Uncertainty analysis in the simulation of an HPGe detector using the Monte Carlo Code MCNP5
Energy Technology Data Exchange (ETDEWEB)
Gallardo, Sergio; Pozuelo, Fausto; Querol, Andrea; Verdu, Gumersindo; Rodenas, Jose, E-mail: sergalbe@upv.es [Universitat Politecnica de Valencia, Valencia, (Spain). Instituto de Seguridad Industrial, Radiofisica y Medioambiental (ISIRYM); Ortiz, J. [Universitat Politecnica de Valencia, Valencia, (Spain). Servicio de Radiaciones. Lab. de Radiactividad Ambiental; Pereira, Claubia [Universidade Federal de Minas Gerais (UFMG), Belo Horizonte, MG (Brazil). Departamento de Engenharia Nuclear
2013-07-01
A gamma spectrometer including an HPGe detector is commonly used for environmental radioactivity measurements. Many works have been focused on the simulation of the HPGe detector using Monte Carlo codes such as MCNP5. However, the simulation of this kind of detectors presents important difficulties due to the lack of information from manufacturers and due to loss of intrinsic properties in aging detectors. Some parameters such as the active volume or the Ge dead layer thickness are many times unknown and are estimated during simulations. In this work, a detailed model of an HPGe detector and a petri dish containing a certified gamma source has been done. The certified gamma source contains nuclides to cover the energy range between 50 and 1800 keV. As a result of the simulation, the Pulse Height Distribution (PHD) is obtained and the efficiency curve can be calculated from net peak areas and taking into account the certified activity of the source. In order to avoid errors due to the net area calculation, the simulated PHD is treated using the GammaVision software. On the other hand, it is proposed to use the Noether-Wilks formula to do an uncertainty analysis of model with the main goal of determining the efficiency curve of this detector and its associated uncertainty. The uncertainty analysis has been focused on dead layer thickness at different positions of the crystal. Results confirm the important role of the dead layer thickness in the low energy range of the efficiency curve. In the high energy range (from 300 to 1800 keV) the main contribution to the absolute uncertainty is due to variations in the active volume. (author)
Energy Technology Data Exchange (ETDEWEB)
Gallardo, S.; Querol, A.; Rodenas, J.; Verdu, G.
2014-07-01
in this paper we propose to perform a simulation model using the MCNP5 code and a registration form meshing to improve the simulation efficiency of the detector in the range of energies ranging from 50 to 2000 keV. This meshing is built by FMESH MCNP5 registration code that allows a mesh with cells of few microns. The photon and electron flow is calculated in the different cells of the mesh which is superimposed on detector geometry. It analyzes the variation of efficiency (related to the variation of energy deposited in the active volume). (Author)
Directory of Open Access Journals (Sweden)
Mehdi Zehtabian
2015-09-01
Full Text Available Introduction Lead-based shields are the most widely used attenuators in X-ray and gamma ray fields. The heavy weight, toxicity and corrosion of lead have led researchers towards the development of non-lead shields. Materials and Methods The purpose of this study was to design multi-layered shields for protection against X-rays and gamma rays in diagnostic radiology and nuclear medicine. In this study, cubic slabs composed of several materials with high atomic numbers, i.e., lead, barium, bismuth, gadolinium, tin and tungsten, were simulated, using MCNP5 Monte Carlo code. Cubic slabs (30×30×0.05 cm3 were simulated at a 50 cm distance from the point photon source. The X-ray spectra of 80 kVp and 120 kVp were obtained, using IPEM Report 78. The photon flux following the use of each shield was obtained inside cubic tally cells (1×1×0.5 cm3 at a 5 cm distance from the shields. The photon attenuation properties of multi-layered shields (i.e., two, three, four and five layers, composed of non-lead radiation materials, were also obtained via Monte Carlo simulations. Results Among different shield designs proposed in this study, the three-layered shield, composed of tungsten, bismuth and gadolinium, showed the most significant attenuation properties in radiology, with acceptable shielding at 140 keV energy in nuclear medicine. Conclusion According to the results, materials with k-edges equal to energies common to diagnostic radiology can be proper substitutes for lead shields.
Energy Technology Data Exchange (ETDEWEB)
Pozuelo, F.; Querol, A.; Gallardo, S.; Rodenas, J.; Verdu, G.
2012-07-01
In this case, used codes PENELOPE MCNP5, based on the Monte Carlo method for x-ray spectrum taking into account the characteristics of the x-ray tube. In order to achieve a greater fit of simulated by the theoretical spectrum. It carried out a sensitivity analysis of the parameters available in both codes. The obtaining of the simulated spectrum could lead to an improvement in quality control of the x-ray tube to incorporate it as a method complementary to techniques.
Directory of Open Access Journals (Sweden)
C.C. Arwui
2011-05-01
Full Text Available Design objectives for brachytherapy treatment facilities require sufficient shielding to reduce primary and scatter radiation to design limit in order to limit exposure to patients, staff and the general public. The primary aim of this study is to verify whether shielding of the brachytherapy unit at the Korle Bu teaching Hospital in Ghana provides adequate protection in order to assess any radiological health and safety impact and also test the suitability of other available sources. The study evaluates the effectiveness of the biological shielding design of a Cs-137 brachytherapy unit at the Korle-Bu Teaching Hospital in Ghana using MCNP5. The facility was modeled based on the design specifications for LDR Cs-137, MDR Cs-137, HDR Co-60 and HDR Ir-192 treatment modalities. The estimated dose rate ranged from (0.01-0.15 μSv/h and (0.37-3.05 μSv/h for the existing initial and decayed activities of LDR Cs-137 for the public and controlled areas respectively, (0.03-0.57 μSv/h and (1.53-8.06 μSv/h for MDR Cs-137, (7.47-59.46 μSv/h and (144.87-178.74 μSv/h for HDR Co- 60, (0.13-6.95 μSv/h and (19.47-242.98 μSv/h for HDR Ir-192 for the public and controlled areas respectively. The results were verified by dose rates measurement for the current LDR setup at the Brachytherapy unit and agreed quiet well. It was also compared with the reference values of 0.5 μSv/h for public areas and 7.5 μSv/h for controlled areas respectively. It can be concluded that the shielding is adequate for the existing source.
VVER-440 Ex-Core Neutron Transport Calculations by MCNP-5 Code and Comparison with Experiment
Energy Technology Data Exchange (ETDEWEB)
Borodkin, Pavel; Khrennikov, Nikolay [Scientific and Engineering Centre for Nuclear and Radiation Safety (SEC NRS) Malaya Krasnoselskaya ul., 2/8, bld. 5, 107140 Moscow (Russian Federation)
2008-07-01
Ex-core neutron transport calculations are needed to evaluate radiation loading parameters (neutron fluence, fluence rate and spectra) on the in-vessel equipment, reactor pressure vessel (RPV) and support constructions of VVER type reactors. Due to these parameters are used for reactor equipment life-time assessment, neutron transport calculations should be carried out by precise and reliable calculation methods. In case of RPVs, especially, of first generation VVER-440s, the neutron fluence plays a key role in the prediction of RPV lifetime. Main part of VVER ex-core neutron transport calculations are performed by deterministic and Monte-Carlo methods. This paper deals with precise calculations of the Russian first generation VVER-440 by MCNP-5 code. The purpose of this work was an application of this code for expert calculations, verification of results by comparison with deterministic calculations and validation by neutron activation measured data. Deterministic discrete ordinates DORT code, widely used for RPV neutron dosimetry and many times tested by experiments, was used for comparison analyses. Ex-vessel neutron activation measurements at the VVER-440 NPP have provided space (in azimuth and height directions) and neutron energy (different activation reactions) distributions data for experimental (E) validation of calculated results. Calculational intercomparison (DORT vs. MCNP-5) and comparison with measured values (MCNP-5 and DORT vs. E) have shown agreement within 10-15% for different space points and reaction rates. The paper submits a discussion of results and makes conclusions about practice use of MCNP-5 code for ex-core neutron transport calculations in expert analysis. (authors)
Simulations of X-ray spectrum and HVL for mammographic equipment using MCNP5 code
Energy Technology Data Exchange (ETDEWEB)
Souza, Rafael Toledo F. de; Alvarez, Matheus; Velo, Alexandre F.; Oliveira, Marcela de; Miranda, Jose Ricardo A. [Universidade Estadual Paulista Julio de mesquita Filho (UNESP), Botucatu, SP (Brazil). Inst. de Biociencias de Botucatu. Dept. de Fisica e Biofisica; Pina, Diana R. [Universidade Estadual Paulista Julio de mesquita Filho (UNESP), Botucatu, SP (Brazil). Fac. de Medicina. Dept. de Doencas Tropicais e Diagnostico por Imagem
2012-07-01
Full text: The main goal of mammography is early detection of breast cancer. Thus, the mammograph should be designed so that the X-ray photons are emitted within an appropriate energy range, to distinguish the normal breast tissue and cancerous tissue. The distribution of the photons amount of X-ray beam, with their respective energies, is called the spectrum. From the spectrum it is possible to estimate the quality of the X-ray beam from the Half Value Layer (HVL). Objectives: This study aims to simulate the Senographe 600T mammography unit, manufactured by General Electric (GE), using the MCNP5 Monte Carlo code, to obtain its spectrum and HVL, and compare the HVL of the simulated model with experimental data. Method: the mammography unit was simulated using a simplified model which a beam of 2x10{sup 8} electrons focuses on a Mo target angled 12 degrees, within a capsule filled with vacuum. The incident electrons were converted into photons. The capsule has a beryllium window, allowing the passage of the X-ray beam. The beam is detected by an air cylinder with 1 cm thickness placed 60 cm from the target. On the path of X-ray beam, is inserted a 0.03 mm Mo filter located 1.6 cm after the beryllium window. The space between the capsule and the detector cylinder was filled with air. The quality of X-ray beam was verified from the HVL using the MCNP5 code and the experimental method for the voltage range typically used in clinical routine (26-31 kVp). Results and discussion: the X-ray spectrum of the mammography device is satisfactorily simulated by MCNP5, showing the characteristic radiation peaks of molybdenum at 17.479 keV and 19.602 keV, the filtered spectrum generated by Bremsstrahlung, and reducing the total number of photons with the decrease in applied tension (kVp). The HVL obtained by MCNP5 and experimental measurements show a maximum difference of 5.31% (for 31 kVp). The result of both methods are within acceptable limits established by national
Image enhancement using MCNP5 code and MATLAB in neutron radiography.
Tharwat, Montaser; Mohamed, Nader; Mongy, T
2014-07-01
This work presents a method that can be used to enhance the neutron radiography (NR) image for objects with high scattering materials like hydrogen, carbon and other light materials. This method used Monte Carlo code, MCNP5, to simulate the NR process and get the flux distribution for each pixel of the image and determines the scattered neutron distribution that caused image blur, and then uses MATLAB to subtract this scattered neutron distribution from the initial image to improve its quality. This work was performed before the commissioning of digital NR system in Jan. 2013. The MATLAB enhancement method is quite a good technique in the case of static based film neutron radiography, while in neutron imaging (NI) technique, image enhancement and quantitative measurement were efficient by using ImageJ software. The enhanced image quality and quantitative measurements were presented in this work.
Gallardo, S; Querol, A; Ródenas, J; Verdú, G
2011-01-01
An accurate knowledge of the photonic spectra emitted by X-ray tubes in radiodiagnostics is essential to better estimate the imparted dose to patients and to improve the image quality obtained with these devices. In this work, several X-ray spectra have been simulated using the MCNP5 code to simulate X-ray production in a commercial device. To validate the Monte Carlo results, simulated spectra have been compared to those extracted from the IPEM 78 database. The uncertainty associated to some geometrical features of the tube and its effect on the simulated spectra has been analyzed using the Noether-Wilks formula. This analysis has been focused on the thickness of collimators, filters, shielding and barrel shutter. Furthermore, results show that the uncertainty due to geometrical parameters (0.98% in terms of Root Mean Squared) is higher than the statistical uncertainty associated to the MCNP5 calculations.
