Research of photon beam dose deposition kernel based on Monte Carlo method
International Nuclear Information System (INIS)
Using Monte Carlo program BEAMnrc to simulate Siemens accelerator 6 MV photon beam, using BEAMdp program to analyse the energy spectrum distribution and mean energy from phase space data of different field sizes, then building beam source, energy spectrum and mono-energy source, to use DOSXYZnrc program to calculate the dose deposition kernels at dmax in standard water phantom with different beam sources and make comparison with different dose deposition kernels. The results show that the dose difference using energy spectrum source is small, the maximum percentage dose discrepancy is 1.47%, but it is large using mono-energy source, which is 6.28%. The maximum dose difference for the kernels derived from energy spectrum source and mono-energy source of the same field is larger than 9%, up to 13.2%. Thus, dose deposition has dependence on photon energy, it can lead to larger errors only using mono-energy source because of the beam spectrum distribution of accelerator. A good method to calculate dose more accurately is to use deposition kernel of energy spectrum source. (authors)
International Nuclear Information System (INIS)
Beta-emitters have proved to be appropriate for radioimmunotherapy. The dosimetric characterization of each radionuclide has to be carefully investigated. One usual and practical dosimetric approach is the calculation of dose distribution from a unit point source emitting particles according to any radionuclide of interest, which is known as dose point kernel. Absorbed dose distributions are due to primary and radiation scattering contributions. This work presented a method capable of performing dose distributions for nuclear medicine dosimetry by means of Monte Carlo methods. Dedicated subroutines have been developed in order to separately compute primary and scattering contributions to the total absorbed dose, performing particle transport up to 1 keV or least. Preliminarily, the suitability of the calculation method has been satisfactory, being tested for monoenergetic sources, and it was further applied to the characterization of different beta-minus radionuclides of nuclear medicine interests for radioimmunotherapy. (author)
International Nuclear Information System (INIS)
Monte Carlo (MC) simulation has been commonly used in the dose evaluation of radiation accidents and for medical purposes. The accuracy of simulated results is affected by the particle-tracking algorithm, cross-sectional database, random number generator and statistical error. The differences among MC simulation software packages must be validated. This study simulated the dose point kernel (DPK) and the cellular S-values of monoenergetic electrons ranging from 0.01 to 2 MeV and the radionuclides of 90Y, 177Lu and 103mmRh, using Fluktuierende Kaskade (FLUKA) and MC N-Particle Transport Code Version 5 (MCNP5). A 6-μm-radius cell model consisting of the cell surface, cytoplasm and cell nucleus was constructed for cellular S-value calculation. The mean absolute percentage errors (MAPEs) of the scaled DPKs, simulated using FLUKA and MCNP5, were 7.92, 9.64, 4.62, 3.71 and 3.84 % for 0.01, 0.1, 0.5, 1 and 2 MeV, respectively. For the three radionuclides, the MAPEs of the scaled DPKs were within 5 %. The maximum deviations of S(N<-N), S(N<-Cy) and S(N<-CS) for the electron energy larger than 10 keV were 6.63, 6.77 and 5.24 %, respectively. The deviations for the self-absorbed S-values and cross-dose S-values of the three radionuclides were within 4 %. On the basis of the results of this study, it was concluded that the simulation results are consistent between FLUKA and MCNP5. However, there is a minor inconsistency for low energy range. The DPK and the cellular S-value should be used as the quality assurance tools before the MC simulation results are adopted as the gold standard. (authors)
Vojtyla, P
2005-01-01
The radiological impact of emissions of radioactive substances from accelerator facilities is characterized by a dominant contribution of the external exposure from short-lived radionuclides in the plume. Ventilation outlets of accelerator facilities are often at low emission heights and receptors reside very close to stacks. Simplified exposure models are not appropriate and integration of the dose kernel over the radioactive plume is required. By using Monte Carlo integration with certain biasing, the integrand can be simplified substantially and an optimum spatial resolution can be achieved. Moreover, long-term releases can be modeled by sampling real weather situations. The mathematical formulation does not depend on any particular atmospheric dispersion model and the applicable code parts can be designed separately, which is another advantage. The obtained results agree within ±10% with results calculated for the semi-infinite cloud model by using detailed particle transport codes and human phantoms.
Energy Technology Data Exchange (ETDEWEB)
Valente, Mauro [CONICET - Consejo Nacional de Investigaciones Cientificas y Tecnicas de La Republica Argentina (Conicet), Buenos Aires, AR (Brazil); Botta, Francesca; Pedroli, Guido [European Institute of Oncology, Milan (Italy). Medical Physics Department; Perez, Pedro, E-mail: valente@famaf.unc.edu.ar [Universidad Nacional de Cordoba, Cordoba (Argentina). Fac. de Matematica, Astronomia y Fisica (FaMAF)
2012-07-01
Beta-emitters have proved to be appropriate for radioimmunotherapy. The dosimetric characterization of each radionuclide has to be carefully investigated. One usual and practical dosimetric approach is the calculation of dose distribution from a unit point source emitting particles according to any radionuclide of interest, which is known as dose point kernel. Absorbed dose distributions are due to primary and radiation scattering contributions. This work presented a method capable of performing dose distributions for nuclear medicine dosimetry by means of Monte Carlo methods. Dedicated subroutines have been developed in order to separately compute primary and scattering contributions to the total absorbed dose, performing particle transport up to 1 keV or least. Preliminarily, the suitability of the calculation method has been satisfactory, being tested for monoenergetic sources, and it was further applied to the characterization of different beta-minus radionuclides of nuclear medicine interests for radioimmunotherapy. (author)
International Nuclear Information System (INIS)
Full text of publication follows. Aim: the aim of this study was to perform a critical comparison of 3 dosimetric approaches in Molecular Radiotherapy: phantom based dosimetry, Dose Voxel Kernels (DVKs) and full Monte Carlo (MC) dosimetry. The objective was to establish the impact of the absorbed dose calculation algorithm on the final result. Materials and Methods: we calculated the absorbed dose to various organs in six healthy volunteers injected with a novel 18F-labelled PET radiotracer from GE Healthcare. Each patient underwent from 8 to 10 whole body 3D PET/CT scans. The first 8 scans were acquired dynamically in order to limit co-registration issues. Eleven organs were segmented on the first PET/CT scan by a physician. We analysed this dataset using the OLINDA/EXM software taking into account actual patient's organ masses; the commercial software Stratos by Philips implementing a DVK approach; and performing full MC dosimetry on the basis of a custom application developed with Gate. The calculations performed with these three techniques were based on the cumulated activities calculated at the voxel level by Stratos. Results: all the absorbed doses calculated with Gate were higher than those calculated with OLINDA. The average ratios between the Gate absorbed dose and OLINDA's was 1.38±0.34 σ (from 0.93 to 2.23) considering all patients. The discrepancy was particularly high for the thyroid, with an average Gate/OLINDA ratio of 1.97±0.83 σ for the 6 patients. The lower absorbed doses in OLINDA may be explained considering the inter-organ distances in the MIRD phantom. These are in general overestimated, leading to lower absorbed doses in target organs. The differences between Stratos and Gate resulted to be the highest. The average ratios between Gate and Stratos absorbed doses were 2.51±1.21 σ (from 1.09 to 6.06). The highest differences were found for lungs (average ratio 4.76±2.13 σ), as expected, since Stratos considers unit density
Mairani, A; Valente, M; Battistoni, G; Botta, F; Pedroli, G; Ferrari, A; Cremonesi, M; Di Dia, A; Ferrari, M; Fasso, A
2011-01-01
Purpose: The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, FLUKA Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, FLUKA has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. Methods: FLUKA DPKS have been calculated in both water and compact bone for monoenergetic electrons (10-3 MeV) and for beta emitting isotopes commonly used for therapy ((89)Sr, (90)Y, (131)I, (153)Sm, (177)Lu, (186)Re, and (188)Re). Point isotropic...
Dose point kernels for beta-emitting radioisotopes
International Nuclear Information System (INIS)
Knowledge of the dose point kernel corresponding to a specific radionuclide is required to calculate the spatial dose distribution produced in a homogeneous medium by a distributed source. Dose point kernels for commonly used radionuclides have been calculated previously using as a basis monoenergetic dose point kernels derived by numerical integration of a model transport equation. The treatment neglects fluctuations in energy deposition, an effect which has been later incorporated in dose point kernels calculated using Monte Carlo methods. This work describes new calculations of dose point kernels using the Monte Carlo results as a basis. An analytic representation of the monoenergetic dose point kernels has been developed. This provides a convenient method both for calculating the dose point kernel associated with a given beta spectrum and for incorporating the effect of internal conversion. An algebraic expression for allowed beta spectra has been accomplished through an extension of the Bethe-Bacher approximation, and tested against the exact expression. Simplified expression for first-forbidden shape factors have also been developed. A comparison of the calculated dose point kernel for 32P with experimental data indicates good agreement with a significant improvement over the earlier results in this respect. An analytic representation of the dose point kernel associated with the spectrum of a single beta group has been formulated. 9 references, 16 figures, 3 tables
International Nuclear Information System (INIS)
Modeling the radio-induced effects in biological medium still requires accurate physics models to describe the interactions induced by all the charged particles present in the irradiated medium in detail. These interactions include inelastic as well as elastic processes. To check the accuracy of the very low energy models recently implemented into the GEANT4 toolkit for modeling the electron slowing-down in liquid water, the simulation of electron dose point kernels remains the preferential test. In this context, we here report normalized radial dose profiles, for mono-energetic point sources, computed in liquid water by using the very low energy “GEANT4-DNA” physics processes available in the GEANT4 toolkit. In the present study, we report an extensive intra-comparison of profiles obtained by a large selection of existing and well-documented Monte-Carlo codes, namely, EGSnrc, PENELOPE, CPA100, FLUKA and MCNPX. - Highlights: ► Normalized radial dose profiles are reported for mono-energetic electron sources. ► The low-energy “GEANT4-DNA” physics package is used for the calculations. ► A comparison with a large number of electron track-structure codes is proposed. ► Evident discrepancies in terms of shape and magnitude are reported. ► Accurate dose profiles have been provided by the GEANT4-DNA code
Energy Technology Data Exchange (ETDEWEB)
Dixon, Robert L.; Boone, John M. [Department of Radiology, Wake Forest University School of Medicine, Winston-Salem, North Carolina 27157 (United States); Departments of Radiology and Biomedical Engineering, University of California Davis, Sacramento, California 95817 (United States)
2011-07-15
Purpose: Knowledge of the complete axial dose profile f(z), including its long scatter tails, provides the most complete (and flexible) description of the accumulated dose in CT scanning. The CTDI paradigm (including CTDI{sub vol}) requires shift-invariance along z (identical dose profiles spaced at equal intervals), and is therefore inapplicable to many of the new and complex shift-variant scan protocols, e.g., high dose perfusion studies using variable (or zero) pitch. In this work, a convolution-based beam model developed by Dixon et al.[Med. Phys. 32, 3712-3728, (2005)] updated with a scatter LSF kernel (or DSF) derived from a Monte Carlo simulation by Boone [Med. Phys. 36, 4547-4554 (2009)] is used to create an analytical equation for the axial dose profile f(z) in a cylindrical phantom. Using f(z), equations are derived which provide the analytical description of conventional (axial and helical) dose, demonstrating its physical underpinnings; and likewise for the peak axial dose f(0) appropriate to stationary phantom cone beam CT, (SCBCT). The methodology can also be applied to dose calculations in shift-variant scan protocols. This paper is an extension of our recent work Dixon and Boone [Med. Phys. 37, 2703-2718 (2010)], which dealt only with the properties of the peak dose f(0), its relationship to CTDI, and its appropriateness to SCBCT. Methods: The experimental beam profile data f(z) of Mori et al.[Med. Phys. 32, 1061-1069 (2005)] from a 256 channel prototype cone beam scanner for beam widths (apertures) ranging from a = 28 to 138 mm are used to corroborate the theoretical axial profiles in a 32 cm PMMA body phantom. Results: The theoretical functions f(z) closely-matched the central axis experimental profile data{sup 11} for all apertures (a = 28 -138 mm). Integration of f(z) likewise yields analytical equations for all the (CTDI-based) dosimetric quantities of conventional CT (including CTDI{sub L} itself) in addition to the peak dose f(0) relevant to
International Nuclear Information System (INIS)
Intracavity instillation of β-emitting colloid pharmaceuticals is a common technique used to treat cystic brain tumors. Most of the dosimetric calculations that have been reported in the literature for this problem are based on empirical formulas derived by Loevinger. Concentration of P-32 radiolabeled solution for the delivery of a prescribed dose (200 Gy to the cyst wall) has been published previously using this formalism in what we refer to as a standard nomogram. The calculations using the Loevinger formulas for calculating the P-32 activity necessary to achieve 200 Gy at the cyst wall is re-evaluated and compared to numerically computed results based on full Monte Carlo simulations (EGSnrc) and the dose-point-kernel (DPK) integration method. For cyst diameters greater than 1 cm, the new calculations agree well with previously published results (the standard nomogram) to within a few percents. However, for cyst diameters of less than 1 cm, it is shown that the standard nomogram results underestimate the therapeutic activity by a factor of ∼3 for very small diameters (∼0.2 cm). New tables based on our calculations are presented and the sources of discrepancies are identified. It is concluded that the new set of data based on our calculations should replace the standard nomogram to administer accurately the target dose to the cyst wall for the smaller diameter cysts (<1 cm)
Beta-ray dose assessment from skin contamination using a point kernel method
International Nuclear Information System (INIS)
In this study, a point kernel method to calculate beta-ray dose rate from skin contamination was introduced. The beta-ray doses rates were computed by performing numerical integration of the radial dose distribution around an isotropic point source of monoenergetic electrons called as point kernel. In this study, in-house code, based on MATLAB version 7.0.4 was developed to perform a numerical integration. The code generated dose distributions for beta-ray emitters from interpolated point kernel, and beta-ray dose rates from skin contamination were calculated by numerical integration. Generated dose distributions for selected beta-ray emitters agreed with those calculated by Cross et al within 20%, except at a longer distance where there are differences up to more than 100%. For a point source, calculated beta-ray doses were agreed well with those derived from Monte Carlo simulation. For a disk source, the differences were up to 17% at a deep region. Point kernel method underestimated beta-ray doses than Monte Carlo simulation. The code will be improved to deal with a three-dimensional source, shielding by cover material, air gap and contribution of photon to skin dose. For the sake of user's convenience, the code will be equipped with graphic user interface. (author)
Monte Carlo dose computation for IMRT optimization*
Laub, W.; Alber, M.; Birkner, M.; Nüsslin, F.
2000-07-01
A method which combines the accuracy of Monte Carlo dose calculation with a finite size pencil-beam based intensity modulation optimization is presented. The pencil-beam algorithm is employed to compute the fluence element updates for a converging sequence of Monte Carlo dose distributions. The combination is shown to improve results over the pencil-beam based optimization in a lung tumour case and a head and neck case. Inhomogeneity effects like a broader penumbra and dose build-up regions can be compensated for by intensity modulation.
Monte Carlo dose mapping on deforming anatomy
Zhong, Hualiang; Siebers, Jeffrey V.
2009-10-01
This paper proposes a Monte Carlo-based energy and mass congruent mapping (EMCM) method to calculate the dose on deforming anatomy. Different from dose interpolation methods, EMCM separately maps each voxel's deposited energy and mass from a source image to a reference image with a displacement vector field (DVF) generated by deformable image registration (DIR). EMCM was compared with other dose mapping methods: energy-based dose interpolation (EBDI) and trilinear dose interpolation (TDI). These methods were implemented in EGSnrc/DOSXYZnrc, validated using a numerical deformable phantom and compared for clinical CT images. On the numerical phantom with an analytically invertible deformation map, EMCM mapped the dose exactly the same as its analytic solution, while EBDI and TDI had average dose errors of 2.5% and 6.0%. For a lung patient's IMRT treatment plan, EBDI and TDI differed from EMCM by 1.96% and 7.3% in the lung patient's entire dose region, respectively. As a 4D Monte Carlo dose calculation technique, EMCM is accurate and its speed is comparable to 3D Monte Carlo simulation. This method may serve as a valuable tool for accurate dose accumulation as well as for 4D dosimetry QA.
Application of Electron Dose Kernels to account for heterogeneities in voxelized phantoms
International Nuclear Information System (INIS)
In this paper, we present work on the application of the Electron Dose Kernel discrete ordinates method (EDK-SN) to compute doses and account for material heterogeneities using high energy external photon beam irradiations in voxelized human phantoms. EDKs are pre-computed using photon pencil 'beamlets' that lead to dose delivery in tissue using highly converged Monte Carlo. Coupling the EDKs to accumulate dose scaled by integral photon fluences computed using SN methods in dose driving voxels (DDVs) allows for the full charged particle physics computed dose to be accumulated throughout the voxelized phantom, and is the basis of the EDK-SN method, which is fully parallelized. For material heterogeneities, a density scaling correction factor is required to yield good agreement. In a fully voxelized phantom, all doses were in agreement with those determined by independent Monte Carlo computations. We are continuing to expand upon the development of this robust approach for rapid and accurate determination of whole body and out of field organ doses due to high energy x-ray beams. (authors)
Monte Carlo dose distributions for radiosurgery
International Nuclear Information System (INIS)
The precision of Radiosurgery Treatment planning systems is limited by the approximations of their algorithms and by their dosimetrical input data. This fact is especially important in small fields. However, the Monte Carlo methods is an accurate alternative as it considers every aspect of particle transport. In this work an acoustic neurinoma is studied by comparing the dose distribution of both a planning system and Monte Carlo. Relative shifts have been measured and furthermore, Dose-Volume Histograms have been calculated for target and adjacent organs at risk. (orig.)
Suitability of point kernel dose calculation techniques in brachytherapy treatment planning
Directory of Open Access Journals (Sweden)
Lakshminarayanan Thilagam
2010-01-01
Full Text Available Brachytherapy treatment planning system (TPS is necessary to estimate the dose to target volume and organ at risk (OAR. TPS is always recommended to account for the effect of tissue, applicator and shielding material heterogeneities exist in applicators. However, most brachytherapy TPS software packages estimate the absorbed dose at a point, taking care of only the contributions of individual sources and the source distribution, neglecting the dose perturbations arising from the applicator design and construction. There are some degrees of uncertainties in dose rate estimations under realistic clinical conditions. In this regard, an attempt is made to explore the suitability of point kernels for brachytherapy dose rate calculations and develop new interactive brachytherapy package, named as BrachyTPS, to suit the clinical conditions. BrachyTPS is an interactive point kernel code package developed to perform independent dose rate calculations by taking into account the effect of these heterogeneities, using two regions build up factors, proposed by Kalos. The primary aim of this study is to validate the developed point kernel code package integrated with treatment planning computational systems against the Monte Carlo (MC results. In the present work, three brachytherapy applicators commonly used in the treatment of uterine cervical carcinoma, namely (i Board of Radiation Isotope and Technology (BRIT low dose rate (LDR applicator and (ii Fletcher Green type LDR applicator (iii Fletcher Williamson high dose rate (HDR applicator, are studied to test the accuracy of the software. Dose rates computed using the developed code are compared with the relevant results of the MC simulations. Further, attempts are also made to study the dose rate distribution around the commercially available shielded vaginal applicator set (Nucletron. The percentage deviations of BrachyTPS computed dose rate values from the MC results are observed to be within plus/minus 5
On optimality of kernels for approximate Bayesian computation using sequential Monte Carlo.
Filippi, Sarah; Barnes, Chris P; Cornebise, Julien; Stumpf, Michael P H
2013-03-01
Approximate Bayesian computation (ABC) has gained popularity over the past few years for the analysis of complex models arising in population genetics, epidemiology and system biology. Sequential Monte Carlo (SMC) approaches have become work-horses in ABC. Here we discuss how to construct the perturbation kernels that are required in ABC SMC approaches, in order to construct a sequence of distributions that start out from a suitably defined prior and converge towards the unknown posterior. We derive optimality criteria for different kernels, which are based on the Kullback-Leibler divergence between a distribution and the distribution of the perturbed particles. We will show that for many complicated posterior distributions, locally adapted kernels tend to show the best performance. We find that the added moderate cost of adapting kernel functions is easily regained in terms of the higher acceptance rate. We demonstrate the computational efficiency gains in a range of toy examples which illustrate some of the challenges faced in real-world applications of ABC, before turning to two demanding parameter inference problems in molecular biology, which highlight the huge increases in efficiency that can be gained from choice of optimal kernels. We conclude with a general discussion of the rational choice of perturbation kernels in ABC SMC settings. PMID:23502346
Optical monitoring of rheumatoid arthritis: Monte Carlo generated reconstruction kernels
Minet, O.; Beuthan, J.; Hielscher, A. H.; Zabarylo, U.
2008-06-01
Optical imaging in biomedicine is governed by the light absorption and scattering interaction on microscopic and macroscopic constituents in the medium. Therefore, light scattering characteristics of human tissue correlate with the stage of some diseases. In the near infrared range the scattering event with the coefficient approximately two orders of magnitude greater than absorption plays a dominant role. When measuring the optical parameters variations were discovered that correlate with the rheumatoid arthritis of a small joint. The potential of an experimental setup for transillumination the finger joint with a laser diode and the pattern of the stray light detection are demonstrated. The scattering caused by skin contains no useful information and it can be removed by a deconvolution technique to enhance the diagnostic value of this non-invasive optical method. Monte Carlo simulations ensure both the construction of the corresponding point spread function and both the theoretical verification of the stray light picture in rather complex geometry.
Generation of a novel phase-space-based cylindrical dose kernel for IMRT optimizationa
Zhong, Hualiang; Chetty, Indrin J.
2012-01-01
Purpose: Improving dose calculation accuracy is crucial in intensity-modulated radiation therapy (IMRT). We have developed a method for generating a phase-space-based dose kernel for IMRT planning of lung cancer patients.
A comparison of Monte Carlo and analytic first scatter dose spread arrays.
McGary, J E; Boyer, A L; Mackie, T R
1999-05-01
We compare first scattered point dose spread arrays generated by Monte Carlo and an analytic method. The analytic method models energy deposition using Klein-Nishina cross sections for Compton scatter and approximations for electron transport. Assumptions in the analytic method are shown to be valid within a region of the point dose spread kernel in which meaningful comparisons can be made. Differences between the models are less than 10% for the forward scatter directions for radii greater than the electron range associated with the first scattered Compton photon. Differences in the backscatter region are discussed and indicate that the analytic model is useful for identifying large errors that might be present in numerically generated first scatter point dose spread arrays. The analytic method is simple and useful for validating first scatter kernels. PMID:10360537
International Nuclear Information System (INIS)
Kernel Density Estimators (KDEs) are a non-parametric density estimation technique that has recently been applied to Monte Carlo radiation transport simulations. Kernel density estimators are an alternative to histogram tallies for obtaining global solutions in Monte Carlo tallies. With KDEs, a single event, either a collision or particle track, can contribute to the score at multiple tally points with the uncertainty at those points being independent of the desired resolution of the solution. Thus, KDEs show potential for obtaining estimates of a global solution with reduced variance when compared to a histogram. Previously, KDEs have been applied to neutronics for one-group reactor physics problems and fixed source shielding applications. However, little work was done to obtain reaction rates using KDEs. This paper introduces a new form of the MFP KDE that is capable of handling general geometries. Furthermore, extending the MFP KDE to 2-D problems in continuous energy introduces inaccuracies to the solution. An ad-hoc solution to these inaccuracies is introduced that produces errors smaller than 4% at material interfaces.
Energy Technology Data Exchange (ETDEWEB)
Burke, TImothy P. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States); Kiedrowski, Brian C. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States); Martin, William R. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States); Brown, Forrest B. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2015-11-19
Kernel Density Estimators (KDEs) are a non-parametric density estimation technique that has recently been applied to Monte Carlo radiation transport simulations. Kernel density estimators are an alternative to histogram tallies for obtaining global solutions in Monte Carlo tallies. With KDEs, a single event, either a collision or particle track, can contribute to the score at multiple tally points with the uncertainty at those points being independent of the desired resolution of the solution. Thus, KDEs show potential for obtaining estimates of a global solution with reduced variance when compared to a histogram. Previously, KDEs have been applied to neutronics for one-group reactor physics problems and fixed source shielding applications. However, little work was done to obtain reaction rates using KDEs. This paper introduces a new form of the MFP KDE that is capable of handling general geometries. Furthermore, extending the MFP KDE to 2-D problems in continuous energy introduces inaccuracies to the solution. An ad-hoc solution to these inaccuracies is introduced that produces errors smaller than 4% at material interfaces.
Fomin, Fedor V.
Preprocessing (data reduction or kernelization) as a strategy of coping with hard problems is universally used in almost every implementation. The history of preprocessing, like applying reduction rules simplifying truth functions, can be traced back to the 1950's [6]. A natural question in this regard is how to measure the quality of preprocessing rules proposed for a specific problem. For a long time the mathematical analysis of polynomial time preprocessing algorithms was neglected. The basic reason for this anomaly was that if we start with an instance I of an NP-hard problem and can show that in polynomial time we can replace this with an equivalent instance I' with |I'| P=NP in classical complexity.
Dose-Volume Histogram Prediction using KernelDensity Estimation
SKARPMAN MUNTER, JOHANNA
2014-01-01
Dose plans developed for stereotactic radiosurgery are assessed by studying so called Dose-Volume Histograms. Since it is hard to compare an individual dose plan with doseplans created for other patients, much experience and knowledge is lost. This thesis therefore investigates a machine learning approach to predicting such Dose-Volume Histograms for a new patient, by learning from previous dose plans.The training set is chosen based on similarity in terms of tumour size. The signed distances...
Monte Carlo dose calculations for dynamic IMRT treatments
International Nuclear Information System (INIS)
Dose calculations for intensity modulated radiation therapy (IMRT) face new challenges due to the complex leaf geometry and time dependent nature of the delivery. A fast method of particle transport through a dynamic multileaf collimator (MLC) geometry that accounts for photon attenuation and first-scattered Compton photon production has been incorporated into an existing Monte Carlo code used for patient dose calculations. Dosimetric agreement between calculation and measurement for two photon energies and MLC types is within experimental error for the sliding window tests. For a patient IMRT field, the Monte Carlo calculations are closer to measured dose than similar superposition or pencil beam calculations. (author)
International Nuclear Information System (INIS)
A new Monte Carlo mesh tally based on a Kernel Density Estimator (KDE) approach using integrated particle tracks is presented. We first derive the KDE integral-track estimator and present a brief overview of its implementation as an alternative to the MCNP fmesh tally. To facilitate a valid quantitative comparison between these two tallies for verification purposes, there are two key issues that must be addressed. The first of these issues involves selecting a good data transfer method to convert the nodal-based KDE results into their cell-averaged equivalents (or vice versa with the cell-averaged MCNP results). The second involves choosing an appropriate resolution of the mesh, since if it is too coarse this can introduce significant errors into the reference MCNP solution. After discussing both of these issues in some detail, we present the results of a convergence analysis that shows the KDE integral-track and MCNP fmesh tallies are indeed capable of producing equivalent results for some simple 3D transport problems. In all cases considered, there was clear convergence from the KDE results to the reference MCNP results as the number of particle histories was increased. (authors)
Diego Azcona, Juan; Barbés, Benigno; Wang, Lilie; Burguete, Javier
2016-01-01
This paper presents a method to obtain the pencil-beam kernels that characterize a megavoltage photon beam generated in a flattening filter free (FFF) linear accelerator (linac) by deconvolution from experimental measurements at different depths. The formalism is applied to perform independent dose calculations in modulated fields. In our previous work a formalism was developed for ideal flat fluences exiting the linac’s head. That framework could not deal with spatially varying energy fluences, so any deviation from the ideal flat fluence was treated as a perturbation. The present work addresses the necessity of implementing an exact analysis where any spatially varying fluence can be used such as those encountered in FFF beams. A major improvement introduced here is to handle the actual fluence in the deconvolution procedure. We studied the uncertainties associated to the kernel derivation with this method. Several Kodak EDR2 radiographic films were irradiated with a 10 MV FFF photon beam from two linacs from different vendors, at the depths of 5, 10, 15, and 20cm in polystyrene (RW3 water-equivalent phantom, PTW Freiburg, Germany). The irradiation field was a 50mm diameter circular field, collimated with a lead block. The 3D kernel for a FFF beam was obtained by deconvolution using the Hankel transform. A correction on the low dose part of the kernel was performed to reproduce accurately the experimental output factors. Error uncertainty in the kernel derivation procedure was estimated to be within 0.2%. Eighteen modulated fields used clinically in different treatment localizations were irradiated at four measurement depths (total of fifty-four film measurements). Comparison through the gamma-index to their corresponding calculated absolute dose distributions showed a number of passing points (3%, 3mm) mostly above 99%. This new procedure is more reliable and robust than the previous one. Its ability to perform accurate independent dose calculations was
Monte Carlo simulation of PET images for injection dose optimization
Czech Academy of Sciences Publication Activity Database
Boldyš, Jiří; Dvořák, Jiří; Bělohlávek, O.; Skopalová, M.
London : Taylor and Francis, 2011 - (Manuel, J.; Tavares, R.; Jorge, N.), s. 1-6 ISBN 978-0-415-68395-1. [VipIMAGE 2011 - third ECCOMAS thematic conference on computational vision and medical image processing. Olhao, Algarve (PT), 12.10.2011-14.10.2011] R&D Projects: GA MŠk(CZ) 1M0572 Institutional research plan: CEZ:AV0Z10750506 Keywords : positron emission tomography * Monte Carlo simulation * biological system modeling * image quality Subject RIV: BD - Theory of Information http://library.utia.cas.cz/separaty/2012/ZOI/boldys-monte carlo simulation of pet images for injection dose optimization.pdf
Monte Carlo dose calculation in dental amalgam phantom
Mohd Zahri Abdul Aziz; Yusoff, A. L.; N D Osman; R. Abdullah; Rabaie, N. A.; M S Salikin
2015-01-01
It has become a great challenge in the modern radiation treatment to ensure the accuracy of treatment delivery in electron beam therapy. Tissue inhomogeneity has become one of the factors for accurate dose calculation, and this requires complex algorithm calculation like Monte Carlo (MC). On the other hand, computed tomography (CT) images used in treatment planning system need to be trustful as they are the input in radiotherapy treatment. However, with the presence of metal amalgam in treatm...
Adjoint Monte Carlo techniques and codes for organ dose calculations
International Nuclear Information System (INIS)
Adjoint Monte Carlo simulations can be effectively used for the estimation of doses in small targets when the sources are extended in large volumes or surfaces. The main features of two computer codes for calculating doses at free points or in organs of an anthropomorphic phantom are described. In the first program (REBEL-3) natural gamma-emitting sources are contained in the walls of a dwelling room; in the second one (POKER-CAMP) the user can specify arbitrary gamma sources with different spatial distributions in the environment: in (or on the surface of) the ground and in the air. 3 figures
Modelling lateral beam quality variations in pencil kernel based photon dose calculations
Nyholm, T.; Olofsson, J.; Ahnesjö, A.; Karlsson, M.
2006-08-01
Standard treatment machines for external radiotherapy are designed to yield flat dose distributions at a representative treatment depth. The common method to reach this goal is to use a flattening filter to decrease the fluence in the centre of the beam. A side effect of this filtering is that the average energy of the beam is generally lower at a distance from the central axis, a phenomenon commonly referred to as off-axis softening. The off-axis softening results in a relative change in beam quality that is almost independent of machine brand and model. Central axis dose calculations using pencil beam kernels show no drastic loss in accuracy when the off-axis beam quality variations are neglected. However, for dose calculated at off-axis positions the effect should be considered, otherwise errors of several per cent can be introduced. This work proposes a method to explicitly include the effect of off-axis softening in pencil kernel based photon dose calculations for arbitrary positions in a radiation field. Variations of pencil kernel values are modelled through a generic relation between half value layer (HVL) thickness and off-axis position for standard treatment machines. The pencil kernel integration for dose calculation is performed through sampling of energy fluence and beam quality in sectors of concentric circles around the calculation point. The method is fully based on generic data and therefore does not require any specific measurements for characterization of the off-axis softening effect, provided that the machine performance is in agreement with the assumed HVL variations. The model is verified versus profile measurements at different depths and through a model self-consistency check, using the dose calculation model to estimate HVL values at off-axis positions. A comparison between calculated and measured profiles at different depths showed a maximum relative error of 4% without explicit modelling of off-axis softening. The maximum relative error
International Nuclear Information System (INIS)
Background. Dosimetry in radionuclide therapy estimates delivered absorbed doses to tumours and ensures that absorbed dose levels to normal organs are below tolerance levels. One procedure is to determine time-activity curves in volumes-of-interests from which the absorbed dose is estimated using SPECT with appropriate corrections for attenuation, scatter and collimator response. From corrected SPECT images the absorbed energy can be calculated by (a) assuming kinetic energy deposited in the same voxel where particles were emitted, (b) convolve with point-dose kernels or (c) use full Monte Carlo (MC) methods. A question arises which dosimetry method is optimal given the limitations in reconstruction- and quantification procedures. Methods. Dosimetry methods (a) and (c) were evaluated by comparing dose-rate volume histograms (DrVHs) from simulated SPECT of 111In, 177Lu, 131I and Bremsstrahlung from 90Y to match true dose rate images. The study used a voxel-based phantom with different tumours in the liver. SPECT reconstruction was made using an iterative OSEM method and MC dosimetry was performed using a charged-particle EGS4 program that also was used to determined true absorbed dose rate distributions for the same phantom geometry but without camera limitations. Results. The DrVHs obtained from SPECT differed from true DrVH mainly due to limited spatial resolution. MC dosimetry had a marginal effect because the SPECT spatial resolution is in the same order as the energy distribution caused by the electron track ranges. For 131I, full MC dosimetry made a difference due to the additional contribution from high-energy photons. SPECT-based DrVHs differ significantly from true DrVHs unless the tumours are considerable larger than the spatial resolution. Conclusion. It is important to understand limitations in quantitative SPECT images and the reasons for apparent heterogeneities since these have an impact on dose-volume histograms. A MC-based dosimetry calculation from
A Monte Carlo dose calculation algorithm for proton therapy
International Nuclear Information System (INIS)
A Monte Carlo (MC) code (VMCpro) for treatment planning in proton beam therapy of cancer is introduced. It is based on ideas of the Voxel Monte Carlo algorithm for photons and electrons and is applicable to human tissue for clinical proton energies. In the present paper the implementation of electromagnetic and nuclear interactions is described. They are modeled by a Class II condensed history algorithm with continuous energy loss, ionization, multiple scattering, range straggling, δ-electron transport, nuclear elastic proton nucleus scattering and inelastic proton nucleus reactions. VMCpro is faster than the general purpose MC codes FLUKA by a factor of 13 and GEANT4 by a factor of 35 for simulations in a phantom with inhomogeneities. For dose calculations in patients the speed improvement is larger, because VMCpro has only a weak dependency on the heterogeneity of the calculation grid. Dose distributions produced with VMCpro are in agreement with GEANT4 results. Integrated or broad beam depth dose curves show maximum deviations not larger than 1% or 0.5 mm in regions with large dose gradients for the examples presented here
Dose calculation of 6 MV Truebeam using Monte Carlo method
International Nuclear Information System (INIS)
The purpose of this work is to simulate 6 MV Varian Truebeam linac dosimeter characteristics using Monte Carlo method and to investigate the availability of phase space file and the accuracy of the simulation. With the phase space file at linac window supplied by Varian to be a source, the patient-dependent part was simulated. Dose distributions in a water phantom with a 10 cm × 10 cm field were calculated and compared with measured data for validation. Evident time reduction was obtained from 4-5 h which a whole simulation cost on the same computer to around 48 minutes. Good agreement between simulations and measurements in water was observed. Dose differences are less than 3% for depth doses in build-up region and also for dose profiles inside the 80% field size, and the effect in penumbra is good. It demonstrate that the simulation using existing phase space file as the EGSnrc source is efficient. Dose differences between calculated data and measured data could meet the requirements for dose calculation. (authors)
A Monte Carlo dose calculation tool for radiotherapy treatment planning
Ma, C.-M.; Li, J. S.; Pawlicki, T.; Jiang, S. B.; Deng, J.; Lee, M. C.; Koumrian, T.; Luxton, M.; Brain, S.
2002-05-01
A Monte Carlo user code, MCDOSE, has been developed for radiotherapy treatment planning (RTP) dose calculations. MCDOSE is designed as a dose calculation module suitable for adaptation to host RTP systems. MCDOSE can be used for both conventional photon/electron beam calculation and intensity modulated radiotherapy (IMRT) treatment planning. MCDOSE uses a multiple-source model to reconstruct the treatment beam phase space. Based on Monte Carlo simulated or measured beam data acquired during commissioning, source-model parameters are adjusted through an automated procedure. Beam modifiers such as jaws, physical and dynamic wedges, compensators, blocks, electron cut-outs and bolus are simulated by MCDOSE together with a 3D rectilinear patient geometry model built from CT data. Dose distributions calculated using MCDOSE agreed well with those calculated by the EGS4/DOSXYZ code using different beam set-ups and beam modifiers. Heterogeneity correction factors for layered-lung or layered-bone phantoms as calculated by both codes were consistent with measured data to within 1%. The effect of energy cut-offs for particle transport was investigated. Variance reduction techniques were implemented in MCDOSE to achieve a speedup factor of 10-30 compared to DOSXYZ.
MCDE: a new Monte Carlo dose engine for IMRT.
Reynaert, N; De Smedt, B; Coghe, M; Paelinck, L; Van Duyse, B; De Gersem, W; De Wagter, C; De Neve, W; Thierens, H
2004-07-21
A new accurate Monte Carlo code for IMRT dose computations, MCDE (Monte Carlo dose engine), is introduced. MCDE is based on BEAMnrc/DOSXYZnrc and consequently the accurate EGSnrc electron transport. DOSXYZnrc is reprogrammed as a component module for BEAMnrc. In this way both codes are interconnected elegantly, while maintaining the BEAM structure and only minimal changes to BEAMnrc.mortran are necessary. The treatment head of the Elekta SLiplus linear accelerator is modelled in detail. CT grids consisting of up to 200 slices of 512 x 512 voxels can be introduced and up to 100 beams can be handled simultaneously. The beams and CT data are imported from the treatment planning system GRATIS via a DICOM interface. To enable the handling of up to 50 x 10(6) voxels the system was programmed in Fortran95 to enable dynamic memory management. All region-dependent arrays (dose, statistics, transport arrays) were redefined. A scoring grid was introduced and superimposed on the geometry grid, to be able to limit the number of scoring voxels. The whole system uses approximately 200 MB of RAM and runs on a PC cluster consisting of 38 1.0 GHz processors. A set of in-house made scripts handle the parallellization and the centralization of the Monte Carlo calculations on a server. As an illustration of MCDE, a clinical example is discussed and compared with collapsed cone convolution calculations. At present, the system is still rather slow and is intended to be a tool for reliable verification of IMRT treatment planning in the case of the presence of tissue inhomogeneities such as air cavities. PMID:15357203
MCDE: a new Monte Carlo dose engine for IMRT
International Nuclear Information System (INIS)
A new accurate Monte Carlo code for IMRT dose computations, MCDE (Monte Carlo dose engine), is introduced. MCDE is based on BEAMnrc/DOSXYZnrc and consequently the accurate EGSnrc electron transport. DOSXYZnrc is reprogrammed as a component module for BEAMnrc. In this way both codes are interconnected elegantly, while maintaining the BEAM structure and only minimal changes to BEAMnrc.mortran are necessary. The treatment head of the Elekta SLiplus linear accelerator is modelled in detail. CT grids consisting of up to 200 slices of 512 x 512 voxels can be introduced and up to 100 beams can be handled simultaneously. The beams and CT data are imported from the treatment planning system GRATIS via a DICOM interface. To enable the handling of up to 50 x 106 voxels the system was programmed in Fortran95 to enable dynamic memory management. All region-dependent arrays (dose, statistics, transport arrays) were redefined. A scoring grid was introduced and superimposed on the geometry grid, to be able to limit the number of scoring voxels. The whole system uses approximately 200 MB of RAM and runs on a PC cluster consisting of 38 1.0 GHz processors. A set of in-house made scripts handle the parallellization and the centralization of the Monte Carlo calculations on a server. As an illustration of MCDE, a clinical example is discussed and compared with collapsed cone convolution calculations. At present, the system is still rather slow and is intended to be a tool for reliable verification of IMRT treatment planning in the case of the presence of tissue inhomogeneities such as air cavities. (note)
Assessment of patient dose in mammography using Monte Carlo simulation
International Nuclear Information System (INIS)
Breast doses due to mammographic examinations were assessed using a MIRD-type female phantom and Monte Carlo simulations. Clinical mammographic data, which vary according to the age group of the subject undergoing the examinations, were obtained from the Korea Cancer Center Hospital in Seoul. The tube potential was fixed to 26 kVp, most commonly used in the mammographic examination, and the source-film distance was kept constant at 65 cm. The breast tissue was assumed to have an even composition between glandular tissue and adipose tissue. The nominal breast equivalent doses were in the range from 0.6 to 1.8 mSv and the resulting effective doses ranged from 0.06 to 0.19 mSv depending on the age group and the projection modes. Lower doses were resulted at older ages. Contributions of organs other than the breast to the effective doses were negligible as long as the X-ray beam was adequately collimated and aligned to avoid exposure of other part of the body than the breast. This means that a simple breast-only phantom can be used in dosimetric calculations for mammography. (author)
The Monte Carlo simulation of the absorbed dose in quartz
Energy Technology Data Exchange (ETDEWEB)
Chen Shaowen [School of Physics Science and Engineering, Sun Yat-Sen University, Guangzhou, Guangdong 510275 (China) and Electron Engineering Department, Dongguan University of Technology, Dongguan 523808 (China)], E-mail: siumon@163.com; Liu Xiaowei; Zhang Chunxiang; Tang Qiang [School of Physics Science and Engineering, Sun Yat-Sen University, Guangzhou, Guangdong 510275 (China)
2009-05-15
Regeneration irradiation is a necessary procedure in TL or OSL dating protocol. The accuracy of measuring the absorbed dose is one of the important factors in dating. Since a beta source is often used in the regeneration irradiation process, the size of the quartz sample, pressure of nitrogen gas and the material of the sample holder may cause significant uncertainties in delivering the absorbed dose. In this work, the effects of the size of the quartz sample, the pressure of nitrogen gas and the material of the sample holder are simulated using the Monte Carlo method, and the uncertainties are discussed in these cases. The results show that they need to be considered in the dating.
Monte Carlo dose calculation in dental amalgam phantom.
Aziz, Mohd Zahri Abdul; Yusoff, A L; Osman, N D; Abdullah, R; Rabaie, N A; Salikin, M S
2015-01-01
It has become a great challenge in the modern radiation treatment to ensure the accuracy of treatment delivery in electron beam therapy. Tissue inhomogeneity has become one of the factors for accurate dose calculation, and this requires complex algorithm calculation like Monte Carlo (MC). On the other hand, computed tomography (CT) images used in treatment planning system need to be trustful as they are the input in radiotherapy treatment. However, with the presence of metal amalgam in treatment volume, the CT images input showed prominent streak artefact, thus, contributed sources of error. Hence, metal amalgam phantom often creates streak artifacts, which cause an error in the dose calculation. Thus, a streak artifact reduction technique was applied to correct the images, and as a result, better images were observed in terms of structure delineation and density assigning. Furthermore, the amalgam density data were corrected to provide amalgam voxel with accurate density value. As for the errors of dose uncertainties due to metal amalgam, they were reduced from 46% to as low as 2% at d80 (depth of the 80% dose beyond Zmax) using the presented strategies. Considering the number of vital and radiosensitive organs in the head and the neck regions, this correction strategy is suggested in reducing calculation uncertainties through MC calculation. PMID:26500401
Monte carlo dose calculation in dental amalgam phantom
Directory of Open Access Journals (Sweden)
Mohd Zahri Abdul Aziz
2015-01-01
Full Text Available It has become a great challenge in the modern radiation treatment to ensure the accuracy of treatment delivery in electron beam therapy. Tissue inhomogeneity has become one of the factors for accurate dose calculation, and this requires complex algorithm calculation like Monte Carlo (MC. On the other hand, computed tomography (CT images used in treatment planning system need to be trustful as they are the input in radiotherapy treatment. However, with the presence of metal amalgam in treatment volume, the CT images input showed prominent streak artefact, thus, contributed sources of error. Hence, metal amalgam phantom often creates streak artifacts, which cause an error in the dose calculation. Thus, a streak artifact reduction technique was applied to correct the images, and as a result, better images were observed in terms of structure delineation and density assigning. Furthermore, the amalgam density data were corrected to provide amalgam voxel with accurate density value. As for the errors of dose uncertainties due to metal amalgam, they were reduced from 46% to as low as 2% at d80 (depth of the 80% dose beyond Zmax using the presented strategies. Considering the number of vital and radiosensitive organs in the head and the neck regions, this correction strategy is suggested in reducing calculation uncertainties through MC calculation.
Monte Carlo N Particle code - Dose distribution of clinical electron beams in inhomogeneous phantoms
H A Nedaie; Mosleh-Shirazi, M. A.; Allahverdi, M.
2013-01-01
Electron dose distributions calculated using the currently available analytical methods can be associated with large uncertainties. The Monte Carlo method is the most accurate method for dose calculation in electron beams. Most of the clinical electron beam simulation studies have been performed using non- MCNP [Monte Carlo N Particle] codes. Given the differences between Monte Carlo codes, this work aims to evaluate the accuracy of MCNP4C-simulated electron dose distributions in a homogenous...
International Nuclear Information System (INIS)
A novel procedure for the generation of a realistic virtual Computed Tomography (CT) image of a patient, using the advanced Boundary RE Presentation (BREP)-based model MASH, has been implemented. This method can be used in radiotherapy assessment. It is shown that it is possible to introduce an artificial cancer, which can be modeled using mesh surfaces. The use of virtual CT images based on BREP models presents several advantages with respect to CT images of actual patients, such as automation, control and flexibility. As an example, two artificial cases, namely a brain and a prostate cancer, were created through the generation of images and tumor/organ contours. As a secondary objective, the described methodology has been used to generate input files for treatment planning system (TPS) and Monte Carlo code dose evaluation. In this paper, we consider treatment plans generated assuming a dose delivery via an active proton beam scanning performed with the INFN-IBA TPS kernel. Additionally, Monte Carlo simulations of the two treatment plans were carried out with GATE/GEANT4. The work demonstrates the feasibility of the approach based on the BREP modeling to produce virtual CT images. In conclusion, this study highlights the benefits in using digital phantom model capable of representing different anatomical structures and varying tumors across different patients. These models could be useful for assessing radiotherapy treatment planning systems (TPS) and computer simulations for the evaluation of the adsorbed dose. (author)
Energy Technology Data Exchange (ETDEWEB)
Milian, F. M.; Attili, A.; Russo, G; Marchetto, F.; Cirio, R., E-mail: felix_mas_milian@yahoo.com, E-mail: attili@to.infn.it, E-mail: russo@to.infn.it, E-mail: fmarchet@to.infn.it, E-mail: cirio@to.infn.it [Istituto Nazionale di Fisica Nucleare (INFN), Torino, TO (Italy); Bourhaleb, F., E-mail: bourhale@to.infn.it [Universita di Torino (UNITO), Torino, TO (Italy)
2013-07-01
A novel procedure for the generation of a realistic virtual Computed Tomography (CT) image of a patient, using the advanced Boundary RE Presentation (BREP)-based model MASH, has been implemented. This method can be used in radiotherapy assessment. It is shown that it is possible to introduce an artificial cancer, which can be modeled using mesh surfaces. The use of virtual CT images based on BREP models presents several advantages with respect to CT images of actual patients, such as automation, control and flexibility. As an example, two artificial cases, namely a brain and a prostate cancer, were created through the generation of images and tumor/organ contours. As a secondary objective, the described methodology has been used to generate input files for treatment planning system (TPS) and Monte Carlo code dose evaluation. In this paper, we consider treatment plans generated assuming a dose delivery via an active proton beam scanning performed with the INFN-IBA TPS kernel. Additionally, Monte Carlo simulations of the two treatment plans were carried out with GATE/GEANT4. The work demonstrates the feasibility of the approach based on the BREP modeling to produce virtual CT images. In conclusion, this study highlights the benefits in using digital phantom model capable of representing different anatomical structures and varying tumors across different patients. These models could be useful for assessing radiotherapy treatment planning systems (TPS) and computer simulations for the evaluation of the adsorbed dose. (author)
International Nuclear Information System (INIS)
An approximate continuous data fitting model for the dose deposition kernel was developed. The model uses a discrete Fourier transform to interpolate dose values in patient space and intensity distribution in treatment space. The continuous kernel was applied to the inverse problem of radiation treatment planning. In the problem a prescribed dose distribution was to be created using intensity modulation of several fields. The Cimmino algorithm suitable for solving large systems of inequalities was adapted. Upper and lower dose constraints for planning target volume (PTV) and organs at risk (OAR) can be implemented into the algorithm. Using continuous and discrete kernels an intensity modulation was computed in a two-dimensional phantom with a PTV and low-dose region, and in the real three-dimensional patient planning. Intensity modulations obtained using continuous and discrete kernels were in good agreement. (author)
International Nuclear Information System (INIS)
A point-kernel integral technique code, PKN, and the related data library have been developed to calculate neutron and secondary gamma-ray dose equivalents in water, concrete and iron shields for neutron sources in 3-dimensional geometry. The comparison between calculational results of the present code and those of the 1-dimensional transport code ANISN = JR, and the 2-dimensional transport code DOT4.2 showed a sufficient accuracy, and the availability of the PKN code has been confirmed. (author)
Clinical implementation of full Monte Carlo dose calculation in proton beam therapy
Energy Technology Data Exchange (ETDEWEB)
Paganetti, Harald; Jiang, Hongyu; Parodi, Katia; Slopsema, Roelf; Engelsman, Martijn [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, MA 02114 (United States)
2008-09-07
The goal of this work was to facilitate the clinical use of Monte Carlo proton dose calculation to support routine treatment planning and delivery. The Monte Carlo code Geant4 was used to simulate the treatment head setup, including a time-dependent simulation of modulator wheels (for broad beam modulation) and magnetic field settings (for beam scanning). Any patient-field-specific setup can be modeled according to the treatment control system of the facility. The code was benchmarked against phantom measurements. Using a simulation of the ionization chamber reading in the treatment head allows the Monte Carlo dose to be specified in absolute units (Gy per ionization chamber reading). Next, the capability of reading CT data information was implemented into the Monte Carlo code to model patient anatomy. To allow time-efficient dose calculation, the standard Geant4 tracking algorithm was modified. Finally, a software link of the Monte Carlo dose engine to the patient database and the commercial planning system was established to allow data exchange, thus completing the implementation of the proton Monte Carlo dose calculation engine ('DoC++'). Monte Carlo re-calculated plans are a valuable tool to revisit decisions in the planning process. Identification of clinically significant differences between Monte Carlo and pencil-beam-based dose calculations may also drive improvements of current pencil-beam methods. As an example, four patients (29 fields in total) with tumors in the head and neck regions were analyzed. Differences between the pencil-beam algorithm and Monte Carlo were identified in particular near the end of range, both due to dose degradation and overall differences in range prediction due to bony anatomy in the beam path. Further, the Monte Carlo reports dose-to-tissue as compared to dose-to-water by the planning system. Our implementation is tailored to a specific Monte Carlo code and the treatment planning system XiO (Computerized Medical
Monte Carlo calculation of ''skyshine'' neutron dose from ALS [Advanced Light Source
International Nuclear Information System (INIS)
This report discusses the following topics on ''skyshine'' neutron dose from ALS: Sources of radiation; ALS modeling for skyshine calculations; MORSE Monte-Carlo; Implementation of MORSE; Results of skyshine calculations from storage ring; and Comparison of MORSE shielding calculations
Monte Carlo calculation of 60Co γ-ray's albedo-dose rate from the air
International Nuclear Information System (INIS)
The Monte Carlo calculation of 60Co γ-ray's albedo-dose rate from the air is reported. A formula is presented with which the relations of the albedo-doserate with some parameters are simulated and fitted
IMRT dose delivery effects in radiotherapy treatment planning using Monte Carlo methods
Tyagi, Neelam
Inter- and intra-leaf transmission and head scatter can play significant roles in Intensity Modulated Radiation Therapy (IMRT)-based treatment deliveries. In order to accurately calculate the dose in the IMRT planning process, it is therefore important that the detailed geometry of the multi-leaf collimator (MLC), in addition to other components in the accelerator treatment head be accurately modeled. In this thesis Monte Carlo (MC) methods have been used to model the treatment head of a Varian linear accelerator. A comprehensive model of the Varian 120-leaf MLC has been developed within the DPM MC code and has been verified against measurements in homogeneous and heterogeneous phantom geometries under different IMRT delivery circumstances. Accuracy of the MLC model in simulating details in the leaf geometry has been established over a range of arbitrarily shaped fields and IMRT fields. A sensitivity analysis of the effect of the electron-on-target parameters and the structure of the flattening filter on the accuracy of calculated dose distributions has been conducted. Adjustment of the electron-on-target parameters resulting in optimal agreement with measurements was an iterative process, with the final parameters representing a tradeoff between small (3x3 cm2) and large (40x40 cm2) field sizes. A novel method based on adaptive kernel density estimation, in the phase space simulation process is also presented as an alternative to particle recycling. Using this model dosimetric differences between MLC-based static (SMLC) and dynamic (DMLC) deliveries have been investigated. Differences between SMLC and DMLC, possibly related to fluence and/or spectral changes, appear to vary systematically with the density of the medium. The effect of fluence modulation due to leaf sequencing shows differences, up to 10% between plans developed with 1% and 10% fluence intervals for both SMLC and DMLC-delivered sequences. Dose differences between planned and delivered leaf sequences
Implementation of Monte Carlo Dose calculation for CyberKnife treatment planning
Ma, C.-M.; Li, J. S.; Deng, J.; Fan, J.
2008-02-01
Accurate dose calculation is essential to advanced stereotactic radiosurgery (SRS) and stereotactic radiotherapy (SRT) especially for treatment planning involving heterogeneous patient anatomy. This paper describes the implementation of a fast Monte Carlo dose calculation algorithm in SRS/SRT treatment planning for the CyberKnife® SRS/SRT system. A superposition Monte Carlo algorithm is developed for this application. Photon mean free paths and interaction types for different materials and energies as well as the tracks of secondary electrons are pre-simulated using the MCSIM system. Photon interaction forcing and splitting are applied to the source photons in the patient calculation and the pre-simulated electron tracks are repeated with proper corrections based on the tissue density and electron stopping powers. Electron energy is deposited along the tracks and accumulated in the simulation geometry. Scattered and bremsstrahlung photons are transported, after applying the Russian roulette technique, in the same way as the primary photons. Dose calculations are compared with full Monte Carlo simulations performed using EGS4/MCSIM and the CyberKnife treatment planning system (TPS) for lung, head & neck and liver treatments. Comparisons with full Monte Carlo simulations show excellent agreement (within 0.5%). More than 10% differences in the target dose are found between Monte Carlo simulations and the CyberKnife TPS for SRS/SRT lung treatment while negligible differences are shown in head and neck and liver for the cases investigated. The calculation time using our superposition Monte Carlo algorithm is reduced up to 62 times (46 times on average for 10 typical clinical cases) compared to full Monte Carlo simulations. SRS/SRT dose distributions calculated by simple dose algorithms may be significantly overestimated for small lung target volumes, which can be improved by accurate Monte Carlo dose calculations.
Implementation of Monte Carlo Dose calculation for CyberKnife treatment planning
International Nuclear Information System (INIS)
Accurate dose calculation is essential to advanced stereotactic radiosurgery (SRS) and stereotactic radiotherapy (SRT) especially for treatment planning involving heterogeneous patient anatomy. This paper describes the implementation of a fast Monte Carlo dose calculation algorithm in SRS/SRT treatment planning for the CyberKnife (registered) SRS/SRT system. A superposition Monte Carlo algorithm is developed for this application. Photon mean free paths and interaction types for different materials and energies as well as the tracks of secondary electrons are pre-simulated using the MCSIM system. Photon interaction forcing and splitting are applied to the source photons in the patient calculation and the pre-simulated electron tracks are repeated with proper corrections based on the tissue density and electron stopping powers. Electron energy is deposited along the tracks and accumulated in the simulation geometry. Scattered and bremsstrahlung photons are transported, after applying the Russian roulette technique, in the same way as the primary photons. Dose calculations are compared with full Monte Carlo simulations performed using EGS4/MCSIM and the CyberKnife treatment planning system (TPS) for lung, head and neck and liver treatments. Comparisons with full Monte Carlo simulations show excellent agreement (within 0.5%). More than 10% differences in the target dose are found between Monte Carlo simulations and the CyberKnife TPS for SRS/SRT lung treatment while negligible differences are shown in head and neck and liver for the cases investigated. The calculation time using our superposition Monte Carlo algorithm is reduced up to 62 times (46 times on average for 10 typical clinical cases) compared to full Monte Carlo simulations. SRS/SRT dose distributions calculated by simple dose algorithms may be significantly overestimated for small lung target volumes, which can be improved by accurate Monte Carlo dose calculations
Implementation of Monte Carlo Dose calculation for CyberKnife treatment planning
Energy Technology Data Exchange (ETDEWEB)
Ma, C-M; Li, J S; Deng, J; Fan, J [Radiation Oncology Department, Fox Chase Cancer Center, Philadelphia, PA (United States)], E-mail: Charlie.ma@fccc.edu
2008-02-01
Accurate dose calculation is essential to advanced stereotactic radiosurgery (SRS) and stereotactic radiotherapy (SRT) especially for treatment planning involving heterogeneous patient anatomy. This paper describes the implementation of a fast Monte Carlo dose calculation algorithm in SRS/SRT treatment planning for the CyberKnife (registered) SRS/SRT system. A superposition Monte Carlo algorithm is developed for this application. Photon mean free paths and interaction types for different materials and energies as well as the tracks of secondary electrons are pre-simulated using the MCSIM system. Photon interaction forcing and splitting are applied to the source photons in the patient calculation and the pre-simulated electron tracks are repeated with proper corrections based on the tissue density and electron stopping powers. Electron energy is deposited along the tracks and accumulated in the simulation geometry. Scattered and bremsstrahlung photons are transported, after applying the Russian roulette technique, in the same way as the primary photons. Dose calculations are compared with full Monte Carlo simulations performed using EGS4/MCSIM and the CyberKnife treatment planning system (TPS) for lung, head and neck and liver treatments. Comparisons with full Monte Carlo simulations show excellent agreement (within 0.5%). More than 10% differences in the target dose are found between Monte Carlo simulations and the CyberKnife TPS for SRS/SRT lung treatment while negligible differences are shown in head and neck and liver for the cases investigated. The calculation time using our superposition Monte Carlo algorithm is reduced up to 62 times (46 times on average for 10 typical clinical cases) compared to full Monte Carlo simulations. SRS/SRT dose distributions calculated by simple dose algorithms may be significantly overestimated for small lung target volumes, which can be improved by accurate Monte Carlo dose calculations.
Ma, C.-M.; Pawlicki, T.; Jiang, S. B.; Li, J. S.; Deng, J.; Mok, E.; Kapur, A.; Xing, L.; Ma, L.; Boyer, A. L.
2000-09-01
The purpose of this work was to use Monte Carlo simulations to verify the accuracy of the dose distributions from a commercial treatment planning optimization system (Corvus, Nomos Corp., Sewickley, PA) for intensity-modulated radiotherapy (IMRT). A Monte Carlo treatment planning system has been implemented clinically to improve and verify the accuracy of radiotherapy dose calculations. Further modifications to the system were made to compute the dose in a patient for multiple fixed-gantry IMRT fields. The dose distributions in the experimental phantoms and in the patients were calculated and used to verify the optimized treatment plans generated by the Corvus system. The Monte Carlo calculated IMRT dose distributions agreed with the measurements to within 2% of the maximum dose for all the beam energies and field sizes for both the homogeneous and heterogeneous phantoms. The dose distributions predicted by the Corvus system, which employs a finite-size pencil beam (FSPB) algorithm, agreed with the Monte Carlo simulations and measurements to within 4% in a cylindrical water phantom with various hypothetical target shapes. Discrepancies of more than 5% (relative to the prescribed target dose) in the target region and over 20% in the critical structures were found in some IMRT patient calculations. The FSPB algorithm as implemented in the Corvus system is adequate for homogeneous phantoms (such as prostate) but may result in significant under- or over-estimation of the dose in some cases involving heterogeneities such as the air-tissue, lung-tissue and tissue-bone interfaces.
Monte Carlo calculation of received dose from ingestion and inhalation of natural uranium
International Nuclear Information System (INIS)
For the purpose of this study eighty samples are taken from the area Bela Crkva and Vrsac. The activity of radionuclide in the soil is determined by gamma- ray spectrometry. Monte Carlo method is used to calculate effective dose received by population resulting from the inhalation and ingestion of natural uranium. The estimated doses were compared with the legally prescribed levels. (author)
International Nuclear Information System (INIS)
The objective of this study was to evaluate the patient effective dose and scattered dose from recently developed dental mobile equipment in Korea. The MCNPX 2.6 (Los Alamos National Laboratory, USA) was used in a Monte Carlo simulation to calculate both the effective and scattered doses. The MCNPX code was constructed identically as in the general use of equipment and the effective dose and scattered dose were calculated using the KTMAN-2 digital phantom. The effective dose was calculated as 906 μSv. The equivalent doses per organ were calculated via the MCNPX code, and were 32 174 and 19 μSv in the salivary gland and oesophagus, respectively. The scattered dose of 22.5-32.6 μSv of the tube side at 25 cm from the centre in anterior and posterior planes was measured as 1.4-3 times higher than the detector side of 10.5-16.0 μSv. (authors)
An energy transfer method for 4D Monte Carlo dose calculation
Siebers, Jeffrey V; Zhong, Hualiang
2008-01-01
This article presents a new method for four-dimensional Monte Carlo dose calculations which properly addresses dose mapping for deforming anatomy. The method, called the energy transfer method (ETM), separates the particle transport and particle scoring geometries: Particle transport takes place in the typical rectilinear coordinate system of the source image, while energy deposition scoring takes place in a desired reference image via use of deformable image registration. Dose is the energy ...
Application of the peregrine Monte Carlo dose calculation system to stereotactic radiosurgery
International Nuclear Information System (INIS)
Purpose/Objective: This work describes the capability to perform Monte Carlo dose calculations for stereotactic radiosurgery within the framework of the PEREGRINE dose calculation system. A future study will use this capability to assess the clinical benefits to this technique of higher accuracy in dose calculation. Materials and Methods: PEREGRINE is a first-principles 3D Monte Carlo dose calculation system for clinical radiation therapy treatment planning (RTP) systems. By taking advantage of recent advances in low-cost computer commodity hardware, modern symmetric multiprocessor architectures and state-of-the-art Monte Carlo transport algorithms, PEREGRINE performs high-resolution (1 mm), high accuracy, Monte Carlo RTP calculations in times that are reasonable for clinical use (< 30 minutes.) The PEREGRINE source model provides a compact, accurate representation of the radiation source and the effects of beam modifiers. Our experience in implementing blocks, wedges, and static MLC ports in PEREGRINE as beam modifiers provides physics models that accurately reproduce the transmitted and scattered fluence at the patient surface. Adapting PEREGRINE to calculate stereotactic radiosurgery dose distributions requires extending the PEREGRINE source model to include stereotactic apertures and treatment arcs. The physics models used for other modifiers will accurately determine stereotactic aperture effects. We only need to provide a new geometry module to describe the physical properties of the apertures. Treatment arcs are easily implemented as a probability distribution in beam direction as a function of delivered dose. Results: A comparison of results from PEREGRINE calculations and experimental measurements made at the University of Wisconsin/Madison is presented. The distribution of direct, transmitted and scattered radiation and the resulting contributions to dose from stereotactic apertures are shown. The accuracy and calculational efficiency of the physics
Patient-specific CT dose determination from CT images using Monte Carlo simulations
Liang, Qing
Radiation dose from computed tomography (CT) has become a public concern with the increasing application of CT as a diagnostic modality, which has generated a demand for patient-specific CT dose determinations. This thesis work aims to provide a clinically applicable Monte-Carlo-based CT dose calculation tool based on patient CT images. The source spectrum was simulated based on half-value layer measurements. Analytical calculations along with the measured flux distribution were used to estimate the bowtie-filter geometry. Relative source output at different points in a cylindrical phantom was measured and compared with Monte Carlo simulations to verify the determined spectrum and bowtie-filter geometry. Sensitivity tests were designed with four spectra with the same kVp and different half-value layers, and showed that the relative output at different locations in a phantom is sensitive to different beam qualities. An mAs-to-dose conversion factor was determined with in-air measurements using an Exradin A1SL ionization chamber. Longitudinal dose profiles were measured with thermoluminescent dosimeters (TLDs) and compared with the Monte-Carlo-simulated dose profiles to verify the mAs-to-dose conversion factor. Using only the CT images to perform Monte Carlo simulations would cause dose underestimation due to the lack of a scatter region. This scenario was demonstrated with a cylindrical phantom study. Four different image extrapolation methods from the existing CT images and the Scout images were proposed. The results show that performing image extrapolation beyond the scan region improves the dose calculation accuracy under both step-shoot scan mode and helical scan mode. Two clinical studies were designed and comparisons were performed between the current CT dose metrics and the Monte-Carlo-based organ dose determination techniques proposed in this work. The results showed that the current CT dosimetry failed to show dose differences between patients with the same
Local dose enhancement in radiation therapy: Monte Carlo simulation study
International Nuclear Information System (INIS)
The development of nanotechnology has boosted the use of nanoparticles in radiation therapy in order to achieve greater therapeutic ratio between tumor and healthy tissues. Gold has been shown to be most suitable to this task due to the high biocompatibility and high atomic number, which contributes to a better in vivo distribution and for the local energy deposition. As a result, this study proposes to study, nanoparticle in the tumor cell. At a range of 11 nm from the nanoparticle surface, results have shown an absorbed dose 141 times higher for the medium with the gold nanoparticle compared to the water for an incident energy spectrum with maximum photon energy of 50 keV. It was also noted that when only scattered radiation is interacting with the gold nanoparticles, the dose was 134 times higher compared to enhanced local dose that remained significant even for scattered radiation. (author)
International Nuclear Information System (INIS)
Highlights: •Develop the novel Multi-Step CADIS (MS-CADIS) hybrid Monte Carlo/deterministic method for multi-step shielding analyses. •Accurately calculate shutdown dose rates using full-scale Monte Carlo models of fusion energy systems. •Demonstrate the dramatic efficiency improvement of the MS-CADIS method for the rigorous two step calculations of the shutdown dose rate in fusion reactors. -- Abstract: The rigorous 2-step (R2S) computational system uses three-dimensional Monte Carlo transport simulations to calculate the shutdown dose rate (SDDR) in fusion reactors. Accurate full-scale R2S calculations are impractical in fusion reactors because they require calculating space- and energy-dependent neutron fluxes everywhere inside the reactor. The use of global Monte Carlo variance reduction techniques was suggested for accelerating the R2S neutron transport calculation. However, the prohibitive computational costs of these approaches, which increase with the problem size and amount of shielding materials, inhibit their ability to accurately predict the SDDR in fusion energy systems using full-scale modeling of an entire fusion plant. This paper describes a novel hybrid Monte Carlo/deterministic methodology that uses the Consistent Adjoint Driven Importance Sampling (CADIS) method but focuses on multi-step shielding calculations. The Multi-Step CADIS (MS-CADIS) methodology speeds up the R2S neutron Monte Carlo calculation using an importance function that represents the neutron importance to the final SDDR. Using a simplified example, preliminary results showed that the use of MS-CADIS enhanced the efficiency of the neutron Monte Carlo simulation of an SDDR calculation by a factor of 550 compared to standard global variance reduction techniques, and that the efficiency enhancement compared to analog Monte Carlo is higher than a factor of 10,000
Experimental validation of Monte Carlo calculations for organ dose
International Nuclear Information System (INIS)
The problem of validating estimates of absorbed dose due to photon energy deposition is examined. The computational approaches used for the estimation of the photon energy deposition is examined. The limited data for validation of these approaches is discussed and suggestions made as to how better validation information might be obtained
International Nuclear Information System (INIS)
Interventional cardiology consists on a set of medical procedures which are mainly focused on diagnosing and treating patients who suffer cardiovascular diseases. Even though the usage of X-ray is justified on this case, it is greatly advised to evaluate the dose which professionals and patients will be exposed due to the fact that the complexity and length of the procedures often require high doses. The objective of this work is to estimate the radiation dose on both a patient and a physician through conversion coefficient (CCs) of effective dose (E) and equivalent dose (H) during a coronary angiography examination.The dose CCs was estimated using the Visual Monte Carlo code (VMC) and a pair of simulators anthropomorphic voxel (Female Adult VoXel). The CCs were normalized in terms of kerma-area product (KAP). As expected, for all situations studied, the patient in anteroposterior projection (AP) obtained the highest conversion coefficient of equivalent dose and effective dose (author)
Monte Carlo calculations of the impact of a hip prosthesis on the dose distribution
International Nuclear Information System (INIS)
Because of the ageing of the population, an increasing number of patients with hip prostheses are undergoing pelvic irradiation. Treatment planning systems (TPS) currently available are not always able to accurately predict the dose distribution around such implants. In fact, only Monte Carlo simulation has the ability to precisely calculate the impact of a hip prosthesis during radiotherapeutic treatment. Monte Carlo phantoms were developed to evaluate the dose perturbations during pelvic irradiation. A first model, constructed with the DOSXYZnrc usercode, was elaborated to determine the dose increase at the tissue-metal interface as well as the impact of the material coating the prosthesis. Next, CT-based phantoms were prepared, using the usercode CTCreate, to estimate the influence of the geometry and the composition of such implants on the beam attenuation. Thanks to a program that we developed, the study was carried out with CT-based phantoms containing a hip prosthesis without metal artefacts. Therefore, anthropomorphic phantoms allowed better definition of both patient anatomy and the hip prosthesis in order to better reproduce the clinical conditions of pelvic irradiation. The Monte Carlo results revealed the impact of certain coatings such as PMMA on dose enhancement at the tissue-metal interface. Monte Carlo calculations in CT-based phantoms highlighted the marked influence of the implant's composition, its geometry as well as its position within the beam on dose distribution
Jeraj, Robert; Keall, Paul
2000-12-01
The effect of the statistical uncertainty, or noise, in inverse treatment planning for intensity modulated radiotherapy (IMRT) based on Monte Carlo dose calculation was studied. Sets of Monte Carlo beamlets were calculated to give uncertainties at Dmax ranging from 0.2% to 4% for a lung tumour plan. The weights of these beamlets were optimized using a previously described procedure based on a simulated annealing optimization algorithm. Several different objective functions were used. It was determined that the use of Monte Carlo dose calculation in inverse treatment planning introduces two errors in the calculated plan. In addition to the statistical error due to the statistical uncertainty of the Monte Carlo calculation, a noise convergence error also appears. For the statistical error it was determined that apparently successfully optimized plans with a noisy dose calculation (3% 1σ at Dmax ), which satisfied the required uniformity of the dose within the tumour, showed as much as 7% underdose when recalculated with a noise-free dose calculation. The statistical error is larger towards the tumour and is only weakly dependent on the choice of objective function. The noise convergence error appears because the optimum weights are determined using a noisy calculation, which is different from the optimum weights determined for a noise-free calculation. Unlike the statistical error, the noise convergence error is generally larger outside the tumour, is case dependent and strongly depends on the required objectives.
