Cauchy and inverse problems
Choulli, Mourad
2016-01-01
This book presents a unified approach to studying the stability of both elliptic Cauchy problems and selected inverse problems. Based on elementary Carleman inequalities, it establishes three-ball inequalities, which are the key to deriving logarithmic stability estimates for elliptic Cauchy problems and are also useful in proving stability estimates for certain elliptic inverse problems. The book presents three inverse problems, the first of which consists in determining the surface impedance of an obstacle from the far field pattern. The second problem investigates the detection of corrosion by electric measurement, while the third concerns the determination of an attenuation coefficient from internal data, which is motivated by a problem encountered in biomedical imaging.
Polarization characteristics of double-clad elliptical fibers.
Zhang, F; Lit, J W
1990-12-20
A scalar variational analysis based on a Gaussian approximation of the fundamental mode of a double-clad elliptical fiber with a depressed inner cladding is studied. The polarization properties and graphic results are presented; they are given in terms of three parameters: the ratio of the major axis to the minor axis of the core, the ratio of the inner cladding major axis to the core major axis, and the difference between the core index and the inner cladding index. The variations of both the spot size and the field intensity with core ellipticity are examined. It is shown that high birefringence and dispersion-free orthogonal polarization modes can be obtained within the single-mode region and that the field intensity distribution may be more confined to the fiber center than in a single-clad elliptical fiber.
System Size, Energy, Pseudorapidity, and Centrality Dependence of Elliptic Flow
Alver, B.; Back, B. B.; Baker, M. D.; Ballintijn, M.; Barton, D. S.; Betts, R. R.; Bickley, A. A.; Bindel, R.; Busza, W.; Carroll, A.; Chai, Z.; Chetluru, V.; Decowski, M. P.; García, E.; Gburek, T.; George, N.; Gulbrandsen, K.; Halliwell, C.; Hamblen, J.; Harnarine, I.; Hauer, M.; Henderson, C.; Hofman, D. J.; Hollis, R. S.; Hołyński, R.; Holzman, B.; Iordanova, A.; Johnson, E.; Kane, J. L.; Khan, N.; Kulinich, P.; Kuo, C. M.; Li, W.; Lin, W. T.; Loizides, C.; Manly, S.; Mignerey, A. C.; Nouicer, R.; Olszewski, A.; Pak, R.; Reed, C.; Richardson, E.; Roland, C.; Roland, G.; Sagerer, J.; Seals, H.; Sedykh, I.; Smith, C. E.; Stankiewicz, M. A.; Steinberg, P.; Stephans, G. S. F.; Sukhanov, A.; Szostak, A.; Tonjes, M. B.; Trzupek, A.; Vale, C.; van Nieuwenhuizen, G. J.; Vaurynovich, S. S.; Verdier, R.; Veres, G. I.; Walters, P.; Wenger, E.; Willhelm, D.; Wolfs, F. L. H.; Wosiek, B.; Woźniak, K.; Wyngaardt, S.; Wysłouch, B.
2007-06-01
This Letter presents measurements of the elliptic flow of charged particles as a function of pseudorapidity and centrality from Cu-Cu collisions at 62.4 and 200 GeV using the PHOBOS detector at the Relativistic Heavy Ion Collider. The elliptic flow in Cu-Cu collisions is found to be significant even for the most central events. For comparison with the Au-Au results, it is found that the detailed way in which the collision geometry (eccentricity) is estimated is of critical importance when scaling out system-size effects. A new form of eccentricity, called the participant eccentricity, is introduced which yields a scaled elliptic flow in the Cu-Cu system that has the same relative magnitude and qualitative features as that in the Au-Au system.
Investigation of collisional effects within the bending magnet region of a DIII-D neutral beamline
International Nuclear Information System (INIS)
Kessler, D.N.; Hong, R.; Kellman, D.H.
1993-10-01
The region between the pole faces of the DIII-D neutral beamline residual ion bending magnets is an area of transient high gas pressure which may cause beam defocusing and increased heating of beamline internal components due to collisional effects. An investigation of these effects helps in understanding residual ion trajectories and in providing information for studying in the beamline capability for operation with increased pulse duration. Examination of collisional effects, and of the possible existence of space charge blow-up, was carried out by injecting deuterium gas into the region between the magnet pole faces with rates varying from 0 to 18 torr-ell/sec. Thermocouple and waterflow calorimetry data were taken to measure the beamline component heating and beam powder deposition on the magnet pole shields, magnet louvers, ion dump, beam collimators, and calorimeter. Data was also taken at gas flow rates varying from 0 to 25 torr-ell/sec into the neutralizer cell and is compared with the magnet region gas injection data obtained. Results show that both collisional effects and space charge blow-up play a role in magnet region component heating and that neutralizer gas flow sufficiently reduces component heating without incurring unacceptable power losses through collisional effects
Wilson, C.; King, J.
The Colorado School of Mines installed a neutron radiography system at the United States Geological Survey TRIGA reactor in 2012. An upgraded beamline could dramatically improve the imaging capabilities of this system. This project performed computational analyses to support the design of a new beamline, with the major goals of minimizing beam divergence and maximizing beam intensity. The new beamline will consist of a square aluminum tube with an 11.43 cm (4.5 in) inner side length and 0.635 cm (0.25 in) thick walls. It is the same length as the original beam tube (8.53 m) and is composed of 1.22 m (4 ft) and 1.52 m (5 ft) flanged sections which bolt together. The bottom 1.22 m of the beamline is a cylindrical aluminum pre-collimator which is 0.635 cm (0.25 in) thick, with an inner diameter of 5.08 cm (2 in). Based on Monte Carlo model results, when a pre-collimator is present, the use of a neutron absorbing liner on the inside surface of the beam tube has almost no effect on the angular distribution of the neutron current at the collimator exit. The use of a pre-collimator may result in a non-uniform flux profile at the image plane; however, as long as the collimator is at least three times longer than the pre-collimator, the flux distortion is acceptably low.
Characteristics and performance of the Los Alamos VUV beamline at the NSLS
International Nuclear Information System (INIS)
Bartlett, R.J.; Trela, W.J.; Michaud, F.D.; Southworth, S.H.; Alkire, R.W.; Roy, P.; Rothe, R.; Walsh, P.J.; Shinn, N.
1988-01-01
We describe the design and performance of the Los Alamos VUV synchrotron radiation beamline, U3C, on the VUV ring of the National Synchrotron Light Source at Brookhaven National Laboratory. The beamline uses separate function optics to collect and focus the horizontally and vertically diverging beam. The monochromator is a grazing incidence Rowland circle instrument of the extended grasshopper design (ERG). A postmonochromator refocusing mirror is used to focus or collimate the diverging beam from the monochromator. The beamline control and diagnostics systems are also discussed. Particular emphasis in the design has been placed on the reduction of stray and harmonic light. Higher order light is reduced by a grazing angle mirror low pass filter installed immediately downstream from the monochromator while stray light is reduced through the use of baffles and thin film filters. Also included in the line is a differential pumping section that permits gas phase and other experiments requiring pressures in the 10 -5 to 10 -4 Torr range to be coupled to the beamline. (orig.)
TIME-RESOLVED INFRARED SPECTROSCOPY IN THE U121R BEAMLINE AT THE NSLS
International Nuclear Information System (INIS)
CARR, G.L.; LAVEIGNE, J.D.; LOBO, R.P.S.M.; REITZE, D.H.; TANNER, D.B.
1999-01-01
A facility for performing time-resolved infrared spectroscopy has been developed at the NSLS, primarily at beamline U12IR. The pulsed IR light from the synchrotron is used to perform pump-probe spectroscopy. The authors present here a description of the facility and results for the relaxation of photoexcitations in both a semiconductor and superconductor
International Nuclear Information System (INIS)
Martin, Michael C.; McKinney, Wayne R.
1999-01-01
Two recently commissioned infrared beamlines on the 1.4 bending magnet port at the Advanced Light Source, LBNL, are described. Using a synchrotron as an IR source provides three primary advantages: increased brightness, very fast light pulses, and enhanced far-IR flux. The considerable brightness advantage manifests itself most beneficially when performing spectroscopy on a microscopic length scale. Beamline (BL) 1.4.3 is a dedicated FTIR spectromicroscopy beamline, where a diffraction-limited spot size using the synchrotron source is utilized. BL 1.4.2 consists of a vacuum FTIR bench with a wide spectral range and step-scan capability. This BL makes use of the pulsed nature of the synchrotron light as well as the far-IR flux. Fast timing is demonstrated by observing the pulses from the electron bunch storage pattern at the ALS. Results from several experiments from both IR beamlines will be presented as an overview of the IR research currently being done at the ALS
CERN’s 2016 Beamline for Schools competition starts on 17 November
2015-01-01
Spread the word: CERN is offering high-school students from around the world the chance to create and perform a scientific experiment on a CERN accelerator beamline. What better way to learn about physics? (Video: Noemi Caraban/CERN ) Now in its third year, the Beamline for Schools competition is open to teams of at least five students aged 16 and with at least one adult supervisor or “coach”. Students can find out about the beamline and facilities via http://cern.ch/bl4s, then think of a simple, creative experiment. They can register their team from 17 November to start receiving e-mail updates. They then submit a written proposal and a short video by 31 March 2016. The winners will be announced in June and will come to CERN, preferably in September 2016. Previous winners have tested webcams and classroom-grown crystals at the beamline, others have studied how particles decay and investigated high-energy gamma rays. All participants will receive...