Simulation of dental intensifying screen for intraoral radiographic using MCNP5 code
Energy Technology Data Exchange (ETDEWEB)
Ferreira, Vanessa M.; Oliveira, Renato C.M., E-mail: vanessamachado@ufmg.br [Curso Superior de Tecnologia em Radiologia. Faculdade de Medicina da Universidade Federal de Minas Gerais, Belo Horizonte, MG (Brazil); Barros, Graiciany P.; Oliveira, Arno H.; Veloso, M. Auxiliadora F. [Departamento de Engenharia Nuclear. Escola de Engenharia. Universidade Federal de Minas Gerais, Belo Horizonte, MG (Brazil)
2011-07-01
One of basic principles for radiological protection is the optimization of techniques for obtain radiographic images, in way that the dose in the patient is kept as low as reasonably achievable (ALARA). Intensifying screens are used in medical radiology, which reduce considerably the dose rates in the production of radiographic images, maintaining the quality of these, while in dental radiology, there is no a intensifying screen available for intraoral examinations. From this technological requirement, this paper evaluates a computational modeling of an intensifying screen for use in intraoral radiography. For this, it was used the Monte Carlo code MCNP5 that allows the radiography simulation through the transport of electrons and photons in the different materials present in this examination. The goal of an intensifying screen is the conversion of X-ray photons to photons in the visible spectrum, knowing that radiographic films are more sensitive to light photons than to X-ray photons. So the screen should be composed of an efficient material for converting x-rays photons in light photons, therefore was made simulations using different materials, thicknesses and positions possible for placing screen in radiographic film in order to find the way more technically feasible. (author)
Benchmark of Atucha-2 PHWR RELAP5-3D control rod model by Monte Carlo MCNP5 core calculation
Energy Technology Data Exchange (ETDEWEB)
Pecchia, M.; D' Auria, F. [San Piero A Grado Nuclear Research Group GRNSPG, Univ. of Pisa, via Diotisalvi, 2, 56122 - Pisa (Italy); Mazzantini, O. [Nucleo-electrica Argentina Societad Anonima NA-SA, Buenos Aires (Argentina)
2012-07-01
Atucha-2 is a Siemens-designed PHWR reactor under construction in the Republic of Argentina. Its geometrical complexity and peculiarities require the adoption of advanced Monte Carlo codes for performing realistic neutronic simulations. Therefore core models of Atucha-2 PHWR were developed using MCNP5. In this work a methodology was set up to collect the flux in the hexagonal mesh by which the Atucha-2 core is represented. The scope of this activity is to evaluate the effect of obliquely inserted control rod on neutron flux in order to validate the RELAP5-3D{sup C}/NESTLE three dimensional neutron kinetic coupled thermal-hydraulic model, applied by GRNSPG/UNIPI for performing selected transients of Chapter 15 FSAR of Atucha-2. (authors)
Energy Technology Data Exchange (ETDEWEB)
Fernandes, Marco A.R., E-mail: marco@cetea.com.b, E-mail: marfernandes@fmb.unesp.b [Universidade Estadual Paulista Julio de Mesquita Filho (FMB/UNESP), Botucatu, SP (Brazil). Fac. de Medicina; Ribeiro, Victor A.B. [Universidade Estadual Paulista Julio de Mesquita Filho (IBB/UNESP), Botucatu, SP (Brazil). Inst. de Biociencias; Viana, Rodrigo S.S.; Coelho, Talita S. [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2011-07-01
The paper illustrates the use of the Monte Carlo method, MCNP-5C code, to analyze the attenuation curve behavior of the 50 kVp radiation beam from superficial radiotherapy equipment as Dermopan2 model. The simulations seek to verify the MCNP-5C code performance to study the variation of the attenuation curve - percentage depth dose (PDD) curve - in function of the radiation field dimension used at radiotherapy of skin tumors with 50 kVp X-ray beams. The PDD curve was calculated for six different radiation field sizes with circular geometry of 1.0, 2.0, 3.0, 4.0, 5.0 and 6.0 cm in diameter. The radiation source was modeled considering a tungsten target with inclination 30 deg, focal point of 6.5 mm in diameter and energy beam of 50 kVp; the X-ray spectrum was calculated with the MCNP-5C code adopting total filtration (beryllium window of 1 mm and aluminum additional filter of 1 mm). The PDD showed decreasing behavior with the attenuation depth similar what is presented on the literature. There was not significant variation at the PDD values for the radiation field between 1.0 and 4.0 cm in diameter. The differences increased for fields of 5.0 and 6.0 cm and at attenuation depth higher than 1.0 cm. When it is compared the PDD values for fields of 3.0 and 6.0 cm in diameter, it verifies the greater difference (12.6 %) at depth of 5.7 cm, proving the scattered radiation effect. The MCNP-5C code showed as an appropriate procedure to analyze the attenuation curves of the superficial radiotherapy beams. (author)
Energy Technology Data Exchange (ETDEWEB)
Koivunoro, Hanna; Siiskonen, Teemu; Kotiluoto, Petri; Auterinen, Iiro; Hippelaeinen, Eero; Savolainen, Sauli [Department of Physics, University of Helsinki, P.O. Box 64, FI-00014 Helsinki University (Finland) and Department of Oncology, Helsinki University Central Hospital, FI-00029 HUS (Finland); STUK-Radiation and Nuclear Safety Authority, P.O. Box 14, FI-00881 Helsinki (Finland); VTT Technical Research Centre of Finland, P.O. Box 1000, FI-02044 VTT (Finland); Department of Physics, University of Helsinki, P.O. Box 64, FI-00014 Helsinki University (Finland); HUS Medical Imaging Centre, Helsinki University Central Hospital, FI-00029 HUS (Finland)
2012-03-15
Purpose: In this work, accuracy of the mcnp5 code in the electron transport calculations and its suitability for ionization chamber (IC) response simulations in photon beams are studied in comparison to egsnrc and penelope codes. Methods: The electron transport is studied by comparing the depth dose distributions in a water phantom subdivided into thin layers using incident energies (0.05, 0.1, 1, and 10 MeV) for the broad parallel electron beams. The IC response simulations are studied in water phantom in three dosimetric gas materials (air, argon, and methane based tissue equivalent gas) for photon beams ({sup 60}Co source, 6 MV linear medical accelerator, and mono-energetic 2 MeV photon source). Two optional electron transport models of mcnp5 are evaluated: the ITS-based electron energy indexing (mcnp5{sub ITS}) and the new detailed electron energy-loss straggling logic (mcnp5{sub new}). The electron substep length (ESTEP parameter) dependency in mcnp5 is investigated as well. Results: For the electron beam studies, large discrepancies (>3%) are observed between the mcnp5 dose distributions and the reference codes at 1 MeV and lower energies. The discrepancy is especially notable for 0.1 and 0.05 MeV electron beams. The boundary crossing artifacts, which are well known for the mcnp5{sub ITS}, are observed for the mcnp5{sub new} only at 0.1 and 0.05 MeV beam energies. If the excessive boundary crossing is eliminated by using single scoring cells, the mcnp5{sub ITS} provides dose distributions that agree better with the reference codes than mcnp5{sub new}. The mcnp5 dose estimates for the gas cavity agree within 1% with the reference codes, if the mcnp5{sub ITS} is applied or electron substep length is set adequately for the gas in the cavity using the mcnp5{sub new}. The mcnp5{sub new} results are found highly dependent on the chosen electron substep length and might lead up to 15% underestimation of the absorbed dose. Conclusions: Since the mcnp5 electron
Energy Technology Data Exchange (ETDEWEB)
Galicia A, J.; Francois L, J. L. [UNAM, Facultad de Ingenieria, Departamento de Sistemas Energeticos, Ciudad Universitaria, 04510 Ciudad de Mexico (Mexico); Aguilar H, F., E-mail: blink19871@hotmail.com [ININ, Carretera Mexico-Toluca s/n, 52750 Ocoyoacac, Estado de Mexico (Mexico)
2015-09-15
The main purpose of this paper is to obtain a model of the reactor core TRIGA Mark III that accurately represents the real operating conditions to 1 M Wth, using the Monte Carlo code MCNP5. To provide a more detailed analysis, different models of the reactor core were realized by simulating the control rods extracted and inserted in conditions in cold (293 K) also including an analysis for shutdown margin, so that satisfied the Operation Technical Specifications. The position they must have the control rods to reach a power equal to 1 M Wth, were obtained from practice entitled Operation in Manual Mode performed at Instituto Nacional de Investigaciones Nucleares (ININ). Later, the behavior of the K{sub eff} was analyzed considering different temperatures in the fuel elements, achieving calculate subsequently the values that best represent the actual reactor operation. Finally, the calculations in the developed model for to obtain the distribution of average flow of thermal, epithermal and fast neutrons in the six new experimental facilities are presented. (Author)
Directory of Open Access Journals (Sweden)
Pešić Milan P.
2012-01-01
Full Text Available A numerical simulation of the radiological consequences of the RB reactor reactivity excursion accident, which occurred on October 15, 1958, and an estimation of the total doses received by the operators were run by the MCNP5 computer code. The simulation was carried out under the same assumptions as those used in the 1960 IAEA-organized experimental simulation of the accident: total fission energy of 80 MJ released in the accident and the frozen positions of the operators. The time interval of exposure to high doses received by the operators has been estimated. Data on the RB1/1958 reactor core relevant to the accident are given. A short summary of the accident scenario has been updated. A 3-D model of the reactor room and the RB reactor tank, with all the details of the core, created. For dose determination, 3-D simplified, homogenised, sexless and faceless phantoms, placed inside the reactor room, have been developed. The code was run for a number of neutron histories which have given a dose rate uncertainty of less than 2%. For the determination of radiation spectra escaping the reactor core and radiation interaction in the tissue of the phantoms, the MCNP5 code was run (in the KCODE option and “mode n p e”, with a 55-group neutron spectra, 35-group gamma ray spectra and a 10-group electron spectra. The doses were determined by using the conversion of flux density (obtained by the F4 tally in the phantoms to doses using factors taken from ICRP-74 and from the deposited energy of neutrons and gamma rays (obtained by the F6 tally in the phantoms’ tissue. A rough estimation of the time moment when the odour of ozone was sensed by the operators is estimated for the first time and given in Appendix A.1. Calculated total absorbed and equivalent doses are compared to the previously reported ones and an attempt to understand and explain the reasons for the obtained differences has been made. A Root Cause Analysis of the accident was done and
Energy Technology Data Exchange (ETDEWEB)
Aredes, Vitor Ottoni; Bitelli, Ulysses d' Utra; Mura, Luiz Ernesto C.; Santos, Diogo Feliciano dos; Lima, Ana Cecilia de Souza, E-mail: ubitelli@ipen.br [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2015-07-01
This study aims to determine the distribution of thermal neutron flux in the IPEN/MB-01 nuclear reactor core assembled with cylindrical core configuration of minor excess of reactivity with 568 fuel rods (28 fuel rods in diameter). The thermal neutron flux at the positions of irradiation derive from the method of reaction rate using gold foils. The experiment consists in inserting gold activations foils with and without cadmium coverage (cadmium boxes with 0.0502 cm thickness) in several positions throughout the active core. After irradiation, activity induced by nuclear reaction rates over gold foils is assessed by gamma ray spectrometry using a high-purity germanium (HPGe) detector. Experimental results are compared to those derived from calculations performed using a three dimensional CITATION diffusion code and MCNP-5 code and a proper nuclear data library. While calculated neutron flux data shows good agreement with experimental values in regions with little disturbance in the neutron flux, also showing that in the region of the reflectors of neutrons and near the control rods, the diffusion theory is not very precise. The average value of thermal neutron flux obtained experimentally compared to the calculated value by CITATION code and MCNP-5 code respectively show a difference of 1.18% and 0.84% at a nuclear power level of 74.65 ± 3.28 % watts. The average measured value of thermal neutron flux is 4.10 10{sup 8} ± 5.25% n/cm{sup 2}s. (author)
Gerardy, I; Ródenas, J; van Dycke, M; Gallardo, S; Ceccolini, Elisa
2010-01-01
A gynaecological applicator consisting of a metallic intra-uterine tube with a plastic vaginal applicator and an HDR Ir-192 source have been simulated with MCNP5 (Monte Carlo code). A solid phantom has been designed to perform measurements around the applicator with radiochromic films. The isodose curves obtained are compared with curves calculated with the F4MESH tally of MCNP5 with a good agreement. A pinpoint ionization chamber has been used to evaluate dose at some reference points.