Monte-Carlo Method Python Library for dose distribution Calculation in Brachytherapy
International Nuclear Information System (INIS)
The Cs-137 Brachytherapy treatment is performed in Madagascar since 2005. Time treatment calculation for prescribed dose is made manually. Monte-Carlo Method Python library written at Madagascar INSTN is experimentally used to calculate the dose distribution on the tumour and around it. The first validation of the code was done by comparing the library curves with the Nucletron company curves. To reduce the duration of the calculation, a Grid of PC's is set up with listner patch run on each PC. The library will be used to modelize the dose distribution in the CT scan patient picture for individual and better accuracy time calculation for a prescribed dose.
Modeling Silicon Diode Dose Response in Radiotherapy Fields using Fluence Pencil Kernels
Eklund, Karin
2010-01-01
In radiotherapy, cancer is treated with ionizing radiation, most commonly bremsstrahlung photons from electrons of several MeV. Secondary electrons produced in photon-interactions results in dose deposition. The treatment response is low for low doses, raises sharply for normal treatment doses and saturates at higher doses. This response pattern applies to both eradication of tumors and to complications in healthy tissues. Well controlled treatments require accurate dosimetry since the uncert...
Feasibility Study of Neutron Dose for Real Time Image Guided Proton Therapy: A Monte Carlo Study
Kim, Jin Sung; Shin, Jung Suk; Kim, Daehyun; Shin, EunHyuk; Chung, Kwangzoo; Cho, Sungkoo; Ahn, Sung Hwan; Ju, Sanggyu; Chung, Yoonsun; Jung, Sang Hoon; Han, Youngyih
2015-01-01
Two full rotating gantry with different nozzles (Multipurpose nozzle with MLC, Scanning Dedicated nozzle) with conventional cyclotron system is installed and under commissioning for various proton treatment options at Samsung Medical Center in Korea. The purpose of this study is to investigate neutron dose equivalent per therapeutic dose, H/D, to x-ray imaging equipment under various treatment conditions with monte carlo simulation. At first, we investigated H/D with the various modifications...
Organ doses from medical x-ray examinations calculated using Monte Carlo techniques
Jones, D G
1985-01-01
Monte Carlo techniques were used to calculate the mean doses received by 20 organs during diagnostic X-ray examinations. Results are presented for 22 commonly used radiographic views and for 45 combinations of tube voltage and filtration ranging from 50 to 140 kVp and 1.5 to 4 mm of aluminium, respectively.
Alem-Bezoubiri, A.; Bezoubiri, F.; Badreddine, A.; Mazrou, H.; Lounis-Mokrani, Z.
2014-04-01
A fully detailed Monte Carlo geometrical model of an 18 MV Varian Clinac 2100C medical linear accelerator, lodged at Blida Anti-Cancer Centre in Algeria, was developed during this study to estimate the photoneutrons spectra and doses at the patient table in a radiotherapy treatment room, for radiation protection purposes.
Gamma irradiator dose mapping: a Monte Carlo simulation and experimental measurements
International Nuclear Information System (INIS)
Gamma irradiator facilities can be used in a wide range of applications such as biological and chemical researches, food treatment and sterilization of medical devices and products. Dose mapping must be performed in these equipment in order to establish plant operational parameters, as dose uniformity, source utilization efficiency and maximum and minimum dose positions. The isodoses curves are generally measured using dosimeters distributed throughout the device, and this procedure often consume a large amount of dosimeters, irradiation time and manpower. However, a detailed curve doses identification of the irradiation facility can be performed using Monte Carlo simulation, which reduces significantly the monitoring with dosimeters. The present work evaluates the absorbed dose in the CDTN/CNEN Gammacell Irradiation Facility, using the Monte Carlo N-particles (MCNP) code. The Gammacell 220, serial number 39, was produced by Atomic Energy of Canada Limited and was loaded with sources of 60Co. Dose measurements using TLD and Fricke dosimeters were also performed to validate the calculations. The good agreement of the results shows that Monte Carlo simulations can be used as a predictive tool of irradiation planning for the CDTN/CNEN Gamma Cell Irradiator. (author)
Energy Technology Data Exchange (ETDEWEB)
Lee, Choonsik; Kim, Kwang Pyo; Long, Daniel; Fisher, Ryan; Tien, Chris; Simon, Steven L.; Bouville, Andre; Bolch, Wesley E. [Division of Cancer Epidemiology and Genetics, National Cancer Institute, National Institute of Health, Bethesda, Maryland 20852 (United States); Department of Nuclear Engineering, Kyung Hee University, Yongin 446-701 (Korea, Republic of); Department of Nuclear and Radiological Engineering, University of Florida, Gainesville, Florida 32611 (United States); Division of Cancer Epidemiology and Genetics, National Cancer Institute, National Institute of Health, Bethesda, Maryland 20852 (United States); Department of Nuclear and Radiological Engineering, University of Florida, Gainesville, Florida 32611 (United States)
2011-03-15
Purpose: To develop a computed tomography (CT) organ dose estimation method designed to readily provide organ doses in a reference adult male and female for different scan ranges to investigate the degree to which existing commercial programs can reasonably match organ doses defined in these more anatomically realistic adult hybrid phantomsMethods: The x-ray fan beam in the SOMATOM Sensation 16 multidetector CT scanner was simulated within the Monte Carlo radiation transport code MCNPX2.6. The simulated CT scanner model was validated through comparison with experimentally measured lateral free-in-air dose profiles and computed tomography dose index (CTDI) values. The reference adult male and female hybrid phantoms were coupled with the established CT scanner model following arm removal to simulate clinical head and other body region scans. A set of organ dose matrices were calculated for a series of consecutive axial scans ranging from the top of the head to the bottom of the phantoms with a beam thickness of 10 mm and the tube potentials of 80, 100, and 120 kVp. The organ doses for head, chest, and abdomen/pelvis examinations were calculated based on the organ dose matrices and compared to those obtained from two commercial programs, CT-EXPO and CTDOSIMETRY. Organ dose calculations were repeated for an adult stylized phantom by using the same simulation method used for the adult hybrid phantom. Results: Comparisons of both lateral free-in-air dose profiles and CTDI values through experimental measurement with the Monte Carlo simulations showed good agreement to within 9%. Organ doses for head, chest, and abdomen/pelvis scans reported in the commercial programs exceeded those from the Monte Carlo calculations in both the hybrid and stylized phantoms in this study, sometimes by orders of magnitude. Conclusions: The organ dose estimation method and dose matrices established in this study readily provides organ doses for a reference adult male and female for different
Monte Carlo angular dose distribution of the microselectron HDR 192Ir brachytherapy source
International Nuclear Information System (INIS)
Polar dose profiles around the Nucletron MicroSelectron high dose rate (HDR) 192Ir brachytherapy source were calculated using the Monte Carlo radiation transport code MCNP (Monte Carlo N Particle) version 4A. The geometry modeled consisted of an identical simulation of the construction of the MicroSelectron HDR source located at the centre of a spherical water phantom of 100cm radius. Doses were calculated using a spherical coordinate system at 5 degree intervals (measured relative to the cable) at radii of 0.25, 0.5,1.0, 3.0, 5.0 and 7.0cm. These polar doses were compared to equivalent profiles from the Nucletron PLATO Brachytherapy Planning System (BPS) version 13.X. At 3.0, 5.0 and 7.0cm radii, the Monte Carlo and BPS profiles are generally within 3%. The near field polar dose profiles however, are in significant disagreement. At 1.0cm radius, the discrepancy can exceed 5%. At 0.5cm this figure rises to 15%, and even 60% at 0.25cm radius
Aguirre, Eder; David, Mariano; deAlmeida, Carlos E
2016-01-01
This work studies the impact of systematic uncertainties associated to interaction cross sections on depth dose curves determined by Monte Carlo simulations. The corresponding sensitivity factors are quantified by changing cross sections in a given amount and determining the variation in the dose. The influence of total cross sections for all particles, photons and only for Compton scattering is addressed. The PENELOPE code was used in all simulations. It was found that photon cross section sensitivity factors depend on depth. In addition, they are positive and negative for depths below and above an equilibrium depth, respectively. At this depth, sensitivity factors are null. The equilibrium depths found in this work agree very well with the mean free path of the corresponding incident photon energy. Using the sensitivity factors reported here, it is possible to estimate the impact of photon cross section uncertainties on the uncertainty of Monte Carlo-determined depth dose curves.
Calculation of radiation dose to the lens of the eye using Monte Carlo simulation
International Nuclear Information System (INIS)
The radiation dose to the lens of the eye of patients undergoing diagnostic and interventional radiological procedures of the lacrimal drainage system has been calculated using a Monte Carlo technique. The technique has also been suggested for the retrospective estimation of the lens dose; when applied to individual patients, good correlation is obtained. In such study, data is required for image acquisition frame numbers and fluoro on-time, mean exposure values for these parameters, and the ratio of lens-to-air dose (viz. the head factor, HF) derived for a standard adult head
SU-E-J-60: Efficient Monte Carlo Dose Calculation On CPU-GPU Heterogeneous Systems
International Nuclear Information System (INIS)
Purpose: It is well-known that the performance of GPU-based Monte Carlo dose calculation implementations is bounded by memory bandwidth. One major cause of this bottleneck is the random memory writing patterns in dose deposition, which leads to several memory efficiency issues on GPU such as un-coalesced writing and atomic operations. We propose a new method to alleviate such issues on CPU-GPU heterogeneous systems, which achieves overall performance improvement for Monte Carlo dose calculation. Methods: Dose deposition is to accumulate dose into the voxels of a dose volume along the trajectories of radiation rays. Our idea is to partition this procedure into the following three steps, which are fine-tuned for CPU or GPU: (1) each GPU thread writes dose results with location information to a buffer on GPU memory, which achieves fully-coalesced and atomic-free memory transactions; (2) the dose results in the buffer are transferred to CPU memory; (3) the dose volume is constructed from the dose buffer on CPU. We organize the processing of all radiation rays into streams. Since the steps within a stream use different hardware resources (i.e., GPU, DMA, CPU), we can overlap the execution of these steps for different streams by pipelining. Results: We evaluated our method using a Monte Carlo Convolution Superposition (MCCS) program and tested our implementation for various clinical cases on a heterogeneous system containing an Intel i7 quad-core CPU and an NVIDIA TITAN GPU. Comparing with a straightforward MCCS implementation on the same system (using both CPU and GPU for radiation ray tracing), our method gained 2-5X speedup without losing dose calculation accuracy. Conclusion: The results show that our new method improves the effective memory bandwidth and overall performance for MCCS on the CPU-GPU systems. Our proposed method can also be applied to accelerate other Monte Carlo dose calculation approaches. This research was supported in part by NSF under Grants CCF
Clouvas, A; Antonopoulos-Domis, M; Silva, J
2000-01-01
The dose rate conversion factors D/sub CF/ (absorbed dose rate in air per unit activity per unit of soil mass, nGy h/sup -1/ per Bq kg/sup -1/) are calculated 1 m above ground for photon emitters of natural radionuclides uniformly distributed in the soil. Three Monte Carlo codes are used: 1) The MCNP code of Los Alamos; 2) The GEANT code of CERN; and 3) a Monte Carlo code developed in the Nuclear Technology Laboratory of the Aristotle University of Thessaloniki. The accuracy of the Monte Carlo results is tested by the comparison of the unscattered flux obtained by the three Monte Carlo codes with an independent straightforward calculation. All codes and particularly the MCNP calculate accurately the absorbed dose rate in air due to the unscattered radiation. For the total radiation (unscattered plus scattered) the D/sub CF/ values calculated from the three codes are in very good agreement between them. The comparison between these results and the results deduced previously by other authors indicates a good ag...
International Nuclear Information System (INIS)
The dose rate conversion factors dot DCF (absorbed dose rate in air per unit activity per unit of soil mass, nGy h-1 per Bq kg-1) are calculated 1 m above ground for photon emitters of natural radionuclides uniformly distributed in the soil. Three Monte Carlo codes are used: (1) The MCNP code of Los Alamos; (2) The GEANT code of CERN; and (3) a Monte Carlo code developed in the Nuclear Technology Laboratory of the Aristotle University of Thessaloniki. The accuracy of the Monte Carlo results is tested by the comparison of the unscattered flux obtained by the three Monte Carlo codes with an independent straightforward calculation. All codes and particularly the MCNP calculate accurately the absorbed dose rate in air due to the unscattered radiation. For the total radiation (unscattered plus scattered) the dot DCF values calculated from the three codes are in very good agreement between them. The comparison between these results and the results deduced previously by other authors indicates a good agreement (less than 15% of difference) for photon energies above 1,500 keV. Antithetically, the agreement is not as good (difference of 20--30%) for the low energy photons
International Nuclear Information System (INIS)
The γ-index test has been commonly adopted to quantify the degree of agreement between a reference dose distribution and an evaluation dose distribution. Monte Carlo (MC) simulation has been widely used for the radiotherapy dose calculation for both clinical and research purposes. The goal of this work is to investigate both theoretically and experimentally the impact of the MC statistical fluctuation on the γ-index test when the fluctuation exists in the reference, the evaluation, or both dose distributions. To the first order approximation, we theoretically demonstrated in a simplified model that the statistical fluctuation tends to overestimate γ-index values when existing in the reference dose distribution and underestimate γ-index values when existing in the evaluation dose distribution given the original γ-index is relatively large for the statistical fluctuation. Our numerical experiments using realistic clinical photon radiation therapy cases have shown that (1) when performing a γ-index test between an MC reference dose and a non-MC evaluation dose, the average γ-index is overestimated and the gamma passing rate decreases with the increase of the statistical noise level in the reference dose; (2) when performing a γ-index test between a non-MC reference dose and an MC evaluation dose, the average γ-index is underestimated when they are within the clinically relevant range and the gamma passing rate increases with the increase of the statistical noise level in the evaluation dose; (3) when performing a γ-index test between an MC reference dose and an MC evaluation dose, the gamma passing rate is overestimated due to the statistical noise in the evaluation dose and underestimated due to the statistical noise in the reference dose. We conclude that the γ-index test should be used with caution when comparing dose distributions computed with MC simulation. (paper)
Poster — Thur Eve — 14: Improving Tissue Segmentation for Monte Carlo Dose Calculation using DECT
Energy Technology Data Exchange (ETDEWEB)
Di Salvio, A.; Bedwani, S.; Carrier, J-F. [Centre hospitalier de l' Université de Montréal (Canada); Bouchard, H. [National Physics Laboratory, Teddington (United Kingdom)
2014-08-15
Purpose: To improve Monte Carlo dose calculation accuracy through a new tissue segmentation technique with dual energy CT (DECT). Methods: Electron density (ED) and effective atomic number (EAN) can be extracted directly from DECT data with a stoichiometric calibration method. Images are acquired with Monte Carlo CT projections using the user code egs-cbct and reconstructed using an FDK backprojection algorithm. Calibration is performed using projections of a numerical RMI phantom. A weighted parameter algorithm then uses both EAN and ED to assign materials to voxels from DECT simulated images. This new method is compared to a standard tissue characterization from single energy CT (SECT) data using a segmented calibrated Hounsfield unit (HU) to ED curve. Both methods are compared to the reference numerical head phantom. Monte Carlo simulations on uniform phantoms of different tissues using dosxyz-nrc show discrepancies in depth-dose distributions. Results: Both SECT and DECT segmentation methods show similar performance assigning soft tissues. Performance is however improved with DECT in regions with higher density, such as bones, where it assigns materials correctly 8% more often than segmentation with SECT, considering the same set of tissues and simulated clinical CT images, i.e. including noise and reconstruction artifacts. Furthermore, Monte Carlo results indicate that kV photon beam depth-dose distributions can double between two tissues of density higher than muscle. Conclusions: A direct acquisition of ED and the added information of EAN with DECT data improves tissue segmentation and increases the accuracy of Monte Carlo dose calculation in kV photon beams.
International Nuclear Information System (INIS)
Contrast-enhanced stereotactic synchrotron radiation therapy (SSRT) is an innovative technique based on localized dose-enhancement effects obtained by reinforced photoelectric absorption in the tumor. Medium energy monochromatic X-rays (50 - 100 keV) are used for irradiating tumors previously loaded with a high-Z element. Clinical trials of SSRT are being prepared at the European Synchrotron Radiation Facility (ESRF), an iodinated contrast agent will be used. In order to compute the energy deposited in the patient (dose), a dedicated treatment planning system (TPS) has been developed for the clinical trials, based on the ISOgray TPS. This work focuses on the SSRT specific modifications of the TPS, especially to the PENELOPE-based Monte Carlo dose engine. The TPS uses a dedicated Monte Carlo simulation of medium energy polarized photons to compute the deposited energy in the patient. Simulations are performed considering the synchrotron source, the modeled beamline geometry and finally the patient. Specific materials were also implemented in the voxelized geometry of the patient, to consider iodine concentrations in the tumor. The computation process has been optimized and parallelized. Finally a specific computation of absolute doses and associated irradiation times (instead of monitor units) was implemented. The dedicated TPS was validated with depth dose curves, dose profiles and absolute dose measurements performed at the ESRF in a water tank and solid water phantoms with or without bone slabs. (author)
Noblet, C.; Chiavassa, S.; Smekens, F.; Sarrut, D.; Passal, V.; Suhard, J.; Lisbona, A.; Paris, F.; Delpon, G.
2016-05-01
In preclinical studies, the absorbed dose calculation accuracy in small animals is fundamental to reliably investigate and understand observed biological effects. This work investigated the use of the split exponential track length estimator (seTLE), a new kerma based Monte Carlo dose calculation method for preclinical radiotherapy using a small animal precision micro irradiator, the X-RAD 225Cx. Monte Carlo modelling of the irradiator with GATE/GEANT4 was extensively evaluated by comparing measurements and simulations for half-value layer, percent depth dose, off-axis profiles and output factors in water and water-equivalent material for seven circular fields, from 20 mm down to 1 mm in diameter. Simulated and measured dose distributions in cylinders of water obtained for a 360° arc were also compared using dose, distance-to-agreement and gamma-index maps. Simulations and measurements agreed within 3% for all static beam configurations, with uncertainties estimated to 1% for the simulation and 3% for the measurements. Distance-to-agreement accuracy was better to 0.14 mm. For the arc irradiations, gamma-index maps of 2D dose distributions showed that the success rate was higher than 98%, except for the 0.1 cm collimator (92%). Using the seTLE method, MC simulations compute 3D dose distributions within minutes for realistic beam configurations with a clinically acceptable accuracy for beam diameter as small as 1 mm.
Monte Carlo calculation of dose to water of a 106Ru COB-type ophthalmic plaque
International Nuclear Information System (INIS)
The concave eye applicators with 106Ru/106Rh or 90Sr/90Y beta-ray sources are worldwide used in brachytherapy for treating intraocular tumors. It raises the need to know the exact dose delivered by beta radiation to tumors but measurement of the dose to water (or tissue) is very difficult due to short range of electrons. The Monte Carlo technique provides a powerful tool for calculation of the dose and dose distributions which helps to predict and determine the doses from different shapes of various types of eye applicators more accurately. The Monte Carlo code MCNPX has been used to calculate dose distributions from a COB-type 106Ru/106Rh ophthalmic applicator manufactured by Eckert and Ziegler BEBIG GmbH. This type of a concave eye applicator has a cut-out whose purpose is to protect the eye nerve which makes the dose distribution more complicated. Several calculations have been performed including depth dose along the applicator central axis and various dose distributions. The depth dose along the applicator central axis and the dose distribution on a spherical surface 1 mm above the plaque inner surface have been compared with measurement data provided by the manufacturer. For distances from 0.5 to 4 mm above the surface, the agreement was within 2.5% and from 5 mm the difference increased from 6% up to 25% at 10 mm whereas the uncertainty on manufacturer data is 20% (2s). It is assumed that the difference is caused by nonuniformly distributed radioactivity over the applicator radioactive layer
Townson, Reid W
2013-01-01
Due to the increasing complexity of radiotherapy delivery, accurate dose verification has become an essential part of the clinical treatment process. The purpose of this work was to develop an electronic portal image (EPI) based pre-treatment verification technique capable of quickly reconstructing 3D dose distributions from both coplanar and non-coplanar treatments. The dose reconstruction is performed in a spherical water phantom by modulating, based on EPID measurements, pre-calculated Monte Carlo (MC) doselets defined on a spherical coordinate system. This is called the spherical doselet modulation (SDM) method. This technique essentially eliminates the statistical uncertainty of the MC dose calculations by exploiting both azimuthal symmetry in a patient-independent phase-space and symmetry of a virtual spherical water phantom. The symmetry also allows the number of doselets necessary for dose reconstruction to be reduced by a factor of about 250. In this work, 51 doselets were used. The SDM method mitiga...
International Nuclear Information System (INIS)
Even with state of the art treatment planning systems the photon dose calculation can be erroneous under certain circumstances. In these cases Monte Carlo methods promise a higher accuracy. We have used the photon transport code CHILD of the GSF-Forschungszentrum, which was developed to calculate dose in diagnostic radiation protection matters. The code was refined for application in radiotherapy for high energy photon irradiation and should serve for dose verification in individual cases. The irradiation phantom can be entered as any desired 3D matrix or be generated automatically from an individual CT database. The particle transport takes into account pair production, photo, and Compton effect with certain approximations. Efficiency is increased by the method of 'fractional photons'. The generated secondary electrons are followed by the unscattered continuous-slowing-down-approximation (CSDA). The developed Monte Carlo code Monaco Matrix was tested with simple homogeneous and heterogeneous phantoms through comparisons with simulations of the well known but slower EGS4 code. The use of a point source with a direction independent energy spectrum as simplest model of the radiation field from the accelerator head is shown to be sufficient for simulation of actual accelerator depth dose curves. Good agreement (<2%) was found for depth dose curves in water and in bone. With complex test phantoms and comparisons with EGS4 calculated dose profiles some drawbacks in the code were found. Thus, the implementation of the electron multiple-scattering should lead us to step by step improvement of the algorithm. (orig.)
International Nuclear Information System (INIS)
The purpose of this study is to perform a clinical evaluation of the first commercial (MDS Nordion, now Nucletron) treatment planning system for electron beams incorporating Monte Carlo dose calculation module. This software implements Kawrakow's VMC++ voxel-based Monte Carlo calculation algorithm. The accuracy of the dose distribution calculations is evaluated by direct comparisons with extensive sets of measured data in homogeneous and heterogeneous phantoms at different source-to-surface distances (SSDs) and gantry angles. We also verify the accuracy of the Monte Carlo module for monitor unit calculations in comparison with independent hand calculations for homogeneous water phantom at two different SSDs. All electron beams in the range 6-20 MeV are from a Siemens KD-2 linear accelerator. We used 10 000 or 50 000 histories/cm2 in our Monte Carlo calculations, which led to about 2.5% and 1% relative standard error of the mean of the calculated dose. The dose calculation time depends on the number of histories, the number of voxels used to map the patient anatomy, the field size, and the beam energy. The typical run time of the Monte Carlo calculations (10 000 histories/cm2) is 1.02 min on a 2.2 GHz Pentium 4 Xeon computer for a 9 MeV beam, 10x10 cm2 field size, incident on the phantom 15x15x10 cm3 consisting of 31 CT slices and voxels size of 3x3x3 mm3 (total of 486 720 voxels). We find good agreement (discrepancies smaller than 5%) for most of the tested dose distributions. We also find excellent agreement (discrepancies of 2.5% or less) for the monitor unit calculations relative to the independent manual calculations. The accuracy of monitor unit calculations does not depend on the SSD used, which allows the use of one virtual machine for each beam energy for all arbitrary SSDs. In some cases the test results are found to be sensitive to the voxel size applied such that bigger systematic errors (>5%) occur when large voxel sizes interfere with the extensions of
Radiation doses in volume-of-interest breast computed tomography—A Monte Carlo simulation study
Energy Technology Data Exchange (ETDEWEB)
Lai, Chao-Jen, E-mail: cjlai3711@gmail.com; Zhong, Yuncheng; Yi, Ying; Wang, Tianpeng; Shaw, Chris C. [Department of Imaging Physics, The University of Texas MD Anderson Cancer Center, Houston, Texas 77030-4009 (United States)
2015-06-15
Purpose: Cone beam breast computed tomography (breast CT) with true three-dimensional, nearly isotropic spatial resolution has been developed and investigated over the past decade to overcome the problem of lesions overlapping with breast anatomical structures on two-dimensional mammographic images. However, the ability of breast CT to detect small objects, such as tissue structure edges and small calcifications, is limited. To resolve this problem, the authors proposed and developed a volume-of-interest (VOI) breast CT technique to image a small VOI using a higher radiation dose to improve that region’s visibility. In this study, the authors performed Monte Carlo simulations to estimate average breast dose and average glandular dose (AGD) for the VOI breast CT technique. Methods: Electron–Gamma-Shower system code-based Monte Carlo codes were used to simulate breast CT. The Monte Carlo codes estimated were validated using physical measurements of air kerma ratios and point doses in phantoms with an ion chamber and optically stimulated luminescence dosimeters. The validated full cone x-ray source was then collimated to simulate half cone beam x-rays to image digital pendant-geometry, hemi-ellipsoidal, homogeneous breast phantoms and to estimate breast doses with full field scans. 13-cm in diameter, 10-cm long hemi-ellipsoidal homogeneous phantoms were used to simulate median breasts. Breast compositions of 25% and 50% volumetric glandular fractions (VGFs) were used to investigate the influence on breast dose. The simulated half cone beam x-rays were then collimated to a narrow x-ray beam with an area of 2.5 × 2.5 cm{sup 2} field of view at the isocenter plane and to perform VOI field scans. The Monte Carlo results for the full field scans and the VOI field scans were then used to estimate the AGD for the VOI breast CT technique. Results: The ratios of air kerma ratios and dose measurement results from the Monte Carlo simulation to those from the physical
International Nuclear Information System (INIS)
The purpose of this study was to compare dose distributions from three different algorithms with the x-ray Voxel Monte Carlo (XVMC) calculations, in actual computed tomography (CT) scans for use in stereotactic radiotherapy (SRT) of small lung cancers. Slow CT scan of 20 patients was performed and the internal target volume (ITV) was delineated on Pinnacle3. All plans were first calculated with a scatter homogeneous mode (SHM) which is compatible with Clarkson algorithm using Pinnacle3 treatment planning system (TPS). The planned dose was 48 Gy in 4 fractions. In a second step, the CT images, structures and beam data were exported to other treatment planning systems (TPSs). Collapsed cone convolution (CCC) from Pinnacle3, superposition (SP) from XiO, and XVMC from Monaco were used for recalculating. The dose distributions and the Dose Volume Histograms (DVHs) were compared with each other. The phantom test revealed that all algorithms could reproduce the measured data within 1% except for the SHM with inhomogeneous phantom. For the patient study, the SHM greatly overestimated the isocenter (IC) doses and the minimal dose received by 95% of the PTV (PTV95) compared to XVMC. The differences in mean doses were 2.96 Gy (6.17%) for IC and 5.02 Gy (11.18%) for PTV95. The DVH's and dose distributions with CCC and SP were in agreement with those obtained by XVMC. The average differences in IC doses between CCC and XVMC, and SP and XVMC were -1.14% (p = 0.17), and -2.67% (p = 0.0036), respectively. Our work clearly confirms that the actual practice of relying solely on a Clarkson algorithm may be inappropriate for SRT planning. Meanwhile, CCC and SP were close to XVMC simulations and actual dose distributions obtained in lung SRT
Effects of CT based Voxel Phantoms on Dose Distribution Calculated with Monte Carlo Method
Institute of Scientific and Technical Information of China (English)
Chen Chaobin; Huang Qunying; Wu Yican
2005-01-01
A few CT-based voxel phantoms were produced to investigate the sensitivity of Monte Carlo simulations of X-ray beam and electron beam to the proportions of elements and the mass densities of the materials used to express the patient's anatomical structure. The human body can be well outlined by air, lung, adipose, muscle, soft bone and hard bone to calculate the dose distribution with Monte Carlo method. The effects of the calibration curves established by using various CT scanners are not clinically significant based on our investigation. The deviation from the values of cumulative dose volume histogram derived from CT-based voxel phantoms is less than 1% for the given target.
Effects of CT based Voxel Phantoms on Dose Distribution Calculated with Monte Carlo Method
Chen, Chaobin; Huang, Qunying; Wu, Yican
2005-04-01
A few CT-based voxel phantoms were produced to investigate the sensitivity of Monte Carlo simulations of x-ray beam and electron beam to the proportions of elements and the mass densities of the materials used to express the patient's anatomical structure. The human body can be well outlined by air, lung, adipose, muscle, soft bone and hard bone to calculate the dose distribution with Monte Carlo method. The effects of the calibration curves established by using various CT scanners are not clinically significant based on our investigation. The deviation from the values of cumulative dose volume histogram derived from CT-based voxel phantoms is less than 1% for the given target.
Dose conversion coefficients for ICRP110 voxel phantom in the Geant4 Monte Carlo code
Martins, M. C.; Cordeiro, T. P. V.; Silva, A. X.; Souza-Santos, D.; Queiroz-Filho, P. P.; Hunt, J. G.
2014-02-01
The reference adult male voxel phantom recommended by International Commission on Radiological Protection no. 110 was implemented in the Geant4 Monte Carlo code. Geant4 was used to calculate Dose Conversion Coefficients (DCCs) expressed as dose deposited in organs per air kerma for photons, electrons and neutrons in the Annals of the ICRP. In this work the AP and PA irradiation geometries of the ICRP male phantom were simulated for the purpose of benchmarking the Geant4 code. Monoenergetic photons were simulated between 15 keV and 10 MeV and the results were compared with ICRP 110, the VMC Monte Carlo code and the literature data available, presenting a good agreement.
International Nuclear Information System (INIS)
A model of a gamma sterilizer was built using the ITS/ACCEPT Monte Carlo code and verified through dosimetry. Individual dosimetry measurements in homogeneous material were pooled to represent larger bodies that could be simulated in a reasonable time. With the assumptions and simplifications described, dose predictions were within 2-5% of dosimetry. The model was used to simulate product movement through the sterilizer and to predict information useful for process optimization and facility design
Monte Carlo N Particle code - Dose distribution of clinical electron beams in inhomogeneous phantoms
Directory of Open Access Journals (Sweden)
H A Nedaie
2013-01-01
Full Text Available Electron dose distributions calculated using the currently available analytical methods can be associated with large uncertainties. The Monte Carlo method is the most accurate method for dose calculation in electron beams. Most of the clinical electron beam simulation studies have been performed using non- MCNP [Monte Carlo N Particle] codes. Given the differences between Monte Carlo codes, this work aims to evaluate the accuracy of MCNP4C-simulated electron dose distributions in a homogenous phantom and around inhomogeneities. Different types of phantoms ranging in complexity were used; namely, a homogeneous water phantom and phantoms made of polymethyl methacrylate slabs containing different-sized, low- and high-density inserts of heterogeneous materials. Electron beams with 8 and 15 MeV nominal energy generated by an Elekta Synergy linear accelerator were investigated. Measurements were performed for a 10 cm × 10 cm applicator at a source-to-surface distance of 100 cm. Individual parts of the beam-defining system were introduced into the simulation one at a time in order to show their effect on depth doses. In contrast to the first scattering foil, the secondary scattering foil, X and Y jaws and applicator provide up to 5% of the dose. A 2%/2 mm agreement between MCNP and measurements was found in the homogenous phantom, and in the presence of heterogeneities in the range of 1-3%, being generally within 2% of the measurements for both energies in a "complex" phantom. A full-component simulation is necessary in order to obtain a realistic model of the beam. The MCNP4C results agree well with the measured electron dose distributions.
International Nuclear Information System (INIS)
Different codes were used for Monte Carlo calculations in radiation therapy. In this study, a new Monte Carlo Simulation Program (MCSP) was developed for the effects of the physical parameters of photons emitted from a Siemens Primus clinical linear accelerator (LINAC) on the dose distribution in water. For MCSP, it was written considering interactions of photons with matter. Here, it was taken into account mainly two interactions: The Compton (or incoherent) scattering and photoelectric effect. Photons which come to water phantom surface emitting from a point source were Bremsstrahlung photons. It should be known the energy distributions of these photons for following photons. Bremsstrahlung photons which have 6 MeV (6 MV photon mode) maximum energies were taken into account. In the 6 MV photon mode, the energies of photons were sampled from using Mohan's experimental energy spectrum (Mohan at al 1985). In order to investigate the performance and accuracy of the simulation, measured and calculated (MCSP) percentage depth dose curves and dose profiles were compared. The Monte Carlo results were shown good agreement with experimental measurements.
International Nuclear Information System (INIS)
Different codes were used for Monte Carlo calculations in radiation therapy. In this study, a new Monte Carlo Simulation Program (MCSP) was developed for the effects of the physical parameters of photons emitted from a Siemens Primus clinical linear accelerator (LINAC) on the dose distribution in water. For MCSP, it was written considering interactions of photons with matter. Here, it was taken into account mainly two interactions: The Compton (or incoherent) scattering and photoelectric effect. Photons which come to water phantom surface emitting from a point source were Bremsstrahlung photons. It should be known the energy distributions of these photons for following photons. Bremsstrahlung photons which have 6 MeV (6 MV photon mode) maximum energies were taken into account. In the 6 MV photon mode, the energies of photons were sampled from using Mohan's experimental energy spectrum (Mohan at al 1985). In order to investigate the performance and accuracy of the simulation, measured and calculated (MCSP) percentage depth dose curves and dose profiles were compared. The Monte Carlo results were shown good agreement with experimental measurements.
Energy Technology Data Exchange (ETDEWEB)
Wieslander, Elinore; Knoeoes, Tommy [Radiation Physics, Lund University Hospital, SE-221 85 Lund (Sweden)
2003-10-21
An increasing number of patients receiving radiation therapy have metallic implants such as hip prostheses. Therefore, beams are normally set up to avoid irradiation through the implant; however, this cannot always be accomplished. In such situations, knowledge of the accuracy of the used treatment planning system (TPS) is required. Two algorithms, the pencil beam (PB) and the collapsed cone (CC), are implemented in the studied TPS. Comparisons are made with Monte Carlo simulations for 6 and 18 MV. The studied materials are steel, CoCrMo, Orthinox(a stainless steel alloy and registered trademark of Stryker Corporation), TiAlV and Ti. Monte Carlo simulated depth dose curves and dose profiles are compared to CC and PB calculated data. The CC algorithm shows overall a better agreement with Monte Carlo than the PB algorithm. Thus, it is recommended to use the CC algorithm to get the most accurate dose calculation both for the planning target volume and for tissues adjacent to the implants when beams are set up to pass through implants.
International Nuclear Information System (INIS)
An increasing number of patients receiving radiation therapy have metallic implants such as hip prostheses. Therefore, beams are normally set up to avoid irradiation through the implant; however, this cannot always be accomplished. In such situations, knowledge of the accuracy of the used treatment planning system (TPS) is required. Two algorithms, the pencil beam (PB) and the collapsed cone (CC), are implemented in the studied TPS. Comparisons are made with Monte Carlo simulations for 6 and 18 MV. The studied materials are steel, CoCrMo, Orthinox(a stainless steel alloy and registered trademark of Stryker Corporation), TiAlV and Ti. Monte Carlo simulated depth dose curves and dose profiles are compared to CC and PB calculated data. The CC algorithm shows overall a better agreement with Monte Carlo than the PB algorithm. Thus, it is recommended to use the CC algorithm to get the most accurate dose calculation both for the planning target volume and for tissues adjacent to the implants when beams are set up to pass through implants
Diagnostic X ray dose profiles in molar teeth using Monte Carlo simulation
International Nuclear Information System (INIS)
The dose profiles in molar teeth from diagnostic X rays was calculated using the Monte Carlo software program MCNP4c2. The information calculated supports needs in EPR retrospective dosimetry to account for diagnostic X ray exposures in teeth. Only tooth positions 6, 7 and 8 were simulated (the three teeth furthest back including the wisdom teeth) using a very detailed model of the pertinent physiology. The lingual and buccal halves of teeth were evaluated as were the crown dentin and roots in tooth position 7. Linear dose profiles through the enamel were also calculated. (author)
Applying graphics processor units to Monte Carlo dose calculation in radiation therapy
Directory of Open Access Journals (Sweden)
Bakhtiari M
2010-01-01
Full Text Available We investigate the potential in using of using a graphics processor unit (GPU for Monte-Carlo (MC-based radiation dose calculations. The percent depth dose (PDD of photons in a medium with known absorption and scattering coefficients is computed using a MC simulation running on both a standard CPU and a GPU. We demonstrate that the GPU′s capability for massive parallel processing provides a significant acceleration in the MC calculation, and offers a significant advantage for distributed stochastic simulations on a single computer. Harnessing this potential of GPUs will help in the early adoption of MC for routine planning in a clinical environment.
International Nuclear Information System (INIS)
This study determined the influence of patient individuality on lung organ doses for chest computed tomography (CT) examinations, viewed in the context of the recommendation of the ICRP 103. Within this current recommendation, a more individualized dose estimation is emphasized. The new ICRP 110 voxelized adult phantom was used and compared to calculation of lung doses for chest CT studies with identical scan parameters (120 kV, 135 mAs, 100 mm collimation, 1.5 pitch). For all patient images, the lung was contoured, and the scanning geometry was simulated using the Monte Carlo method. The lungs were completely included in the scan area. A user code was developed for the Monte Carlo package EGSnrc, which enables the simulation of a CT examination procedure and allows an efficient dose scoring within a patient geometry. All simulations were calculated with the same CT source model and calibrated to a realistic CTDIair value. Simulation values were grouped into 1 mSv classes. The organ dose classes fit well to a Gaussian distribution (adjusted correlation coefficient R2 = 0.95). The mean value of the fit was 10 mSv, with a standard deviation of 2 mSv. The variability was about ±30% with a minimum of 8 mSv and maximum of 13 mSv. The calculated lung dose of the ICRP adult female phantom was approximately 11 mSv and thus within the calculated standard deviation of the patient pool. The correlation between lung volume and dose was weak (adjusted correlation coefficient R2 = 0.33). Gender specific differences between the ICRP male and female phantoms were about 17%. In comparison, the differences between the female and a limited set of male patient studies were not statistically significant. Further, the relation between the HU values of CT scans and material/density necessary for the Monte Carlo simulations was investigated. It resulted that the simple but commonly employed relationship leads to significant deviations compared to definite materials in the ICRP phantoms
International Nuclear Information System (INIS)
A common approach to implementing the Monte Carlo method for the calculation of brachytherapy radiation dose deposition is to use a phase space file containing information on particles emitted from a brachytherapy source. However, the loading of the phase space file during the dose calculation consumes a large amount of computer random access memory, imposing a higher requirement for computer hardware. In this study, we propose a method to parameterize the information (e.g., particle location, direction and energy) stored in the phase space file by using several probability distributions. This method was implemented for dose calculations of a commercial Ir-192 high dose rate source. Dose calculation accuracy of the parameterized source was compared to the results observed using the full phase space file in a simple water phantom and in a clinical breast cancer case. The results showed the parameterized source at a size of 200 kB was as accurate as the phase space file represented source of 1.1 GB. By using the parameterized source representation, a compact Monte Carlo job can be designed, which allows an easy setup for parallel computing in brachytherapy planning. (paper)
Zhang, M.; Zou, W.; Chen, T.; Kim, L.; Khan, A.; Haffty, B.; Yue, N. J.
2014-01-01
A common approach to implementing the Monte Carlo method for the calculation of brachytherapy radiation dose deposition is to use a phase space file containing information on particles emitted from a brachytherapy source. However, the loading of the phase space file during the dose calculation consumes a large amount of computer random access memory, imposing a higher requirement for computer hardware. In this study, we propose a method to parameterize the information (e.g., particle location, direction and energy) stored in the phase space file by using several probability distributions. This method was implemented for dose calculations of a commercial Ir-192 high dose rate source. Dose calculation accuracy of the parameterized source was compared to the results observed using the full phase space file in a simple water phantom and in a clinical breast cancer case. The results showed the parameterized source at a size of 200 kB was as accurate as the phase space file represented source of 1.1 GB. By using the parameterized source representation, a compact Monte Carlo job can be designed, which allows an easy setup for parallel computing in brachytherapy planning.
Inverse treatment planning for radiation therapy based on fast Monte Carlo dose calculation
International Nuclear Information System (INIS)
An inverse treatment planning system based on fast Monte Carlo (MC) dose calculation is presented. It allows optimisation of intensity modulated dose distributions in 15 to 60 minutes on present day personal computers. If a multi-processor machine is available, parallel simulation of particle histories is also possible, leading to further calculation time reductions. The optimisation process is divided into two stages. The first stage results influence profiles based on pencil beam (PB) dose calculation. The second stage starts with MC verification and post-optimisation of the PB dose and fluence distributions. Because of the potential to accurately model beam modifiers, MC based inverse planning systems are able to optimise compensator thicknesses and leaf trajectories instead of intensity profiles only. The corresponding techniques, whose implementation is the subject for future work, are also presented here. (orig.)
International Nuclear Information System (INIS)
In some postmastectomy breast radiotherapy, patients are often irradiated with a temporary tissue expander. Most tissue expanders present a high density metallic disk inside which produce severe streaking artifacts in CT images and, as a consequence, is expected to affect the dose calculations . With the implementation of complex technique such as the IMRT technique, more rigorous verification is required in order to ensure the accurate determination of the absorbed dose before the treatment delivery. Monte Carlo (MC) algorithms have shown to be a reliable tool to provide improved dose accuracy in such situations. The aim of this work is to assess the accuracy of the dose calculation performed with a commercial TPS for breast IMRT radiotherapy in presence of metallic expanders (model McGhan Style 150). A MC method is used as gold standard for this evaluation. (Author)
Effects of human model configuration in Monte Carlo calculations on organ doses from CT examinations
International Nuclear Information System (INIS)
A new dosimetry system, WAZA-ARI, is being developed to estimate radiation dose from Computed Tomography (CT) examination in Japan. The dose estimation in WAZA-ARI utilizes organ dose data, which have been derived by Monte Carlo calculations using Particle and Heavy Ion Transport code System, PHITS. A Japanese adult male phantom, JM phantom, is adapted as a reference human model in the calculations, because the physique and inner organ masses agree well with the average values for Japanese adult males. On the other hand, each patient has arbitrary physical characteristics. Thus, the effects of human body configuration on organ doses are studied by applying another Japanese male model and the reference phantom by the International Commission on Radiological Protection (ICRP) to PHITS. In addition, this paper describes computation conditions for the three human models, which are constructed in the format of voxel phantom with different resolutions. (author)
Energy Technology Data Exchange (ETDEWEB)
Zarza-Moreno, M.; Calvo Ortega, J. F.; Jesus, A. P.; Casals Farran, J.
2013-07-01
In some postmastectomy breast radiotherapy, patients are often irradiated with a temporary tissue expander. Most tissue expanders present a high density metallic disk inside which produce severe streaking artifacts in CT images and, as a consequence, is expected to affect the dose calculations . With the implementation of complex technique such as the IMRT technique, more rigorous verification is required in order to ensure the accurate determination of the absorbed dose before the treatment delivery. Monte Carlo (MC) algorithms have shown to be a reliable tool to provide improved dose accuracy in such situations. The aim of this work is to assess the accuracy of the dose calculation performed with a commercial TPS for breast IMRT radiotherapy in presence of metallic expanders (model McGhan Style 150). A MC method is used as gold standard for this evaluation. (Author)
Energy Technology Data Exchange (ETDEWEB)
Chibani, Omar, E-mail: omar.chibani@fccc.edu; C-M Ma, Charlie [Fox Chase Cancer Center, Philadelphia, Pennsylvania 19111 (United States)
2014-05-15
Purpose: To present a new accelerated Monte Carlo code for CT-based dose calculations in high dose rate (HDR) brachytherapy. The new code (HDRMC) accounts for both tissue and nontissue heterogeneities (applicator and contrast medium). Methods: HDRMC uses a fast ray-tracing technique and detailed physics algorithms to transport photons through a 3D mesh of voxels representing the patient anatomy with applicator and contrast medium included. A precalculated phase space file for the{sup 192}Ir source is used as source term. HDRM is calibrated to calculated absolute dose for real plans. A postprocessing technique is used to include the exact density and composition of nontissue heterogeneities in the 3D phantom. Dwell positions and angular orientations of the source are reconstructed using data from the treatment planning system (TPS). Structure contours are also imported from the TPS to recalculate dose-volume histograms. Results: HDRMC was first benchmarked against the MCNP5 code for a single source in homogenous water and for a loaded gynecologic applicator in water. The accuracy of the voxel-based applicator model used in HDRMC was also verified by comparing 3D dose distributions and dose-volume parameters obtained using 1-mm{sup 3} versus 2-mm{sup 3} phantom resolutions. HDRMC can calculate the 3D dose distribution for a typical HDR cervix case with 2-mm resolution in 5 min on a single CPU. Examples of heterogeneity effects for two clinical cases (cervix and esophagus) were demonstrated using HDRMC. The neglect of tissue heterogeneity for the esophageal case leads to the overestimate of CTV D90, CTV D100, and spinal cord maximum dose by 3.2%, 3.9%, and 3.6%, respectively. Conclusions: A fast Monte Carlo code for CT-based dose calculations which does not require a prebuilt applicator model is developed for those HDR brachytherapy treatments that use CT-compatible applicators. Tissue and nontissue heterogeneities should be taken into account in modern HDR
Monte Carlo study of electron dose distributions produced by the elekta precise linear accelerator
International Nuclear Information System (INIS)
Background: Monte Carlo simulation of radiation transport is considered to be one of the most accurate methods of radiation therapy dose calculation and has ability to reduce the uncertainty in the calculated dose to a few percent. Aims: (1) To study the efficacy of the MCNP4C Monte Carlo code to simulate the dose distribution in a homogeneous medium produced by electron beams from the Elekta Precise linear accelerator. (2) To quantify the effect of introduction of various components to the simulated geometry for the above machine. Materials/Methods: Full Monte Carlo simulation of the detailed geometry of the Precise treatment head for 8 and 15 MeV energies and 10 x 10 applicator was performed. Experimental depth dose and lateral profiles at 2 cm depth were measured using a P-type diode detector with a 2.5 mm diameter. To quantify the effects of different parts of the treatment head, seven cases were simulated for a 15 MeV beam to reflect increasing levels of complexity, by step-wise introduction of beam divergence, primary and secondary scattering foils, secondary collimators, applicator, Mirror and Mylar screen. Results: The discrepancy between measured and calculated data is within 2 %/2 mm at both 8 and 15 MeV. In terms of the mean and most probable energies at the surface, the difference was < 0.2 MeV for the majority of cases and the maximum deviation was no more than 0.3 MeV. Conclusions: The results obtained with MCNP4C agree well with measured electron dose distributions. Inclusion of all the main components of the treatment head in the simulated geometry is necessary to avoid discrepancies of about 5 % compared to measurements. (authors)
Makarevich, K. O.; Minenko, V. F.; Verenich, K. A.; Kuten, S. A.
2016-05-01
This work is dedicated to modeling dental radiographic examinations to assess the absorbed doses of patients and effective doses. For simulating X-ray spectra, the TASMIP empirical model is used. Doses are assessed on the basis of the Monte Carlo method by using MCNP code for voxel phantoms of ICRP. The results of the assessment of doses to individual organs and effective doses for different types of dental examinations and features of X-ray tube are presented.
An energy transfer method for 4D Monte Carlo dose calculation.
Siebers, Jeffrey V; Zhong, Hualiang
2008-09-01
This article presents a new method for four-dimensional Monte Carlo dose calculations which properly addresses dose mapping for deforming anatomy. The method, called the energy transfer method (ETM), separates the particle transport and particle scoring geometries: Particle transport takes place in the typical rectilinear coordinate system of the source image, while energy deposition scoring takes place in a desired reference image via use of deformable image registration. Dose is the energy deposited per unit mass in the reference image. ETM has been implemented into DOSXYZnrc and compared with a conventional dose interpolation method (DIM) on deformable phantoms. For voxels whose contents merge in the deforming phantom, the doses calculated by ETM are exactly the same as an analytical solution, contrasting to the DIM which has an average 1.1% dose discrepancy in the beam direction with a maximum error of 24.9% found in the penumbra of a 6 MV beam. The DIM error observed persists even if voxel subdivision is used. The ETM is computationally efficient and will be useful for 4D dose addition and benchmarking alternative 4D dose addition algorithms. PMID:18841862
Evaluation of a new commercial Monte Carlo dose calculation algorithm for electron beams
International Nuclear Information System (INIS)
Purpose: In this report the authors present the validation of a Monte Carlo dose calculation algorithm (XiO EMC from Elekta Software) for electron beams. Methods: Calculated and measured dose distributions were compared for homogeneous water phantoms and for a 3D heterogeneous phantom meant to approximate the geometry of a trachea and spine. Comparisons of measurements and calculated data were performed using 2D and 3D gamma index dose comparison metrics. Results: Measured outputs agree with calculated values within estimated uncertainties for standard and extended SSDs for open applicators, and for cutouts, with the exception of the 17 MeV electron beam at extended SSD for cutout sizes smaller than 5 × 5 cm2. Good agreement was obtained between calculated and experimental depth dose curves and dose profiles (minimum number of measurements that pass a 2%/2 mm agreement 2D gamma index criteria for any applicator or energy was 97%). Dose calculations in a heterogeneous phantom agree with radiochromic film measurements (>98% of pixels pass a 3 dimensional 3%/2 mm γ-criteria) provided that the steep dose gradient in the depth direction is considered. Conclusions: Clinically acceptable agreement (at the 2%/2 mm level) between the measurements and calculated data for measurements in water are obtained for this dose calculation algorithm. Radiochromic film is a useful tool to evaluate the accuracy of electron MC treatment planning systems in heterogeneous media
A clinical study of lung cancer dose calculation accuracy with Monte Carlo simulation
International Nuclear Information System (INIS)
The accuracy of dose calculation is crucial to the quality of treatment planning and, consequently, to the dose delivered to patients undergoing radiation therapy. Current general calculation algorithms such as Pencil Beam Convolution (PBC) and Collapsed Cone Convolution (CCC) have shortcomings in regard to severe inhomogeneities, particularly in those regions where charged particle equilibrium does not hold. The aim of this study was to evaluate the accuracy of the PBC and CCC algorithms in lung cancer radiotherapy using Monte Carlo (MC) technology. Four treatment plans were designed using Oncentra Masterplan TPS for each patient. Two intensity-modulated radiation therapy (IMRT) plans were developed using the PBC and CCC algorithms, and two three-dimensional conformal therapy (3DCRT) plans were developed using the PBC and CCC algorithms. The DICOM-RT files of the treatment plans were exported to the Monte Carlo system to recalculate. The dose distributions of GTV, PTV and ipsilateral lung calculated by the TPS and MC were compared. For 3DCRT and IMRT plans, the mean dose differences for GTV between the CCC and MC increased with decreasing of the GTV volume. For IMRT, the mean dose differences were found to be higher than that of 3DCRT. The CCC algorithm overestimated the GTV mean dose by approximately 3% for IMRT. For 3DCRT plans, when the volume of the GTV was greater than 100 cm3, the mean doses calculated by CCC and MC almost have no difference. PBC shows large deviations from the MC algorithm. For the dose to the ipsilateral lung, the CCC algorithm overestimated the dose to the entire lung, and the PBC algorithm overestimated V20 but underestimated V5; the difference in V10 was not statistically significant. PBC substantially overestimates the dose to the tumour, but the CCC is similar to the MC simulation. It is recommended that the treatment plans for lung cancer be developed using an advanced dose calculation algorithm other than PBC. MC can accurately
Energy Technology Data Exchange (ETDEWEB)
Lee, Choonsik; Kim, Kwang Pyo; Long, Daniel J.; Bolch, Wesley E. [Division of Cancer Epidemiology and Genetics, National Cancer Institute, National Institute of Health, Bethesda, Maryland 20852 (United States); Department of Nuclear Engineering, Kyung Hee University, Gyeonggi-do, 446906 (Korea, Republic of); J. Crayton Pruitt Family Department of Biomedical Engineering, University of Florida, Gainesville, Florida 32611 (United States)
2012-04-15
Purpose: To establish an organ dose database for pediatric and adolescent reference individuals undergoing computed tomography (CT) examinations by using Monte Carlo simulation. The data will permit rapid estimates of organ and effective doses for patients of different age, gender, examination type, and CT scanner model. Methods: The Monte Carlo simulation model of a Siemens Sensation 16 CT scanner previously published was employed as a base CT scanner model. A set of absorbed doses for 33 organs/tissues normalized to the product of 100 mAs and CTDI{sub vol} (mGy/100 mAs mGy) was established by coupling the CT scanner model with age-dependent reference pediatric hybrid phantoms. A series of single axial scans from the top of head to the feet of the phantoms was performed at a slice thickness of 10 mm, and at tube potentials of 80, 100, and 120 kVp. Using the established CTDI{sub vol}- and 100 mAs-normalized dose matrix, organ doses for different pediatric phantoms undergoing head, chest, abdomen-pelvis, and chest-abdomen-pelvis (CAP) scans with the Siemens Sensation 16 scanner were estimated and analyzed. The results were then compared with the values obtained from three independent published methods: CT-Expo software, organ dose for abdominal CT scan derived empirically from patient abdominal circumference, and effective dose per dose-length product (DLP). Results: Organ and effective doses were calculated and normalized to 100 mAs and CTDI{sub vol} for different CT examinations. At the same technical setting, dose to the organs, which were entirely included in the CT beam coverage, were higher by from 40 to 80% for newborn phantoms compared to those of 15-year phantoms. An increase of tube potential from 80 to 120 kVp resulted in 2.5-2.9-fold greater brain dose for head scans. The results from this study were compared with three different published studies and/or techniques. First, organ doses were compared to those given by CT-Expo which revealed dose
Uncertainties in Monte Carlo-based absorbed dose calculations for an experimental benchmark
International Nuclear Information System (INIS)
There is a need to verify the accuracy of general purpose Monte Carlo codes like EGSnrc, which are commonly employed for investigations of dosimetric problems in radiation therapy. A number of experimental benchmarks have been published to compare calculated values of absorbed dose to experimentally determined values. However, there is a lack of absolute benchmarks, i.e. benchmarks without involved normalization which may cause some quantities to be cancelled. Therefore, at the Physikalisch-Technische Bundesanstalt a benchmark experiment was performed, which aimed at the absolute verification of radiation transport calculations for dosimetry in radiation therapy. A thimble-type ionization chamber in a solid phantom was irradiated by high-energy bremsstrahlung and the mean absorbed dose in the sensitive volume was measured per incident electron of the target. The characteristics of the accelerator and experimental setup were precisely determined and the results of a corresponding Monte Carlo simulation with EGSnrc are presented within this study. For a meaningful comparison, an analysis of the uncertainty of the Monte Carlo simulation is necessary. In this study uncertainties with regard to the simulation geometry, the radiation source, transport options of the Monte Carlo code and specific interaction cross sections are investigated, applying the general methodology of the Guide to the expression of uncertainty in measurement. Besides studying the general influence of changes in transport options of the EGSnrc code, uncertainties are analyzed by estimating the sensitivity coefficients of various input quantities in a first step. Secondly, standard uncertainties are assigned to each quantity which are known from the experiment, e.g. uncertainties for geometric dimensions. Data for more fundamental quantities such as photon cross sections and the I-value of electron stopping powers are taken from literature. The significant uncertainty contributions are identified as
Uncertainties in Monte Carlo-based absorbed dose calculations for an experimental benchmark
Renner, F.; Wulff, J.; Kapsch, R.-P.; Zink, K.
2015-10-01
There is a need to verify the accuracy of general purpose Monte Carlo codes like EGSnrc, which are commonly employed for investigations of dosimetric problems in radiation therapy. A number of experimental benchmarks have been published to compare calculated values of absorbed dose to experimentally determined values. However, there is a lack of absolute benchmarks, i.e. benchmarks without involved normalization which may cause some quantities to be cancelled. Therefore, at the Physikalisch-Technische Bundesanstalt a benchmark experiment was performed, which aimed at the absolute verification of radiation transport calculations for dosimetry in radiation therapy. A thimble-type ionization chamber in a solid phantom was irradiated by high-energy bremsstrahlung and the mean absorbed dose in the sensitive volume was measured per incident electron of the target. The characteristics of the accelerator and experimental setup were precisely determined and the results of a corresponding Monte Carlo simulation with EGSnrc are presented within this study. For a meaningful comparison, an analysis of the uncertainty of the Monte Carlo simulation is necessary. In this study uncertainties with regard to the simulation geometry, the radiation source, transport options of the Monte Carlo code and specific interaction cross sections are investigated, applying the general methodology of the Guide to the expression of uncertainty in measurement. Besides studying the general influence of changes in transport options of the EGSnrc code, uncertainties are analyzed by estimating the sensitivity coefficients of various input quantities in a first step. Secondly, standard uncertainties are assigned to each quantity which are known from the experiment, e.g. uncertainties for geometric dimensions. Data for more fundamental quantities such as photon cross sections and the I-value of electron stopping powers are taken from literature. The significant uncertainty contributions are identified as
GPUMCD: a new GPU-oriented Monte Carlo dose calculation platform
Hissoiny, Sami; Ozell, Benoît; Després, Philippe
2011-01-01
Purpose: Monte Carlo methods are considered the gold standard for dosimetric computations in radiotherapy. Their execution time is however still an obstacle to the routine use of Monte Carlo packages in a clinical setting. To address this problem, a completely new, and designed from the ground up for the GPU, Monte Carlo dose calculation package for voxelized geometries is proposed: GPUMCD. Method : GPUMCD implements a coupled photon-electron Monte Carlo simulation for energies in the range 0.01 MeV to 20 MeV. An analogue simulation of photon interactions is used and a Class II condensed history method has been implemented for the simulation of electrons. A new GPU random number generator, some divergence reduction methods as well as other optimization strategies are also described. GPUMCD was run on a NVIDIA GTX480 while single threaded implementations of EGSnrc and DPM were run on an Intel Core i7 860. Results : Dosimetric results obtained with GPUMCD were compared to EGSnrc. In all but one test case, 98% o...
Fast Monte Carlo Simulation for Patient-specific CT/CBCT Imaging Dose Calculation
Jia, Xun; Gu, Xuejun; Jiang, Steve B
2011-01-01
Recently, X-ray imaging dose from computed tomography (CT) or cone beam CT (CBCT) scans has become a serious concern. Patient-specific imaging dose calculation has been proposed for the purpose of dose management. While Monte Carlo (MC) dose calculation can be quite accurate for this purpose, it suffers from low computational efficiency. In response to this problem, we have successfully developed a MC dose calculation package, gCTD, on GPU architecture under the NVIDIA CUDA platform for fast and accurate estimation of the x-ray imaging dose received by a patient during a CT or CBCT scan. Techniques have been developed particularly for the GPU architecture to achieve high computational efficiency. Dose calculations using CBCT scanning geometry in a homogeneous water phantom and a heterogeneous Zubal head phantom have shown good agreement between gCTD and EGSnrc, indicating the accuracy of our code. In terms of improved efficiency, it is found that gCTD attains a speed-up of ~400 times in the homogeneous water ...
MO-G-BRF-09: Investigating Magnetic Field Dose Effects in Mice: A Monte Carlo Study
Energy Technology Data Exchange (ETDEWEB)
Rubinstein, A; Guindani, M; Followill, D; Melancon, A; Hazle, J; Court, L [UT MD Anderson Cancer Center, Houston, TX (United States)
2014-06-15
Purpose: In MRI-linac treatments, radiation dose distributions are affected by magnetic fields, especially at high-density/low-density interfaces. Radiobiological consequences of magnetic field dose effects are presently unknown; therefore, preclinical studies are needed to ensure the safe clinical use of MRI-linacs. This study investigates the optimal combination of beam energy and magnetic field strength needed for preclinical murine studies. Methods: The Monte Carlo code MCNP6 was used to simulate the effects of a magnetic field when irradiating a mouse-sized lung phantom with a 1.0cmx1.0cm photon beam. Magnetic field effects were examined using various beam energies (225kVp, 662keV[Cs-137], and 1.25MeV[Co-60]) and magnetic field strengths (0.75T, 1.5T, and 3T). The resulting dose distributions were compared to Monte Carlo results for humans with various field sizes and patient geometries using a 6MV/1.5T MRI-linac. Results: In human simulations, the addition of a 1.5T magnetic field caused an average dose increase of 49% (range:36%–60%) to lung at the soft tissue-to-lung interface and an average dose decrease of 30% (range:25%–36%) at the lung-to-soft tissue interface. In mouse simulations, the magnetic fields had no effect on the 225kVp dose distribution. The dose increases for the Cs-137 beam were 12%, 33%, and 49% for 0.75T, 1.5T, and 3.0T magnetic fields, respectively while the dose decreases were 7%, 23%, and 33%. For the Co-60 beam, the dose increases were 14%, 45%, and 41%, and the dose decreases were 18%, 35%, and 35%. Conclusion: The magnetic field dose effects observed in mouse phantoms using a Co-60 beam with 1.5T or 3T fields and a Cs-137 beam with a 3T field compare well with those seen in simulated human treatments with an MRI-linac. These irradiator/magnet combinations are suitable for preclinical studies investigating potential biological effects of delivering radiation therapy in the presence of a magnetic field. Partially funded by Elekta.
International Nuclear Information System (INIS)
To report the result of independent absorbed-dose calculations based on a Monte Carlo (MC) algorithm in volumetric modulated arc therapy (VMAT) for various treatment sites. All treatment plans were created by the superposition/convolution (SC) algorithm of SmartArc (Pinnacle V9.2, Philips). The beam information was converted into the format of the Monaco V3.3 (Elekta), which uses the X-ray voxel-based MC (XVMC) algorithm. The dose distribution was independently recalculated in the Monaco. The dose for the planning target volume (PTV) and the organ at risk (OAR) were analyzed via comparisons with those of the treatment plan. Before performing an independent absorbed-dose calculation, the validation was conducted via irradiation from 3 different gantry angles with a 10- × 10-cm2 field. For the independent absorbed-dose calculation, 15 patients with cancer (prostate, 5; lung, 5; head and neck, 3; rectal, 1; and esophageal, 1) who were treated with single-arc VMAT were selected. To classify the cause of the dose difference between the Pinnacle and Monaco TPSs, their calculations were also compared with the measurement data. In validation, the dose in Pinnacle agreed with that in Monaco within 1.5%. The agreement in VMAT calculations between Pinnacle and Monaco using phantoms was exceptional; at the isocenter, the difference was less than 1.5% for all the patients. For independent absorbed-dose calculations, the agreement was also extremely good. For the mean dose for the PTV in particular, the agreement was within 2.0% in all the patients; specifically, no large difference was observed for high-dose regions. Conversely, a significant difference was observed in the mean dose for the OAR. For patients with prostate cancer, the mean rectal dose calculated in Monaco was significantly smaller than that calculated in Pinnacle. There was no remarkable difference between the SC and XVMC calculations in the high-dose regions. The difference observed in the low-dose regions may
Tetrahedral-mesh-based computational human phantom for fast Monte Carlo dose calculations
International Nuclear Information System (INIS)
Although polygonal-surface computational human phantoms can address several critical limitations of conventional voxel phantoms, their Monte Carlo simulation speeds are much slower than those of voxel phantoms. In this study, we sought to overcome this problem by developing a new type of computational human phantom, a tetrahedral mesh phantom, by converting a polygonal surface phantom to a tetrahedral mesh geometry. The constructed phantom was implemented in the Geant4 Monte Carlo code to calculate organ doses as well as to measure computation speed, the values were then compared with those for the original polygonal surface phantom. It was found that using the tetrahedral mesh phantom significantly improved the computation speed by factors of between 150 and 832 considering all of the particles and simulated energies other than the low-energy neutrons (0.01 and 1 MeV), for which the improvement was less significant (17.2 and 8.8 times, respectively). (paper)
International Nuclear Information System (INIS)
By the help of a Monte-Carlo program the dose that single organs, organ groups and bigger or smaller parts of body would receive on an average, caused by an irradiation definitely fixed by the geometry of irradiation and photon energy, can be determined. Thus the phantom in connection with the Monte-Carlo program can be used for several considerations as for example - calculation of dose from occupational exposures - calculation of dose from diagnostic procedures - calculation of dose from radiotherapy procedures. (orig.)
SU-E-T-238: Monte Carlo Estimation of Cerenkov Dose for Photo-Dynamic Radiotherapy
Energy Technology Data Exchange (ETDEWEB)
Chibani, O; Price, R; Ma, C [Fox Chase Cancer Center, Philadelphia, PA (United States); Eldib, A [Fox Chase Cancer Center, Philadelphia, PA (United States); University Cairo (Egypt); Mora, G [de Lisboa, Codex, Lisboa (Portugal)
2014-06-01
Purpose: Estimation of Cerenkov dose from high-energy megavoltage photon and electron beams in tissue and its impact on the radiosensitization using Protoporphyrine IX (PpIX) for tumor targeting enhancement in radiotherapy. Methods: The GEPTS Monte Carlo code is used to generate dose distributions from 18MV Varian photon beam and generic high-energy (45-MV) photon and (45-MeV) electron beams in a voxel-based tissueequivalent phantom. In addition to calculating the ionization dose, the code scores Cerenkov energy released in the wavelength range 375–425 nm corresponding to the pick of the PpIX absorption spectrum (Fig. 1) using the Frank-Tamm formula. Results: The simulations shows that the produced Cerenkov dose suitable for activating PpIX is 4000 to 5500 times lower than the overall radiation dose for all considered beams (18MV, 45 MV and 45 MeV). These results were contradictory to the recent experimental studies by Axelsson et al. (Med. Phys. 38 (2011) p 4127), where Cerenkov dose was reported to be only two orders of magnitude lower than the radiation dose. Note that our simulation results can be corroborated by a simple model where the Frank and Tamm formula is applied for electrons with 2 MeV/cm stopping power generating Cerenkov photons in the 375–425 nm range and assuming these photons have less than 1mm penetration in tissue. Conclusion: The Cerenkov dose generated by high-energy photon and electron beams may produce minimal clinical effect in comparison with the photon fluence (or dose) commonly used for photo-dynamic therapy. At the present time, it is unclear whether Cerenkov radiation is a significant contributor to the recently observed tumor regression for patients receiving radiotherapy and PpIX versus patients receiving radiotherapy only. The ongoing study will include animal experimentation and investigation of dose rate effects on PpIX response.