The current status of small-angle x-ray scattering beamline at Diamond Light Source
International Nuclear Information System (INIS)
Inoue, Katsuaki; Doutch, James; Terrill, Nick
2013-01-01
The small-angle X-ray scattering (SAXS) covers the major disciplines of biology, chemistry and physics delivering structural and dynamic information in nanoscience, mesoscopic architectures, supramolecular structures, and nucleation/growth of crystals. SAXS is also proving to be important in archaeological, environmental, and conservation sciences, and has further indicated its ability to span wide-ranging scientific disciplines. Thus, strong needs for SAXS studies are increasing significantly in a broad range of scientific fields year by year. Based on such a background, the demand for high throughput SAXS experiments is increasing. At the synchrotron facility, Diamond Light Source, one SAXS beamline, Non-crystalline diffraction I22 is now operational and highly automated throughput small-angle X-ray scattering (HATSAXS) beamline B21 is now under construction. I22 is the Undulator beamline and wide varieties of experiments, including time-resolved experiments are attempted. Based on the concept of HATSAXS, the key feature of B21 will focuses on the automation of end-station equipment. A automated sample changer has been purchased for solution SAXS measurements on biomolecules. A robotic-arm-type automated sample changer that is capable of handling several kinds of samples in material science is also being constructed. B21 is expected to successfully provide all users highly automated throughput measurements with the highest possible reliability and accuracy. Construction of this beamline will end in the second half of 2012, and will be open for users in the early summer of 2013 after commissioning. (author)
Photon-ion spectrometer PIPE at the Variable Polarization XUV Beamline of PETRA III
International Nuclear Information System (INIS)
Schippers, S; Ricz, S; Buhr, T; Hellhund, J; Müller, A; Klumpp, S; Martins, M; Flesch, R; Rühl, E; Lower, J; Jahnke, T; Metz, D; Schmidt, L Ph H; Dörner, R; Ullrich, J; Wolf, A
2012-01-01
The photon-ion spectrometer PIPE is currently being installed as a permanent end station at beamline P04 of the PETRA III synchrotron radiation source. Various state-of-the-art experimental techniques will be available for studies of gaseous matter with circularly and linearly polarized synchrotron radiation with photon energies in range the 100–3000 eV.
International Nuclear Information System (INIS)
Cipriani, Florent; Felisaz, Franck; Lavault, Bernard; Brockhauser, Sandor; Ravelli, Raimond; Launer, Ludovic; Leonard, Gordon; Renier, Michel
2007-01-01
While routine Macromolecular x-ray (MX) crystallography has relied on well established techniques for some years all the synchrotrons around the world are improving the throughput of their MX beamlines. Third generation synchrotrons provide small intense beams that make data collection of 5-10 microns sized crystals possible. The EMBL/ESRF MX Group in Grenoble has developed a new generation of instruments to easily collect data on 10 μm size crystals in an automated environment. This work is part of the Grenoble automation program that enables FedEx like crystallography using fully automated data collection and web monitored experiments. Seven ESRF beamlines and the MRC BM14 ESRF/CRG beamline are currently equipped with these latest instruments. We describe here the main features of the MD2x diffractometer family and the SC3 sample changer robot. Although the SC3 was primarily designed to increase the throughput of MX beamlines, it has also been shown to be efficient in improving the quality of the data collected. Strategies in screening a large number of crystals, selecting the best, and collecting a full data set from several re-oriented micro-crystals can now be run with minimum time and effort. The MD2x and SC3 instruments are now commercialised by the company ACCEL GmbH
The materials science synchrotron beamline EDDI for energy-dispersive diffraction analysis
International Nuclear Information System (INIS)
Genzel, Ch.; Denks, I.A.; Gibmeier, J.; Klaus, M.; Wagener, G.
2007-01-01
In April 2005 the materials science beamline EDDI (Energy Dispersive DIffraction) at the Berlin synchrotron storage ring BESSY started operation. The beamline is operated in the energy-dispersive mode of diffraction using the high energy white photon beam provided by a superconducting 7 T multipole wiggler. Starting from basic information on the beamline set-up, its measuring facilities and data processing concept, the wide range of applications for energy-dispersive diffraction is demonstrated by a series of examples coming from different fields in materials sciences. It will be shown, that the EDDI beamline is especially suitable for the investigation of structural properties and gradients in the near surface region of polycrystalline materials. In particular, this concerns the analysis of multiaxial residual stress fields in the highly stressed surface zone of technical parts. The high photon flux further facilitates fast in situ experiments at room as well as high temperature to monitor for example the growth kinetics and reaction in thin film growth
Neutrinos from the NuMI beamline in the MiniBooNE detector
International Nuclear Information System (INIS)
Aguilar-Arevalo, Alexis A.
2006-01-01
With the startup of the NuMI beamline early in 2005, the MiniBooNE detector has the unique opportunity to be the first user of an off-axis neutrino beam (110 mrad off-axis). MiniBooNE is assembling a rich sample of neutrino interactions from this source
Blume, H; Bourenkov, G P; Kosciesza, D; Bartunik, H D
2001-01-01
The wiggler beamline BW6 at DORIS has been optimized for de-novo solution of protein structures on the basis of MAD phasing. Facilities for automatic data collection, rapid data transfer and storage, and online processing have been developed which provide adequate conditions for high-throughput applications, e.g., in structural genomics.
Investigations of Solid State Dynamics at the NRSSR Beamline at PETRA II. An Overview
Energy Technology Data Exchange (ETDEWEB)
Franz, H. [HAmburger SYnchrotronstrahlungs LABor, DESY (Germany); Asthalter, T. [Universitaet Stuttgart (Germany); Dommach, M.; Ehnes, A. [HAmburger SYnchrotronstrahlungs LABor, DESY (Germany); Messel, K.; Sergueev, I. [Technische Universitaet Muenchen, James-Franck-Strasse (Germany)
2002-06-15
The present status of the new nuclear resonance beamline PETRA 1 at HASYLAB, DESY, Hamburg is described. Besides an overview of the experimental setup some examples of recent experiments are given. Those cover the main applications, i.e., inelastic scattering from iron alloys and quasielastic scattering from glass-forming liquids.
Investigations of Solid State Dynamics at the NRSSR Beamline at PETRA II. An Overview
International Nuclear Information System (INIS)
Franz, H.; Asthalter, T.; Dommach, M.; Ehnes, A.; Messel, K.; Sergueev, I.
2002-01-01
The present status of the new nuclear resonance beamline PETRA 1 at HASYLAB, DESY, Hamburg is described. Besides an overview of the experimental setup some examples of recent experiments are given. Those cover the main applications, i.e., inelastic scattering from iron alloys and quasielastic scattering from glass-forming liquids.
Vacuum ultraviolet beamline at the Swiss Light Source for chemical dynamics studies
International Nuclear Information System (INIS)
Johnson, Melanie; Bodi, Andras; Schulz, Lothar; Gerber, Thomas
2009-01-01
A bend-magnet vacuum ultraviolet (VUV) beamline, intended for chemical dynamics studies, was constructed and brought into operation at the Swiss Light Source (SLS) of the Paul Scherrer Institut. The beamline delivers synchrotron radiation in the 5-30 eV photon energy range with a photon flux of 10 11 photons/s at 10 eV and 10 12 photons/s at 20 eV with a resolving power of 2500. The resolving power increases to 10 4 at the cost of photon flux. An in-house designed rare gas filter is used to suppress higher harmonic radiation by a factor of >10 4 , yielding purely monochromatic light in the energy range of 5-21.6 eV. The filter is compact, easy to align, requires a total pumping power of less than 645 l/s and consumes only 3 normal l/h of filter gas. It is located at the end of the beamline, right in front of the experimental endstation. It is usually operated at a higher pressure than the endstation, which offers the additional benefit of protecting the beamline vacuum from sample contamination.
Current status of the hard x-ray nanoprobe beamline at the SSRF
Li, Aiguo; Jiang, Hui; Wang, Hua; Zhang, Zhaohong; He, Yan; Zhao, Gaofeng; Shu, Deming
2017-09-01
The hard X-ray nanoprobe beamline (HXN) designed at the Shanghai Synchrotron Radiation facility (SSRF) will be of capability to realize a focal spot size of 10 nm for hard X-rays to satisfy requirements in biology, environmental, material sciences and etc.. The beamline includes two modes of operation, high energy resolution mode and high flux mode respectively. High flux mode utilizes the multilayer KB system to obtain high-flux diffraction-limited focusing of 10nm. An ultra-high-precision figure fabrication for diffraction-limited focusing is required to meet the Rayleigh Criterion. An idea to overcome this problem is to introduce a phase compensator upstream of the KB system to compensate the wavefront errors in the beamline. At wavelength speckle-based method will be used to measure the wavefront error in the beamline and feedback to the phase compensator. Vibration measurements have been carried out at the secondary source and endstation hutch. The flexure hinge mechanisms and high-precision actuators ensure the KB system and sample manipulator working with high stability. The building of HXN has been designed and is under construction at present.