Institute of Scientific and Technical Information of China (English)
曹振; 阮锡超; 孟贝蒂; 石翠燕
2014-01-01
根据AAPM TG43U1的推荐,使用MCNP5与EGSnrc两种蒙特卡罗程序计算6711型125I种子源剂量计算参数,并将两者计算结果和AAPM推荐值比较,得到相对偏差结果如下:剂量率常数,MCNP5为0.62％,EGSnrc为2.07％;径向剂量函数,MNCP5为0.15％-5.12％,EGSnrc为0％-2.18％.两者计算结果均与推荐值符合得很好,而EGSnrc的计算结果更具优势.
MCNP5 study on kinetics parameters of coupled fast-thermal system HERBE
Directory of Open Access Journals (Sweden)
Pešić Milan P.
2011-01-01
Full Text Available New validation of the well-known Monte Carlo code MCNP5 against measured criticality and kinetics data for the coupled fast-thermal HERBE System at the Reactor B critical assembly is shown in this paper. Results of earlier calculations of these criticality and kinetics parameters, done by combination of transport and diffusion codes using two-dimension geometry model are compared to results of new calculations carried out by the MCNP5 code in three-dimension geometry. Satisfactory agreements in comparison of new results with experimental data, in spite complex heterogeneous composition of the HERBE core, are achieved confirming that MCNP5 code could apply successfully to study on HERBE kinetics parameters after uncertainties in impurities in material compositions and positions of fuel elements in fast zone were removed.
Uncertainty analysis in MCNP5 calculations for brachytherapy treatment
Energy Technology Data Exchange (ETDEWEB)
Gerardy, I., E-mail: gerardy@isib.be [Institut Superieur Industriel de Bruxelles, 150, Rue Royale, B-1000 Brussels (Belgium); Rodenas, J.; Gallardo, S. [Departamento de Ingenieria Quimica y Nuclear, Universidad Politecnica de Valencia (Spain)
2011-08-15
The Monte Carlo (MC) method can be applied to simulate brachytherapy treatment planning. The MCNP5 code gives, together with results, a statistical uncertainty associated with them. However, the latter is not the only existing uncertainty related to the simulation and other uncertainties must be taken into account. A complete analysis of all sources of uncertainty having some influence on results of the simulation of brachytherapy treatment is presented in this paper. This analysis has been based on the recommendations of the American Association for Physicist in Medicine (AAPM) and of the International Standard Organisation (ISO).
Energy Technology Data Exchange (ETDEWEB)
Corral B, J. R.
2015-07-01
Humans should avoid exposure to radiation, because the consequences are harmful to health. Although there are different emission sources of radiation, generated by medical devices they are usually of great interest, since people who attend hospitals are exposed in one way or another to ionizing radiation. Therefore, is important to conduct studies on radioactive levels that are generated in hospitals, as a result of the use of medical equipment. To determine levels of exposure speed of a radioactive facility there are different methods, including the radiation detector and computational method. This thesis uses the computational method. With the program MCNP5 was determined the speed of the radiation exposure in the radiotherapy room of Cancer Center of ABC Hospital in Mexico City. In the application of computational method, first the thicknesses of the shields were calculated, using variables as: 1) distance from the shield to the source; 2) desired weekly equivalent dose; 3) weekly total dose equivalent emitted by the equipment; 4) occupation and use factors. Once obtained thicknesses, we proceeded to model the bunker using the mentioned program. The program uses the Monte Carlo code to probabilistic ally determine the phenomena of interaction of radiation with the shield, which will be held during the X-ray emission from the linear accelerator. The results of computational analysis were compared with those obtained experimentally with the detection method, for which was required the use of a Geiger-Muller counter and the linear accelerator was programmed with an energy of 19 MV with 500 units monitor positioning the detector in the corresponding boundary. (Author)
Evaluation of Geometric Progression (GP Buildup Factors using MCNP Codes (MCNP6.1 and MCNP5-1.60
Directory of Open Access Journals (Sweden)
Kim Kyung-O
2016-01-01
Full Text Available The gamma-ray buildup factors of three-dimensional point kernel code (QAD-CGGP are re-evaluated by using MCNP codes (MCNP6.1 and MCNPX5-1.60 and ENDF/B-VI.8 photoatomic data, which cover an energy range of 0.015–15 MeV and an iron thickness of 0.5–40 Mean Free Path (MFP. These new data are fitted to the Geometric Progression (GP fitting function and are then compared with ANS standard data equipped with QAD-CGGP. In addition, a simple benchmark calculation was performed to compare the QAD-CGGP results applied with new and existing buildup factors based on the MCNP codes. In the case of the buildup factors of low-energy gamma-rays, new data are evaluated to be about 5% higher than the existing data. In other cases, these new data present a similar trend based on the specific penetration depth, while existing data continuously increase beyond that depth. In a simple benchmark, the calculations using the existing data were slightly underestimated compared to the reference data at a deep penetration depth. On the other hand, the calculations with new data were stabilized with an increasing penetration depth, despite a slight overestimation at a shallow penetration depth.
Comparative Criticality Analysis of Two Monte Carlo Codes on Centrifugal Atomizer: MCNPS and SCALE
Energy Technology Data Exchange (ETDEWEB)
Kang, H-S; Jang, M-S; Kim, S-R [NESS, Daejeon (Korea, Republic of); Park, J-M; Kim, K-N [Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of)
2015-10-15
There are two well-known Monte Carlo codes for criticality analysis, MCNP5 and SCALE. MCNP5 is a general-purpose Monte Carlo N-Particle code that can be used for neutron, photon, electron or coupled neutron / photon / electron transport, including the capability to calculate eigenvalues for critical system as a main analysis code. SCALE provides a comprehensive, verified and validated, user-friendly tool set for criticality safety, reactor physics, radiation shielding, radioactive source term characterization, and sensitivity and uncertainty analysis. SCALE was conceived and funded by US NRC to perform standardized computer analysis for licensing evaluation and is used widely in the world. We performed a validation test of MCNP5 and a comparative analysis of Monte Carlo codes, MCNP5 and SCALE, in terms of the critical analysis of centrifugal atomizer. In the criticality analysis using MCNP5 code, we obtained the statistically reliable results by using a large number of source histories per cycle and performing of uncertainty analysis.
Energy Technology Data Exchange (ETDEWEB)
Cramer, S.N.
1984-01-01
The MORSE code is a large general-use multigroup Monte Carlo code system. Although no claims can be made regarding its superiority in either theoretical details or Monte Carlo techniques, MORSE has been, since its inception at ORNL in the late 1960s, the most widely used Monte Carlo radiation transport code. The principal reason for this popularity is that MORSE is relatively easy to use, independent of any installation or distribution center, and it can be easily customized to fit almost any specific need. Features of the MORSE code are described.
Energy Technology Data Exchange (ETDEWEB)
Boffie, J., E-mail: jboffie@yahoo.com [Department of Nuclear Engineering and Material Science, School of Nuclear and Allied Sciences (SNAS), University of Ghana, P.O. Box AE 1, Atomic Energy, Accra (Ghana); National Nuclear Research Institute, Ghana Atomic Energy Commission, P.O. Box LG 80, Legon, Accra (Ghana); Akaho, E.H.K. [Department of Nuclear Engineering and Material Science, School of Nuclear and Allied Sciences (SNAS), University of Ghana, P.O. Box AE 1, Atomic Energy, Accra (Ghana); Nyarko, B.J.B.; Odoi, H.C.; Tuffour-Achampong, K.; Abrefah, R.G. [Department of Nuclear Engineering and Material Science, School of Nuclear and Allied Sciences (SNAS), University of Ghana, P.O. Box AE 1, Atomic Energy, Accra (Ghana); National Nuclear Research Institute, Ghana Atomic Energy Commission, P.O. Box LG 80, Legon, Accra (Ghana)
2013-12-15
Highlights: • The operating parameters for both the HEU core and proposed LEU core were similar. • The length of the Cd in the capsules must be increased for its use in the LEU core. • Cd rabbits can emergently be used to shut down MNSRs. - Abstract: Miniature Neutron Source Reactors (MNSRs) are noted to be among highly safe research reactors. However, because of its use of one control rod for reactivity control and shutdown purposes, alternative methods of shutting it down are important. The Ghana MNSR uses four cadmium rabbits of approximate dimensions 6.5 cm × 5.0 cm × 0.1 cm and mass of 9.48 g each to emergently shut down the reactor. The Monte Carlo N-Particle code; version 5, (MCNP5) was used to design the high enriched uranium (HEU) and low enriched uranium (LEU) cores of the MNSR with four cadmium rabbits inserted in four inner irradiation sites of each core. The operating parameters and shutdown parameters for both cores with the central control rod (CCR) either fully withdrawn or fully inserted had similar results with the HEU core having slightly better results in terms of safety. However, the results show that the four inserted cadmium rabbits make the HEU core subcritical whiles in the LEU core, it still remains critical (k{sub eff} = 1.00005 ± 0.00007). The length of the cadmium material in each cadmium rabbit must therefore be increased by at least 0.5 cm in order to attain subcriticality (k{sub eff} = 0.99989 ± 0.00006) and shutdown margin of 0.11 mk when inserted in the LEU core.
THE MCNPX MONTE CARLO RADIATION TRANSPORT CODE
Energy Technology Data Exchange (ETDEWEB)
WATERS, LAURIE S. [Los Alamos National Laboratory; MCKINNEY, GREGG W. [Los Alamos National Laboratory; DURKEE, JOE W. [Los Alamos National Laboratory; FENSIN, MICHAEL L. [Los Alamos National Laboratory; JAMES, MICHAEL R. [Los Alamos National Laboratory; JOHNS, RUSSELL C. [Los Alamos National Laboratory; PELOWITZ, DENISE B. [Los Alamos National Laboratory
2007-01-10
MCNPX (Monte Carlo N-Particle eXtended) is a general-purpose Monte Carlo radiation transport code with three-dimensional geometry and continuous-energy transport of 34 particles and light ions. It contains flexible source and tally options, interactive graphics, and support for both sequential and multi-processing computer platforms. MCNPX is based on MCNP4B, and has been upgraded to most MCNP5 capabilities. MCNP is a highly stable code tracking neutrons, photons and electrons, and using evaluated nuclear data libraries for low-energy interaction probabilities. MCNPX has extended this base to a comprehensive set of particles and light ions, with heavy ion transport in development. Models have been included to calculate interaction probabilities when libraries are not available. Recent additions focus on the time evolution of residual nuclei decay, allowing calculation of transmutation and delayed particle emission. MCNPX is now a code of great dynamic range, and the excellent neutronics capabilities allow new opportunities to simulate devices of interest to experimental particle physics; particularly calorimetry. This paper describes the capabilities of the current MCNPX version 2.6.C, and also discusses ongoing code development.