SU-E-T-238: Monte Carlo Estimation of Cerenkov Dose for Photo-Dynamic Radiotherapy
International Nuclear Information System (INIS)
Purpose: Estimation of Cerenkov dose from high-energy megavoltage photon and electron beams in tissue and its impact on the radiosensitization using Protoporphyrine IX (PpIX) for tumor targeting enhancement in radiotherapy. Methods: The GEPTS Monte Carlo code is used to generate dose distributions from 18MV Varian photon beam and generic high-energy (45-MV) photon and (45-MeV) electron beams in a voxel-based tissueequivalent phantom. In addition to calculating the ionization dose, the code scores Cerenkov energy released in the wavelength range 375–425 nm corresponding to the pick of the PpIX absorption spectrum (Fig. 1) using the Frank-Tamm formula. Results: The simulations shows that the produced Cerenkov dose suitable for activating PpIX is 4000 to 5500 times lower than the overall radiation dose for all considered beams (18MV, 45 MV and 45 MeV). These results were contradictory to the recent experimental studies by Axelsson et al. (Med. Phys. 38 (2011) p 4127), where Cerenkov dose was reported to be only two orders of magnitude lower than the radiation dose. Note that our simulation results can be corroborated by a simple model where the Frank and Tamm formula is applied for electrons with 2 MeV/cm stopping power generating Cerenkov photons in the 375–425 nm range and assuming these photons have less than 1mm penetration in tissue. Conclusion: The Cerenkov dose generated by high-energy photon and electron beams may produce minimal clinical effect in comparison with the photon fluence (or dose) commonly used for photo-dynamic therapy. At the present time, it is unclear whether Cerenkov radiation is a significant contributor to the recently observed tumor regression for patients receiving radiotherapy and PpIX versus patients receiving radiotherapy only. The ongoing study will include animal experimentation and investigation of dose rate effects on PpIX response
SU-E-T-358: Monte Carlo Dose Calculation of Small Field Electron Beams
International Nuclear Information System (INIS)
Purpose: Dynamic radiotherapy involving electron beams such as Dynamic Electron Arc Radiotherapy (DEAR) requires accurate dose modelling of small field sizes, similar to the requirement of IMRT field on the small photon field. The current commercial electron Monte Carlo algorithms such as eMC v11 in Eclipse were developed for standard field sizes and do not support the planning of dynamic therapy yet. The purpose of this study is to develop a method to accurately model small field electron beam dosimetry using Monte Carlo simulations. Methods: Comparison between eMC, phantom measurements (diode), and Monte Carlo (MC) simulations (BEAMnrc/DOSYZnrc) were performed for a Varian TrueBeam linac. MC simulations utilized Varian TrueBeam phase space files which had been validated in another study. Static single small field was assessed by comparing dose distributions in water for a 16 MeV beam for circular (2 cm diameter) and rectangular (1×10 cm2) cut-out. MC was performed with a resolution of 2.5×2.5×2 mm2 and statistical uncertainty < 4%. The dose distribution was averaged over adjacent bins to improve precision. Depth dose and orthogonal profiles were evaluated. Results: Small field PDDs differ from those with standard cones. For both circular and rectangular cutouts, the difference in range R8 0-R1 0 is less than 2 mm and in dose within 2%. For the orthogonal profiles, field size and penumbra differences were within 1 mm at depth of maximum dose. The eMC displayed a distinctive “step” in the out-field dose profile in disagreement with both measurement and MC results and needs further investigation. Conclusion: MC was able to characterize the small field dosimetry with good agreement with the measurement data, and thus offers the opportunity for treatment planning of dynamic radiotherapy. Analyses for all other electron energies and cut-out sizes are under way and results will be included in the presentation
Monte Carlo calculation of synchrotron x-ray beam dose profiles in a lung phantom
International Nuclear Information System (INIS)
Full text: Recent advances in synchrotron generated X-ray beams with high fluence rate permit investigation of the application of an array of closely spaced, parallel or converging microplanar beams in radiotherapy. The proposed technique takes advantage of the hypothesized repair mechanism of capillary cells between alternate microbeam zones, which regenerates the lethally irradiated endothelial cells. The lateral and depth doses of 100 keV microplanar beams are investigated for different beam dimensions and spacings in a tissue, lung and tissue/lung/tissue phantom. The EGS4 Monte Carlo code is used to calculate dose profiles at different depth and bundles of beams (up to 20x20 cm square cross section). The maximum dose on the beam axis (peak) and the minimum interbeam dose (valley) are compared at different depths, bundles, heights, widths and beam spacings. Relatively high peak to valley ratios are observed in the lung region, suggesting an ideal environment for microbeam radiotherapy. For a single field, the ratio at the tissue/lung interface will set the maximum dose to the target volume. However, in clinical application, several fields would be involved allowing much greater doses to be applied for the elimination of cancer cells. We conclude therefore that multifield microbeam therapy has the potential to achieve useful therapeutic ratios for the treatment of lung cancer
Postimplant Dosimetry Using a Monte Carlo Dose Calculation Engine: A New Clinical Standard
International Nuclear Information System (INIS)
Purpose: To use the Monte Carlo (MC) method as a dose calculation engine for postimplant dosimetry. To compare the results with clinically approved data for a sample of 28 patients. Two effects not taken into account by the clinical calculation, interseed attenuation and tissue composition, are being specifically investigated. Methods and Materials: An automated MC program was developed. The dose distributions were calculated for the target volume and organs at risk (OAR) for 28 patients. Additional MC techniques were developed to focus specifically on the interseed attenuation and tissue effects. Results: For the clinical target volume (CTV) D90 parameter, the mean difference between the clinical technique and the complete MC method is 10.7 Gy, with cases reaching up to 17 Gy. For all cases, the clinical technique overestimates the deposited dose in the CTV. This overestimation is mainly from a combination of two effects: the interseed attenuation (average, 6.8 Gy) and tissue composition (average, 4.1 Gy). The deposited dose in the OARs is also overestimated in the clinical calculation. Conclusions: The clinical technique systematically overestimates the deposited dose in the prostate and in the OARs. To reduce this systematic inaccuracy, the MC method should be considered in establishing a new standard for clinical postimplant dosimetry and dose-outcome studies in a near future
Monte Carlo simulation of red bone marrow dose from CT examination
International Nuclear Information System (INIS)
To evaluate the methods of calculating red bone marrow dose from CT scan, simulating red bone marrow do ses from different CT scan protocols using different energy can provide the basic dose data for patient radiation protection. Method: Monte Carlo software MCNPX and RPI voxel phantom were used for the simulation, by mass absorption coefficient (MEAC) method, energy including 80 kV, 100 kV, 120 kV and 140 kV of the CT device were simulated, and different CT protocols such as chest scan, abdomen scan and body scan were taken into consideration when simulating the red bone marrow dose (mGy/100 mAs). Results: Under the same other conditions, the larger beam energy caused larger red bone marrow dose, the results of 140 kV was two times larger than that of 80 kV for the same protocol; while under the same beam energy, the difference among different protocol was less than 10%. Conclusion: Under the same conditions, the red bone marrow dose from CT scan depends on beam energy (tube voltage) and total effective mAs; if the total effective mAs was constant, the influence of scan protocol to red bone marrow dose was not much. (authors)
International Nuclear Information System (INIS)
The conversion coefficients, H'(d,α)/φ, for monoenergetic positrons and positron-emitting radionuclides were calculated by using the user code UCICRPM of the Monte Carlo code EGS5 to estimate the radiation dose for medical staff involved in positron emission tomography examinations. From these coefficients, the dose equivalent rates per unit activity at 0.07 and 10 mm depths in a soft tissue for a straight-line source of 2-deoxy-2-[18F]fluoro-d-glucose (18F-FDG) were calculated by using the developed user code UCF18DOSE. The dose equivalent rates per unit activity at 0.07 and 10 mm depths were measured by using a personal dosemeter (DOSE 3) under the same conditions as those considered in the calculation. The calculated dose equivalent rates per unit activity at 0.07 and 10 mm depths were 0.116 and 0.0352 pSv min-1 Bq-1, respectively, at 20 cm from the 18F-FDG injection tube. (authors)
International Nuclear Information System (INIS)
After an accidental release of radionuclides to the inhabited environment the external gamma irradiation from deposited radioactivity contributes significantly to the radiation exposure of the population for extended periods. For evaluating this exposure pathway, three main model requirements are needed: (i) to calculate the air kerma value per photon emitted per unit source area, based on Monte Carlo (MC) simulations; (ii) to describe the distribution and dynamics of radionuclides on the diverse urban surfaces; and (iii) to combine all these elements in a relevant urban model to calculate the resulting doses according to the actual scenario. This paper provides an overview about the different approaches to calculate photon transport in urban areas and about several dose calculation codes published. Two types of Monte Carlo simulations are presented using the global and the local approaches of photon transport. Moreover, two different philosophies of the dose calculation, the 'location factor method' and a combination of relative contamination of surfaces with air kerma values are described. The main features of six codes (ECOSYS, EDEM2M, EXPURT, PARATI, TEMAS, URGENT) are highlighted together with a short model-model features intercomparison
Environmental dose rate assessment of ITER using the Monte Carlo method
Directory of Open Access Journals (Sweden)
Karimian Alireza
2014-01-01
Full Text Available Exposure to radiation is one of the main sources of risk to staff employed in reactor facilities. The staff of a tokamak is exposed to a wide range of neutrons and photons around the tokamak hall. The International Thermonuclear Experimental Reactor (ITER is a nuclear fusion engineering project and the most advanced experimental tokamak in the world. From the radiobiological point of view, ITER dose rates assessment is particularly important. The aim of this study is the assessment of the amount of radiation in ITER during its normal operation in a radial direction from the plasma chamber to the tokamak hall. To achieve this goal, the ITER system and its components were simulated by the Monte Carlo method using the MCNPX 2.6.0 code. Furthermore, the equivalent dose rates of some radiosensitive organs of the human body were calculated by using the medical internal radiation dose phantom. Our study is based on the deuterium-tritium plasma burning by 14.1 MeV neutron production and also photon radiation due to neutron activation. As our results show, the total equivalent dose rate on the outside of the bioshield wall of the tokamak hall is about 1 mSv per year, which is less than the annual occupational dose rate limit during the normal operation of ITER. Also, equivalent dose rates of radiosensitive organs have shown that the maximum dose rate belongs to the kidney. The data may help calculate how long the staff can stay in such an environment, before the equivalent dose rates reach the whole-body dose limits.
A Monte Carlo tool for raster-scanning particle therapy dose computation
Jelen, U.; Radon, M.; Santiago, A.; Wittig, A.; Ammazzalorso, F.
2014-03-01
Purpose of this work was to implement Monte Carlo (MC) dose computation in realistic patient geometries with raster-scanning, the most advanced ion beam delivery technique, combining magnetic beam deflection with energy variation. FLUKA, a Monte Carlo package well-established in particle therapy applications, was extended to simulate raster-scanning delivery with clinical data, unavailable as built-in feature. A new complex beam source, compatible with FLUKA public programming interface, was implemented in Fortran to model the specific properties of raster-scanning, i.e. delivery by means of multiple spot sources with variable spatial distributions, energies and numbers of particles. The source was plugged into the MC engine through the user hook system provided by FLUKA. Additionally, routines were provided to populate the beam source with treatment plan data, stored as DICOM RTPlan or TRiP98's RST format, enabling MC recomputation of clinical plans. Finally, facilities were integrated to read computerised tomography (CT) data into FLUKA. The tool was used to recompute two representative carbon ion treatment plans, a skull base and a prostate case, prepared with analytical dose calculation (TRiP98). Selected, clinically relevant issues influencing the dose distributions were investigated: (1) presence of positioning errors, (2) influence of fiducial markers and (3) variations in pencil beam width. Notable differences in modelling of these challenging situations were observed between the analytical and Monte Carlo results. In conclusion, a tool was developed, to support particle therapy research and treatment, when high precision MC calculations are required, e.g. in presence of severe density heterogeneities or in quality assurance procedures.
SU-E-I-28: Evaluating the Organ Dose From Computed Tomography Using Monte Carlo Calculations
International Nuclear Information System (INIS)
Purpose: To evaluate organ doses from computed tomography (CT) using Monte Carlo (MC) calculations. Methods: A Philips Brilliance CT scanner (64 slice) was simulated using the GMctdospp (IMPS, Germany) based on the EGSnrc user code. The X-ray spectra and a bowtie filter for MC simulations were determined to coincide with measurements of half-value layer (HVL) and off-center ratio (OCR) profile in air. The MC dose was calibrated from absorbed dose measurements using a Farmer chamber and a cylindrical water phantom. The dose distribution from CT was calculated using patient CT images and organ doses were evaluated from dose volume histograms. Results: The HVLs of Al at 80, 100, and 120 kV were 6.3, 7.7, and 8.7 mm, respectively. The calculated HVLs agreed with measurements within 0.3%. The calculated and measured OCR profiles agreed within 3%. For adult head scans (CTDIvol) =51.4 mGy), mean doses for brain stem, eye, and eye lens were 23.2, 34.2, and 37.6 mGy, respectively. For pediatric head scans (CTDIvol =35.6 mGy), mean doses for brain stem, eye, and eye lens were 19.3, 24.5, and 26.8 mGy, respectively. For adult chest scans (CTDIvol=19.0 mGy), mean doses for lung, heart, and spinal cord were 21.1, 22.0, and 15.5 mGy, respectively. For adult abdominal scans (CTDIvol=14.4 mGy), the mean doses for kidney, liver, pancreas, spleen, and spinal cord were 17.4, 16.5, 16.8, 16.8, and 13.1 mGy, respectively. For pediatric abdominal scans (CTDIvol=6.76 mGy), mean doses for kidney, liver, pancreas, spleen, and spinal cord were 8.24, 8.90, 8.17, 8.31, and 6.73 mGy, respectively. In head scan, organ doses were considerably different from CTDIvol values. Conclusion: MC dose distributions calculated by using patient CT images are useful to evaluate organ doses absorbed to individual patients
SU-E-I-28: Evaluating the Organ Dose From Computed Tomography Using Monte Carlo Calculations
Energy Technology Data Exchange (ETDEWEB)
Ono, T; Araki, F [Faculty of Life Sciences, Kumamoto University, Kumamoto (Japan)
2014-06-01
Purpose: To evaluate organ doses from computed tomography (CT) using Monte Carlo (MC) calculations. Methods: A Philips Brilliance CT scanner (64 slice) was simulated using the GMctdospp (IMPS, Germany) based on the EGSnrc user code. The X-ray spectra and a bowtie filter for MC simulations were determined to coincide with measurements of half-value layer (HVL) and off-center ratio (OCR) profile in air. The MC dose was calibrated from absorbed dose measurements using a Farmer chamber and a cylindrical water phantom. The dose distribution from CT was calculated using patient CT images and organ doses were evaluated from dose volume histograms. Results: The HVLs of Al at 80, 100, and 120 kV were 6.3, 7.7, and 8.7 mm, respectively. The calculated HVLs agreed with measurements within 0.3%. The calculated and measured OCR profiles agreed within 3%. For adult head scans (CTDIvol) =51.4 mGy), mean doses for brain stem, eye, and eye lens were 23.2, 34.2, and 37.6 mGy, respectively. For pediatric head scans (CTDIvol =35.6 mGy), mean doses for brain stem, eye, and eye lens were 19.3, 24.5, and 26.8 mGy, respectively. For adult chest scans (CTDIvol=19.0 mGy), mean doses for lung, heart, and spinal cord were 21.1, 22.0, and 15.5 mGy, respectively. For adult abdominal scans (CTDIvol=14.4 mGy), the mean doses for kidney, liver, pancreas, spleen, and spinal cord were 17.4, 16.5, 16.8, 16.8, and 13.1 mGy, respectively. For pediatric abdominal scans (CTDIvol=6.76 mGy), mean doses for kidney, liver, pancreas, spleen, and spinal cord were 8.24, 8.90, 8.17, 8.31, and 6.73 mGy, respectively. In head scan, organ doses were considerably different from CTDIvol values. Conclusion: MC dose distributions calculated by using patient CT images are useful to evaluate organ doses absorbed to individual patients.
Application of Monte Carlo method for dose calculation in thyroid follicle
International Nuclear Information System (INIS)
The Monte Carlo method is an important tool to simulate radioactive particles interaction with biologic medium. The principal advantage of the method when compared with deterministic methods is the ability to simulate a complex geometry. Several computational codes use the Monte Carlo method to simulate the particles transport and they have the capacity to simulate energy deposition in models of organs and/or tissues, as well models of cells of human body. Thus, the calculation of the absorbed dose to thyroid's follicles (compound of colloid and follicles' cells) have a fundamental importance to dosimetry, because these cells are radiosensitive due to ionizing radiation exposition, in particular, exposition due to radioisotopes of iodine, because a great amount of radioiodine may be released into the environment in case of a nuclear accidents. In this case, the goal of this work was use the code of particles transport MNCP4C to calculate absorbed doses in models of thyroid's follicles, for Auger electrons, internal conversion electrons and beta particles, by iodine-131 and short-lived iodines (131, 132, 133, 134 e 135), with diameters varying from 30 to 500 μm. The results obtained from simulation with the MCNP4C code shown an average percentage of the 25% of total absorbed dose by colloid to iodine- 131 and 75% to short-lived iodine's. For follicular cells, this percentage was of 13% to iodine-131 and 87% to short-lived iodine's. The contributions from particles with low energies, like Auger and internal conversion electrons should not be neglected, to assessment the absorbed dose in cellular level. Agglomerative hierarchical clustering was used to compare doses obtained by codes MCNP4C, EPOTRAN, EGS4 and by deterministic methods. (author)
Study of dose distribution in dental radiology using the Monte Carlo Simulation
International Nuclear Information System (INIS)
Full text: The purpose of this study was to study the absorbed dose in mouth due to scattering in teeth in dental radiography using the monte carlo simulation. The Electron Gamma Shower (EGS-4) system of computer codes was used, which is a general purpose package for monte carlo simulation of the coupled transport of electrons and photons in an arbitrary geometry for particles with energies above a few keV up to several TeV. In the case of a X ray dental the low energy photons beam, are removed of the spectrum by the filtration. These low energy photons beam do not contribute in the obtaining of the radiographic image, but they will be contribute in the dose to the patient, however when the incident radiation crosses the tooth it generates a scattering radiation that contributes in the dose received by the patient in the oral cavity (cheek, tooth and oral cavity). Dental radiography is one of the largest single groups of radiographic examination accounting for 32% of radiographs taken in the Brazil. A number of relatively recent improvements in technology, equipment and techniques have the potential to reduce patient radiation dose and improve image quality. To optimize radiation protection all reasonable means should be employed to minimize the dose of each exposure. Dentists therefore need to keep up to date with changes in techniques and equipment and modify their own practice. In preliminary analysis could be notice that the energy below the 30 keV (low energy) is deposited in the cheek. To 30 keV photons there is the maximum absorbed energy in the tooth (about 60%). In 40 keV could be notice that deposited energy is same to teeth and cheek, but up to 40 keV just a small part of energy is deposited, e.g., the great part of energy is transmitted to the inner mouth (oral cavity). (orig.)
Monte Carlo 20 and 45 MeV Bremsstrahlung and dose-reduction calculations
International Nuclear Information System (INIS)
The SANDYL electron-photon coupled Monte Carlo code has been compared with previously published experimental bremsstrahlung data at 20.9 MeV electron energy. The code was then used to calculate forward-directed spectra, angular distributions and dose-reduction factors for three practical configurations. These are: 20 MeV electrons incident on 1 mm of W + 59 mm of Be, 45 MeV electrons of 1 mm of W and 45 MeV electrons on 1 mm of W + 147 mm of Be. The application of these results to flash radiography is discussed. 7 references, 12 figures, 1 table
Dose measurement using radiochromic lms and Monte Carlo simulation for hadron-therapy
International Nuclear Information System (INIS)
Because of the increase in dose at the end of the range of ions, dose delivery during patient treatment with hadron-therapy should be controlled with high precision. Monte Carlo codes are now considered mandatory for validation of clinical treatment planning and as a new tool for dosimetry of ion beams. In this work, we aimed to calculate the absorbed dose using Monte Carlo simulation Geant4/Gate. The effect on the dose calculation accuracy of different Geant4 parameters has been studied for mono-energetic carbon ion beams of 300 MeV/u in water. The parameters are: the production threshold of secondary particles and the maximum step limiter of the particle track. Tolerated criterion were chosen to meet the precision required in radiotherapy in term of value and dose localisation (2%, 2 mm respectively) and to obtain the best compromise on dose distribution and computational time. We propose here the values of parameters in order to satisfy the precision required. In the second part of this work, we study the response of radiochromic films MD-v2-55 for quality control in proton and carbon ion beams. We have particularly observed and studied the quenching effect of dosimetric films for high LET (≥20 keV/μm) irradiation in homogeneous and heterogeneous media. This effect is due to the high ionization density around the track of the particle. We have developed a method to predict the response of radiochromic films taking into account the saturation effect. This model is called the RADIS model for 'Radiochromic films Dosimetry for Ions using Simulations'. It is based on the response of films under photon irradiations and the saturation of films due to high linear energy deposit calculated by Monte Carlo. Different beams were used in this study and aimed to validate the model for hadron-therapy applications: carbon ions, protons and photons at different energies. Experiments were performed at Grand Accelerateur National d'Ions Lourds (GANIL), Proton therapy center of
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources
International Nuclear Information System (INIS)
A novel phase-space source implementation has been designed for graphics processing unit (GPU)-based Monte Carlo dose calculation engines. Short of full simulation of the linac head, using a phase-space source is the most accurate method to model a clinical radiation beam in dose calculations. However, in GPU-based Monte Carlo dose calculations where the computation efficiency is very high, the time required to read and process a large phase-space file becomes comparable to the particle transport time. Moreover, due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel source implementation utilizing pre-processed patient-independent phase-spaces that are sorted by particle type, energy and position. Position bins located outside a rectangular region of interest enclosing the treatment field are ignored, substantially decreasing simulation time with little effect on the final dose distribution. The three methods were validated in absolute dose against BEAMnrc/DOSXYZnrc and compared using gamma-index tests (2%/2 mm above the 10% isodose). It was found that the PSL method has the optimal balance between accuracy and efficiency and thus is used as the default method in gDPM v3.0. Using the PSL method, open fields of 4 × 4, 10 × 10 and 30 × 30 cm
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources.
Townson, Reid W; Jia, Xun; Tian, Zhen; Graves, Yan Jiang; Zavgorodni, Sergei; Jiang, Steve B
2013-06-21
A novel phase-space source implementation has been designed for graphics processing unit (GPU)-based Monte Carlo dose calculation engines. Short of full simulation of the linac head, using a phase-space source is the most accurate method to model a clinical radiation beam in dose calculations. However, in GPU-based Monte Carlo dose calculations where the computation efficiency is very high, the time required to read and process a large phase-space file becomes comparable to the particle transport time. Moreover, due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel source implementation utilizing pre-processed patient-independent phase-spaces that are sorted by particle type, energy and position. Position bins located outside a rectangular region of interest enclosing the treatment field are ignored, substantially decreasing simulation time with little effect on the final dose distribution. The three methods were validated in absolute dose against BEAMnrc/DOSXYZnrc and compared using gamma-index tests (2%/2 mm above the 10% isodose). It was found that the PSL method has the optimal balance between accuracy and efficiency and thus is used as the default method in gDPM v3.0. Using the PSL method, open fields of 4 × 4, 10 × 10 and 30 × 30 cm
Monte Carlo Dosimetry of the 60Co BEBIG High Dose Rate for Brachytherapy.
Directory of Open Access Journals (Sweden)
Luciana Tourinho Campos
Full Text Available The use of high-dose-rate brachytherapy is currently a widespread practice worldwide. The most common isotope source is 192Ir, but 60Co is also becoming available for HDR. One of main advantages of 60Co compared to 192Ir is the economic and practical benefit because of its longer half-live, which is 5.27 years. Recently, Eckert & Ziegler BEBIG, Germany, introduced a new afterloading brachytherapy machine (MultiSource®; it has the option to use either the 60Co or 192Ir HDR source. The source for the Monte Carlo calculations is the new 60Co source (model Co0.A86, which is referred to as the new BEBIG 60Co HDR source and is a modified version of the 60Co source (model GK60M21, which is also from BEBIG.The purpose of this work is to obtain the dosimetry parameters in accordance with the AAPM TG-43U1 formalism with Monte Carlo calculations regarding the BEBIG 60Co high-dose-rate brachytherapy to investigate the required treatment-planning parameters. The geometric design and material details of the source was provided by the manufacturer and was used to define the Monte Carlo geometry. To validate the source geometry, a few dosimetry parameters had to be calculated according to the AAPM TG-43U1 formalism. The dosimetry studies included the calculation of the air kerma strength Sk, collision kerma in water along the transverse axis with an unbounded phantom, dose rate constant and radial dose function. The Monte Carlo code system that was used was EGSnrc with a new cavity code, which is a part of EGS++ that allows calculating the radial dose function around the source. The spectrum to simulate 60Co was composed of two photon energies, 1.17 and 1.33 MeV. Only the gamma part of the spectrum was used; the contribution of the electrons to the dose is negligible because of the full absorption by the stainless-steel wall around the metallic 60Co. The XCOM photon cross-section library was used in subsequent simulations, and the photoelectric effect, pair
Estimation of skyshine dose from turbine building of BWR plant using Monte Carlo code
International Nuclear Information System (INIS)
The Monte Carlo N-Particle transport code (MCNP) was adopted to calculate the skyshine dose from the turbine building of a BWR plant for obtaining precise estimations at the site boundary. In MCNP calculation, the equipment and piping arranged on the operating floor of the turbine building were considered and modeled in detail. The inner and outer walls of the turbine building, the shielding materials around the high-pressure turbine, and the piping connected from the moisture separator to the low-pressure turbine were all considered. A three-step study was conducted to estimate the applicability of MCNP code. The first step is confirming the propriety of calculation models. The atmospheric relief diaphragms, which are installed on top of the low-pressure turbine exhaust hood, are not considered in the calculation model. There was little difference between the skyshine dose distributions that were considered when using and not using the atmospheric relief diaphragms. The calculated dose rates agreed well with the measurements taken around the turbine. The second step is estimating the dose rates on the outer roof surface of the turbine building. This calculation was made to confirm the dose distribution of gamma-rays on the turbine roof before being scattered into the air. The calculated dose rates agreed well with the measured data. The third step is making a final confirmation by comparing the calculations and measurements of skyshine dose rates around the turbine building. The source terms of the main steam system are based on the measured activity data of N-16 and C-15. As a conclusion, we were able to calculate reasonable skyshine dose rates by using MCNP code. (authors)
An improved Monte Carlo (MC) dose simulation for charged particle cancer therapy
Energy Technology Data Exchange (ETDEWEB)
Ying, C. K. [Advanced Medical and Dental Institute, AMDI, Universiti Sains Malaysia, Penang, Malaysia and School of Medical Sciences, Universiti Sains Malaysia, Kota Bharu (Malaysia); Kamil, W. A. [Advanced Medical and Dental Institute, AMDI, Universiti Sains Malaysia, Penang, Malaysia and Radiology Department, Hospital USM, Kota Bharu (Malaysia); Shuaib, I. L. [Advanced Medical and Dental Institute, AMDI, Universiti Sains Malaysia, Penang (Malaysia); Matsufuji, Naruhiro [Research Centre of Charged Particle Therapy, National Institute of Radiological Sciences, NIRS, Chiba (Japan)
2014-02-12
Heavy-particle therapy such as carbon ion therapy are more popular nowadays because of the nature characteristics of charged particle and almost no side effect to patients. An effective treatment is achieved with high precision of dose calculation, in this research work, Geant4 based Monte Carlo simulation method has been used to calculate the radiation transport and dose distribution. The simulation have the same setting with the treatment room in Heavy Ion Medical Accelerator, HIMAC. The carbon ion beam at the isocentric gantry nozzle for the therapeutic energy of 290 MeV/u was simulated, experimental work was carried out in National Institute of Radiological Sciences, NIRS, Chiba, Japan by using the HIMAC to confirm the accuracy and qualities dose distribution by MC methods. The Geant4 based simulated dose distribution were verified with measurements for Bragg peak and spread out Bragg peak (SOBP) respectively. The verification of results shows that the Bragg peak depth-dose and SOBP distributions in simulation has good agreement with measurements. In overall, the study showed that Geant4 based can be fully applied in the heavy-ion therapy field for simulation, further works need to be carry on to refine and improve the Geant4 MC simulations.
An improved Monte Carlo (MC) dose simulation for charged particle cancer therapy
Ying, C. K.; Kamil, W. A.; Shuaib, I. L.; Matsufuji, Naruhiro
2014-02-01
Heavy-particle therapy such as carbon ion therapy are more popular nowadays because of the nature characteristics of charged particle and almost no side effect to patients. An effective treatment is achieved with high precision of dose calculation, in this research work, Geant4 based Monte Carlo simulation method has been used to calculate the radiation transport and dose distribution. The simulation have the same setting with the treatment room in Heavy Ion Medical Accelerator, HIMAC. The carbon ion beam at the isocentric gantry nozzle for the therapeutic energy of 290 MeV/u was simulated, experimental work was carried out in National Institute of Radiological Sciences, NIRS, Chiba, Japan by using the HIMAC to confirm the accuracy and qualities dose distribution by MC methods. The Geant4 based simulated dose distribution were verified with measurements for Bragg peak and spread out Bragg peak (SOBP) respectively. The verification of results shows that the Bragg peak depth-dose and SOBP distributions in simulation has good agreement with measurements. In overall, the study showed that Geant4 based can be fully applied in the heavy-ion therapy field for simulation, further works need to be carry on to refine and improve the Geant4 MC simulations.
Monte Carlo simulations of absorbed dose in a mouse phantom from 18-fluorine compounds
International Nuclear Information System (INIS)
The purpose of this study was to calculate internal absorbed dose distribution in mice from preclinical small animal PET imaging procedures with fluorine-18 labeled compounds (18FDG, 18FLT, and fluoride ion). The GATE Monte Carlo software and a realistic, voxel-based mouse phantom that included a subcutaneous tumor were used to perform simulations. Discretized time-activity curves obtained from dynamic in vivo studies with each of the compounds were used to set the activity concentration in the simulations. For 18FDG, a realistic range of uptake ratios was considered for the heart and tumor. For each simulated time frame, the biodistribution of the radionuclide in the phantom was considered constant, and a sufficient number of decays were simulated to achieve low statistical uncertainty. Absorbed dose, which was scaled to take into account radioactive decay, integration with time, and changes in biological distribution was reported in mGy per MBq of administered activity for several organs and uptake scenarios. The mean absorbed dose ranged from a few mGy/MBq to hundreds of mGy/MBq. Major organs receive an absorbed dose in a range for which biological effects have been reported. The effects on a given investigation are hard to predict; however, investigators should be aware of potential perturbations especially when the studied organ receives high absorbed dose and when longitudinal imaging protocols are considered
Tsai, Hui-Yu; Lin, Yung-Chieh; Tyan, Yeu-Sheng
2014-11-01
The purpose of this study was to evaluate organ doses for individual patients undergoing interventional transcatheter arterial embolization (TAE) for hepatocellular carcinoma (HCC) using measurement-based Monte Carlo simulation and adaptive organ segmentation. Five patients were enrolled in this study after institutional ethical approval and informed consent. Gafchromic XR-RV3 films were used to measure entrance surface dose to reconstruct the nonuniform fluence distribution field as the input data in the Monte Carlo simulation. XR-RV3 films were used to measure entrance surface doses due to their lower energy dependence compared with that of XR-RV2 films. To calculate organ doses, each patient's three-dimensional dose distribution was incorporated into CT DICOM images with image segmentation using thresholding and k-means clustering. Organ doses for all patients were estimated. Our dose evaluation system not only evaluated entrance surface doses based on measurements, but also evaluated the 3D dose distribution within patients using simulations. When film measurements were unavailable, the peak skin dose (between 0.68 and 0.82 of a fraction of the cumulative dose) can be calculated from the cumulative dose obtained from TAE dose reports. Successful implementation of this dose evaluation system will aid radiologists and technologists in determining the actual dose distributions within patients undergoing TAE.
Characterization of 60Co dose distribution using BEAMnrc Monte Carlo code
International Nuclear Information System (INIS)
In this study BEAMnrc based on EGSnrc as Monte Carlo code has been used for modeling and simulating 60Co machine in radioisotope centre of Khartoum (RICK), Two fields size ( 5 cm x 5 cm and 35 cm x 35 cm), were been studied, to define the characterization of 60Co machine and to investigate the effect of increasing the surface to skin distance (SSD) on the 60Co machine properties, e.g.; beam profile and percentage depth dose (Pdd). For the narrow field size there is a small change observed in the curves representing beam profile and the percentage depth dose when increasing the distance by 5 cm, for the wide fi ld size there relatively clear different in curves. The study results been compared with other previous studies and clear consistence observed. (Author)
The use of Monte Carlo technique to optimize the dose distribution in total skin irradiation
Energy Technology Data Exchange (ETDEWEB)
Poli, M.E.R. E-mail: esmeraldapoli@hotmail.com; Pereira, S.A.; Yoriyaz, H
2001-06-01
Cutaneous T-cell lymphoma (mycosis fungoides) is an indolent disease with a low percentage of cure. Total skin irradiation using an electron beam has become an efficient treatment of mycosis fungoides with curative intention, with success in almost 40% of the patients. In this work, we propose the use of a Monte Carlo technique to simulate the dose distribution in the patients during total skin irradiation treatments. Use was made of MCNP-4B, a well known and established code used to simulate transport of electrons, photons and neutrons through matter, especially in the area of reactor physics, and also finding increasing utility in medical physics. The goal of our work is to simulate different angles between each beam with a fixed treatment distance in order to obtain a uniform dose distribution in the patient.
Monte Carlo calculations for doses in organs and tissues to oral radiography
International Nuclear Information System (INIS)
Using the MIRD 5 phantom and Monte Carlo technique, organ doses in patients undergoing external dental examination were calculated taking into account the different x-ray beam geometries and the various possible positions of x-ray source with regard to the head of the patient. It was necessary to introduce in the original computer program a new source description specific for dental examinations. To have a realistic evaluation of organ doses during dental examination it was necessary to introduce a new region in the phantom heat which characterizes the teeth and salivary glands. The attenuation of the x-ray beam by the lead shield of the radiographic film was also introduced in the calculation. (author)
Absorbed dose measurements in mammography using Monte Carlo method and ZrO2+PTFE dosemeters
International Nuclear Information System (INIS)
Mammography test is a central tool for breast cancer diagnostic. In addition, programs are conducted periodically to detect the asymptomatic women in certain age groups; these programs have shown a reduction on breast cancer mortality. Early detection of breast cancer is achieved through a mammography, which contrasts the glandular and adipose tissue with a probable calcification. The parameters used for mammography are based on the thickness and density of the breast, their values depend on the voltage, current, focal spot and anode-filter combination. To achieve an image clear and a minimum dose must be chosen appropriate irradiation conditions. Risk associated with mammography should not be ignored. This study was performed in the General Hospital No. 1 IMSS in Zacatecas. Was used a glucose phantom and measured air Kerma at the entrance of the breast that was calculated using Monte Carlo methods and ZrO2+PTFE thermoluminescent dosemeters, this calculation was completed with calculating the absorbed dose. (author)
GPU-based fast Monte Carlo simulation for radiotherapy dose calculation
Jia, Xun; Graves, Yan Jiang; Folkerts, Michael; Jiang, Steve B
2011-01-01
Monte Carlo (MC) simulation is commonly considered to be the most accurate dose calculation method in radiotherapy. However, its efficiency still requires improvement for many routine clinical applications. In this paper, we present our recent progress towards the development a GPU-based MC dose calculation package, gDPM v2.0. It utilizes the parallel computation ability of a GPU to achieve high efficiency, while maintaining the same particle transport physics as in the original DPM code and hence the same level of simulation accuracy. In GPU computing, divergence of execution paths between threads can considerably reduce the efficiency. Since photons and electrons undergo different physics and hence attain different execution paths, we use a simulation scheme where photon transport and electron transport are separated to partially relieve the thread divergence issue. High performance random number generator and hardware linear interpolation are also utilized. We have also developed various components to hand...
International Nuclear Information System (INIS)
This research thesis addresses the dosimetric control of radiotherapy treatments by using amorphous silicon digital portal imagery. In a first part, the author reports the analysis of the dosimetric abilities of the imager (iViewGT) which is used in the radiotherapy department. The stability of the imager response on a short and on a long term has been studied. A relationship between the image grey level and the dose has been established for a reference irradiation field. The influence of irradiation parameters on the grey level variation with respect to the dose has been assessed. The obtained results show the possibility to use this system for dosimetry provided that a precise calibration is performed while taking the most influencing irradiation parameters into account, i.e. photon beam nominal energy, field size, and patient thickness. The author reports the development of a Monte Carlo simulation to model the imager response. It models the accelerator head by a generalized source point. Space and energy distributions of photons are calculated. This modelling can also be applied to the calculation of dose distribution within a patient, or to study physical interactions in the accelerator head. Then, the author explores a new approach to dose portal image prediction within the frame of an in vivo dosimetric control. He computes the image transmitted through the patient by Monte Carlo simulation, and measures the portal image of the irradiation field without the patient. Validation experiments are reported, and problems to be solved are highlighted (computation time, improvement of the collimator simulation)
Development of a GPU-based Monte Carlo dose calculation code for coupled electron-photon transport
Jia, Xun; Sempau, Josep; Choi, Dongju; Majumdar, Amitava; Jiang, Steve B
2009-01-01
Monte Carlo simulation is the most accurate method for absorbed dose calculations in radiotherapy. Its efficiency still requires improvement for routine clinical applications, especially for online adaptive radiotherapy. In this paper, we report our recent development on a GPU-based Monte Carlo dose calculation code for coupled electron-photon transport. We have implemented the Dose Planning Method (DPM) Monte Carlo dose calculation package (Sempau et al, Phys. Med. Biol., 45(2000)2263-2291) on GPU architecture under CUDA platform. The implementation has been tested with respect to the original sequential DPM code on CPU in two cases. Our results demonstrate the adequate accuracy of the GPU implementation for both electron and photon beams in radiotherapy energy range. A speed up factor of 4.5 and 5.5 times have been observed for electron and photon testing cases, respectively, using an NVIDIA Tesla C1060 GPU card against a 2.27GHz Intel Xeon CPU processor .
Adaptation of GEANT4 to Monte Carlo dose calculations based on CT data
International Nuclear Information System (INIS)
The GEANT4 Monte Carlo code provides many powerful functions for conducting particle transport simulations with great reliability and flexibility. However, as a general purpose Monte Carlo code, not all the functions were specifically designed and fully optimized for applications in radiation therapy. One of the primary issues is the computational efficiency, which is especially critical when patient CT data have to be imported into the simulation model. In this paper we summarize the relevant aspects of the GEANT4 tracking and geometry algorithms and introduce our work on using the code to conduct dose calculations based on CT data. The emphasis is focused on modifications of the GEANT4 source code to meet the requirements for fast dose calculations. The major features include a quick voxel search algorithm, fast volume optimization, and the dynamic assignment of material density. These features are ready to be used for tracking the primary types of particles employed in radiation therapy such as photons, electrons, and heavy charged particles. Re-calculation of a proton therapy treatment plan generated by a commercial treatment planning program for a paranasal sinus case is presented as an example
GPU-based Monte Carlo radiotherapy dose calculation using phase-space sources
Townson, Reid; Tian, Zhen; Graves, Yan Jiang; Zavgorodni, Sergei; Jiang, Steve B
2013-01-01
A novel phase-space source implementation has been designed for GPU-based Monte Carlo dose calculation engines. Due to the parallelized nature of GPU hardware, it is essential to simultaneously transport particles of the same type and similar energies but separated spatially to yield a high efficiency. We present three methods for phase-space implementation that have been integrated into the most recent version of the GPU-based Monte Carlo radiotherapy dose calculation package gDPM v3.0. The first method is to sequentially read particles from a patient-dependent phase-space and sort them on-the-fly based on particle type and energy. The second method supplements this with a simple secondary collimator model and fluence map implementation so that patient-independent phase-space sources can be used. Finally, as the third method (called the phase-space-let, or PSL, method) we introduce a novel strategy to pre-process patient-independent phase-spaces and bin particles by type, energy and position. Position bins l...
International Nuclear Information System (INIS)
Purpose: Current procedure guidelines for whole body [18F]fluoro-2-deoxy-D-glucose (FDG)-positron emission tomography (PET) state that studies with visible dose extravasations should be rejected for quantification protocols. Our work is focused on the development and validation of methods for estimating extravasated doses in order to correct standard uptake value (SUV) values for this effect in clinical routine. Methods: One thousand three hundred sixty-seven consecutive whole body FDG-PET studies were visually inspected looking for extravasation cases. Two methods for estimating the extravasated dose were proposed and validated in different scenarios using Monte Carlo simulations. All visible extravasations were retrospectively evaluated using a manual ROI based method. In addition, the 50 patients with higher extravasated doses were also evaluated using a threshold-based method. Results: Simulation studies showed that the proposed methods for estimating extravasated doses allow us to compensate the impact of extravasations on SUV values with an error below 5%. The quantitative evaluation of patient studies revealed that paravenous injection is a relatively frequent effect (18%) with a small fraction of patients presenting considerable extravasations ranging from 1% to a maximum of 22% of the injected dose. A criterion based on the extravasated volume and maximum concentration was established in order to identify this fraction of patients that might be corrected for paravenous injection effect. Conclusions: The authors propose the use of a manual ROI based method for estimating the effectively administered FDG dose and then correct SUV quantification in those patients fulfilling the proposed criterion
Energy Technology Data Exchange (ETDEWEB)
Silva-Rodríguez, Jesús, E-mail: jesus.silva.rodriguez@sergas.es; Aguiar, Pablo, E-mail: pablo.aguiar.fernandez@sergas.es [Fundación Ramón Domínguez, Santiago de Compostela, Galicia (Spain); Servicio de Medicina Nuclear, Complexo Hospitalario Universidade de Santiago de Compostela (USC), 15782, Galicia (Spain); Grupo de Imaxe Molecular, Instituto de Investigación Sanitarias (IDIS), Santiago de Compostela, 15706, Galicia (Spain); Sánchez, Manuel; Mosquera, Javier; Luna-Vega, Víctor [Servicio de Radiofísica y Protección Radiológica, Complexo Hospitalario Universidade de Santiago de Compostela (USC), 15782, Galicia (Spain); Cortés, Julia; Garrido, Miguel [Servicio de Medicina Nuclear, Complexo Hospitalario Universitario de Santiago de Compostela, 15706, Galicia, Spain and Grupo de Imaxe Molecular, Instituto de Investigación Sanitarias (IDIS), Santiago de Compostela, 15706, Galicia (Spain); Pombar, Miguel [Servicio de Radiofísica y Protección Radiológica, Complexo Hospitalario Universitario de Santiago de Compostela, 15706, Galicia (Spain); Ruibal, Álvaro [Servicio de Medicina Nuclear, Complexo Hospitalario Universidade de Santiago de Compostela (USC), 15782, Galicia (Spain); Grupo de Imaxe Molecular, Instituto de Investigación Sanitarias (IDIS), Santiago de Compostela, 15706, Galicia (Spain); Fundación Tejerina, 28003, Madrid (Spain)
2014-05-15
Purpose: Current procedure guidelines for whole body [18F]fluoro-2-deoxy-D-glucose (FDG)-positron emission tomography (PET) state that studies with visible dose extravasations should be rejected for quantification protocols. Our work is focused on the development and validation of methods for estimating extravasated doses in order to correct standard uptake value (SUV) values for this effect in clinical routine. Methods: One thousand three hundred sixty-seven consecutive whole body FDG-PET studies were visually inspected looking for extravasation cases. Two methods for estimating the extravasated dose were proposed and validated in different scenarios using Monte Carlo simulations. All visible extravasations were retrospectively evaluated using a manual ROI based method. In addition, the 50 patients with higher extravasated doses were also evaluated using a threshold-based method. Results: Simulation studies showed that the proposed methods for estimating extravasated doses allow us to compensate the impact of extravasations on SUV values with an error below 5%. The quantitative evaluation of patient studies revealed that paravenous injection is a relatively frequent effect (18%) with a small fraction of patients presenting considerable extravasations ranging from 1% to a maximum of 22% of the injected dose. A criterion based on the extravasated volume and maximum concentration was established in order to identify this fraction of patients that might be corrected for paravenous injection effect. Conclusions: The authors propose the use of a manual ROI based method for estimating the effectively administered FDG dose and then correct SUV quantification in those patients fulfilling the proposed criterion.
International Nuclear Information System (INIS)
To evaluate the dosimetric differences between Superposition/Convolution (SC) and Monte Carlo (MC) calculated dose distributions for simultaneous integrated boost (SIB) prostate cancer intensity modulated radiotherapy (IMRT) compared to experimental (film) measurements and the implications for clinical treatments. Twenty-two prostate patients treated with an in-house SIB-IMRT protocol were selected. SC-based plans used for treatment were re-evaluated with EGS4-based MC calculations for treatment verification. Accuracy was evaluated with-respect-to film-based dosimetry. Comparisons used gamma (γ)-index, distance-to-agreement (DTA), and superimposed dose distributions. The treatment plans were also compared based on dose-volume indices and 3-D γ index for targets and critical structures. Flat-phantom comparisons demonstrated that the MC algorithm predicted measurements better than the SC algorithm. The average PTVprostate D98 agreement between SC and MC was 1.2% ± 1.1. For rectum, the average differences in SC and MC calculated D50 ranged from -3.6% to 3.4%. For small bowel, there were up to 30.2% ± 40.7 (range: 0.2%, 115%) differences between SC and MC calculated average D50 index. For femurs, the differences in average D50 reached up to 8.6% ± 3.6 (range: 1.2%, 14.5%). For PTVprostate and PTVnodes, the average gamma scores were >95.0%. MC agrees better with film measurements than SC. Although, on average, SC-calculated doses agreed with MC calculations within the targets within 2%, there were deviations up to 5% for some patient's treatment plans. For some patients, the magnitude of such deviations might decrease the intended target dose levels that are required for the treatment protocol, placing the patients in different dose levels that do not satisfy the protocol dose requirements
Dosimetry and dose planning in boron neutron capture therapy : Monte Carlo studies
Energy Technology Data Exchange (ETDEWEB)
Koivunoro, H.
2012-07-01
Boron neutron capture therapy (BNCT) is a biologically targeted radiotherapy modality. So far, 249 cancer patients have received BNCT at the Finnish Research Reactor 1 (FiR 1) in Finland. The effectiveness and safety of radiotherapy are dependent on the radiation dose delivered to the tumor and healthy tissues, and on the accuracy of the doses. At FiR 1, patient dose calculations are performed with the Monte Carlo (MC) -based treatmentplanning system (TPS), Simulation Environment for Radiotherapy Applications (SERA). Initially, BNCT was applied to head and neck cancer, brain tumors, and malignant melanoma. To evaluate the applicability of the new target tumors for BNCT, calculation dosimetry studies are needed. So far, clinical BNCT has been performed with the neutrons from a nuclear reactor, while an accelerator based neutron sources applicable for hospital operation would be preferable. In this thesis, BNCT patient dose calculation practice in Finland was evaluated against reference calculations and experimental data in several cases. Calculations with two TPSs applied in clinical BNCT were compared. The suitability of the deuterium-deuterium (DD) and deuterium-tritium (D-T) fusion reaction-based compact neutron sources for BNCT were evaluated. In addition, feasibility of BNCT for noninvasive liver tumor treatments was examined. The deviation between SERA and the reference calculations was within 4% in the phantoms studied and in a brain cancer patient model elsewhere, except on the phantom or skin surface, for the boron, nitrogen, and photon dose components. These dose components produce 99% of the tumor dose and > 90% of the healthy tissue dose at points of relevance for treatment at the FiR 1 facility. The reduced voxel cell size ({<=} 0.5 cm) in the SERA edit mesh improved calculation accuracy on the surface. The erratic biased fastneutron run option in SERA led to significant underestimation (up to 30-60%) of the fastneutron dose, while more accurate fast
Monte Carlo simulations of dose distributions with necrotic tumor targeted radioimmunotherapy
International Nuclear Information System (INIS)
Radio-resistant hypoxic tumor cells are significant contributors to the locoregional recurrences and distant metastases that mark failure of radiotherapy. Due to restricted tissue oxygenation, chronically hypoxic tumor cells frequently become necrotic and thus there is often an association between chronically hypoxic and necrotic tumor regions. This simulation study is the first in a series to determine the feasibility of hypoxic cell killing after first targeting adjacent areas of necrosis with either an α- or β-emitting radioimmunoconjugate. - Highlights: • A representative necrotic tumor geometry was created in the Geant4 Monte Carlo toolkit. • Custom designed particle tracking was performed allowing for separation of deposited doses from different decay particles. • Post-processing of the data included relative biological effectiveness of the different decay particles and effects of cell oxygenation. • Physical and equivalent doses resulting from 177Lu and 212Pb were compared by means of dose maps and dose profiles. • 212Pb appears to be a promising isotope for necrotic tumor targeted α-therapy and will be pursued in future in vivo studies
PCXMC, a Monte Carlo program for calculating patient doses in medical x-ray examinations
International Nuclear Information System (INIS)
PCXMC is a Monte Carlo program for calculating patients' organ doses and effective doses in medical x-ray examinations. The organs and tissues considered in the program are: active bone marrow, adrenals, brain, breasts, colon (upper and lower large intestine), extrathoracic airways, gall bladder, heart, kidneys, liver, lungs, lymph nodes, muscle, oesophagus, oral mucosa, ovaries, pancreas, prostate, salivary glands, skeleton, skin, small intestine, spleen, stomach, testicles, thymus, thyroid, urinary bladder and uterus. The program calculates the effective dose with both the present tissue weighting factors of ICRP Publication 103 (2007) and the old tissue weighting factors of ICRP Publication 60 (1991). The anatomical data are based on the mathematical hermaphrodite phantom models of Cristy and Eckerman (1987), which describe patients of six different ages: new-born, 1, 5, 10, 15-year-old and adult patients. Some changes are made to these phantoms in order to make them more realistic for external irradiation conditions and to enable the calculation of the effective dose according to the new ICRP Publication 103 tissue weighting factors. The phantom sizes are adjustable to mimic patients of an arbitrary weight and height. PCXMC allows a free adjustment of the x-ray beam projection and other examination conditions of projection radiography and fluoroscopy
International Nuclear Information System (INIS)
Lung brachytherapy using high-dose rate 192Ir technique is a well-established technique of radiation therapy. However, many commercial treatment planning systems do not have the ability to consider the inhomogeneity of lung in relation to normal tissue. Under such circumstances dose calculations for tissues and organs at risk close to the target are inaccurate. The purpose of the current study was to estimate the dose difference due to tissue inhomogeneity using the Monte Carlo simulation code MCNP-5. Results showed that there was a relative sub dosage by treatment planning systems calculations in neighbouring tissues around the radioactive source due to inhomogeneity ignorance. The presence of lung instead of normal tissue resulted in an increase in relative dose, which approached 8 % at 4-cm distance from the source. Additionally, the relative increase was small for the lung (2.1 %) and larger for organs at risk such as the heart (6.8 %) and bone marrow (7.6 %). (authors)
Feasibility Study of Neutron Dose for Real Time Image Guided Proton Therapy: A Monte Carlo Study
Kim, Jin Sung; Kim, Daehyun; Shin, EunHyuk; Chung, Kwangzoo; Cho, Sungkoo; Ahn, Sung Hwan; Ju, Sanggyu; Chung, Yoonsun; Jung, Sang Hoon; Han, Youngyih
2015-01-01
Two full rotating gantry with different nozzles (Multipurpose nozzle with MLC, Scanning Dedicated nozzle) with conventional cyclotron system is installed and under commissioning for various proton treatment options at Samsung Medical Center in Korea. The purpose of this study is to investigate neutron dose equivalent per therapeutic dose, H/D, to x-ray imaging equipment under various treatment conditions with monte carlo simulation. At first, we investigated H/D with the various modifications of the beam line devices (Scattering, Scanning, Multi-leaf collimator, Aperture, Compensator) at isocenter, 20, 40, 60 cm distance from isocenter and compared with other research groups. Next, we investigated the neutron dose at x-ray equipments used for real time imaging with various treatment conditions. Our investigation showed the 0.07 ~ 0.19 mSv/Gy at x-ray imaging equipments according to various treatment options and intestingly 50% neutron dose reduction effect of flat panel detector was observed due to multi- lea...
Monte Carlo simulation methods of determining red bone marrow dose from external radiation
International Nuclear Information System (INIS)
Objective: To provide evidence for a more reasonable method of determining red bone marrow dose by analyzing and comparing existing simulation methods. Methods: By utilizing Monte Carlo simulation software MCNPX, the absorbed doses of red hone marrow of Rensselaer Polytechnic Institute (RPI) adult female voxel phantom were calculated through 4 different methods: direct energy deposition.dose response function (DRF), King-Spiers factor method and mass-energy absorption coefficient (MEAC). The radiation sources were defined as infinite plate.sources with the energy ranging from 20 keV to 10 MeV, and 23 sources with different energies were simulated in total. The source was placed right next to the front of the RPI model to achieve a homogeneous anteroposterior radiation scenario. The results of different simulated photon energy sources through different methods were compared. Results: When the photon energy was lower than 100 key, the direct energy deposition method gave the highest result while the MEAC and King-Spiers factor methods showed more reasonable results. When the photon energy was higher than 150 keV taking into account of the higher absorption ability of red bone marrow at higher photon energy, the result of the King-Spiers factor method was larger than those of other methods. Conclusions: The King-Spiers factor method might be the most reasonable method to estimate the red bone marrow dose from external radiation. (authors)
Fast CPU-based Monte Carlo simulation for radiotherapy dose calculation
Ziegenhein, Peter; Pirner, Sven; Kamerling, Cornelis Ph; Oelfke, Uwe
2015-08-01
Monte-Carlo (MC) simulations are considered to be the most accurate method for calculating dose distributions in radiotherapy. Its clinical application, however, still is limited by the long runtimes conventional implementations of MC algorithms require to deliver sufficiently accurate results on high resolution imaging data. In order to overcome this obstacle we developed the software-package PhiMC, which is capable of computing precise dose distributions in a sub-minute time-frame by leveraging the potential of modern many- and multi-core CPU-based computers. PhiMC is based on the well verified dose planning method (DPM). We could demonstrate that PhiMC delivers dose distributions which are in excellent agreement to DPM. The multi-core implementation of PhiMC scales well between different computer architectures and achieves a speed-up of up to 37× compared to the original DPM code executed on a modern system. Furthermore, we could show that our CPU-based implementation on a modern workstation is between 1.25× and 1.95× faster than a well-known GPU implementation of the same simulation method on a NVIDIA Tesla C2050. Since CPUs work on several hundreds of GB RAM the typical GPU memory limitation does not apply for our implementation and high resolution clinical plans can be calculated.
Ding, George X.; Duggan, Dennis M.; Coffey, Charles W.; Shokrani, Parvaneh; Cygler, Joanna E.
2006-06-01
The purpose of this study is to present our experience of commissioning, testing and use of the first commercial macro Monte Carlo based dose calculation algorithm for electron beam treatment planning and to investigate new issues regarding dose reporting (dose-to-water versus dose-to-medium) as well as statistical uncertainties for the calculations arising when Monte Carlo based systems are used in patient dose calculations. All phantoms studied were obtained by CT scan. The calculated dose distributions and monitor units were validated against measurements with film and ionization chambers in phantoms containing two-dimensional (2D) and three-dimensional (3D) type low- and high-density inhomogeneities at different source-to-surface distances. Beam energies ranged from 6 to 18 MeV. New required experimental input data for commissioning are presented. The result of validation shows an excellent agreement between calculated and measured dose distributions. The calculated monitor units were within 2% of measured values except in the case of a 6 MeV beam and small cutout fields at extended SSDs (>110 cm). The investigation on the new issue of dose reporting demonstrates the differences up to 4% for lung and 12% for bone when 'dose-to-medium' is calculated and reported instead of 'dose-to-water' as done in a conventional system. The accuracy of the Monte Carlo calculation is shown to be clinically acceptable even for very complex 3D-type inhomogeneities. As Monte Carlo based treatment planning systems begin to enter clinical practice, new issues, such as dose reporting and statistical variations, may be clinically significant. Therefore it is imperative that a consistent approach to dose reporting is used.
International Nuclear Information System (INIS)
The purpose of this study is to present our experience of commissioning, testing and use of the first commercial macro Monte Carlo based dose calculation algorithm for electron beam treatment planning and to investigate new issues regarding dose reporting (dose-to-water versus dose-to-medium) as well as statistical uncertainties for the calculations arising when Monte Carlo based systems are used in patient dose calculations. All phantoms studied were obtained by CT scan. The calculated dose distributions and monitor units were validated against measurements with film and ionization chambers in phantoms containing two-dimensional (2D) and three-dimensional (3D) type low- and high-density inhomogeneities at different source-to-surface distances. Beam energies ranged from 6 to 18 MeV. New required experimental input data for commissioning are presented. The result of validation shows an excellent agreement between calculated and measured dose distributions. The calculated monitor units were within 2% of measured values except in the case of a 6 MeV beam and small cutout fields at extended SSDs (>110 cm). The investigation on the new issue of dose reporting demonstrates the differences up to 4% for lung and 12% for bone when 'dose-to-medium' is calculated and reported instead of 'dose-to-water' as done in a conventional system. The accuracy of the Monte Carlo calculation is shown to be clinically acceptable even for very complex 3D-type inhomogeneities. As Monte Carlo based treatment planning systems begin to enter clinical practice, new issues, such as dose reporting and statistical variations, may be clinically significant. Therefore it is imperative that a consistent approach to dose reporting is used
International Nuclear Information System (INIS)
International standards and guidelines for calibrating high-dose dosimetry systems to be used in industrial radiation processing recommend that dose-rate effects on dosimeters be evaluated under conditions of use. This is important when the irradiation relies on high-current electron accelerators, which usually provide very high dose-rates. However, most dosimeter calibration facilities use low-intensity gamma radiation or low-current electron accelerators, which deliver comparatively low dose-rates. Because of issues of thermal conductivity and response, portable calorimeters cannot be practically used with high-current accelerators, where product conveyor speeds under an electron beam can exceed several meters per second and the calorimeter is not suitable for use with product handling systems. As an alternative, Monte Carlo calculations can give theoretical estimates of the absorbed dose in materials with flat or complex configurations such that the results are independent of dose-rate. Monte Carlo results can then be compared to experimental dose determinations to see whether dose-rate effects in the dosimeters are significant. A Monte Carlo code has been used in this study to calculate the absorbed doses in alanine film dosimeters supported by flat sheets of plywood irradiated with electrons using incident energies extending from 1.0 MeV to 10 MeV with beam currents up to 30 mA. The same process conditions have been used for dose determinations with high-current electron beams using low dose-rate gamma calibrated alanine film dosimeters. The close agreement between these calculations and the dosimeter determinations indicates that the response of this type of dosimeter system is independent of the dose-rate, and provides assurance that Monte Carlo calculations can yield results with sufficient accuracy for many industrial applications
Energy Technology Data Exchange (ETDEWEB)
Dimitriadis, A; Gialousis, G; Karlatira, M; Karaiskos, P; Georgiou, E; Yakoumakis, E [Medical Physics Department, Medical School, University of Athens, 75 Mikras Asias Str., Goudi 11527, Athens (Greece); Makri, T; Papaodysseas, S, E-mail: anestisdim@yahoo.com [Radiological Imaging Department, Ag. Sofia Hospital, Lebadias and Thibon, Goudi 11527, Athens (Greece)
2011-01-21
Organ doses are important quantities in assessing the radiation risk. In the case of children, estimation of this risk is of particular concern due to their significant radiosensitivity and the greater health detriment. The purpose of this study is to estimate the organ doses to paediatric patients undergoing barium meal and micturating cystourethrography examinations by clinical measurements and Monte Carlo simulation. In clinical measurements, dose-area products (DAPs) were assessed during examination of 50 patients undergoing barium meal and 90 patients undergoing cystourethrography examinations, separated equally within three age categories: namely newborn, 1 year and 5 years old. Monte Carlo simulation of photon transport in male and female mathematical phantoms was applied using the MCNP5 code in order to estimate the equivalent organ doses. Regarding the micturating cystourethrography examinations, the organs receiving considerable amounts of radiation doses were the urinary bladder (1.87, 2.43 and 4.7 mSv, the first, second and third value in the parentheses corresponds to neonatal, 1 year old and 5 year old patients, respectively), the large intestines (1.54, 1.8, 3.1 mSv), the small intestines (1.34, 1.56, 2.78 mSv), the stomach (1.46, 1.02, 2.01 mSv) and the gall bladder (1.46, 1.66, 2.18 mSv), depending upon the age of the child. Organs receiving considerable amounts of radiation during barium meal examinations were the stomach (9.81, 9.92, 11.5 mSv), the gall bladder (3.05, 5.74, 7.15 mSv), the rib bones (9.82, 10.1, 11.1 mSv) and the pancreas (5.8, 5.93, 6.65 mSv), depending upon the age of the child. DAPs to organ/effective doses conversion factors were derived for each age and examination in order to be compared with other studies.
GPU-based fast Monte Carlo dose calculation for proton therapy
Jia, Xun; Schümann, Jan; Paganetti, Harald; Jiang, Steve B.
2012-12-01
Accurate radiation dose calculation is essential for successful proton radiotherapy. Monte Carlo (MC) simulation is considered to be the most accurate method. However, the long computation time limits it from routine clinical applications. Recently, graphics processing units (GPUs) have been widely used to accelerate computationally intensive tasks in radiotherapy. We have developed a fast MC dose calculation package, gPMC, for proton dose calculation on a GPU. In gPMC, proton transport is modeled by the class II condensed history simulation scheme with a continuous slowing down approximation. Ionization, elastic and inelastic proton nucleus interactions are considered. Energy straggling and multiple scattering are modeled. Secondary electrons are not transported and their energies are locally deposited. After an inelastic nuclear interaction event, a variety of products are generated using an empirical model. Among them, charged nuclear fragments are terminated with energy locally deposited. Secondary protons are stored in a stack and transported after finishing transport of the primary protons, while secondary neutral particles are neglected. gPMC is implemented on the GPU under the CUDA platform. We have validated gPMC using the TOPAS/Geant4 MC code as the gold standard. For various cases including homogeneous and inhomogeneous phantoms as well as a patient case, good agreements between gPMC and TOPAS/Geant4 are observed. The gamma passing rate for the 2%/2 mm criterion is over 98.7% in the region with dose greater than 10% maximum dose in all cases, excluding low-density air regions. With gPMC it takes only 6-22 s to simulate 10 million source protons to achieve ˜1% relative statistical uncertainty, depending on the phantoms and energy. This is an extremely high efficiency compared to the computational time of tens of CPU hours for TOPAS/Geant4. Our fast GPU-based code can thus facilitate the routine use of MC dose calculation in proton therapy.
Jansen, Jan T. M.; Shrimpton, Paul C.
2016-07-01
The ImPACT (imaging performance assessment of CT scanners) CT patient dosimetry calculator is still used world-wide to estimate organ and effective doses (E) for computed tomography (CT) examinations, although the tool is based on Monte Carlo calculations reflecting practice in the early 1990’s. Subsequent developments in CT scanners, definitions of E, anthropomorphic phantoms, computers and radiation transport codes, have all fuelled an urgent need for updated organ dose conversion factors for contemporary CT. A new system for such simulations has been developed and satisfactorily tested. Benchmark comparisons of normalised organ doses presently derived for three old scanners (General Electric 9800, Philips Tomoscan LX and Siemens Somatom DRH) are within 5% of published values. Moreover, calculated normalised values of CT Dose Index for these scanners are in reasonable agreement (within measurement and computational uncertainties of ±6% and ±1%, respectively) with reported standard measurements. Organ dose coefficients calculated for a contemporary CT scanner (Siemens Somatom Sensation 16) demonstrate potential deviations by up to around 30% from the surrogate values presently assumed (through a scanner matching process) when using the ImPACT CT Dosimetry tool for newer scanners. Also, illustrative estimates of E for some typical examinations and a range of anthropomorphic phantoms demonstrate the significant differences (by some 10’s of percent) that can arise when changing from the previously adopted stylised mathematical phantom to the voxel phantoms presently recommended by the International Commission on Radiological Protection (ICRP), and when following the 2007 ICRP recommendations (updated from 1990) concerning tissue weighting factors. Further simulations with the validated dosimetry system will provide updated series of dose coefficients for a wide range of contemporary scanners.
Comparison between Acuros XB and Brainlab Monte Carlo algorithms for photon dose calculation
Energy Technology Data Exchange (ETDEWEB)
Misslbeck, M.; Kneschaurek, P. [Klinikum rechts der Isar der Technischen Univ. Muenchen (Germany). Klinik und Poliklinik fuer Strahlentherapie und Radiologische Onkologie
2012-07-15
Purpose: The Acuros {sup registered} XB dose calculation algorithm by Varian and the Monte Carlo algorithm XVMC by Brainlab were compared with each other and with the well-established AAA algorithm, which is also from Varian. Methods: First, square fields to two different artificial phantoms were applied: (1) a 'slab phantom' with a 3 cm water layer, followed by a 2 cm bone layer, a 7 cm lung layer, and another 18 cm water layer and (2) a 'lung phantom' with water surrounding an eccentric lung block. For the slab phantom, depth-dose curves along central beam axis were compared. The lung phantom was used to compare profiles at depths of 6 and 14 cm. As clinical cases, the CTs of three different patients were used. The original AAA plans with all three algorithms using open fields were recalculated. Results: There were only minor differences between Acuros and XVMC in all artificial phantom depth doses and profiles; however, this was different for AAA, which had deviations of up to 13% in depth dose and a few percent for profiles in the lung phantom. These deviations did not translate into the clinical cases, where the dose-volume histograms of all algorithms were close to each other for open fields. Conclusion: Only within artificial phantoms with clearly separated layers of simulated tissue does AAA show differences at layer boundaries compared to XVMC or Acuros. In real patient CTs, these differences in the dose-volume histogram of the planning target volume were not observed. (orig.)
The use of Monte Carlo technique to optimize the dose distribution in total skin irradiation
International Nuclear Information System (INIS)
Cutaneous T-cell lymphoma (Mycosis Fungoides) is an indolent disease with a low percentage of cure. The total skin irradiation with electron beam became an efficient treatment of mycosis fungoides with curative intention, with success in almost 40% of the patients. The classical technique was developed in Stanford, where patients are irradiated in all longitudinal extension. Two angled fields of about 20 degrees are used, above and below the horizontal line, passing through from the waistline of the patient on foot, distant 3 meters from the focus. During the treatment, 4 or 6 pairs of these beams irradiate the patient; being rotated 90 or 60 degrees on his longitudinal axis, respectively. The energy normally used is about 3-7 MeV. The determination of the dose distribution in the patient can be made in many different ways: using three parallel-plate ionization chambers (Victoreen-Holt plus CNMC 206 electrometer) in the air, using dosimetric films and others. In the first case, at least three ionization chambers are necessary, which is unusual in Brazilian hospitals. In the second case, a lot of dosimetric films are necessary, which is too expensive. In this work, we propose the use of the Monte Carlo technique to simulate the dose distribution in the patients during Total Skin Irradiation treatments. The Monte Carlo code used is the MCNP4B, a well known and established code used to perform the transport of electrons, photons and neutrons through matter, especially in reactor physics, but with increasing utilization in medical physics in recent years. The goals of our work is to simulate different angles between each beam with a fixed treatment distance, which is characteristic of each treatment room, in order to obtain a uniform dose distribution in the patient. (author)
Evaluation of an electron Monte Carlo dose calculation algorithm for treatment planning.
Chamberland, Eve; Beaulieu, Luc; Lachance, Bernard
2015-01-01
The purpose of this study is to evaluate the accuracy of the electron Monte Carlo (eMC) dose calculation algorithm included in a commercial treatment planning system and compare its performance against an electron pencil beam algorithm. Several tests were performed to explore the system's behavior in simple geometries and in configurations encountered in clinical practice. The first series of tests were executed in a homogeneous water phantom, where experimental measurements and eMC-calculated dose distributions were compared for various combinations of energy and applicator. More specifically, we compared beam profiles and depth-dose curves at different source-to-surface distances (SSDs) and gantry angles, by using dose difference and distance to agreement. Also, we compared output factors, we studied the effects of algorithm input parameters, which are the random number generator seed, as well as the calculation grid size, and we performed a calculation time evaluation. Three different inhomogeneous solid phantoms were built, using high- and low-density materials inserts, to clinically simulate relevant heterogeneity conditions: a small air cylinder within a homogeneous phantom, a lung phantom, and a chest wall phantom. We also used an anthropomorphic phantom to perform comparison of eMC calculations to measurements. Finally, we proceeded with an evaluation of the eMC algorithm on a clinical case of nose cancer. In all mentioned cases, measurements, carried out by means of XV-2 films, radiographic films or EBT2 Gafchromic films. were used to compare eMC calculations with dose distributions obtained from an electron pencil beam algorithm. eMC calculations in the water phantom were accurate. Discrepancies for depth-dose curves and beam profiles were under 2.5% and 2 mm. Dose calculations with eMC for the small air cylinder and the lung phantom agreed within 2% and 4%, respectively. eMC calculations for the chest wall phantom and the anthropomorphic phantom also
International Nuclear Information System (INIS)
The aim of this study was to validate the computed tomography dose index (CTDI) and organ doses evaluated by Monte Carlo simulations through comparisons with doses evaluated by in-phantom dosimetry. Organ doses were measured with radio-photoluminescence glass dosemeter (RGD) set at various organ positions within adult and 1-y-old anthropomorphic phantoms. For the dose simulations, the X-ray spectrum and bow-tie filter shape of a CT scanner were estimated and 3D voxelised data of the CTDI and anthropomorphic phantoms from the acquired CT images were derived. Organ dose simulations and measurements were performed with chest and abdomen -pelvis CT examination scan parameters. Relative differences between the simulated and measured doses were within 5 % for the volume CTDI and 13 % for organ doses for organs within the scan range in adult and paediatric CT examinations. The simulation results were considered to be in good agreement with the measured doses. (authors)
Fujii, K; Nomura, K; Muramatsu, Y; Takahashi, K; Obara, S; Akahane, K; Satake, M
2015-07-01
The aim of this study was to validate the computed tomography dose index (CTDI) and organ doses evaluated by Monte Carlo simulations through comparisons with doses evaluated by in-phantom dosimetry. Organ doses were measured with radio-photoluminescence glass dosemeter (RGD) set at various organ positions within adult and 1-y-old anthropomorphic phantoms. For the dose simulations, the X-ray spectrum and bow-tie filter shape of a CT scanner were estimated and 3D voxelised data of the CTDI and anthropomorphic phantoms from the acquired CT images were derived. Organ dose simulations and measurements were performed with chest and abdomen-pelvis CT examination scan parameters. Relative differences between the simulated and measured doses were within 5 % for the volume CTDI and 13 % for organ doses for organs within the scan range in adult and paediatric CT examinations. The simulation results were considered to be in good agreement with the measured doses. PMID:25848103
International Nuclear Information System (INIS)
To investigate the radiation dose to the fetus using retrospective tube current modulation (TCM) data selected from archived clinical records. This paper describes the calculation of fetal doses using retrospective TCM data and Monte Carlo (MC) simulations. Three TCM schemes were adopted for use with three pregnant patient phantoms. MC simulations were used to model CT scanners, TCM schemes and pregnant patients. Comparisons between organ doses from TCM schemes and those from non-TCM schemes show that these three TCM schemes reduced fetal doses by 14, 18 and 25 %, respectively. These organ doses were also compared with those from ImPACT calculation. It is found that the difference between the calculated fetal dose and the ImPACT reported dose is as high as 46 %. This work demonstrates methods to study organ doses from various TCM protocols and potential ways to improve the accuracy of CT dose calculation for pregnant patients. (authors)
International Nuclear Information System (INIS)
The treatment planning systems (T.P.S.) occupy a key position in the radiotherapy service: they realize the projected calculation of the dose distribution and the treatment duration. Traditionally, the quality control of the calculated distribution doses relies on their comparisons with dose distributions measured under the device of treatment. This thesis proposes to substitute these dosimetry measures to the profile of reference dosimetry calculations got by the Penelope Monte-Carlo code. The Monte-Carlo simulations give a broad choice of test configurations and allow to envisage a quality control of dosimetry aspects of T.P.S. without monopolizing the treatment devices. This quality control, based on the Monte-Carlo simulations has been tested on a clinical T.P.S. and has allowed to simplify the quality procedures of the T.P.S.. This quality control, in depth, more precise and simpler to implement could be generalized to every center of radiotherapy. (N.C.)