Transfer coefficients for plate fin and elliptical tube heat exchangers
International Nuclear Information System (INIS)
Saboya, S.M.; Saboya, F.E.M.
1981-01-01
In order to determine transfer coefficients for plate fin and elliptical tube exchangers, mass transfer experiments have been performed using the naphthalene sublimation technique. By means of the heat-mass transfer analogy, the results can be converted to heat transfer results. The transfer coefficients were compared with those for circular tube exchangers and the comparison revealed no major differences. This is a positive outcome, since the use of elliptical tubes may reduce substantially the pressure drop, without affecting the transfer characteristics.(Author) [pt
Landau-Ginzburg Orbifolds, Mirror Symmetry and the Elliptic Genus
Berglund, P.; Henningson, M.
1994-01-01
We compute the elliptic genus for arbitrary two dimensional $N=2$ Landau-Ginzburg orbifolds. This is used to search for possible mirror pairs of such models. We show that if two Landau-Ginzburg models are conjugate to each other in a certain sense, then to every orbifold of the first theory corresponds an orbifold of the second theory with the same elliptic genus (up to a sign) and with the roles of the chiral and anti-chiral rings interchanged. These orbifolds thus constitute a possible mirr...
FDM for Elliptic Equations with Bitsadze-Samarskii-Dirichlet Conditions
Directory of Open Access Journals (Sweden)
Allaberen Ashyralyev
2012-01-01
Full Text Available A numerical method is proposed for solving nonlocal boundary value problem for the multidimensional elliptic partial differential equation with the Bitsadze-Samarskii-Dirichlet condition. The first and second-orders of accuracy stable difference schemes for the approximate solution of this nonlocal boundary value problem are presented. The stability estimates, coercivity, and almost coercivity inequalities for solution of these schemes are established. The theoretical statements for the solutions of these nonlocal elliptic problems are supported by results of numerical examples.
Coexistence of a General Elliptic System in Population Dynamics
DEFF Research Database (Denmark)
Pedersen, Michael
2004-01-01
This paper is concerned with a strongly-coupled elliptic system representing a competitive interaction between two species. We give a sufficient condition for the existence of positive solutions. An example is also given to show that there is a coexistence of a steady state if the cross-diffusion......This paper is concerned with a strongly-coupled elliptic system representing a competitive interaction between two species. We give a sufficient condition for the existence of positive solutions. An example is also given to show that there is a coexistence of a steady state if the cross...
UV Visibility of Moderate-Redshift Giant Elliptical Galaxies
Directory of Open Access Journals (Sweden)
Myung-Hyun Rhee
1998-06-01
Full Text Available We show quantitatively whether giant elliptical galaxies would be visible at far UV wavelengths if they were placed at moderate redshift of 0.4-0.5. On the basis of simple cosmological tests, we conclude that giant elliptical galaxies can be detectable upto the redshift of 0.4-0.5 in the proposed GALEX (Galaxy Evolution Explorer Deep Imaging Survey. We also show that obtaining UV color index such as m_1550 - V from upcoming GALEX and SDSS (Sloan Digital Sky Survey observations should be feasible.
Plasma blob generation due to cooperative elliptic instability.
Manz, P; Xu, M; Müller, S H; Fedorczak, N; Thakur, S C; Yu, J H; Tynan, G R
2011-11-04
Using fast-camera measurements the generation mechanism of plasma blobs is investigated in the linear device CSDX. During the ejection of plasma blobs the plasma is dominated by an m=1 mode, which is a counterrotating vortex pair. These flows are known to be subject to the cooperative elliptic instability, which is characterized by a cooperative disturbance of the vortex cores and results in a three-dimensional breakdown of two-dimensional flows. The first experimental evidence of a cooperative elliptic instability preceding the blob-ejection is provided in terms of the qualitative evolution of the vortex geometries and internal wave patterns.
Inflation of polymer melts into elliptic and circular cylinders
DEFF Research Database (Denmark)
Rasmussen, Henrik Koblitz; Christensen, Jens Horslund; Gøttsche, Søren
2000-01-01
A thin sheet (membrane) of the polymeric material is clamped between a Teflon-coated thermostated plate and a thermostated aluminium cylinder. By applying thermostated air through the plate, the polymer membrane deforms into an elliptic or a circular cylinder. The position of the top of the infla......A thin sheet (membrane) of the polymeric material is clamped between a Teflon-coated thermostated plate and a thermostated aluminium cylinder. By applying thermostated air through the plate, the polymer membrane deforms into an elliptic or a circular cylinder. The position of the top...
Large N elliptic genus and AdS/CFT Correspondence
International Nuclear Information System (INIS)
Boer, Jan de
1998-01-01
According to one of Maldacena's dualities, type IIB string theory on AdS 3 x S 3 x K3 is equivalent to a certain N = (4, 4) superconformal field theory. In this note we compute the elliptic genus of the boundary theory in the supergravity approximation. A finite quantity is obtained once we introduce a particular exclusion principle. In the regime where the supergravity approximation is reliable, we find exact agreement with the elliptic genus of a sigma model with target space K3 N /S N
Hörmander spaces, interpolation, and elliptic problems
Mikhailets, Vladimir A; Malyshev, Peter V
2014-01-01
The monograph gives a detailed exposition of the theory of general elliptic operators (scalar and matrix) and elliptic boundary value problems in Hilbert scales of Hörmander function spaces. This theory was constructed by the authors in a number of papers published in 2005-2009. It is distinguished by a systematic use of the method of interpolation with a functional parameter of abstract Hilbert spaces and Sobolev inner product spaces. This method, the theory and their applications are expounded for the first time in the monographic literature. The monograph is written in detail and in a
Elliptic interpretation of black holes and quantum mechanics
International Nuclear Information System (INIS)
Gibbons, G.W.
1987-01-01
The lectures as delivered contained an elementary introduction to the classical theory of black holes together with an account of Hawking's original derivation of the thermal emission from black holes in the quantum theory. Also described here is what is here called the elliptic interpretation partly because of its possible relevance to the lectures of Professor 't Hooft. A rather more detailed account of the elliptic interpretation is given and the reader is referred to the original literature for the elementary material. 22 references
An electrostatic elliptical mirror for neutral polar molecules.
González Flórez, A Isabel; Meek, Samuel A; Haak, Henrik; Conrad, Horst; Santambrogio, Gabriele; Meijer, Gerard
2011-11-14
Focusing optics for neutral molecules finds application in shaping and steering molecular beams. Here we present an electrostatic elliptical mirror for polar molecules consisting of an array of microstructured gold electrodes deposited on a glass substrate. Alternating positive and negative voltages applied to the electrodes create a repulsive potential for molecules in low-field-seeking states. The equipotential lines are parallel to the substrate surface, which is bent in an elliptical shape. The mirror is characterized by focusing a beam of metastable CO molecules and the results are compared to the outcome of trajectory simulations.
Elliptic Genera of Symmetric Products and Second Quantized Strings
Dijkgraaf, R; Verlinde, Erik; Verlinde, Herman L
1997-01-01
In this note we prove an identity that equates the elliptic genus partition function of a supersymmetric sigma model on the $N$-fold symmetric product $M^N/S_N$ of a manifold $M$ to the partition function of a second quantized string theory on the space $M \\times S^1$. The generating function of these elliptic genera is shown to be (almost) an automorphic form for $O(3,2,\\Z)$. In the context of D-brane dynamics, this result gives a precise computation of the free energy of a gas of D-strings inside a higher-dimensional brane.
Event-by-Event Elliptic Flow Fluctuations from PHOBOS
Wosiek, B.; Alver, B.; Back, B. B.; Baker, M. D.; Ballintijn, M.; Barton, D. S.; Betts, R. R.; Bickley, A. A.; Bindel, R.; Busza, W.; Carroll, A.; Chai, Z.; Chetluru, V.; Decowski, M. P.; García, E.; Gburek, T.; George, N.; Gulbrandsen, K.; Halliwell, C.; Hamblen, J.; Harnarine, I.; Hauer, M.; Henderson, C.; Hofman, D. J.; Hollis, R. S.; Hołyński, R.; Holzman, B.; Iordanova, A.; Johnson, E.; Kane, J. L.; Khan, N.; Kulinich, P.; Kuo, C. M.; Li, W.; Lin, W. T.; Loizides, C.; Manly, S.; Mignerey, A. C.; Nouicer, R.; Olszewski, A.; Pak, R.; Reed, C.; Richardson, E.; Roland, C.; Roland, G.; Sagerer, J.; Seals, H.; Sedykh, I.; Smith, C. E.; Stankiewicz, M. A.; Steinberg, P.; Stephans, G. S. F.; Sukhanov, A.; Szostak, A.; Tonjes, M. B.; Trzupek, A.; Vale, C.; van Nieuwenhuizen, G. J.; Vaurynovich, S. S.; Verdier, R.; Veres, G. I.; Walters, P.; Wenger, E.; Willhelm, D.; Wolfs, F. L. H.; Woźniak, K.; Wyngaardt, S.; Wysłouch, B.