MCOR - Monte Carlo depletion code for reference LWR calculations
Energy Technology Data Exchange (ETDEWEB)
Puente Espel, Federico, E-mail: fup104@psu.edu [Department of Mechanical and Nuclear Engineering, Pennsylvania State University (United States); Tippayakul, Chanatip, E-mail: cut110@psu.edu [Department of Mechanical and Nuclear Engineering, Pennsylvania State University (United States); Ivanov, Kostadin, E-mail: kni1@psu.edu [Department of Mechanical and Nuclear Engineering, Pennsylvania State University (United States); Misu, Stefan, E-mail: Stefan.Misu@areva.com [AREVA, AREVA NP GmbH, Erlangen (Germany)
2011-04-15
Research highlights: > Introduction of a reference Monte Carlo based depletion code with extended capabilities. > Verification and validation results for MCOR. > Utilization of MCOR for benchmarking deterministic lattice physics (spectral) codes. - Abstract: The MCOR (MCnp-kORigen) code system is a Monte Carlo based depletion system for reference fuel assembly and core calculations. The MCOR code is designed as an interfacing code that provides depletion capability to the LANL Monte Carlo code by coupling two codes: MCNP5 with the AREVA NP depletion code, KORIGEN. The physical quality of both codes is unchanged. The MCOR code system has been maintained and continuously enhanced since it was initially developed and validated. The verification of the coupling was made by evaluating the MCOR code against similar sophisticated code systems like MONTEBURNS, OCTOPUS and TRIPOLI-PEPIN. After its validation, the MCOR code has been further improved with important features. The MCOR code presents several valuable capabilities such as: (a) a predictor-corrector depletion algorithm, (b) utilization of KORIGEN as the depletion module, (c) individual depletion calculation of each burnup zone (no burnup zone grouping is required, which is particularly important for the modeling of gadolinium rings), and (d) on-line burnup cross-section generation by the Monte Carlo calculation for 88 isotopes and usage of the KORIGEN libraries for PWR and BWR typical spectra for the remaining isotopes. Besides the just mentioned capabilities, the MCOR code newest enhancements focus on the possibility of executing the MCNP5 calculation in sequential or parallel mode, a user-friendly automatic re-start capability, a modification of the burnup step size evaluation, and a post-processor and test-matrix, just to name the most important. The article describes the capabilities of the MCOR code system; from its design and development to its latest improvements and further ameliorations. Additionally
Energy Technology Data Exchange (ETDEWEB)
Paxton, Adam B.; Davis, Stephen D.; DeWerd, Larry A. [Department of Medical Physics, University of Wisconsin-Madison, Madison, Wisconsin 53705 (United States); Department of Medical Physics, University of Wisconsin-Madison, Madison, Wisconsin 53705 and McGill University Health Centre, Department of Medical Physics, Montreal, Quebec H3G 1A4 (Canada); Department of Medical Physics, University of Wisconsin-Madison, Madison, Wisconsin 53705 (United States)
2012-03-15
Purpose: Recent advances in the imaging of {sup 90}Y using positron emission tomography (PET) and improved uncertainty in the branching ratio for the internal pair production component of {sup 90}Y decay allow for a more accurate determination of the activity distribution of {sup 90}Y microspheres within a patient. This improved activity distribution can be convolved with the dose kernel of {sup 90}Y to calculate the dose distribution within a patient. This work investigates the effects of microsphere and surrounding material composition on {sup 90}Y dose kernels using egsnrc and mcnp5 and compares the results of these two transport codes. Methods: Monte Carlo simulations were performed with egsnrc and mcnp5 to calculate the dose rate at multiple radial distances around various {sup 90}Y sources. Point source simulations were completed with mcnp5 to determine the optimal electron transport settings for this work. After determining the optimal settings, point source simulations were completed using egsnrc (user code edknrc) and mcnp5 in water and liver [as defined by the International Commission on Radiation Units and Measurements (ICRU) Report 44]. The results were compared to ICRU Report 72 reference data. Point source simulations were also completed in water with a density of 1.06 g{center_dot}cm{sup -3} to evaluate the effect of the density of the surrounding material. Glass and resin microsphere simulations were performed with average and maximum diameter and density values (based on values given in the literature) in water and in liver. The results were compared to point source simulation results using the same transport code and in the same surrounding material. All simulations had statistical uncertainties less than 1%. Results: The optimal transport settings in mcnp5 for this work included using the energy-and step-specific algorithm (DBCN 17J 2) and ESTEP set to 10. These settings were used for all subsequent simulations with mcnp5. The point source
MCNP5 CRITICALITY VALIDATION AND BIAS FOR INTERMEDIATE ENRICHED URANIUM SYSTEMS
Energy Technology Data Exchange (ETDEWEB)
FINFROCK SH
2009-12-10
The purpose of this analysis is to validate the Monte Carlo N-Particle 5 (MCNP5) code Version 1.40 (LA-UR-03-1987, 2005) and its cross-section database for k-code calculations of intermediate enriched uranium systems on INTEL{reg_sign} processor based PC's running any version of the WINDOWS operating system. Configurations with intermediate enriched uranium were modeled with the moderator range of 39 {le} H/Fissile {le} 1438. See Table 2-1 for brief descriptions of selected cases and Table 3-1 for the range of applicability for this validation. A total of 167 input cases were evaluated including bare and reflected systems in a single body or arrays. The 167 cases were taken directly from the previous (Version 4C [Lan 2005]) validation database. Section 2.0 list data used to calculate k-effective (k{sub eff}) for the 167 experimental criticality benchmark cases using the MCNP5 code v1.40 and its cross section database. Appendix B lists the MCNP cross-section database entries validated for use in evaluating the intermediate enriched uranium systems for criticality safety. The dimensions and atom densities for the intermediate enriched uranium experiments were taken from NEA/NSC/DOC(95)03, September 2005, which will be referred to as the benchmark handbook throughout the report. For these input values, the experimental benchmark k{sub eff} is approximately 1.0. The MCNP validation computer runs ran to an accuracy of approximately {+-} 0.001. For the cases where the reported benchmark k{sub eff} was not equal to 1.0000 the MCNP calculational results were normalized. The difference between the MCNP validation computer runs and the experimentally measured k{sub eff} is the MCNP5 v1.40 bias. The USLSTATS code (ORNL 1998) was utilized to perform the statistical analysis and generate an acceptable maximum k{sub eff} limit for calculations of the intermediate enriched uranium type systems.
An Advanced Neutronic Analysis Toolkit with Inline Monte Carlo capability for BHTR Analysis
Energy Technology Data Exchange (ETDEWEB)
William R. Martin; John C. Lee
2009-12-30
Monte Carlo capability has been combined with a production LWR lattice physics code to allow analysis of high temperature gas reactor configurations, accounting for the double heterogeneity due to the TRISO fuel. The Monte Carlo code MCNP5 has been used in conjunction with CPM3, which was the testbench lattice physics code for this project. MCNP5 is used to perform two calculations for the geometry of interest, one with homogenized fuel compacts and the other with heterogeneous fuel compacts, where the TRISO fuel kernels are resolved by MCNP5.
TRIPOLI-4{sup ®} Monte Carlo code ITER A-lite neutronic model validation
Energy Technology Data Exchange (ETDEWEB)
Jaboulay, Jean-Charles, E-mail: jean-charles.jaboulay@cea.fr [CEA, DEN, Saclay, DM2S, SERMA, F-91191 Gif-sur-Yvette (France); Cayla, Pierre-Yves; Fausser, Clement [MILLENNIUM, 16 Av du Québec Silic 628, F-91945 Villebon sur Yvette (France); Damian, Frederic; Lee, Yi-Kang; Puma, Antonella Li; Trama, Jean-Christophe [CEA, DEN, Saclay, DM2S, SERMA, F-91191 Gif-sur-Yvette (France)
2014-10-15
3D Monte Carlo transport codes are extensively used in neutronic analysis, especially in radiation protection and shielding analyses for fission and fusion reactors. TRIPOLI-4{sup ®} is a Monte Carlo code developed by CEA. The aim of this paper is to show its capability to model a large-scale fusion reactor with complex neutron source and geometry. A benchmark between MCNP5 and TRIPOLI-4{sup ®}, on the ITER A-lite model was carried out; neutron flux, nuclear heating in the blankets and tritium production rate in the European TBMs were evaluated and compared. The methodology to build the TRIPOLI-4{sup ®} A-lite model is based on MCAM and the MCNP A-lite model. Simplified TBMs, from KIT, were integrated in the equatorial-port. A good agreement between MCNP and TRIPOLI-4{sup ®} is shown; discrepancies are mainly included in the statistical error.
Energy Technology Data Exchange (ETDEWEB)
Lin, Yi-Chun [Department of Biomedical Engineering and Environmental Sciences, National Tsing Hua University, Taiwan (China); Liu, Yuan-Hao, E-mail: yhl.taiwan@gmail.com [Boron Neutron Capture Therapy Center, Nuclear Science and Technology Development Center, National Tsing Hua University, No. 101, Section 2, Kuang-Fu Road, Hsinchu City 30013, Taiwan (China); Nievaart, Sander [Institute for Energy, Joint Research Centre, European Commission, Petten (Netherlands); Chen, Yen-Fu [Department of Engineering and System Science, National Tsing Hua University, Taiwan (China); Wu, Shu-Wei; Chou, Wen-Tsae [Department of Biomedical Engineering and Environmental Sciences, National Tsing Hua University, Taiwan (China); Jiang, Shiang-Huei [Institute of Nuclear Engineering and Science, National Tsing Hua University, Taiwan (China)
2011-10-01
High energy photon (over 10 MeV) and neutron beams adopted in radiobiology and radiotherapy always produce mixed neutron/gamma-ray fields. The Mg(Ar) ionization chambers are commonly applied to determine the gamma-ray dose because of its neutron insensitive characteristic. Nowadays, many perturbation corrections for accurate dose estimation and lots of treatment planning systems are based on Monte Carlo technique. The Monte Carlo codes EGSnrc, FLUKA, GEANT4, MCNP5, and MCNPX were used to evaluate energy dependent response functions of the Exradin M2 Mg(Ar) ionization chamber to a parallel photon beam with mono-energies from 20 keV to 20 MeV. For the sake of validation, measurements were carefully performed in well-defined (a) primary M-100 X-ray calibration field, (b) primary {sup 60}Co calibration beam, (c) 6-MV, and (d) 10-MV therapeutic beams in hospital. At energy region below 100 keV, MCNP5 and MCNPX both had lower responses than other codes. For energies above 1 MeV, the MCNP ITS-mode greatly resembled other three codes and the differences were within 5%. Comparing to the measured currents, MCNP5 and MCNPX using ITS-mode had perfect agreement with the {sup 60}Co, and 10-MV beams. But at X-ray energy region, the derivations reached 17%. This work shows us a better insight into the performance of different Monte Carlo codes in photon-electron transport calculation. Regarding the application of the mixed field dosimetry like BNCT, MCNP with ITS-mode is recognized as the most suitable tool by this work.
Directory of Open Access Journals (Sweden)
Dinan A.