A virtual photon energy fluence model for Monte Carlo dose calculation
International Nuclear Information System (INIS)
The presented virtual energy fluence (VEF) model of the patient-independent part of the medical linear accelerator heads, consists of two Gaussian-shaped photon sources and one uniform electron source. The planar photon sources are located close to the bremsstrahlung target (primary source) and to the flattening filter (secondary source), respectively. The electron contamination source is located in the plane defining the lower end of the filter. The standard deviations or widths and the relative weights of each source are free parameters. Five other parameters correct for fluence variations, i.e., the horn or central depression effect. If these parameters and the field widths in the X and Y directions are given, the corresponding energy fluence distribution can be calculated analytically and compared to measured dose distributions in air. This provides a method of fitting the free parameters using the measurements for various square and rectangular fields and a fixed number of monitor units. The next step in generating the whole set of base data is to calculate monoenergetic central axis depth dose distributions in water which are used to derive the energy spectrum by deconvolving the measured depth dose curves. This spectrum is also corrected to take the off-axis softening into account. The VEF model is implemented together with geometry modules for the patient specific part of the treatment head (jaws, multileaf collimator) into the XVMC dose calculation engine. The implementation into other Monte Carlo codes is possible based on the information in this paper. Experiments are performed to verify the model by comparing measured and calculated dose distributions and output factors in water. It is demonstrated that open photon beams of linear accelerators from two different vendors are accurately simulated using the VEF model. The commissioning procedure of the VEF model is clinically feasible because it is based on standard measurements in air and water. It is
Dosimetric validation of Acuros XB with Monte Carlo methods for photon dose calculations
International Nuclear Information System (INIS)
Purpose: The dosimetric accuracy of the recently released Acuros XB advanced dose calculation algorithm (Varian Medical Systems, Palo Alto, CA) is investigated for single radiation fields incident on homogeneous and heterogeneous geometries, and a comparison is made to the analytical anisotropic algorithm (AAA). Methods: Ion chamber measurements for the 6 and 18 MV beams within a range of field sizes (from 4.0x4.0 to 30.0x30.0 cm2) are used to validate Acuros XB dose calculations within a unit density phantom. The dosimetric accuracy of Acuros XB in the presence of lung, low-density lung, air, and bone is determined using BEAMnrc/DOSXYZnrc calculations as a benchmark. Calculations using the AAA are included for reference to a current superposition/convolution standard. Results: Basic open field tests in a homogeneous phantom reveal an Acuros XB agreement with measurement to within ±1.9% in the inner field region for all field sizes and energies. Calculations on a heterogeneous interface phantom were found to agree with Monte Carlo calculations to within ±2.0%(σMC=0.8%) in lung (ρ=0.24 g cm-3) and within ±2.9%(σMC=0.8%) in low-density lung (ρ=0.1 g cm-3). In comparison, differences of up to 10.2% and 17.5% in lung and low-density lung were observed in the equivalent AAA calculations. Acuros XB dose calculations performed on a phantom containing an air cavity (ρ=0.001 g cm-3) were found to be within the range of ±1.5% to ±4.5% of the BEAMnrc/DOSXYZnrc calculated benchmark (σMC=0.8%) in the tissue above and below the air cavity. A comparison of Acuros XB dose calculations performed on a lung CT dataset with a BEAMnrc/DOSXYZnrc benchmark shows agreement within ±2%/2mm and indicates that the remaining differences are primarily a result of differences in physical material assignments within a CT dataset. Conclusions: By considering the fundamental particle interactions in matter based on theoretical interaction cross sections, the Acuros XB algorithm is
Koger, B.; Kirkby, C.
2016-03-01
Gold nanoparticles (GNPs) have shown potential in recent years as a means of therapeutic dose enhancement in radiation therapy. However, a major challenge in moving towards clinical implementation is the exact characterisation of the dose enhancement they provide. Monte Carlo studies attempt to explore this property, but they often face computational limitations when examining macroscopic scenarios. In this study, a method of converting dose from macroscopic simulations, where the medium is defined as a mixture containing both gold and tissue components, to a mean dose-to-tissue on a microscopic scale was established. Monte Carlo simulations were run for both explicitly-modeled GNPs in tissue and a homogeneous mixture of tissue and gold. A dose ratio was obtained for the conversion of dose scored in a mixture medium to dose-to-tissue in each case. Dose ratios varied from 0.69 to 1.04 for photon sources and 0.97 to 1.03 for electron sources. The dose ratio is highly dependent on the source energy as well as GNP diameter and concentration, though this effect is less pronounced for electron sources. By appropriately weighting the monoenergetic dose ratios obtained, the dose ratio for any arbitrary spectrum can be determined. This allows complex scenarios to be modeled accurately without explicitly simulating each individual GNP.
International Nuclear Information System (INIS)
The purpose of this study was to evaluate organ doses for individual patients undergoing interventional transcatheter arterial embolization (TAE) for hepatocellular carcinoma (HCC) using measurement-based Monte Carlo simulation and adaptive organ segmentation. Five patients were enrolled in this study after institutional ethical approval and informed consent. Gafchromic XR-RV3 films were used to measure entrance surface dose to reconstruct the nonuniform fluence distribution field as the input data in the Monte Carlo simulation. XR-RV3 films were used to measure entrance surface doses due to their lower energy dependence compared with that of XR-RV2 films. To calculate organ doses, each patient's three-dimensional dose distribution was incorporated into CT DICOM images with image segmentation using thresholding and k-means clustering. Organ doses for all patients were estimated. Our dose evaluation system not only evaluated entrance surface doses based on measurements, but also evaluated the 3D dose distribution within patients using simulations. When film measurements were unavailable, the peak skin dose (between 0.68 and 0.82 of a fraction of the cumulative dose) can be calculated from the cumulative dose obtained from TAE dose reports. Successful implementation of this dose evaluation system will aid radiologists and technologists in determining the actual dose distributions within patients undergoing TAE. - Highlights: • An organ dose evaluation system for interventional procedures was successfully established. • Response variation of XR-RV3 due to energy dependence was 2.7%. • Three-dimensional image segmentations utilize thresholding and k-means clustering. • Peak skin doses are the fractions from 0.68 to 0.82 of cumulated doses for interventional TAE of HCC
Monte Carlo dosimetric study of the medium dose rate CSM40 source
International Nuclear Information System (INIS)
The 137Cs medium dose rate (MDR) CSM40 source model (Eckert and Ziegler BEBIG, Germany) is in clinical use but no dosimetric dataset has been published. This study aims to obtain dosimetric data for the CSM40 source for its use in clinical practice as required by the American Association of Physicists in Medicine (AAPM) and the European Society for Radiotherapy and Oncology (ESTRO). Penelope2008 and Geant4 Monte Carlo codes were used to characterize this source dosimetrically. It was located in an unbounded water phantom with composition and mass density as recommended by AAPM and ESTRO. Due to the low photon energies of 137Cs, absorbed dose was approximated by collisional kerma. Additional simulations were performed to obtain the air-kerma strength, sK. Mass–energy absorption coefficients in water and air were consistently derived and used to calculate collisional kerma. Results performed with both radiation transport codes showed agreement typically within 0.05%. Dose rate constant, radial dose function and anisotropy function are provided for the CSM40 and compared with published data for other commercially available 137Cs sources. An uncertainty analysis has been performed. The data provided by this study can be used as input data and verification in the treatment planning systems. - Highlights: • A dosimetric dataset is obtained for the 137Cs medium dose rate CSM40 source model. • Along-away table and TG-43 formalism parameters and functions are derived as recommended by AAPM-ESTRO. • This can be used as input data and verification in the treatment planning systems used in clinical practice
Tian, Zhen; Folkerts, Michael; Shi, Feng; Jiang, Steve B; Jia, Xun
2015-01-01
Monte Carlo (MC) simulation is considered as the most accurate method for radiation dose calculations. Accuracy of a source model for a linear accelerator is critical for the overall dose calculation accuracy. In this paper, we presented an analytical source model that we recently developed for GPU-based MC dose calculations. A key concept called phase-space-ring (PSR) was proposed. It contained a group of particles that are of the same type and close in energy and radial distance to the center of the phase-space plane. The model parameterized probability densities of particle location, direction and energy for each primary photon PSR, scattered photon PSR and electron PSR. For a primary photon PSRs, the particle direction is assumed to be from the beam spot. A finite spot size is modeled with a 2D Gaussian distribution. For a scattered photon PSR, multiple Gaussian components were used to model the particle direction. The direction distribution of an electron PSRs was also modeled as a 2D Gaussian distributi...
Application of a Monte Carlo linac model in routine verifications of dose calculations
International Nuclear Information System (INIS)
The analysis of some parameters of interest in Radiotherapy Medical Physics based on an experimentally validated Monte Carlo model of an Elekta Precise lineal accelerator, was performed for 6 and 15 Mv photon beams. The simulations were performed using the EGSnrc code. As reference for simulations, the optimal beam parameters values (energy and FWHM) previously obtained were used. Deposited dose calculations in water phantoms were done, on typical complex geometries commonly are used in acceptance and quality control tests, such as irregular and asymmetric fields. Parameters such as MLC scatter, maximum opening or closing position, and the separation between them were analyzed from calculations in water. Similarly simulations were performed on phantoms obtained from CT studies of real patients, making comparisons of the dose distribution calculated with EGSnrc and the dose distribution obtained from the computerized treatment planning systems (TPS) used in routine clinical plans. All the results showed a great agreement with measurements, finding all of them within tolerance limits. These results allowed the possibility of using the developed model as a robust verification tool for validating calculations in very complex situation, where the accuracy of the available TPS could be questionable. (Author)
International Nuclear Information System (INIS)
The purpose of this work was to study and quantify the differences in dose distributions computed with some of the newest dose calculation algorithms available in commercial planning systems. The study was done for clinical cases originally calculated with pencil beam convolution (PBC) where large density inhomogeneities were present. Three other dose algorithms were used: a pencil beam like algorithm, the anisotropic analytic algorithm (AAA), a convolution superposition algorithm, collapsed cone convolution (CCC), and a Monte Carlo program, voxel Monte Carlo (VMC++). The dose calculation algorithms were compared under static field irradiations at 6 MV and 15 MV using multileaf collimators and hard wedges where necessary. Five clinical cases were studied: three lung and two breast cases. We found that, in terms of accuracy, the CCC algorithm performed better overall than AAA compared to VMC++, but AAA remains an attractive option for routine use in the clinic due to its short computation times. Dose differences between the different algorithms and VMC++ for the median value of the planning target volume (PTV) were typically 0.4% (range: 0.0 to 1.4%) in the lung and -1.3% (range: -2.1 to -0.6%) in the breast for the few cases we analysed. As expected, PTV coverage and dose homogeneity turned out to be more critical in the lung than in the breast cases with respect to the accuracy of the dose calculation. This was observed in the dose volume histograms obtained from the Monte Carlo simulations
International Nuclear Information System (INIS)
The purpose of this work was to study and quantify the differences in dose distributions computed with some of the newest dose calculation algorithms available in commercial planning systems. The study was done for clinical cases where large density inhomogeneities were present. Three dose algorithms were used: a pencil beam like algorithm, the anisotropic analytic algorithm (AAA), a convolution superposition algorithm, collapsed cone convolution (CCC) and a Monte Carlo program, voxel Monte Carlo (VMC++). The dose calculation algorithms were compared under static field irradiations at 6 MV and 15 MV using multileaf collimators and hard wedges where necessary. Five clinical cases were studied: three lung and two breast cases. We found that the CCC algorithm performed overall better than AAA compared to VMC++, but AAA remains an attractive option for routine use in the clinic due to its short computation times. Dose differences between the different algorithms for the median value of the planning target volume (PTV) were typically 0.4% (range: 0.0-1.4%) in the lung and -1.3% (range: -2.1--0.6%) in the breast for the few cases we analysed. As expected, PTV coverage and dose homogeneity turned out to be more critical in the lung than in the breast cases with respect to the accuracy of the dose calculation. This was observed in the dose volume histograms obtained from the Monte Carlo simulations
Zhang, Aizhen; Wen, Ning; Nurushev, Teamour; Burmeister, Jay; Chetty, Indrin J
2013-01-01
A commercial electron Monte Carlo (eMC) dose calculation algorithm has become available in Eclipse treatment planning system. The purpose of this work was to evaluate the eMC algorithm and investigate the clinical implementation of this system. The beam modeling of the eMC algorithm was performed for beam energies of 6, 9, 12, 16, and 20 MeV for a Varian Trilogy and all available applicator sizes in the Eclipse treatment planning system. The accuracy of the eMC algorithm was evaluated in a homogeneous water phantom, solid water phantoms containing lung and bone materials, and an anthropomorphic phantom. In addition, dose calculation accuracy was compared between pencil beam (PB) and eMC algorithms in the same treatment planning system for heterogeneous phantoms. The overall agreement between eMC calculations and measurements was within 3%/2 mm, while the PB algorithm had large errors (up to 25%) in predicting dose distributions in the presence of inhomogeneities such as bone and lung. The clinical implementation of the eMC algorithm was investigated by performing treatment planning for 15 patients with lesions in the head and neck, breast, chest wall, and sternum. The dose distributions were calculated using PB and eMC algorithms with no smoothing and all three levels of 3D Gaussian smoothing for comparison. Based on a routine electron beam therapy prescription method, the number of eMC calculated monitor units (MUs) was found to increase with increased 3D Gaussian smoothing levels. 3D Gaussian smoothing greatly improved the visual usability of dose distributions and produced better target coverage. Differences of calculated MUs and dose distributions between eMC and PB algorithms could be significant when oblique beam incidence, surface irregularities, and heterogeneous tissues were present in the treatment plans. In our patient cases, monitor unit differences of up to 7% were observed between PB and eMC algorithms. Monitor unit calculations were also preformed
Patient-specific Monte Carlo dose calculations for 103Pd breast brachytherapy
Miksys, N.; Cygler, J. E.; Caudrelier, J. M.; Thomson, R. M.
2016-04-01
This work retrospectively investigates patient-specific Monte Carlo (MC) dose calculations for 103Pd permanent implant breast brachytherapy, exploring various necessary assumptions for deriving virtual patient models: post-implant CT image metallic artifact reduction (MAR), tissue assignment schemes (TAS), and elemental tissue compositions. Three MAR methods (thresholding, 3D median filter, virtual sinogram) are applied to CT images; resulting images are compared to each other and to uncorrected images. Virtual patient models are then derived by application of different TAS ranging from TG-186 basic recommendations (mixed adipose and gland tissue at uniform literature-derived density) to detailed schemes (segmented adipose and gland with CT-derived densities). For detailed schemes, alternate mass density segmentation thresholds between adipose and gland are considered. Several literature-derived elemental compositions for adipose, gland and skin are compared. MC models derived from uncorrected CT images can yield large errors in dose calculations especially when used with detailed TAS. Differences in MAR method result in large differences in local doses when variations in CT number cause differences in tissue assignment. Between different MAR models (same TAS), PTV {{D}90} and skin {{D}1~\\text{c{{\\text{m}}3}}} each vary by up to 6%. Basic TAS (mixed adipose/gland tissue) generally yield higher dose metrics than detailed segmented schemes: PTV {{D}90} and skin {{D}1~\\text{c{{\\text{m}}3}}} are higher by up to 13% and 9% respectively. Employing alternate adipose, gland and skin elemental compositions can cause variations in PTV {{D}90} of up to 11% and skin {{D}1~\\text{c{{\\text{m}}3}}} of up to 30%. Overall, AAPM TG-43 overestimates dose to the PTV ({{D}90} on average 10% and up to 27%) and underestimates dose to the skin ({{D}1~\\text{c{{\\text{m}}3}}} on average 29% and up to 48%) compared to the various MC models derived using the post-MAR CT images studied
Monte Carlo dose calculation using a cell processor based PlayStation 3 system
International Nuclear Information System (INIS)
This study investigates the performance of the EGSnrc computer code coupled with a Cell-based hardware in Monte Carlo simulation of radiation dose in radiotherapy. Performance evaluations of two processor-intensive functions namely, HOWNEAR and RANMARGET in the EGSnrc code were carried out basing on the 20-80 rule (Pareto principle). The execution speeds of the two functions were measured by the profiler gprof specifying the number of executions and total time spent on the functions. A testing architecture designed for Cell processor was implemented in the evaluation using a PlayStation3 (PS3) system. The evaluation results show that the algorithms examined are readily parallelizable on the Cell platform, provided that an architectural change of the EGSnrc was made. However, as the EGSnrc performance was limited by the PowerPC Processing Element in the PS3, PC coupled with graphics processing units or GPCPU may provide a more viable avenue for acceleration.
Vishwanathan, S V N; Kondor, Imre Risi; Schraudolph, Nicol N
2008-01-01
We present a unified framework to study graph kernels, special cases of which include the random walk graph kernel \\citep{GaeFlaWro03,BorOngSchVisetal05}, marginalized graph kernel \\citep{KasTsuIno03,KasTsuIno04,MahUedAkuPeretal04}, and geometric kernel on graphs \\citep{Gaertner02}. Through extensions of linear algebra to Reproducing Kernel Hilbert Spaces (RKHS) and reduction to a Sylvester equation, we construct an algorithm that improves the time complexity of kernel computation from $O(n^6)$ to $O(n^3)$. When the graphs are sparse, conjugate gradient solvers or fixed-point iterations bring our algorithm into the sub-cubic domain. Experiments on graphs from bioinformatics and other application domains show that it is often more than a thousand times faster than previous approaches. We then explore connections between diffusion kernels \\citep{KonLaf02}, regularization on graphs \\citep{SmoKon03}, and graph kernels, and use these connections to propose new graph kernels. Finally, we show that rational kernels ...
International Nuclear Information System (INIS)
Isotope production and Application Division of Bhabha Atomic Research Center developed 32P patch sources for treatment of superficial tumors. Surface dose rate of a newly developed 32P patch source of nominal diameter 25 mm was measured experimentally using standard extrapolation ionization chamber and Gafchromic EBT film. Monte Carlo model of the 32P patch source along with the extrapolation chamber was also developed to estimate the surface dose rates from these sources. The surface dose rates to tissue (cGy/min) measured using extrapolation chamber and radiochromic films are 82.03±4.18 (k=2) and 79.13±2.53 (k=2) respectively. The two values of the surface dose rates measured using the two independent experimental methods are in good agreement to each other within a variation of 3.5%. The surface dose rate to tissue (cGy/min) estimated using the MCNP Monte Carlo code works out to be 77.78±1.16 (k=2). The maximum deviation between the surface dose rates to tissue obtained by Monte Carlo and the extrapolation chamber method is 5.2% whereas the difference between the surface dose rates obtained by radiochromic film measurement and the Monte Carlo simulation is 1.7%. The three values of the surface dose rates of the 32P patch source obtained by three independent methods are in good agreement to one another within the uncertainties associated with their measurements and calculation. This work has demonstrated that MCNP based electron transport simulations are accurate enough for determining the dosimetry parameters of the indigenously developed 32P patch sources for contact brachytherapy applications. - Highlights: • Surface dose rates of 25 mm nominal diameter newly developed 32P patch sources were measured experimentally using extrapolation chamber and Gafchromic EBT2 film. Monte Carlo model of the 32P patch source along with the extrapolation chamber was also developed. • The surface dose rates to tissue (cGy/min) measured using extrapolation chamber and
Longo, Mariaconcetta; Marchioni, Chiara; Insero, Teresa; Donnarumma, Raffaella; D'Adamo, Alessandro; Lucatelli, Pierleone; Fanelli, Fabrizio; Salvatori, Filippo Maria; Cannavale, Alessandro; Di Castro, Elisabetta
2016-03-01
This study evaluates X-ray exposure in patient undergoing abdominal extra-vascular interventional procedures by means of Digital Imaging and COmmunications in Medicine (DICOM) image headers and Monte Carlo simulation. The main aim was to assess the effective and equivalent doses, under the hypothesis of their correlation with the dose area product (DAP) measured during each examination. This allows to collect dosimetric information about each patient and to evaluate associated risks without resorting to in vivo dosimetry. The dose calculation was performed in 79 procedures through the Monte Carlo simulator PCXMC (A PC-based Monte Carlo program for calculating patient doses in medical X-ray examinations), by using the real geometrical and dosimetric irradiation conditions, automatically extracted from DICOM headers. The DAP measurements were also validated by using thermoluminescent dosemeters on an anthropomorphic phantom. The expected linear correlation between effective doses and DAP was confirmed with an R(2) of 0.974. Moreover, in order to easily calculate patient doses, conversion coefficients that relate equivalent doses to measurable quantities, such as DAP, were obtained. PMID:26211013
International Nuclear Information System (INIS)
This study evaluates X-ray exposure in patient undergoing abdominal extra-vascular interventional procedures by means of Digital Imaging and Communications in Medicine (DICOM) image headers and Monte Carlo simulation. The main aim was to assess the effective and equivalent doses, under the hypothesis of their correlation with the dose area product (DAP) measured during each examination. This allows to collect dosimetric information about each patient and to evaluate associated risks without resorting to in vivo dosimetry. The dose calculation was performed in 79 procedures through the Monte Carlo simulator PCXMC (A PC-based Monte Carlo program for calculating patient doses in medical X-ray examinations), by using the real geometrical and dosimetric irradiation conditions, automatically extracted from DICOM headers. The DAP measurements were also validated by using thermoluminescent dosemeters on an anthropomorphic phantom. The expected linear correlation between effective doses and DAP was confirmed with an R2 of 0.974. Moreover, in order to easily calculate patient doses, conversion coefficients that relate equivalent doses to measurable quantities, such as DAP, were obtained. (authors)
International Nuclear Information System (INIS)
Purpose: The objective of this work is to assess the sensitivity of Monte Carlo (MC) dose calculations to uncertainties in human tissue composition for a range of low photon energy brachytherapy sources: 125I, 103Pd, 131Cs, and an electronic brachytherapy source (EBS). The low energy photons emitted by these sources make the dosimetry sensitive to variations in tissue atomic number due to the dominance of the photoelectric effect. This work reports dose to a small mass of water in medium Dw,m as opposed to dose to a small mass of medium in medium Dm,m. Methods: Mean adipose, mammary gland, and breast tissues (as uniform mixture of the aforementioned tissues) are investigated as well as compositions corresponding to one standard deviation from the mean. Prostate mean compositions from three different literature sources are also investigated. Three sets of MC simulations are performed with the GEANT4 code: (1) Dose calculations for idealized TG-43-like spherical geometries using point sources. Radial dose profiles obtained in different media are compared to assess the influence of compositional uncertainties. (2) Dose calculations for four clinical prostate LDR brachytherapy permanent seed implants using 125I seeds (Model 2301, Best Medical, Springfield, VA). The effect of varying the prostate composition in the planning target volume (PTV) is investigated by comparing PTV D90 values. (3) Dose calculations for four clinical breast LDR brachytherapy permanent seed implants using 103Pd seeds (Model 2335, Best Medical). The effects of varying the adipose/gland ratio in the PTV and of varying the elemental composition of adipose and gland within one standard deviation of the assumed mean composition are investigated by comparing PTV D90 values. For (2) and (3), the influence of using the mass density from CT scans instead of unit mass density is also assessed. Results: Results from simulation (1) show that variations in the mean compositions of tissues affect low energy
Choi, Chang Heon; Jung, Seongmoon; Choi, Kanghyuk; Son, Kwang-Jae; Lee, Jun Sig; Ye, Sung-Joon
2016-04-01
This study aims to determine the activity of a sealed pure beta-source by measuring the surface dose rate using an extrapolation chamber. A conversion factor (cGy s-1 Bq-1), which was defined as the ratio of surface dose rate to activity, can be calculated by Monte Carlo simulations of the extrapolation chamber measurement. To validate this hypothesis the certified activities of two standard pure beta-sources of Sr/Y-90 and Si/P-32 were compared with those determined by this method. In addition, a sealed test source of Sr/Y-90 was manufactured by the HANARO reactor group of KAERI (Korea Atomic Energy Research Institute) and used to further validate this method. The measured surface dose rates of the Sr/Y-90 and Si/P-32 standard sources were 4.615×10-5 cGy s-1 and 2.259×10-5 cGy s-1, respectively. The calculated conversion factors of the two sources were 1.213×10-8 cGy s-1 Bq-1 and 1.071×10-8 cGy s-1 Bq-1, respectively. Therefore, the activity of the standard Sr/Y-90 source was determined to be 3.995 kBq, which was 2.0% less than the certified value (4.077 kBq). For Si/P-32 the determined activity was 2.102 kBq, which was 6.6% larger than the certified activity (1.971 kBq). The activity of the Sr/Y-90 test source was determined to be 4.166 kBq, while the apparent activity reported by KAERI was 5.803 kBq. This large difference might be due to evaporation and diffusion of the source liquid during preparation and uncertainty in the amount of weighed aliquot of source liquid. The overall uncertainty involved in this method was determined to be 7.3%. We demonstrated that the activity of a sealed pure beta-source could be conveniently determined by complementary combination of measuring the surface dose rate and Monte Carlo simulations.
International Nuclear Information System (INIS)
Proton beam radiotherapy is an effective and non-invasive treatment for uveal melanoma. Recent research efforts have focused on improving the dosimetric accuracy of treatment planning and overcoming the present limitation of relative analytical dose calculations. Monte Carlo algorithms have been shown to accurately predict dose per monitor unit (D/MU) values, but this has yet to be shown for analytical algorithms dedicated to ocular proton therapy, which are typically less computationally expensive than Monte Carlo algorithms. The objective of this study was to determine if an analytical method could predict absolute dose distributions and D/MU values for a variety of treatment fields like those used in ocular proton therapy. To accomplish this objective, we used a previously validated Monte Carlo model of an ocular nozzle to develop an analytical algorithm to predict three-dimensional distributions of D/MU values from pristine Bragg peaks and therapeutically useful spread-out Bragg peaks (SOBPs). Results demonstrated generally good agreement between the analytical and Monte Carlo absolute dose calculations. While agreement in the proximal region decreased for beams with less penetrating Bragg peaks compared with the open-beam condition, the difference was shown to be largely attributable to edge-scattered protons. A method for including this effect in any future analytical algorithm was proposed. Comparisons of D/MU values showed typical agreement to within 0.5%. We conclude that analytical algorithms can be employed to accurately predict absolute proton dose distributions delivered by an ocular nozzle.
Spatial fractionation of the dose using neon and heavier ions: A Monte Carlo study
Energy Technology Data Exchange (ETDEWEB)
Peucelle, C.; Martínez-Rovira, I.; Prezado, Y., E-mail: prezado@imnc.in2p3.fr [IMNC-UMR 8165, CNRS Paris 7 and Paris 11 Universities, 15 rue Georges Clemenceau, Orsay Cedex 91406 (France)
2015-10-15
Purpose: This work explores a new radiation therapy approach which might trigger a renewed use of neon and heavier ions to treat cancers. These ions were shown to be extremely efficient in radioresistant tumor killing. Unfortunately, the efficient region also extends into the normal tissue in front of the tumor. The strategy the authors propose is to profit from the well-established sparing effect of thin spatially fractionated beams, so that the impact on normal tissues might be minimized while a high tumor control is achieved. The main goal of this work is to provide a proof of concept of this new approach. With that aim, a dosimetric study was carried out as a first step to evaluate the interest of further explorations of this avenue. Methods: The GATE/GEANT4 v.6.1 Monte Carlo simulation platform was employed to simulate arrays of rectangular minibeams (700 μm × 2 cm) of four ions (Ne, Si, Ar, and Fe). The irradiations were performed with a 2 cm-long spread-out Bragg peak centered at 7 cm-depth. Dose distributions in a water phantom were scored considering two minibeams center-to-center distances: 1400 and 3500 μm. Peak and valley doses, peak-to-valley dose ratios (PVDRs), beam penumbras, and relative contribution of nuclear fragments and electromagnetic processes were assessed as figures of merit. In addition, the type and proportion of the secondary nuclear fragments were evaluated in both peak and valley regions. Results: Extremely high PVDR values (>100) and low valley doses were obtained. The higher the atomic number (Z) of the primary ion is, the lower the valleys and the narrower the penumbras. Although the yield of secondary nuclear products increases with Z, the actual dose being deposited by the secondary nuclear fragments in the valleys starts to be the dominant contribution at deeper points, helping in the sparing of proximal normal tissues. Additionally, a wider center-to-center distance leads to a minimized contribution of heavier secondary
Omar, Artur; Benmakhlouf, Hamza; Marteinsdottir, Maria; Bujila, Robert; Nowik, Patrik; Andreo, Pedro
2014-03-01
Complex interventional and diagnostic x-ray angiographic (XA) procedures may yield patient skin doses exceeding the threshold for radiation induced skin injuries. Skin dose is conventionally determined by converting the incident air kerma free-in-air into entrance surface air kerma, a process that requires the use of backscatter factors. Subsequently, the entrance surface air kerma is converted into skin kerma using mass energy-absorption coefficient ratios tissue-to-air, which for the photon energies used in XA is identical to the skin dose. The purpose of this work was to investigate how the cranial bone affects backscatter factors for the dosimetry of interventional neuroradiology procedures. The PENELOPE Monte Carlo system was used to calculate backscatter factors at the entrance surface of a spherical and a cubic water phantom that includes a cranial bone layer. The simulations were performed for different clinical x-ray spectra, field sizes, and thicknesses of the bone layer. The results show a reduction of up to 15% when a cranial bone layer is included in the simulations, compared with conventional backscatter factors calculated for a homogeneous water phantom. The reduction increases for thicker bone layers, softer incident beam qualities, and larger field sizes, indicating that, due to the increased photoelectric crosssection of cranial bone compared to water, the bone layer acts primarily as an absorber of low-energy photons. For neurointerventional radiology procedures, backscatter factors calculated at the entrance surface of a water phantom containing a cranial bone layer increase the accuracy of the skin dose determination.
Vishwanathan, S. V. N.; Borgwardt, Karsten M.; Kondor, Imre Risi; Schraudolph, Nicol N.
2010-01-01
We present a unified framework to study graph kernels, special cases of which include the random walk (Gärtner et al., 2003; Borgwardt et al., 2005) and marginalized (Kashima et al., 2003, 2004; Mahé et al., 2004) graph kernels. Through reduction to a Sylvester equation we improve the time complexity of kernel computation between unlabeled graphs with n vertices from O(n^6) to O(n^3). We find a spectral decomposition approach even more efficient when computing entire kernel matric...
International Nuclear Information System (INIS)
This study determined the influence of patient individuality on lung organ doses for chest CT examinations. The aim was a statistical statement on the variability as well as the uncertainty caused by the patient individuality. Furthermore, the reproducibility of the mean organ dose value of the lung using the new ICRP 110 voxelized adult female phantom was determined. Calculation of lung doses for 61 female chest CT studies with identical scan parameters (120 kV, 135 mAs, 100 mm collimation, 1.5 pitch) were done. For all patients, the lung was contoured and the geometry was simulated using the Monte Carlo method without patient table and with its original voxel size. The lungs were completely included in the scan area. A so-called user code CTDOSPP was developed which extends the Monte Carlo package EGSnrc and enables rotational simulation of CT X-ray sources. A developed graphical user interface GMctdospp allows easy handling of simulation parameters and CT studies, which are loaded in the DicomRT struct format. The transformation of CT values to material and density values is carried out with a standard relationship. The ICRP adult female material composition of all organs were directly taken from the publication. The patient table and bed and pillow were assumed to be air in order to be similar to patient pool. All simulations were calibrated for better handling and visualisation to a CTDIair value of 22.9 mGy. Simulation values were grouped into 1 mSv classes. The organ dose classes fit well to a Gaussian distribution (correlation coefficient R2 = 0.97). The fit's mean value is 10 mSv with a standard deviation of 2 mSv. The variability is about ± 30 % with minimum at 8 mSv and maximum at 13 mSv. The calculated organ dose to the lungs of the ICRP adult female phantom is about 11 mSv and thus within the calculated standard deviation of the patient pool. For all simulations the statistical uncertainty was between 2 and 3.5 %. This present study shows good
International Nuclear Information System (INIS)
The interpretation of data obtained from fixed, ground-based dose rate monitoring stations of environmental networks, in terms of deposited radionuclide activity per unit area, requires not only the knowledge of the nuclide spectrum and the deposition mechanism, but also the knowledge of the situation in the vicinity at the probe if it significantly differs from ideal conditions, which are defined to be an infinitely-extended plane surface. Distance-dependent calibration factors for different gamma-ray energies and depth profiles are calculated with the new Monte Carlo code PENELOPE. How these distance-dependent calibration factors can take inhomogeneous surface types in the vicinity of the probe into account will also be discussed. In addition, calibration factors for different detector heights and calibration factors for gamma sources in the air will also be calculated. The main irregularities that, in practice, occur in the vicinity of such probes are discussed. Their impact on the representativeness of the site is assessed. For some typical irregularities parameterized calibration factors are calculated and discussed. Sewage plants and sandboxes are discussed as further special examples. The application of these results to real sites allows for the improved interpretation of data and the quantitative assessment of the representativeness of the site. A semi-quantitative scoring scheme helps to decide to what extent irregularities can be tolerated. Its application is straightforward and provides a coarse but objective description of the site-specific conditions of a dose rate probe. (orig.)
Dose to drivers during drive-through cargo scanning using GEANT4 Monte Carlo simulation
Energy Technology Data Exchange (ETDEWEB)
Gomes, Rogerio S.; Gomes, Joana D' Arc R.L.; Costa, Mara Lucia L., E-mail: rogeriog@cnen.gov.br, E-mail: jlopes@cnen.gov.br, E-mail: mara@cnen.gov.br [Comissao Nacional de Energia Nuclear (CNEN-RJ), Rio de Janeiro, RJ (Brazil). Diretoria de Radioprotecao e Seguranca muclear; Miranda, Valeria F.E.S., E-mail: mvaleria@cnen.gov.br [Centro Regional de Ciencias Nucleares do Nordeste (CRCN-NE/CNEN-PE), Recife, PE (Brazil)
2013-07-01
The use of radiation technologies to perform screening for cargo containers has been increased due to security issues, mainly, as a consequence of the United States (US) legislation which requires, from 2013, the scanning of all intermodal cargo containers which arrive at US ports. Currently, systems to cargo inspections, using accelerator-driven high energy X-rays, between 4 and 9 MeV, are available for scanning operations. It is expected that, in the future, the use of these systems will be widely spread on roads, ports and airports in Brazil. However, in order to improve the productivity and reduce the costs of acquisition, operation and maintenance these systems require that the driver drives its vehicle through irradiation area, in a situation where members of the public (the truck drivers) enter in controlled area and are deliberately exposed to high-energy beam. Some manufacturers justifies this procedure arguing that the drivers are exposed briefly, and only to the scattered beam, since there are safety systems in order to avoid that the drivers are exposed to direct beam. In this work, it is presented the preliminary results of Monte Carlo simulations concerning the dose of drivers during scanning operations, including the dose due to a failure of safety system, producing an exposure of drivers to the direct beam, as well as, an analysis of the justification of practice, mainly related to the drive-through operational procedure. (author)
PCXMC. A PC-based Monte Carlo program for calculating patient doses in medical x-ray examinations
International Nuclear Information System (INIS)
The report describes PCXMC, a Monte Carlo program for calculating patients' organ doses and the effective dose in medical x-ray examinations. The organs considered are: the active bone marrow, adrenals, brain, breasts, colon (upper and lower large intestine), gall bladder, heats, kidneys, liver, lungs, muscle, oesophagus, ovaries, pancreas, skeleton, skin, small intestine, spleen, stomach, testes, thymes, thyroid, urinary bladder, and uterus. (42 refs.)
Kumar, Sudhir; Srinivasan, P; Sharma, S D; Saxena, Sanjay Kumar; Bakshi, A K; Dash, Ashutosh; Babu, D A R; Sharma, D N
2015-09-01
Isotope production and Application Division of Bhabha Atomic Research Center developed (32)P patch sources for treatment of superficial tumors. Surface dose rate of a newly developed (32)P patch source of nominal diameter 25 mm was measured experimentally using standard extrapolation ionization chamber and Gafchromic EBT film. Monte Carlo model of the (32)P patch source along with the extrapolation chamber was also developed to estimate the surface dose rates from these sources. The surface dose rates to tissue (cGy/min) measured using extrapolation chamber and radiochromic films are 82.03±4.18 (k=2) and 79.13±2.53 (k=2) respectively. The two values of the surface dose rates measured using the two independent experimental methods are in good agreement to each other within a variation of 3.5%. The surface dose rate to tissue (cGy/min) estimated using the MCNP Monte Carlo code works out to be 77.78±1.16 (k=2). The maximum deviation between the surface dose rates to tissue obtained by Monte Carlo and the extrapolation chamber method is 5.2% whereas the difference between the surface dose rates obtained by radiochromic film measurement and the Monte Carlo simulation is 1.7%. The three values of the surface dose rates of the (32)P patch source obtained by three independent methods are in good agreement to one another within the uncertainties associated with their measurements and calculation. This work has demonstrated that MCNP based electron transport simulations are accurate enough for determining the dosimetry parameters of the indigenously developed (32)P patch sources for contact brachytherapy applications. PMID:26086681
International Nuclear Information System (INIS)
MCNP and Monte Carlo method was used to calculate dose rate in the air-space of irradiation room at Hanoi Irradiation Center. Experiment measurements were also carried out to investigate the real distribution of dose field in air of the irradiator as well as the distribution of absorbed dose in sample product containers. The results show that there is a deviation between calculated data given by MCNP and measurements. The data of MCNP give a symmetric distribution of dose field against the axes going through the center of the source rack meanwhile the experiment data show that dose rate get higher values in the lower part of the space. Going to lower position to the floor dose rate getting higher value. This phenomenon was also occurred for the measurements of absorbed dose in sample product container. (author)
International Nuclear Information System (INIS)
There has been some evidence that cervical cancer patients who were treated by radiotherapy, had an increased incidence of second primary cancers noticeable 15 years or more after the radiotherapy. The data suggested that high dose pelvic irradiation was associated with increase in cancers of the bladder, kidneys, rectum, ovaries, corpus uteri, and non-Hodgkin's lymphoma but not leukemia (Kleinerman et al., 1982, Morton 1973). The aim of the present work is to estimate the absorbed dose, due to radiotherapy treatment for cervival cancer, to various organs and tissues in the body. Monte Carlo calculations were performed to calculate the organ absorbed doses resulting from intracavitary sources such as ovoids and applicators filled or loaded with radium, Co-60 and Cs-137. For that purpose a routine which simulates an internal source was constructed and added to the existing Monte Carlo code (GSF-Bericht S-885, Kramer et al.). Calculations were also made for external beam therapy. Various anterior, posterior and lateral fields were applied, resulting from megavoltage, Co-60 and Cs-137 therapy machines. The calculated organ doses are tabulated in three different ways: as organ dose per air Kerma in the reference field, according to the recommendations of the International Commission on Radiation Units and Measurements (ICRU Report No 38, 1985); as organ dose per surface dose and as organ dose per tissue dose at Point B. (orig.)
International Nuclear Information System (INIS)
A method for the calculation of the transit doses in HDR brachytherapy based on Monte Carlo simulations has been presented. The transit doses resulting from a linear implant with seven dwell positions is simulated by performing calculations at all positions in which, the moving 192Ir source, instantaneously, had its geometrical centre located exactly between two adjacent dwell positions. Discrete step sizes of 0.25 cm were used to calculate the dose rates and the total transit dose at any of the calculation points evaluated. By comparing this method to the results obtained from Sievert Integrals, we observed dose calculation errors ranging from 32 to 21% for the examples considered. The errors could be much higher for longer treatment lengths where contributions from points near the longitudinal axis of the source become more important. To date, the most accurate method of calculating doses in radiotherapy is by Monte Carlo Simulations but the long computational times associated with it renders its use in treatment planning impracticable. The Sievert Integral algorithms on the other hand are simple, versatile and very easy to use but its accuracy had been repeatedly put into question for low energy isotopes like iridium. We therefore advocate a modification of the Sievert Integral algorithms by superimposing the output from Monte Carlo Simulations on the Sievert Integrals when dealing with low energy isotopes. In this way, we would be combining accuracy, simplicity and reasonable computational times (author)
An analytic linear accelerator source model for GPU-based Monte Carlo dose calculations
Tian, Zhen; Li, Yongbao; Folkerts, Michael; Shi, Feng; Jiang, Steve B.; Jia, Xun
2015-10-01
Recently, there has been a lot of research interest in developing fast Monte Carlo (MC) dose calculation methods on graphics processing unit (GPU) platforms. A good linear accelerator (linac) source model is critical for both accuracy and efficiency considerations. In principle, an analytical source model should be more preferred for GPU-based MC dose engines than a phase-space file-based model, in that data loading and CPU-GPU data transfer can be avoided. In this paper, we presented an analytical field-independent source model specifically developed for GPU-based MC dose calculations, associated with a GPU-friendly sampling scheme. A key concept called phase-space-ring (PSR) was proposed. Each PSR contained a group of particles that were of the same type, close in energy and reside in a narrow ring on the phase-space plane located just above the upper jaws. The model parameterized the probability densities of particle location, direction and energy for each primary photon PSR, scattered photon PSR and electron PSR. Models of one 2D Gaussian distribution or multiple Gaussian components were employed to represent the particle direction distributions of these PSRs. A method was developed to analyze a reference phase-space file and derive corresponding model parameters. To efficiently use our model in MC dose calculations on GPU, we proposed a GPU-friendly sampling strategy, which ensured that the particles sampled and transported simultaneously are of the same type and close in energy to alleviate GPU thread divergences. To test the accuracy of our model, dose distributions of a set of open fields in a water phantom were calculated using our source model and compared to those calculated using the reference phase-space files. For the high dose gradient regions, the average distance-to-agreement (DTA) was within 1 mm and the maximum DTA within 2 mm. For relatively low dose gradient regions, the root-mean-square (RMS) dose difference was within 1.1% and the maximum
Liu, Han; Zhuang, Tingliang; Stephans, Kevin; Videtic, Gregory; Raithel, Stephen; Djemil, Toufik; Xia, Ping
2015-01-01
For patients with medically inoperable early-stage non-small cell lung cancer (NSCLC) treated with stereotactic body radiation therapy, early treatment plans were based on a simpler dose calculation algorithm, the pencil beam (PB) calculation. Because these patients had the longest treatment follow-up, identifying dose differences between the PB calculated dose and Monte Carlo calculated dose is clinically important for understanding of treatment outcomes. Previous studies found significant dose differences between the PB dose calculation and more accurate dose calculation algorithms, such as convolution-based or Monte Carlo (MC), mostly for three-dimensional conformal radiotherapy (3D CRT) plans. The aim of this study is to investigate whether these observed dose differences also exist for intensity-modulated radiotherapy (IMRT) plans for both centrally and peripherally located tumors. Seventy patients (35 central and 35 peripheral) were retrospectively selected for this study. The clinical IMRT plans that were initially calculated with the PB algorithm were recalculated with the MC algorithm. Among these paired plans, dosimetric parameters were compared for the targets and critical organs. When compared to MC calculation, PB calculation overestimated doses to the planning target volumes (PTVs) of central and peripheral tumors with different magnitudes. The doses to 95% of the central and peripheral PTVs were overestimated by 9.7% ± 5.6% and 12.0% ± 7.3%, respectively. This dose overestimation did not affect doses to the critical organs, such as the spinal cord and lung. In conclusion, for NSCLC treated with IMRT, dose differences between the PB and MC calculations were different from that of 3D CRT. No significant dose differences in critical organs were observed between the two calculations. PMID:26699560
A GPU-based Monte Carlo dose calculation code for photon transport in a voxel phantom
Energy Technology Data Exchange (ETDEWEB)
Bellezzo, M.; Do Nascimento, E.; Yoriyaz, H., E-mail: mbellezzo@gmail.br [Instituto de Pesquisas Energeticas e Nucleares / CNEN, Av. Lineu Prestes 2242, Cidade Universitaria, 05508-000 Sao Paulo (Brazil)
2014-08-15
As the most accurate method to estimate absorbed dose in radiotherapy, Monte Carlo method has been widely used in radiotherapy treatment planning. Nevertheless, its efficiency can be improved for clinical routine applications. In this paper, we present the CUBMC code, a GPU-based Mc photon transport algorithm for dose calculation under the Compute Unified Device Architecture platform. The simulation of physical events is based on the algorithm used in Penelope, and the cross section table used is the one generated by the Material routine, als present in Penelope code. Photons are transported in voxel-based geometries with different compositions. To demonstrate the capabilities of the algorithm developed in the present work four 128 x 128 x 128 voxel phantoms have been considered. One of them is composed by a homogeneous water-based media, the second is composed by bone, the third is composed by lung and the fourth is composed by a heterogeneous bone and vacuum geometry. Simulations were done considering a 6 MeV monoenergetic photon point source. There are two distinct approaches that were used for transport simulation. The first of them forces the photon to stop at every voxel frontier, the second one is the Woodcock method, where the photon stop in the frontier will be considered depending on the material changing across the photon travel line. Dose calculations using these methods are compared for validation with Penelope and MCNP5 codes. Speed-up factors are compared using a NVidia GTX 560-Ti GPU card against a 2.27 GHz Intel Xeon CPU processor. (Author)
A GPU-based Monte Carlo dose calculation code for photon transport in a voxel phantom
International Nuclear Information System (INIS)
As the most accurate method to estimate absorbed dose in radiotherapy, Monte Carlo method has been widely used in radiotherapy treatment planning. Nevertheless, its efficiency can be improved for clinical routine applications. In this paper, we present the CUBMC code, a GPU-based Mc photon transport algorithm for dose calculation under the Compute Unified Device Architecture platform. The simulation of physical events is based on the algorithm used in Penelope, and the cross section table used is the one generated by the Material routine, als present in Penelope code. Photons are transported in voxel-based geometries with different compositions. To demonstrate the capabilities of the algorithm developed in the present work four 128 x 128 x 128 voxel phantoms have been considered. One of them is composed by a homogeneous water-based media, the second is composed by bone, the third is composed by lung and the fourth is composed by a heterogeneous bone and vacuum geometry. Simulations were done considering a 6 MeV monoenergetic photon point source. There are two distinct approaches that were used for transport simulation. The first of them forces the photon to stop at every voxel frontier, the second one is the Woodcock method, where the photon stop in the frontier will be considered depending on the material changing across the photon travel line. Dose calculations using these methods are compared for validation with Penelope and MCNP5 codes. Speed-up factors are compared using a NVidia GTX 560-Ti GPU card against a 2.27 GHz Intel Xeon CPU processor. (Author)
International Nuclear Information System (INIS)
The analysis of depth-dose distributions in bricks sampled from walls in areas with nuclear waste or accident contamination has the potential of providing information on the energy and source configuration of the γ-radiation that had been incident on the brick. In this study, a brick from a mill facing a shallow water reservoir of the contaminated Techa river in the South Ural region is investigated. Thermoluminescene (TL) methods were used to measure the accumulated dose at several depths in the brick. The accidental external γ-dose is obtained by subtracting the natural radiation background dose from the total accumulated dose. In the first segment of the brick, at a depth of about 1.5 cm, the accident dose was found to be roughly 3.5 Gy. Monte-Carlo simulations of the photon transport form the reservoir bed contaminated with 137Cs were calculated for different depths in the brick. The calculations were made assuming different attenuating water levels. It is found that the depth-dose distribution determined by measurements corresponds to a water level between 20 and 50 cm. The results indicate that the TL measurements combined with Monte-Carlo modelling calculations are highly promising for external γ-dose reconstruction applications. (Author)
Dual-energy CT-based material extraction for tissue segmentation in Monte Carlo dose calculations
Bazalova, Magdalena; Carrier, Jean-François; Beaulieu, Luc; Verhaegen, Frank
2008-05-01
Monte Carlo (MC) dose calculations are performed on patient geometries derived from computed tomography (CT) images. For most available MC codes, the Hounsfield units (HU) in each voxel of a CT image have to be converted into mass density (ρ) and material type. This is typically done with a (HU; ρ) calibration curve which may lead to mis-assignment of media. In this work, an improved material segmentation using dual-energy CT-based material extraction is presented. For this purpose, the differences in extracted effective atomic numbers Z and the relative electron densities ρe of each voxel are used. Dual-energy CT material extraction based on parametrization of the linear attenuation coefficient for 17 tissue-equivalent inserts inside a solid water phantom was done. Scans of the phantom were acquired at 100 kVp and 140 kVp from which Z and ρe values of each insert were derived. The mean errors on Z and ρe extraction were 2.8% and 1.8%, respectively. Phantom dose calculations were performed for 250 kVp and 18 MV photon beams and an 18 MeV electron beam in the EGSnrc/DOSXYZnrc code. Two material assignments were used: the conventional (HU; ρ) and the novel (HU; ρ, Z) dual-energy CT tissue segmentation. The dose calculation errors using the conventional tissue segmentation were as high as 17% in a mis-assigned soft bone tissue-equivalent material for the 250 kVp photon beam. Similarly, the errors for the 18 MeV electron beam and the 18 MV photon beam were up to 6% and 3% in some mis-assigned media. The assignment of all tissue-equivalent inserts was accurate using the novel dual-energy CT material assignment. As a result, the dose calculation errors were below 1% in all beam arrangements. Comparable improvement in dose calculation accuracy is expected for human tissues. The dual-energy tissue segmentation offers a significantly higher accuracy compared to the conventional single-energy segmentation.
Effect of elemental compositions on Monte Carlo dose calculations in proton therapy of eye tumors
Rasouli, Fatemeh S.; Farhad Masoudi, S.; Keshazare, Shiva; Jette, David
2015-12-01
Recent studies in eye plaque brachytherapy have found considerable differences between the dosimetric results by using a water phantom, and a complete human eye model. Since the eye continues to be simulated as water-equivalent tissue in the proton therapy literature, a similar study for investigating such a difference in treating eye tumors by protons is indispensable. The present study inquires into this effect in proton therapy utilizing Monte Carlo simulations. A three-dimensional eye model with elemental compositions is simulated and used to examine the dose deposition to the phantom. The beam is planned to pass through a designed beam line to moderate the protons to the desired energies for ocular treatments. The results are compared with similar irradiation to a water phantom, as well as to a material with uniform density throughout the whole volume. Spread-out Bragg peaks (SOBPs) are created by adding pristine peaks to cover a typical tumor volume. Moreover, the corresponding beam parameters recommended by the ICRU are calculated, and the isodose curves are computed. The results show that the maximum dose deposited in ocular media is approximately 5-7% more than in the water phantom, and about 1-1.5% less than in the homogenized material of density 1.05 g cm-3. Furthermore, there is about a 0.2 mm shift in the Bragg peak due to the tissue composition difference between the models. It is found that using the weighted dose profiles optimized in a water phantom for the realistic eye model leads to a small disturbance of the SOBP plateau dose. In spite of the plaque brachytherapy results for treatment of eye tumors, it is found that the differences between the simplified models presented in this work, especially the phantom containing the homogenized material, are not clinically significant in proton therapy. Taking into account the intrinsic uncertainty of the patient dose calculation for protons, and practical problems corresponding to applying patient
International Nuclear Information System (INIS)
Purpose: The aim of the study was to evaluate the dosimetric impact of low-Z and high-Z metallic implants on IMRT plans. Methods: Computed tomography (CT) scans of three patients were analyzed to study effects due to the presence of Titanium (low-Z), Platinum and Gold (high-Z) inserts. To eliminate artifacts in CT images, a sinogram-based metal artifact reduction algorithm was applied. IMRT dose calculations were performed on both the uncorrected and corrected images using a commercial planning system (convolution/superposition algorithm) and an in-house Monte Carlo platform. Dose differences between uncorrected and corrected datasets were computed and analyzed using gamma index (Pγ) and setting 2 mm and 2% as distance to agreement and dose difference criteria, respectively. Beam specific depth dose profiles across the metal were also examined. Results: Dose discrepancies between corrected and uncorrected datasets were not significant for low-Z material. High-Z materials caused under-dosage of 20%–25% in the region surrounding the metal and over dosage of 10%–15% downstream of the hardware. Gamma index test yielded Pγ>99% for all low-Z cases; while for high-Z cases it returned 91% < 99%. Analysis of the depth dose curve of a single beam for low-Z cases revealed that, although the dose attenuation is altered inside the metal, it does not differ downstream of the insert. However, for high-Z metal implants the dose is increased up to 10%–12% around the insert. In addition, Monte Carlo method was more sensitive to the presence of metal inserts than superposition/convolution algorithm. Conclusions: The reduction in terms of dose of metal artifacts in CT images is relevant for high-Z implants. In this case, dose distribution should be calculated using Monte Carlo algorithms, given their superior accuracy in dose modeling in and around the metal. In addition, the knowledge of the composition of metal inserts improves the accuracy of the Monte Carlo dose
TH-A-19A-06: Site-Specific Comparison of Analytical and Monte Carlo Based Dose Calculations
Energy Technology Data Exchange (ETDEWEB)
Schuemann, J; Grassberger, C; Paganetti, H [Massachusetts General Hospital and Harvard Medical School, Boston, MA (United States); Dowdell, S [Illawarra Shoalhaven Local Health District, Wollongong (Australia)
2014-06-15
Purpose: To investigate the impact of complex patient geometries on the capability of analytical dose calculation algorithms to accurately predict dose distributions and to verify currently used uncertainty margins in proton therapy. Methods: Dose distributions predicted by an analytical pencilbeam algorithm were compared with Monte Carlo simulations (MCS) using TOPAS. 79 complete patient treatment plans were investigated for 7 disease sites (liver, prostate, breast, medulloblastoma spine and whole brain, lung and head and neck). A total of 508 individual passively scattered treatment fields were analyzed for field specific properties. Comparisons based on target coverage indices (EUD, D95, D90 and D50) were performed. Range differences were estimated for the distal position of the 90% dose level (R90) and the 50% dose level (R50). Two-dimensional distal dose surfaces were calculated and the root mean square differences (RMSD), average range difference (ARD) and average distal dose degradation (ADD), the distance between the distal position of the 80% and 20% dose levels (R80- R20), were analyzed. Results: We found target coverage indices calculated by TOPAS to generally be around 1–2% lower than predicted by the analytical algorithm. Differences in R90 predicted by TOPAS and the planning system can be larger than currently applied range margins in proton therapy for small regions distal to the target volume. We estimate new site-specific range margins (R90) for analytical dose calculations considering total range uncertainties and uncertainties from dose calculation alone based on the RMSD. Our results demonstrate that a reduction of currently used uncertainty margins is feasible for liver, prostate and whole brain fields even without introducing MC dose calculations. Conclusion: Analytical dose calculation algorithms predict dose distributions within clinical limits for more homogeneous patients sites (liver, prostate, whole brain). However, we recommend
Directory of Open Access Journals (Sweden)
Ahad Ollah Ezzati
2014-08-01
Full Text Available Introduction In this study, we aimed to calculate dose enhancement factor (DEF for gold (Au and iron (Fe nanoparticles (NPs in brachytherapy and teletherapy, using Monte Carlo (MC method. Materials and Methods In this study, a new algorithm was introduced to calculate dose enhancement by AuNPs and FeNPs for Iridium-192 (Ir-192 brachytherapy and Cobalt-60 (Co-60 teletherapy sources, using the MC method. In this algorithm, the semi-random distribution of NPs was used instead of the regular distribution. Diameters were assumed to be 15, 30, and 100 nm in brachytherapy and 15 and 30 nm in teletherapy. Monte Carlo MCNP4C code was used for simulations, and NP density values were 0.107 mg/ml and 0.112 mg/ml in brachytherapy and teletherapy, respectively. Results AuNPs significantly enhanced the radiation dose in brachytherapy (approximately 60%, and 100 nm diameter NPs showed the most uniform dose distribution. AuNPs had an insignificant effect on teletherapy radiation field, with a dose enhancement ratio of 3% (about the calculation uncertainty or less. In addition, FeNPs had an insignificant effect on both brachytherapy and teletherapy radiation fields. FeNPs dose enhancement was 3% in brachytherapy and 6% (about the calculation uncertainty or less in teletherapy. Conclusion It can be concluded that AuNPs can significantly increase the absorbed dose in brachytherapy; however, FeNPs do not have a noticeable effect on the absorbed dose
Kim, Sangroh; Yoshizumi, Terry T; Toncheva, Greta; Frush, Donald P; Yin, Fang-Fang
2010-03-01
The purpose of this study was to establish a dose estimation tool with Monte Carlo (MC) simulations. A 5-y-old paediatric anthropomorphic phantom was computed tomography (CT) scanned to create a voxelised phantom and used as an input for the abdominal cone-beam CT in a BEAMnrc/EGSnrc MC system. An X-ray tube model of the Varian On-Board Imager((R)) was built in the MC system. To validate the model, the absorbed doses at each organ location for standard-dose and low-dose modes were measured in the physical phantom with MOSFET detectors; effective doses were also calculated. In the results, the MC simulations were comparable to the MOSFET measurements. This voxelised phantom approach could produce a more accurate dose estimation than the stylised phantom method. This model can be easily applied to multi-detector CT dosimetry. PMID:19889800
International Nuclear Information System (INIS)
In order to apply the Monte Carlo simulation technique for usual radiological examinations we developed a Pc program, 'IradMed', written entirely in Java. The main purpose of this program is to compute the organ doses and the effective dose of patients, which are exposed at a X-ray beam having photon energies in 10 to 150 keV radiodiagnostic range. Three major radiological procedures are considered, namely mammography, radiography and CT. The fluoroscopy implies an irregular geometry and therefore it is neglected. Nevertheless, a gross estimation of patient doses can be made taking into account the fluoroscopy as being composed of several radiographic examinations applied in different anatomical regions. The interactions between radiation and matter are well-known, and the accuracy of the calculation is limited by the accuracy of the anatomical model used to describe actual patients and by characterisation of the radiation field applied. In this version of IradMed, it is assumed that the absorbed dose is equal with kerma for all tissues. No procedure has been used to take account of the finite range of the secondary electrons that are produced by photoelectric or Compton interactions. These ranges are small compared with the dimensions of the organs, and the absorbed dose will not change abruptly with distance except at boundary where composition and density change. However these boundary effects would have little effect in the determination of the average doses to almost all organs, except the active bone marrow which is treated separately. Another justification for this kerma approximation is the fact that the sum of all electron energies that exit the organ is statistically equal with the sum of all electron energies that enter in that particular organ. In this version of program, it is considered the following interactions: the Rayleigh scattering, the Compton scattering and the photoelectric effect. The Compton scattering is modeled by several methods which
Energy Technology Data Exchange (ETDEWEB)
Fulea, D. [Institute of Public Health ' Prof.Dr.Iuliu Moldovan' , Cluj-Napoca (Romania); Cosma, C. [Babes-Bolyai Univ., Faculty of Physics, Cluj-Napoca (Romania)
2006-07-01
In order to apply the Monte Carlo simulation technique for usual radiological examinations we developed a Pc program, 'IradMed', written entirely in Java. The main purpose of this program is to compute the organ doses and the effective dose of patients, which are exposed at a X-ray beam having photon energies in 10 to 150 keV radiodiagnostic range. Three major radiological procedures are considered, namely mammography, radiography and CT. The fluoroscopy implies an irregular geometry and therefore it is neglected. Nevertheless, a gross estimation of patient doses can be made taking into account the fluoroscopy as being composed of several radiographic examinations applied in different anatomical regions. The interactions between radiation and matter are well-known, and the accuracy of the calculation is limited by the accuracy of the anatomical model used to describe actual patients and by characterisation of the radiation field applied. In this version of IradMed, it is assumed that the absorbed dose is equal with kerma for all tissues. No procedure has been used to take account of the finite range of the secondary electrons that are produced by photoelectric or Compton interactions. These ranges are small compared with the dimensions of the organs, and the absorbed dose will not change abruptly with distance except at boundary where composition and density change. However these boundary effects would have little effect in the determination of the average doses to almost all organs, except the active bone marrow which is treated separately. Another justification for this kerma approximation is the fact that the sum of all electron energies that exit the organ is statistically equal with the sum of all electron energies that enter in that particular organ. In this version of program, it is considered the following interactions: the Rayleigh scattering, the Compton scattering and the photoelectric effect. The Compton scattering is modeled by several
Energy Technology Data Exchange (ETDEWEB)
Çatlı, Serap, E-mail: serapcatli@hotmail.com [Gazi University, Faculty of Sciences, 06500 Teknikokullar, Ankara (Turkey); Tanır, Güneş [Gazi University, Faculty of Sciences, 06500 Teknikokullar, Ankara (Turkey)
2013-10-01
The present study aimed to investigate the effects of titanium, titanium alloy, and stainless steel hip prostheses on dose distribution based on the Monte Carlo simulation method, as well as the accuracy of the Eclipse treatment planning system (TPS) at 6 and 18 MV photon energies. In the present study the pencil beam convolution (PBC) method implemented in the Eclipse TPS was compared to the Monte Carlo method and ionization chamber measurements. The present findings show that if high-Z material is used in prosthesis, large dose changes can occur due to scattering. The variance in dose observed in the present study was dependent on material type, density, and atomic number, as well as photon energy; as photon energy increased back scattering decreased. The dose perturbation effect of hip prostheses was significant and could not be predicted accurately by the PBC method for hip prostheses. The findings show that for accurate dose calculation the Monte Carlo-based TPS should be used in patients with hip prostheses.
New approach based on tetrahedral-mesh geometry for accurate 4D Monte Carlo patient-dose calculation
International Nuclear Information System (INIS)
In the present study, to achieve accurate 4D Monte Carlo dose calculation in radiation therapy, we devised a new approach that combines (1) modeling of the patient body using tetrahedral-mesh geometry based on the patient’s 4D CT data, (2) continuous movement/deformation of the tetrahedral patient model by interpolation of deformation vector fields acquired through deformable image registration, and (3) direct transportation of radiation particles during the movement and deformation of the tetrahedral patient model. The results of our feasibility study show that it is certainly possible to construct 4D patient models (= phantoms) with sufficient accuracy using the tetrahedral-mesh geometry and to directly transport radiation particles during continuous movement and deformation of the tetrahedral patient model. This new approach not only produces more accurate dose distribution in the patient but also replaces the current practice of using multiple 3D voxel phantoms and combining multiple dose distributions after Monte Carlo simulations. For routine clinical application of our new approach, the use of fast automatic segmentation algorithms is a must. In order to achieve, simultaneously, both dose accuracy and computation speed, the number of tetrahedrons for the lungs should be optimized. Although the current computation speed of our new 4D Monte Carlo simulation approach is slow (i.e. ∼40 times slower than that of the conventional dose accumulation approach), this problem is resolvable by developing, in Geant4, a dedicated navigation class optimized for particle transportation in tetrahedral-mesh geometry. (paper)
International Nuclear Information System (INIS)
The present report describes a computer code DEEP which calculates the organ dose equivalents and the effective dose equivalent for external photon exposure by the Monte Carlo method. MORSE-CG, Monte Carlo radiation transport code, is incorporated into the DEEP code to simulate photon transport phenomena in and around a human body. The code treats an anthropomorphic phantom represented by mathematical formulae and user has a choice for the phantom sex: male, female and unisex. The phantom can wear personal dosimeters on it and user can specify their location and dimension. This document includes instruction and sample problem for the code as well as the general description of dose calculation, human phantom and computer code. (author)
Brons, S; Elsässer, T; Ferrari, A; Gadioli, E; Mairani, A; Parodi, K; Sala, P; Scholz, M; Sommerer, F
2010-01-01
Monte Carlo codes are rapidly spreading among hadron therapy community due to their sophisticated nuclear/electromagnetic models which allow an improved description of the complex mixed radiation field produced by nuclear reactions in therapeutic irradiation. In this contribution results obtained with the Monte Carlo code FLUKA are presented focusing on the production of secondary fragments in carbon ion interaction with water and on CT-based calculations of absorbed and biological effective dose for typical clinical situations. The results of the simulations are compared with the available experimental data and with the predictions of the GSI analytical treatment planning code TRiP.
A study of potential numerical pitfalls in GPU-based Monte Carlo dose calculation
Magnoux, Vincent; Ozell, Benoît; Bonenfant, Éric; Després, Philippe
2015-07-01
The purpose of this study was to evaluate the impact of numerical errors caused by the floating point representation of real numbers in a GPU-based Monte Carlo code used for dose calculation in radiation oncology, and to identify situations where this type of error arises. The program used as a benchmark was bGPUMCD. Three tests were performed on the code, which was divided into three functional components: energy accumulation, particle tracking and physical interactions. First, the impact of single-precision calculations was assessed for each functional component. Second, a GPU-specific compilation option that reduces execution time as well as precision was examined. Third, a specific function used for tracking and potentially more sensitive to precision errors was tested by comparing it to a very high-precision implementation. Numerical errors were found in two components of the program. Because of the energy accumulation process, a few voxels surrounding a radiation source end up with a lower computed dose than they should. The tracking system contained a series of operations that abnormally amplify rounding errors in some situations. This resulted in some rare instances (less than 0.1%) of computed distances that are exceedingly far from what they should have been. Most errors detected had no significant effects on the result of a simulation due to its random nature, either because they cancel each other out or because they only affect a small fraction of particles. The results of this work can be extended to other types of GPU-based programs and be used as guidelines to avoid numerical errors on the GPU computing platform.
Directory of Open Access Journals (Sweden)
Karimian Alireza
2014-01-01
Full Text Available Nowadays, dual energy X-ray absorptiometry is used in bone mineral density systems to assess the amount of osteoporosis. The purpose of this research is to evaluate patient organ doses from dual X-ray absorptiometry by thermoluminescence dosimeters chips and Monte Carlo method. To achieve this goal, in the first step, the surface dose of the cervix, kidney, abdomen region, and thyroid were measured by using TLD-GR 200 at various organ locations. Then, to evaluate the absorbed dose by simulation, the BMD system, patient's body, X-ray source and radiosensitive tissues were simulated by the Monte Carlo method. The results showed, for the spine (left femur bone mineral density scan by using thermoluminescence dosimeters, the absorbed doses of the cervix and kidney were 4.5 (5.64 and 162.17 (3.99(mGy, respectively. For spine (left femur bone mineral density scan in simulation, the absorbed doses of the cervix and kidney were 4.19 (5.88 and 175 (3.68(mGy, respectively. The data obtained showed that the absorbed dose of the kidney in the spine scan is noticeable. Furthermore, because of the small relative difference between the simulation and experimental results, the radiation absorbed dose may be assessed by simulation and software, especially for internal organs, and at different depths of otherwise inaccessible organs which is not possible in experiments.
International Nuclear Information System (INIS)
Monte Carlo simulation has been used by many researchers to calculate organ and effective dose of patients arising from conventional X-ray examinations. In this study the radiation transport code, MCNP4C, has been used to perform Monte Carlo simulations to estimate radiation dose delivered to different organs in conventional X-ray examinations. Materials and Methods: In this work we have made use of ORNL mathematical phantoms with few modifications which have been made. The source has been defined as a point source, emitting photons into a solid angle. The X-ray beam was shaped by a collimator to produce a rectangular field at the midline of the phantom. Results: to validate the simulation executed in this study normalized organs doses to unit ESD for hermaphrodite phantom were computed. Our results were compared with corresponding values presented by NRPB. In general organs doses obtained by application of MCNP-4C (present study) and corresponding values presented in NRPB were in good agreement. For further evaluation of our phantom, the values acquired for organ and effective doses by MCNP-4C and ODS-60 were compared. Conclusion: the technique we have developed is capable of estimating organ and effective doses with a better accuracy than dose values obtained by employment of NRPB and ODS-60 technique
International Nuclear Information System (INIS)
Water in soil affects the dose rate in sedimentary media. First it dilutes the radioelements, and second the mass stopping power as well as photon interaction cross sections are greater in water than in typical sediment constituents (). Here the effect of moisture on the gamma dose rate is investigated using Monte Carlo simulations based on the GEANT4 toolkit. Dose deposition processes are studied on the scale of individual grains in modelled well-sorted sediments: detailed tracking of gamma rays and secondary electrons shed light on the relative importance of the different interaction modes at stake, as a function of grain size, porosity and fraction of saturation of the sediments, as well as the energy of gamma rays. A description of the specifically designed GEANT4 codes is provided, followed by a comprehensive analysis of the phenomena. It is shown that geometry effects, specifically grain size and compactness, have a great impact on gamma dose rates received by sediment grains and that these can be quantified in the case of very simple grain assemblages. For a better accuracy in retrospective dosimetry dating methods, morphological features of the sediments that influence dose rates should be characterized and their influence on dose rate studied; Monte Carlo transport codes seem to be the favoured tools for that purpose.
Directory of Open Access Journals (Sweden)
Lucas Paixão
2015-12-01
Full Text Available Abstract Objective: Derive filtered tungsten X-ray spectra used in digital mammography systems by means of Monte Carlo simulations. Materials and Methods: Filtered spectra for rhodium filter were obtained for tube potentials between 26 and 32 kV. The half-value layer (HVL of simulated filtered spectra were compared with those obtained experimentally with a solid state detector Unfors model 8202031-H Xi R/F & MAM Detector Platinum and 8201023-C Xi Base unit Platinum Plus w mAs in a Hologic Selenia Dimensions system using a direct radiography mode. Results: Calculated HVL values showed good agreement as compared with those obtained experimentally. The greatest relative difference between the Monte Carlo calculated HVL values and experimental HVL values was 4%. Conclusion: The results show that the filtered tungsten anode X-ray spectra and the EGSnrc Monte Carlo code can be used for mean glandular dose determination in mammography.
Paixão, Lucas; Oliveira, Bruno Beraldo; Viloria, Carolina; de Oliveira, Marcio Alves; Teixeira, Maria Helena Araújo; Nogueira, Maria do Socorro
2015-01-01
Objective Derive filtered tungsten X-ray spectra used in digital mammography systems by means of Monte Carlo simulations. Materials and Methods Filtered spectra for rhodium filter were obtained for tube potentials between 26 and 32 kV. The half-value layer (HVL) of simulated filtered spectra were compared with those obtained experimentally with a solid state detector Unfors model 8202031-H Xi R/F & MAM Detector Platinum and 8201023-C Xi Base unit Platinum Plus w mAs in a Hologic Selenia Dimensions system using a direct radiography mode. Results Calculated HVL values showed good agreement as compared with those obtained experimentally. The greatest relative difference between the Monte Carlo calculated HVL values and experimental HVL values was 4%. Conclusion The results show that the filtered tungsten anode X-ray spectra and the EGSnrc Monte Carlo code can be used for mean glandular dose determination in mammography. PMID:26811553
International Nuclear Information System (INIS)
Organ doses from environmental γ-rays (U-238, Th-232, K-40) were calculated using Monte Carlo methods for three typical sources of a semi-infinite volume source in the air, an infinite plane source in the ground and a volume source in the ground. γ-ray fields in the natural environment were simulated rigourously without approximations or simplifications in the intermediate steps except for the disturbance of the radiation field by the human body which was neglected. Organ doses were calculated for four anthropomorphic phantoms representing a baby, a child, a female and a male adult. The dose of a fetus is given by the dose to the uterus of the adult female. Air kerma and dose conversion factors normalised to air kerma and to source intensity are given for monoenergetic sources and for the natural radionuclides. (orig./HP)
International Nuclear Information System (INIS)
Consideration is given to the implications of parameter uncertainty on the output of a mathematical model describing the atmospheric dispersion of 131I, transport through the food chain into milk and the subsequent dose to the thyroid gland of infants resulting from milk ingestion. The approach used was a Monte-Carlo procedure which calculates the dose to the thyroid based on parameters whose values were randomly selected from a normal distribution about their means, stores the results and then repeats the process for other randomly selected parameter values. Predicted doses to the thyroid are log-normally distributed, with the mode strongly shifting to the left and a long tail extending to higher doses. Given parameter uncertainties, the present model predicts that it is possible to receive a dose that is an order of magnitude higher than the prediction obtained by inserting the mean value for each parameter into the model, i.e. the deterministic prediction. (U.K.)