2009-04-01
Recently PHOBOS has focused on the study of fluctuations and correlations in particle production in heavy-ion collisions at the highest energies delivered by the Relativistic Heavy Ion Collider (RHIC). In this report, we present results on event-by-event elliptic flow fluctuations in (Au+Au) collisions at sqrt {sNN}=200 GeV. A data-driven method was used to estimate the dominant contribution from non-flow correlations. Over the broad range of collision centralities, the observed large elliptic flow fluctuations are in agreement with the fluctuations in the initial source eccentricity.
Elliptic flow in Au+Au collisions at RHIC
Vale, Carla M.; PHOBOS Collaboration; Back, B. B.; Baker, M. D.; Ballintijn, M.; Barton, D. S.; Betts, R. R.; Bickley, A. A.; Bindel, R.; Budzanowski, A.; Busza, W.; Carroll, A.; Decowski, M. P.; García, E.; George, N.; Gulbrandsen, K.; Gushue, S.; Halliwell, C.; Hamblen, J.; Heintzelman, G. A.; Henderson, C.; Hofman, D. J.; Hollis, R. S.; Holynski, R.; Holzman, B.; Iordanova, A.; Johnson, E.; Kane, J. L.; Katzy, J.; Khan, N.; Kucewicz, W.; Kulinich, P.; Kuo, C. M.; Lin, W. T.; Manly, S.; McLeod, D.; Mignerey, A. C.; Ngyuen, M.; Nouicer, R.; Olszewski, A.; Pak, R.; Park, I. C.; Pernegger, H.; Reed, C.; Remsberg, L. P.; Reuter, M.; Roland, C.; Roland, G.; Rosenberg, L.; Sagerer, J.; Sarin, P.; Sawicki, P.; Skulski, W.; Steinberg, P.; Stephans, G. S. F.; Sukhanov, A.; Tang, J.-L.; Tonjes, M. B.; Trzupek, A.; van Nieuwenhuizen, G. J.; Verdier, R.; Veres, G.; Wolfs, F. L. H.; Wosiek, B.; Wozniak, K.; Wuosmaa, A. H.; Wyslouch, B.
2005-04-01
Elliptic flow is an interesting probe of the dynamical evolution of the dense system formed in the ultrarelativistic heavy ion collisions at the relativistic heavy ion collider (RHIC). The elliptic flow dependences on transverse momentum, centrality and pseudorapidity were measured using data collected by the PHOBOS detector, which offers a unique opportunity to study the azimuthal anisotropies of charged particles over a wide range of pseudorapidity. These measurements are presented, together with an overview of the analysis methods and a discussion of the results.
Rossendorf Beamline at ESRF (ROBL-CRG). Bi-annual report 2009/2010
International Nuclear Information System (INIS)
Scheinost, Andreas C.; Baehtz, Carsten
2011-01-01
The Rossendorf Beamline (ROBL) - located at BM20 of the European Synchrotron Radiation Facility (ESRF) in Grenoble, France - is in operation since 1998. This 7th report covers the period from January 2009 to December 2010. In these two years, 67 peer- reviewed papers have been published based on experiments done at the beamline, more than in any biannual period before. Six highlight reports have been selected for this report to demonstrate the scientific strength and diversity of the experiments performed on the two end-stations of the beamline, dedicated to Radiochemistry (RCH) and Materials Research (MRH). The beamtime was more heavily overbooked than ever before, with an acceptance rate of only 25% experiments. We would like to thank our external proposal review members, Prof. Andre Maes (KU Leuven, Belgium), Prof. Laurent Charlet (UJF Grenoble, France), Dr. Andreas Leinweber (MPI Metallforschung, Stuttgart, Germany), Prof. David Rafaja (TU Bergakademie Freiberg, Germany), Prof. Dirk Meyer (TU Dresden, Germany), who evaluated the inhouse proposals in a thorough manner, thereby ensuring that beamtime was distributed according to scientific merit. The period was not only characterized by very successful science, but also by intense work on the optics upgrade. In spring 2009, a workshop was held at ROBL, assembling beamline experts from German, Spanish and Swiss synchrotrons, to evaluate the best setup for the new optics. These suggestions was used to prepare the call for tender published in July 2009. From the tender acceptance in November 2009 on, a series of design review meetings and factory acceptance tests followed. Already in July 2010, the first piece of equipment was delivered, the new double-crystal, double-multilayer monochromator. The disassembly of the old optics components started end of July, 2011, followed by the installation of the new components. As of December 2011, the new optics have seen the first test beam and thorough hot commissioning will
Rossendorf Beamline at ESRF (ROBL-CRG). Bi-annual report 2009/2010
Energy Technology Data Exchange (ETDEWEB)
Scheinost, Andreas C; Baehtz, Carsten [eds.
2011-07-01
The Rossendorf Beamline (ROBL) - located at BM20 of the European Synchrotron Radiation Facility (ESRF) in Grenoble, France - is in operation since 1998. This 7th report covers the period from January 2009 to December 2010. In these two years, 67 peer- reviewed papers have been published based on experiments done at the beamline, more than in any biannual period before. Six highlight reports have been selected for this report to demonstrate the scientific strength and diversity of the experiments performed on the two end-stations of the beamline, dedicated to Radiochemistry (RCH) and Materials Research (MRH). The beamtime was more heavily overbooked than ever before, with an acceptance rate of only 25% experiments. We would like to thank our external proposal review members, Prof. Andre Maes (KU Leuven, Belgium), Prof. Laurent Charlet (UJF Grenoble, France), Dr. Andreas Leinweber (MPI Metallforschung, Stuttgart, Germany), Prof. David Rafaja (TU Bergakademie Freiberg, Germany), Prof. Dirk Meyer (TU Dresden, Germany), who evaluated the inhouse proposals in a thorough manner, thereby ensuring that beamtime was distributed according to scientific merit. The period was not only characterized by very successful science, but also by intense work on the optics upgrade. In spring 2009, a workshop was held at ROBL, assembling beamline experts from German, Spanish and Swiss synchrotrons, to evaluate the best setup for the new optics. These suggestions was used to prepare the call for tender published in July 2009. From the tender acceptance in November 2009 on, a series of design review meetings and factory acceptance tests followed. Already in July 2010, the first piece of equipment was delivered, the new double-crystal, double-multilayer monochromator. The disassembly of the old optics components started end of July, 2011, followed by the installation of the new components. As of December 2011, the new optics have seen the first test beam and thorough hot commissioning will
Workshop on cooling of x-ray monochromators on high power beamlines
International Nuclear Information System (INIS)
Matsushita, T.; Ishikawa, T.
1989-03-01
This report is a Workshop on Cooling of X-ray Monochromators on High Power Beamlines held on August 31, 1988 at the Photon Factory during the Third International Conference on Synchrotron Radiation Instrumentation (SRI88). On high power beamlines, especially on insertion device beamlines, heating of crystal monochromators is becoming a serious problem: Researchers observe that the intensity of the X-ray beam on the sample is not proportional to the source intensity because of thermal distortion of the monochromator crystal. This problem will be even more serious on beamlines for the next generation X-ray rings. In the very tight program of the SRI88 conference, only 2 speakers were able to give invited talks closely related to this problem in the session of OPTICAL COMPONENTS FOR HIGH POWER BEAMLINES on Wednesday morning of August 31, 1988. We held this workshop in the afternoon of the same day with the intention of offering further opportunities to exchange information on efforts underway at various laboratories and to discuss ideas how to solve this problem. We also intended that the workshop would be a 'follow-up' to the X-ray optics workshop held at ESRF, Grenoble in September 1987, where the importance of crystal cooling was strongly pointed out. There were 32 participants from 7 countries. 12 people represented their experiences and ideas for reducing thermal distortion of crystal monochromators. Following those presentations, there were discussions on collaborations for solving this important problem. The attendees agreed that exchange of information should be continued by holding such meetings at reasonable intervals. (J.P.N.)