2011-12-01
Full Text Available Banyak perangkat kritik dibangun untuk memenuhi kebutuhan studi fenomena kecelakaan kritikalitas pada larutan fisil di fasilitas daur bahan bakar nuklir. Salah satu diantaranya adalah perangkat kritik SCAMP. Di perangkat ini dikerjakan eksperimen kritikalitas menggunakan bejana silindris stainless steel berisi larutan plutonium uranium nitrat (Pu ditambah U nitrat. Sebanyak 7 eksperimen didemonstrasikan dengan reflektor air di semua sisi permukaan bejana larutan kecuali di bagian atas bejana. Makalah ini membahas simulasi transport Monte Carlo MCNP5 dalam eksperimen kritikalitas larutan Pu ditambah U nitrat dengan reflektor air dan polyethylene. Simulasi MCNP5 dengan pustaka ENDF/BVI memberikan hasil yang paling dekat dengan data eksperimen terutama pada kasus A untuk varian geometri 4. Dibandingkan pustaka ENDF/BV, perhitungan kritikalitas dengan pustaka ENDF/B-VI memberikan hasil lebih dekat dengan perhitungan MONK dimana bias perhitungannya kurang dari 0,44%, khususnya pada kasus A namun pada kasus B dan C simulasi MCNP5dengan pustaka ENDF/BV memberikan hasil dengan kecenderungan lebih baik dibandingkan pustaka ENDFB/VI dengan bias perhitungan kurang dari 2,67% dan kurang dari 1,13%. Secara keseluruhan dapat disimpulkan bahwa MCNP5 telah menunjukkan reliabilitasnya dalam simulasi kritikalitas larutan Pu ditambah U nitrat.
Mehdi Zehtabian; Elham Piruzan; Zahra Molaiemanesh; Sedigheh Sina
2015-01-01
Introduction Lead-based shields are the most widely used attenuators in X-ray and gamma ray fields. The heavy weight, toxicity and corrosion of lead have led researchers towards the development of non-lead shields. Materials and Methods The purpose of this study was to design multi-layered shields for protection against X-rays and gamma rays in diagnostic radiology and nuclear medicine. In this study, cubic slabs composed of several materials with high atomic numbers, i.e., lead, barium, bism...
Adjei, Christian Amevi
The main objective of this thesis is twofold. The starting objective was to develop a model for meaningful benchmarking of different versions of GEANT4 against an experimental set-up and MCNP5 pertaining to photon transport and interactions. The following objective was to develop a preliminary design of a Fast Neutron Pencil Beam (FNPB) Facility to be applicable for the University of Utah research reactor (UUTR) using MCNP5 and GEANT4. The three various GEANT4 code versions, GEANT4.9.4, GEANT4.9.3, and GEANT4.9.2, were compared to MCNP5 and the experimental measurements of gamma attenuation in air. The average gamma dose rate was measured in the laboratory experiment at various distances from a shielded cesium source using a Ludlum model 19 portable NaI detector. As it was expected, the gamma dose rate decreased with distance. All three GEANT4 code versions agreed well with both the experimental data and the MCNP5 simulation. Additionally, a simple GEANT4 and MCNP5 model was developed to compare the code agreements for neutron interactions in various materials. Preliminary FNPB design was developed using MCNP5; a semi-accurate model was developed using GEANT4 (because GEANT4 does not support the reactor physics modeling, the reactor was represented as a surface neutron source, thus a semi-accurate model). Based on the MCNP5 model, the fast neutron flux in a sample holder of the FNPB is obtained to be 6.52×107 n/cm2s, which is one order of magnitude lower than gigantic fast neutron pencil beam facilities existing elsewhere. The MCNP5 model-based neutron spectrum indicates that the maximum expected fast neutron flux is at a neutron energy of ~1 MeV. In addition, the MCNP5 model provided information on gamma flux to be expected in this preliminary FNPB design; specifically, in the sample holder, the gamma flux is to be expected to be around 108 γ/cm 2s, delivering a gamma dose of 4.54×103 rem/hr. This value is one to two orders of magnitudes below the gamma
The MC21 Monte Carlo Transport Code
Energy Technology Data Exchange (ETDEWEB)
Sutton TM, Donovan TJ, Trumbull TH, Dobreff PS, Caro E, Griesheimer DP, Tyburski LJ, Carpenter DC, Joo H
2007-01-09
MC21 is a new Monte Carlo neutron and photon transport code currently under joint development at the Knolls Atomic Power Laboratory and the Bettis Atomic Power Laboratory. MC21 is the Monte Carlo transport kernel of the broader Common Monte Carlo Design Tool (CMCDT), which is also currently under development. The vision for CMCDT is to provide an automated, computer-aided modeling and post-processing environment integrated with a Monte Carlo solver that is optimized for reactor analysis. CMCDT represents a strategy to push the Monte Carlo method beyond its traditional role as a benchmarking tool or ''tool of last resort'' and into a dominant design role. This paper describes various aspects of the code, including the neutron physics and nuclear data treatments, the geometry representation, and the tally and depletion capabilities.
Monte Carlo simulation code modernization
CERN. Geneva
2015-01-01
The continual development of sophisticated transport simulation algorithms allows increasingly accurate description of the effect of the passage of particles through matter. This modelling capability finds applications in a large spectrum of fields from medicine to astrophysics, and of course HEP. These new capabilities however come at the cost of a greater computational intensity of the new models, which has the effect of increasing the demands of computing resources. This is particularly true for HEP, where the demand for more simulation are driven by the need of both more accuracy and more precision, i.e. better models and more events. Usually HEP has relied on the "Moore's law" evolution, but since almost ten years the increase in clock speed has withered and computing capacity comes in the form of hardware architectures of many-core or accelerated processors. To harness these opportunities we need to adapt our code to concurrent programming models taking advantages of both SIMD and SIMT architectures. Th...
Ralind Re Marla; Yohannes Sardjono; Supardi Supardi
2015-01-01
Telah dilakukan desain teras Pembangkit Listrik Tenaga Nuklir (PLTN) untuk jenis Pebble Bed Modular Reactor (PBMR) dengan daya 70 MWe untuk keperluan proses smelter pada keadaan beginning of life (BOL). Analisis ini bertujuan untuk mengetahui persen pengkayaan, distribusi suhu dan nilai keselamatan dengan koefisien reaktivitas teras yang negatif pada reaktor jenis PBMR apabila daya reaktor 70 MWe. Analisis menggunakan program Monte Carlo N-Particle-5 (MCNP5) dan dari hasil analisis ini dihara...
X-ray simulation with the Monte Carlo code PENELOPE. Application to Quality Control.
Pozuelo, F; Gallardo, S; Querol, A; Verdú, G; Ródenas, J
2012-01-01
A realistic knowledge of the energy spectrum is very important in Quality Control (QC) of X-ray tubes in order to reduce dose to patients. However, due to the implicit difficulties to measure the X-ray spectrum accurately, it is not normally obtained in routine QC. Instead, some parameters are measured and/or calculated. PENELOPE and MCNP5 codes, based on the Monte Carlo method, can be used as complementary tools to verify parameters measured in QC. These codes allow estimating Bremsstrahlung and characteristic lines from the anode taking into account specific characteristics of equipment. They have been applied to simulate an X-ray spectrum. Results are compared with theoretical IPEM 78 spectrum. A sensitivity analysis has been developed to estimate the influence on simulated spectra of important parameters used in simulation codes. With this analysis it has been obtained that the FORCE factor is the most important parameter in PENELOPE simulations. FORCE factor, which is a variance reduction method, improves the simulation but produces hard increases of computer time. The value of FORCE should be optimized so that a good agreement of simulated and theoretical spectra is reached, but with a reduction of computer time. Quality parameters such as Half Value Layer (HVL) can be obtained with the PENELOPE model developed, but FORCE takes such a high value that computer time is hardly increased. On the other hand, depth dose assessment can be achieved with acceptable results for small values of FORCE.
Morse Monte Carlo Radiation Transport Code System
Energy Technology Data Exchange (ETDEWEB)
Emmett, M.B.
1975-02-01
The report contains sections containing descriptions of the MORSE and PICTURE codes, input descriptions, sample problems, deviations of the physical equations and explanations of the various error messages. The MORSE code is a multipurpose neutron and gamma-ray transport Monte Carlo code. Time dependence for both shielding and criticality problems is provided. General three-dimensional geometry may be used with an albedo option available at any material surface. The PICTURE code provide aid in preparing correct input data for the combinatorial geometry package CG. It provides a printed view of arbitrary two-dimensional slices through the geometry. By inspecting these pictures one may determine if the geometry specified by the input cards is indeed the desired geometry. 23 refs. (WRF)
Monte Carlo based radial shield design of typical PWR reactor
Energy Technology Data Exchange (ETDEWEB)
Gul, Anas; Khan, Rustam; Qureshi, M. Ayub; Azeem, Muhammad Waqar; Raza, S.A. [Pakistan Institute of Engineering and Applied Sciences, Islamabad (Pakistan). Dept. of Nuclear Engineering; Stummer, Thomas [Technische Univ. Wien (Austria). Atominst.
2016-11-15
Neutron and gamma flux and dose equivalent rate distribution are analysed in radial and shields of a typical PWR type reactor based on the Monte Carlo radiation transport computer code MCNP5. The ENDF/B-VI continuous energy cross-section library has been employed for the criticality and shielding analysis. The computed results are in good agreement with the reference results (maximum difference is less than 56 %). It implies that MCNP5 a good tool for accurate prediction of neutron and gamma flux and dose rates in radial shield around the core of PWR type reactors.
Energy Technology Data Exchange (ETDEWEB)
Pozuelo, F.; Querol, A.; Juste, B.; Gallardo, S.; Rodenas, J.; Verdu, G.
2012-07-01
Obtaining the primary spectrum of X-rays to determine the quality of a photon beam produced by an X-ray tube, since the dosimetric characteristics of a radiation beam to have a direct relation to the primary X-ray spectrum. In this work are studied, the depth dose curves obtained in the energy range of diagnostic radiology, between 40 and 130 keV.
Nuclear reactions in Monte Carlo codes.
Ferrari, A; Sala, P R
2002-01-01
The physics foundations of hadronic interactions as implemented in most Monte Carlo codes are presented together with a few practical examples. The description of the relevant physics is presented schematically split into the major steps in order to stress the different approaches required for the full understanding of nuclear reactions at intermediate and high energies. Due to the complexity of the problem, only a few semi-qualitative arguments are developed in this paper. The description will be necessarily schematic and somewhat incomplete, but hopefully it will be useful for a first introduction into this topic. Examples are shown mostly for the high energy regime, where all mechanisms mentioned in the paper are at work and to which perhaps most of the readers are less accustomed. Examples for lower energies can be found in the references.
A GPU-based Monte Carlo dose calculation code for photon transport in a voxel phantom
Energy Technology Data Exchange (ETDEWEB)
Bellezzo, M.; Do Nascimento, E.; Yoriyaz, H., E-mail: mbellezzo@gmail.br [Instituto de Pesquisas Energeticas e Nucleares / CNEN, Av. Lineu Prestes 2242, Cidade Universitaria, 05508-000 Sao Paulo (Brazil)
2014-08-15
As the most accurate method to estimate absorbed dose in radiotherapy, Monte Carlo method has been widely used in radiotherapy treatment planning. Nevertheless, its efficiency can be improved for clinical routine applications. In this paper, we present the CUBMC code, a GPU-based Mc photon transport algorithm for dose calculation under the Compute Unified Device Architecture platform. The simulation of physical events is based on the algorithm used in Penelope, and the cross section table used is the one generated by the Material routine, als present in Penelope code. Photons are transported in voxel-based geometries with different compositions. To demonstrate the capabilities of the algorithm developed in the present work four 128 x 128 x 128 voxel phantoms have been considered. One of them is composed by a homogeneous water-based media, the second is composed by bone, the third is composed by lung and the fourth is composed by a heterogeneous bone and vacuum geometry. Simulations were done considering a 6 MeV monoenergetic photon point source. There are two distinct approaches that were used for transport simulation. The first of them forces the photon to stop at every voxel frontier, the second one is the Woodcock method, where the photon stop in the frontier will be considered depending on the material changing across the photon travel line. Dose calculations using these methods are compared for validation with Penelope and MCNP5 codes. Speed-up factors are compared using a NVidia GTX 560-Ti GPU card against a 2.27 GHz Intel Xeon CPU processor. (Author)
Energy Technology Data Exchange (ETDEWEB)
Querol, A.; Gallardo, S.; Rodenas, J.; Verdu, G.