Indian Academy of Sciences (India)
V C Petwal; J N Rao; Jishnu Dwivedi; V K Senecha; K V Subbaiah
2010-03-01
A prototype pulsed electron beam irradiation facility for radiation processing of food and medical products is being commissioned at our centre in Indore, India. Analysis of surface dose and uniformity for a pulsed beam facility is of crucial importance because it is influenced by various operating parameters such as beam current, pulse repetition rate (PRR), scanning current profile and frequency, scanning width and product conveying speed. A large number of experiments are required to determine the harmonized setting of these operating parameters for achieving uniform dose. Since there is no readily available tool to set these parameters, use of Monte Carlo methods and computational tools can prove to be the most viable and time saving technique to support the assessment of the dose distribution. In the present study, Monte Carlo code, MCNP, is used to simulate the transport of 10 MeV electron beam through various mediums coming into the beam path and generate an equivalent dose profile in a polystyrene phantom for stationary state. These results have been verified with experimentally measured dose profile, showing that results are in good agreement within 4%. The Monte Carlo simulation further has been used to optimize the overlapping between the successive pulses of a scan to achieve ± 5% dose uniformity along the scanning direction. A mathematical model, which uses the stationary state data, is developed to include the effect of conveyor speed. The algorithm of the model is discussed and the results are compared with the experimentally measured values, which show that the agreement is better than 15%. Finally, harmonized setting for operating parameters of the accelerator are derived to deliver uniform surface dose in the range of 1–13 kGy/pass.
International Nuclear Information System (INIS)
Purpose: A unique method of delivering radiation dose to the coronary vessel wall to prevent restenosis is by direct injection of radioactive compounds into the vessel wall using a specially designed angioplasty balloon catheter. The radiation dose distribution resulting from such intramural delivery was investigated using Monte Carlo simulations. Materials and methods: The radioisotope source distribution was modeled for two configurations within the vessel wall: (1) uniform to a depth of 0.5 mm and (2) confined to discrete pools surrounding the delivery injection ports. Monte Carlo MCNP4B computer simulations were utilized to estimate the associated radiation dose distribution for the following radioisotopes: 188Re, 186Re, 32P, 153Sm, 111In, 123I, and 99mTc. Results: For the uniform case where the radioisotopes are distributed uniformly to the depth of 0.5 mm into the vessel wall, an essentially constant radiation dose is delivered within the source distribution. Outside of the source volume, the dose falls off at a rate depending on the emission properties of the particular radioisotope. The nonuniform case involving discrete pools of activity showed the dose distribution being confined largely to the regions surrounding the delivery ports with significant regions between these ports receiving very little dose. Conclusions: Direct injection of selected radioisotopes into the arterial wall appears to represent a potentially effective method for delivering radiation dose for the prevention of restenosis. Sufficiently high doses may be obtained from relatively low activity and the dose falls off rapidly outside of the target area for certain radioisotopes
International Nuclear Information System (INIS)
With the rapidly growing number of CT examinations, the consequential radiation risk has aroused more and more attention. The average dose in each organ during CT scans can only be obtained by using Monte Carlo simulation with computational phantoms. Since children tend to have higher radiation sensitivity than adults, the radiation dose of pediatric CT examinations requires special attention and needs to be assessed accurately. So far, studies on organ doses from CT exposures for pediatric patients are still limited. In this work, a 1-year-old computational phantom was constructed. The body contour was obtained from the CT images of a 1-year-old physical phantom and the internal organs were deformed from an existing Chinese reference adult phantom. To ensure the organ locations in the 1-year-old computational phantom were consistent with those of the physical phantom, the organ locations in 1-year-old computational phantom were manually adjusted one by one, and the organ masses were adjusted to the corresponding Chinese reference values. Moreover, a CT scanner model was developed using the Monte Carlo technique and the 1-year-old computational phantom was applied to estimate organ doses derived from simulated CT exposures. As a result, a database including doses to 36 organs and tissues from 47 single axial scans was built. It has been verified by calculation that doses of axial scans are close to those of helical scans; therefore, this database could be applied to helical scans as well. Organ doses were calculated using the database and compared with those obtained from the measurements made in the physical phantom for helical scans. The differences between simulation and measurement were less than 25% for all organs. The result shows that the 1-year-old phantom developed in this work can be used to calculate organ doses in CT exposures, and the dose database provides a method for the estimation of 1-year-old patient doses in a variety of CT examinations. (paper)
International Nuclear Information System (INIS)
This report presents a tabulation of organ and tissue equivalent dose as well as effective dose conversion coefficients, normalised to air kerma free in air, for occupational exposures and environmental exposures of the public to external photon radiation. For occupational exposures, whole-body irradiation with idealised geometries, i.e. broad parallel beams and fully isotropic radiation incidence, is considered. The directions of incidence for the parallel beams are anterior-posterior, posterior-anterior, left lateral, right lateral and a full 360 rotation around the body's longitudinal axis. The influence of beam divergence on the body doses is also considered as well as the dependence of effective dose on the angle of radiation incidence. Regarding exposure of the public to environmental sources, three source geometries are considered: exposure from a radioactive cloud, from ground contamination and from the natural radionuclides distributed homogeneously in the ground. The precise angular and energy distributions of the gamma rays incident on the human body were taken into account. The organ dose conversion coefficients given in this catalogue were calculated using a Monte Carlo code simulating the photon transport in mathematical models of an adult male and an adult female, respectively. Conversion coefficients are given for the equivalent dose of 23 organs and tissues as well as for effective dose and the equivalent dose of the so-called 'remainder'. The organ equivalent dose conversion coefficients are given separately for the adult male and female models and - as arithmetic mean of the conversion coefficients of both - for an average adult. Fitted data of the coefficients are presented in tables; the primary raw data as resulting from the Monte Carlo calculation are shown in figures together with the fitted data. (orig.)
International Nuclear Information System (INIS)
The aim of this work was to investigate how dose distributions calculated with the collapsed cone (CC) algorithm depend on the size of the water phantom used in deriving the point kernel for multiple scatter.A research version of the CC algorithm equipped with a set of selectable point kernels for multiple-scatter dose that had initially been derived in water phantoms of various dimensions was used. The new point kernels were generated using EGSnrc in spherical water phantoms of radii 5 cm, 7.5 cm, 10 cm, 15 cm, 20 cm, 30 cm and 50 cm. Dose distributions derived with CC in water phantoms of different dimensions and in a CT-based clinical breast geometry were compared to Monte Carlo (MC) simulations using the Geant4-based brachytherapy specific MC code Algebra.Agreement with MC within 1% was obtained when the dimensions of the phantom used to derive the multiple-scatter kernel were similar to those of the calculation phantom. Doses are overestimated at phantom edges when kernels are derived in larger phantoms and underestimated when derived in smaller phantoms (by around 2% to 7% depending on distance from source and phantom dimensions). CC agrees well with MC in the high dose region of a breast implant and is superior to TG43 in determining skin doses for all multiple-scatter point kernel sizes. Increased agreement between CC and MC is achieved when the point kernel is comparable to breast dimensions.The investigated approximation in multiple scatter dose depends on the choice of point kernel in relation to phantom size and yields a significant fraction of the total dose only at distances of several centimeters from a source/implant which correspond to volumes of low doses. The current implementation of the CC algorithm utilizes a point kernel derived in a comparatively large (radius 20 cm) water phantom. A fixed point kernel leads to predictable behaviour of the algorithm with the worst case being a source/implant located well within a patient
International Nuclear Information System (INIS)
Knowing the depth dose at the central axis is fundamental for the accurate planning of medical treatment systems involving ionizing radiation. With the evolution of the informatics it is possible the utilization of various computational tools such as GEANT4 and the MCNPX, which use the Monte Carlo Method for simulation of such situations, This paper makes a comparative between the two tools for the this type of application
Energy Technology Data Exchange (ETDEWEB)
Long, Daniel J.; Lee, Choonsik; Tien, Christopher; Fisher, Ryan; Hoerner, Matthew R.; Hintenlang, David; Bolch, Wesley E. [J Crayton Pruitt Family Department of Biomedical Engineering, University of Florida, Gainesville, Florida 32611-6131 (United States); National Cancer Institute, National Institute of Health, Bethesda, Maryland 20892-1502 (United States); J Crayton Pruitt Family Department of Biomedical Engineering, University of Florida, Gainesville, Florida 32611-6131 (United States); Department of Radiology, University of Florida, Gainesville, Florida 32610-0374 (United States); J Crayton Pruitt Family Department of Biomedical Engineering, University of Florida, Gainesville, Florida 32611-6131 (United States)
2013-01-15
Purpose: To validate the accuracy of a Monte Carlo source model of the Siemens SOMATOM Sensation 16 CT scanner using organ doses measured in physical anthropomorphic phantoms. Methods: The x-ray output of the Siemens SOMATOM Sensation 16 multidetector CT scanner was simulated within the Monte Carlo radiation transport code, MCNPX version 2.6. The resulting source model was able to perform various simulated axial and helical computed tomographic (CT) scans of varying scan parameters, including beam energy, filtration, pitch, and beam collimation. Two custom-built anthropomorphic phantoms were used to take dose measurements on the CT scanner: an adult male and a 9-month-old. The adult male is a physical replica of University of Florida reference adult male hybrid computational phantom, while the 9-month-old is a replica of University of Florida Series B 9-month-old voxel computational phantom. Each phantom underwent a series of axial and helical CT scans, during which organ doses were measured using fiber-optic coupled plastic scintillator dosimeters developed at University of Florida. The physical setup was reproduced and simulated in MCNPX using the CT source model and the computational phantoms upon which the anthropomorphic phantoms were constructed. Average organ doses were then calculated based upon these MCNPX results. Results: For all CT scans, good agreement was seen between measured and simulated organ doses. For the adult male, the percent differences were within 16% for axial scans, and within 18% for helical scans. For the 9-month-old, the percent differences were all within 15% for both the axial and helical scans. These results are comparable to previously published validation studies using GE scanners and commercially available anthropomorphic phantoms. Conclusions: Overall results of this study show that the Monte Carlo source model can be used to accurately and reliably calculate organ doses for patients undergoing a variety of axial or helical CT
International Nuclear Information System (INIS)
Purpose: To validate the accuracy of a Monte Carlo source model of the Siemens SOMATOM Sensation 16 CT scanner using organ doses measured in physical anthropomorphic phantoms. Methods: The x-ray output of the Siemens SOMATOM Sensation 16 multidetector CT scanner was simulated within the Monte Carlo radiation transport code, MCNPX version 2.6. The resulting source model was able to perform various simulated axial and helical computed tomographic (CT) scans of varying scan parameters, including beam energy, filtration, pitch, and beam collimation. Two custom-built anthropomorphic phantoms were used to take dose measurements on the CT scanner: an adult male and a 9-month-old. The adult male is a physical replica of University of Florida reference adult male hybrid computational phantom, while the 9-month-old is a replica of University of Florida Series B 9-month-old voxel computational phantom. Each phantom underwent a series of axial and helical CT scans, during which organ doses were measured using fiber-optic coupled plastic scintillator dosimeters developed at University of Florida. The physical setup was reproduced and simulated in MCNPX using the CT source model and the computational phantoms upon which the anthropomorphic phantoms were constructed. Average organ doses were then calculated based upon these MCNPX results. Results: For all CT scans, good agreement was seen between measured and simulated organ doses. For the adult male, the percent differences were within 16% for axial scans, and within 18% for helical scans. For the 9-month-old, the percent differences were all within 15% for both the axial and helical scans. These results are comparable to previously published validation studies using GE scanners and commercially available anthropomorphic phantoms. Conclusions: Overall results of this study show that the Monte Carlo source model can be used to accurately and reliably calculate organ doses for patients undergoing a variety of axial or helical CT
International Nuclear Information System (INIS)
Computed tomography (CT) is a technique which offers a high diagnostic capability; however, the dose to the patient is high compared to conventional radiography. This report provides a catalogue of organ doses resulting from CT examinations. The organ doses were calculated for the type of CT scanners most commonly used in the FRG and for three different radiation qualities. For the dose calculations, the patients were represented by the adult mathematical phantoms Adam and Eva. The radiation transport in the body was simulated using a Monte Carlo method. The doses were calculated as conversion factors of mean organ doses per air kerma free in air on the axis of rotation. Mean organ dose conversion factors are given per organ and per single CT slice of 1 cm width. The mean dose to an organ resulting from a particular CT examination can be estimated by summing up the contribution to the organ dose from each relevant slice. In order to facilitate the selection of the appropriate slices, a table is given which relates the mathematical phantoms' coordinates to certain anatomical landmarks in the human body. (orig.)
International Nuclear Information System (INIS)
Digital breast tomosynthesis (DBT) is a screening and diagnostic modality that acquires images of the breast at multiple angles during a short scan. The Selenia Dimensions (Hologic, Bedford, Mass) DBT system can perform both full-field digital mammography and DBT. The system acquires 15 projections over a 15 deg. angular range (from -7.5 deg. to +7.5 deg.). An important factor in determining the optimal imaging technique for breast tomosynthesis is the radiation dose. In breast imaging, the radiation dose of concern is that deposited in the glandular tissue of the breast because this is the tissue that has a risk of developing cancer. The concept of the normalised mean glandular dose (DgN) has been introduced as the metric for the dose in breast imaging. The DgN is difficult to measure. The Monte Carlo techniques offer an alternative method for a realistic estimation of the radiation dose. The purpose of this work was to use the Monte Carlo code MCNPX technique to generate monoenergetic glandular dose data for estimating the breast tissue dose in tomosynthesis for arbitrary spectra as well as to observe the deposited radiation dose by projection on the glandular portion of the breast in a Selenia Dimensions DBT system. A Monte Carlo simulation of the system was developed to compute the DgN in a craniocaudal view. Monoenergetic X-ray beams from 10 to 49 keV in 1-keV increments were used. The simulation utilised the assumption of a homogeneous breast composition and three compositions (0 % glandular, 50 % glandular and 100 % glandular). The glandular and adipose tissue compositions were specified according ICRU Report 44. A skin layer of 4 mm was assumed to encapsulate the breast on all surfaces. The breast size was varied using the chest wall-to-nipple distance (CND) and compressed breast thickness (t). In this work, the authors assumed a CND of 5 cm and the thicknesses ranged from 2 to 8 cm, in steps of 2 cm. The fractional energy absorption increases (up to 44
International Nuclear Information System (INIS)
The four primary radiation measurement systems considered to be necessary for the modern radionuclide measurement laboratory are gas-flow proportional counters, liquid scintillation counters, Si spectrometer systems, and Ge spectrometer systems. A national institute of standards recognized by BIPM (Bureau International des Poids et Mesures) use these measurement systems have some limitation. For a gas-filled detector, the source size should not be larger than that determined by the counter geometry and the measurements are limited to sources with the activities of less than 20,000 Bq. A liquid scintillation detector can only be used to measure an activity of liquid mixture isotope. Therefore it can't be used to measure activities of sealed sources. Further it is quite difficult to measure a radioactivity of beta-isotope accurately due to self-absorption and scattering. In particular an radioactivity generated by a nuclear reactor is approximately calculated by using target material compositions, cross-sections, and neutron flux, which the nuclear reactor. However, the results from this approach involve a high uncertainty. We develop a Monte Carlo applied radioactivity determination for beta sources. The extrapolation ion-chamber was used to measure the surface dose rate for a standard source. The Sr/Y-90 standard source is calibrated by NIST used for this study. There was about 1.7% difference in the reference dose rates measured by the two techniques. These difference and relative errors were comparable to those of other studies. It should be noticed that the area of collecting electrode was nine times larger than that of radiation field. It the diameter of the source is smaller than the area of collecting electrode, the source should be located at a sufficiently large distance from the detector surface so that the radiation field at the detector surface was larger than the area of collecting electrode. In addition, beta particles from the applicator interact
International Nuclear Information System (INIS)
Monte-Carlo simulation is one of the most essential computational tools to study the particle transport and interaction of radiation with matter as well as radiation protection and dosimetry. In this paper it was used to calculate percent depth doses in the water phantom for two Co-60 beam irradiation cases with using the MCNP-4C2 code. The simulation results was validated by comparison with those of measurements. Application of the MCNP-4C2 code for dose calculations in Co-60 beam treatment planning was recommended. (author)
Gharibian, Katherine N; Mueller, Bruce A
2016-07-01
Fluconazole is a renally-eliminated antifungal commonly used to treat Candida species infections. In critically-ill patients receiving prolonged intermittent renal replacement therapy (PIRRT), limited pharmacokinetic (PK) data are available to guide fluconazole dosing. We used previously-published fluconazole clearance data and PK data of critically-ill patients with acute kidney injury to develop a PK model with the goal of determining a therapeutic dosing regimen for critically-ill patients receiving PIRRT. Monte Carlo simulations were performed to create a virtual cohort of patients receiving different fluconazole dosing regimens. Plasma drug concentration-time profiles were evaluated on the probability of attaining a mean 24-hour area under the drug concentration-time curve to minimum inhibitory concentration ratio (AUC24h : MIC) of 100 during the initial 48 hours of antifungal therapy. At the susceptibility breakpoint of Candida albicans (2 mg/L), 93 - 96% of simulated subjects receiving PIRRT attained the pharmacodynamic target with a fluconazole 800-mg loading dose plus 400 mg twice daily (q12h or pre and post PIRRT) regimen. Monte Carlo simulations of a PK model of PIRRT provided a basis for the development of an informed fluconazole dosing recommendation when PK data was limited. This finding should be validated in the clinical setting. PMID:27251341
International Nuclear Information System (INIS)
The aim of this study was to quantify the dose enhancement by gadolinium and gold nanoparticles in brachytherapy. MCNPX Monte Carlo code was used to simulate four brachytherapy sources: 60Co, 198Au, 192Ir, 169Yb. To verify the accuracy of our simulations, the obtained values of dose rate constants and radial dose functions were compared with corresponding published values for these sources. To study dose enhancements, a spherical soft tissue phantom with 15 cm in radius was simulated. Gadolinium and gold nanoparticles at 10, 20 and 30 mg/ml concentrations were separately assumed in a 1 × 1 × 1 cm3 volume simulating tumour. The simulated dose to the tumour with the impurity was compared to the dose without impurity, as a function of radial distance and concentration of the impurity, to determine the enhancement of dose due to the presence of the impurity. Dose enhancements in the tumour obtained in the presence of gadolinium and gold nanoparticles with concentration of 30 mg/ml, were found to be in the range of −0.5–106.1 and 0.4–153.1 % respectively. In addition, at higher radial distances from the source center, higher dose enhancements were observed. GdNPs can be used as a high atomic number material to enhance dose in tumour volume with dose enhancements up to 106.1 % when used in brachytherapy. Regardless considering the clinical limitations of the here-in presented model, for a similar source and concentration of nanoparticles, gold nanoparticles show higher dose enhancement than gadolinium nanoparticles and can have more clinical usefulness as dose enhancer material.
International Nuclear Information System (INIS)
This report considers the contribution from scattered radiation to the dose to organs and tissues which lie outside the useful therapy beams. The results presented are the product of Monte Carlo studies used to determine the tissue doses due to internal scattering of the useful beams only. General cases are calculated in which central target volumes in the trunk are treated with 10 x 14 cm2 and 14 x 14 cm2 fields from 200 kV, Co-60, 8 MV and 25 MV therapy equipment. Target volumes in the neck are considered to be treated with 5 x 5 cm2 fields. Different treatment plans are calculated including rotational therapy. Also two specific cases are more fully analysed, namely for Ankylosing Spondylitis and central abdomen malignant disease in the region of the head of the pancreas. The calculated organ doses are presented in tables as a percentage of the target volume dose. (orig.)
International Nuclear Information System (INIS)
The purpose of this paper is to report a set of experimental values of patient and staff doses in a cardiac catheterisation laboratory using the range of radiographic and geometric parameters from routine clinical practice. The data obtained will be available for validation of Monte Carlo calculations and for training purposes. They will also help optimise radiation protection for patients and staff. Experimental measurements were made with an anthropomorphic phantom, and a monoplane flat detector-based X-ray system was used for interventional cardiology procedures. Standard operational protocols used in clinical practice were applied. Around 1000 patient dose and 5000 staff dose values were measured for different operational conditions (angulations, distances, collimation and wedge filter, magnification, phantom thicknesses, using Copper absorber, etc.). Uncertainties were also estimated. Increase factors of 3-10 for patients and staff doses were measured for the different C-arm angulations. (authors)
Energy Technology Data Exchange (ETDEWEB)
Silva, Laura E. da; Nicolucci, Patricia, E-mail: laura.emilia.fm@gmail.com [Universidade de Sao Paulo (USP), Ribeirao Preto, SP (Brazil). Faculdade de Filosofia, Ciencias e Letras
2014-04-15
The development of nanotechnology has boosted the use of nanoparticles in radiation therapy in order to achieve greater therapeutic ratio between tumor and healthy tissues. Gold has been shown to be most suitable to this task due to the high biocompatibility and high atomic number, which contributes to a better in vivo distribution and for the local energy deposition. As a result, this study proposes to study, nanoparticle in the tumor cell. At a range of 11 nm from the nanoparticle surface, results have shown an absorbed dose 141 times higher for the medium with the gold nanoparticle compared to the water for an incident energy spectrum with maximum photon energy of 50 keV. It was also noted that when only scattered radiation is interacting with the gold nanoparticles, the dose was 134 times higher compared to enhanced local dose that remained significant even for scattered radiation. (author)
International Nuclear Information System (INIS)
Highlights: → The MCNP5 code has been used to model a radiotherapy room of a 18 MV linear accelerator. → The neutron and the secondary gamma ray dose equivalents were evaluated at various points inside the treatment room and along the the maze. → To reduce the neutron and gamma ray doses, we have also investigated the radiotherapy room shielding performance. → The use of paraffin wax containing boron carbide indicates much better shielding effects. - Abstract: Medical accelerators operating above 10 MV are a source of undesirable neutron radiations which contaminate the therapeutic photon beam. These photoneutrons can also generate secondary gamma rays which increases undesirable dose to the patient body and to personnel and general public. In this study, the Monte Carlo N-Particle MCNP5 code has been used to model the radiotherapy room of a medical linear accelerator operating at 18 MV and to calculate the neutron and the secondary gamma ray energy spectra and the dose equivalents at various points inside the treatment room and along the maze. To validate our Monte Carlo simulation we compared our results with those evaluated by the recommended analytical methods of IAEA Report No. 47, and with experimental and simulated values published in the literature. After validation, the Monte Carlo simulation has been used to evaluate the shielding performance of the radiotherapy room. The obtained results showed that the use of paraffin wax containing boron carbide, in the lining of the radiotherapy room walls, presents enough effectiveness to reduce both neutron and gamma ray doses inside the treatment room and at the maze entrance. Such evaluation cannot be performed by the analytical methods since room material and wall surface lining are not taken into consideration.
Kim, Sangroh; Yoshizumi, Terry; Toncheva, Greta; Yoo, Sua; Yin, Fang-Fang; Frush, Donald
2010-05-01
To address the lack of accurate dose estimation method in cone beam computed tomography (CBCT), we performed point dose metal oxide semiconductor field-effect transistor (MOSFET) measurements and Monte Carlo (MC) simulations. A Varian On-Board Imager (OBI) was employed to measure point doses in the polymethyl methacrylate (PMMA) CT phantoms with MOSFETs for standard and low dose modes. A MC model of the OBI x-ray tube was developed using BEAMnrc/EGSnrc MC system and validated by the half value layer, x-ray spectrum and lateral and depth dose profiles. We compared the weighted computed tomography dose index (CTDIw) between MOSFET measurements and MC simulations. The CTDIw was found to be 8.39 cGy for the head scan and 4.58 cGy for the body scan from the MOSFET measurements in standard dose mode, and 1.89 cGy for the head and 1.11 cGy for the body in low dose mode, respectively. The CTDIw from MC compared well to the MOSFET measurements within 5% differences. In conclusion, a MC model for Varian CBCT has been established and this approach may be easily extended from the CBCT geometry to multi-detector CT geometry. PMID:20386198
Parodi, K.; Ferrari, A.; Sommerer, F.; Paganetti, H.
2007-07-01
Clinical investigations on post-irradiation PET/CT (positron emission tomography/computed tomography) imaging for in vivo verification of treatment delivery and, in particular, beam range in proton therapy are underway at Massachusetts General Hospital (MGH). Within this project, we have developed a Monte Carlo framework for CT-based calculation of dose and irradiation-induced positron emitter distributions. Initial proton beam information is provided by a separate Geant4 Monte Carlo simulation modelling the treatment head. Particle transport in the patient is performed in the CT voxel geometry using the FLUKA Monte Carlo code. The implementation uses a discrete number of different tissue types with composition and mean density deduced from the CT scan. Scaling factors are introduced to account for the continuous Hounsfield unit dependence of the mass density and of the relative stopping power ratio to water used by the treatment planning system (XiO (Computerized Medical Systems Inc.)). Resulting Monte Carlo dose distributions are generally found in good correspondence with calculations of the treatment planning program, except a few cases (e.g. in the presence of air/tissue interfaces). Whereas dose is computed using standard FLUKA utilities, positron emitter distributions are calculated by internally combining proton fluence with experimental and evaluated cross-sections yielding 11C, 15O, 14O, 13N, 38K and 30P. Simulated positron emitter distributions yield PET images in good agreement with measurements. In this paper, we describe in detail the specific implementation of the FLUKA calculation framework, which may be easily adapted to handle arbitrary phase spaces of proton beams delivered by other facilities or include more reaction channels based on additional cross-section data. Further, we demonstrate the effects of different acquisition time regimes (e.g., PET imaging during or after irradiation) on the intensity and spatial distribution of the irradiation
Vanderstraeten, Barbara; Reynaert, Nick; Paelinck, Leen; Madani, Indira; De Wagter, Carlos; De Gersem, Werner; De Neve, Wilfried; Thierens, Hubert
2006-09-01
The accuracy of dose computation within the lungs depends strongly on the performance of the calculation algorithm in regions of electronic disequilibrium that arise near tissue inhomogeneities with large density variations. There is a lack of data evaluating the performance of highly developed analytical dose calculation algorithms compared to Monte Carlo computations in a clinical setting. We compared full Monte Carlo calculations (performed by our Monte Carlo dose engine MCDE) with two different commercial convolution/superposition (CS) implementations (Pinnacle-CS and Helax-TMS's collapsed cone model Helax-CC) and one pencil beam algorithm (Helax-TMS's pencil beam model Helax-PB) for 10 intensity modulated radiation therapy (IMRT) lung cancer patients. Treatment plans were created for two photon beam qualities (6 and 18 MV). For each dose calculation algorithm, patient, and beam quality, the following set of clinically relevant dose-volume values was reported: (i) minimal, median, and maximal dose (Dmin, D50, and Dmax) for the gross tumor and planning target volumes (GTV and PTV); (ii) the volume of the lungs (excluding the GTV) receiving at least 20 and 30 Gy (V20 and V30) and the mean lung dose; (iii) the 33rd percentile dose (D33) and Dmax delivered to the heart and the expanded esophagus; and (iv) Dmax for the expanded spinal cord. Statistical analysis was performed by means of one-way analysis of variance for repeated measurements and Tukey pairwise comparison of means. Pinnacle-CS showed an excellent agreement with MCDE within the target structures, whereas the best correspondence for the organs at risk (OARs) was found between Helax-CC and MCDE. Results from Helax-PB were unsatisfying for both targets and OARs. Additionally, individual patient results were analyzed. Within the target structures, deviations above 5% were found in one patient for the comparison of MCDE and Helax-CC, while all differences between MCDE and Pinnacle-CS were below 5%. For both
International Nuclear Information System (INIS)
The accuracy of dose computation within the lungs depends strongly on the performance of the calculation algorithm in regions of electronic disequilibrium that arise near tissue inhomogeneities with large density variations. There is a lack of data evaluating the performance of highly developed analytical dose calculation algorithms compared to Monte Carlo computations in a clinical setting. We compared full Monte Carlo calculations (performed by our Monte Carlo dose engine MCDE) with two different commercial convolution/superposition (CS) implementations (Pinnacle-CS and Helax-TMS's collapsed cone model Helax-CC) and one pencil beam algorithm (Helax-TMS's pencil beam model Helax-PB) for 10 intensity modulated radiation therapy (IMRT) lung cancer patients. Treatment plans were created for two photon beam qualities (6 and 18 MV). For each dose calculation algorithm, patient, and beam quality, the following set of clinically relevant dose-volume values was reported: (i) minimal, median, and maximal dose (Dmin, D50, and Dmax) for the gross tumor and planning target volumes (GTV and PTV); (ii) the volume of the lungs (excluding the GTV) receiving at least 20 and 30 Gy (V20 and V30) and the mean lung dose; (iii) the 33rd percentile dose (D33) and Dmax delivered to the heart and the expanded esophagus; and (iv) Dmax for the expanded spinal cord. Statistical analysis was performed by means of one-way analysis of variance for repeated measurements and Tukey pairwise comparison of means. Pinnacle-CS showed an excellent agreement with MCDE within the target structures, whereas the best correspondence for the organs at risk (OARs) was found between Helax-CC and MCDE. Results from Helax-PB were unsatisfying for both targets and OARs. Additionally, individual patient results were analyzed. Within the target structures, deviations above 5% were found in one patient for the comparison of MCDE and Helax-CC, while all differences between MCDE and Pinnacle-CS were below 5%. For both
Directory of Open Access Journals (Sweden)
Hammam Oktajianto
2015-01-01
Full Text Available Gas-cooled nuclear reactor is a Generation IV reactor which has been receiving significant attention due to many desired characteristics such as inherent safety, modularity, relatively low cost, short construction period, and easy financing. High temperature reactor (HTR pebble-bed as one of type of gas-cooled reactor concept is getting attention. In HTR pebble-bed design, radius and enrichment of the fuel kernel are the key parameter that can be chosen freely to determine the desired value of criticality. This paper models HTR pebble-bed 10 MW and determines an effective of enrichment and radius of the fuel (Kernel to get criticality value of reactor. The TRISO particle coated fuel particle which was modelled explicitly and distributed in the fuelled region of the fuel pebbles using a Simple-Cubic (SC lattice. The pebble-bed balls and moderator balls distributed in the core zone using a Body-Centred Cubic lattice with assumption of a fresh fuel by the fuel enrichment was 7-17% at 1% range and the size of the fuel radius was 175-300 µm at 25 µm ranges. The geometrical model of the full reactor is obtained by using lattice and universe facilities provided by MCNP4C. The details of model are discussed with necessary simplifications. Criticality calculations were conducted by Monte Carlo transport code MCNP4C and continuous energy nuclear data library ENDF/B-VI. From calculation results can be concluded that an effective of enrichment and radius of fuel (Kernel to achieve a critical condition was the enrichment of 15-17% at a radius of 200 µm, the enrichment of 13-17% at a radius of 225 µm, the enrichments of 12-15% at radius of 250 µm, the enrichments of 11-14% at a radius of 275 µm and the enrichment of 10-13% at a radius of 300 µm, so that the effective of enrichments and radii of fuel (Kernel can be considered in the HTR 10 MW.
TU-F-18A-03: Improving Tissue Segmentation for Monte Carlo Dose Calculation Using DECT Data
Energy Technology Data Exchange (ETDEWEB)
Di, Salvio A; Bedwani, S; Carrier, J [CHUM - Notre-Dame, Montreal, QC (Canada)
2014-06-15
Purpose: To develop a new segmentation technique using dual energy CT (DECT) to overcome limitations related to segmentation from a standard Hounsfield unit (HU) to electron density (ED) calibration curve. Both methods are compared with a Monte Carlo analysis of dose distribution. Methods: DECT allows a direct calculation of both ED and effective atomic number (EAN) within a given voxel. The EAN is here defined as a function of the total electron cross-section of a medium. These values can be effectively acquired using a calibrated method from scans at two different energies. A prior stoichiometric calibration on a Gammex RMI phantom allows us to find the parameters to calculate EAN and ED within a voxel. Scans from a Siemens SOMATOM Definition Flash dual source system provided the data for our study. A Monte Carlo analysis compares dose distribution simulated by dosxyz-nrc, considering a head phantom defined by both segmentation techniques. Results: Results from depth dose and dose profile calculations show that materials with different atomic compositions but similar EAN present differences of less than 1%. Therefore, it is possible to define a short list of basis materials from which density can be adapted to imitate interaction behavior of any tissue. Comparison of the dose distributions on both segmentations shows a difference of 50% in dose in areas surrounding bone at low energy. Conclusion: The presented segmentation technique allows a more accurate medium definition in each voxel, especially in areas of tissue transition. Since the behavior of human tissues is highly sensitive at low energies, this reduces the errors on calculated dose distribution. This method could be further developed to optimize the tissue characterization based on anatomic site.
TU-F-18A-03: Improving Tissue Segmentation for Monte Carlo Dose Calculation Using DECT Data
International Nuclear Information System (INIS)
Purpose: To develop a new segmentation technique using dual energy CT (DECT) to overcome limitations related to segmentation from a standard Hounsfield unit (HU) to electron density (ED) calibration curve. Both methods are compared with a Monte Carlo analysis of dose distribution. Methods: DECT allows a direct calculation of both ED and effective atomic number (EAN) within a given voxel. The EAN is here defined as a function of the total electron cross-section of a medium. These values can be effectively acquired using a calibrated method from scans at two different energies. A prior stoichiometric calibration on a Gammex RMI phantom allows us to find the parameters to calculate EAN and ED within a voxel. Scans from a Siemens SOMATOM Definition Flash dual source system provided the data for our study. A Monte Carlo analysis compares dose distribution simulated by dosxyz-nrc, considering a head phantom defined by both segmentation techniques. Results: Results from depth dose and dose profile calculations show that materials with different atomic compositions but similar EAN present differences of less than 1%. Therefore, it is possible to define a short list of basis materials from which density can be adapted to imitate interaction behavior of any tissue. Comparison of the dose distributions on both segmentations shows a difference of 50% in dose in areas surrounding bone at low energy. Conclusion: The presented segmentation technique allows a more accurate medium definition in each voxel, especially in areas of tissue transition. Since the behavior of human tissues is highly sensitive at low energies, this reduces the errors on calculated dose distribution. This method could be further developed to optimize the tissue characterization based on anatomic site
Oliver, Mike; Staruch, Robert; Gladwish, Adam; Craig, Jeff; Chen, Jeff; Wong, Eugene
2008-05-01
Respiratory gating is emerging as a tool to limit the effect of motion for liver and lung tumors. In order to study the impact of target motion and gated intensity modulated radiation therapy (IMRT) delivery, a computer program was developed to simulate segmental IMRT delivery to a moving phantom. Two distinct plans were delivered to a rigid-motion phantom with a film insert in place under four conditions: static, sinusoidal motion, gated sinusoidal motion with a duty cycle of 25% and gated sinusoidal motion with duty cycle of 50% under motion conditions of a typical patient (A = 1 cm, T = 4 s). The MLC controller log files and gating log files were retained to perform a retrospective Monte Carlo dose calculation of the plans. Comparison of the 2D planar dose distributions between simulation and measurement demonstrated that our technique had at least 94% of the points passing gamma criteria of 3% for dose difference and 3 mm as the distance to agreement. This note demonstrates that the use of dynamic multi-leaf collimator and respiratory monitoring system log files together with a fast Monte Carlo dose calculation algorithm is an accurate and efficient way to study the dosimetric effect of motion for gated or non-gated IMRT delivery on a rigidly-moving body.
Künzler, Thomas; Fotina, Irina; Stock, Markus; Georg, Dietmar
2009-12-01
The dosimetric performance of a Monte Carlo algorithm as implemented in a commercial treatment planning system (iPlan, BrainLAB) was investigated. After commissioning and basic beam data tests in homogenous phantoms, a variety of single regular beams and clinical field arrangements were tested in heterogeneous conditions (conformal therapy, arc therapy and intensity-modulated radiotherapy including simultaneous integrated boosts). More specifically, a cork phantom containing a concave-shaped target was designed to challenge the Monte Carlo algorithm in more complex treatment cases. All test irradiations were performed on an Elekta linac providing 6, 10 and 18 MV photon beams. Absolute and relative dose measurements were performed with ion chambers and near tissue equivalent radiochromic films which were placed within a transverse plane of the cork phantom. For simple fields, a 1D gamma (γ) procedure with a 2% dose difference and a 2 mm distance to agreement (DTA) was applied to depth dose curves, as well as to inplane and crossplane profiles. The average gamma value was 0.21 for all energies of simple test cases. For depth dose curves in asymmetric beams similar gamma results as for symmetric beams were obtained. Simple regular fields showed excellent absolute dosimetric agreement to measurement values with a dose difference of 0.1% ± 0.9% (1 standard deviation) at the dose prescription point. A more detailed analysis at tissue interfaces revealed dose discrepancies of 2.9% for an 18 MV energy 10 × 10 cm2 field at the first density interface from tissue to lung equivalent material. Small fields (2 × 2 cm2) have their largest discrepancy in the re-build-up at the second interface (from lung to tissue equivalent material), with a local dose difference of about 9% and a DTA of 1.1 mm for 18 MV. Conformal field arrangements, arc therapy, as well as IMRT beams and simultaneous integrated boosts were in good agreement with absolute dose measurements in the
Investigation of geometrical and scoring grid resolution for Monte Carlo dose calculations for IMRT
DeSmedt, B.; Vanderstraeten, B.; Reynaert, N.; DeNeve, W.; Thierens, H.
2005-09-01
Monte Carlo based treatment planning of two different patient groups treated with step-and-shoot IMRT (head-and-neck and lung treatments) with different CT resolutions and scoring methods is performed to determine the effect of geometrical and scoring voxel sizes on DVHs and calculation times. Dose scoring is performed in two different ways: directly into geometrical voxels (or in a number of grouped geometrical voxels) or into scoring voxels defined by a separate scoring grid superimposed on the geometrical grid. For the head-and-neck cancer patients, more than 2% difference is noted in the right optical nerve when using voxel dimensions of 4 × 4 × 4 mm3 compared to the reference calculation with 1 × 1 × 2 mm3 voxel dimensions. For the lung cancer patients, 2% difference is noted in the spinal cord when using voxel dimensions of 4 × 4 × 10 mm3 compared to the 1 × 1 × 5 mm3 calculation. An independent scoring grid introduces several advantages. In cases where a relatively high geometrical resolution is required and where the scoring resolution is less important, the number of scoring voxels can be limited while maintaining a high geometrical resolution. This can be achieved either by grouping several geometrical voxels together into scoring voxels or by superimposing a separate scoring grid of spherical voxels with a user-defined radius on the geometrical grid. For the studied lung cancer cases, both methods produce accurate results and introduce a speed increase by a factor of 10-36. In cases where a low geometrical resolution is allowed, but where a high scoring resolution is required, superimposing a separate scoring grid on the geometrical grid allows a reduction in geometrical voxels while maintaining a high scoring resolution. For the studied head-and-neck cancer cases, calculations performed with a geometrical resolution of 2 × 2 × 2 mm3 and a separate scoring grid containing spherical scoring voxels with a radius of 2 mm produce accurate results
WE-A-17A-04: Development of An Ultra-Fast Monte Carlo Dose Engine for High Dose Rate Brachytherapy
International Nuclear Information System (INIS)
Purpose: Current clinical brachytherapy dose calculations are based on AAPM TG43 guidelines, which approximate the patient geometry as a large water phantom. This ignores heterogeneities and tends to overestimate skin dose. Although Monte Carlo (MC) dose calculations have been recognized as the most accurate method, its associated long computational time is a major bottleneck for routine clinical applications. This work aims to develop a GPU-based ultra-fast MC dose engine (gBMC) for HDR brachytherapy to provide clinical users with accurate sub-minute dose calculations. Methods: Standard photon transport with discrete events including: Compton scattering, Rayleigh scattering and Photoelectric effect, was implemented. Secondary electrons were transported under the continuous slowing down approximation. To reduce the GPU thread divergence, photons and electrons were separately transported. Transport of photons was grouped according to energy. The source model in gBMC can be either a phase-space file generated using Geant4 or a parameterized source model. This dose engine was validated against TG43 in a water phantom and against Geant4 calculations in heterogeneous patient geometries. Results: A phase space file was generated for the Varian VS2000 Ir-192 source. In a water phantom, the calculated radial dose function was within 0.6% of the TG43 calculations for radial distances from 1 cm to 20 cm. The anisotropy functions were within 1% for radial distances from 1 cm to 20 cm except for polar angles larger than 173°. Local point-dose differences were within 2%. In a Mammosite breast cancer case with 22 dwell locations, gBMC and Geant4 isodose lines compared well. The computation time was about 28 seconds using the phase-space file source and 20 seconds using the parameterized source to simulate 1 billion particles, yielding less than 1% statistical uncertainty. Conclusion: The gBMC dose engine makes it possible to use fast and accurate MC dose calculations for
Matsumoto, Shinnosuke; Koba, Yusuke; Kohno, Ryosuke; Lee, Choonsik; Bolch, Wesley E; Kai, Michiaki
2016-04-01
Proton therapy has the physical advantage of a Bragg peak that can provide a better dose distribution than conventional x-ray therapy. However, radiation exposure of normal tissues cannot be ignored because it is likely to increase the risk of secondary cancer. Evaluating secondary neutrons generated by the interaction of the proton beam with the treatment beam-line structure is necessary; thus, performing the optimization of radiation protection in proton therapy is required. In this research, the organ dose and energy spectrum were calculated from secondary neutrons using Monte Carlo simulations. The Monte Carlo code known as the Particle and Heavy Ion Transport code System (PHITS) was used to simulate the transport proton and its interaction with the treatment beam-line structure that modeled the double scattering body of the treatment nozzle at the National Cancer Center Hospital East. The doses of the organs in a hybrid computational phantom simulating a 5-y-old boy were calculated. In general, secondary neutron doses were found to decrease with increasing distance to the treatment field. Secondary neutron energy spectra were characterized by incident neutrons with three energy peaks: 1×10, 1, and 100 MeV. A block collimator and a patient collimator contributed significantly to organ doses. In particular, the secondary neutrons from the patient collimator were 30 times higher than those from the first scatter. These results suggested that proactive protection will be required in the design of the treatment beam-line structures and that organ doses from secondary neutrons may be able to be reduced. PMID:26910030
Feasibility study of the neutron dose for real-time image-guided proton therapy: A Monte Carlo study
Kim, Jin Sung; Shin, Jung Suk; Kim, Daehyun; Shin, Eunhyuk; Chung, Kwangzoo; Cho, Sungkoo; Ahn, Sung Hwan; Ju, Sanggyu; Chung, Yoonsun; Jung, Sang Hoon; Han, Youngyih
2015-07-01
Two full rotating gantries with different nozzles (multipurpose nozzle with MLC, scanning dedicated nozzle) for a conventional cyclotron system are installed and being commissioned for various proton treatment options at Samsung Medical Center in Korea. The purpose of this study is to use Monte Carlo simulation to investigate the neutron dose equivalent per therapeutic dose, H/D, for X-ray imaging equipment under various treatment conditions. At first, we investigated the H/D for various modifications of the beamline devices (scattering, scanning, multi-leaf collimator, aperture, compensator) at the isocenter and at 20, 40 and 60 cm distances from the isocenter, and we compared our results with those of other research groups. Next, we investigated the neutron dose at the X-ray equipment used for real-time imaging under various treatment conditions. Our investigation showed doses of 0.07 ~ 0.19 mSv/Gy at the X-ray imaging equipment, depending on the treatment option and interestingly, the 50% neutron dose reduction was observed due to multileaf collimator during proton scanning treatment with the multipurpose nozzle. In future studies, we plan to measure the neutron dose experimentally and to validate the simulation data for X-ray imaging equipment for use as an additional neutron dose reduction method.
Energy Technology Data Exchange (ETDEWEB)
Habib, B.; Poumarede, B.; Tola, F.; Barthe, J. [CEA, LIST, Dept Technol Capteur et Signal, F-91191 Gif Sur Yvette, (France)
2010-07-01
The aim of the present study is to demonstrate the potential of accelerated dose calculations, using the fast Monte Carlo (MC) code referred to as PENFAST, rather than the conventional MC code PENELOPE, without losing accuracy in the computed dose. For this purpose, experimental measurements of dose distributions in homogeneous and inhomogeneous phantoms were compared with simulated results using both PENELOPE and PENFAST. The simulations and experiments were performed using a Saturne 43 linac operated at 12 MV (photons), and at 18 MeV (electrons). Pre-calculated phase space files (PSFs) were used as input data to both the PENELOPE and PENFAST dose simulations. Since depth-dose and dose profile comparisons between simulations and measurements in water were found to be in good agreement (within {+-} 1% to 1 mm), the PSF calculation is considered to have been validated. In addition, measured dose distributions were compared to simulated results in a set of clinically relevant, inhomogeneous phantoms, consisting of lung and bone heterogeneities in a water tank. In general, the PENFAST results agree to within a 1% to 1 mm difference with those produced by PENELOPE, and to within a 2% to 2 mm difference with measured values. Our study thus provides a pre-clinical validation of the PENFAST code. It also demonstrates that PENFAST provides accurate results for both photon and electron beams, equivalent to those obtained with PENELOPE. CPU time comparisons between both MC codes show that PENFAST is generally about 9-21 times faster than PENELOPE. (authors)
On-board Dose Measurement and its Monte Carlo Analysis in a Low Level Waste Shipping Vessel
International Nuclear Information System (INIS)
On-board dose measurements were made in a shipping vessel for low level radioactive wastes, the Seiei Maru. The measured values are much smaller than the regulation values both on the hatch covers and in the accommodation area. The dose equivalent rates on the hatch cover are analysed by using a continuous energy Monte Carlo code, MCNP 4B, with two kinds of calculational models. One is the detailed model with the geometry of containers and LLW drums, and an asymmetrical source distribution. The results of the detailed calculation approached the shape of the measured dose rate distribution graphs. The other is the simplified model that mixes source volume uniformly. The calculated values obtained with the simplified model are twice as large as those calculated with the detailed model. (author)
Influencing factors of dose equivalence for X and γ rays with different energy based on Monte Carlo
International Nuclear Information System (INIS)
Background: The accuracy of dosimeter measurement of X and γ rays needs to be resolved. Purpose: The aim is to study the correction term of the equivalent process of low energy X-ray and the natural radioactive source. Methods: Instead of the standard sources, X-ray machine was adopted on the dose instrument calibration. The influence factors of the equivalence between low-energy X-ray and high-energy X or γ rays were simulated using Monte Carlo (MCNP) software. Results: The influences of distance, space scattering, response of detector on dose equivalence were obtained. The simulation results were also analyzed. Conclusion: The method can be used in dose equivalent correction for low-energy X-ray, high-energy X or γ rays, which is significant for the widespread use of X rays. (authors)
SU-E-T-117: Dose to Organs Outside of CT Scan Range- Monte Carlo and Hybrid Phantom Approach
International Nuclear Information System (INIS)
Purpose: Epidemiological study of second cancer risk for cancer survivors often requires the dose to normal tissues located outside the anatomy covered by radiological imaging, which is usually limited to tumor and organs at risk. We have investigated the feasibility of using whole body computational human phantoms for estimating out-of-field organ doses for patients treated by Intensity Modulated Radiation Therapy (IMRT). Methods: Identical 7-field IMRT prostate plans were performed using X-ray Voxel Monte Carlo (XVMC), a radiotherapy-specific Monte Carlo transport code, on the computed tomography (CT) images of the torso of an adult male patient (175 cm height, 66 kg weight) and an adult male hybrid computational phantom with the equivalent body size. Dose to the liver, right lung, and left lung were calculated and compared. Results: Considerable differences are seen between the doses calculated by XVMC for the patient CT and the hybrid phantom. One major contributing factor is the treatment method, deep inspiration breath hold (DIBH), used for this patient. This leads to significant differences in the organ position relative to the treatment isocenter. The transverse distances from the treatment isocenter to the inferior border of the liver, left lung, and right lung are 19.5cm, 29.5cm, and 30.0cm, respectively for the patient CT, compared with 24.3cm, 36.6cm, and 39.1cm, respectively, for the hybrid phantom. When corrected for the distance, the mean doses calculated using the hybrid phantom are within 28% of those calculated using the patient CT. Conclusion: This study showed that mean dose to the organs located in the missing CT coverage can be reconstructed by using whole body computational human phantoms within reasonable dosimetric uncertainty, however appropriate corrections may be necessary if the patient is treated with a technique that will significantly deform the size or location of the organs relative to the hybrid phantom
Energy Technology Data Exchange (ETDEWEB)
Brualla, Lorenzo, E-mail: lorenzo.brualla@uni-due.de [NCTeam, Strahlenklinik, Universitaetsklinikum Essen, Essen (Germany); Zaragoza, Francisco J.; Sempau, Josep [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Barcelona (Spain); Wittig, Andrea [Department of Radiation Oncology, University Hospital Giessen and Marburg, Philipps-University Marburg, Marburg (Germany); Sauerwein, Wolfgang [NCTeam, Strahlenklinik, Universitaetsklinikum Essen, Essen (Germany)
2012-07-15
Purpose: External beam radiotherapy is the only conservative curative approach for Stage I non-Hodgkin lymphomas of the conjunctiva. The target volume is geometrically complex because it includes the eyeball and lid conjunctiva. Furthermore, the target volume is adjacent to radiosensitive structures, including the lens, lacrimal glands, cornea, retina, and papilla. The radiotherapy planning and optimization requires accurate calculation of the dose in these anatomical structures that are much smaller than the structures traditionally considered in radiotherapy. Neither conventional treatment planning systems nor dosimetric measurements can reliably determine the dose distribution in these small irradiated volumes. Methods and Materials: The Monte Carlo simulations of a Varian Clinac 2100 C/D and human eye were performed using the PENELOPE and PENEASYLINAC codes. Dose distributions and dose volume histograms were calculated for the bulbar conjunctiva, cornea, lens, retina, papilla, lacrimal gland, and anterior and posterior hemispheres. Results: The simulated results allow choosing the most adequate treatment setup configuration, which is an electron beam energy of 6 MeV with additional bolus and collimation by a cerrobend block with a central cylindrical hole of 3.0 cm diameter and central cylindrical rod of 1.0 cm diameter. Conclusions: Monte Carlo simulation is a useful method to calculate the minute dose distribution in ocular tissue and to optimize the electron irradiation technique in highly critical structures. Using a voxelized eye phantom based on patient computed tomography images, the dose distribution can be estimated with a standard statistical uncertainty of less than 2.4% in 3 min using a computing cluster with 30 cores, which makes this planning technique clinically relevant.
International Nuclear Information System (INIS)
Purpose: External beam radiotherapy is the only conservative curative approach for Stage I non-Hodgkin lymphomas of the conjunctiva. The target volume is geometrically complex because it includes the eyeball and lid conjunctiva. Furthermore, the target volume is adjacent to radiosensitive structures, including the lens, lacrimal glands, cornea, retina, and papilla. The radiotherapy planning and optimization requires accurate calculation of the dose in these anatomical structures that are much smaller than the structures traditionally considered in radiotherapy. Neither conventional treatment planning systems nor dosimetric measurements can reliably determine the dose distribution in these small irradiated volumes. Methods and Materials: The Monte Carlo simulations of a Varian Clinac 2100 C/D and human eye were performed using the PENELOPE and PENEASYLINAC codes. Dose distributions and dose volume histograms were calculated for the bulbar conjunctiva, cornea, lens, retina, papilla, lacrimal gland, and anterior and posterior hemispheres. Results: The simulated results allow choosing the most adequate treatment setup configuration, which is an electron beam energy of 6 MeV with additional bolus and collimation by a cerrobend block with a central cylindrical hole of 3.0 cm diameter and central cylindrical rod of 1.0 cm diameter. Conclusions: Monte Carlo simulation is a useful method to calculate the minute dose distribution in ocular tissue and to optimize the electron irradiation technique in highly critical structures. Using a voxelized eye phantom based on patient computed tomography images, the dose distribution can be estimated with a standard statistical uncertainty of less than 2.4% in 3 min using a computing cluster with 30 cores, which makes this planning technique clinically relevant.
SU-E-T-117: Dose to Organs Outside of CT Scan Range- Monte Carlo and Hybrid Phantom Approach
Energy Technology Data Exchange (ETDEWEB)
Pelletier, C; Jung, J [East Carolina University, Greenville, NC (United States); Lee, C [University of Michigan, Ann Arbor, MI (United States); Kim, J [University of Pittsburgh Medical Center, Pittsburgh, PA (United States); Lee, C [National Cancer Institute, Rockville, MD (United States)
2014-06-01
Purpose: Epidemiological study of second cancer risk for cancer survivors often requires the dose to normal tissues located outside the anatomy covered by radiological imaging, which is usually limited to tumor and organs at risk. We have investigated the feasibility of using whole body computational human phantoms for estimating out-of-field organ doses for patients treated by Intensity Modulated Radiation Therapy (IMRT). Methods: Identical 7-field IMRT prostate plans were performed using X-ray Voxel Monte Carlo (XVMC), a radiotherapy-specific Monte Carlo transport code, on the computed tomography (CT) images of the torso of an adult male patient (175 cm height, 66 kg weight) and an adult male hybrid computational phantom with the equivalent body size. Dose to the liver, right lung, and left lung were calculated and compared. Results: Considerable differences are seen between the doses calculated by XVMC for the patient CT and the hybrid phantom. One major contributing factor is the treatment method, deep inspiration breath hold (DIBH), used for this patient. This leads to significant differences in the organ position relative to the treatment isocenter. The transverse distances from the treatment isocenter to the inferior border of the liver, left lung, and right lung are 19.5cm, 29.5cm, and 30.0cm, respectively for the patient CT, compared with 24.3cm, 36.6cm, and 39.1cm, respectively, for the hybrid phantom. When corrected for the distance, the mean doses calculated using the hybrid phantom are within 28% of those calculated using the patient CT. Conclusion: This study showed that mean dose to the organs located in the missing CT coverage can be reconstructed by using whole body computational human phantoms within reasonable dosimetric uncertainty, however appropriate corrections may be necessary if the patient is treated with a technique that will significantly deform the size or location of the organs relative to the hybrid phantom.
Directory of Open Access Journals (Sweden)
Nilseia Aparecida Barbosa
2014-08-01
Full Text Available Purpose: Melanoma at the choroid region is the most common primary cancer that affects the eye in adult patients. Concave ophthalmic applicators with 106Ru/106Rh beta sources are the more used for treatment of these eye lesions, mainly lesions with small and medium dimensions. The available treatment planning system for 106Ru applicators is based on dose distributions on a homogeneous water sphere eye model, resulting in a lack of data in the literature of dose distributions in the eye radiosensitive structures, information that may be crucial to improve the treatment planning process, aiming the maintenance of visual acuity. Methods: The Monte Carlo code MCNPX was used to calculate the dose distribution in a complete mathematical model of the human eye containing a choroid melanoma; considering the eye actual dimensions and its various component structures, due to an ophthalmic brachytherapy treatment, using 106Ru/106Rh beta-ray sources. Two possibilities were analyzed; a simple water eye and a heterogeneous eye considering all its structures. Two concave applicators, CCA and CCB manufactured by BEBIG and a complete mathematical model of the human eye were modeled using the MCNPX code. Results and Conclusion: For both eye models, namely water model and heterogeneous model, mean dose values simulated for the same eye regions are, in general, very similar, excepting for regions very distant from the applicator, where mean dose values are very low, uncertainties are higher and relative differences may reach 20.4%. For the tumor base and the eye structures closest to the applicator, such as sclera, choroid and retina, the maximum difference observed was 4%, presenting the heterogeneous model higher mean dose values. For the other eye regions, the higher doses were obtained when the homogeneous water eye model is taken into consideration. Mean dose distributions determined for the homogeneous water eye model are similar to those obtained for the
International Nuclear Information System (INIS)
This study aims to compare the doses received for patients submitted to brachytherapy High Dose Rate (HDR) brachytherapy, a method of treatment of the cervix carcinoma, performed in the planning system PLATO BPS with the doses obtained by Monte Carlo simulation using the radiation transport code MCNP 5 and one female anthropomorphic phantom based on voxel, the FAX. The implementation of HDR brachytherapy treatment for the cervix carcinoma consists of the insertion of an intrauterine probe and an intravaginal probe (ring or ovoid) and then two radiographs are obtained, anteroposterior (AP) and lateral (LAT) to confirm the position of the applicators in the patient and to allow the treatment planning and the determination of the absorbed dose at points of interest: rectum, bladder, sigmoid and point A, which corresponds anatomically to the crossings of the uterine arteries with ureters The absorbed doses obtained with the code MCNP 5, with the exception of the absorbed dose in the rectum and sigmoid for the simulation considering a point source of 192Ir, are lower than the absorbed doses from PLATO BPS calculations because the MCNP 5 considers the chemical compositions and densities of FAX body, not considering the medium as water. When considering the Monte Carlo simulation for a source with dimensions equal to that used in the brachytherapy irradiator used in this study, the values of calculated absorbed dose to the bladder, to the rectum, to the right point A and to the left point A were respectively lower than those determined by the treatment planning system in 33.29, 5.01, 22.93 and 19.04%. These values are almost all larger than the maximum acceptable deviation between patient planned and administered doses (5 %). With regard to the rectum and bladder, which are organs that must be protected, the present results are in favor of the radiological protection of patients. The point A, that is on the isodose of 100%, used to tumor treatment, the results indicate an
International Nuclear Information System (INIS)
Full text of publication follows. Purpose: a 3D-Personalized Monte Carlo Dosimetry (PMCD) was developed for treatment planning in nuclear medicine. The method was applied to Selective Internal Radiation Therapy (SIRT) using 90Y-microspheres for unresectable hepatic cancers. Methods: The PMCD method was evaluated for 20 patients treated for hepatic metastases or hepatocellular carcinoma at the European Hospital Georges Pompidou (Paris). First, regions of interest were outlined on the patient CT images. Using the OEDIPE software, patient-specific voxel phantoms were created. 99mTc-MAA SPECT data were then used to generate 3D-matrices of cumulated activity. Absorbed doses and Biologically Effective Dose (BED) were calculated at the voxel scale using the MCNPX Monte Carlo transport code. Finally, OEDIPE was used to determine the maximum injectable activity (MIA) for tolerance criteria on organs at risk (OARs), i.e. the lungs and non tumoral liver (NTL). Tolerance criteria based on mean absorbed doses, mean BED, Dose-Volume Histograms (DVHs) or BED-Volume Histograms (BVHs) were considered. Those MIAs were compared to the Partition Model with tolerance criteria on mean absorbed doses, which is a conventional method applied in clinical practice. Results: compared to Partition Model recommendations, performing dosimetry using the PMCD method enables to increase the activity prescription while ensuring OARs' radiation protection. Moreover, tolerance criteria based on DVHs allow us to enhance treatment planning efficiency by taking advantage of the parallel characteristic of the liver and the lungs, whose functions are not impaired if the level of irradiation to a fraction of the organ is kept sufficiently low. Finally, multi-cycle treatments based on tolerance criteria on mean BED and BVHs, were considered to go further in the dose optimization, taking into account biological considerations such as cell repair or radiosensitivity. Conclusion: besides its feasibility
International Nuclear Information System (INIS)
This paper discusses the simulation of contemporary computed tomography (CT) scanners using Monte Carlo calculation methods to derive normalized organ doses, which enable hospital physicists to estimate typical organ and effective doses for CT examinations. The hardware used in a small PC-cluster at the Health Protection Agency (HPA) for these calculations is described. Investigations concerning optimization of software, including the radiation transport codes MCNP5 and MCNPX, and the Intel and PGI FORTRAN compilers, are presented in relation to results and calculation speed. Differences in approach for modelling the X-ray source are described and their influences are analysed. Comparisons with previously published calculations at HPA from the early 1990's proved satisfactory for the purposes of quality assurance and are presented in terms of organ dose ratios for whole body exposure and differences in organ location. Influences on normalized effective dose are discussed in relation to choice of cross section library, CT scanner technology (contemporary multi slice versus single slice), definition for effective dose (1990 and 2007 versions) and anthropomorphic phantom (mathematical and voxel). The results illustrate the practical need for the updated scanner-specific dose coefficients presently being calculated at HPA, in order to facilitate improved dosimetry for contemporary CT practice. (authors)
An OpenCL-based Monte Carlo dose calculation engine (oclMC) for coupled photon-electron transport
Tian, Zhen; Folkerts, Michael; Qin, Nan; Jiang, Steve B; Jia, Xun
2015-01-01
Monte Carlo (MC) method has been recognized the most accurate dose calculation method for radiotherapy. However, its extremely long computation time impedes clinical applications. Recently, a lot of efforts have been made to realize fast MC dose calculation on GPUs. Nonetheless, most of the GPU-based MC dose engines were developed in NVidia CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a fast cross-platform MC dose engine oclMC using OpenCL environment for external beam photon and electron radiotherapy in MeV energy range. Coupled photon-electron MC simulation was implemented with analogue simulations for photon transports and a Class II condensed history scheme for electron transports. To test the accuracy and efficiency of our dose engine oclMC, we compared dose calculation results of oclMC and gDPM, our previously developed GPU-based MC code, for a 15 MeV electron ...
International Nuclear Information System (INIS)
This study aims to investigate the effects of oblique incidence, small field size and inhomogeneous media on the electron dose distribution, and to compare calculated (Elekta/CMS XiO) and measured results. All comparisons were done in terms of absolute dose. A new measuring method was developed for high resolution, absolute dose measurement of non-standard beams using Gafchromic® EBT3 film. A portable U-shaped holder was designed and constructed to hold EBT3 films vertically in a reproducible setup submerged in a water phantom. The experimental film method was verified with ionisation chamber measurements and agreed to within 2% or 1 mm. Agreement between XiO electron Monte Carlo (eMC) and EBT3 was within 2% or 2 mm for most standard fields and 3% or 3 mm for the non-standard fields. Larger differences were seen in the build-up region where XiO eMC overestimates dose by up to 10% for obliquely incident fields and underestimates the dose for small circular fields by up to 5% when compared to measurement. Calculations with inhomogeneous media mimicking ribs, lung and skull tissue placed at the side of the film in water agreed with measurement to within 3% or 3 mm. Gafchromic film in water proved to be a convenient high spatial resolution method to verify dose distributions from electrons in non-standard conditions including irradiation in inhomogeneous media. (paper)
Stratis, Andreas; Zhang, Guozhi; Jacobs, Reinhilde; Bogaerts, Ria; Bosmans, Hilde
2015-03-01
The aim of this work was to investigate the influence of backscatter radiation from the orbital bone and the intraorbital fat on the eye lens dose in the dental CBCT energy range. To this end we conducted three different yet interrelated studies; A preliminary simulation study was conducted to examine the impact of a bony layer situated underneath a soft tissue layer on the amount of backscatter radiation. We compared the Percentage Depth Dose (PDD) curves in soft tissue with and without the bone layer and we estimated the depth in tissue where the decrease in backscatter caused by the presence of the bone is noticeable. In a supplementary study, an eye voxel phantom was designed with the DOSxyznrc code. Simulations were performed exposing the phantom at different x-ray energies sequentially in air, in fat tissue and in realistic anatomy with the incident beam perpendicular to the phantom. Finally, a virtual head phantom was implemented into a validated hybrid Monte Carlo (MC) framework to simulate a large Field of View protocol of a real CBCT scanner and examine the influence of scattered dose to the eye lens during the whole rotation of the paired tube-detector system. The results indicated an increase in the dose to the lens due to the fatty tissue in the surrounding anatomy. There is a noticeable dose reduction close to the bone-tissue interface which weakens with increasing distance from the interface, such that the impact of the orbital bone in the eye lens dose becomes small.
Monte Carlo evaluation of hand and finger doses due to exposure to 18F in PET procedures
International Nuclear Information System (INIS)
The increasing number of PET procedures performed in nuclear medicine, and, consequently, of workers handling radiopharmaceuticals, is a potential hazard in radiation protection. It is then necessary to evaluate the doses of workers employed in the practice of PET. In this work, the Geant4 Monte Carlo code was used to evaluate doses to fingers and hands of those workers. A geometric phantom, representing the hand of the professional inserted in the clinical procedure, was implemented in the simulation code, with dimensions of a standard man's forearm, which in this case will assess the exposure of the extremities. The geometric phantom is designed so that a simple definition of joint angles configures the fingers, allowing investigations into alternative configurations. Thus, it was possible the placement of the phantom fingers, to simulate all forms of manipulation of a syringe, and subsequently obtain exposure data, relating to the administration procedure of the PET radiopharmaceutical to the patient. The simulation was validated by the irradiation of a REMABR hand phantom, consisting of a human skeleton hand covered by a tenite II shell, which can be filled with water. Air Kerma values were obtained from the beam dosimetry, which was done with a calibrated ionization chamber. The reading of TLD's, placed on certain points of the surface of the phantom, were compared with the values obtained in the Monte Carlo simulation. After validation of the program, we obtained dose values for the PET procedure, simulating syringes with and without shielding. (author)
Directory of Open Access Journals (Sweden)
Ilić Radovan D.
2004-01-01
Full Text Available This paper describes the application of SRNA Monte Carlo package for proton transport simulations in complex geometry and different material composition. SRNA package was developed for 3D dose distribution calculation in proton therapy and dosimetry and it was based on the theory of multiple scattering. The compound nuclei decay was simulated by our own and the Russian MSDM models using ICRU 63 data. The developed package consists of two codes SRNA-2KG, which simulates proton transport in the combinatorial geometry and SRNA-VOX, which uses the voxelized geometry using the CT data and conversion of the Hounsfield’s data to tissue elemental composition. Transition probabilities for both codes are prepared by the SRNADAT code. The simulation of proton beam characterization by Multi-Layer Faraday Cup, spatial distribution of positron emitters obtained by SRNA-2KG code, and intercomparison of computational codes in radiation dosimetry, indicate the immediate application of the Monte Carlo techniques in clinical practice. In this paper, we briefly present the physical model implemented in SRNA pack age, the ISTAR proton dose planning software, as well as the results of the numerical experiments with proton beams to obtain 3D dose distribution in the eye and breast tumor.
Monte Carlo evaluation of hand and finger doses due to exposure to {sup 18}F in PET procedures
Energy Technology Data Exchange (ETDEWEB)
Pessanha, Paula R.; Queiroz Filho, Pedro P.; Santos, Denison S.; Mauricio, Claudia L.P., E-mail: paula@ird.gov.br, E-mail: queiroz@ird.gov.br, E-mail: santosd@ird.gov.br, E-mail: claudia@ird.gov.br [Instituto de Radioprotecao e Dosimetria (IRD/CNEN-RJ), Rio de Janeiro, RJ (Brazil)
2011-07-01
The increasing number of PET procedures performed in nuclear medicine, and, consequently, of workers handling radiopharmaceuticals, is a potential hazard in radiation protection. It is then necessary to evaluate the doses of workers employed in the practice of PET. In this work, the Geant4 Monte Carlo code was used to evaluate doses to fingers and hands of those workers. A geometric phantom, representing the hand of the professional inserted in the clinical procedure, was implemented in the simulation code, with dimensions of a standard man's forearm, which in this case will assess the exposure of the extremities. The geometric phantom is designed so that a simple definition of joint angles configures the fingers, allowing investigations into alternative configurations. Thus, it was possible the placement of the phantom fingers, to simulate all forms of manipulation of a syringe, and subsequently obtain exposure data, relating to the administration procedure of the PET radiopharmaceutical to the patient. The simulation was validated by the irradiation of a REMAB{sup R} hand phantom, consisting of a human skeleton hand covered by a tenite II shell, which can be filled with water. Air Kerma values were obtained from the beam dosimetry, which was done with a calibrated ionization chamber. The reading of TLD's, placed on certain points of the surface of the phantom, were compared with the values obtained in the Monte Carlo simulation. After validation of the program, we obtained dose values for the PET procedure, simulating syringes with and without shielding. (author)
International Nuclear Information System (INIS)
This paper describes the application of SRNA Monte Carlo package for proton transport simulations in complex geometry and different material composition. SRNA package was developed for 3D dose distribution calculation in proton therapy and dosimetry and it was based on the theory of multiple scattering. The compound nuclei decay was simulated by our own and the Russian MSDM models using ICRU 63 data. The developed package consists of two codes: SRNA-2KG, which simulates proton transport in the combinatorial geometry and SRNA-VOX, which uses the voxelized geometry using the CT data and conversion of the Hounsfield's data to tissue elemental composition. Transition probabilities for both codes are prepared by the SRNADAT code. The simulation of proton beam characterization by Multi-Layer Faraday Cup, spatial distribution of positron emitters obtained by SRNA-2KG code, and intercomparison of computational codes in radiation dosimetry, indicate the immediate application of the Monte Carlo techniques in clinical practice. In this paper, we briefly present the physical model implemented in SRNA package, the ISTAR proton dose planning software, as well as the results of the numerical experiments with proton beams to obtain 3D dose distribution in the eye and breast tumor. (author)
The Monte Carlo SRNA-VOX code for 3D proton dose distribution in voxelized geometry using CT data
Energy Technology Data Exchange (ETDEWEB)
Ilic, Radovan D [Laboratory of Physics (010), Vinca Institute of Nuclear Sciences, PO Box 522, 11001 Belgrade (Serbia and Montenegro); Spasic-Jokic, Vesna [Laboratory of Physics (010), Vinca Institute of Nuclear Sciences, PO Box 522, 11001 Belgrade (Serbia and Montenegro); Belicev, Petar [Laboratory of Physics (010), Vinca Institute of Nuclear Sciences, PO Box 522, 11001 Belgrade (Serbia and Montenegro); Dragovic, Milos [Center for Nuclear Medicine MEDICA NUCLEARE, Bulevar Despota Stefana 69, 11000 Belgrade (Serbia and Montenegro)
2005-03-07
This paper describes the application of the SRNA Monte Carlo package for proton transport simulations in complex geometry and different material compositions. The SRNA package was developed for 3D dose distribution calculation in proton therapy and dosimetry and it was based on the theory of multiple scattering. The decay of proton induced compound nuclei was simulated by the Russian MSDM model and our own using ICRU 63 data. The developed package consists of two codes: the SRNA-2KG, which simulates proton transport in combinatorial geometry and the SRNA-VOX, which uses the voxelized geometry using the CT data and conversion of the Hounsfield's data to tissue elemental composition. Transition probabilities for both codes are prepared by the SRNADAT code. The simulation of the proton beam characterization by multi-layer Faraday cup, spatial distribution of positron emitters obtained by the SRNA-2KG code and intercomparison of computational codes in radiation dosimetry, indicate immediate application of the Monte Carlo techniques in clinical practice. In this paper, we briefly present the physical model implemented in the SRNA package, the ISTAR proton dose planning software, as well as the results of the numerical experiments with proton beams to obtain 3D dose distribution in the eye and breast tumour.