Development of the XFP beamline for x-ray footprinting at NSLS-II
Energy Technology Data Exchange (ETDEWEB)
Bohon, Jen, E-mail: jbohon@bnl.gov; Sullivan, Michael; Abel, Don; Toomey, John; Chance, Mark R., E-mail: mark.chance@case.edu [Case Center for Synchrotron Biosciences, Center for Proteomics and Bioinformatics, Case Western Reserve University, Cleveland, OH (United States); Dvorak, Joseph [Brookhaven National Laboratory, Upton, NY (United States)
2016-07-27
For over a decade, synchrotron-based footprinting studies at the NSLS X28C beamline have provided unique insights and approaches for examining the solution-state structures of large macromolecular assemblies, membrane proteins, and soluble proteins, for time-resolved studies of macromolecular dynamics, and most recently for in vivo studies of RNA-protein complexes. The transition from NSLS to NSLS-II has provided the opportunity to create an upgraded facility for the study of increasingly complex systems; progress on the development of the XFP (X-ray Footprinting for In Vitro and In Vivo Structural Studies of Biological Macromolecules) beamline at NSLS-II is presented here. The XFP beamline will utilize a focused 3-pole wiggler source to deliver a high flux density x-ray beam, where dynamics can be studied on the microsecond to millisecond timescales appropriate for probing biological macromolecules while minimizing sample perturbation. The beamline optics and diagnostics enable adaptation of the beam size and shape to accommodate a variety of sample morphologies with accurate measurement of the incident beam, and the upgrades in sample handling and environment control will allow study of highly sensitive or unstable samples. The XFP beamline is expected to enhance relevant flux densities more than an order of magnitude from that previously available at X28C, allowing static and time-resolved structural analysis of highly complex samples that have previously pushed the boundaries of x-ray footprinting technology. XFP, located at NSLS-II 17-BM, is anticipated to become available for users in 2016.
7-GeV advanced photon source beamline initiative: Conceptual design report
International Nuclear Information System (INIS)
1993-05-01
The DOE is building a new generation 6-7 GeV Synchrotron Radiation Source known as the Advanced Photon Source (APS) at Argonne National Laboratory. This facility, to be completed in FY 1996, can provide 70 x-ray sources of unprecedented brightness to meet the research needs of virtually all scientific disciplines and numerous technologies. The technological research capability of the APS in the areas of energy, communications and health will enable a new partnership between the DOE and US industry. Current funding for the APS will complete the current phase of construction so that scientists can begin their applications in FY 1996. Comprehensive utilization of the unique properties of APS beams will enable cutting-edge research not currently possible. It is now appropriate to plan to construct additional radiation sources and beamline standard components to meet the excess demands of the APS users. In this APS Beamline Initiative, 2.5-m-long insertion-device x-ray sources will be built on four straight sections of the APS storage ring, and an additional four bending-magnet sources will also be put in use. The front ends for these eight x-ray sources will be built to contain and safeguard access to these bright x-ray beams. In addition, funds will be provided to build standard beamline components to meet scientific and technological research demands of the Collaborative Access Teams. The Conceptual Design Report (CDR) for the APS Beamline Initiative describes the scope of all the above technical and conventional construction and provides a detailed cost and schedule for these activities. The document also describes the preconstruction R ampersand D plans for the Beamline Initiative activities and provides the cost estimates for the required R ampersand D
New developments at the INE-Beamline for actinide research at ANKA
Dardenne, K.; Brendebach, B.; Denecke, M. A.; Liu, X.; Rothe, J.; Vitova, T.
2009-11-01
The INE-Beamline for actinide research at the synchrotron source ANKA is operated by the Institut für Nukleare Entsorgung (INE) at the Forschungszentrum Karlsruhe. Experiments on radioactive samples with activities up to 106 times the limit of exemption inside a safe and flexible double containment concept are possible. One great advantage of the beamline is its close proximity to INE's active laboratories with its equipment for manipulation of actinide materials and state-of-the-art spectroscopic, analytical, and microscopic instrumentation. This constellation is unique in Europe. The INE-Beamline is built primarily to serve INE in-house research associated with safe disposal of high level nuclear waste such as actinide speciation or coordination-, redox-, and geo-chemistry of actinides. A wide energy range from around 2.1 keV to 25 keV covering the K-edges from P to Pd and the L3, L2, and L1 edges for actinides from Th to Cm can be used. The INE-Beamline is optimized for X-ray absorption spectroscopy techniques (XANES/EXAFS), but x-ray fluorescence (XRF) analysis and powder diffraction (XRD) are also possible, as well as surface sensitive measurements in grazing incidence geometry (GI-XAFS). Upgrades of instrumentation and extension of experimental capabilities at the INE-Beamline are driven by user needs. Two of the recent upgrades are presented: 1) installation of a microfocus option for spatially resolved studies (μ-XRF, μ-XANES, μ-XRD) and investigations of small volumes (e.g., heterogeneous natural samples and diamond anvil high pressure cells); 2) construction, and commissioning of a high resolution x-ray emission spectrometer (HRXES); 3) availability of an electrochemical cell for investigation of redox sensitive systems.
Nanoscopium: a Scanning Hard X-ray Nanoprobe Beamline at Synchrotron Soleil
Somogyi, A.; Polack, F.; Moreno, T.
2010-06-01
Nanoscopium is the single scanning hard X-ray nano-probe beamline planned at SOLEIL. This ˜155 m long beamline will fully exploit the high brilliance and coherence characteristics of the X-ray beam both for diffraction limited focusing and for contrast formation. It will offer the most advanced imaging techniques in multimodal mode and will be a research tool for a wide user community working in the fields of earth-, environmental-, and life-sciences. The different μ-μnano-probe techniques offered by the beamline will permit elemental mapping at trace (ppm) levels (scanning XRF), speciation mapping (XANES), phase gradient mapping (scanning differential phase contrast), and density-contrast based imaging of internal structures (coherent diffraction imaging) in the 30 nm to 1 μm spatial resolution range, also in "in situ conditions". Nanoscopium will cover the 5-20 keV energy range. The stability of the nanobeam will be ensured by horizontally reflecting beamline optics (a sagitally and a tangentially pre-focusing mirror, horizontally reflecting monochromators) in front of the overfilled secondary source. Trade-off between high energy resolution (ΔE/E˜10-4) and high flux (1011 ph/s with ΔE/E˜10-2) will be achieved by two interchangeable monochromators (a double crystal and a double multilayer one). KB mirror and FZP lenses will be used as focusing devices. The beamline is in the design and construction phase. It is foreseen to be open for users at the beginning of 2013.
Tugendhat, Tim M.; Schäfer, Björn Malte
2018-05-01
We investigate a physical, composite alignment model for both spiral and elliptical galaxies and its impact on cosmological parameter estimation from weak lensing for a tomographic survey. Ellipticity correlation functions and angular ellipticity spectra for spiral and elliptical galaxies are derived on the basis of tidal interactions with the cosmic large-scale structure and compared to the tomographic weak-lensing signal. We find that elliptical galaxies cause a contribution to the weak-lensing dominated ellipticity correlation on intermediate angular scales between ℓ ≃ 40 and ℓ ≃ 400 before that of spiral galaxies dominates on higher multipoles. The predominant term on intermediate scales is the negative cross-correlation between intrinsic alignments and weak gravitational lensing (GI-alignment). We simulate parameter inference from weak gravitational lensing with intrinsic alignments unaccounted; the bias induced by ignoring intrinsic alignments in a survey like Euclid is shown to be several times larger than the statistical error and can lead to faulty conclusions when comparing to other observations. The biases generally point into different directions in parameter space, such that in some cases one can observe a partial cancellation effect. Furthermore, it is shown that the biases increase with the number of tomographic bins used for the parameter estimation process. We quantify this parameter estimation bias in units of the statistical error and compute the loss of Bayesian evidence for a model due to the presence of systematic errors as well as the Kullback-Leibler divergence to quantify the distance between the true model and the wrongly inferred one.
The Closest Point Method and Multigrid Solvers for Elliptic Equations on Surfaces
Chen, Yujia; Macdonald, Colin B.
2015-01-01
© 2015 Society for Industrial and Applied Mathematics. Elliptic partial differential equations are important from both application and analysis points of view. In this paper we apply the closest point method to solve elliptic equations on general
Efficient method for finding square roots for elliptic curves over OEF
CSIR Research Space (South Africa)
Abu-Mahfouz, Adnan M
2009-01-01
Full Text Available Elliptic curve cryptosystems like others public key encryption schemes, require computing a square roots modulo a prime number. The arithmetic operations in elliptic curve schemes over Optimal Extension Fields (OEF) can be efficiently computed...
International Nuclear Information System (INIS)
Irick, S.C.; Jung, C.R.