2011-07-01
The characterization of the primary X-ray spectrum is a very useful tool for quality control of radiology equipment, as it allows a better estimation of the dose received by the patient, and better image quality. In this work, we have developed a Monte Carlo model with MCNP5 program to simulate X-ray spectra from an electron source and the characteristics of a commercial X-ray tube.
Daures, J; Gouriou, J; Bordy, J M
2011-03-01
This work has been performed within the frame of the European Union ORAMED project (Optimisation of RAdiation protection for MEDical staff). The main goal of the project is to improve standards of protection for medical staff for procedures resulting in potentially high exposures and to develop methodologies for better assessing and for reducing, exposures to medical staff. The Work Package WP2 is involved in the development of practical eye-lens dosimetry in interventional radiology. This study is complementary of the part of the ENEA report concerning the calculations with the MCNP-4C code of the conversion factors related to the operational quantity H(p)(3). In this study, a set of energy- and angular-dependent conversion coefficients (H(p)(3)/K(a)), in the newly proposed square cylindrical phantom made of ICRU tissue, have been calculated with the Monte-Carlo code PENELOPE and MCNP5. The H(p)(3) values have been determined in terms of absorbed dose, according to the definition of this quantity, and also with the kerma approximation as formerly reported in ICRU reports. At a low-photon energy (up to 1 MeV), the two results obtained with the two methods are consistent. Nevertheless, large differences are showed at a higher energy. This is mainly due to the lack of electronic equilibrium, especially for small angle incidences. The values of the conversion coefficients obtained with the MCNP-4C code published by ENEA quite agree with the kerma approximation calculations obtained with PENELOPE. We also performed the same calculations with the code MCNP5 with two types of tallies: F6 for kerma approximation and *F8 for estimating the absorbed dose that is, as known, due to secondary electrons. PENELOPE and MCNP5 results agree for the kerma approximation and for the absorbed dose calculation of H(p)(3) and prove that, for photon energies larger than 1 MeV, the transport of the secondary electrons has to be taken into account.
A Patch to MCNP5 for Multiplication Inference: Description and User Guide
Energy Technology Data Exchange (ETDEWEB)
Solomon, Jr., Clell J. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2014-05-05
A patch to MCNP5 has been written to allow generation of multiple neutrons from a spontaneous-fission event and generate list-mode output. This report documents the implementation and usage of this patch.
Criticality benchmarks validation of the Monte Carlo code TRIPOLI-2
Energy Technology Data Exchange (ETDEWEB)
Maubert, L. (Commissariat a l' Energie Atomique, Inst. de Protection et de Surete Nucleaire, Service d' Etudes de Criticite, 92 - Fontenay-aux-Roses (France)); Nouri, A. (Commissariat a l' Energie Atomique, Inst. de Protection et de Surete Nucleaire, Service d' Etudes de Criticite, 92 - Fontenay-aux-Roses (France)); Vergnaud, T. (Commissariat a l' Energie Atomique, Direction des Reacteurs Nucleaires, Service d' Etudes des Reacteurs et de Mathematique Appliquees, 91 - Gif-sur-Yvette (France))
1993-04-01
The three-dimensional energy pointwise Monte-Carlo code TRIPOLI-2 includes metallic spheres of uranium and plutonium, nitrate plutonium solutions, square and triangular pitch assemblies of uranium oxide. Results show good agreements between experiments and calculations, and avoid a part of the code and its ENDF-B4 library validation. (orig./DG)
An Overview of the Monte Carlo Methods, Codes, & Applications Group
Energy Technology Data Exchange (ETDEWEB)
Trahan, Travis John [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2016-08-30
This report sketches the work of the Group to deliver first-principle Monte Carlo methods, production quality codes, and radiation transport-based computational and experimental assessments using the codes MCNP and MCATK for such applications as criticality safety, non-proliferation, nuclear energy, nuclear threat reduction and response, radiation detection and measurement, radiation health protection, and stockpile stewardship.
Usage of burnt fuel isotopic compositions from engineering codes in Monte-Carlo code calculations
Energy Technology Data Exchange (ETDEWEB)
Aleshin, Sergey S.; Gorodkov, Sergey S.; Shcherenko, Anna I. [Nuclear Research Centre ' ' Kurchatov Institute' ' , Moscow (Russian Federation)
2015-09-15
A burn-up calculation of VVER's cores by Monte-Carlo code is complex process and requires large computational costs. This fact makes Monte-Carlo codes usage complicated for project and operating calculations. Previously prepared isotopic compositions are proposed to use for the Monte-Carlo code (MCU) calculations of different states of VVER's core with burnt fuel. Isotopic compositions are proposed to calculate by an approximation method. The approximation method is based on usage of a spectral functionality and reference isotopic compositions, that are calculated by engineering codes (TVS-M, PERMAK-A). The multiplication factors and power distributions of FA and VVER with infinite height are calculated in this work by the Monte-Carlo code MCU using earlier prepared isotopic compositions. The MCU calculation data were compared with the data which were obtained by engineering codes.
Energy Technology Data Exchange (ETDEWEB)
Both, J.P.; Lee, Y.K.; Mazzolo, A.; Peneliau, Y.; Petit, O.; Roesslinger, B. [CEA Saclay, Dir. de l' Energie Nucleaire (DEN), Service d' Etudes de Reacteurs et de Modelisation Avancee, 91 - Gif sur Yvette (France)
2003-07-01
Tripoli-4 is a three dimensional calculations code using the Monte Carlo method to simulate the transport of neutrons, photons, electrons and positrons. This code is used in four application fields: the protection studies, the criticality studies, the core studies and the instrumentation studies. Geometry, cross sections, description of sources, principle. (N.C.)
Parallelization of Monte Carlo codes MVP/GMVP
Energy Technology Data Exchange (ETDEWEB)
Nagaya, Yasunobu; Mori, Takamasa; Nakagawa, Masayuki [Japan Atomic Energy Research Inst., Tokai, Ibaraki (Japan). Tokai Research Establishment; Sasaki, Makoto
1998-03-01
General-purpose Monte Carlo codes MVP/GMVP are well-vectorized and thus enable us to perform high-speed Monte Carlo calculations. In order to achieve more speedups, we parallelized the codes on the different types of the parallel processing platforms. The platforms reported are a distributed-memory vector-parallel computer Fujitsu VPP500, a distributed-memory massively parallel computer Intel Paragon and a distributed-memory scalar-parallel computer Hitachi SR2201. As mentioned generally, ideal speedup could be obtained for large-scale problems but parallelization efficiency got worse as the batch size per a processing element (PE) was smaller. (author)
A Monte Carlo code for ion beam therapy
Anaïs Schaeffer
2012-01-01
Initially developed for applications in detector and accelerator physics, the modern Fluka Monte Carlo code is now used in many different areas of nuclear science. Over the last 25 years, the code has evolved to include new features, such as ion beam simulations. Given the growing use of these beams in cancer treatment, Fluka simulations are being used to design treatment plans in several hadron-therapy centres in Europe. Fluka calculates the dose distribution for a patient treated at CNAO with proton beams. The colour-bar displays the normalized dose values. Fluka is a Monte Carlo code that very accurately simulates electromagnetic and nuclear interactions in matter. In the 1990s, in collaboration with NASA, the code was developed to predict potential radiation hazards received by space crews during possible future trips to Mars. Over the years, it has become the standard tool to investigate beam-machine interactions, radiation damage and radioprotection issues in the CERN accelerator com...
Directory of Open Access Journals (Sweden)
P. Deatanyah
2012-03-01
Full Text Available The primary objective with radiation sources and facilities is the protection of both radiation workers and the general public. The biological shields of the Secondary Standard Dosimetry Laboratory of the Radiation Protection Institute (RPI Ghana had been evaluated for a collimated isotropic cesium-137 source for calibration purpose using MCNP5 code. The dose rate at supervised areas ranged from 0.57 to 8.35 :Sv/h and 0.26 to 10.22 :Sv/h at control areas when the source was panoramic. When the source was collimated, the dose rate ranged from 0.05 to 0.30 :Sv/h at supervised areas and 0.23 to 8.88 :Sv/h at control areas for 22.2 GBq of the cesium-137 source. The scatter contribution from the surfaces of the walls and roofs were also accounted for. The scatter radiation in the room decreased to 400 :Sv/h when the source was first collimated and to 3.5 :Sv/h when the source was further collimated. These results agreed quite well with experimental measurement. To effectively protect the staff, a narrow beam of 1.2 cm diameter which was defined at 1.0 m by the total surface of the ISO slab phantom was recommended to reduce the dose rate to less than 1.5 :Sv/h outside the calibration bunker even when the current activity is doubled. It was concluded that the 4.7 cm diameter of the existing narrow beam should be decreased to 1.2 cm by further collimation of the beam.
MCNP5 CALCULATIONS REPLICATING ARH-600 NITRATE DATA
Energy Technology Data Exchange (ETDEWEB)
FINFROCK SH
2011-10-25
This report serves to extend the previous document: 'MCNP Calculations Replicating ARH-600 Data' by replicating the nitrate curves found in ARH-600. This report includes the MCNP models used, the calculated critical dimension for each analyzed parameter set, and the resulting data libraries for use with the CritView code. As with the ARH-600 data, this report is not meant to replace the analysis of the fissile systems by qualified criticality personnel. The M CNP data is presented without accounting for the statistical uncertainty (although this is typically less than 0.001) or bias and, as such, the application of a reasonable safety margin is required. The data that follows pertains to the uranyl nitrate and plutonium nitrate spheres, infinite cylinders, and infinite slabs of varying isotopic composition, reflector thickness, and molarity. Each of the cases was modeled in MCNP (version 5.1.40), using the ENDF/B-VI cross section set. Given a molarity, isotopic composition, and reflector thickness, the fissile concentration and diameter (or thicknesses in the case of the slab geometries) were varied. The diameter for which k-effective equals 1.00 for a given concentration could then be calculated and graphed. These graphs are included in this report. The pages that follow describe the regions modeled, formulas for calculating the various parameters, a list of cross-sections used in the calculations, a description of the automation routine and data, and finally the data output. The data of most interest are the critical dimensions of the various systems analyzed. This is presented graphically, and in table format, in Appendix B. Appendix C provides a text listing of the same data in a format that is compatible with the CritView code. Appendices D and E provide listing of example Template files and MCNP input files (these are discussed further in Section 4). Appendix F is a complete listing of all of the output data (i.e., all of the analyzed dimensions and
A semianalytic Monte Carlo code for modelling LIDAR measurements
Palazzi, Elisa; Kostadinov, Ivan; Petritoli, Andrea; Ravegnani, Fabrizio; Bortoli, Daniele; Masieri, Samuele; Premuda, Margherita; Giovanelli, Giorgio
2007-10-01
LIDAR (LIght Detection and Ranging) is an optical active remote sensing technology with many applications in atmospheric physics. Modelling of LIDAR measurements appears useful approach for evaluating the effects of various environmental variables and scenarios as well as of different measurement geometries and instrumental characteristics. In this regard a Monte Carlo simulation model can provide a reliable answer to these important requirements. A semianalytic Monte Carlo code for modelling LIDAR measurements has been developed at ISAC-CNR. The backscattered laser signal detected by the LIDAR system is calculated in the code taking into account the contributions due to the main atmospheric molecular constituents and aerosol particles through processes of single and multiple scattering. The contributions by molecular absorption, ground and clouds reflection are evaluated too. The code can perform simulations of both monostatic and bistatic LIDAR systems. To enhance the efficiency of the Monte Carlo simulation, analytical estimates and expected value calculations are performed. Artificial devices (such as forced collision, local forced collision, splitting and russian roulette) are moreover foreseen by the code, which can enable the user to drastically reduce the variance of the calculation.