The Monte Carlo SRNA-VOX code for 3D proton dose distribution in voxelized geometry using CT data
Ilic, Radovan D.; Spasic-Jokic, Vesna; Belicev, Petar; Dragovic, Milos
2005-03-01
This paper describes the application of the SRNA Monte Carlo package for proton transport simulations in complex geometry and different material compositions. The SRNA package was developed for 3D dose distribution calculation in proton therapy and dosimetry and it was based on the theory of multiple scattering. The decay of proton induced compound nuclei was simulated by the Russian MSDM model and our own using ICRU 63 data. The developed package consists of two codes: the SRNA-2KG, which simulates proton transport in combinatorial geometry and the SRNA-VOX, which uses the voxelized geometry using the CT data and conversion of the Hounsfield's data to tissue elemental composition. Transition probabilities for both codes are prepared by the SRNADAT code. The simulation of the proton beam characterization by multi-layer Faraday cup, spatial distribution of positron emitters obtained by the SRNA-2KG code and intercomparison of computational codes in radiation dosimetry, indicate immediate application of the Monte Carlo techniques in clinical practice. In this paper, we briefly present the physical model implemented in the SRNA package, the ISTAR proton dose planning software, as well as the results of the numerical experiments with proton beams to obtain 3D dose distribution in the eye and breast tumour.
The Monte Carlo SRNA-VOX code for 3D proton dose distribution in voxelized geometry using CT data
International Nuclear Information System (INIS)
This paper describes the application of the SRNA Monte Carlo package for proton transport simulations in complex geometry and different material compositions. The SRNA package was developed for 3D dose distribution calculation in proton therapy and dosimetry and it was based on the theory of multiple scattering. The decay of proton induced compound nuclei was simulated by the Russian MSDM model and our own using ICRU 63 data. The developed package consists of two codes: the SRNA-2KG, which simulates proton transport in combinatorial geometry and the SRNA-VOX, which uses the voxelized geometry using the CT data and conversion of the Hounsfield's data to tissue elemental composition. Transition probabilities for both codes are prepared by the SRNADAT code. The simulation of the proton beam characterization by multi-layer Faraday cup, spatial distribution of positron emitters obtained by the SRNA-2KG code and intercomparison of computational codes in radiation dosimetry, indicate immediate application of the Monte Carlo techniques in clinical practice. In this paper, we briefly present the physical model implemented in the SRNA package, the ISTAR proton dose planning software, as well as the results of the numerical experiments with proton beams to obtain 3D dose distribution in the eye and breast tumour
Energy Technology Data Exchange (ETDEWEB)
Kang, Sei-Kwon; Yoon, Jai-Woong; Hwang, Taejin; Park, Soah; Cheong, Kwang-Ho; Jin Han, Tae; Kim, Haeyoung; Lee, Me-Yeon; Ju Kim, Kyoung, E-mail: kjkim@hallym.or.kr; Bae, Hoonsik
2015-10-01
A metallic contact eye shield has sometimes been used for eyelid treatment, but dose distribution has never been reported for a patient case. This study aimed to show the shield-incorporated CT-based dose distribution using the Pinnacle system and Monte Carlo (MC) calculation for 3 patient cases. For the artifact-free CT scan, an acrylic shield machined as the same size as that of the tungsten shield was used. For the MC calculation, BEAMnrc and DOSXYZnrc were used for the 6-MeV electron beam of the Varian 21EX, in which information for the tungsten, stainless steel, and aluminum material for the eye shield was used. The same plan was generated on the Pinnacle system and both were compared. The use of the acrylic shield produced clear CT images, enabling delineation of the regions of interest, and yielded CT-based dose calculation for the metallic shield. Both the MC and the Pinnacle systems showed a similar dose distribution downstream of the eye shield, reflecting the blocking effect of the metallic eye shield. The major difference between the MC and the Pinnacle results was the target eyelid dose upstream of the shield such that the Pinnacle system underestimated the dose by 19 to 28% and 11 to 18% for the maximum and the mean doses, respectively. The pattern of dose difference between the MC and the Pinnacle systems was similar to that in the previous phantom study. In conclusion, the metallic eye shield was successfully incorporated into the CT-based planning, and the accurate dose calculation requires MC simulation.
International Nuclear Information System (INIS)
A metallic contact eye shield has sometimes been used for eyelid treatment, but dose distribution has never been reported for a patient case. This study aimed to show the shield-incorporated CT-based dose distribution using the Pinnacle system and Monte Carlo (MC) calculation for 3 patient cases. For the artifact-free CT scan, an acrylic shield machined as the same size as that of the tungsten shield was used. For the MC calculation, BEAMnrc and DOSXYZnrc were used for the 6-MeV electron beam of the Varian 21EX, in which information for the tungsten, stainless steel, and aluminum material for the eye shield was used. The same plan was generated on the Pinnacle system and both were compared. The use of the acrylic shield produced clear CT images, enabling delineation of the regions of interest, and yielded CT-based dose calculation for the metallic shield. Both the MC and the Pinnacle systems showed a similar dose distribution downstream of the eye shield, reflecting the blocking effect of the metallic eye shield. The major difference between the MC and the Pinnacle results was the target eyelid dose upstream of the shield such that the Pinnacle system underestimated the dose by 19 to 28% and 11 to 18% for the maximum and the mean doses, respectively. The pattern of dose difference between the MC and the Pinnacle systems was similar to that in the previous phantom study. In conclusion, the metallic eye shield was successfully incorporated into the CT-based planning, and the accurate dose calculation requires MC simulation
Energy Technology Data Exchange (ETDEWEB)
Rodrigues, Anna; Yin, Fang-Fang; Wu, Qiuwen, E-mail: Qiuwen.Wu@Duke.edu [Department of Radiation Oncology, Duke University Medical Center, Durham, North Carolina 27710 and Medical Physics Graduate Program, Duke University Medical Center, Durham, North Carolina 27705 (United States); Sawkey, Daren [Varian Medical Systems, Palo Alto, California 94304 (United States)
2015-05-15
Purpose: To develop a framework for accurate electron Monte Carlo dose calculation. In this study, comprehensive validations of vendor provided electron beam phase space files for Varian TrueBeam Linacs against measurement data are presented. Methods: In this framework, the Monte Carlo generated phase space files were provided by the vendor and used as input to the downstream plan-specific simulations including jaws, electron applicators, and water phantom computed in the EGSnrc environment. The phase space files were generated based on open field commissioning data. A subset of electron energies of 6, 9, 12, 16, and 20 MeV and open and collimated field sizes 3 × 3, 4 × 4, 5 × 5, 6 × 6, 10 × 10, 15 × 15, 20 × 20, and 25 × 25 cm{sup 2} were evaluated. Measurements acquired with a CC13 cylindrical ionization chamber and electron diode detector and simulations from this framework were compared for a water phantom geometry. The evaluation metrics include percent depth dose, orthogonal and diagonal profiles at depths R{sub 100}, R{sub 50}, R{sub p}, and R{sub p+} for standard and extended source-to-surface distances (SSD), as well as cone and cut-out output factors. Results: Agreement for the percent depth dose and orthogonal profiles between measurement and Monte Carlo was generally within 2% or 1 mm. The largest discrepancies were observed within depths of 5 mm from phantom surface. Differences in field size, penumbra, and flatness for the orthogonal profiles at depths R{sub 100}, R{sub 50}, and R{sub p} were within 1 mm, 1 mm, and 2%, respectively. Orthogonal profiles at SSDs of 100 and 120 cm showed the same level of agreement. Cone and cut-out output factors agreed well with maximum differences within 2.5% for 6 MeV and 1% for all other energies. Cone output factors at extended SSDs of 105, 110, 115, and 120 cm exhibited similar levels of agreement. Conclusions: We have presented a Monte Carlo simulation framework for electron beam dose calculations for
International Nuclear Information System (INIS)
Purpose: To develop a framework for accurate electron Monte Carlo dose calculation. In this study, comprehensive validations of vendor provided electron beam phase space files for Varian TrueBeam Linacs against measurement data are presented. Methods: In this framework, the Monte Carlo generated phase space files were provided by the vendor and used as input to the downstream plan-specific simulations including jaws, electron applicators, and water phantom computed in the EGSnrc environment. The phase space files were generated based on open field commissioning data. A subset of electron energies of 6, 9, 12, 16, and 20 MeV and open and collimated field sizes 3 × 3, 4 × 4, 5 × 5, 6 × 6, 10 × 10, 15 × 15, 20 × 20, and 25 × 25 cm2 were evaluated. Measurements acquired with a CC13 cylindrical ionization chamber and electron diode detector and simulations from this framework were compared for a water phantom geometry. The evaluation metrics include percent depth dose, orthogonal and diagonal profiles at depths R100, R50, Rp, and Rp+ for standard and extended source-to-surface distances (SSD), as well as cone and cut-out output factors. Results: Agreement for the percent depth dose and orthogonal profiles between measurement and Monte Carlo was generally within 2% or 1 mm. The largest discrepancies were observed within depths of 5 mm from phantom surface. Differences in field size, penumbra, and flatness for the orthogonal profiles at depths R100, R50, and Rp were within 1 mm, 1 mm, and 2%, respectively. Orthogonal profiles at SSDs of 100 and 120 cm showed the same level of agreement. Cone and cut-out output factors agreed well with maximum differences within 2.5% for 6 MeV and 1% for all other energies. Cone output factors at extended SSDs of 105, 110, 115, and 120 cm exhibited similar levels of agreement. Conclusions: We have presented a Monte Carlo simulation framework for electron beam dose calculations for Varian TrueBeam Linacs. Electron beam energies of 6
Monte Carlo simulations of the dose from imaging with GE eXplore 120 micro-CT using GATE
Energy Technology Data Exchange (ETDEWEB)
Bretin, Florian; Bahri, Mohamed Ali; Luxen, André; Phillips, Christophe; Plenevaux, Alain; Seret, Alain, E-mail: aseret@ulg.ac.be [Cyclotron Research Centre, University of Liège, Sart Tilman B30, Liège 4000 (Belgium)
2015-10-15
Purpose: Small animals are increasingly used as translational models in preclinical imaging studies involving microCT, during which the subjects can be exposed to large amounts of radiation. While the radiation levels are generally sublethal, studies have shown that low-level radiation can change physiological parameters in mice. In order to rule out any influence of radiation on the outcome of such experiments, or resulting deterministic effects in the subjects, the levels of radiation involved need to be addressed. The aim of this study was to investigate the radiation dose delivered by the GE eXplore 120 microCT non-invasively using Monte Carlo simulations in GATE and to compare results to previously obtained experimental values. Methods: Tungsten X-ray spectra were simulated at 70, 80, and 97 kVp using an analytical tool and their half-value layers were simulated for spectra validation against experimentally measured values of the physical X-ray tube. A Monte Carlo model of the microCT system was set up and four protocols that are regularly applied to live animal scanning were implemented. The computed tomography dose index (CTDI) inside a PMMA phantom was derived and multiple field of view acquisitions were simulated using the PMMA phantom, a representative mouse and rat. Results: Simulated half-value layers agreed with experimentally obtained results within a 7% error window. The CTDI ranged from 20 to 56 mGy and closely matched experimental values. Derived organ doses in mice reached 459 mGy in bones and up to 200 mGy in soft tissue organs using the highest energy protocol. Dose levels in rats were lower due to the increased mass of the animal compared to mice. The uncertainty of all dose simulations was below 14%. Conclusions: Monte Carlo simulations proved a valuable tool to investigate the 3D dose distribution in animals from microCT. Small animals, especially mice (due to their small volume), receive large amounts of radiation from the GE eXplore 120
Monte Carlo simulations of the dose from imaging with GE eXplore 120 micro-CT using GATE
International Nuclear Information System (INIS)
Purpose: Small animals are increasingly used as translational models in preclinical imaging studies involving microCT, during which the subjects can be exposed to large amounts of radiation. While the radiation levels are generally sublethal, studies have shown that low-level radiation can change physiological parameters in mice. In order to rule out any influence of radiation on the outcome of such experiments, or resulting deterministic effects in the subjects, the levels of radiation involved need to be addressed. The aim of this study was to investigate the radiation dose delivered by the GE eXplore 120 microCT non-invasively using Monte Carlo simulations in GATE and to compare results to previously obtained experimental values. Methods: Tungsten X-ray spectra were simulated at 70, 80, and 97 kVp using an analytical tool and their half-value layers were simulated for spectra validation against experimentally measured values of the physical X-ray tube. A Monte Carlo model of the microCT system was set up and four protocols that are regularly applied to live animal scanning were implemented. The computed tomography dose index (CTDI) inside a PMMA phantom was derived and multiple field of view acquisitions were simulated using the PMMA phantom, a representative mouse and rat. Results: Simulated half-value layers agreed with experimentally obtained results within a 7% error window. The CTDI ranged from 20 to 56 mGy and closely matched experimental values. Derived organ doses in mice reached 459 mGy in bones and up to 200 mGy in soft tissue organs using the highest energy protocol. Dose levels in rats were lower due to the increased mass of the animal compared to mice. The uncertainty of all dose simulations was below 14%. Conclusions: Monte Carlo simulations proved a valuable tool to investigate the 3D dose distribution in animals from microCT. Small animals, especially mice (due to their small volume), receive large amounts of radiation from the GE eXplore 120
International Nuclear Information System (INIS)
Purpose: To develop a comprehensive end-to-end test for Varian's TrueBeam linear accelerator for head and neck IMRT using a custom phantom designed to utilize multiple dosimetry devices. Purpose: To commission a multiple-source Monte Carlo model of Elekta linear accelerator beams of nominal energies 6MV and 10MV. Methods: A three source, Monte Carlo model of Elekta 6 and 10MV therapeutic x-ray beams was developed. Energy spectra of two photon sources corresponding to primary photons created in the target and scattered photons originating in the linear accelerator head were determined by an optimization process that fit the relative fluence of 0.25 MeV energy bins to the product of Fatigue-Life and Fermi functions to match calculated percent depth dose (PDD) data with that measured in a water tank for a 10x10cm2 field. Off-axis effects were modeled by a 3rd degree polynomial used to describe the off-axis half-value layer as a function of off-axis angle and fitting the off-axis fluence to a piecewise linear function to match calculated dose profiles with measured dose profiles for a 40×40cm2 field. The model was validated by comparing calculated PDDs and dose profiles for field sizes ranging from 3×3cm2 to 30×30cm2 to those obtained from measurements. A benchmarking study compared calculated data to measurements for IMRT plans delivered to anthropomorphic phantoms. Results: Along the central axis of the beam 99.6% and 99.7% of all data passed the 2%/2mm gamma criterion for 6 and 10MV models, respectively. Dose profiles at depths of dmax, through 25cm agreed with measured data for 99.4% and 99.6% of data tested for 6 and 10MV models, respectively. A comparison of calculated dose to film measurement in a head and neck phantom showed an average of 85.3% and 90.5% of pixels passing a 3%/2mm gamma criterion for 6 and 10MV models respectively. Conclusion: A Monte Carlo multiple-source model for Elekta 6 and 10MV therapeutic x-ray beams has been developed as a
Monte Carlo treatment planning for photon and electron beams
Reynaert, N.; van der Marck, S. C.; Schaart, D. R.; Van der Zee, W.; Van Vliet-Vroegindeweij, C.; Tomsej, M.; Jansen, J.; Heijmen, B.; Coghe, M.; De Wagter, C.
2007-04-01
During the last few decades, accuracy in photon and electron radiotherapy has increased substantially. This is partly due to enhanced linear accelerator technology, providing more flexibility in field definition (e.g. the usage of computer-controlled dynamic multileaf collimators), which led to intensity modulated radiotherapy (IMRT). Important improvements have also been made in the treatment planning process, more specifically in the dose calculations. Originally, dose calculations relied heavily on analytic, semi-analytic and empirical algorithms. The more accurate convolution/superposition codes use pre-calculated Monte Carlo dose "kernels" partly accounting for tissue density heterogeneities. It is generally recognized that the Monte Carlo method is able to increase accuracy even further. Since the second half of the 1990s, several Monte Carlo dose engines for radiotherapy treatment planning have been introduced. To enable the use of a Monte Carlo treatment planning (MCTP) dose engine in clinical circumstances, approximations have been introduced to limit the calculation time. In this paper, the literature on MCTP is reviewed, focussing on patient modeling, approximations in linear accelerator modeling and variance reduction techniques. An overview of published comparisons between MC dose engines and conventional dose calculations is provided for phantom studies and clinical examples, evaluating the added value of MCTP in the clinic. An overview of existing Monte Carlo dose engines and commercial MCTP systems is presented and some specific issues concerning the commissioning of a MCTP system are discussed.
Energy Technology Data Exchange (ETDEWEB)
Blazy-Aubignac, L
2007-09-15
The treatment planning systems (T.P.S.) occupy a key position in the radiotherapy service: they realize the projected calculation of the dose distribution and the treatment duration. Traditionally, the quality control of the calculated distribution doses relies on their comparisons with dose distributions measured under the device of treatment. This thesis proposes to substitute these dosimetry measures to the profile of reference dosimetry calculations got by the Penelope Monte-Carlo code. The Monte-Carlo simulations give a broad choice of test configurations and allow to envisage a quality control of dosimetry aspects of T.P.S. without monopolizing the treatment devices. This quality control, based on the Monte-Carlo simulations has been tested on a clinical T.P.S. and has allowed to simplify the quality procedures of the T.P.S.. This quality control, in depth, more precise and simpler to implement could be generalized to every center of radiotherapy. (N.C.)
Karimian, A; Nikparvar, B; Jabbari, I
2014-11-01
Renal angiography is one of the medical imaging methods in which patient and physician receive high equivalent doses due to long duration of fluoroscopy. In this research, equivalent doses of some radiosensitive tissues of patient (adult and child) and physician during renal angiography have been calculated by using adult and child Oak Ridge National Laboratory phantoms and Monte Carlo method (MCNPX). The results showed, in angiography of right kidney in a child and adult patient, that gall bladder with the amounts of 2.32 and 0.35 mSv, respectively, has received the most equivalent dose. About the physician, left hand, left eye and thymus absorbed the most amounts of doses, means 0.020 mSv. In addition, equivalent doses of the physician's lens eye, thyroid and knees were 0.023, 0.007 and 7.9E-4 mSv, respectively. Although these values are less than the reported thresholds by ICRP 103, it should be noted that these amounts are related to one examination. PMID:25063788
International Nuclear Information System (INIS)
Renal angiography is one of the medical imaging methods in which patient and physician receive high equivalent doses due to long duration of fluoroscopy. In this research, equivalent doses of some radiosensitive tissues of patient (adult and child) and physician during renal angiography have been calculated by using adult and child Oak Ridge National Laboratory phantoms and Monte Carlo method (MCNPX). The results showed, in angiography of right kidney in a child and adult patient, that gall bladder with the amounts of 2.32 and 0.35 mSv, respectively, has received the most equivalent dose. About the physician, left hand, left eye and thymus absorbed the most amounts of doses, means 0.020 mSv. In addition, equivalent doses of the physician's lens eye, thyroid and knees were 0.023, 0.007 and 7.9 - 4 mSv, respectively. Although these values are less than the reported thresholds by ICRP 103, it should be noted that these amounts are related to one examination. (authors)
Energy Technology Data Exchange (ETDEWEB)
Garcia, Marie-Paule, E-mail: marie-paule.garcia@univ-brest.fr; Villoing, Daphnée [UMR 1037 INSERM/UPS, CRCT, 133 Route de Narbonne, 31062 Toulouse (France); McKay, Erin [St George Hospital, Gray Street, Kogarah, New South Wales 2217 (Australia); Ferrer, Ludovic [ICO René Gauducheau, Boulevard Jacques Monod, St Herblain 44805 (France); Cremonesi, Marta; Botta, Francesca; Ferrari, Mahila [European Institute of Oncology, Via Ripamonti 435, Milano 20141 (Italy); Bardiès, Manuel [UMR 1037 INSERM/UPS, CRCT, 133 Route de Narbonne, Toulouse 31062 (France)
2015-12-15
Purpose: The TestDose platform was developed to generate scintigraphic imaging protocols and associated dosimetry by Monte Carlo modeling. TestDose is part of a broader project (www.dositest.com) whose aim is to identify the biases induced by different clinical dosimetry protocols. Methods: The TestDose software allows handling the whole pipeline from virtual patient generation to resulting planar and SPECT images and dosimetry calculations. The originality of their approach relies on the implementation of functional segmentation for the anthropomorphic model representing a virtual patient. Two anthropomorphic models are currently available: 4D XCAT and ICRP 110. A pharmacokinetic model describes the biodistribution of a given radiopharmaceutical in each defined compartment at various time-points. The Monte Carlo simulation toolkit GATE offers the possibility to accurately simulate scintigraphic images and absorbed doses in volumes of interest. The TestDose platform relies on GATE to reproduce precisely any imaging protocol and to provide reference dosimetry. For image generation, TestDose stores user’s imaging requirements and generates automatically command files used as input for GATE. Each compartment is simulated only once and the resulting output is weighted using pharmacokinetic data. Resulting compartment projections are aggregated to obtain the final image. For dosimetry computation, emission data are stored in the platform database and relevant GATE input files are generated for the virtual patient model and associated pharmacokinetics. Results: Two samples of software runs are given to demonstrate the potential of TestDose. A clinical imaging protocol for the Octreoscan™ therapeutical treatment was implemented using the 4D XCAT model. Whole-body “step and shoot” acquisitions at different times postinjection and one SPECT acquisition were generated within reasonable computation times. Based on the same Octreoscan™ kinetics, a dosimetry
International Nuclear Information System (INIS)
Purpose: The TestDose platform was developed to generate scintigraphic imaging protocols and associated dosimetry by Monte Carlo modeling. TestDose is part of a broader project (www.dositest.com) whose aim is to identify the biases induced by different clinical dosimetry protocols. Methods: The TestDose software allows handling the whole pipeline from virtual patient generation to resulting planar and SPECT images and dosimetry calculations. The originality of their approach relies on the implementation of functional segmentation for the anthropomorphic model representing a virtual patient. Two anthropomorphic models are currently available: 4D XCAT and ICRP 110. A pharmacokinetic model describes the biodistribution of a given radiopharmaceutical in each defined compartment at various time-points. The Monte Carlo simulation toolkit GATE offers the possibility to accurately simulate scintigraphic images and absorbed doses in volumes of interest. The TestDose platform relies on GATE to reproduce precisely any imaging protocol and to provide reference dosimetry. For image generation, TestDose stores user’s imaging requirements and generates automatically command files used as input for GATE. Each compartment is simulated only once and the resulting output is weighted using pharmacokinetic data. Resulting compartment projections are aggregated to obtain the final image. For dosimetry computation, emission data are stored in the platform database and relevant GATE input files are generated for the virtual patient model and associated pharmacokinetics. Results: Two samples of software runs are given to demonstrate the potential of TestDose. A clinical imaging protocol for the Octreoscan™ therapeutical treatment was implemented using the 4D XCAT model. Whole-body “step and shoot” acquisitions at different times postinjection and one SPECT acquisition were generated within reasonable computation times. Based on the same Octreoscan™ kinetics, a dosimetry
International Nuclear Information System (INIS)
Over the last decade, there was a substantial increase in the number of interventional cardiology procedures worldwide, and the corresponding ionizing radiation doses for both the medical staff and patients became a subject of concern. Interventional procedures in cardiology are normally very complex, resulting in long exposure times. Also, these interventions require the operator to work near the patient and, consequently, close to the primary X-ray beam. Moreover, due to the scattered radiation from the patient and the equipment, the medical staff is also exposed to a non-uniform radiation field that can lead to a significant exposure of sensitive body organs and tissues, such as the eye lens, the thyroid and the extremities. In order to better understand the spatial variation of the dose and dose rate distributions during an interventional cardiology procedure, the dose distribution around a C-arm fluoroscopic system, in operation in a cardiac cath lab at Portuguese Hospital, was estimated using both Monte Carlo (MC) simulations and dosimetric measurements. To model and simulate the cardiac cath lab, including the fluoroscopic equipment used to execute interventional procedures, the state-of-the-art MC radiation transport code MCNPX 2.7.0 was used. Subsequently, Thermo-Luminescent Detector (TLD) measurements were performed, in order to validate and support the simulation results obtained for the cath lab model. The preliminary results presented in this study reveal that the cardiac cath lab model was successfully validated, taking into account the good agreement between MC calculations and TLD measurements. The simulated results for the isodose curves related to the C-arm fluoroscopic system are also consistent with the dosimetric information provided by the equipment manufacturer (Siemens). The adequacy of the implemented computational model used to simulate complex procedures and map dose distributions around the operator and the medical staff is discussed, in
International Nuclear Information System (INIS)
The dose distribution calculation is one of major steps in cancer radiotherapy. This paper applies Monte Carlo code, MCNP5, in simulation 15 MV photon beams from linear accelerator of General Hospital of Kien Giang in a case treatment of lung cancer. The settings for beam direction, field size and isocenter position used in MCNP5 must be the same as in treatment plan at hospital to ensure the results from MCNP5 are accurate. We also built a program CODIM by using MATLAB® programming software. This program is used to construct digital voxel phantoms from lung CT images obtained from cancer treatment cases at Kien Giang hospital and then simulate the delivered dose of linac in these phantoms by using MCNP5 simulation code. The results show that there is a difference of 5% in comparison to Prowess Panther program - a semi-empirical simulation program which is being used for treatment planning in Kien Giang hospital. (author)
International Nuclear Information System (INIS)
The present paper reports on the structure and first results from a new Monte Carlo programme for calculations of energy distributions within tissue equivalent phantoms irradiated from π--beams. Each pion or generated secondary particle is transported until to the complete loss of its kinetic energy taking into account pion processes like multiple Coulomb scattering, pion reactions in flight and absorption of stopped pions. The code uses mainly data from experiments, and physical models have been added only in cases of lacking data. Depth dose curves for a pensil beam of 170 MeV/c within a water phantom are discussed as a function of various parameters. Isodose contours are plotted resulting from a convolution of an extended beam profile and the dose distribution of a pencil beams. (orig.)
A GPU OpenCL based cross-platform Monte Carlo dose calculation engine (goMC)
International Nuclear Information System (INIS)
Monte Carlo (MC) simulation has been recognized as the most accurate dose calculation method for radiotherapy. However, the extremely long computation time impedes its clinical application. Recently, a lot of effort has been made to realize fast MC dose calculation on graphic processing units (GPUs). However, most of the GPU-based MC dose engines have been developed under NVidia’s CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a GPU OpenCL based cross-platform MC dose engine named goMC with coupled photon–electron simulation for external photon and electron radiotherapy in the MeV energy range. Compared to our previously developed GPU-based MC code named gDPM (Jia et al 2012 Phys. Med. Biol. 57 7783–97), goMC has two major differences. First, it was developed under the OpenCL environment for high code portability and hence could be run not only on different GPU cards but also on CPU platforms. Second, we adopted the electron transport model used in EGSnrc MC package and PENELOPE’s random hinge method in our new dose engine, instead of the dose planning method employed in gDPM. Dose distributions were calculated for a 15 MeV electron beam and a 6 MV photon beam in a homogenous water phantom, a water-bone-lung-water slab phantom and a half-slab phantom. Satisfactory agreement between the two MC dose engines goMC and gDPM was observed in all cases. The average dose differences in the regions that received a dose higher than 10% of the maximum dose were 0.48–0.53% for the electron beam cases and 0.15–0.17% for the photon beam cases. In terms of efficiency, goMC was ∼4–16% slower than gDPM when running on the same NVidia TITAN card for all the cases we tested, due to both the different electron transport models and the different development environments. The code portability of our new dose engine goMC was
A GPU OpenCL based cross-platform Monte Carlo dose calculation engine (goMC)
Tian, Zhen; Shi, Feng; Folkerts, Michael; Qin, Nan; Jiang, Steve B.; Jia, Xun
2015-09-01
Monte Carlo (MC) simulation has been recognized as the most accurate dose calculation method for radiotherapy. However, the extremely long computation time impedes its clinical application. Recently, a lot of effort has been made to realize fast MC dose calculation on graphic processing units (GPUs). However, most of the GPU-based MC dose engines have been developed under NVidia’s CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a GPU OpenCL based cross-platform MC dose engine named goMC with coupled photon-electron simulation for external photon and electron radiotherapy in the MeV energy range. Compared to our previously developed GPU-based MC code named gDPM (Jia et al 2012 Phys. Med. Biol. 57 7783-97), goMC has two major differences. First, it was developed under the OpenCL environment for high code portability and hence could be run not only on different GPU cards but also on CPU platforms. Second, we adopted the electron transport model used in EGSnrc MC package and PENELOPE’s random hinge method in our new dose engine, instead of the dose planning method employed in gDPM. Dose distributions were calculated for a 15 MeV electron beam and a 6 MV photon beam in a homogenous water phantom, a water-bone-lung-water slab phantom and a half-slab phantom. Satisfactory agreement between the two MC dose engines goMC and gDPM was observed in all cases. The average dose differences in the regions that received a dose higher than 10% of the maximum dose were 0.48-0.53% for the electron beam cases and 0.15-0.17% for the photon beam cases. In terms of efficiency, goMC was ~4-16% slower than gDPM when running on the same NVidia TITAN card for all the cases we tested, due to both the different electron transport models and the different development environments. The code portability of our new dose engine goMC was validated by
A GPU OpenCL based cross-platform Monte Carlo dose calculation engine (goMC).
Tian, Zhen; Shi, Feng; Folkerts, Michael; Qin, Nan; Jiang, Steve B; Jia, Xun
2015-10-01
Monte Carlo (MC) simulation has been recognized as the most accurate dose calculation method for radiotherapy. However, the extremely long computation time impedes its clinical application. Recently, a lot of effort has been made to realize fast MC dose calculation on graphic processing units (GPUs). However, most of the GPU-based MC dose engines have been developed under NVidia's CUDA environment. This limits the code portability to other platforms, hindering the introduction of GPU-based MC simulations to clinical practice. The objective of this paper is to develop a GPU OpenCL based cross-platform MC dose engine named goMC with coupled photon-electron simulation for external photon and electron radiotherapy in the MeV energy range. Compared to our previously developed GPU-based MC code named gDPM (Jia et al 2012 Phys. Med. Biol. 57 7783-97), goMC has two major differences. First, it was developed under the OpenCL environment for high code portability and hence could be run not only on different GPU cards but also on CPU platforms. Second, we adopted the electron transport model used in EGSnrc MC package and PENELOPE's random hinge method in our new dose engine, instead of the dose planning method employed in gDPM. Dose distributions were calculated for a 15 MeV electron beam and a 6 MV photon beam in a homogenous water phantom, a water-bone-lung-water slab phantom and a half-slab phantom. Satisfactory agreement between the two MC dose engines goMC and gDPM was observed in all cases. The average dose differences in the regions that received a dose higher than 10% of the maximum dose were 0.48-0.53% for the electron beam cases and 0.15-0.17% for the photon beam cases. In terms of efficiency, goMC was ~4-16% slower than gDPM when running on the same NVidia TITAN card for all the cases we tested, due to both the different electron transport models and the different development environments. The code portability of our new dose engine goMC was validated by
International Nuclear Information System (INIS)
Radiation exposure of patients during endovascular aneurysm repair (EVAR) procedures ranks in the upper sector of medical exposure. Thus, estimation of radiation doses achieved during EVAR is of great importance. Organ doses (OD) and effective doses (ED) administered to 17 patients receiving EVAR were determined (1) from the exposure parameters by performing Monte Carlo simulations in mathematical phantoms and (2) by measurements with thermoluminescent dosimeters in a physical anthropomorphic phantom. The mean fluoroscopy time was 26 min, the mean dose area product was 24995 cGy cm2. The mean ED was 34.8 mSv, ODs up to 626 mSv were found. Whereas digital subtraction angiographies (DSA) and fluoroscopies each contributed about 50 % to the cumulative ED, the ED rates of DSAs were found to be ten times higher than those of fluoroscopies. Doubling of the field size caused an ED rate enhancement up to a factor of 3. EVAR procedures cause high radiation exposure levels that exceed the values published thus far. As a consequence, (1) DSAs should be only performed when necessary and with a low image rate, (2) fluoroscopies should be kept as short as possible, and (3) field sizes should be minimized. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Foerth, Monika; Treitl, Karla Maria; Treitl, Marcus [Ludwig Maximilians University of Munich, Institute for Clinical Radiology, Munich (Germany); Seidenbusch, Michael C. [Ludwig Maximilians University of Munich, Institute for Clinical Radiology, Munich (Germany); Clinical Centre of the Ludwig Maximilian University of Munich, Institute for Clinical Radiology, Munich (Germany); Sadeghi-Azandaryani, Mojtaba [Clinical Centre of the County of Erding, Department of Vascular Surgery, Erding (Germany); Lechel, Ursula [Federal Office for Radiation Protection, Department of Medical and Occupational Radiation Protection, Oberschleissheim (Germany)
2015-09-15
Radiation exposure of patients during endovascular aneurysm repair (EVAR) procedures ranks in the upper sector of medical exposure. Thus, estimation of radiation doses achieved during EVAR is of great importance. Organ doses (OD) and effective doses (ED) administered to 17 patients receiving EVAR were determined (1) from the exposure parameters by performing Monte Carlo simulations in mathematical phantoms and (2) by measurements with thermoluminescent dosimeters in a physical anthropomorphic phantom. The mean fluoroscopy time was 26 min, the mean dose area product was 24995 cGy cm2. The mean ED was 34.8 mSv, ODs up to 626 mSv were found. Whereas digital subtraction angiographies (DSA) and fluoroscopies each contributed about 50 % to the cumulative ED, the ED rates of DSAs were found to be ten times higher than those of fluoroscopies. Doubling of the field size caused an ED rate enhancement up to a factor of 3. EVAR procedures cause high radiation exposure levels that exceed the values published thus far. As a consequence, (1) DSAs should be only performed when necessary and with a low image rate, (2) fluoroscopies should be kept as short as possible, and (3) field sizes should be minimized. (orig.)
Directory of Open Access Journals (Sweden)
Nitin Ramesh Kakade
2015-01-01
Full Text Available Background: Gold nanoparticle (GNP-aided radiation therapy (RT is useful to make the tumor more sensitive to radiation damage because of the enhancement in the dose inside the tumor region. Polymer gel dosimeter (PGD can be a good choice for the physical measurement of dose enhancement produced by GNP inside the gel. Materials and Methods: The present study uses EGSnrc Monte Carlo code to estimate dose enhancement factor (DEF due to the introduction of GNPs inside the PGD at different concentrations (7 and 18 mg Au/g of gel when irradiated by therapeutic X-rays of energy 100 kVp, 150 kVp, 6 MV, and 15 MV. The simulation was also carried out to quantify the dose enhancement in PAGAT gel and tumor for 100 kVp X-rays. Results: For 100 kVp X-rays, average DEF of 1.86 and 2.91 is observed in the PAGAT gel dosimeter with 7 and 18 mg Au/g of gel, respectively. Average DEF of 1.69 and 2.61 is recorded for 150 kVp X-rays with 7 and 18 mg Au/g of gel, respectively. No clinically meaningful DEF was observed for 6 and 15 MV photon beams. Furthermore, the dose enhancement within the PAGAT gel dosimeter and tumor closely matches with each other. Conclusion: The polymer gel dosimetry can be a suitable method of dose estimation and verification for clinical implementation of GNP-aided RT. GNP-aided RT has the potential of delivering high localized tumoricidal dose with significant sparing of normal structures when the treatment is delivered with low energy X-rays.
Çatli, Serap
2015-01-01
High atomic number and density of dental implants leads to major problems at providing an accurate dose distribution in radiotherapy and contouring tumors and organs caused by the artifact in head and neck tumors. The limits and deficiencies of the algorithms using in the treatment planning systems can lead to large errors in dose calculation, and this may adversely affect the patient's treatment. In the present study, four commercial dental implants were used: pure titanium, titanium alloy (Ti-6Al-4V), amalgam, and crown. The effects of dental implants on dose distribution are determined with two methods: pencil beam convolution (PBC) algorithm and Monte Carlo code for 6 MV photon beam. The central axis depth doses were calculated on the phantom for a source-skin distance (SSD) of 100 cm and a 10 × 10 cm2 field using both of algorithms. The results of Monte Carlo method and Eclipse TPS were compared to each other and to those previously reported. In the present study, dose increases in tissue at a distance of 2 mm in front of the dental implants were seen due to the backscatter of electrons for dental implants at 6 MV using the Monte Carlo method. The Eclipse treatment planning system (TPS) couldn't precisely account for the backscatter radiation caused by the dental prostheses. TPS underestimated the back scatter dose and overestimated the dose after the dental implants. The large errors found for TPS in this study are due to the limits and deficiencies of the algorithms. The accuracy of the PBC algorithm of Eclipse TPS was evaluated in comparison to Monte Carlo calculations in consideration of the recommendations of the American Association of Physicists in Medicine Radiation Therapy Committee Task Group 65. From the comparisons of the TPS and Monte Carlo calculations, it is verified that the Monte Carlo simulation is a good approach to derive the dose distribution in heterogeneous media. PMID:26699323
Monte Carlo Calculations of Dose to Medium and Dose to Water for Carbon Ion Beams in Various Media
DEFF Research Database (Denmark)
Herrmann, Rochus; Petersen, Jørgen B.B.; Jäkel, Oliver;
materials exposed to carbon ion beams. The scored track-length fluence spectrum Φi for a given particle i at the energy E, is multiplied with the mass stopping power for target material for calculating Dm . Similarly, Dw is calculated by multiplying the same fluence spectrum with the mass stopping power...... the PSTAR, ASTAR stopping power routines available at NIST1 and MSTAR2 provided by H. Paul et al. 3 Results For a pristine carbon ion beam we encountered a maximum deviation between Dw and Dm up to 8% for bone. In addition we investigate spread out Bragg peak configurations which dilutes the effect......1 Background In clinical practice the quantity dose to water (Dw ) is used as a reference standard for dosimeters and treatment planning systems. Treatment planning systems usually rely on analytical representation of the particle beam, which are normally expressed as dose with respect to water...
Estimation of staff doses in complex radiological examinations using a Monte Carlo computer code
International Nuclear Information System (INIS)
The protection of medical personnel in interventional radiology is an important issue of radiological protection. The irradiation of the worker is largely non-uniform, and a large part of his body is shielded by a lead apron. The estimation of effective dose (E) under these conditions is difficult and several approaches are used to estimate effective dose involving such a protective apron. This study presents a summary from an extensive series of simulations to determine scatter-dose distribution around the patient and staff effective dose from personal dosimeter readings. The influence of different parameters (like beam energy and size, patient size, irradiated region, worker position and orientation) on the staff doses has been determined. Published algorithms that combine readings of an unshielded and a shielded dosimeter to estimate effective dose have been applied and a new algorithm, that gives more accurate dose estimates for a wide range of situations was proposed. A computational approach was used to determine the dose distribution in the worker's body. The radiation transport and energy deposition was simulated using the MCNP4B code. The human bodies of the patient and radiologist were generated with the Body Builder anthropomorphic model-generating tool. The radiologist is protected with a lead apron (0.5 mm lead equivalent in the front and 0.25 mm lead equivalent in the back and sides) and a thyroid collar (0.35 mm lead equivalent). The lower-arms of the worker were folded to simulate the arms position during clinical examinations. This realistic situation of the folded arms affects the effective dose to the worker. Depending on the worker position and orientation (and of course the beam energy), the difference can go up to 25 percent. A total of 12 Hp(10) dosimeters were positioned above and under the lead apron at the neck, chest and waist levels. Extra dosimeters for the skin dose were positioned at the forehead, the forearms and the front surface of
Mohammadyari, Parvin; Faghihi, Reza; Mosleh-Shirazi, Mohammad Amin; Lotfi, Mehrzad; Hematiyan, Mohammad Rahim; Koontz, Craig; Meigooni, Ali S
2015-12-01
Compression is a technique to immobilize the target or improve the dose distribution within the treatment volume during different irradiation techniques such as AccuBoost(®) brachytherapy. However, there is no systematic method for determination of dose distribution for uncompressed tissue after irradiation under compression. In this study, the mechanical behavior of breast tissue between compressed and uncompressed states was investigated. With that, a novel method was developed to determine the dose distribution in uncompressed tissue after irradiation of compressed breast tissue. Dosimetry was performed using two different methods, namely, Monte Carlo simulations using the MCNP5 code and measurements using thermoluminescent dosimeters (TLD). The displacement of the breast elements was simulated using a finite element model and calculated using ABAQUS software. From these results, the 3D dose distribution in uncompressed tissue was determined. The geometry of the model was constructed from magnetic resonance images of six different women volunteers. The mechanical properties were modeled by using the Mooney-Rivlin hyperelastic material model. Experimental dosimetry was performed by placing the TLD chips into the polyvinyl alcohol breast equivalent phantom. The results determined that the nodal displacements, due to the gravitational force and the 60 Newton compression forces (with 43% contraction in the loading direction and 37% expansion in the orthogonal direction) were determined. Finally, a comparison of the experimental data and the simulated data showed agreement within 11.5% ± 5.9%. PMID:26572554
International Nuclear Information System (INIS)
Compression is a technique to immobilize the target or improve the dose distribution within the treatment volume during different irradiation techniques such as AccuBoost® brachytherapy. However, there is no systematic method for determination of dose distribution for uncompressed tissue after irradiation under compression. In this study, the mechanical behavior of breast tissue between compressed and uncompressed states was investigated. With that, a novel method was developed to determine the dose distribution in uncompressed tissue after irradiation of compressed breast tissue. Dosimetry was performed using two different methods, namely, Monte Carlo simulations using the MCNP5 code and measurements using thermoluminescent dosimeters (TLD). The displacement of the breast elements was simulated using a finite element model and calculated using ABAQUS software. From these results, the 3D dose distribution in uncompressed tissue was determined. The geometry of the model was constructed from magnetic resonance images of six different women volunteers. The mechanical properties were modeled by using the Mooney–Rivlin hyperelastic material model. Experimental dosimetry was performed by placing the TLD chips into the polyvinyl alcohol breast equivalent phantom. The results determined that the nodal displacements, due to the gravitational force and the 60 Newton compression forces (with 43% contraction in the loading direction and 37% expansion in the orthogonal direction) were determined. Finally, a comparison of the experimental data and the simulated data showed agreement within 11.5% ± 5.9%. (paper)
Mohammadyari, Parvin; Faghihi, Reza; Mosleh-Shirazi, Mohammad Amin; Lotfi, Mehrzad; Rahim Hematiyan, Mohammad; Koontz, Craig; Meigooni, Ali S.
2015-12-01
Compression is a technique to immobilize the target or improve the dose distribution within the treatment volume during different irradiation techniques such as AccuBoost® brachytherapy. However, there is no systematic method for determination of dose distribution for uncompressed tissue after irradiation under compression. In this study, the mechanical behavior of breast tissue between compressed and uncompressed states was investigated. With that, a novel method was developed to determine the dose distribution in uncompressed tissue after irradiation of compressed breast tissue. Dosimetry was performed using two different methods, namely, Monte Carlo simulations using the MCNP5 code and measurements using thermoluminescent dosimeters (TLD). The displacement of the breast elements was simulated using a finite element model and calculated using ABAQUS software. From these results, the 3D dose distribution in uncompressed tissue was determined. The geometry of the model was constructed from magnetic resonance images of six different women volunteers. The mechanical properties were modeled by using the Mooney-Rivlin hyperelastic material model. Experimental dosimetry was performed by placing the TLD chips into the polyvinyl alcohol breast equivalent phantom. The results determined that the nodal displacements, due to the gravitational force and the 60 Newton compression forces (with 43% contraction in the loading direction and 37% expansion in the orthogonal direction) were determined. Finally, a comparison of the experimental data and the simulated data showed agreement within 11.5% ± 5.9%.
International Nuclear Information System (INIS)
Full text: In this study irradiation geometry applicable to PCXMC and the consequent calculation of effective dose in applications of cone beam computed tomography (CBCT) was developed. Two different CBCT equipment s for dental applications were evaluated: Care Stream Cs-9000 3-Dimensional and Gendex GXCB-500 tomographs. Each protocol initially was characterized by measuring the surface kerma input and the product air kerma-area, PKA. Then, technical parameters of each of the predetermined protocols and geometric conditions in the PCXMC software were introduced to obtain the values of effective dose. The calculated effective dose is within the range of 9.0 to 15.7 μSv for Cs 9000 3-D and in the range 44.5 to 89 mSv for GXCB-500 equipment. These values were compared with dosimetric results obtained using thermoluminescent dosimeters implanted in anthropomorphic mannequin and were considered consistent. The effective dose results are very sensitive to the radiation geometry (beam position); this represents a factor of fragility software usage, but on the other hand, turns out to be a very useful tool for quick conclusions regarding the optimization process of protocols. We can conclude that the use of Monte Carlo simulation software PCXMC is useful in the evaluation of test protocols of CBCT in dental applications. (Author)
International Nuclear Information System (INIS)
This work studies the effects of corrections in the calculation of dose distribution for brachytherapy sources when they are inserted in a male human voxel phantom. The sources studied here are the Best Industries 125I 2301 model for low dose rate and the Amersham Buchler G0814 model 192Ir seed for high dose rate, in the simulation of prostate treatments. The presence of organs around the interest point scatters radiation in a different form than a water cube, the situation that is usually configured in these calculations. The insertion of the sources in an anthropomorphic phantom brings results closer to the real situation. The chosen phantom was the head and torso voxel phantom created by Zubal. The Geant4 Monte Carlo toolkit was used to simulate the radiation transportation along the source shielding and the human organs of the voxel phantom. After inserting the source in the phantom, the energy deposition in each voxel is computed, allowing the construction of isodose curves. The source insertion in the anthropomorphic phantom aims also at a further knowledge about the brachytherapy treatment planning and additional information such as the target volume dose and in neighbor organs, data that will be useful for medical staff working with this technique. (author)
Directory of Open Access Journals (Sweden)
Ghavami Seyed Mostafa
2016-01-01
Full Text Available Using the nano-scaled radionuclides in the radionuclide therapy significantly reduces the particles trapping in the organs vessels and avoids thrombosis formations. Additionally, uniform distribution in the target organ may be another benefit of the nanoradionuclides in the radionuclide therapy. Monte Carlo simulation was conducted to model a mathematical humanoid phantom and the liver cells of the simulated phantom were filled with the 90Y nanospheres. Healthy organs doses, fatal and nonfatal risks of the surrounding organs were estimated. The estimations and calculations were made in four different distribution patterns of the radionuclide seeds. Maximum doses and risks estimated for the surrounding organs were obtained in the high edge concentrated distribution model of the liver including the nanoradionuclides. For the dose equivalent, effective dose, fatal and non-fatal risks, the values obtained as 7.51E-03 Sv/Bq, 3.01E-01 Sv/Bq, and 9.16E-01 cases/104 persons for the bladder, colon, and kidney of the modeled phantom, respectively. The mentioned values were the maximum values among the studied modeled distributions. Maximum values of Normal Tissue Complication Probability for the healthy organs calculated as 5.9-8.9 %. Result of using nanoparticles of the 90Y provides promising dosimetric properties in MC simulation results considering non-toxicity reports for the radionuclide.
Energy Technology Data Exchange (ETDEWEB)
Hardiansyah, D.; Haryanto, F. [Nuclear Physics and Biophysics Research Laboratory, Physics Department, Institut Teknologi Bandung (ITB) (Indonesia); Male, S. [Radiotherapy Division, Research Hospital of Hassanudin University (Indonesia)
2014-09-30
Prism is a non-commercial Radiotherapy Treatment Planning System (RTPS) develop by Ira J. Kalet from Washington University. Inhomogeneity factor is included in Prism TPS dose calculation. The aim of this study is to investigate the sensitivity of dose calculation on Prism using Monte Carlo simulation. Phase space source from head linear accelerator (LINAC) for Monte Carlo simulation is implemented. To achieve this aim, Prism dose calculation is compared with EGSnrc Monte Carlo simulation. Percentage depth dose (PDD) and R50 from both calculations are observed. BEAMnrc is simulated electron transport in LINAC head and produced phase space file. This file is used as DOSXYZnrc input to simulated electron transport in phantom. This study is started with commissioning process in water phantom. Commissioning process is adjusted Monte Carlo simulation with Prism RTPS. Commissioning result is used for study of inhomogeneity phantom. Physical parameters of inhomogeneity phantom that varied in this study are: density, location and thickness of tissue. Commissioning result is shown that optimum energy of Monte Carlo simulation for 6 MeV electron beam is 6.8 MeV. This commissioning is used R50 and PDD with Practical length (R{sub p}) as references. From inhomogeneity study, the average deviation for all case on interest region is below 5 %. Based on ICRU recommendations, Prism has good ability to calculate the radiation dose in inhomogeneity tissue.
Hardiansyah, D.; Male, S.; Haryanto, F.
2014-09-01
Prism is a non-commercial Radiotherapy Treatment Planning System (RTPS) develop by Ira J. Kalet from Washington University. Inhomogeneity factor is included in Prism TPS dose calculation. The aim of this study is to investigate the sensitivity of dose calculation on Prism using Monte Carlo simulation. Phase space source from head linear accelerator (LINAC) for Monte Carlo simulation is implemented. To achieve this aim, Prism dose calculation is compared with EGSnrc Monte Carlo simulation. Percentage depth dose (PDD) and R50 from both calculations are observed. BEAMnrc is simulated electron transport in LINAC head and produced phase space file. This file is used as DOSXYZnrc input to simulated electron transport in phantom. This study is started with commissioning process in water phantom. Commissioning process is adjusted Monte Carlo simulation with Prism RTPS. Commissioning result is used for study of inhomogeneity phantom. Physical parameters of inhomogeneity phantom that varied in this study are: density, location and thickness of tissue. Commissioning result is shown that optimum energy of Monte Carlo simulation for 6 MeV electron beam is 6.8 MeV. This commissioning is used R50 and PDD with Practical length (Rp) as references. From inhomogeneity study, the average deviation for all case on interest region is below 5 %. Based on ICRU recommendations, Prism has good ability to calculate the radiation dose in inhomogeneity tissue.
International Nuclear Information System (INIS)
Prism is a non-commercial Radiotherapy Treatment Planning System (RTPS) develop by Ira J. Kalet from Washington University. Inhomogeneity factor is included in Prism TPS dose calculation. The aim of this study is to investigate the sensitivity of dose calculation on Prism using Monte Carlo simulation. Phase space source from head linear accelerator (LINAC) for Monte Carlo simulation is implemented. To achieve this aim, Prism dose calculation is compared with EGSnrc Monte Carlo simulation. Percentage depth dose (PDD) and R50 from both calculations are observed. BEAMnrc is simulated electron transport in LINAC head and produced phase space file. This file is used as DOSXYZnrc input to simulated electron transport in phantom. This study is started with commissioning process in water phantom. Commissioning process is adjusted Monte Carlo simulation with Prism RTPS. Commissioning result is used for study of inhomogeneity phantom. Physical parameters of inhomogeneity phantom that varied in this study are: density, location and thickness of tissue. Commissioning result is shown that optimum energy of Monte Carlo simulation for 6 MeV electron beam is 6.8 MeV. This commissioning is used R50 and PDD with Practical length (Rp) as references. From inhomogeneity study, the average deviation for all case on interest region is below 5 %. Based on ICRU recommendations, Prism has good ability to calculate the radiation dose in inhomogeneity tissue
Bousis, C; Emfietzoglou, D; Hadjidoukas, P; Nikjoo, H
2008-07-21
Monte Carlo transport calculations of dose point kernels (DPKs) and depth dose profiles (DDPs) in both the vapor and liquid phases of water are presented for electrons with initial energy between 10 keV and 1 MeV. The results are obtained by the MC4 code using three different implementations of the condensed-history technique for inelastic collisions, namely the continuous slowing down approximation, the mixed-simulation with delta-ray transport and the addition of straggling distributions for soft collisions derived from accurate relativistic Born cross sections. In all schemes, elastic collisions are simulated individually based on single-scattering cross sections. Electron transport below 10 keV is performed in an event-by-event mode. Differences on inelastic interactions between the vapor and liquid phase are treated explicitly using our recently developed dielectric response function which is supplemented by relativistic corrections and the transverse contribution. On the whole, the interaction coefficients used agree to better than approximately 5% with NIST/ICRU values. It is shown that condensed phase effects in both DPKs and DDPs practically vanish above 100 keV. The effect of delta-rays, although decreases with energy, is sizeable leading to more diffused distributions, especially for DPKs. The addition of straggling for soft collisions is practically inconsequential above a few hundred keV. An extensive benchmarking with other condensed-history codes is provided. PMID:18574312
International Nuclear Information System (INIS)
Computed tomography is a specialized x-ray imaging technique that has found increasing use since its introduction into diagnostic radiology in the early 1979 's. Although CT is a diagnostic imaging modality giving higher patient dose in comparison with other radiological procedure, but the good resolution of its images, increases the use of this modality in diagnostic imaging with introduction of spiral CT in 1989 and multi slice spiral CT in the early 1990 's, some radiation protection committee tried to calculate the patient dose during CT exam. Since the direct measurement of patient dose and its relation with scan techniques such as mA, KV, Pitch and slice thickness is very difficult , the Monte Carlo method was introduced as the most powerful tool for measuring the organ and patient absorbed dose. Nowadays , the predominant method for assessment of patient's organ absorbed dose is the application of conversion coefficients established by use of Monte Carlo simulation, one of the most famous dose calculation software is CT dosimetry which makes use of the NRPB Monte Carlo dose data sets produced in report SR 250. In this work we use CT Dosimetry Monte Carlo package for measurement of the relationship between the patient dose and scan techniques in mathematically standard phantom( Hermaphrodite 170 cm / 70 Kg) for an abdomen -pelvis scan in single and multi slice spiral CT. As we expected, by increasing the mA and KV the patient dose increases but the relation between the effective dose and mA is linear, also in both kind of scanners the patient dose increases with decreasing the pitch factor. And finally although the multi slice CT is faster than single slice systems, but the patient dose in multi slice systems is more than single slice scanners
Energy Technology Data Exchange (ETDEWEB)
Lima, Alexandre Roza de
2014-07-01
The International Commission on Radiological Protection, ICRP, in its publication 103, reviewed recent epidemiological evidence and indicated that, for the eye lens, the absorbed dose threshold for induction of late detriment is around 0.5 Gy. On this basis, on April 21, 2011, the ICRP recommended changes to the occupational dose limit in planned exposure situations, reducing the eye lens equivalent dose limit from 150 mSv to 20 mSv per year, on average, during the period of 5 years, with exposure not to exceed 50 mSv in a single year. This paper presents the dose estimation to eye lens, H{sub p}(10), effective dose and doses to important organs in the body, received by industrial gamma radiography workers, during planned or accidental exposure situations. The computer program Visual Monte Carlo was used and two relevant scenarios were postulated. The first is a planned exposure situation scenario where the operator is directly exposed to radiation during the operation. 12 radiographic exposures per day for 250 days per year, which leads to an exposure of 36,000 seconds or 10 hours per year were considered. The simulation was carried out using the following parameters: a {sup 192}Ir source with 1.0 TBq of activity, the source/operator distance varying from 5 m to 10 m at three different heights of 0.2 m, 1.0 m and 2.0 m. The eyes lens doses were estimated as being between 16.9 mSv/year and 66.9 mSv/year and for H{sub p}(10) the doses were between 17.7 mSv/year and 74.2 mSv/year. For the accidental exposure situation scenario, the same radionuclide and activity were used, but in this case the doses were calculated with and without a collimator. The heights above ground considered were 1.0 m, 1.5 m e 2.0 m, the source/operator distance was 40 cm and, the exposure time 74 seconds. The eyes lens doses, for 1.5 m, were 12.3 mGy and 0.28 mGy without and with a collimator, respectively. Three conclusions resulted from this work. The first was that the estimated doses show
Bozkurt, Ahmet
The distribution of absorbed doses in the body can be computationally determined using mathematical or tomographic representations of human anatomy. A whole- body model was developed from the color images of the National Library of Medicine's Visible Human Project® for simulating the transport of radiation in the human body. The model, called Visible Photographic Man (VIP-Man), has sixty-one organs and tissues represented in the Monte Carlo code MCNPX at 4-mm voxel resolution. Organ dose calculations from external neutron sources were carried out using VIP-man and MCNPX to determine a new set of dose conversion coefficients to be used in radiation protection. Monoenergetic neutron beams between 10-9 MeV and 10 GeV were studied under six different irradiation geometries: anterior-posterior, posterior-anterior, right lateral, left lateral, rotational and isotropic. The results for absorbed doses in twenty-four organs and the effective doses based on twelve critical organs are presented in tabular form. A comprehensive comparison of the results with those from the mathematical models show discrepancies that can be attributed to the variations in body modeling (size, location and shape of the individual organs) and the use of different nuclear datasets or models to derive the reaction cross sections, as well as the use of different transport packages for simulation radiation effects. The organ dose results based on the realistic VIP-Man body model allow the existing radiation protection dosimetry on neutrons to be re-evaluated and improved.
International Nuclear Information System (INIS)
Computed tomography has advanced very rapidly, resulting in a rapid increase in the number of examinations performed, as well as in the range of clinical applications now based on this technique. CT is a procedure generally associated with relatively high dose levels. It constitutes 5% of worldwide radiological examinations; yet contributes one third of the collective dose associated with medical imaging. Therefore, studies on the issue of dose optimization are welcome. Computer simulation uses theoretical models to predict performance of real systems. This kind of simulation is able, among other tasks, to evaluate the impact of geometric parameters of a tomograph on spatial resolution, for example: detector size; source-detector distance and source-isocentre distance. The development of new pre-process and image reconstruction algorithms and the evaluation of dose in organs and tissues of human body are tasks that have been developed by means of computer simulations. In this work, conventional CT acquisitions were completely simulated. Monte Carlo techniques were applied in radiation transport simulation to obtain projection data and to calculate conversion coefficients for doses in organs and tissues of a female anthropomorphic phantom. The computer simulation of the entire measurement and reconstruction process is a valuable and very effective means to evaluate the effects of individual parameters. A graphical reconstruction algorithm was used to obtain virtual tomographic images, the attenuation values of each projection were added to the related pixel of the reconstruction matrix. Several different geometries were simulated and the effect of geometric parameters on image quality and dose in organs and tissues was evaluated. (author)
International Nuclear Information System (INIS)
Monte Carlo calculations were used to investigate the efficiency of radiation protection equipment in reducing eye and whole body doses during fluoroscopically guided interventional procedures. Eye lens doses were determined considering different models of eyewear with various shapes, sizes and lead thickness. The origin of scattered radiation reaching the eyes was also assessed to explain the variation in the protection efficiency of the different eyewear models with exposure conditions. The work also investigates the variation of eye and whole body doses with ceiling-suspended shields of various shapes and positioning. For all simulations, a broad spectrum of configurations typical for most interventional procedures was considered. Calculations showed that ‘wrap around’ glasses are the most efficient eyewear models reducing, on average, the dose by 74% and 21% for the left and right eyes respectively. The air gap between the glasses and the eyes was found to be the primary source of scattered radiation reaching the eyes. The ceiling-suspended screens were more efficient when positioned close to the patient’s skin and to the x-ray field. With the use of such shields, the Hp(10) values recorded at the collar, chest and waist level and the Hp(3) values for both eyes were reduced on average by 47%, 37%, 20% and 56% respectively. Finally, simulations proved that beam quality and lead thickness have little influence on eye dose while beam projection, the position and head orientation of the operator as well as the distance between the image detector and the patient are key parameters affecting eye and whole body doses. (paper)
TH-A-19A-10: Fast Four Dimensional Monte Carlo Dose Computations for Proton Therapy of Lung Cancer
International Nuclear Information System (INIS)
Purpose: To develop and validate a fast and accurate four dimensional (4D) Monte Carlo (MC) dose computation system for proton therapy of lung cancer and other thoracic and abdominal malignancies in which the delivered dose distributions can be affected by respiratory motion of the patient. Methods: A 4D computer tomography (CT) scan for a lung cancer patient treated with protons in our clinic was used to create a time dependent patient model using our in-house, MCNPX-based Monte Carlo system (“MC2”). The beam line configurations for two passively scattered proton beams used in the actual treatment were extracted from the clinical treatment plan and a set of input files was created automatically using MC2. A full MC simulation of the beam line was computed using MCNPX and a set of phase space files for each beam was collected at the distal surface of the range compensator. The particles from these phase space files were transported through the 10 voxelized patient models corresponding to the 10 phases of the breathing cycle in the 4DCT, using MCNPX and an accelerated (fast) MC code called “FDC”, developed by us and which is based on the track repeating algorithm. The accuracy of the fast algorithm was assessed by comparing the two time dependent dose distributions. Results: The error of less than 1% in 100% of the voxels in all phases of the breathing cycle was achieved using this method with a speedup of more than 1000 times. Conclusion: The proposed method, which uses full MC to simulate the beam line and the accelerated MC code FDC for the time consuming particle transport inside the complex, time dependent, geometry of the patient shows excellent accuracy together with an extraordinary speed
TH-A-19A-10: Fast Four Dimensional Monte Carlo Dose Computations for Proton Therapy of Lung Cancer
Energy Technology Data Exchange (ETDEWEB)
Mirkovic, D; Titt, U; Mohan, R [U.T M.D. Anderson Cancer Center, Houston, TX (United States); Yepes, P [Rice University, Houston, TX (United States)
2014-06-15
Purpose: To develop and validate a fast and accurate four dimensional (4D) Monte Carlo (MC) dose computation system for proton therapy of lung cancer and other thoracic and abdominal malignancies in which the delivered dose distributions can be affected by respiratory motion of the patient. Methods: A 4D computer tomography (CT) scan for a lung cancer patient treated with protons in our clinic was used to create a time dependent patient model using our in-house, MCNPX-based Monte Carlo system (“MC{sup 2}”). The beam line configurations for two passively scattered proton beams used in the actual treatment were extracted from the clinical treatment plan and a set of input files was created automatically using MC{sup 2}. A full MC simulation of the beam line was computed using MCNPX and a set of phase space files for each beam was collected at the distal surface of the range compensator. The particles from these phase space files were transported through the 10 voxelized patient models corresponding to the 10 phases of the breathing cycle in the 4DCT, using MCNPX and an accelerated (fast) MC code called “FDC”, developed by us and which is based on the track repeating algorithm. The accuracy of the fast algorithm was assessed by comparing the two time dependent dose distributions. Results: The error of less than 1% in 100% of the voxels in all phases of the breathing cycle was achieved using this method with a speedup of more than 1000 times. Conclusion: The proposed method, which uses full MC to simulate the beam line and the accelerated MC code FDC for the time consuming particle transport inside the complex, time dependent, geometry of the patient shows excellent accuracy together with an extraordinary speed.
International Nuclear Information System (INIS)
A hybrid approach, combining deterministic and Monte Carlo (MC) calculations, is proposed to compute the distribution of dose deposited during stereotactic synchrotron radiation therapy treatment. The proposed approach divides the computation into two parts: (i) the dose deposited by primary radiation (coming directly from the incident x-ray beam) is calculated in a deterministic way using ray casting techniques and energy-absorption coefficient tables and (ii) the dose deposited by secondary radiation (Rayleigh and Compton scattering, fluorescence) is computed using a hybrid algorithm combining MC and deterministic calculations. In the MC part, a small number of particle histories are simulated. Every time a scattering or fluorescence event takes place, a splitting mechanism is applied, so that multiple secondary photons are generated with a reduced weight. The secondary events are further processed in a deterministic way, using ray casting techniques. The whole simulation, carried out within the framework of the Monte Carlo code Geant4, is shown to converge towards the same results as the full MC simulation. The speed of convergence is found to depend notably on the splitting multiplicity, which can easily be optimized. To assess the performance of the proposed algorithm, we compare it to state-of-the-art MC simulations, accelerated by the track length estimator technique (TLE), considering a clinically realistic test case. It is found that the hybrid approach is significantly faster than the MC/TLE method. The gain in speed in a test case was about 25 for a constant precision. Therefore, this method appears to be suitable for treatment planning applications.
Energy Technology Data Exchange (ETDEWEB)
Smekens, F; Freud, N; Letang, J M; Babot, D [CNDRI (Nondestructive Testing using Ionizing Radiations) Laboratory, INSA-Lyon, 69621 Villeurbanne Cedex (France); Adam, J-F; Elleaume, H; Esteve, F [INSERM U-836, Equipe 6 ' Rayonnement Synchrotron et Recherche Medicale' , Institut des Neurosciences de Grenoble (France); Ferrero, C; Bravin, A [European Synchrotron Radiation Facility, Grenoble (France)], E-mail: francois.smekens@insa-lyon.fr
2009-08-07
A hybrid approach, combining deterministic and Monte Carlo (MC) calculations, is proposed to compute the distribution of dose deposited during stereotactic synchrotron radiation therapy treatment. The proposed approach divides the computation into two parts: (i) the dose deposited by primary radiation (coming directly from the incident x-ray beam) is calculated in a deterministic way using ray casting techniques and energy-absorption coefficient tables and (ii) the dose deposited by secondary radiation (Rayleigh and Compton scattering, fluorescence) is computed using a hybrid algorithm combining MC and deterministic calculations. In the MC part, a small number of particle histories are simulated. Every time a scattering or fluorescence event takes place, a splitting mechanism is applied, so that multiple secondary photons are generated with a reduced weight. The secondary events are further processed in a deterministic way, using ray casting techniques. The whole simulation, carried out within the framework of the Monte Carlo code Geant4, is shown to converge towards the same results as the full MC simulation. The speed of convergence is found to depend notably on the splitting multiplicity, which can easily be optimized. To assess the performance of the proposed algorithm, we compare it to state-of-the-art MC simulations, accelerated by the track length estimator technique (TLE), considering a clinically realistic test case. It is found that the hybrid approach is significantly faster than the MC/TLE method. The gain in speed in a test case was about 25 for a constant precision. Therefore, this method appears to be suitable for treatment planning applications.
International Nuclear Information System (INIS)
Field gamma spectrometers are widely used to determine gamma dose rates in sedimentary media. However the most widely used technique-the 'window technique'-is time consuming and introduces important statistical uncertainty in the determination of the radioelement contents, and finally on the gamma dose rate. The threshold technique directly relates the number of counts recorded above certain threshold energy to the gamma dose rate. Recently new experimental measurements further investigated this technique but it has not been tested in various sedimentary media. In this paper, numerical simulations using a specifically designed Geant4 code allow to test the sensitivity of this technique to changes of sediments nature, humidity content and disequilibrium in the U-series. Finally another threshold technique, relating the gamma dose rate to the energy per unit time deposited above another threshold energy, is investigated. It is shown than the latter has a number of advantages compared to the classical techniques. Experimental results testing this approach are presented.
A clinical study of lung cancer dose calculation accuracy with Monte Carlo simulation
Zhao, Yanqun; Qi, Guohai; Yin, Gang; Wang, Xianliang; Wang, Pei; Li, Jian; Xiao, Mingyong; Li, Jie; Kang, Shengwei; Liao, Xiongfei
2014-01-01
Background The accuracy of dose calculation is crucial to the quality of treatment planning and, consequently, to the dose delivered to patients undergoing radiation therapy. Current general calculation algorithms such as Pencil Beam Convolution (PBC) and Collapsed Cone Convolution (CCC) have shortcomings in regard to severe inhomogeneities, particularly in those regions where charged particle equilibrium does not hold. The aim of this study was to evaluate the accuracy of the PBC and CCC alg...
SU-E-T-584: Commissioning of the MC2 Monte Carlo Dose Computation Engine
Energy Technology Data Exchange (ETDEWEB)
Titt, U; Mirkovic, D; Liu, A; Ciangaru, G; Mohan, R [MD Anderson Cancer Center (United States); Anand, A [Mayo Clinic Arizona, Phoenix (United States); Perles, L [Scripps Proton Therapy Center, San Diego, CA (United States)
2014-06-01
Purpose: An automated system, MC2, was developed to convert DICOM proton therapy treatment plans into a sequence MCNPX input files, and submit these to a computing cluster. MC2 converts the results into DICOM format, and any treatment planning system can import the data for comparison vs. conventional dose predictions. This work describes the data and the efforts made to validate the MC2 system against measured dose profiles and how the system was calibrated to predict the correct number of monitor units (MUs) to deliver the prescribed dose. Methods: A set of simulated lateral and longitudinal profiles was compared to data measured for commissioning purposes and during annual quality assurance efforts. Acceptance criteria were relative dose differences smaller than 3% and differences in range (in water) of less than 2 mm. For two out of three double scattering beam lines validation results were already published. Spot checks were performed to assure proper performance. For the small snout, all available measurements were used for validation vs. simulated data. To calibrate the dose per MU, the energy deposition per source proton at the center of the spread out Bragg peaks (SOBPs) was recorded for a set of SOBPs from each option. Subsequently these were then scaled to the results of dose per MU determination based on published methods. The simulations of the doses in the magnetically scanned beam line were also validated vs. measured longitudinal and lateral profiles. The source parameters were fine tuned to achieve maximum agreement with measured data. The dosimetric calibration was performed by scoring energy deposition per proton, and scaling the results to a standard dose measurement of a 10 x 10 x 10 cm3 volume irradiation using 100 MU. Results: All simulated data passed the acceptance criteria. Conclusion: MC2 is fully validated and ready for clinical application.