1997-07-01
There are a number of ray trace programs currently used for the design of synchrotron beamlines. While several of these programs have been written and used mostly within the programmer''s institution, many have also been available to the general public. This paper discusses three such programs. One is a commercial product oriented for the general optical designer (not specifically for synchrotron beamlines). One is designed for synchrotron beamlines and is free with restricted availability. Finally, one is designed for synchrotron beamlines and is used primarily in one institution. The wealth of information from general optical materials and components catalogs is readily available in the commercial program for general optical designs. This makes the design of an infrared beamline easier from the standpoint of component selection. However, this program is not easily configured for synchrotron beamline designs, particularly for a bending magnet source. The synchrotron ray trace programs offer a variety of sources, but generally are not as easy to use from the standpoint of the user interface. This paper shows ray traces of the same beamline Optikwerks, SHADOW, and RAY, and compares the results
Energy Technology Data Exchange (ETDEWEB)
Chung, B. H.; Yoon, S. P.; Seol, K. T.; Kim, H. S.; Kwon, H. J.; Cho, Y. S. [Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of)
2015-10-15
The 100-MeV beamline consist of 5 target room, a TR 103 as one of these is operating beamline, and a TR 101 as the other beamline is under construction as shown in Fig. 1. The TR 101 as beamline target room will be used for the high value-added medical isotope production and increased utilization of the proton accelerator. The optical system of the beamline consisted of dipole and quadrupole, and it included beam position monitor (BPM) and current transformer (CT) for beam diagnostics. The beamline was inserted into the carbon block and the aluminum collimator, the end of pipe as beam window was used for the aluminum to reduce the radioactive of materials. The target transfer equipment is being installed for RI production. The RI Beamline was aligned using the laser tracker, and vacuum leak was not detected by the helium leak detector. This facility is expected to the high value-added medical isotope production and increased utilization of the proton accelerator.
Design of an elliptical solenoid magnet for transverse beam matching to the spiral inflector
International Nuclear Information System (INIS)
Goswami, A.; Sing Babu, P.; Pandit, V.S.
2013-01-01
In this work, we present the design study of an elliptical solenoid magnet to be used for transverse beam matching at the input of a spiral inflector for efficient transmission. We have studied the dependence of axial field and gradients in the transverse directions of the elliptical solenoid magnet with ellipticity of the aperture. Using the beam envelope equations we have studied the feasibility of using an elliptical solenoid for transverse beam matching to the acceptance of a spiral inflector. (author)
Optimization of elliptic neutron guides for triple-axis spectroscopy
International Nuclear Information System (INIS)
Janoschek, M.; Boeni, P.; Braden, M.
2010-01-01
In the last decade the performance of neutron guides for the transport of neutrons has been significantly increased. The most recent developments have shown that elliptic guide systems can be used to focus neutron beams while simultaneously reducing the number of neutron reflections, hence, leading to considerable gains in neutron flux. We have carried out Monte-Carlo simulations for a new triple-axis spectrometer that will be built at the end position of the conventional cold guide NL-1 in the neutron guide hall of the research reactor FRM-II in Munich, Germany. Our results demonstrate that an elliptic guide section at the end of a conventional guide can be used to at least maintain the total neutron flux onto the sample, while significantly improving the energy resolution of the spectrometer. The simulation further allows detailed insight how the defining parameters of an elliptic guide have to be chosen to obtain optimum results. Finally, we show that the elliptic guide limits losses in the neutron flux that generally arise at the gaps, where the monochromator system of the upstream instrument is situated.
The dynamical fingerprint of core scouring in massive elliptical galaxies
International Nuclear Information System (INIS)
Thomas, J.; Saglia, R. P.; Bender, R.; Erwin, P.; Fabricius, M.
2014-01-01
The most massive elliptical galaxies have low-density centers or cores that differ dramatically from the high-density centers of less massive ellipticals and bulges of disk galaxies. These cores have been interpreted as the result of mergers of supermassive black hole binaries, which depopulate galaxy centers by gravitationally slingshotting central stars toward large radii. Such binaries naturally form in mergers of luminous galaxies. Here, we analyze the population of central stellar orbits in 11 massive elliptical galaxies that we observed with the integral field spectrograph SINFONI at the European Southern Observatory Very Large Telescope. Our dynamical analysis is orbit-based and includes the effects of a central black hole, the mass distribution of the stars, and a dark matter halo. We show that the use of integral field kinematics and the inclusion of dark matter is important to conclude on the distribution of stellar orbits in galaxy centers. Six of our galaxies are core galaxies. In these six galaxies, but not in the galaxies without cores, we detect a coherent lack of stars on radial orbits in the core region and a uniform excess of radial orbits outside of it: when scaled by the core radius r b , the radial profiles of the classical anisotropy parameter β(r) are nearly identical in core galaxies. Moreover, they quantitatively match the predictions of black hole binary simulations, providing the first convincing dynamical evidence for core scouring in the most massive elliptical galaxies.
Monotone difference schemes for weakly coupled elliptic and parabolic systems
P. Matus (Piotr); F.J. Gaspar Lorenz (Franscisco); L. M. Hieu (Le Minh); V.T.K. Tuyen (Vo Thi Kim)
2017-01-01
textabstractThe present paper is devoted to the development of the theory of monotone difference schemes, approximating the so-called weakly coupled system of linear elliptic and quasilinear parabolic equations. Similarly to the scalar case, the canonical form of the vector-difference schemes is
Global weighted estimates for second-order nondivergence elliptic ...
Indian Academy of Sciences (India)
Fengping Yao
2018-03-21
Mar 21, 2018 ... One of the key a priori estimates in the theory of second-order elliptic .... It is well known that the maximal functions satisfy strong p–p .... Here we prove the following auxiliary result, which will be a crucial ingredient in the proof.
Acoustic scattering by multiple elliptical cylinders using collocation multipole method
International Nuclear Information System (INIS)
Lee, Wei-Ming
2012-01-01
This paper presents the collocation multipole method for the acoustic scattering induced by multiple elliptical cylinders subjected to an incident plane sound wave. To satisfy the Helmholtz equation in the elliptical coordinate system, the scattered acoustic field is formulated in terms of angular and radial Mathieu functions which also satisfy the radiation condition at infinity. The sound-soft or sound-hard boundary condition is satisfied by uniformly collocating points on the boundaries. For the sound-hard or Neumann conditions, the normal derivative of the acoustic pressure is determined by using the appropriate directional derivative without requiring the addition theorem of Mathieu functions. By truncating the multipole expansion, a finite linear algebraic system is derived and the scattered field can then be determined according to the given incident acoustic wave. Once the total field is calculated as the sum of the incident field and the scattered field, the near field acoustic pressure along the scatterers and the far field scattering pattern can be determined. For the acoustic scattering of one elliptical cylinder, the proposed results match well with the analytical solutions. The proposed scattered fields induced by two and three elliptical–cylindrical scatterers are critically compared with those provided by the boundary element method to validate the present method. Finally, the effects of the convexity of an elliptical scatterer, the separation between scatterers and the incident wave number and angle on the acoustic scattering are investigated.
Dynamic stress intensity factors for a longitudinal semi-elliptical ...
African Journals Online (AJOL)
elliptical crack in a thick-walled cylinder subjected to transient dynamic stresses. First, the problem of dynamic elasticity in a thick-walled cylinder is solved analytically using the finite Hankel transform. Transient pressure is assumed to act on ...
Three dimensional alignment of molecules using elliptically polarized laser fields
DEFF Research Database (Denmark)
Larsen, J.J.; Bjerre, N.; Hald, K.
2000-01-01
We demonstrate, theoretically and experimentally, that an intense, elliptically polarized, nonresonant laser field can simultaneously force all three axes of a molecule to align along given axes fixed in space, thus inhibiting the free rotation in all three Euler angles. Theoretically, the effect...
Generation of Elliptically Polarized Terahertz Waves from Antiferromagnetic Sandwiched Structure.
Zhou, Sheng; Zhang, Qiang; Fu, Shu-Fang; Wang, Xuan-Zhang; Song, Yu-Ling; Wang, Xiang-Guang; Qu, Xiu-Rong
2018-04-01
The generation of elliptically polarized electromagnetic wave of an antiferromagnetic (AF)/dielectric sandwiched structure in the terahertz range is studied. The frequency and external magnetic field can change the AF optical response, resulting in the generation of elliptical polarization. An especially useful geometry with high levels of the generation of elliptical polarization is found in the case where an incident electromagnetic wave perpendicularly illuminates the sandwiched structure, the AF anisotropy axis is vertical to the wave-vector and the external magnetic field is pointed along the wave-vector. In numerical calculations, the AF layer is FeF2 and the dielectric layers are ZnF2. Although the effect originates from the AF layer, it can be also influenced by the sandwiched structure. We found that the ZnF2/FeF2/ZnF2 structure possesses optimal rotation of the principal axis and ellipticity, which can reach up to about thrice that of a single FeF2 layer.
Eliminating line of sight in elliptic guides using gravitational curving
DEFF Research Database (Denmark)
Klenø, Kaspar H.; Willendrup, Peter Kjær; Bergbäck Knudsen, Erik
2011-01-01
result in a breakdown of the geometrical focusing mechanism inherent to the elliptical shape, resulting in unwanted reflections and loss of transmission. We present a new and yet untried idea by curving a guide in such a way as to follow the ballistic curve of a neutron in the gravitational field, while...