TRIPOLI-3: a neutron/photon Monte Carlo transport code
Energy Technology Data Exchange (ETDEWEB)
Nimal, J.C.; Vergnaud, T. [Commissariat a l' Energie Atomique, Gif-sur-Yvette (France). Service d' Etudes de Reacteurs et de Mathematiques Appliquees
2001-07-01
The present version of TRIPOLI-3 solves the transport equation for coupled neutron and gamma ray problems in three dimensional geometries by using the Monte Carlo method. This code is devoted both to shielding and criticality problems. The most important feature for particle transport equation solving is the fine treatment of the physical phenomena and sophisticated biasing technics useful for deep penetrations. The code is used either for shielding design studies or for reference and benchmark to validate cross sections. Neutronic studies are essentially cell or small core calculations and criticality problems. TRIPOLI-3 has been used as reference method, for example, for resonance self shielding qualification. (orig.)
SPAMCART: a code for smoothed particle Monte Carlo radiative transfer
Lomax, O.; Whitworth, A. P.
2016-10-01
We present a code for generating synthetic spectral energy distributions and intensity maps from smoothed particle hydrodynamics simulation snapshots. The code is based on the Lucy Monte Carlo radiative transfer method, i.e. it follows discrete luminosity packets as they propagate through a density field, and then uses their trajectories to compute the radiative equilibrium temperature of the ambient dust. The sources can be extended and/or embedded, and discrete and/or diffuse. The density is not mapped on to a grid, and therefore the calculation is performed at exactly the same resolution as the hydrodynamics. We present two example calculations using this method. First, we demonstrate that the code strictly adheres to Kirchhoff's law of radiation. Secondly, we present synthetic intensity maps and spectra of an embedded protostellar multiple system. The algorithm uses data structures that are already constructed for other purposes in modern particle codes. It is therefore relatively simple to implement.
SPAMCART: a code for smoothed particle Monte Carlo radiative transfer
Lomax, O
2016-01-01
We present a code for generating synthetic SEDs and intensity maps from Smoothed Particle Hydrodynamics simulation snapshots. The code is based on the Lucy (1999) Monte Carlo Radiative Transfer method, i.e. it follows discrete luminosity packets, emitted from external and/or embedded sources, as they propagate through a density field, and then uses their trajectories to compute the radiative equilibrium temperature of the ambient dust. The density is not mapped onto a grid, and therefore the calculation is performed at exactly the same resolution as the hydrodynamics. We present two example calculations using this method. First, we demonstrate that the code strictly adheres to Kirchhoff's law of radiation. Second, we present synthetic intensity maps and spectra of an embedded protostellar multiple system. The algorithm uses data structures that are already constructed for other purposes in modern particle codes. It is therefore relatively simple to implement.
Geometric Templates for Improved Tracking Performance in Monte Carlo Codes
Nease, Brian R.; Millman, David L.; Griesheimer, David P.; Gill, Daniel F.
2014-06-01
One of the most fundamental parts of a Monte Carlo code is its geometry kernel. This kernel not only affects particle tracking (i.e., run-time performance), but also shapes how users will input models and collect results for later analyses. A new framework based on geometric templates is proposed that optimizes performance (in terms of tracking speed and memory usage) and simplifies user input for large scale models. While some aspects of this approach currently exist in different Monte Carlo codes, the optimization aspect has not been investigated or applied. If Monte Carlo codes are to be realistically used for full core analysis and design, this type of optimization will be necessary. This paper describes the new approach and the implementation of two template types in MC21: a repeated ellipse template and a box template. Several different models are tested to highlight the performance gains that can be achieved using these templates. Though the exact gains are naturally problem dependent, results show that runtime and memory usage can be significantly reduced when using templates, even as problems reach realistic model sizes.
Applications guide to the MORSE Monte Carlo code
Energy Technology Data Exchange (ETDEWEB)
Cramer, S.N.
1985-08-01
A practical guide for the implementation of the MORESE-CG Monte Carlo radiation transport computer code system is presented. The various versions of the MORSE code are compared and contrasted, and the many references dealing explicitly with the MORSE-CG code are reviewed. The treatment of angular scattering is discussed, and procedures for obtaining increased differentiality of results in terms of reaction types and nuclides from a multigroup Monte Carlo code are explained in terms of cross-section and geometry data manipulation. Examples of standard cross-section data input and output are shown. Many other features of the code system are also reviewed, including (1) the concept of primary and secondary particles, (2) fission neutron generation, (3) albedo data capability, (4) DOMINO coupling, (5) history file use for post-processing of results, (6) adjoint mode operation, (7) variance reduction, and (8) input/output. In addition, examples of the combinatorial geometry are given, and the new array of arrays geometry feature (MARS) and its three-dimensional plotting code (JUNEBUG) are presented. Realistic examples of user routines for source, estimation, path-length stretching, and cross-section data manipulation are given. A deatiled explanation of the coupling between the random walk and estimation procedure is given in terms of both code parameters and physical analogies. The operation of the code in the adjoint mode is covered extensively. The basic concepts of adjoint theory and dimensionality are discussed and examples of adjoint source and estimator user routines are given for all common situations. Adjoint source normalization is explained, a few sample problems are given, and the concept of obtaining forward differential results from adjoint calculations is covered. Finally, the documentation of the standard MORSE-CG sample problem package is reviewed and on-going and future work is discussed.
Proton therapy Monte Carlo SRNA-VOX code
Directory of Open Access Journals (Sweden)
Ilić Radovan D.
2012-01-01
Full Text Available The most powerful feature of the Monte Carlo method is the possibility of simulating all individual particle interactions in three dimensions and performing numerical experiments with a preset error. These facts were the motivation behind the development of a general-purpose Monte Carlo SRNA program for proton transport simulation in technical systems described by standard geometrical forms (plane, sphere, cone, cylinder, cube. Some of the possible applications of the SRNA program are: (a a general code for proton transport modeling, (b design of accelerator-driven systems, (c simulation of proton scattering and degrading shapes and composition, (d research on proton detectors; and (e radiation protection at accelerator installations. This wide range of possible applications of the program demands the development of various versions of SRNA-VOX codes for proton transport modeling in voxelized geometries and has, finally, resulted in the ISTAR package for the calculation of deposited energy distribution in patients on the basis of CT data in radiotherapy. All of the said codes are capable of using 3-D proton sources with an arbitrary energy spectrum in an interval of 100 keV to 250 MeV.
Validation of the Monte Carlo code MCNP-DSP
Energy Technology Data Exchange (ETDEWEB)
Valentine, T.E.; Mihalczo, J.T. [Oak Ridge National Lab., TN (United States)
1996-09-12
Several calculations were performed to validate MCNP-DSP, which is a Monte Carlo code that calculates all the time and frequency analysis parameters associated with the {sup 252}Cf-source-driven time and frequency analysis method. The frequency analysis parameters are obtained in two ways: directly by Fourier transforming the detector responses and indirectly by taking the Fourier transform of the autocorrelation and cross-correlation functions. The direct and indirect Fourier processing methods were shown to produce the same frequency spectra and convergence, thus verifying the way to obtain the frequency analysis parameters from the time sequences of detector pulses. (Author).
Computed radiography simulation using the Monte Carlo code MCNPX
Energy Technology Data Exchange (ETDEWEB)
Correa, S.C.A. [Programa de Engenharia Nuclear/COPPE, Universidade Federal do Rio de Janeiro, Ilha do Fundao, Caixa Postal 68509, 21945-970, Rio de Janeiro, RJ (Brazil); Centro Universitario Estadual da Zona Oeste (CCMAT)/UEZO, Av. Manuel Caldeira de Alvarenga, 1203, Campo Grande, 23070-200, Rio de Janeiro, RJ (Brazil); Souza, E.M. [Programa de Engenharia Nuclear/COPPE, Universidade Federal do Rio de Janeiro, Ilha do Fundao, Caixa Postal 68509, 21945-970, Rio de Janeiro, RJ (Brazil); Silva, A.X., E-mail: ademir@con.ufrj.b [PEN/COPPE-DNC/Poli CT, Universidade Federal do Rio de Janeiro, Ilha do Fundao, Caixa Postal 68509, 21945-970, Rio de Janeiro, RJ (Brazil); Cassiano, D.H. [Instituto de Radioprotecao e Dosimetria/CNEN Av. Salvador Allende, s/n, Recreio, 22780-160, Rio de Janeiro, RJ (Brazil); Lopes, R.T. [Programa de Engenharia Nuclear/COPPE, Universidade Federal do Rio de Janeiro, Ilha do Fundao, Caixa Postal 68509, 21945-970, Rio de Janeiro, RJ (Brazil)
2010-09-15
Simulating X-ray images has been of great interest in recent years as it makes possible an analysis of how X-ray images are affected owing to relevant operating parameters. In this paper, a procedure for simulating computed radiographic images using the Monte Carlo code MCNPX is proposed. The sensitivity curve of the BaFBr image plate detector as well as the characteristic noise of a 16-bit computed radiography system were considered during the methodology's development. The results obtained confirm that the proposed procedure for simulating computed radiographic images is satisfactory, as it allows obtaining results comparable with experimental data.
Monte Carlo calculation for the development of a BNCT neutron source (1eV-10KeV) using MCNP code.
El Moussaoui, F; El Bardouni, T; Azahra, M; Kamili, A; Boukhal, H
2008-09-01
Different materials have been studied in order to produce the epithermal neutron beam between 1eV and 10KeV, which are extensively used to irradiate patients with brain tumors such as GBM. For this purpose, we have studied three different neutrons moderators (H(2)O, D(2)O and BeO) and their combinations, four reflectors (Al(2)O(3), C, Bi, and Pb) and two filters (Cd and Bi). Results of calculation showed that the best obtained assembly configuration corresponds to the combination of the three moderators H(2)O, BeO and D(2)O jointly to Al(2)O(3) reflector and two filter Cd+Bi optimize the spectrum of the epithermal neutron at 72%, and minimize the thermal neutron to 4% and thus it can be used to treat the deep tumor brain. The calculations have been performed by means of the Monte Carlo N (particle code MCNP 5C). Our results strongly encourage further studying of irradiation of the head with epithermal neutron fields.
Parallelization of a Monte Carlo particle transport simulation code
Hadjidoukas, P.; Bousis, C.; Emfietzoglou, D.
2010-05-01
We have developed a high performance version of the Monte Carlo particle transport simulation code MC4. The original application code, developed in Visual Basic for Applications (VBA) for Microsoft Excel, was first rewritten in the C programming language for improving code portability. Several pseudo-random number generators have been also integrated and studied. The new MC4 version was then parallelized for shared and distributed-memory multiprocessor systems using the Message Passing Interface. Two parallel pseudo-random number generator libraries (SPRNG and DCMT) have been seamlessly integrated. The performance speedup of parallel MC4 has been studied on a variety of parallel computing architectures including an Intel Xeon server with 4 dual-core processors, a Sun cluster consisting of 16 nodes of 2 dual-core AMD Opteron processors and a 200 dual-processor HP cluster. For large problem size, which is limited only by the physical memory of the multiprocessor server, the speedup results are almost linear on all systems. We have validated the parallel implementation against the serial VBA and C implementations using the same random number generator. Our experimental results on the transport and energy loss of electrons in a water medium show that the serial and parallel codes are equivalent in accuracy. The present improvements allow for studying of higher particle energies with the use of more accurate physical models, and improve statistics as more particles tracks can be simulated in low response time.