International Nuclear Information System (INIS)
The objective of this study was to estimate doses in the physician and the nurse assistant at different positions during interventional radiology procedures. In this study, effective doses obtained for the physician and at points occupied by other workers were normalised by air kerma-area product (KAP). The simulations were performed for two X-ray spectra (70 kVp and 87 kVp) using the radiation transport code MCNPX (version 2.7.0), and a pair of anthropomorphic voxel phantoms (MASH/FASH) used to represent both the patient and the medical professional at positions from 7 cm to 47 cm from the patient. The X-ray tube was represented by a point source positioned in the anterior posterior (AP) and posterior anterior (PA) projections. The CC can be useful to calculate effective doses, which in turn are related to stochastic effects. With the knowledge of the values of CCs and KAP measured in an X-ray equipment, at a similar exposure, medical professionals will be able to know their own effective dose. - Highlights: ► This study presents a series of simulations to determine scatter-dose in IR. ► Irradiation of the worker is non-uniform and a part of his body is shielded. ► With the CCs it is possible to estimate the occupational doses in the CA examination. ► Protection of medical personnel in IR is an important issue of radiological protection
International Nuclear Information System (INIS)
International Commission on Radiological Protection (ICRP) reported comprehensive dose conversion coefficients for adult population, which is exposed to external photon sources in the Publication 74. However, those quantities were calculated from so-called stylized (or mathematical) phantoms composed of simplified mathematical surface equations so that the discrepancy between the phantoms and real human anatomy has been investigated by several authors using Caucasian-based voxel phantoms. To address anatomical and racial limitations of the stylized phantoms, several Asian-based voxel phantoms have been developed by Korean and Japanese investigators, independently. In the current study, photon dose conversion coefficients of ICRP 74 were compared with those from a total of five Asian-based male voxel phantoms, whose body dimensions were almost identical. Those of representative radio-sensitive organs (testes, red bone marrow, colon, lungs, and stomach), and effective dose conversion coefficients were obtained for comparison. Even though organ doses for testes, colon and lungs, and effective doses from ICRP 74 agreed well with those from Asian voxel phantoms within 10%, absorbed doses for red bone marrow and stomach showed significant discrepancies up to 30% which was mainly attributed to difference of phantom description between stylized and voxel phantoms. This study showed that the ICRP 74 dosimetry data, which have been reported to be unrealistic compared to those from Caucasian-based voxel phantoms, are also not appropriate for Asian population
Energy Technology Data Exchange (ETDEWEB)
David, Mariano G.; Pires, Evandro J.; Magalhaes, Luis A.; Almeida, Carlos E. de; Alves, Carlos F.E., E-mail: marianogd08@gmail.com [Universidade do Estado do Rio de Janeiro (UERJ), RJ (Brazil). Lab. Ciencias Radiologicas; Albuquerque, Marcos A. [Coordenacao dos Programas de Pos-Graduacao de Engenharia (COPPE/UFRJ), RJ (Brazil). Instituto Alberto Luiz Coimbra; Bernal, Mario A. [Universidade Estadual de Campinas (UNICAMP), SP (Brazil). Instituto de Fisica Gleb Wataghin; Peixoto, Jose G. [Instituto de Radioprotecao e Dosimetria (IRD/CNEN-RJ), Rio de Janeiro, RJ (Brazil)
2012-08-15
This paper focuses on the obtainment, using experimental and Monte Carlo-simulated (MMC) methods, of the photon spectra at various depths and depth-dose deposition curves for x-rays beams used in mammography, obtained on a polymethylmethacrylate (PMMA) breast phantom. Spectra were obtained for 28 and 30 kV quality-beams and the corresponding average energy values (Emed) were calculated. For the experimental acquisition was used a Si-PIN photodiode spectrometer and for the MMC simulations the PENELOPE code was employed. The simulated and the experimental spectra show a very good agreement, which was corroborated by the low differences found between the Emed values. An increase in the Emed values and a strong attenuation of the beam through the depth of the PMMA phantom was also observed. (author)
International Nuclear Information System (INIS)
The aim of this work is to investigate the influence of the statistical fluctuations of Monte Carlo (MC) dose distributions on the dose volume histograms (DVHs) and radiobiological models, in particular the Poisson model for tumour control probability (tcp). The MC matrix is characterized by a mean dose in each scoring voxel, d, and a statistical error on the mean dose, σd; whilst the quantities d and σd depend on many statistical and physical parameters, here we consider only their dependence on the phantom voxel size and the number of histories from the radiation source. Dose distributions from high-energy photon beams have been analysed. It has been found that the DVH broadens when increasing the statistical noise of the dose distribution, and the tcp calculation systematically underestimates the real tumour control value, defined here as the value of tumour control when the statistical error of the dose distribution tends to zero. When increasing the number of energy deposition events, either by increasing the voxel dimensions or increasing the number of histories from the source, the DVH broadening decreases and tcp converges to the 'correct' value. It is shown that the underestimation of the tcp due to the noise in the dose distribution depends on the degree of heterogeneity of the radiobiological parameters over the population; in particular this error decreases with increasing the biological heterogeneity, whereas it becomes significant in the hypothesis of a radiosensitivity assay for single patients, or for subgroups of patients. It has been found, for example, that when the voxel dimension is changed from a cube with sides of 0.5 cm to a cube with sides of 0.25 cm (with a fixed number of histories of 108 from the source), the systematic error in the tcp calculation is about 75% in the homogeneous hypothesis, and it decreases to a minimum value of about 15% in a case of high radiobiological heterogeneity. The possibility of using the error on the tcp to
International Nuclear Information System (INIS)
Purpose: Due to its capability to more accurately detect deep lesions inside the breast by removing the effect of overlying anatomy, digital breast tomosynthesis (DBT) has the potential to replace the standard mammography technique in clinical screening exams. However, the European Guidelines for DBT dosimetry are still a work in progress and there are little data available on organ doses other than to the breast. It is, therefore, of great importance to assess the dosimetric performance of DBT with respect to the one obtained with standard digital mammography (DM) systems. The aim of this work is twofold: (i) to study the dosimetric properties of a combined DBT/DM system (MAMMOMAT Inspiration Siemens®) for a tungsten/rhodium (W/Rh) anode/filter combination and (ii) to evaluate organs doses during a DBT examination. Methods: For the first task, measurements were performed in manual and automatic exposure control (AEC) modes, using two homogeneous breast phantoms: a PMMA slab phantom and a 4 cm thick breast-shaped rigid phantom, with 50% of glandular tissue in its composition. Monte Carlo (MC) simulations were performed using Monte Carlo N-Particle eXtended v.2.7.0. A MC model was implemented to mimic DM and DBT acquisitions for a wide range of x-ray spectra (24 –34 kV). This was used to calculate mean glandular dose (MGD) and to compute series of backscatter factors (BSFs) that could be inserted into the DBT dosimetric formalism proposed by Dance et al. Regarding the second aim of the study, the implemented MC model of the clinical equipment, together with a female voxel phantom (“Laura”), was used to calculate organ doses considering a typical DBT acquisition. Results were compared with a standard two-view mammography craniocaudal (CC) acquisition. Results: Considering the AEC mode, the acquisition of a single CC view results in a MGD ranging from 0.53 ± 0.07 mGy to 2.41 ± 0.31 mGy in DM mode and from 0.77 ± 0.11 mGy to 2.28 ± 0.32 mGy in DBT mode
Energy Technology Data Exchange (ETDEWEB)
Baptista, Mariana, E-mail: marianabaptista@ctn.ist.utl.pt; Di Maria, Salvatore; Barros, Sílvia; Vaz, Pedro [Centro de Ciências e Tecnologias Nucleares, Instituto Superior Técnico, Universidade de Lisboa, Estrada Nacional 10, km 139,7, Bobadela LRS 2695-066 (Portugal); Figueira, Catarina [Centre for Plasma Physics, School of Mathematics and Physics, Queen’s University, Belfast BT7 1NN (United Kingdom); Sarmento, Marta; Orvalho, Lurdes [Serviço de Imagiologia, Hospital da Luz, Avenida Lusíada, 100, Lisboa 1500-650 (Portugal)
2015-07-15
Purpose: Due to its capability to more accurately detect deep lesions inside the breast by removing the effect of overlying anatomy, digital breast tomosynthesis (DBT) has the potential to replace the standard mammography technique in clinical screening exams. However, the European Guidelines for DBT dosimetry are still a work in progress and there are little data available on organ doses other than to the breast. It is, therefore, of great importance to assess the dosimetric performance of DBT with respect to the one obtained with standard digital mammography (DM) systems. The aim of this work is twofold: (i) to study the dosimetric properties of a combined DBT/DM system (MAMMOMAT Inspiration Siemens{sup ®}) for a tungsten/rhodium (W/Rh) anode/filter combination and (ii) to evaluate organs doses during a DBT examination. Methods: For the first task, measurements were performed in manual and automatic exposure control (AEC) modes, using two homogeneous breast phantoms: a PMMA slab phantom and a 4 cm thick breast-shaped rigid phantom, with 50% of glandular tissue in its composition. Monte Carlo (MC) simulations were performed using Monte Carlo N-Particle eXtended v.2.7.0. A MC model was implemented to mimic DM and DBT acquisitions for a wide range of x-ray spectra (24 –34 kV). This was used to calculate mean glandular dose (MGD) and to compute series of backscatter factors (BSFs) that could be inserted into the DBT dosimetric formalism proposed by Dance et al. Regarding the second aim of the study, the implemented MC model of the clinical equipment, together with a female voxel phantom (“Laura”), was used to calculate organ doses considering a typical DBT acquisition. Results were compared with a standard two-view mammography craniocaudal (CC) acquisition. Results: Considering the AEC mode, the acquisition of a single CC view results in a MGD ranging from 0.53 ± 0.07 mGy to 2.41 ± 0.31 mGy in DM mode and from 0.77 ± 0.11 mGy to 2.28 ± 0.32 mGy in DBT mode
A Monte Carlo evaluation of RapidArc dose calculations for oropharynx radiotherapy.
Gagne, I M; Ansbacher, W; Zavgorodni, S; Popescu, C; Beckham, W A
2008-12-21
RapidArc, recently released by Varian Medical Systems, is a novel extension of IMRT in which an optimized 3D dose distribution may be delivered in a single gantry rotation of 360 degrees or less. The purpose of this study was to investigate the accuracy of the analytical anisotropic algorithm (AAA), the sole algorithm for photon dose calculations of RapidArc treatment plans. The clinical site chosen was oropharynx and the associated nodes involved. The VIMC-Arc system, which utilizes BEAMnrc and DOSXYZnrc for particle transport through the linac head and patient CT phantom, was used as a benchmarking tool. As part of this study, the dose for a single static aperture, typical for RapidArc delivery, was calculated by the AAA, MC and compared with the film. This film measurement confirmed MC modeling of the beam aperture in water. It also demonstrated that the AAA dosimetric error can be as high as 12% near isolated leaf edges and up to 5% at the leaf end. The composite effect of these errors in a full RapidArc calculation in water involving a C-shaped target and the associated organ at risk produced a 1.5% overprediction of the mean target dose. In our cohort of six patients, the AAA was found, on average, to overestimate the PTV60 coverage at the 95% level in the presence of air cavities by 1.0% (SD = 1.1%). Removing the air cavities from the target volumes reduced these differences by about a factor of 2. The dose to critical structures was also overestimated by the AAA. The mean dose to the spinal cord was higher by 1.8% (SD = 0.8%), while the effective maximum dose (D2%) was only 0.2% higher (SD = 0.6%). The mean dose to the parotid glands was overestimated by approximately 9%. This study has shown that the accuracy of the AAA for RapidArc dose calculations, performed at a resolution of 2.5 mm or better, is adequate for clinical use. PMID:19033640
Baptista, M.; Teles, P.; Cardoso, G.; Vaz, P.
2014-11-01
Over the last decade, there was a substantial increase in the number of interventional cardiology procedures worldwide, and the corresponding ionizing radiation doses for both the medical staff and patients became a subject of concern. Interventional procedures in cardiology are normally very complex, resulting in long exposure times. Also, these interventions require the operator to work near the patient and, consequently, close to the primary X-ray beam. Moreover, due to the scattered radiation from the patient and the equipment, the medical staff is also exposed to a non-uniform radiation field that can lead to a significant exposure of sensitive body organs and tissues, such as the eye lens, the thyroid and the extremities. In order to better understand the spatial variation of the dose and dose rate distributions during an interventional cardiology procedure, the dose distribution around a C-arm fluoroscopic system, in operation in a cardiac cath lab at Portuguese Hospital, was estimated using both Monte Carlo (MC) simulations and dosimetric measurements. To model and simulate the cardiac cath lab, including the fluoroscopic equipment used to execute interventional procedures, the state-of-the-art MC radiation transport code MCNPX 2.7.0 was used. Subsequently, Thermo-Luminescent Detector (TLD) measurements were performed, in order to validate and support the simulation results obtained for the cath lab model. The preliminary results presented in this study reveal that the cardiac cath lab model was successfully validated, taking into account the good agreement between MC calculations and TLD measurements. The simulated results for the isodose curves related to the C-arm fluoroscopic system are also consistent with the dosimetric information provided by the equipment manufacturer (Siemens). The adequacy of the implemented computational model used to simulate complex procedures and map dose distributions around the operator and the medical staff is discussed, in
Monte Carlo simulation and X-ray dose measurement of 9 MeV RF electron LINAC for cargo scanning
International Nuclear Information System (INIS)
A 9 MeV RF Electron Linear Accelerator is operational at ECIL, Hyderabad to demonstrate cargo scanning applications to prevent the transport of contraband objects. The interrogation is done by a fanned X-ray beam produced by impinging an energetic electron beam on tantalum (Ta) target. The desired dose rate for this purpose is 24Gy/min at 1 m from the X-ray source. It is required to determine the X-ray dose rate at one meter distance from the X-ray converter for a given set of beam parameters. Monte Carlo simulations have been done by using Geant4 software toolkit. X-ray dose rate values at a meter distance from X-ray target at various polar angles with respect to the direction of the incident beam. The plot shows angular distribution of significant X-ray intensity in a cone of 15 deg half angle. The maximum X-ray emission is in the forward direction of the beam and the dose-rate is estimated to be 23.4 Gy/min when the beam parameters are i) 9 MeV energy ii) 60 mA peak beam current and iii) 250 Hz pulse repetition frequency (PRF). Experiments have also been carried out using air ionization chamber to obtain X-ray dose-rate at a meter distance from target. The measured value is 24 Gy/min at 1m distance from the target and it agrees well with our estimation. (author)
MAGIK: a Monte Carlo system for computing induced residual activation dose rates
International Nuclear Information System (INIS)
The photon dose rate from the induced activity produced by sustained bombardment of materials by neutrons and charged particles may present a significant radiation hazard. To minimize this hazard, the material configuration must be so designed that the photon dose rate decays to an acceptable level soon after the source beam is turned off. MAGIK calculates the time-independent photon dose rates that result from activities produced by nucleon-nucleus and meson-nucleus collisions over a wide range of energies. The system has been used both for high-energy accelerator studies and for fusion reactor studies. In the MAGIK system the lengthy photon transport calculations are carried out independent of time, and the time dependence is introduced in the final program, thereby permitting study of various operating scenarios with a minimum computing cost
International Nuclear Information System (INIS)
Mammography is a standard procedure that facilitates breast cancer detection. Initial results of contrast-enhanced digital mammography (CEDM) are promising. The purpose of this study is to assess the CEDM radiation dose using a Monte Carlo code. EGSnrc MC code was used to simulate the interaction of photons with matter and estimate the glandular dose (Dg). A voxel female human phantom with a 2-8-cm breast thickness range and a breast glandular composition of 50 % was applied. Dg values ranged between 0.96 and 1.45 mGy (low and high energy). Dg values for a breast thickness of 5.0 cm and a glandular fraction of 50 % for craniocaudal and mediolateral oblique view were 1.12 (low energy image contribution is 0.98 mGy) and 1.07 (low energy image contribution is 0.95 mGy), respectively. The low kV part of CEDM is the main contributor to total glandular breast dose. (authors)
Yuan, Jiankui; Lo, Simon S; Zheng, Yiran; Sohn, Jason W; Sloan, Andrew E; Ellis, Rodney; Machtay, Mitchell; Wessels, Barry
2016-01-01
Detailed Monte Carlo (MC) modeling of the Leksell Gamma Knife (GK) Perfexion (PFX) collimator system is the only accurate ab initio approach appearing in the literature. As a different approach, in this work, we present a MC model based on film measurement. By adjusting the model parameters and fine-tuning the derived fluence map for each individual source to match the manufacturer's ring output factors, we created a reasonable virtual source model for MC simulations to verify treatment planning dose for the GK PFX radiosurgery system. The MC simulation model was commissioned by simple single shots. Dose profiles and both ring and collimator output factors were compared with the treatment planning system (TPS). Good agreement was achieved for dose profiles especially for the region of plateau (< 2%), while larger difference (< 5%) came from the penumbra region. The maximum difference of the calculated output factor was within 0.7%. The model was further validated by a clinical test case. Good agreement was obtained. The DVHs for brainstem and the skull were almost identical and, for the target, the volume covered by the prescription (12.5 Gy to 50% isodose line) was 95.6% from MC calculation versus 100% from the TPS. PMID:27455497
International Nuclear Information System (INIS)
Currently ICRP (International Commission on Radiological Protection) develops a new recommendation to estimate the natural radiation exposure of an agreed set of animals and reference plants. For estimating effective dose in humans and animals, the incorporated activities of natural and artificial radionuclides in body tissues and contents of the digestive system have to be known. It was the aim of this investigation to calculate energy doses caused by Cs 137 and K 40 in the reproductive organs (uterus, ovaries) of a cow. During its whole lifetime from 1986 to 1992, the cow incorporated continuously Cs 137 which was due to the fallout following the Chernobyl accident. K 40 occurs naturally in the cow's fodder. The cow was born in a highly contaminated region of Styria, Austria, and was infertile since 1990. The activities of Cs 137 and K 40 in the cow's fodder and in tissues, organs and contents of the digestive system of the carcass were measured simultaneously with the help of semiconductor detectors. To calculate the specific absorbed fractions by means of the Monte Carlo code MCNP, an appropriate simulation model for the reproductive organs and their surrounding tissues was developed. The contents of rectum and urinary bladder account for the main part of the energy dose in the reproductive organs. Comparison of our results with data from other investigations showed, that lifetime accumulation of Cs 137 and K 40 was too low to cause radiation inferred infertility. (orig.)
International Nuclear Information System (INIS)
Purpose: The purpose of this work was to evaluate the absorbed dose to Al2O3 dosimeter at various depths of water phantom in radiotherapy photon beams by Monte Carlo simulation and evaluate the beam quality dependence. Methods: The simulations were done using EGSnrc. The cylindrical Al2O3 dosimeter (Φ4 mmx1 mm) was placed at the central axis of the water phantom (Φ16 cmx16 cm) at depths between 0.5 and 8 cm. The incident beams included monoenergetic photon beams ranging from 1 to 18 MeV, 60Co γ beams, Varian 6 MV beams using phase space files based on a full simulation of the linac, and Varian beams between 4 and 24 MV using Mohan's spectra. The absorbed dose to the dosimeter and the water at the corresponding position in the absence of the dosimeter, as well as absorbed dose ratio factor fmd, was calculated. Results: The results show that fmd depends obviously on the photon energy at the shallow depths. However, as the depth increases, the change in fmd becomes small, beyond the buildup region, the maximum discrepancy of fmd to the average value is not more than 1%. Conclusions: These simulation results confirm the use of Al2O3 dosimeter in radiotherapy photon beams and clearly indicate that more attention should be paid when using such a dosimeter in the buildup region of high-energy radiotherapy photon beams.
International Nuclear Information System (INIS)
In the present study, a number of brachytherapy sources and activation media were simulated using MCNPX code and the results were analyzed based on the dose enhancement factor values. Furthermore, two new brachytherapy sources (131Cs and a hypothetical 170Tm) were evaluated for their application in photon activation therapy (PAT). 125I, 103Pd, 131Cs and hypothetical 170Tm brachytherapy sources were simulated in water and their dose rate constant and the radial dose functions were compared with previously published data. The sources were then simulated in a soft tissue phantom which was composed of Ag, I, Pt or Au as activation media uniformly distributed in the tumour volume. These simulations were performed using the MCNPX code, and dose enhancement factor (DEF) was obtained for 7, 18 and 30 mg/ml concentrations of the activation media. Each source, activation medium and concentration was evaluated separately in a separate simulation. The calculated dose rate constant and radial dose functions were in agreement with the published data for the aforementioned sources. The maximum DEF was found to be 5.58 for a combination of the 170Tm source with 30 mg/ml concentration of I. The DEFs for 131Cs and 170Tm sources for all the four activation media were higher than those for other sources and activation media. From this point of view, these two sources can be more useful in photon activation therapy with photon emitter sources. Furthermore, 131Cs and 170Tm brachytherapy sources can be proposed as new options for use in the field of PAT.
International Nuclear Information System (INIS)
With the purpose of determining the radiation field in the accelerator environment in an efficient way, it was developed an approximate methodology which uses multigroup albedo coefficients to describe the particle reflection by the walls. This method avoids the particle transport calculation inside the walls, which spends much of the processing time. The Monte Carlo code MCNP was suitably modified to allow the simulation of such calculations. To assess the accuracy achieved with this methodology, very realistic calculations considering the transport of particles inside all the walls are performed. The results showed that the use of albedo coefficients for some walls while allowing the transport of particles inside the other walls in the same calculation gives accurate results, saving significant computational time. The results obtained for the accelerator showed an excellent agreement with the realistic calculation, and that the technique is applicable to large environments. (author)
Monte Carlo simulation of dose distribution in tubing using the integrated TIGER series
International Nuclear Information System (INIS)
One of the difficulties in the irradiation of nonslab geometry, such as a tube, is the uneven penetration of electrons. Distribution of dose in a tube or cable in relationship to voltage, composition, wall thickness, and diameter vary greatly but can be mapped using the ITS/ACCEPT code, complete with automatic subzoning. Other elements of three-dimensional geometry that affect dose distribution including effects of window foil, backscattering fixtures, and beam-scanning angles can also be accounted for by using ITS/ACCEPT without the need to run time-consuming and costly factory experiments
SU-E-T-386: A Monte Carlo Dose Calculation Framework for Electron Beams On Varian TrueBeam
International Nuclear Information System (INIS)
Purpose: The design of the linac head is different for TrueBeam than Clinac, and there are differences in measured dose distributions in water phantoms between TrueBeam and Clinac for electron beams. Therefore, MC models for Clinac may not be applied directly to the Truebeam linac. The purpose of this study is to validate a Monte Carlo (MC) dose calculation framework for electron beams on Varian TrueBeam with phase space files provided by Varian. Methods: The particle histories from the phase space file were used as input for the down-stream simulation including jaws, applicators, and water phantom. MC packages BEAMnrc/DOSYXZnrc were used. The down-stream beam components were modeled according to manufacturer specifications and the dose distributions were compared with the measured data of standard cones. The measurements were performed in a water phantom with a p-type electron field diode (diameter 0.2cm) and ion chamber (CC13). Depth dose and orthogonal profiles at depths defined by R1 0 0, R5 0, Rp were compared. Results: Preliminary results for a 16 MeV phase space and 10x10, 15x15, and 20x20 cm2 applicator are presented. Simulations were run for a statistical uncertainty of <2% at depth of maximum dose for a voxel resolution of 0.5x0.5x0.2cm2. Dose and range differences for the PDD profiles were within 2% and 1 mm, respectively. Dose differences within the central 80% of the beam width for the orthogonal profiles at depth of maximum dose were less than 2% for the 10x10, 15x15, and 20x20 cm2 applicator, respectively. Conclusion: Varian electron phase space files simulations are in agreement with measured commissioning data. These phase space files can be used in the simulation of TrueBeam linacs, and will provide reproducibility across publications. Analyses for all electron energies and standard applicators are under way and results will be included in the presentation
Santos, W. S.; Carvalho, A. B., Jr.; Hunt, J. G.; Maia, A. F.
2014-02-01
The objective of this study was to estimate doses in the physician and the nurse assistant at different positions during interventional radiology procedures. In this study, effective doses obtained for the physician and at points occupied by other workers were normalised by air kerma-area product (KAP). The simulations were performed for two X-ray spectra (70 kVp and 87 kVp) using the radiation transport code MCNPX (version 2.7.0), and a pair of anthropomorphic voxel phantoms (MASH/FASH) used to represent both the patient and the medical professional at positions from 7 cm to 47 cm from the patient. The X-ray tube was represented by a point source positioned in the anterior posterior (AP) and posterior anterior (PA) projections. The CC can be useful to calculate effective doses, which in turn are related to stochastic effects. With the knowledge of the values of CCs and KAP measured in an X-ray equipment, at a similar exposure, medical professionals will be able to know their own effective dose.
Energy Technology Data Exchange (ETDEWEB)
Paixao, L.; Oliveira, B. B.; Nogueira, M. do S. [Centro de Desenvolvimento da Tecnologia Nuclear, Post-graduation in Science and Technology of Radiations, Minerals and Materials, Pte. Antonio Carlos 6.627, Pampulha, 31270-901 Belo Horizonte (Brazil); Viloria, C. [UFMG, Departamento de Engenharia Nuclear, Post-graduation in Nuclear Sciences and Techniques, Pte. Antonio Carlos 6.627, Pampulha, 31270-901 Belo Horizonte (Brazil); Alves de O, M. [UFMG, Department of Anatomy and Imaging, Prof. Alfredo Balena 190, 30130-100 Belo Horizonte (Brazil); Araujo T, M. H., E-mail: lpr@cdtn.br [Dr Maria Helena Araujo Teixeira Clinic, Guajajaras 40, 30180-100 Belo Horizonte (Brazil)
2014-08-15
It is widely accepted that the mean glandular dose (D{sub G}) for the glandular tissue is the more useful magnitude for characterizing the breast cancer risk. The procedure to estimate the D{sub G}, for being difficult to measure it directly in the breast, it is to make the use of conversion factors that relate incident air kerma (K{sub i}) at this dose. Generally, the conversion factors vary with the x-ray spectrum half-value layer and the breast composition and thickness. Several authors through computer simulations have calculated such factors by the Monte Carlo (Mc) method. Many spectral models for D{sub G} computer simulations purposes are available in the diagnostic range. One of the models available generates unfiltered spectra. In this work, the Monte Carlo EGSnrc code package with the C++ class library (eg spp) was employed to derive filtered tungsten x-ray spectra used in digital mammography systems. Filtered spectra for rhodium and aluminium filters were obtained for tube potentials between 26 and 32 kV. The half-value layer of simulated filtered spectra were compared with those obtained experimentally with a solid state detector Unfors model 8202031-H Xi R/F and Mam Detector Platinum and 8201023-C Xi Base unit Platinum Plus w m As in a Hologic Selenia Dimensions system using a Direct Radiography mode. Calculated half-value layer values showed good agreement compared to those obtained experimentally. These results show that the filtered tungsten anode x-ray spectra and the EGSnrc Mc code can be used for D{sub G} determination in mammography. (Author)
International Nuclear Information System (INIS)
It is widely accepted that the mean glandular dose (DG) for the glandular tissue is the more useful magnitude for characterizing the breast cancer risk. The procedure to estimate the DG, for being difficult to measure it directly in the breast, it is to make the use of conversion factors that relate incident air kerma (Ki) at this dose. Generally, the conversion factors vary with the x-ray spectrum half-value layer and the breast composition and thickness. Several authors through computer simulations have calculated such factors by the Monte Carlo (Mc) method. Many spectral models for DG computer simulations purposes are available in the diagnostic range. One of the models available generates unfiltered spectra. In this work, the Monte Carlo EGSnrc code package with the C++ class library (eg spp) was employed to derive filtered tungsten x-ray spectra used in digital mammography systems. Filtered spectra for rhodium and aluminium filters were obtained for tube potentials between 26 and 32 kV. The half-value layer of simulated filtered spectra were compared with those obtained experimentally with a solid state detector Unfors model 8202031-H Xi R/F and Mam Detector Platinum and 8201023-C Xi Base unit Platinum Plus w m As in a Hologic Selenia Dimensions system using a Direct Radiography mode. Calculated half-value layer values showed good agreement compared to those obtained experimentally. These results show that the filtered tungsten anode x-ray spectra and the EGSnrc Mc code can be used for DG determination in mammography. (Author)
An estimation of the percentage of dose in intraoral radiology exams using Monte Carlo simulation
International Nuclear Information System (INIS)
In this work we used the EGS4 code in a simulated study of dose percentage in intraoral examination to 10 energy range to 140 keV. The simulation was carried out on a model consisting of different geometry (cheek, tooth and mouth cavity) under normal incidence X-ray beam over the surface of the various simulated materials. It was observed that for energy smaller than 30 keV most of the energy is deposited on the cheek. In 30 keV there is a point of maximum radiation absorption in the tooth (approximately 60% of the energy of the incident radiation is deposited on the tooth) in relation to other simulated materials. It means that in this energy there is a better contrast in the radiographic image of the tooth and a smaller dose on the cheek. In 40 keV the deposited energy in the tooth is roughly equal to the energy that is transmitted (to the radiographic film or buccal cavity) causing a degradation in the radiographic image and/or a higher dose in the oral cavity. For energies above 40 keV, the amount of energy transmitted (to the oral cavity and/or radiographic film) is higher than the energy deposited in other materials, i.e, it only contributes to increasing of dose in the regions close to the oral cavity and the radiographic image degradation. These results can provide important information for radiological procedures applied in dentistry where the image quality is a relevant factor to a dental evaluation needs as well as reducing dose in the oral cavity.
International Nuclear Information System (INIS)
Purpose: Various cancers of the eye are successfully treated with radiotherapy utilizing one anterior-posterior (A/P) beam that encompasses the entire content of the orbit. In such cases, a hanging lens shield can be used to spare dose to the radiosensitive lens of the eye to prevent cataracts. Methods: This research focused on Monte Carlo characterization of dose distributions resulting from a single A-P field to the orbit with a hanging shield in place. Monte Carlo codes were developed which calculated dose distributions for various electron radiation energies, hanging lens shield radii, shield heights above the eye, and beam spoiler configurations. Film dosimetry was used to benchmark the coding to ensure it was calculating relative dose accurately. Results: The Monte Carlo dose calculations indicated that lateral and depth dose profiles are insensitive to changes in shield height and electron beam energy. Dose deposition was sensitive to shield radius and beam spoiler composition and height above the eye. Conclusion: The use of a single A/P electron beam to treat cancers of the eye while maintaining adequate lens sparing is feasible. Shield radius should be customized to have the same radius as the patient’s lens. A beam spoiler should be used if it is desired to substantially dose the eye tissues lying posterior to the lens in the shadow of the lens shield. The compromise between lens sparing and dose to diseased tissues surrounding the lens can be modulated by varying the beam spoiler thickness, spoiler material composition, and spoiler height above the eye. The sparing ratio is a metric that can be used to evaluate the compromise between lens sparing and dose to surrounding tissues. The higher the ratio, the more dose received by the tissues immediately posterior to the lens relative to the dose received by the lens
Absorbed dose estimations of 131I for critical organs using the GEANT4 Monte Carlo simulation code
Institute of Scientific and Technical Information of China (English)
Ziaur Rahman; Shakeel ur Rehman; Waheed Arshed; Nasir M Mirza; Abdul Rashid; Jahan Zeb
2012-01-01
The aim of this study is to compare the absorbed doses of critical organs of 131I using the MIRD (Medical Internal Radiation Dose) with the corresponding predictions made by GEANT4 simulations.S-values (mean absorbed dose rate per unit activity) and energy deposition per decay for critical organs of 131I for various ages,using standard cylindrical phantom comprising water and ICRP soft-tissue material,have also been estimated.In this study the effect of volume reduction of thyroid,during radiation therapy,on the calculation of absorbed dose is also being estimated using GEANT4.Photon specific energy deposition in the other organs of the neck,due to 131I decay in the thyroid organ,has also been estimated.The maximum relative difference of MIRD with the GEANT4 simulated results is 5.64％ for an adult's critical organs of 131I.Excellent agreement was found between the results of water and ICRP soft tissue using the cylindrical model.S-values are tabulated for critical organs of 131I,using 1,5,10,15 and 18 years (adults) individuals.S-values for a cylindrical thyroid of different sizes,having 3.07％ relative differences of GEANT4 with Siegel & Stabin results.Comparison of the experimentally measured values at 0.5 and 1 m away from neck of the ionization chamber with GEANT4 based Monte Carlo simulations results show good agreement.This study shows that GEANT4 code is an important tool for the internal dosimetry calculations.
Monte Carlo calculations of the depth-dose distribution in skin contaminated by hot particles
Energy Technology Data Exchange (ETDEWEB)
Patau, J.-P. (Toulouse-3 Univ., 31 (France))
1991-01-01
Accurate computer programs were developed in order to calculate the spatial distribution of absorbed radiation doses in the skin, near high activity particles (''hot particles''). With a view to ascertaining the reliability of the codes the transport of beta particles was simulated in a complex configuration used for dosimetric measurements: spherical {sup 60}Co sources of 10-1000 {mu}m fastened to an aluminium support with a tissue-equivalent adhesive overlaid with 10 {mu}m thick aluminium foil. Behind it an infinite polystyrene medium including an extrapolation chamber was assumed. The exact energy spectrum of beta emission was sampled. Production and transport of secondary knock-on electrons were also simulated. Energy depositions in polystyrene were calculated with a high spatial resolution. Finally, depth-dose distributions were calculated for hot particles placed on the skin. The calculations will be continued for other radionuclides and for a configuration suited to TLD measurements. (author).
Environmental dose rate assessment of ITER using the Monte Carlo method
Karimian Alireza; Beheshti Amir; Abdi Mohammadreza; Jabbari Iraj
2014-01-01
Exposure to radiation is one of the main sources of risk to staff employed in reactor facilities. The staff of a tokamak is exposed to a wide range of neutrons and photons around the tokamak hall. The International Thermonuclear Experimental Reactor (ITER) is a nuclear fusion engineering project and the most advanced experimental tokamak in the world. From the radiobiological point of view, ITER dose rates assessment is particularly important. The aim of th...
Energy Technology Data Exchange (ETDEWEB)
Silva, Aneli Oliveira da
2010-07-01
This study aims to compare the doses received for patients submitted to brachytherapy High Dose Rate (HDR) brachytherapy, a method of treatment of the cervix carcinoma, performed in the planning system PLATO BPS with the doses obtained by Monte Carlo simulation using the radiation transport code MCNP 5 and one female anthropomorphic phantom based on voxel, the FAX. The implementation of HDR brachytherapy treatment for the cervix carcinoma consists of the insertion of an intrauterine probe and an intravaginal probe (ring or ovoid) and then two radiographs are obtained, anteroposterior (AP) and lateral (LAT) to confirm the position of the applicators in the patient and to allow the treatment planning and the determination of the absorbed dose at points of interest: rectum, bladder, sigmoid and point A, which corresponds anatomically to the crossings of the uterine arteries with ureters The absorbed doses obtained with the code MCNP 5, with the exception of the absorbed dose in the rectum and sigmoid for the simulation considering a point source of {sup 192}Ir, are lower than the absorbed doses from PLATO BPS calculations because the MCNP 5 considers the chemical compositions and densities of FAX body, not considering the medium as water. When considering the Monte Carlo simulation for a source with dimensions equal to that used in the brachytherapy irradiator used in this study, the values of calculated absorbed dose to the bladder, to the rectum, to the right point A and to the left point A were respectively lower than those determined by the treatment planning system in 33.29, 5.01, 22.93 and 19.04%. These values are almost all larger than the maximum acceptable deviation between patient planned and administered doses (5 %). With regard to the rectum and bladder, which are organs that must be protected, the present results are in favor of the radiological protection of patients. The point A, that is on the isodose of 100%, used to tumor treatment, the results
Adjustment of a Korean Voxel Phantom and Its Effect on Monte Carlo Dose Calculations
International Nuclear Information System (INIS)
Many different voxel models have been developed for use in the calculation of the radiation doses to organs and tissues in the human body. Most of these voxel models are constructed from the CT and MR images of individuals. The organs and tissues are, however, not always discernable on CT and MR images, making it difficult to segment some organs and tissues. Furthermore, these models do not represent the average radiation workers in Korea because the height, weight, and organ masses are different from those of the average Korean workers. This paper compares two Korean voxel models (VKH-Man, HDRK-Man), which have been constructed from the same data set of serially-sectioned color images, in terms of calculated organs doses. The HDRK-Man has been developed by adjusting the VKH Man to the data of the 'Reference Korean'. In this study, organs doses were calculated with these developed models and the calculated values were compared each other to see the effect of the adjustment
Adjustment of a Korean Voxel Phantom and Its Effect on Monte Carlo Dose Calculations
Energy Technology Data Exchange (ETDEWEB)
Jeong, Jong Hwi; Choi, Sang Hyoun; Cho, Sung Koo [Hanyang University, Seoul (Korea, Republic of); Lee, Choon Sik [University of Florida, Gainesville (United States); Chung, Min Suk; Kim, Chan Hyeong [Ajou University School of Medicine, Suwon (Korea, Republic of)
2007-07-01
Many different voxel models have been developed for use in the calculation of the radiation doses to organs and tissues in the human body. Most of these voxel models are constructed from the CT and MR images of individuals. The organs and tissues are, however, not always discernable on CT and MR images, making it difficult to segment some organs and tissues. Furthermore, these models do not represent the average radiation workers in Korea because the height, weight, and organ masses are different from those of the average Korean workers. This paper compares two Korean voxel models (VKH-Man, HDRK-Man), which have been constructed from the same data set of serially-sectioned color images, in terms of calculated organs doses. The HDRK-Man has been developed by adjusting the VKH Man to the data of the 'Reference Korean'. In this study, organs doses were calculated with these developed models and the calculated values were compared each other to see the effect of the adjustment.
International Nuclear Information System (INIS)
An electron-photon coupled Monte Carlo code ARCHER - Accelerated Radiation-transport Computations in Heterogeneous EnviRonments - is being developed at Rensselaer Polytechnic Institute as a software test-bed for emerging heterogeneous high performance computers that utilize accelerators such as GPUs (Graphics Processing Units). This paper presents the preliminary code development and the testing involving radiation dose related problems. In particular, the paper discusses the electron transport simulations using the class-II condensed history method. The considered electron energy ranges from a few hundreds of keV to 30 MeV. As for photon part, photoelectric effect, Compton scattering and pair production were simulated. Voxelized geometry was supported. A serial CPU (Central Processing Unit)code was first written in C++. The code was then transplanted to the GPU using the CUDA C 5.0 standards. The hardware involved a desktop PC with an Intel Xeon X5660 CPU and six NVIDIA Tesla M2090 GPUs. The code was tested for a case of 20 MeV electron beam incident perpendicularly on a water-aluminum-water phantom. The depth and later dose profiles were found to agree with results obtained from well tested MC codes. Using six GPU cards, 6*106 electron histories were simulated within 2 seconds. In comparison, the same case running the EGSnrc and MCNPX codes required 1645 seconds and 9213 seconds, respectively. On-going work continues to test the code for different medical applications such as radiotherapy and brachytherapy. (authors)
Electron absorbed dose comparison between MCNP5 and Penelope Monte Carlo code for microdosimetry
International Nuclear Information System (INIS)
The objective of the present work was to compare electron absorbed dose results between two widespread used codes in international scientific community: MCNP5 and Penelope-2003. Individual water spheres with masses between 10-9 g up to 10-3 g immersed in an infinite water medium (density of 1g/cm3) and monoenergetic electron sources with energy from 0.002 MeV to 0.1 MeV have been considered. The absorbed dose in the spheres was evaluated by both codes and the relative differences have been quantified. The results shown that Penelope gives, in general, higher results that, in some cases saturate or reach a maximum point and then rapidly drops. Particularly, for the 40 keV electron source we have done additional tests in three different scenarios: more points in the region of lower masses to a better definition of the curve behavior; MCNP used 200 substeps and Penelope was set to a full detail history methodology, and almost same parameters of case B but with the density of exterior medium increased to 10 g/cm3. The three cases show the influence of the backscattering that contribute with an important fraction of absorbed dose, finally we can infer a range of reliability to use the codes in this kind of simulations: both codes can calculate close results for up to 10-4 g.Even though MCNP5 uses the condensed history method, if simulation parameters are chosen carefully it can reproduce results very close to those obtained using detailed history mode. In some cases, the use of higher number of electron substeps causes significant differences in the result. (author)
Monte Carlo Simulation Of Absorbed Dose From LINAC On VOXEL Phantom By Using MCNP5 Code
International Nuclear Information System (INIS)
In this work, we use MCNP5 code for simulating dose distribution calculation from LINAC on phantom CT. CT images obtained from cancer treatment cases at Cho Ray hospital. In order to transform CT images into data of MCNP5 input file we also build a program CODIM by using MATLAB programming software. The results show that there is a difference of 5% in comparison to DSS program - a semi-empirical simulation program which is being used for treatment planning in Cho Ray hospital. (author)
A voxel-based mouse for internal dose calculations using Monte Carlo simulations (MCNP)
International Nuclear Information System (INIS)
Murine models are useful for targeted radiotherapy pre-clinical experiments. These models can help to assess the potential interest of new radiopharmaceuticals. In this study, we developed a voxel-based mouse for dosimetric estimates. A female nude mouse (30 g) was frozen and cut into slices. High-resolution digital photographs were taken directly on the frozen block after each section. Images were segmented manually. Monoenergetic photon or electron sources were simulated using the MCNP4c2 Monte Carlo code for each source organ, in order to give tables of S-factors (in Gy Bq sup - sup 1 s sup - sup 1) for all target organs. Results obtained from monoenergetic particles were then used to generate S-factors for several radionuclides of potential interest in targeted radiotherapy. Thirteen source and 25 target regions were considered in this study. For each source region, 16 photon and 16 electron energies were simulated. Absorbed fractions, specific absorbed fractions and S-factors were calculated for 16 radionuclides of interest for targeted radiotherapy. The results obtained generally agree well with data published previously. For electron energies ranging from 0.1 to 2.5 MeV, the self-absorbed fraction varies from 0.98 to 0.376 for the liver, and from 0.89 to 0.04 for the thyroid. Electrons cannot be considered as 'non-penetrating' radiation for energies above 0.5 MeV for mouse organs. This observation can be generalized to radionuclides: for example, the beta self-absorbed fraction for the thyroid was 0.616 for I-131; absorbed fractions for Y-90 for left kidney-to-left kidney and for left kidney-to-spleen were 0.486 and 0.058, respectively. Our voxel-based mouse allowed us to generate a dosimetric database for use in preclinical targeted radiotherapy experiments. (author)
International Nuclear Information System (INIS)
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent for isotropic exposure of an adult female and an adult male to deuterons (2H+) in the energy range 10 MeV-1 TeV (0.01-1000 GeV). Coefficients were calculated using the Monte Carlo transport code MCNPX 2.7.C and BodyBuilderTM 1.3 anthropomorphic phantoms. Phantoms were modified to allow calculation of the effective dose to a Reference Person using tissues and tissue weighting factors from 1990 and 2007 recommendations of the International Commission on Radiological Protection (ICRP) and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. Coefficients for the equivalent and effective dose incorporated a radiation weighting factor of 2. At 15 of 19 energies for which coefficients for the effective dose were calculated, coefficients based on ICRP 1990 and 2007 recommendations differed by < 3 %. The greatest difference, 47 %, occurred at 30 MeV. (authors)
Monte Carlo simulation to study the doses in an accelerator BNCT treatment
International Nuclear Information System (INIS)
In this work the 7Li(p, n)7Be reaction has been studied as a neutron source for accelerator-based BNCT (Boron Neutron Capture Therapy). In order to optimize the design of the neutron production target and the beam shaping assembly, extensive MCNP simulations have been performed. These simulations include a thick Li metal target, a whole-body phantom, a moderator-reflector assembly (Al/AlF3 as moderator and graphite as reflector) and the treatment room. The doses were evaluated for two proton bombarding energies of 1.92 MeV (near to the threshold of the reaction) and 2.3 MeV (near to the resonance of the reaction) and for three Al/ALF3 moderator thicknesses (18, 26 and 34 cm). To assess the doses, a comparison using a Tumor Control Probability (TCP) model was done. In a second instance, the effect of the specific skin radiosensitivity (an RBE of 2.5 for the 10B(n,α)7Li reaction) and a 10B uptake of 17 ppm was considered for the scalp. Finally, the simulations show the advantage of irradiating with near-resonance-energy protons (2.3 MeV) because of the high neutron yield at this energy, leading to the lowest treatment times. Moreover, the 26 cm Al/AlF3 moderator has shown the best performance among the studied cases. (author)
International Nuclear Information System (INIS)
Radiation therapy treatment planning requires accurate determination of absorbed dose in the patient. Monte Carlo simulation is the most accurate method for solving the transport problem of particles in matter. This thesis is the first study dealing with the validation of the Monte Carlo simulation platform GATE (GEANT4 Application for Tomographic Emission), based on GEANT4 (Geometry And Tracking) libraries, for the computation of absorbed dose deposited by electron beams. This thesis aims at demonstrating that GATE/GEANT4 calculations are able to reach treatment planning requirements in situations where analytical algorithms are not satisfactory. The goal is to prove that GATE/GEANT4 is useful for treatment planning using electrons and competes with well validated Monte Carlo codes. This is demonstrated by the simulations with GATE/GEANT4 of realistic electron beams and electron sources used for external radiation therapy or targeted radiation therapy. The computed absorbed dose distributions are in agreement with experimental measurements and/or calculations from other Monte Carlo codes. Furthermore, guidelines are proposed to fix the physics parameters of the GATE/GEANT4 simulations in order to ensure the accuracy of absorbed dose calculations according to radiation therapy requirements. (author)
Petoukhova, A. L.; van Wingerden, K.; Wiggenraad, R. G. J.; van de Vaart, P. J. M.; van Egmond, J.; Franken, E. M.; van Santvoort, J. P. C.
2010-08-01
This study presents data for verification of the iPlan RT Monte Carlo (MC) dose algorithm (BrainLAB, Feldkirchen, Germany). MC calculations were compared with pencil beam (PB) calculations and verification measurements in phantoms with lung-equivalent material, air cavities or bone-equivalent material to mimic head and neck and thorax and in an Alderson anthropomorphic phantom. Dosimetric accuracy of MC for the micro-multileaf collimator (MLC) simulation was tested in a homogeneous phantom. All measurements were performed using an ionization chamber and Kodak EDR2 films with Novalis 6 MV photon beams. Dose distributions measured with film and calculated with MC in the homogeneous phantom are in excellent agreement for oval, C and squiggle-shaped fields and for a clinical IMRT plan. For a field with completely closed MLC, MC is much closer to the experimental result than the PB calculations. For fields larger than the dimensions of the inhomogeneities the MC calculations show excellent agreement (within 3%/1 mm) with the experimental data. MC calculations in the anthropomorphic phantom show good agreement with measurements for conformal beam plans and reasonable agreement for dynamic conformal arc and IMRT plans. For 6 head and neck and 15 lung patients a comparison of the MC plan with the PB plan was performed. Our results demonstrate that MC is able to accurately predict the dose in the presence of inhomogeneities typical for head and neck and thorax regions with reasonable calculation times (5-20 min). Lateral electron transport was well reproduced in MC calculations. We are planning to implement MC calculations for head and neck and lung cancer patients.
Monte-Carlo determination of dose rates in spherical PWR shield
International Nuclear Information System (INIS)
An operating nuclear reactor is a source of potentially dangerous nuclear radiation. It emits many different types of nuclear radiations. However, the neutrons and gamma rays are the main sources of radiation to give the contributions to the radiation situation in the reactor during operation, and the principal concern of reactor shielding. In current study the neutron and gamma radiation dose rates at different depths in concrete bio-shield of a PWR are calculated in spherical model by M-C simulation with using MCNP4C2. The simulation results are compared with the results obtained from similar calculations based on S8P3 spherical approximation with using the ANISN code. (author)
International Nuclear Information System (INIS)
In deuterium-deuterium (D-D) and deuterium-tritium (D-T) fusion plasmas neutrons are produced causing activation of JET machine components. For safe operation and maintenance it is important to be able to predict the induced activation and the resulting shut down dose rates. This requires a suitable system of codes which is capable of simulating both the neutron induced material activation during operation and the decay gamma radiation transport after shut-down in the proper 3-D geometry. Two methodologies to calculate the dose rate in fusion devices have been developed recently and applied to fusion machines, both using the MCNP Monte Carlo code. FZK has developed a more classical approach, the rigorous 2-step (R2S) system in which MCNP is coupled to the FISPACT inventory code with an automated routing. ENEA, in collaboration with the ITER Team, has developed an alternative approach, the direct 1 step method (D1S). Neutron and decay gamma transport are handled in one single MCNP run, using an ad hoc cross section library. The intention was to tightly couple the neutron induced production of a radio-isotope and the emission of its decay gammas for an accurate spatial distribution and a reliable calculated statistical error. The two methods have been used by the two Associations to calculate the dose rate in five positions of JET machine, two inside the vacuum chamber and three outside, at cooling times between 1 second and 1 year after shutdown. The same MCNP model and irradiation conditions have been assumed. The exercise has been proposed and financed in the frame of the Fusion Technological Program of the JET machine. The scope is to supply the designers with the most reliable tool and data to calculate the dose rate on fusion machines. Results showed that there is a good agreement: the differences range between 5-35%. The next step to be considered in 2003 will be an exercise in which the comparison will be done with dose-rate data from JET taken during and
Energy Technology Data Exchange (ETDEWEB)
Wang, L; Fourkal, E; Hayes, S; Jin, L; Ma, C [Fox Chase Cancer Center, Philadelphia, PA (United States)
2014-06-01
Purpose: To study the dosimetric difference resulted in using the pencil beam algorithm instead of Monte Carlo (MC) methods for tumors adjacent to the skull. Methods: We retrospectively calculated the dosimetric differences between RT and MC algorithms for brain tumors treated with CyberKnife located adjacent to the skull for 18 patients (total of 27 tumors). The median tumor sizes was 0.53-cc (range 0.018-cc to 26.2-cc). The absolute mean distance from the tumor to the skull was 2.11 mm (range - 17.0 mm to 9.2 mm). The dosimetric variables examined include the mean, maximum, and minimum doses to the target, the target coverage (TC) and conformality index. The MC calculation used the same MUs as the RT dose calculation without further normalization and 1% statistical uncertainty. The differences were analyzed by tumor size and distance from the skull. Results: The TC was generally reduced with the MC calculation (24 out of 27 cases). The average difference in TC between RT and MC was 3.3% (range 0.0% to 23.5%). When the TC was deemed unacceptable, the plans were re-normalized in order to increase the TC to 99%. This resulted in a 6.9% maximum change in the prescription isodose line. The maximum changes in the mean, maximum, and minimum doses were 5.4 %, 7.7%, and 8.4%, respectively, before re-normalization. When the TC was analyzed with regards to target size, it was found that the worst coverage occurred with the smaller targets (0.018-cc). When the TC was analyzed with regards to the distance to the skull, there was no correlation between proximity to the skull and TC between the RT and MC plans. Conclusions: For smaller targets (< 4.0-cc), MC should be used to re-evaluate the dose coverage after RT is used for the initial dose calculation in order to ensure target coverage.
Energy Technology Data Exchange (ETDEWEB)
Petrizzi, L.; Batistoni, P.; Migliori, S. [Associazione EURATOM ENEA sulla Fusione, Frascati (Roma) (Italy); Chen, Y.; Fischer, U.; Pereslavtsev, P. [Association FZK-EURATOM Forschungszentrum Karlsruhe (Germany); Loughlin, M. [EURATOM/UKAEA Fusion Association, Culham Science Centre, Abingdon, Oxfordshire, OX (United Kingdom); Secco, A. [Nice Srl Via Serra 33 Camerano Casasco AT (Italy)
2003-07-01
In deuterium-deuterium (D-D) and deuterium-tritium (D-T) fusion plasmas neutrons are produced causing activation of JET machine components. For safe operation and maintenance it is important to be able to predict the induced activation and the resulting shut down dose rates. This requires a suitable system of codes which is capable of simulating both the neutron induced material activation during operation and the decay gamma radiation transport after shut-down in the proper 3-D geometry. Two methodologies to calculate the dose rate in fusion devices have been developed recently and applied to fusion machines, both using the MCNP Monte Carlo code. FZK has developed a more classical approach, the rigorous 2-step (R2S) system in which MCNP is coupled to the FISPACT inventory code with an automated routing. ENEA, in collaboration with the ITER Team, has developed an alternative approach, the direct 1 step method (D1S). Neutron and decay gamma transport are handled in one single MCNP run, using an ad hoc cross section library. The intention was to tightly couple the neutron induced production of a radio-isotope and the emission of its decay gammas for an accurate spatial distribution and a reliable calculated statistical error. The two methods have been used by the two Associations to calculate the dose rate in five positions of JET machine, two inside the vacuum chamber and three outside, at cooling times between 1 second and 1 year after shutdown. The same MCNP model and irradiation conditions have been assumed. The exercise has been proposed and financed in the frame of the Fusion Technological Program of the JET machine. The scope is to supply the designers with the most reliable tool and data to calculate the dose rate on fusion machines. Results showed that there is a good agreement: the differences range between 5-35%. The next step to be considered in 2003 will be an exercise in which the comparison will be done with dose-rate data from JET taken during and
International Nuclear Information System (INIS)
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent, for isotropic exposure of an adult male and an adult female to helions (3He2+) in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). Calculations were performed using Monte Carlo transport code MCNPX 2.7.C and BodyBuilderTM 1.3 anthropomorphic phantoms modified to allow calculation of effective dose using tissues and tissue weighting factors from either the 1990 or 2007 recommendations of the International Commission on Radiological Protection (ICRP), and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. At 15 of the 19 energies for which coefficients for effective dose were calculated, coefficients based on ICRP 2007 and 1990 recommendations differed by less than 2%. The greatest difference, 62%, occurred at 100 MeV. Published by Oxford Univ. Press on behalf of the U.S. Government 2010. (authors)
International Nuclear Information System (INIS)
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent for isotropic exposure of an adult female and an adult male to tritons (3H+) in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). Coefficients were calculated using Monte Carlo transport code MCNPX 2.7.C and BodyBuilderTM 1.3 anthropomorphic phantoms. Phantoms were modified to allow calculation of effective dose to a Reference Person using tissues and tissue weighting factors from 1990 and 2007 recommendations of the International Commission on Radiological Protection (ICRP) and calculation of gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. At 15 of the 19 energies for which coefficients for effective dose were calculated, coefficients based on ICRP 2007 and 1990 recommendations differed by less than 3%. The greatest difference, 43%, occurred at 30 MeV. Published by Oxford Univ. Press on behalf of the US Government 2010. (authors)
Improvement and benchmarking of the new shutdown dose estimation method by Monte Carlo code
International Nuclear Information System (INIS)
In the ITER (international thermonuclear experimental reactor,) project calculations of the dose rate after shutdown are very important and their results are critical for the machine design. A new method has been proposed which makes use of MCNP also for decay gamma-ray transport calculations. The objective is to have an easy tool giving results affected by low uncertainty due to the modeling or simplifications in the flux shape assumptions. Further improvements to this method are here presented. This methodology has been developed, in the ITER frame, in a limited case in which the radioactivity comes only from the Vacuum Vessel (made of stainless steel) till a time around few days after ITER shutdown. Further improvement is required to make it applicable to more general cases (at different times and/or with different materials). Some benchmark results are shown. Discrepancies between the different methods are due mainly to the different cross section used. Agreement with available ad hoc experiment is very good. (orig.)
International Nuclear Information System (INIS)
Our aim in the study was to assess the eye lens dose reduction resulting from the use of radioprotective bismuth garments to shield the eyes of pediatric patients undergoing head CT. The Monte Carlo N-particle transport code and mathematical humanoid phantoms representing the average individual at different ages were used to determine eye lens dose reduction accomplished with bismuth shielding of the eye in the following simulated CT scans: (a) scanning of the orbits, (b) scanning of the whole head, and (c) 20 deg. angled scanning of the brain excluding the orbits. The effect of bismuth shielding on the eye lens dose was also investigated using an anthropomorphic phantom and thermoluminescence dosimetry (TLD). Eye lens dose reduction achieved by bismuth shielding was measured in 16 patients undergoing multiphase CT scanning of the head. The patient's scans were divided in the following: CT examinations where the eye globes were entirely included (n=5), partly included (n=6) and excluded (n=5) from the scanned region. The eye lens dose reduction depended mainly on the scan boundaries set by an operator. The average eye lens dose reduction determined by Monte Carlo simulation was 38.2%, 33.0% and <1% for CT scans of the orbits, whole head, and brain with an angled gantry, respectively. The difference between the Monte Carlo derived eye lens dose reduction factor values and corresponding values determined directly by using the anthropomorphic phantom head was found less than 5%. The mean eye lens dose reduction achieved by bismuth shielding in pediatric patients were 34%, 20% and <2% when eye globes were entirely included, partly included and excluded from the scanned region, respectively. A significant reduction in eye lens dose may be achieved by using superficial orbital bismuth shielding during pediatric head CT scans. However, bismuth garments should not be used in children when the eyes are excluded from the primarily exposed region
International Nuclear Information System (INIS)
A pair of walking phantoms was developed from deformable mesh phantoms to represent individuals walking on contaminated ground. The Monte Carlo N-particle extended version code was used to calculate organ doses from ground contamination scenarios involving parallel and isotropic planar sources of Cs-137 and Co-60 with concentrations of 30 kBqm-2. For the parallel plane source case, the organ doses were up to 78% greater for walking phantoms than those for the standing phantoms. The dose difference is because the widely open legs during walking provide less shielding to several organs, especially the kidneys, ovaries and liver, from parallel sources. The effective doses of the walking phantoms were on average 15% higher than the standing phantoms. On the other hand, when isotropic planar sources were considered, no significant dose difference was observed. This study demonstrated the feasibility of using deformable phantoms to represent realistic postures for organ dose calculations in environmental dosimetry studies. (authors)
International Nuclear Information System (INIS)
Mathematical phantoms of representing the adult female at 0,3,6 and 9 months of gestation were constructed, and organ doses and effective doses were calculated in standard irradiation environment and abdomen X-ray examination. Phantoms were based on the data set of ORNL and MCNP4B, a general-purposed Monte Carlo code was used for dose calculation. Firstly, organ doses and effective doses of pregnant female and fetus for 0.4 and 0.8MeV broad parallel beam incident from AP and PA direction were calculated. Then, the same calculations were performed in abdomen AP X-ray examination. As gestation period went by, effective doses of pregnant woman decreased because major organs were shielded by expanded uterus. Fetus of 9 month is lower than that of 6 month because of shielding effect of placement for AP irradiation
Calculation of equivalent dose index for electrons from 5,0 to 22,0 MeV by the Monte Carlo method
International Nuclear Information System (INIS)
The index of equivalent dose in depth and in a sphere surface of a soft tissue equivalent material were determined by Monte Carlo method for electron irradiations from 5,0 to 22.00 MeV. The effect of different irradiation geometries which simulate the incidence of onedirectional opposite rotational and isotropic beams was studied. It is also shown that the detector of wall thickness with 0.5g/cm2 and isotropic response com be used to measure index of equivalent dose for fast electrons. The alternative concept of average equivalent dose for radiation protection is discussed. (M.C.K.)
Energy Technology Data Exchange (ETDEWEB)
Rojas C, E.L.; Varon T, C.F.; Pedraza N, R. [ININ, 52750 La Marquesa, Estado de Mexico (Mexico)]. e-mail: elrc@nuclear.inin.mx
2007-07-01
The treatment of the breast cancer at early stages is of vital importance. For that, most of the investigations are dedicated to the early detection of the suffering and their treatment. As investigation consequence and clinical practice, in 2002 it was developed in U.S.A. an irradiation system of high dose rate known as Mammosite. In this work we carry out dose calculations for a simplified Mammosite system with the Monte Carlo Penelope simulation code and MCNPX, varying the concentration of the contrast material that it is used in the one. (Author)
International Nuclear Information System (INIS)
Purpose: To validate the feasibility of developing a radiotherapy unit with kilovoltage X-rays through actual irradiation of live rabbit lungs, and to explore the practical issues anticipated in future clinical application to humans through Monte Carlo dose simulation. Methods and Materials: A converging stereotactic irradiation unit was developed, consisting of a modified diagnostic computed tomography (CT) scanner. A tiny cylindrical volume in 13 normal rabbit lungs was individually irradiated with single fractional absorbed doses of 15, 30, 45, and 60 Gy. Observational CT scanning of the whole lung was performed every 2 weeks for 30 weeks after irradiation. After 30 weeks, histopathologic specimens of the lungs were examined. Dose distribution was simulated using the Monte Carlo method, and dose-volume histograms were calculated according to the data. A trial estimation of the effect of respiratory movement on dose distribution was made. Results: A localized hypodense change and subsequent reticular opacity around the planning target volume (PTV) were observed in CT images of rabbit lungs. Dose-volume histograms of the PTVs and organs at risk showed a focused dose distribution to the target and sufficient dose lowering in the organs at risk. Our estimate of the dose distribution, taking respiratory movement into account, revealed dose reduction in the PTV. Conclusions: A converging stereotactic irradiation unit using kilovoltage X-rays was able to generate a focused radiobiologic reaction in rabbit lungs. Dose-volume histogram analysis and estimated sagittal dose distribution, considering respiratory movement, clarified the characteristics of the irradiation received from this type of unit.
International Nuclear Information System (INIS)
-Small animal PET allows qualitative assessment and quantitative measurement of biochemical processes in vivo, but the accuracy and reproducibility of imaging results can be affected by several parameters. The first aim of this study was to investigate the performance of different CT-based attenuation correction strategies and assess the resulting impact on PET images. The absorbed dose in different tissues caused by scanning procedures was also discussed to minimize biologic damage generated by radiation exposure due to PET/CT scanning. A small animal PET/CT system was modeled based on Monte Carlo simulation to generate imaging results and dose distribution. Three energy mapping methods, including the bilinear scaling method, the dual-energy method and the hybrid method which combines the kVp conversion and the dual-energy method, were investigated comparatively through assessing the accuracy of estimating linear attenuation coefficient at 511 keV and the bias introduced into PET quantification results due to CT-based attenuation correction. Our results showed that the hybrid method outperformed the bilinear scaling method, while the dual-energy method achieved the highest accuracy among the three energy mapping methods. Overall, the accuracy of PET quantification results have similar trend as that for the estimation of linear attenuation coefficients, whereas the differences between the three methods are more obvious in the estimation of linear attenuation coefficients than in the PET quantification results. With regards to radiation exposure from CT, the absorbed dose ranged between 7.29-45.58 mGy for 50-kVp scan and between 6.61-39.28 mGy for 80-kVp scan. For 18F radioactivity concentration of 1.86x105 Bq/ml, the PET absorbed dose was around 24 cGy for tumor with a target-to-background ratio of 8. The radiation levels for CT scans are not lethal to the animal, but concurrent use of PET in longitudinal study can increase the risk of biological effects. The
Monte Carlo design of a system for the detection of explosive materials and analysis of the dose
International Nuclear Information System (INIS)
The problems associated with insecurity and terrorism have forced to designing systems for detecting nuclear materials, drugs and explosives that are installed on roads, ports and airports. Organic materials are composed of C, H, O and N; similarly the explosive materials are manufactured which can be distinguished by the concentration of these elements. Its elemental composition, particularly the concentration of hydrogen and oxygen, allow distinguish them from other organic substances. When these materials are irradiated with neutrons nuclear reactions (n, γ) are produced, where the emitted photons are ready gamma rays whose energy is characteristic of each element and its abundance allows estimating their concentration. The aim of this study was designed using Monte Carlo methods a system with neutron source, gamma rays detector and moderator able to distinguish the presence of Rdx and urea. In design were used as moderators: paraffin, light water, polyethylene and graphite; as detectors were used HPGe and the NaI(Tl). The design that showed the best performance was the moderator of light water and HPGe, with a source of 241AmBe. For this design, the values of ambient dose equivalent around the system were calculated. (Author)
International Nuclear Information System (INIS)
In this research, we used a 135 MeV/nucleon carbon-ion beam to irradiate a biological sample composed of fresh chicken meat and bones, which was placed in front of a PAGAT gel dosimeter, and compared the measured and simulated transverse-relaxation-rate (R2) distributions in the gel dosimeter. We experimentally measured the three-dimensional R2 distribution, which records the dose induced by particles penetrating the sample, by using magnetic resonance imaging. The obtained R2 distribution reflected the heterogeneity of the biological sample. We also conducted Monte Carlo simulations using the PHITS code by reconstructing the elemental composition of the biological sample from its computed tomography images while taking into account the dependence of the gel response on the linear energy transfer. The simulation reproduced the experimental distal edge structure of the R2 distribution with an accuracy under about 2 mm, which is approximately the same as the voxel size currently used in treatment planning. (paper)
Furuta, T.; Maeyama, T.; Ishikawa, K. L.; Fukunishi, N.; Fukasaku, K.; Takagi, S.; Noda, S.; Himeno, R.; Hayashi, S.
2015-08-01
In this research, we used a 135 MeV/nucleon carbon-ion beam to irradiate a biological sample composed of fresh chicken meat and bones, which was placed in front of a PAGAT gel dosimeter, and compared the measured and simulated transverse-relaxation-rate (R2) distributions in the gel dosimeter. We experimentally measured the three-dimensional R2 distribution, which records the dose induced by particles penetrating the sample, by using magnetic resonance imaging. The obtained R2 distribution reflected the heterogeneity of the biological sample. We also conducted Monte Carlo simulations using the PHITS code by reconstructing the elemental composition of the biological sample from its computed tomography images while taking into account the dependence of the gel response on the linear energy transfer. The simulation reproduced the experimental distal edge structure of the R2 distribution with an accuracy under about 2 mm, which is approximately the same as the voxel size currently used in treatment planning.
DS86 neutron dose: Monte Carlo analysis for depth profile of 152Eu activity in a large stone sample.
Endo, S; Iwatani, K; Oka, T; Hoshi, M; Shizuma, K; Imanaka, T; Takada, J; Fujita, S; Hasai, H
1999-06-01
The depth profile of 152Eu activity induced in a large granite stone pillar by Hiroshima atomic bomb neutrons was calculated by a Monte Carlo N-Particle Transport Code (MCNP). The pillar was on the Motoyasu Bridge, located at a distance of 132 m (WSW) from the hypocenter. It was a square column with a horizontal sectional size of 82.5 cm x 82.5 cm and height of 179 cm. Twenty-one cells from the north to south surface at the central height of the column were specified for the calculation and 152Eu activities for each cell were calculated. The incident neutron spectrum was assumed to be the angular fluence data of the Dosimetry System 1986 (DS86). The angular dependence of the spectrum was taken into account by dividing the whole solid angle into twenty-six directions. The calculated depth profile of specific activity did not agree with the measured profile. A discrepancy was found in the absolute values at each depth with a mean multiplication factor of 0.58 and also in the shape of the relative profile. The results indicated that a reassessment of the neutron energy spectrum in DS86 is required for correct dose estimation. PMID:10494148
Li, Ping
2014-01-01
The term "CoRE kernel" stands for correlation-resemblance kernel. In many applications (e.g., vision), the data are often high-dimensional, sparse, and non-binary. We propose two types of (nonlinear) CoRE kernels for non-binary sparse data and demonstrate the effectiveness of the new kernels through a classification experiment. CoRE kernels are simple with no tuning parameters. However, training nonlinear kernel SVM can be (very) costly in time and memory and may not be suitable for truly lar...
Energy Technology Data Exchange (ETDEWEB)
May, Matthias S.; Kuettner, Axel; Lell, Michael M.; Wuest, Wolfgang; Scharf, Michael; Uder, Michael [University of Erlangen, Department of Radiology, Erlangen (Germany); Deak, Paul; Kalender, Willi A. [University of Erlangen, Department of Medical Physics, Erlangen (Germany); Keller, Andrea K.; Haeberle, Lothar [University of Erlangen, Department of Medical Informatics, Biometry and Epidemiology, Erlangen (Germany); Achenbach, Stephan; Seltmann, Martin [University of Erlangen, Department of Cardiology, Erlangen (Germany)
2012-03-15
To evaluate radiation dose levels in patients undergoing spiral coronary computed tomography angiography (CTA) on a dual-source system in clinical routine. Coronary CTA was performed for 56 patients with electrocardiogram-triggered tube current modulation (TCM) and heart-rate (HR) dependent pitch adaptation. Individual Monte Carlo (MC) simulations were performed for dose assessment. Retrospective simulations with constant tube current (CTC) served as reference. Lung tissue was segmented and used for organ and effective dose (ED) calculation. Estimates for mean relative ED was 7.1 {+-} 2.1 mSv/100 mAs for TCM and 12.5 {+-} 5.3 mSv/100 mAs for CTC (P < 0.001). Relative dose reduction at low HR ({<=}60 bpm) was highest (49 {+-} 5%) compared to intermediate (60-70 bpm, 33 {+-} 12%) and high HR (>70 bpm, 29 {+-} 12%). However lowest ED is achieved at high HR (5.2 {+-} 1.5 mSv/100 mAs), compared with intermediate (6.7 {+-} 1.6 mSv/100 mAs) and low (8.3 {+-} 2.1 mSv/100 mAs) HR when automated pitch adaptation is applied. Radiation dose savings up to 52% are achievable by TCM at low and regular HR. However lowest ED is attained at high HR by pitch adaptation despite inferior radiation dose reduction by TCM. circle Monte Carlo simulations allow for individual radiation dose calculations. (orig.)
Montanari, Davide; Silvestri, Chiara; Graves, Yan J; Yan, Hao; Cervino, Laura; Rice, Roger; Jiang, Steve B; Jia, Xun
2013-01-01
Cone beam CT (CBCT) has been widely used for patient setup in image guided radiation therapy (IGRT). Radiation dose from CBCT scans has become a clinical concern. The purposes of this study are 1) to commission a GPU-based Monte Carlo (MC) dose calculation package gCTD for Varian On-Board Imaging (OBI) system and test the calculation accuracy, and 2) to quantitatively evaluate CBCT dose from the OBI system in typical IGRT scan protocols. We first conducted dose measurements in a water phantom. X-ray source model parameters used in gCTD are obtained through a commissioning process. gCTD accuracy is demonstrated by comparing calculations with measurements in water and in CTDI phantoms. 25 brain cancer patients are used to study dose in a standard-dose head protocol, and 25 prostate cancer patients are used to study dose in pelvis protocol and pelvis spotlight protocol. Mean dose to each organ is calculated. Mean dose to 2% voxels that have the highest dose is also computed to quantify the maximum dose. It is fo...