Refined functional relations for the elliptic SOS model
Energy Technology Data Exchange (ETDEWEB)
Galleas, W., E-mail: w.galleas@uu.nl [ARC Centre of Excellence for the Mathematics and Statistics of Complex Systems, University of Melbourne, VIC 3010 (Australia)
2013-02-21
In this work we refine the method presented in Galleas (2012) [1] and obtain a novel kind of functional equation determining the partition function of the elliptic SOS model with domain wall boundaries. This functional relation arises from the dynamical Yang-Baxter relation and its solution is given in terms of multiple contour integrals.
Influence of Some Variable Parameters on Horizontal Elliptic Micro ...
African Journals Online (AJOL)
The study investigates the laminar flow and heat transfer characteristics in elliptic micro-channels of varying axis ratios and with internal longitudinal fins, operating in a region that is hydrodynamically and thermally fully developed; purposely to determine the effects of some salient fluid and geometry parameters such as ...
Recombination plus fragmentation model at RHIC: elliptic flow
Energy Technology Data Exchange (ETDEWEB)
Nonaka, C [Department of Physics, Duke University, Durham, NC 27708 (United States); Fries, R J [School of Physics and Astronomy, University of Minnesota, Minneapolis, MN 55455 (United States); Mueller, B [Department of Physics, Duke University, Durham, NC 27708 (United States); Bass, S A [Department of Physics, Duke University, Durham, NC 27708 (United States); RIKEN BNL Research Center, Brookhaven National Laboratory, Upton, NY 11973 (United States); Asakawa, M [Department of Physics, Osaka University, Toyonaka 560-0043 (Japan)
2005-04-01
We discuss hadron production in relativistic heavy-ion collisions in the framework of the recombination and fragmentation model. We propose elliptic flow as a useful tool for exploring final interactions of resonances, the hadron structure of exotic particles and the phase structure of the reaction.
Tearing mode stability of tokamak plasmas with elliptical cross section
International Nuclear Information System (INIS)
Carreras, B.A.; Holmes, J.A.; Hicks, H.R.; Lynch, V.E.
1981-02-01
The effect of the ellipticity of the plasma cross section on tearing mode stability is investigated. The induced coupling between modes is shown to be destabilizing; however, the modification of the equilibrium tends to stabilize the tearing modes. The net effect depends on the manner in which the equilibrium is modified as the plasma cross-section shape is changed
The shortage of long-period comets in elliptical orbits
International Nuclear Information System (INIS)
Everhart, E.
1979-01-01
Based on the number of 'new' comets seen on near-parabolic orbits, one can predict the number of comets that should be found on definitely elliptical orbits on their subsequent returns. The author shows that about three out of four of these returning comets are not observed. (Auth.)
On nonlocal semi linear elliptic problem with an indefinite term
International Nuclear Information System (INIS)
Yechoui, Akila
2007-08-01
The aim of this paper is to investigate the existence of solutions of a nonlocal semi linear elliptic equation with an indefinite term. The monotone method, the method of upper and lower solutions and the classical maximum principle are used to obtain our results. (author)
Existence of positive solutions to semilinear elliptic problems with ...
Indian Academy of Sciences (India)
57
In mathematical modeling, elliptic partial differential equations are used together with boundary conditions specifying the .... Note that the trace map X ↩→ Lq(∂Ω) is compact for q ∈ [1, 2∗) (see, e.g., [4, ..... [2] Ambrosetti A and Rabinowitz P H, Dual variational methods in critical point theory and applications, J. Functional ...
Flexible hardware design for RSA and Elliptic Curve Cryptosystems
Batina, L.; Bruin - Muurling, G.; Örs, S.B.; Okamoto, T.
2004-01-01
This paper presents a scalable hardware implementation of both commonly used public key cryptosystems, RSA and Elliptic Curve Cryptosystem (ECC) on the same platform. The introduced hardware accelerator features a design which can be varied from very small (less than 20 Kgates) targeting wireless
Diffraction and Dirchlet problem for parameter-elliptic convolution ...
African Journals Online (AJOL)
In this paper we evaluate the difference between the inverse operators of a Dirichlet problem and of a diffraction problem for parameter-elliptic convolution operators with constant symbols. We prove that the inverse operator of a Dirichlet problem can be obtained as a limit case of such a diffraction problem. Quaestiones ...
Fast elliptic-curve cryptography on the Cell Broadband Engine
Costigan, N.; Schwabe, P.; Preneel, B.
2009-01-01
This paper is the first to investigate the power of the Cell Broadband Engine for state-of-the-art public-key cryptography. We present a high-speed implementation of elliptic-curve Diffie-Hellman (ECDH) key exchange for this processor, which needs 697080 cycles on one Synergistic Processor Unit for
Nonconforming h-p spectral element methods for elliptic problems
Indian Academy of Sciences (India)
In [6,7,13,14] h-p spectral element methods for solving elliptic boundary value problems on polygonal ... Let M denote the number of corner layers and W denote the number of degrees of .... β is given by Theorem 2.2 of [3] which can be stated.
hp Spectral element methods for three dimensional elliptic problems
Indian Academy of Sciences (India)
elliptic boundary value problems on non-smooth domains in R3. For Dirichlet problems, ... of variable degree bounded by W. Let N denote the number of layers in the geomet- ric mesh ... We prove a stability theorem for mixed problems when the spectral element functions vanish ..... Applying Theorem 3.1,. ∫ r l. |Mu|2dx −.
Implementing parallel elliptic solver on a Beowulf cluster
Directory of Open Access Journals (Sweden)
Marcin Paprzycki
1999-12-01
Full Text Available In a recent paper cite{zara} a parallel direct solver for the linear systems arising from elliptic partial differential equations has been proposed. The aim of this note is to present the initial evaluation of the performance characteristics of this algorithm on Beowulf-type cluster. In this context the performance of PVM and MPI based implementations is compared.
Radial solutions to semilinear elliptic equations via linearized operators
Directory of Open Access Journals (Sweden)
Phuong Le
2017-04-01
Full Text Available Let $u$ be a classical solution of semilinear elliptic equations in a ball or an annulus in $\\mathbb{R}^N$ with zero Dirichlet boundary condition where the nonlinearity has a convex first derivative. In this note, we prove that if the $N$-th eigenvalue of the linearized operator at $u$ is positive, then $u$ must be radially symmetric.
Nonlinear anisotropic elliptic equations with variable exponents and degenerate coercivity
Directory of Open Access Journals (Sweden)
Hocine Ayadi
2018-02-01
Full Text Available In this article, we prove the existence and the regularity of distributional solutions for a class of nonlinear anisotropic elliptic equations with $p_i(x$ growth conditions, degenerate coercivity and $L^{m(\\cdot}$ data, with $m(\\cdot$ being small, in appropriate Lebesgue-Sobolev spaces with variable exponents. The obtained results extend some existing ones [8,10].
The use of MACSYMA for solving elliptic boundary value problems
Thejll, Peter; Gilbert, Robert P.
1990-01-01
A boundary method is presented for the solution of elliptic boundary value problems. An approach based on the use of complete systems of solutions is emphasized. The discussion is limited to the Dirichlet problem, even though the present method can possibly be adapted to treat other boundary value problems.
An elliptic analogue of the Franklin-Schneider theorem
Bijlsma, A.
1980-01-01
Let p be a Weierstrass elliptic function with algebraic invariants g2 and g3. Let a and b be complex numbers such that a and b are not among the poles of p. A lower bound is given for the simultaneous approximation of p(a), b and p(ab) by algebraic numbers, expressed in their heights and degrees. By
Transfer coefficients in elliptical tubes and plate fin heat exchangers
International Nuclear Information System (INIS)
Saboya, S.M.
1979-09-01
Mean transfer coefficients in elliptical tubes and plate fin heat exchangers were determined by application of heat and mass transfer analogy in conjunction with the naphthalene sublimation technique. The transfer coefficients are presented in a dimensionless form as functions of the Reynolds number. By using the least squares method analytical expressions for the transfer coefficients were determined with low scattering. (E.G.) [pt
Refined functional relations for the elliptic SOS model
International Nuclear Information System (INIS)
Galleas, W.
2013-01-01
In this work we refine the method presented in Galleas (2012) [1] and obtain a novel kind of functional equation determining the partition function of the elliptic SOS model with domain wall boundaries. This functional relation arises from the dynamical Yang–Baxter relation and its solution is given in terms of multiple contour integrals.
Elliptic hypergeometric integrals and 't Hooft anomaly matching conditions
International Nuclear Information System (INIS)
Spiridonov, V.P.; Vartanov, G.S.
2012-03-01
Elliptic hypergeometric integrals describe superconformal indices of 4d supersymmetric field theories. We show that all 't Hooft anomaly matching conditions for Seiberg dual theories can be derived from SL(3, Z)-modular transformation properties of the kernels of dual indices.
Multiple solutions for inhomogeneous nonlinear elliptic problems arising in astrophyiscs
Directory of Open Access Journals (Sweden)
Marco Calahorrano
2004-04-01
Full Text Available Using variational methods we prove the existence and multiplicity of solutions for some nonlinear inhomogeneous elliptic problems on a bounded domain in $mathbb{R}^n$, with $ngeq 2$ and a smooth boundary, and when the domain is $mathbb{R}_+^n$
A Primer on Elliptic Functions with Applications in Classical Mechanics
Brizard, Alain J.
2009-01-01
The Jacobi and Weierstrass elliptic functions used to be part of the standard mathematical arsenal of physics students. They appear as solutions of many important problems in classical mechanics: the motion of a planar pendulum (Jacobi), the motion of a force-free asymmetric top (Jacobi), the motion of a spherical pendulum (Weierstrass) and the…
On the elliptic flow for nearly symmetric collisions and nuclear ...
Indian Academy of Sciences (India)
10Ne20+13Al27, 18Ar40+21Sc45, 30Zn64+28Ni58, 36Kr86+41Nb93) using the quantum molecular dynamics (QMD) model. General features of elliptic ﬂow are investigated with the help of theoretical simulations. The simulations are ...
Stellar populations as a function of radius in giant elliptical galaxies
Peletier, Reynier F.; Valentijn, Edwin A.
Accurate surface photometry has been obtained in J and K for 12 giant elliptical galaxies. Ellipses have been fitted, to obtain luminosity, ellipticity, and major axis position angle profiles. The results have been combined with visual profiles from CCD observations. It is found that elliptical
International Nuclear Information System (INIS)
Franke, H.G. III.
1987-12-01
A fixed-target test beam facility has been designed and constructed at the Meson Test (MT) site to support studies of components of the Collider Detector at Fermi National Accelerator Laboratory (CDF). I assisted in the design and constuction of the test beam facility gas delivery system, and I conducted the initial studies to document the ability of the MT beamline to meet the needs of CDF. Analysis of the preliminary performance data on MT beamline components and beam tunes at required particle energies is presented. Preliminary studies show that the MT beamline has the necessary flexibility to satisfy most CDF requirements now
Structure and stellar content of dwarf elliptical galaxies
International Nuclear Information System (INIS)
Caldwell, N.
1983-01-01
A small number of low-luminosity elliptical galaxies in the Virgo cluster and around other prominent galaxies have been studied using photoelectric and photographic techniques. The color-magnitude relation for ellipticals now extends from M/sub v/ = -23 to -15, and is linear over that range with a slope of 0.10 in U-V per visual magnitude. Galaxies which are known to contain a large number of young stars (''extreme cases'') are from 0.10 to 0.20 mag bluer in U-V than the lower envelope of the dwarf elliptical color-magnitude relation. This difference can be accounted for if the dwarf elliptical galaxies are young, but do not contain the massive blue stars that probably exist in the young populations of the extreme cases. Surface brightness profiles of the dwarfs have revealed some interesting distinctions between themselves and the brighter E's. In general, their intensity profiles are shallower than those of the bright E's, meaning they are of lower mean density. These mean densities are also a function of the total luminosity. Unlike the bright E's, the surface brightnesses near the centers are also a strong function of the total luminosity. The presence of a nucleation, which can be as much as 2 mag brighter than what the outer envelope would predict, does not appear to depend on any other measurable property of the galaxies. The variation in surface brightness profiles at the same total luminosity is suggestive that the low-luminosity dwarfs formed in more than one way. The flattening distribution of the dwarfs is like that of the bright ellipticals, and is also similar to the flattening distribution of field irregular galaxies
Structure and Formation of Elliptical and Spheroidal Galaxies
Kormendy, John; Fisher, David B.; Cornell, Mark E.; Bender, Ralf
2009-05-01
New surface photometry of all known elliptical galaxies in the Virgo cluster is combined with published data to derive composite profiles of brightness, ellipticity, position angle, isophote shape, and color over large radius ranges. These provide enough leverage to show that Sérsic log I vprop r 1/n functions fit the brightness profiles I(r) of nearly all ellipticals remarkably well over large dynamic ranges. Therefore, we can confidently identify departures from these profiles that are diagnostic of galaxy formation. Two kinds of departures are seen at small radii. All 10 of our ellipticals with total absolute magnitudes MVT 4 uncorrelated with MVT . They also are α-element enhanced, implying short star-formation timescales. And their stellar populations have a variety of ages but mostly are very old. Extra light ellipticals generally rotate rapidly, are more isotropic than core Es, and have disky isophotes. We show that they have n sime 3 ± 1 almost uncorrelated with MVT and younger and less α-enhanced stellar populations. These are new clues to galaxy formation. We suggest that extra light ellipticals got their low Sérsic indices by forming in relatively few binary mergers, whereas giant ellipticals have n > 4 because they formed in larger numbers of mergers of more galaxies at once plus later heating during hierarchical clustering. We confirm that core Es contain X-ray-emitting gas whereas extra light Es generally do not. This leads us to suggest why the E-E dichotomy arose. If energy feedback from active galactic nuclei (AGNs) requires a "working surface" of hot gas, then this is present in core galaxies but absent in extra light galaxies. We suggest that AGN energy feedback is a strong function of galaxy mass: it is weak enough in small Es not to prevent merger starbursts but strong enough in giant Es and their progenitors to make dry mergers dry and to protect old stellar populations from late star formation. Finally, we verify that there is a strong
Wagner, U. H.; Alcock, S.; Ludbrook, G.; Wiatryzk, J.; Rau, C.
2012-05-01
I13L is a 250m long hard x-ray beamline (6 keV to 35 keV) currently under construction at the Diamond Light Source. The beamline comprises of two independent experimental endstations: one for imaging in direct space using x-ray microscopy and one for imaging in reciprocal space using coherent diffraction based imaging techniques. To minimise the impact of thermal fluctuations and vibrations onto the beamline performance, we are developing a new generation of ultra-stable beamline instrumentation with highly repeatable adjustment mechanisms using low thermal expansion materials like granite and large piezo-driven flexure stages. For minimising the beam distortion we use very high quality optical components like large ion-beam polished mirrors. In this paper we present the first metrology results on a newly designed mirror system following this design philosophy.
PARALLEL MEASUREMENT AND MODELING OF TRANSPORT IN THE DARHT II BEAMLINE ON ETA II
International Nuclear Information System (INIS)
Chambers, F W; Raymond, B A; Falabella, S; Lee, B S; Richardson, R A; Weir, J T; Davis, H A; Schultze, M E
2005-01-01
To successfully tune the DARHT II transport beamline requires the close coupling of a model of the beam transport and the measurement of the beam observables as the beam conditions and magnet settings are varied. For the ETA II experiment using the DARHT II beamline components this was achieved using the SUICIDE (Simple User Interface Connecting to an Integrated Data Environment) data analysis environment and the FITS (Fully Integrated Transport Simulation) model. The SUICIDE environment has direct access to the experimental beam transport data at acquisition and the FITS predictions of the transport for immediate comparison. The FITS model is coupled into the control system where it can read magnet current settings for real time modeling. We find this integrated coupling is essential for model verification and the successful development of a tuning aid for the efficient convergence on a useable tune. We show the real time comparisons of simulation and experiment and explore the successes and limitations of this close coupled approach
BSRF-3B3 Medium Energy X-ray Beamline and Its Application for XAFS Research
International Nuclear Information System (INIS)
Ma Chenyan; Cui Mingqi; Zhou Kejin; Zhao Yidong; Tian Yulian; Wu Ziyu; Zheng Lei; Zhu Jie; Zhao Jia; Chen Kai; Sun Lijuan
2007-01-01
A new medium X-ray beamline 3B3 covering energy from 1.2 keV up to 6.0 keV was built at Beijing Synchrotron Radiation Facility (BSRF) in 2005. With perfect capability of high monochromaticity, good focus and low harmonics, it could be applied to study metrology, optic component characteristics and medium X-ray absorption spectroscopy (XAS). A simple XAFS apparatus has been set up and some measurements such as S, P, Cl, Ca, Al, Mg K-edge X-ray absorption fine structure (XAFS) in their compounds have also been carried out. The results show that it is feasible to do XAFS research at 3B3 beamline under present condition. The fabrication of a more delicate medium XAFS spectrometer is underway including transmission, fluorescence and electronic yield modes
Review of scientific cases and beamline technology for x-ray free-electron laser
International Nuclear Information System (INIS)
Yabashi, Makina
2012-01-01
We summarize the scientific cases and beamline technology for utilizing X-ray Free-Electron Lasers (XFELs). The scientific targets are classified into two categories: observation with ultrahigh spatio-temporal resolution and the creation of extreme states. As a representative of the former, the coherent diffraction imaging (CDI) method reveals the structure of non-crystalline samples with a single-shot exposure of the XFEL pulse. For the latter, nonlinear interactions between matter and short-wavelength radiation have been investigated. Combining optical lasers and XFEL with a pump and probe technique is a powerful tool for studying the ultrafast dynamics of materials. The state-of-the-art beamline technologies developed as SACLA are also introduced. (author)