International Nuclear Information System (INIS)
Purpose: The commercial release of volumetric modulated arc therapy techniques using a conventional linear accelerator and the growing number of helical tomotherapy users have triggered renewed interest in dose verification methods, and also in tools for exploring the impact of machine tolerance and patient motion on dose distributions without the need to approximate time-varying parameters such as gantry position, MLC leaf motion, or patient motion. To this end we have developed a Monte Carlo-based calculation method capable of simulating a wide variety of treatment techniques without the need to resort to discretization approximations. Methods: The ability to perform complete position-probability-sampled Monte Carlo dose calculations was implemented in the BEAMnrc/DOSXZYnrc user codes of EGSnrc. The method includes full accelerator head simulations of our tomotherapy and Elekta linacs, and a realistic representation of continous motion via the sampling of a time variable. The functionality of this algorithm was tested via comparisons with both measurements and treatment planning dose distributions for four types of treatment techniques: 3D conformal, step-shoot intensity modulated radiation therapy, helical tomotherapy, and volumetric modulated arc therapy. Results: For static fields, the absolute dose agreement between the EGSnrc Monte Carlo calculations and measurements is within 2%/1 mm. Absolute dose agreement between Monte Carlo calculations and treatment planning system for the four different treatment techniques is within 3%/3 mm. Discrepancies with the tomotherapy TPS on the order of 10%/5 mm were observed for the extreme example of a small target located 15 cm off-axis and planned with a low modulation factor. The increase in simulation time associated with using position-probability sampling, as opposed to the discretization approach, was less than 2% in most cases. Conclusions: A single Monte Carlo simulation method can be used to calculate patient
Energy Technology Data Exchange (ETDEWEB)
Yani, Sitti, E-mail: sitti.yani@s.itb.ac.id [Nuclear Physics and Biophysics Division, Physics Department, Institut Teknologi Bandung (Indonesia); Akademi Kebidanan Pelita Ibu, Kendari (Indonesia); Dirgayussa, I Gde E.; Haryanto, Freddy; Arif, Idam [Nuclear Physics and Biophysics Division, Physics Department, Institut Teknologi Bandung (Indonesia); Rhani, Moh. Fadhillah [Tan Tock Seng Hospital (Singapore)
2015-09-30
Recently, Monte Carlo (MC) calculation method has reported as the most accurate method of predicting dose distributions in radiotherapy. The MC code system (especially DOSXYZnrc) has been used to investigate the different voxel (volume elements) sizes effect on the accuracy of dose distributions. To investigate this effect on dosimetry parameters, calculations were made with three different voxel sizes. The effects were investigated with dose distribution calculations for seven voxel sizes: 1 × 1 × 0.1 cm{sup 3}, 1 × 1 × 0.5 cm{sup 3}, and 1 × 1 × 0.8 cm{sup 3}. The 1 × 10{sup 9} histories were simulated in order to get statistical uncertainties of 2%. This simulation takes about 9-10 hours to complete. Measurements are made with field sizes 10 × 10 cm2 for the 6 MV photon beams with Gaussian intensity distribution FWHM 0.1 cm and SSD 100.1 cm. MC simulated and measured dose distributions in a water phantom. The output of this simulation i.e. the percent depth dose and dose profile in d{sub max} from the three sets of calculations are presented and comparisons are made with the experiment data from TTSH (Tan Tock Seng Hospital, Singapore) in 0-5 cm depth. Dose that scored in voxels is a volume averaged estimate of the dose at the center of a voxel. The results in this study show that the difference between Monte Carlo simulation and experiment data depend on the voxel size both for percent depth dose (PDD) and profile dose. PDD scan on Z axis (depth) of water phantom, the big difference obtain in the voxel size 1 × 1 × 0.8 cm{sup 3} about 17%. In this study, the profile dose focused on high gradient dose area. Profile dose scan on Y axis and the big difference get in the voxel size 1 × 1 × 0.1 cm{sup 3} about 12%. This study demonstrated that the arrange voxel in Monte Carlo simulation becomes important.
International Nuclear Information System (INIS)
Recently, Monte Carlo (MC) calculation method has reported as the most accurate method of predicting dose distributions in radiotherapy. The MC code system (especially DOSXYZnrc) has been used to investigate the different voxel (volume elements) sizes effect on the accuracy of dose distributions. To investigate this effect on dosimetry parameters, calculations were made with three different voxel sizes. The effects were investigated with dose distribution calculations for seven voxel sizes: 1 × 1 × 0.1 cm3, 1 × 1 × 0.5 cm3, and 1 × 1 × 0.8 cm3. The 1 × 109 histories were simulated in order to get statistical uncertainties of 2%. This simulation takes about 9-10 hours to complete. Measurements are made with field sizes 10 × 10 cm2 for the 6 MV photon beams with Gaussian intensity distribution FWHM 0.1 cm and SSD 100.1 cm. MC simulated and measured dose distributions in a water phantom. The output of this simulation i.e. the percent depth dose and dose profile in dmax from the three sets of calculations are presented and comparisons are made with the experiment data from TTSH (Tan Tock Seng Hospital, Singapore) in 0-5 cm depth. Dose that scored in voxels is a volume averaged estimate of the dose at the center of a voxel. The results in this study show that the difference between Monte Carlo simulation and experiment data depend on the voxel size both for percent depth dose (PDD) and profile dose. PDD scan on Z axis (depth) of water phantom, the big difference obtain in the voxel size 1 × 1 × 0.8 cm3 about 17%. In this study, the profile dose focused on high gradient dose area. Profile dose scan on Y axis and the big difference get in the voxel size 1 × 1 × 0.1 cm3 about 12%. This study demonstrated that the arrange voxel in Monte Carlo simulation becomes important
Yani, Sitti; Dirgayussa, I. Gde E.; Rhani, Moh. Fadhillah; Haryanto, Freddy; Arif, Idam
2015-09-01
Recently, Monte Carlo (MC) calculation method has reported as the most accurate method of predicting dose distributions in radiotherapy. The MC code system (especially DOSXYZnrc) has been used to investigate the different voxel (volume elements) sizes effect on the accuracy of dose distributions. To investigate this effect on dosimetry parameters, calculations were made with three different voxel sizes. The effects were investigated with dose distribution calculations for seven voxel sizes: 1 × 1 × 0.1 cm3, 1 × 1 × 0.5 cm3, and 1 × 1 × 0.8 cm3. The 1 × 109 histories were simulated in order to get statistical uncertainties of 2%. This simulation takes about 9-10 hours to complete. Measurements are made with field sizes 10 × 10 cm2 for the 6 MV photon beams with Gaussian intensity distribution FWHM 0.1 cm and SSD 100.1 cm. MC simulated and measured dose distributions in a water phantom. The output of this simulation i.e. the percent depth dose and dose profile in dmax from the three sets of calculations are presented and comparisons are made with the experiment data from TTSH (Tan Tock Seng Hospital, Singapore) in 0-5 cm depth. Dose that scored in voxels is a volume averaged estimate of the dose at the center of a voxel. The results in this study show that the difference between Monte Carlo simulation and experiment data depend on the voxel size both for percent depth dose (PDD) and profile dose. PDD scan on Z axis (depth) of water phantom, the big difference obtain in the voxel size 1 × 1 × 0.8 cm3 about 17%. In this study, the profile dose focused on high gradient dose area. Profile dose scan on Y axis and the big difference get in the voxel size 1 × 1 × 0.1 cm3 about 12%. This study demonstrated that the arrange voxel in Monte Carlo simulation becomes important.
International Nuclear Information System (INIS)
Purpose: the lens of an eye is a particularly radiosensitive organ. This study investigates two different materials for eye shielding during CT scanning, i.e. a commercially available bismuth protector and a newly developed material for eye shielding, comprised of an alloy of Bi/Sb/Gd/W. Materials and methods: the radiation dose during head CT scanning was measured using thermoluminescence dosimeters and an anthropomorphic Alderson-RANDO phantom. A radiation dose reduction was compared to two shielding materials and to the condition without any eye shielding. The effect of gantry angulation that excludes the eyes from beam path was also investigated. Radiation dose measurements were validated using a Monte-Carlo simulation. For this simulation we used the EGSsnr code system, and a new application CTDOSPP was developed for simulation of the computed tomography examination. Eight radiologists evaluated the diagnostic quality of the images. Results: dose measurements and Monte-Carlo simulations are in good agreement. If the eye shields are placed in the primary beam path, bismuth eye shielding and the new material reduce the dose by up to 38% and 48%, respectively. Angling the gantry causes an 88% reduction in radiation dose. All shielding materials generate beam hardening artifacts located close to the protector, but the artifacts do not spread into the brain. (orig.)
International Nuclear Information System (INIS)
To evaluate radiation dose levels in patients undergoing spiral coronary computed tomography angiography (CTA) on a dual-source system in clinical routine. Coronary CTA was performed for 56 patients with electrocardiogram-triggered tube current modulation (TCM) and heart-rate (HR) dependent pitch adaptation. Individual Monte Carlo (MC) simulations were performed for dose assessment. Retrospective simulations with constant tube current (CTC) served as reference. Lung tissue was segmented and used for organ and effective dose (ED) calculation. Estimates for mean relative ED was 7.1 ± 2.1 mSv/100 mAs for TCM and 12.5 ± 5.3 mSv/100 mAs for CTC (P 70 bpm, 29 ± 12%). However lowest ED is achieved at high HR (5.2 ± 1.5 mSv/100 mAs), compared with intermediate (6.7 ± 1.6 mSv/100 mAs) and low (8.3 ± 2.1 mSv/100 mAs) HR when automated pitch adaptation is applied. Radiation dose savings up to 52% are achievable by TCM at low and regular HR. However lowest ED is attained at high HR by pitch adaptation despite inferior radiation dose reduction by TCM. circle Monte Carlo simulations allow for individual radiation dose calculations. (orig.)
Directory of Open Access Journals (Sweden)
Biniam Yohannes Tesfamicael
2014-03-01
Full Text Available Purpose: To construct a dose monitoring system based on an endorectal balloon coupled to thin scintillating fibers to study the dose to the rectum in proton therapy of prostate cancer.Method: A Geant4 Monte Carlo toolkit was used to simulate the proton therapy of prostate cancer, with an endorectal balloon and a set of scintillating fibers for immobilization and dosimetry measurements, respectively.Results: A linear response of the fibers to the dose delivered was observed to within less than 2%. Results obtained show that fibers close to the prostate recorded higher dose, with the closest fiber recording about one-third of the dose to the target. A 1/r2 (r is defined as center-to-center distance between the prostate and the fibers decrease was observed as one goes toward the frontal and distal regions. A very low dose was recorded by the fibers beneath the balloon which is a clear indication that the overall volume of the rectal wall that is exposed to a higher dose is relatively minimized. Further analysis showed a relatively linear relationship between the dose to the target and the dose to the top fibers (total 17, with a slope of (-0.07 ± 0.07 at large number of events per degree of rotation of the modulator wheel (i.e., dose.Conclusion: Thin (1 mm × 1 mm, long (1 m scintillating fibers were found to be ideal for real time in-vivo dose measurement to the rectum during proton therapy of prostate cancer. The linear response of the fibers to the dose delivered makes them good candidates as dosimeters. With thorough calibration and the ability to define a good correlation between the dose to the target and the dose to the fibers, such dosimeters can be used for real time dose verification to the target.-----------------------------------Cite this article as: Tesfamicael BY, Avery S, Gueye P, Lyons D, Mahesh M. Scintillating fiber based in-vivo dose monitoring system to the rectum in proton therapy of prostate cancer: A Geant4 Monte Carlo
International Nuclear Information System (INIS)
Cell dosimetry is relevant regarding new efforts in specific molecular radiotherapy using Auger, CE and beta emitters. Absorbed dose in cells can be obtained by means of the dose per unit cumulated activity (S-values), together with the activity distribution. In this work, Monte Carlo simulation codes PENELOPE and MCNPX were used to obtain cellular S-values for point and extended sources of electrons and beta emitting radionuclides in the nucleus of breast (MDA-MB231, MCF7) and prostate (PC3) cancer cell models. - Highlights: • Cellular S-values were calculated using Penelope and MCPNX Monte Carlo codes. • S-values were obtained for e− and beta emitting radionuclides in cancer cell models. • Breast (MDA-MB231, MCF7) and prostate (PC3) cancer cell models were investigated. • Results are relevant for specific targeted molecular radiotherapy cell dosimetry
Energy Technology Data Exchange (ETDEWEB)
Rojas C, E.L. [ININ, 52045 Ocoyoacac, Estado de Mexico (Mexico); Al-Dweri, F.M.O.; Lallena R, A.M. [Universidad de Granada, Granada (Spain)]. e-mail: elrc@nuclear.inin.mx
2005-07-01
In this work they are studied, by means of Monte Carlo simulation, the effects that take place in the dose profiles that are obtained with the Leksell Gamma Knife (R), when they are kept in account heterogeneities. The considered heterogeneities simulate the skull and the spaces of air that are in the head, like they can be the nasal breasts or the auditory conduits. The calculations were made using the Monte Carlo Penelope simulation code (v. 2003). The geometry of each one of the 201 sources that this instrument is composed, as well as of the corresponding channels of collimation of the Gamma Knife (R), it was described by means of a simplified model of geometry that has been recently studied. The obtained results when they are kept in mind the heterogeneities they present non worthless differences regarding those obtained when those are not considered. These differences are maximum in the proximities of the interfaces among different materials. (Author)
International Nuclear Information System (INIS)
Purpose: To provide a prescription dose for Monte Carlo (MC) treatment planning in patients with non-small-cell lung cancer according to tumor size and location. Methods: Fifty-three stereotactic radiotherapy plans designed using the equivalent path-length (EPL) algorithm were re-calculated using MC. Plans were compared by the minimum dose to 95% of the PTV (D95), the heterogeneity index (HI) and the mean dose to organs at risk (OARs). Based on changes in D95, the prescription dose was converted from EPL to MC. Based on changes in HI, we examined the feasibility of MC prescription to plans re-calculated but not re-optimized with MC. Results: The MC fraction dose for peripheral tumors is 16-18 Gy depending on tumor size. For central tumors the MC dose was reduced less than for peripheral tumors. The HI decreased on average by 4-9% in peripheral tumors and 3-5% in central tumors. The mean dose to OARs was lower for MC than EPL, and correlated strongly (R2 = 0.98-0.99). Conclusion: For the conversion from EPL to MC we recommend a separate prescription dose according to tumor size. MC optimization is not required if a HI ≥ 70% is accepted. Dose constraints to OARs can be easily converted due to the high EPL-MC correlation.
Nasir, M.; Pratama, D.; Anam, C.; Haryanto, F.
2016-03-01
The aim of this research was to calculate Size Specific Dose Estimates (SSDE) generated by the varian OBI CBCT v1.4 X-ray tube working at 100 kV using EGSnrc Monte Carlo simulations. The EGSnrc Monte Carlo code used in this simulation was divided into two parts. Phase space file data resulted by the first part simulation became an input to the second part. This research was performed with varying phantom diameters of 5 to 35 cm and varying phantom lengths of 10 to 25 cm. Dose distribution data were used to calculate SSDE values using trapezoidal rule (trapz) function in a Matlab program. SSDE obtained from this calculation was compared to that in AAPM report and experimental data. It was obtained that the normalization of SSDE value for each phantom diameter was between 1.00 and 3.19. The normalization of SSDE value for each phantom length was between 0.96 and 1.07. The statistical error in this simulation was 4.98% for varying phantom diameters and 5.20% for varying phantom lengths. This study demonstrated the accuracy of the Monte Carlo technique in simulating the dose calculation. In the future, the influence of cylindrical phantom material to SSDE would be studied.
Toossi Mohammad Taghi Bahreyni; Ghorbani Mahdi; Sabet Leila Sobhkhiz; Akbari Fateme; Mehrpouyan Mohammad
2015-01-01
The aim of this study is the evaluation of electron dose enhancement and photon contamination production by various nanoparticles in the electron mode of a medical linac. MCNPX Monte Carlo code was used for simulation of Siemens Primus linac as well as a phantom and a tumor loaded with nanoparticles. Electron dose enhancement by Au, Ag, I and Fe2O3 nanoparticles of 7, 18 and 30 mg/ml concentrations for 8, 12 and 14 MeV electrons was calculated. The increase in photon contamination due to the ...
Edimo, Paul
2012-01-01
The present study is focused on the clinical validation of two electron Monte Carlo (eMC) based treatment planning systems (TPS), Oncentra MasterPlan TPS (OMTPS) and XiO eMC. We present a new approach on the commissioning process based on, (a) homogeneous water phantom validation, (b) heterogeneous phantom validation with film measurements and, (c) Full MC validation. As a first step, MC models of electron beams (4, 8, 12 and 18 MeV) from an Elekta SL25 medical linear accelerator were buil...
International Nuclear Information System (INIS)
This study was conducted to analyze the radiological impact in the experimental area of the Hard X-Ray beamline at Canadian Light Source under beam loss scenario during Top-up injection. The radiation doses were calculated using Monte Carlo code: FLUKA. The physical size, location, and material of the beamline components were adopted from the technical drawings and were incorporated in the FLUKA model. Three (03) beam loss scenarios were simulated: (i) Beam was miss-steered in the storage ring (ii) Beam hit misaligned components inside the ring and (iii) Beam was lost inside the primary optical enclosure (POE). Total ambient dose was calculated at several observation points for each scenario considering the injected beam as the primary source. The results and impacts were discussed. - Highlights: • Monte Carlo method was used to calculate radiation dose for a beamline at Canadian Light Source. • Three possible beam loss scenarios were studied. • The predicted worst dose was found below the regulatory dose limit
International Nuclear Information System (INIS)
This paper addresses issues in the application of the electron paramagnetic resonance (EPR) retrospective dosimetry with dental tissues exposed by radionuclides accumulated in the dentin. A simple dosimetric model of a tooth incorporating 90Sr is presented. The tooth is modelled as two concentric cylinders: the inner cylinder composed of dentin, and the outer cylindrical shell of enamel. Extensive Monte Carlo calculations were done to obtain the distributions of absorbed dose in dentin and enamel for teeth of different sizes. The results were used to calculate the mean absorbed doses in enamel that are directly measurable by EPR. A relationship between such measured doses and the specific activity of 90Sr in dentin was derived based on a simple model of 90Sr accumulation. The roles of different tooth tissues as dose detectors are analysed, and the importance of dentin as a dosimetric material for internal exposure is pointed out. (author)
Energy Technology Data Exchange (ETDEWEB)
Isambert, A.; Lefkopoulos, D. [Institut Gustave-Roussy, Medical Physics Dept., 94 - Villejuif (France); Brualla, L. [NCTeam, Strahlenklinik, Universitatsklinikum Essen (Germany); Benkebil, M. [DOSIsoft, 94 - Cachan (France)
2010-04-15
Purpose of study Monte Carlo based treatment planning system are known to be more accurate than analytical methods for performing absorbed dose estimation, particularly in and near heterogeneities. However, the required computation time can still be an issue. The present study focused on the determination of the optimum statistical uncertainty in order to minimise computation time while keeping the reliability of the absorbed dose estimation in treatments planned with electron-beams. Materials and methods Three radiotherapy plans (medulloblastoma, breast and gynaecological) were used to investigate the influence of the statistical uncertainty of the absorbed dose on the target volume dose-volume histograms (spinal cord, intra-mammary nodes and pelvic lymph nodes, respectively). Results The study of the dose-volume histograms showed that for statistical uncertainty levels (1 S.D.) above 2 to 3%, the standard deviation of the mean dose in the target volume calculated from the dose-volume histograms increases by at least 6%, reflecting the gradual flattening of the dose-volume histograms. Conclusions This work suggests that, in clinical context, Monte Carlo based absorbed dose estimations should be performed with a maximum statistical uncertainty of 2 to 3%. (authors)
International Nuclear Information System (INIS)
The purpose of this study is to provide detailed characteristics of incident photon beams for different field sizes and beam energies. This information is critical to the future development of accurate treatment planning systems. It also enhances our knowledge of radiotherapy photon beams. The EGS4 Monte Carlo code, BEAM, has been used to simulate 6 and 18 MV photon beams from a Varian Clinac-2100EX accelerator. A simulated realistic beam is stored in a phase space data file, which contains details of each particle's complete history including where it has been and where it has interacted. The phase space files are analysed to obtain energy spectra, angular distribution, fluence profile and mean energy profiles at the phantom surface for particles separated according to their charge and history. The accuracy of a simulated beam is validated by the excellent agreement between the Monte Carlo calculated and measured dose distributions. Measured depth-dose curves are obtained from depth-ionization curves by accounting for newly introduced chamber fluence corrections and the stopping-power ratios for realistic beams. The study presents calculated depth-dose components from different particles as well as calculated surface dose and contribution from different particles to surface dose across the field. It is shown that the increase of surface dose with the increase of the field size is mainly due to the increase of incident contaminant charged particles. At 6 MV, the incident charged particles contribute 7% to 21% of maximum dose at the surface when the field size increases from 10x10 to 40x40 cm2. At 18 MV, their contributions are up to 11% and 29% of maximum dose at the surface for 10x10 cm2 and 40x40 cm2 fields respectively. However, the fluence of these incident charged particles is less than 1% of incident photon fluence in all cases. (author)
International Nuclear Information System (INIS)
Purpose: To evaluate the accuracy of electron dose distribution calculated by the Varian Eclipse electron Monte Carlo (eMC) algorithm for use with recent commercially available bolus electron conformal therapy (ECT). Methods: eMC-calculated electron dose distributions for bolus ECT have been compared to those previously measured for cylindrical phantoms (retromolar trigone and nose), whose axial cross sections were based on the mid-PTV CT anatomy for each site. The phantoms consisted of SR4 muscle substitute, SR4 bone substitute, and air. The bolus ECT treatment plans were imported into the Eclipse treatment planning system and calculated using the maximum allowable histories (2×109), resulting in a statistical error of <0.2%. Smoothing was not used for these calculations. Differences between eMC-calculated and measured dose distributions were evaluated in terms of absolute dose difference as well as distance to agreement (DTA). Results: Results from the eMC for the retromolar trigone phantom showed 89% (41/46) of dose points within 3% dose difference or 3 mm DTA. There was an average dose difference of −0.12% with a standard deviation of 2.56%. Results for the nose phantom showed 95% (54/57) of dose points within 3% dose difference or 3 mm DTA. There was an average dose difference of 1.12% with a standard deviation of 3.03%. Dose calculation times for the retromolar trigone and nose treatment plans were 15 min and 22 min, respectively, using 16 processors (Intel Xeon E5-2690, 2.9 GHz) on a Varian Eclipse framework agent server (FAS). Results of this study were consistent with those previously reported for accuracy of the eMC electron dose algorithm and for the .decimal, Inc. pencil beam redefinition algorithm used to plan the bolus. Conclusion: These results show that the accuracy of the Eclipse eMC algorithm is suitable for clinical implementation of bolus ECT
International Nuclear Information System (INIS)
Purpose: The main focus of this work is to continue investigations into the Monte Carlo predicted skin doses seen in MRI-guided radiotherapy. In particular, the authors aim to characterize the 70 μm skin doses over a larger range of magnetic field strength and x-ray field size than in the current literature. The effect of surface orientation on both the entry and exit sides is also studied. Finally, the use of exit bolus is also investigated for minimizing the negative effects of the electron return effect (ERE) on the exit skin dose. Methods: High resolution GEANT4 Monte Carlo simulations of a water phantom exposed to a 6 MV x-ray beam (Varian 2100C) have been performed. Transverse magnetic fields of strengths between 0 and 3 T have been applied to a 30x30x20 cm3 phantom. This phantom is also altered to have variable entry and exit surfaces with respect to the beam central axis and they range from -75 deg. to +75 deg. The exit bolus simulated is a 1 cm thick (water equivalent) slab located on the beam exit side. Results: On the entry side, significant skin doses at the beam central axis are reported for large positive surface angles and strong magnetic fields. However, over the entry surface angle range of -30 deg. to -60 deg., the entry skin dose is comparable to or less than the zero magnetic field skin dose, regardless of magnetic field strength and field size. On the exit side, moderate to high central axis skin dose increases are expected except at large positive surface angles. For exit bolus of 1 cm thickness, the central axis exit skin dose becomes an almost consistent value regardless of magnetic field strength or exit surface angle. This is due to the almost complete absorption of the ERE electrons by the bolus. Conclusions: There is an ideal entry angle range of -30 deg. to -60 deg. where entry skin dose is comparable to or less than the zero magnetic field skin dose. Other than this, the entry skin dose increases are significant, especially at higher
Energy Technology Data Exchange (ETDEWEB)
Oborn, B. M.; Metcalfe, P. E.; Butson, M. J.; Rosenfeld, A. B. [Illawarra Cancer Care Centre (ICCC), Wollongong, New South Wales 2500 (Australia) and Centre for Medical Radiation Physics (CMRP), University of Wollongong, Wollongong, New South Wales 2500 (Australia); Centre for Medical Radiation Physics (CMRP), University of Wollongong, Wollongong, New South Wales 2500 (Australia); Illawarra Cancer Care Centre (ICCC), Wollongong, New South Wales 2500 (Australia); Centre for Medical Radiation Physics (CMRP), University of Wollongong, Wollongong, New South Wales 2500 (Australia)
2010-10-15
Purpose: The main focus of this work is to continue investigations into the Monte Carlo predicted skin doses seen in MRI-guided radiotherapy. In particular, the authors aim to characterize the 70 {mu}m skin doses over a larger range of magnetic field strength and x-ray field size than in the current literature. The effect of surface orientation on both the entry and exit sides is also studied. Finally, the use of exit bolus is also investigated for minimizing the negative effects of the electron return effect (ERE) on the exit skin dose. Methods: High resolution GEANT4 Monte Carlo simulations of a water phantom exposed to a 6 MV x-ray beam (Varian 2100C) have been performed. Transverse magnetic fields of strengths between 0 and 3 T have been applied to a 30x30x20 cm{sup 3} phantom. This phantom is also altered to have variable entry and exit surfaces with respect to the beam central axis and they range from -75 deg. to +75 deg. The exit bolus simulated is a 1 cm thick (water equivalent) slab located on the beam exit side. Results: On the entry side, significant skin doses at the beam central axis are reported for large positive surface angles and strong magnetic fields. However, over the entry surface angle range of -30 deg. to -60 deg., the entry skin dose is comparable to or less than the zero magnetic field skin dose, regardless of magnetic field strength and field size. On the exit side, moderate to high central axis skin dose increases are expected except at large positive surface angles. For exit bolus of 1 cm thickness, the central axis exit skin dose becomes an almost consistent value regardless of magnetic field strength or exit surface angle. This is due to the almost complete absorption of the ERE electrons by the bolus. Conclusions: There is an ideal entry angle range of -30 deg. to -60 deg. where entry skin dose is comparable to or less than the zero magnetic field skin dose. Other than this, the entry skin dose increases are significant, especially at
International Nuclear Information System (INIS)
The recommended target dose in radioiodine therapy of solitary hyperfunctioning thyroid nodules is 300-400 Gy and therefore higher than in other radiotherapies. This is due to the fact that an unknown, yet significant portion of the activity is stored in extranodular areas but is neglected in the calculatory dosimetry. We investigate the feasibility of determining the ratio of nodular and extranodular activity concentrations (uptakes) from post-therapeutically acquired planar scintigrams with Monte Carlo simulations in GATE. The geometry of a gamma camera with a high energy collimator was emulated in GATE (Version 5). A geometrical thyroid-neck phantom (GP) and the ICRP reference voxel phantoms 'Adult Female' (AF, 16 ml thyroid) and 'Adult Male' (AM, 19 ml thyroid) were used as source regions. Nodules of 1 ml and 3 ml volume were placed in the phantoms. For each phantom and each nodule 200 scintigraphic acquisitions were simulated. Uptake ratios of nodule and rest of thyroid ranging from 1 to 20 could be created by summation. Quantitative image analysis was performed by investigating the number of simulated counts in regions of interest (ROIs). ROIs were created by perpendicular projection of the phantom onto the camera plane to avoid a user dependant bias. The ratio of count densities in ROIs over the nodule and over the contralateral lobe, which should be least affected by nodular activity, was taken to be the best available measure for the uptake ratios. However, the predefined uptake ratios are underestimated by these count density ratios: For an uptake ratio of 20 the count ratios range from 4.5 (AF, 1 ml nodule) to 15.3 (AM, 3 ml nodule). Furthermore, the contralateral ROI is more strongly affected by nodular activity than expected: For an uptake ratio of 20 between nodule and rest of thyroid up to 29% of total counts in the ROI over the contralateral lobe are caused by decays in the nodule (AF 3 ml). In the case of the 1 ml nodules this effect is smaller: 9
Energy Technology Data Exchange (ETDEWEB)
Hammes, Jochen; Schmidt, Matthias; Schicha, Harald; Eschner, Wolfgang [Universitaetsklinikum Koeln (Germany). Klinik und Poliklinik fuer Nuklearmedizin; Pietrzyk, Uwe [Forschungszentrum Juelich GmbH (Germany). Inst. fuer Neurowissenschaften und Medizin (INM-4); Wuppertal Univ. (Germany). Fachbereich C - Physik
2011-07-01
The recommended target dose in radioiodine therapy of solitary hyperfunctioning thyroid nodules is 300-400 Gy and therefore higher than in other radiotherapies. This is due to the fact that an unknown, yet significant portion of the activity is stored in extranodular areas but is neglected in the calculatory dosimetry. We investigate the feasibility of determining the ratio of nodular and extranodular activity concentrations (uptakes) from post-therapeutically acquired planar scintigrams with Monte Carlo simulations in GATE. The geometry of a gamma camera with a high energy collimator was emulated in GATE (Version 5). A geometrical thyroid-neck phantom (GP) and the ICRP reference voxel phantoms 'Adult Female' (AF, 16 ml thyroid) and 'Adult Male' (AM, 19 ml thyroid) were used as source regions. Nodules of 1 ml and 3 ml volume were placed in the phantoms. For each phantom and each nodule 200 scintigraphic acquisitions were simulated. Uptake ratios of nodule and rest of thyroid ranging from 1 to 20 could be created by summation. Quantitative image analysis was performed by investigating the number of simulated counts in regions of interest (ROIs). ROIs were created by perpendicular projection of the phantom onto the camera plane to avoid a user dependant bias. The ratio of count densities in ROIs over the nodule and over the contralateral lobe, which should be least affected by nodular activity, was taken to be the best available measure for the uptake ratios. However, the predefined uptake ratios are underestimated by these count density ratios: For an uptake ratio of 20 the count ratios range from 4.5 (AF, 1 ml nodule) to 15.3 (AM, 3 ml nodule). Furthermore, the contralateral ROI is more strongly affected by nodular activity than expected: For an uptake ratio of 20 between nodule and rest of thyroid up to 29% of total counts in the ROI over the contralateral lobe are caused by decays in the nodule (AF 3 ml). In the case of the 1 ml nodules this
International Nuclear Information System (INIS)
Radiation dose mapping of the Spent Fuel Storage Building (SFSB) housing large number of Pressurised Heavy Water Reactor spent fuel bundles is carried out using MCNP (Monte Carlo N-Particle) code. The general methodology followed in this study involves postulation of severe accident sequences and their evaluation for on-site consequences. These accidents are extremely improbable accidents that might occur due to a combination of human error, violation of operating procedure, and failure of maintenance procedure. Probabilities were not calculated in this exercise. Two severe accident scenarios are postulated for this analysis: 1. Accidental falling of the fuel transfer cask containing irradiated fuel bundles, and 2. Loss of water cover to the spent fuels in the storage pool. Since the analysis involves for the most part shielding of large number of sources and their complex geometries, an elaborate Monte Carlo model of the SFSB has been generated. Sources in the fuel rods are estimated using the inventory code, ORIGEN2, and the dose rates at specified location of interest are calculated using MCNP code. The maximum dose rates are found to be 0.43 Gy/hr for Accident-1 (with ten-bundle fall), and 0.0254 Gy/h (with one-bundle fall). The maximum dose rate for Accident-2 is 2.104 Gy/h near the edge of the pool. (author)
Energy Technology Data Exchange (ETDEWEB)
Gargett, Maegan, E-mail: mg406@uowmail.edu.au; Rosenfeld, Anatoly [Centre for Medical Radiation Physics, University of Wollongong, NSW 2522 (Australia); Oborn, Brad [Illawarra Cancer Care Centre, Wollongong Hospital, NSW 2500, Australia and Centre for Medical Radiation Physics, University of Wollongong, NSW 2522 (Australia); Metcalfe, Peter [Centre for Medical Radiation Physics, University of Wollongong, NSW 2522, Australia and Liverpool Cancer Therapy Centre and Ingham Institute, NSW 2170 (Australia)
2015-02-15
Purpose: MRI-guided radiation therapy systems (MRIgRT) are being developed to improve online imaging during treatment delivery. At present, the operation of single point dosimeters and an ionization chamber array have been characterized in such systems. This work investigates a novel 2D diode array, named “magic plate,” for both single point calibration and 2D positional performance, the latter being a key element of modern radiotherapy techniques that will be delivered by these systems. Methods: GEANT4 Monte Carlo methods have been employed to study the dose response of a silicon diode array to 6 MV photon beams, in the presence of in-line and perpendicularly aligned uniform magnetic fields. The array consists of 121 silicon diodes (dimensions 1.5 × 1.5 × 0.38 mm{sup 3}) embedded in kapton substrate with 1 cm pitch, spanning a 10 × 10 cm{sup 2} area in total. A geometrically identical, water equivalent volume was simulated concurrently for comparison. The dose response of the silicon diode array was assessed for various photon beam field shapes and sizes, including an IMRT field, at 1 T. The dose response was further investigated at larger magnetic field strengths (1.5 and 3 T) for a 4 × 4 cm{sup 2} photon field size. Results: The magic plate diode array shows excellent correspondence (< ± 1%) to water dose in the in-line orientation, for all beam arrangements and magnetic field strengths investigated. The perpendicular orientation, however, exhibits a dose shift with respect to water at the high-dose-gradient beam edge of jaw-defined fields [maximum (4.3 ± 0.8)% over-response, maximum (1.8 ± 0.8)% under-response on opposing side for 1 T, uncertainty 1σ]. The trend is not evident in areas with in-field dose gradients typical of IMRT dose maps. Conclusions: A novel 121 pixel silicon diode array detector has been characterized by Monte Carlo simulation for its performance inside magnetic fields representative of current prototype and proposed MRI
Flampouri, Stella; Jiang, Steve B.; Sharp, Greg C.; Wolfgang, John; Patel, Abhijit A.; Choi, Noah C.
2006-06-01
The purpose of this study is to accurately estimate the difference between the planned and the delivered dose due to respiratory motion and free breathing helical CT artefacts for lung IMRT treatments, and to estimate the impact of this difference on clinical outcome. Six patients with representative tumour motion, size and position were selected for this retrospective study. For each patient, we had acquired both a free breathing helical CT and a ten-phase 4D-CT scan. A commercial treatment planning system was used to create four IMRT plans for each patient. The first two plans were based on the GTV as contoured on the free breathing helical CT set, with a GTV to PTV expansion of 1.5 cm and 2.0 cm, respectively. The third plan was based on the ITV, a composite volume formed by the union of the CTV volumes contoured on free breathing helical CT, end-of-inhale (EOI) and end-of-exhale (EOE) 4D-CT. The fourth plan was based on GTV contoured on the EOE 4D-CT. The prescribed dose was 60 Gy for all four plans. Fluence maps and beam setup parameters of the IMRT plans were used by the Monte Carlo dose calculation engine MCSIM for absolute dose calculation on both the free breathing CT and 4D-CT data. CT deformable registration between the breathing phases was performed to estimate the motion trajectory for both the tumour and healthy tissue. Then, a composite dose distribution over the whole breathing cycle was calculated as a final estimate of the delivered dose. EUD values were computed on the basis of the composite dose for all four plans. For the patient with the largest motion effect, the difference in the EUD of CTV between the planed and the delivered doses was 33, 11, 1 and 0 Gy for the first, second, third and fourth plan, respectively. The number of breathing phases required for accurate dose prediction was also investigated. With the advent of 4D-CT, deformable registration and Monte Carlo simulations, it is feasible to perform an accurate calculation of the
Energy Technology Data Exchange (ETDEWEB)
Flampouri, Stella; Jiang, Steve B; Sharp, Greg C; Wolfgang, John; Patel, Abhijit A; Choi, Noah C [Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, MA 02114 (United States)
2006-06-07
The purpose of this study is to accurately estimate the difference between the planned and the delivered dose due to respiratory motion and free breathing helical CT artefacts for lung IMRT treatments, and to estimate the impact of this difference on clinical outcome. Six patients with representative tumour motion, size and position were selected for this retrospective study. For each patient, we had acquired both a free breathing helical CT and a ten-phase 4D-CT scan. A commercial treatment planning system was used to create four IMRT plans for each patient. The first two plans were based on the GTV as contoured on the free breathing helical CT set, with a GTV to PTV expansion of 1.5 cm and 2.0 cm, respectively. The third plan was based on the ITV, a composite volume formed by the union of the CTV volumes contoured on free breathing helical CT, end-of-inhale (EOI) and end-of-exhale (EOE) 4D-CT. The fourth plan was based on GTV contoured on the EOE 4D-CT. The prescribed dose was 60 Gy for all four plans. Fluence maps and beam setup parameters of the IMRT plans were used by the Monte Carlo dose calculation engine MCSIM for absolute dose calculation on both the free breathing CT and 4D-CT data. CT deformable registration between the breathing phases was performed to estimate the motion trajectory for both the tumour and healthy tissue. Then, a composite dose distribution over the whole breathing cycle was calculated as a final estimate of the delivered dose. EUD values were computed on the basis of the composite dose for all four plans. For the patient with the largest motion effect, the difference in the EUD of CTV between the planed and the delivered doses was 33, 11, 1 and 0 Gy for the first, second, third and fourth plan, respectively. The number of breathing phases required for accurate dose prediction was also investigated. With the advent of 4D-CT, deformable registration and Monte Carlo simulations, it is feasible to perform an accurate calculation of the
International Nuclear Information System (INIS)
The purpose of this study is to accurately estimate the difference between the planned and the delivered dose due to respiratory motion and free breathing helical CT artefacts for lung IMRT treatments, and to estimate the impact of this difference on clinical outcome. Six patients with representative tumour motion, size and position were selected for this retrospective study. For each patient, we had acquired both a free breathing helical CT and a ten-phase 4D-CT scan. A commercial treatment planning system was used to create four IMRT plans for each patient. The first two plans were based on the GTV as contoured on the free breathing helical CT set, with a GTV to PTV expansion of 1.5 cm and 2.0 cm, respectively. The third plan was based on the ITV, a composite volume formed by the union of the CTV volumes contoured on free breathing helical CT, end-of-inhale (EOI) and end-of-exhale (EOE) 4D-CT. The fourth plan was based on GTV contoured on the EOE 4D-CT. The prescribed dose was 60 Gy for all four plans. Fluence maps and beam setup parameters of the IMRT plans were used by the Monte Carlo dose calculation engine MCSIM for absolute dose calculation on both the free breathing CT and 4D-CT data. CT deformable registration between the breathing phases was performed to estimate the motion trajectory for both the tumour and healthy tissue. Then, a composite dose distribution over the whole breathing cycle was calculated as a final estimate of the delivered dose. EUD values were computed on the basis of the composite dose for all four plans. For the patient with the largest motion effect, the difference in the EUD of CTV between the planed and the delivered doses was 33, 11, 1 and 0 Gy for the first, second, third and fourth plan, respectively. The number of breathing phases required for accurate dose prediction was also investigated. With the advent of 4D-CT, deformable registration and Monte Carlo simulations, it is feasible to perform an accurate calculation of the
SU-E-T-29: A Web Application for GPU-Based Monte Carlo IMRT/VMAT QA with Delivered Dose Verification
Energy Technology Data Exchange (ETDEWEB)
Folkerts, M [The University of Texas Southwestern Medical Ctr, Dallas, TX (United States); University of California, San Diego, La Jolla, CA (United States); Graves, Y [University of California, San Diego, La Jolla, CA (United States); Tian, Z; Gu, X; Jia, X; Jiang, S [The University of Texas Southwestern Medical Ctr, Dallas, TX (United States)
2014-06-01
Purpose: To enable an existing web application for GPU-based Monte Carlo (MC) 3D dosimetry quality assurance (QA) to compute “delivered dose” from linac logfile data. Methods: We added significant features to an IMRT/VMAT QA web application which is based on existing technologies (HTML5, Python, and Django). This tool interfaces with python, c-code libraries, and command line-based GPU applications to perform a MC-based IMRT/VMAT QA. The web app automates many complicated aspects of interfacing clinical DICOM and logfile data with cutting-edge GPU software to run a MC dose calculation. The resultant web app is powerful, easy to use, and is able to re-compute both plan dose (from DICOM data) and delivered dose (from logfile data). Both dynalog and trajectorylog file formats are supported. Users upload zipped DICOM RP, CT, and RD data and set the expected statistic uncertainty for the MC dose calculation. A 3D gamma index map, 3D dose distribution, gamma histogram, dosimetric statistics, and DVH curves are displayed to the user. Additional the user may upload the delivery logfile data from the linac to compute a 'delivered dose' calculation and corresponding gamma tests. A comprehensive PDF QA report summarizing the results can also be downloaded. Results: We successfully improved a web app for a GPU-based QA tool that consists of logfile parcing, fluence map generation, CT image processing, GPU based MC dose calculation, gamma index calculation, and DVH calculation. The result is an IMRT and VMAT QA tool that conducts an independent dose calculation for a given treatment plan and delivery log file. The system takes both DICOM data and logfile data to compute plan dose and delivered dose respectively. Conclusion: We sucessfully improved a GPU-based MC QA tool to allow for logfile dose calculation. The high efficiency and accessibility will greatly facilitate IMRT and VMAT QA.
SU-E-T-29: A Web Application for GPU-Based Monte Carlo IMRT/VMAT QA with Delivered Dose Verification
International Nuclear Information System (INIS)
Purpose: To enable an existing web application for GPU-based Monte Carlo (MC) 3D dosimetry quality assurance (QA) to compute “delivered dose” from linac logfile data. Methods: We added significant features to an IMRT/VMAT QA web application which is based on existing technologies (HTML5, Python, and Django). This tool interfaces with python, c-code libraries, and command line-based GPU applications to perform a MC-based IMRT/VMAT QA. The web app automates many complicated aspects of interfacing clinical DICOM and logfile data with cutting-edge GPU software to run a MC dose calculation. The resultant web app is powerful, easy to use, and is able to re-compute both plan dose (from DICOM data) and delivered dose (from logfile data). Both dynalog and trajectorylog file formats are supported. Users upload zipped DICOM RP, CT, and RD data and set the expected statistic uncertainty for the MC dose calculation. A 3D gamma index map, 3D dose distribution, gamma histogram, dosimetric statistics, and DVH curves are displayed to the user. Additional the user may upload the delivery logfile data from the linac to compute a 'delivered dose' calculation and corresponding gamma tests. A comprehensive PDF QA report summarizing the results can also be downloaded. Results: We successfully improved a web app for a GPU-based QA tool that consists of logfile parcing, fluence map generation, CT image processing, GPU based MC dose calculation, gamma index calculation, and DVH calculation. The result is an IMRT and VMAT QA tool that conducts an independent dose calculation for a given treatment plan and delivery log file. The system takes both DICOM data and logfile data to compute plan dose and delivered dose respectively. Conclusion: We sucessfully improved a GPU-based MC QA tool to allow for logfile dose calculation. The high efficiency and accessibility will greatly facilitate IMRT and VMAT QA
Tian, Zhen; Li, Yongbao; Shi, Feng; Jiang, Steve B; Jia, Xun
2015-01-01
We recently built an analytical source model for GPU-based MC dose engine. In this paper, we present a sampling strategy to efficiently utilize this source model in GPU-based dose calculation. Our source model was based on a concept of phase-space-ring (PSR). This ring structure makes it effective to account for beam rotational symmetry, but not suitable for dose calculations due to rectangular jaw settings. Hence, we first convert PSR source model to its phase-space let (PSL) representation. Then in dose calculation, different types of sub-sources were separately sampled. Source sampling and particle transport were iterated. So that the particles being sampled and transported simultaneously are of same type and close in energy to alleviate GPU thread divergence. We also present an automatic commissioning approach to adjust the model for a good representation of a clinical linear accelerator . Weighting factors were introduced to adjust relative weights of PSRs, determined by solving a quadratic minimization ...
Hashikin, N. A. A.; Yeong, C. H.; Guatelli, S.; Abdullah, B. J. J.; Ng, K. H.; Malaroda, A.; Rosenfeld, A. B.; Perkins, A. C.
2016-03-01
90Y-radioembolization is a palliative treatment for liver cancer. 90Y decays via beta emission, making imaging difficult due to absence of gamma radiation. Since post-procedure imaging is crucial, several theranostic radionuclides have been explored as alternatives. However, exposures to gamma radiation throughout the treatment caused concern for the organs near the liver. Geant4 Monte Carlo simulation using MIRD Pamphlet 5 reference phantom was carried out. A spherical tumour with 4.3cm radius was modelled within the liver. 1.82GBq of 90Y sources were isotropically distributed within the tumour, with no extrahepatic shunting. The simulation was repeated with 153Sm, 166Ho and 177Lu. The estimated tumour doses for all radionuclides were 262.9Gy. Tumour dose equivalent to 1.82GBq 90Y can be achieved with 8.32, 5.83, and 4.44GBq for 153Sm, 166Ho and 177Lu, respectively. Normal liver doses by the other radionuclides were lower than 90Y, hence beneficial for normal tissue sparing. The organ doses from 153Sm and 177Lu were relatively higher due to higher gamma energy, but were still well below 1Gy. 166Ho, 177Lu and 153Sm offer useful gamma emission for post-procedure imaging. They show potential as 90Y substitutes, delivering comparable tumour doses, lower normal liver doses and other organs doses far below the tolerance limit.
International Nuclear Information System (INIS)
Calculations of normal tissue complication probability (NTCP) values for the rectum are difficult because it is a hollow, non-rigid, organ. Finding the true cumulative dose distribution for a number of treatment fractions requires a CT scan before each treatment fraction. This is labour intensive, and several surrogate distributions have therefore been suggested, such as dose wall histograms, dose surface histograms and histograms for the solid rectum, with and without margins. In this study, a Monte Carlo method is used to investigate the relationships between the cumulative dose distributions based on all treatment fractions and the above-mentioned histograms that are based on one CT scan only, in terms of equivalent uniform dose. Furthermore, the effect of a specific choice of histogram on estimates of the volume parameter of the probit NTCP model was investigated. It was found that the solid rectum and the rectum wall histograms (without margins) gave equivalent uniform doses with an expected value close to the values calculated from the cumulative dose distributions in the rectum wall. With the number of patients available in this study the standard deviations of the estimates of the volume parameter were large, and it was not possible to decide which volume gave the best estimates of the volume parameter, but there were distinct differences in the mean values of the values obtained
Energy Technology Data Exchange (ETDEWEB)
Schwarcke, Marcelo; Marques, Tatiana; Nicolucci, Patricia; Baffa, Oswaldo, E-mail: mschwarcke@usp.b [Universidade de Sao Paulo (FFCLRP/USP), Ribeirao Preto, SP (Brazil). Faculdade de Filosofia, Ciencias e Letras. Dept. de Fisica e Matematica; Bornemann, Clarissa [Hospital de Caridade Astrogildo de Azevedo, Santa Maria, RS (Brazil). Servico de Medicina Nuclear de Santa Maria
2010-06-15
Patients with Graves disease have a high hormonal disorder, which causes behavioral changes. One way to treat this disease is the use of high doses of {sup 131} Iodine, requiring that the patient carries out the examination of {sup 131}I uptake to estimate the activity to be administered. Using these data capture and compared with the simulated data using the Monte Carlo code PENELOPE is possible to determine a distribution of dose to the region surrounding the thyroid. As noted the difference between the simulated values and the experimentally obtained were 10.36%, thus showing the code of simulation for accurate determination of absorbed dose in tissue near the thyroid. (author)
International Nuclear Information System (INIS)
The purpose of this study was to examine dose distribution of a skull base tumor and surrounding critical structures in response to high dose intensity-modulated radiosurgery (IMRS) with Monte Carlo (MC) simulation using a dual resolution sandwich phantom. The measurement-based Monte Carlo (MBMC) method (Lin et al., 2009) was adopted for the study. The major components of the MBMC technique involve (1) the BEAMnrc code for beam transport through the treatment head of a Varian 21EX linear accelerator, (2) the DOSXYZnrc code for patient dose simulation and (3) an EPID-measured efficiency map which describes non-uniform fluence distribution of the IMRS treatment beam. For the simulated case, five isocentric 6 MV photon beams were designed to deliver a total dose of 1200 cGy in two fractions to the skull base tumor. A sandwich phantom for the MBMC simulation was created based on the patient's CT scan of a skull base tumor [gross tumor volume (GTV)=8.4 cm3] near the right 8th cranial nerve. The phantom, consisted of a 1.2-cm thick skull base region, had a voxel resolution of 0.05×0.05×0.1 cm3 and was sandwiched in between 0.05×0.05×0.3 cm3 slices of a head phantom. A coarser 0.2×0.2×0.3 cm3 single resolution (SR) phantom was also created for comparison with the sandwich phantom. A particle history of 3×108 for each beam was used for simulations of both the SR and the sandwich phantoms to achieve a statistical uncertainty of <2%. Our study showed that the planning target volume (PTV) receiving at least 95% of the prescribed dose (VPTV95) was 96.9%, 96.7% and 99.9% for the TPS, SR, and sandwich phantom, respectively. The maximum and mean doses to large organs such as the PTV, brain stem, and parotid gland for the TPS, SR and sandwich MC simulations did not show any significant difference; however, significant dose differences were observed for very small structures like the right 8th cranial nerve, right cochlea, right malleus and right semicircular canal
Kernel Based Regularization Netowrks
Czech Academy of Sciences Publication Activity Database
Kudová, Petra
Praha: MATFYZPRESS, 2005 - (Hakl, F.), s. 65-74 ISBN 80-86732-56-8. [Doktorandský den '05. Nový Dvůr (CZ), 05.10.2005-07.10.2005] R&D Projects: GA ČR GA201/05/0557 Institutional research plan: CEZ:AV0Z10300504 Keywords : learning from examples * regularization networks * kernel methods * product kernels * sum kernels Subject RIV: BA - General Mathematics
Almansa, J F; Anguiano, M; Guerrero, R; Lallena, A M; Al-Dweri, Feras M.O.; Almansa, Julio F.; Guerrero, Rafael
2006-01-01
Monte Carlo calculations using the codes PENELOPE and GEANT4 have been performed to characterize the dosimetric properties of monoenergetic photon point sources in water. The dose rate in water has been calculated for energies of interest in brachytherapy, ranging between 10 keV and 2 MeV. A comparison of the results obtained using the two codes with the available data calculated with other Monte Carlo codes is carried out. A chi2-like statistical test is proposed for these comparisons. PENELOPE and GEANT4 show a reasonable agreement for all energies analyzed and distances to the source larger than 1 cm. Significant differences are found at distances from the source up to 1 cm. A similar situation occurs between PENELOPE and EGS4.
International Nuclear Information System (INIS)
Purpose: It is well known that pacemaker implantation is sometimes needed in pregnant patients with symptomatic bradycardia. To our knowledge, there is no reported experience regarding radiation doses to the unborn child resulting from fluoroscopy during pacemaker implantation. The purpose of the current study was to develop a method for estimating embryo/fetus dose from fluoroscopically guided pacemaker implantation procedures performed on pregnant patients during all trimesters of gestation. Methods: The Monte Carlo N-Particle (MCNP) radiation transport code was employed in this study. Three mathematical anthropomorphic phantoms representing the average pregnant patient at the first, second and third trimesters of gestation were generated using Bodybuilder software (White Rock science, White Rock, NM). The normalized embryo/fetus dose from the posteroanterior (PA), the 30° left-anterior oblique (LAO) and the 30° right-anterior oblique (RAO) projections were calculated for a wide range of kVp (50–120 kVp) and total filtration values (2.5–9.0 mm Al). Results: The results consist of radiation doses normalized to a) entrance skin dose (ESD) and b) dose area product (DAP) so that the dose to the unborn child from any fluoroscopic technique and x-ray device used can be calculated. ESD normalized doses ranged from 0.008 (PA, first trimester) to 2.519 μGy/mGy (RAO, third trimester). DAP normalized doses ranged from 0.051 (PA, first trimester) to 12.852 μGy/Gycm2 (RAO, third trimester). Conclusion: Embryo/fetus doses from fluoroscopically guided pacemaker implantation procedures performed on pregnant patients during all stages of gestation can be estimated using the method developed in this study. This study was supported by the Greek Ministry of Education and Religious Affairs, General Secretariat for Research and Technology, Operational Program ‘Education and Lifelong Learning’, ARISTIA (Research project: CONCERT)
Energy Technology Data Exchange (ETDEWEB)
Damilakis, J; Stratakis, J; Solomou, G [University of Crete, Heraklion (Greece)
2014-06-01
Purpose: It is well known that pacemaker implantation is sometimes needed in pregnant patients with symptomatic bradycardia. To our knowledge, there is no reported experience regarding radiation doses to the unborn child resulting from fluoroscopy during pacemaker implantation. The purpose of the current study was to develop a method for estimating embryo/fetus dose from fluoroscopically guided pacemaker implantation procedures performed on pregnant patients during all trimesters of gestation. Methods: The Monte Carlo N-Particle (MCNP) radiation transport code was employed in this study. Three mathematical anthropomorphic phantoms representing the average pregnant patient at the first, second and third trimesters of gestation were generated using Bodybuilder software (White Rock science, White Rock, NM). The normalized embryo/fetus dose from the posteroanterior (PA), the 30° left-anterior oblique (LAO) and the 30° right-anterior oblique (RAO) projections were calculated for a wide range of kVp (50–120 kVp) and total filtration values (2.5–9.0 mm Al). Results: The results consist of radiation doses normalized to a) entrance skin dose (ESD) and b) dose area product (DAP) so that the dose to the unborn child from any fluoroscopic technique and x-ray device used can be calculated. ESD normalized doses ranged from 0.008 (PA, first trimester) to 2.519 μGy/mGy (RAO, third trimester). DAP normalized doses ranged from 0.051 (PA, first trimester) to 12.852 μGy/Gycm2 (RAO, third trimester). Conclusion: Embryo/fetus doses from fluoroscopically guided pacemaker implantation procedures performed on pregnant patients during all stages of gestation can be estimated using the method developed in this study. This study was supported by the Greek Ministry of Education and Religious Affairs, General Secretariat for Research and Technology, Operational Program ‘Education and Lifelong Learning’, ARISTIA (Research project: CONCERT)
Energy Technology Data Exchange (ETDEWEB)
Jung, J; Pelletier, C [East Carolina University, Greenville, NC (United States); Lee, C [University of Michigan, Ann Arbor, MI (United States); Kim, J [University of Pittsburgh Medical Center, Pittsburgh, PA (United States); Pyakuryal, A; Lee, C [National Cancer Institute, Rockville, MD (United States)
2015-06-15
Purpose: Organ doses for the Hodgkin’s lymphoma patients treated with cobalt-60 radiation were estimated using an anthropomorphic model and Monte Carlo modeling. Methods: A cobalt-60 treatment unit modeled in the BEAMnrc Monte Carlo code was used to produce phase space data. The Monte Carlo simulation was verified with percent depth dose measurement in water at various field sizes. Radiation transport through the lung blocks were modeled by adjusting the weights of phase space data. We imported a precontoured adult female hybrid model and generated a treatment plan. The adjusted phase space data and the human model were imported to the XVMC Monte Carlo code for dose calculation. The organ mean doses were estimated and dose volume histograms were plotted. Results: The percent depth dose agreement between measurement and calculation in water phantom was within 2% for all field sizes. The mean organ doses of heart, left breast, right breast, and spleen for the selected case were 44.3, 24.1, 14.6 and 3.4 Gy, respectively with the midline prescription dose of 40.0 Gy. Conclusion: Organ doses were estimated for the patient group whose threedimensional images are not available. This development may open the door to more accurate dose reconstruction and estimates of uncertainties in secondary cancer risk for Hodgkin’s lymphoma patients. This work was partially supported by the intramural research program of the National Institutes of Health, National Cancer Institute, Division of Cancer Epidemiology and Genetics.
International Nuclear Information System (INIS)
Purpose: Organ doses for the Hodgkin’s lymphoma patients treated with cobalt-60 radiation were estimated using an anthropomorphic model and Monte Carlo modeling. Methods: A cobalt-60 treatment unit modeled in the BEAMnrc Monte Carlo code was used to produce phase space data. The Monte Carlo simulation was verified with percent depth dose measurement in water at various field sizes. Radiation transport through the lung blocks were modeled by adjusting the weights of phase space data. We imported a precontoured adult female hybrid model and generated a treatment plan. The adjusted phase space data and the human model were imported to the XVMC Monte Carlo code for dose calculation. The organ mean doses were estimated and dose volume histograms were plotted. Results: The percent depth dose agreement between measurement and calculation in water phantom was within 2% for all field sizes. The mean organ doses of heart, left breast, right breast, and spleen for the selected case were 44.3, 24.1, 14.6 and 3.4 Gy, respectively with the midline prescription dose of 40.0 Gy. Conclusion: Organ doses were estimated for the patient group whose threedimensional images are not available. This development may open the door to more accurate dose reconstruction and estimates of uncertainties in secondary cancer risk for Hodgkin’s lymphoma patients. This work was partially supported by the intramural research program of the National Institutes of Health, National Cancer Institute, Division of Cancer Epidemiology and Genetics
International Nuclear Information System (INIS)
Purpose: The motivation of this study was to find and eliminate the cause of errors in dose-averaged linear energy transfer (LET) calculations from therapeutic protons in small targets, such as biological cell layers, calculated using the GEANT 4 Monte Carlo code. Furthermore, the purpose was also to provide a recommendation to select an appropriate LET quantity from GEANT 4 simulations to correlate with biological effectiveness of therapeutic protons. Methods: The authors developed a particle tracking step based strategy to calculate the average LET quantities (track-averaged LET, LETt and dose-averaged LET, LETd) using GEANT 4 for different tracking step size limits. A step size limit refers to the maximally allowable tracking step length. The authors investigated how the tracking step size limit influenced the calculated LETt and LETd of protons with six different step limits ranging from 1 to 500 μm in a water phantom irradiated by a 79.7-MeV clinical proton beam. In addition, the authors analyzed the detailed stochastic energy deposition information including fluence spectra and dose spectra of the energy-deposition-per-step of protons. As a reference, the authors also calculated the averaged LET and analyzed the LET spectra combining the Monte Carlo method and the deterministic method. Relative biological effectiveness (RBE) calculations were performed to illustrate the impact of different LET calculation methods on the RBE-weighted dose. Results: Simulation results showed that the step limit effect was small for LETt but significant for LETd. This resulted from differences in the energy-deposition-per-step between the fluence spectra and dose spectra at different depths in the phantom. Using the Monte Carlo particle tracking method in GEANT 4 can result in incorrect LETd calculation results in the dose plateau region for small step limits. The erroneous LETd results can be attributed to the algorithm to determine fluctuations in energy deposition along the
International Nuclear Information System (INIS)
Purpose: To measure radiation dose in a water-equivalent medium from very high-energy electron (VHEE) beams and make comparisons to Monte Carlo (MC) simulation results. Methods: Dose in a polystyrene phantom delivered by an experimental VHEE beam line was measured with Gafchromic films for three 50 MeV and two 70 MeV Gaussian beams of 4.0–6.9 mm FWHM and compared to corresponding MC-simulated dose distributions. MC dose in the polystyrene phantom was calculated with the EGSnrc/BEAMnrc and DOSXYZnrc codes based on the experimental setup. Additionally, the effect of 2% beam energy measurement uncertainty and possible non-zero beam angular spread on MC dose distributions was evaluated. Results: MC simulated percentage depth dose (PDD) curves agreed with measurements within 4% for all beam sizes at both 50 and 70 MeV VHEE beams. Central axis PDD at 8 cm depth ranged from 14% to 19% for the 5.4–6.9 mm 50 MeV beams and it ranged from 14% to 18% for the 4.0–4.5 mm 70 MeV beams. MC simulated relative beam profiles of regularly shaped Gaussian beams evaluated at depths of 0.64 to 7.46 cm agreed with measurements to within 5%. A 2% beam energy uncertainty and 0.286° beam angular spread corresponded to a maximum 3.0% and 3.8% difference in depth dose curves of the 50 and 70 MeV electron beams, respectively. Absolute dose differences between MC simulations and film measurements of regularly shaped Gaussian beams were between 10% and 42%. Conclusions: The authors demonstrate that relative dose distributions for VHEE beams of 50–70 MeV can be measured with Gafchromic films and modeled with Monte Carlo simulations to an accuracy of 5%. The reported absolute dose differences likely caused by imperfect beam steering and subsequent charge loss revealed the importance of accurate VHEE beam control and diagnostics
Energy Technology Data Exchange (ETDEWEB)
Bazalova-Carter, Magdalena; Liu, Michael; Palma, Bianey; Koong, Albert C.; Maxim, Peter G., E-mail: Peter.Maxim@Stanford.edu, E-mail: BWLoo@Stanford.edu; Loo, Billy W., E-mail: Peter.Maxim@Stanford.edu, E-mail: BWLoo@Stanford.edu [Department of Radiation Oncology, Stanford University, Stanford, California 94305-5847 (United States); Dunning, Michael; McCormick, Doug; Hemsing, Erik; Nelson, Janice; Jobe, Keith; Colby, Eric; Tantawi, Sami; Dolgashev, Valery [SLAC National Accelerator Laboratory, Menlo Park, California 94025 (United States)
2015-04-15
Purpose: To measure radiation dose in a water-equivalent medium from very high-energy electron (VHEE) beams and make comparisons to Monte Carlo (MC) simulation results. Methods: Dose in a polystyrene phantom delivered by an experimental VHEE beam line was measured with Gafchromic films for three 50 MeV and two 70 MeV Gaussian beams of 4.0–6.9 mm FWHM and compared to corresponding MC-simulated dose distributions. MC dose in the polystyrene phantom was calculated with the EGSnrc/BEAMnrc and DOSXYZnrc codes based on the experimental setup. Additionally, the effect of 2% beam energy measurement uncertainty and possible non-zero beam angular spread on MC dose distributions was evaluated. Results: MC simulated percentage depth dose (PDD) curves agreed with measurements within 4% for all beam sizes at both 50 and 70 MeV VHEE beams. Central axis PDD at 8 cm depth ranged from 14% to 19% for the 5.4–6.9 mm 50 MeV beams and it ranged from 14% to 18% for the 4.0–4.5 mm 70 MeV beams. MC simulated relative beam profiles of regularly shaped Gaussian beams evaluated at depths of 0.64 to 7.46 cm agreed with measurements to within 5%. A 2% beam energy uncertainty and 0.286° beam angular spread corresponded to a maximum 3.0% and 3.8% difference in depth dose curves of the 50 and 70 MeV electron beams, respectively. Absolute dose differences between MC simulations and film measurements of regularly shaped Gaussian beams were between 10% and 42%. Conclusions: The authors demonstrate that relative dose distributions for VHEE beams of 50–70 MeV can be measured with Gafchromic films and modeled with Monte Carlo simulations to an accuracy of 5%. The reported absolute dose differences likely caused by imperfect beam steering and subsequent charge loss revealed the importance of accurate VHEE beam control and diagnostics.
Energy Technology Data Exchange (ETDEWEB)
Gomes B, W. O., E-mail: wilsonottobatista@gmail.com [Instituto Federal da Bahia, Rua Emidio dos Santos s/n, Bardalho, 40301-015 Salvador, Bahia (Brazil)
2015-10-15
Full text: In this study irradiation geometry applicable to PCXMC and the consequent calculation of effective dose in applications of cone beam computed tomography (CBCT) was developed. Two different CBCT equipment s for dental applications were evaluated: Care Stream Cs-9000 3-Dimensional and Gendex GXCB-500 tomographs. Each protocol initially was characterized by measuring the surface kerma input and the product air kerma-area, P{sub KA}. Then, technical parameters of each of the predetermined protocols and geometric conditions in the PCXMC software were introduced to obtain the values of effective dose. The calculated effective dose is within the range of 9.0 to 15.7 μSv for Cs 9000 3-D and in the range 44.5 to 89 mSv for GXCB-500 equipment. These values were compared with dosimetric results obtained using thermoluminescent dosimeters implanted in anthropomorphic mannequin and were considered consistent. The effective dose results are very sensitive to the radiation geometry (beam position); this represents a factor of fragility software usage, but on the other hand, turns out to be a very useful tool for quick conclusions regarding the optimization process of protocols. We can conclude that the use of Monte Carlo simulation software PCXMC is useful in the evaluation of test protocols of CBCT in dental applications. (Author)
Energy Technology Data Exchange (ETDEWEB)
Kvinnsland, Y.; Skretting, A.; Bruland, Oe.S. [Department of Nuclear Medicine, The Norwegian Radium Hospital, 0310 Oslo (Norway)
2001-04-01
The purpose of the present work was to investigate how haematopoietic stem cell survival is affected by the differences in the dose distribution that arise from different radionuclides contained in bone-seeking radiopharmaceuticals. This was carried out in three steps: (a) calculations of representative dose distributions in individual bone marrow cavities that are irradiated by sources of {sup 89}Sr, {sup 186}Re, {sup 117}mSn or {sup 153}Sm, uniformly distributed on the bone surfaces; (b) assessment of the corresponding haematopoietic stem cell survival and (c) a comparison of these results with results obtained using the assumption of a uniform dose distribution. Two different idealized models of the geometry of trabecular bone were formulated, each consisting of an infinite array of identical elements. Monte Carlo simulations were used to generate dose-volume histograms that were used to assess haematopoietic stem cell survival with two different assumptions about spatial cell distributions. Compared with a homogeneous dose distribution, the estimated cell survival was markedly higher for {sup 117}mSn and {sup 153}Sm, and only slightly different for {sup 89}Sr and {sup 186}Re. The quantitative results differed between the two geometric models and the assumptions about spatial cell distribution, but the trends were the same. The results imply that it is necessary to include dose distributions for individual bone marrow cavities in considerations concerning bone marrow toxicity. (author)
International Nuclear Information System (INIS)
Background and purpose: To recompute clinical intensity-modulated treatment plans for ethmoid sinus cancer and to compare quantitatively the dose-volume histograms (DVHs) of the planning target volume (PTV) and the optic organs at risk. Material and methods: Ten step-and-shoot intensity-modulated treatment plans were enrolled in this study. Large natural and surgical air cavities challenged the calculation systems. Each optimized treatment plan was recalculated by two superposition convolution (TMS and Pinnacle) and a Monte Carlo system (MCDE). To compare the resulting DVHs, a one-way ANOVA for repeated measurements was performed and multiple pairwise comparisons were made. Results: The tails of the PTV-DVHs were significantly higher for the Monte Carlo system. The DVHs of the critical organs displayed some statistically but not always clinically significant differences. For the individual patients, the three planning systems sometimes reproduced clinically discrepant DVHs that were not significantly different when averaged over all patients. Conclusions: Dose to air cavities contains computational uncertainty. As this dose is clinically irrelevant and optimizing it is meaningless, we recommended extracting the air from the PTV when constructing the PTV-DVH. The planning systems considered reproduce DVHs that are significantly different, especially in the tail region of PTV-DVHs
International Nuclear Information System (INIS)
Measurement-based Monte Carlo (MBMC) simulation using a high definition (HD) phantom was used to evaluate the dose distribution in nasopharyngeal cancer (NPC) patients treated with intensity modulated radiation therapy (IMRT). Around nasopharyngeal cavity, there exists many small volume organs-at-risk (OARs) such as the optic nerves, auditory nerves, cochlea, and semicircular canal which necessitate the use of a high definition phantom for accurate and correct dose evaluation. The aim of this research was to study the advantages of using an HD phantom for MBMC simulation in NPC patients treated with IMRT. The MBMC simulation in this study was based on the IMRT treatment plan of three NPC patients generated by the anisotropic analytical algorithm (AAA) of the Eclipse treatment planning system (Varian Medical Systems, Palo Alto, CA, USA) using a calculation grid of 2 mm2. The NPC tumor was treated to a cumulative dose of 7000 cGy in 35 fractions using the shrinking-field sequential IMRT (SIMRT) method. The BEAMnrc MC Code was used to simulate a Varian EX21 linear accelerator treatment head. The HD phantom contained 0.5 × 0.5 × 1 mm3 voxels for the nasopharyngeal area and 0.5 × 0.5 × 3 mm3 for the rest of the head area. An efficiency map was obtained for the amorphous silicon aS1000 electronic portal imaging device (EPID) to adjust the weighting of each particle in the phase-space file for each IMRT beam. Our analysis revealed that small volume organs such as the eighth cranial nerve, semicircular canal, cochlea and external auditory canal showed an absolute dose difference of ≥200 cGy, while the dose difference for larger organs such as the parotid glands and tumor was negligible for the MBMC simulation using the HD phantom. The HD phantom was found to be suitable for Monte Carlo dose volume analysis of small volume organs. - Highlights: • HD dose evaluation for IMRT of NPC patients have been verified by the MC method. • MC results shows higher
Yeh, Chi-Yuan; Tung, Chuan-Jung; Chao, Tsi-Chain; Lin, Mu-Han; Lee, Chung-Chi
2014-11-01
The purpose of this study was to examine dose distribution of a skull base tumor and surrounding critical structures in response to high dose intensity-modulated radiosurgery (IMRS) with Monte Carlo (MC) simulation using a dual resolution sandwich phantom. The measurement-based Monte Carlo (MBMC) method (Lin et al., 2009) was adopted for the study. The major components of the MBMC technique involve (1) the BEAMnrc code for beam transport through the treatment head of a Varian 21EX linear accelerator, (2) the DOSXYZnrc code for patient dose simulation and (3) an EPID-measured efficiency map which describes non-uniform fluence distribution of the IMRS treatment beam. For the simulated case, five isocentric 6 MV photon beams were designed to deliver a total dose of 1200 cGy in two fractions to the skull base tumor. A sandwich phantom for the MBMC simulation was created based on the patient's CT scan of a skull base tumor [gross tumor volume (GTV)=8.4 cm3] near the right 8th cranial nerve. The phantom, consisted of a 1.2-cm thick skull base region, had a voxel resolution of 0.05×0.05×0.1 cm3 and was sandwiched in between 0.05×0.05×0.3 cm3 slices of a head phantom. A coarser 0.2×0.2×0.3 cm3 single resolution (SR) phantom was also created for comparison with the sandwich phantom. A particle history of 3×108 for each beam was used for simulations of both the SR and the sandwich phantoms to achieve a statistical uncertainty of <2%. Our study showed that the planning target volume (PTV) receiving at least 95% of the prescribed dose (VPTV95) was 96.9%, 96.7% and 99.9% for the TPS, SR, and sandwich phantom, respectively. The maximum and mean doses to large organs such as the PTV, brain stem, and parotid gland for the TPS, SR and sandwich MC simulations did not show any significant difference; however, significant dose differences were observed for very small structures like the right 8th cranial nerve, right cochlea, right malleus and right semicircular canal. Dose
Melacci, Stefano; Gori, Marco
2013-11-01
Supervised examples and prior knowledge on regions of the input space have been profitably integrated in kernel machines to improve the performance of classifiers in different real-world contexts. The proposed solutions, which rely on the unified supervision of points and sets, have been mostly based on specific optimization schemes in which, as usual, the kernel function operates on points only. In this paper, arguments from variational calculus are used to support the choice of a special class of kernels, referred to as box kernels, which emerges directly from the choice of the kernel function associated with a regularization operator. It is proven that there is no need to search for kernels to incorporate the structure deriving from the supervision of regions of the input space, because the optimal kernel arises as a consequence of the chosen regularization operator. Although most of the given results hold for sets, we focus attention on boxes, whose labeling is associated with their propositional description. Based on different assumptions, some representer theorems are given that dictate the structure of the solution in terms of box kernel expansion. Successful results are given for problems of medical diagnosis, image, and text categorization. PMID:24051728
Energy Technology Data Exchange (ETDEWEB)
Duff, I.
1994-12-31
This workshop focuses on kernels for iterative software packages. Specifically, the three speakers discuss various aspects of sparse BLAS kernels. Their topics are: `Current status of user lever sparse BLAS`; Current status of the sparse BLAS toolkit`; and `Adding matrix-matrix and matrix-matrix-matrix multiply to the sparse BLAS toolkit`.