A comparison between the Monte Carlo radiation transport codes MCNP and MCBEND
Energy Technology Data Exchange (ETDEWEB)
Sawamura, Hidenori; Nishimura, Kazuya [Computer Software Development Co., Ltd., Tokyo (Japan)
2001-01-01
In Japan, almost of all radiation analysts are using the MCNP code and MVP code on there studies. But these codes have not had automatic variance reduction. MCBEND code made by UKAEA have automatic variance reduction. And, MCBEND code is user friendly more than other Monte Carlo Radiation Transport Codes. Our company was first introduced MCBEND code in Japan. Therefore, we compared with MCBEND code and MCNP code about functions and production capacity. (author)
Firstenberg, H.
1971-01-01
The statistics are considered of the Monte Carlo method relative to the interpretation of the NUGAM2 and NUGAM3 computer code results. A numerical experiment using the NUGAM2 code is presented and the results are statistically interpreted.
The Monte Carlo code MCSHAPE: Main features and recent developments
Energy Technology Data Exchange (ETDEWEB)
Scot, Viviana, E-mail: viviana.scot@unibo.it; Fernandez, Jorge E.
2015-06-01
MCSHAPE is a general purpose Monte Carlo code developed at the University of Bologna to simulate the diffusion of X- and gamma-ray photons with the special feature of describing the full evolution of the photon polarization state along the interactions with the target. The prevailing photon–matter interactions in the energy range 1–1000 keV, Compton and Rayleigh scattering and photoelectric effect, are considered. All the parameters that characterize the photon transport can be suitably defined: (i) the source intensity, (ii) its full polarization state as a function of energy, (iii) the number of collisions, and (iv) the energy interval and resolution of the simulation. It is possible to visualize the results for selected groups of interactions. MCSHAPE simulates the propagation in heterogeneous media of polarized photons (from synchrotron sources) or of partially polarized sources (from X-ray tubes). In this paper, the main features of MCSHAPE are illustrated with some examples and a comparison with experimental data. - Highlights: • MCSHAPE is an MC code for the simulation of the diffusion of photons in the matter. • It includes the proper description of the evolution of the photon polarization state. • The polarization state is described by means of the Stokes vector, I, Q, U, V. • MCSHAPE includes the computation of the detector influence in the measured spectrum. • MCSHAPE features are illustrated with examples and comparison with experiments.
Comparison of KENO-VI and MCNP5 Criticality Analyses for a Lunar Regolith Clustered-Reactor System
Bess, John Darrell
2008-01-01
The Lunar Regolith Clustered-Reactor System design has been presented as an alternative method for providing surface power to a lunar facility using a fast-fission, heatpipe-cooled nuclear reactor. The reactor system is divided into subcritical units that can be safely launched into orbit without risk of inadvertent criticality in the event of a launch accident. The reactor subunits are emplaced into the lunar surface to form a clustered-reactor system, utilizing the regolith as both radiation shielding and neutron-reflector material. Coordinated placement of multiple subunits can provision a critical reactor system proportional to localized lunar surface power demand. Reactor units assembled using proven and tested materials in radiation environments such as UO2 fuel, stainless-steel cladding and support, and compatible liquid-metal heatpipes promote safety and reliability, with ease of manufacture and testing. Reactor power levels of approximately 100 kWth per subunit significantly reduces the negative effects of elevated temperature and radiation environments associated with single nuclear power reactors operated at higher power levels. The analysis of subunit criticality in various accident scenarios differs by up to 4% (~$6 in reactivity) between results generated using conventional criticality analysis codes, MCNP5 and KENO-VI. A demonstrated trend exists between results of the two criticality codes as accident conditions approach a multiplication factor of one. Code comparison of a tri-cluster system on the lunar surface provides comparable results with calculated system reactivity within 0.5%. Iron concentration is confirmed as the dominant element in the lunar regolith influencing system reactivity.
Directory of Open Access Journals (Sweden)
Ralind Re Marla
2015-03-01
Full Text Available Telah dilakukan desain teras Pembangkit Listrik Tenaga Nuklir (PLTN untuk jenis Pebble Bed Modular Reactor (PBMR dengan daya 70 MWe untuk keperluan proses smelter pada keadaan beginning of life (BOL. Analisis ini bertujuan untuk mengetahui persen pengkayaan, distribusi suhu dan nilai keselamatan dengan koefisien reaktivitas teras yang negatif pada reaktor jenis PBMR apabila daya reaktor 70 MWe. Analisis menggunakan program Monte Carlo N-Particle-5 (MCNP5 dan dari hasil analisis ini diharapkan dapat memenuhi syarat dalam mendukung program percepatan pembangunan kelistrikan batubara 10.000 MWe khususnya untuk proses smelter, yang tersebar merata di wilayah Indonesia. Hasil penelitian menunjukkan bahwa, faktor perlipatan efektif (k-eff Reaktor jenis PBMR daya 70 MWe mengalami kondisi kritis pada pengkayaan 5,626 % dengan nilai faktor perlipatan efektif 1,00031±0,00087 dan nilai koefisien reaktivitas suhu pada -10,0006 pcm/K. Dari hasil analisis daat disimpulkan bahwa reaktor jenis PBMR daya 70 MWe adalah aman. ABSTRACT The core design of Nuclear Power Plant for Pebble Bed Modular Reactor (PBMR type with 70 MWe capacity power in Beginning of Life (BOL has been performed. The aim of this analysis, to know percent enrichment, temperature distribution and safety value by negative temperature coefficient at type PBMR if reactor power become lower equal to 70 MWe. This analysis was expected become one part of overview project development the power plant with 10.000 MWe of total capacity, spread evenly in territory of Indonesia especially to support of smelter industries. The results showed that, effective multiplication factor (keff with power 70 MWe critical condition at enrichment 5,626 %is 1,00031±0,00087, based on enrichment result, a value of the temperature coefficient reactivity is - 10,0006 pcm/K. Based on the results of these studies, it can beconcluded that the PBMR 70 MWe design is theoritically safe.
Qin, Jianguo; Liu, Rong; Zhu, Tonghua; Zhang, Xinwei; Ye, Bangjiao
2015-01-01
To overcome the problem of inefficient computing time and unreliable results in MCNP5 calculation, a two-step method is adopted to calculate the energy deposition of prompt gamma-rays in detectors for depleted uranium spherical shells under D-T neutrons irradiation. In the first step, the gamma-ray spectrum for energy below 7 MeV is calculated by MCNP5 code; secondly, the electron recoil spectrum in a BC501A liquid scintillator detector is simulated based on EGSnrc Monte Carlo Code with the gamma-ray spectrum from the first step as input. The comparison of calculated results with experimental ones shows that the simulations agree well with experiment in the energy region 0.4-3 MeV for the prompt gamma-ray spectrum and below 4 MeVee for the electron recoil spectrum. The reliability of the two-step method in this work is validated.
Energy Technology Data Exchange (ETDEWEB)
NONE
2001-01-01
In the report, research results discussed in 1999 fiscal year at Nuclear Code Evaluation Committee of Nuclear Code Research Committee were summarized. Present status of Monte Carlo simulation on nuclear energy study was described. Especially, besides of criticality, shielding and core analyses, present status of applications to risk and radiation damage analyses, high energy transport and nuclear theory calculations of Monte Carlo Method was described. The 18 papers are indexed individually. (J.P.N.)
TART97 a coupled neutron-photon 3-D, combinatorial geometry Monte Carlo transport code
Energy Technology Data Exchange (ETDEWEB)
Cullen, D.E.
1997-11-22
TART97 is a coupled neutron-photon, 3 Dimensional, combinatorial geometry, time dependent Monte Carlo transport code. This code can on any modern computer. It is a complete system to assist you with input preparation, running Monte Carlo calculations, and analysis of output results. TART97 is also incredibly FAST; if you have used similar codes, you will be amazed at how fast this code is compared to other similar codes. Use of the entire system can save you a great deal of time and energy. TART97 is distributed on CD. This CD contains on- line documentation for all codes included in the system, the codes configured to run on a variety of computers, and many example problems that you can use to familiarize yourself with the system. TART97 completely supersedes all older versions of TART, and it is strongly recommended that users only use the most recent version of TART97 and its data riles.
Data libraries as a collaborative tool across Monte Carlo codes
Augelli, Mauro; Han, Mincheol; Hauf, Steffen; Kim, Chan-Hyeung; Kuster, Markus; Pia, Maria Grazia; Quintieri, Lina; Saracco, Paolo; Seo, Hee; Sudhakar, Manju; Eidenspointner, Georg; Zoglauer, Andreas
2010-01-01
The role of data libraries in Monte Carlo simulation is discussed. A number of data libraries currently in preparation are reviewed; their data are critically examined with respect to the state-of-the-art in the respective fields. Extensive tests with respect to experimental data have been performed for the validation of their content.
Energy Technology Data Exchange (ETDEWEB)
ElAgib, I. [College of Science, King Saud University, P.O. Box 2455 (Saudi Arabia)], E-mail: elagib@ksu.edu.sa; Elsheikh, N. [College of Applied and Industrial Science, University of Juba, Khartoum, P.O. Box 321 (Sudan); AlSewaidan, H. [College of Science, King Saud University, P.O. Box 2455 (Saudi Arabia); Habbani, F. [Faculty of Science, Physics Department, University of Khartoum, Khartoum, P.O. Box 321 (Sudan)
2009-01-15
Calculations of elastically backscattered (EBS) neutrons from hidden explosives buried in soil were performed using Monte-Carlo N-particle transport code MCNP5. Three different neutron sources were used in the study. The study re-examines the performance of the neutron backscattering methods in providing identification of hidden explosives through their chemical composition. The EBS neutron energy spectra of fast and slow neutrons of the major constituent elements in soil and an explosive material in form of TNT have shown definite structures that can be used for the identification of a buried landmine.
Study of an extrapolation chamber in a standard diagnostic radiology beam by Monte Carlo simulation
Energy Technology Data Exchange (ETDEWEB)
Vedovato, Uly Pita; Silva, Rayre Janaina Vieira; Neves, Lucio Pereira; Santos, William S.; Perini, Ana Paula, E-mail: anapaula.perini@ufu.br [Universidade Federal de Uberlandia (INFIS/UFU), MG (Brazil). Instituto de Fisica; Caldas, Linda V.E. [Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil); Belinato, Walmir [Instituto Federal de Educacao, Ciencia e Tecnologia da Bahia (IFBA), Vitoria da Conquista, BA (Brazil)
2016-07-01
In this work, we studied the influence of the components of an extrapolation ionization chamber in its response. This study was undertaken using the MCNP-5 Monte Carlo code, and the standard diagnostic radiology quality for direct beams (RQR5). Using tally F6 and 2.1 x 10{sup 9} simulated histories, the results showed that the chamber design and material not alter significantly the energy deposited in its sensitive volume. The collecting electrode and support board were the components with more influence on the chamber response. (author)
Energy Technology Data Exchange (ETDEWEB)
Nimal, J.C.; Vergnaud, T. (CEA Centre d' Etudes Nucleaires de Saclay, 91 - Gif-sur-Yvette (France))
1990-01-01
This paper describes the most important features of the Monte Carlo code TRIPOLI-2. This code solves the Boltzmann equation in three-dimensional geometries for coupled neutron and gamma rays problems. A particular emphasis is devoted to the biasing techniques, which are very important for deep penetration. Future developments in TRIPOLI are described in the conclusion. (author).
FREYA-a new Monte Carlo code for improved modeling of fission chains
Energy Technology Data Exchange (ETDEWEB)
Hagmann, C A; Randrup, J; Vogt, R L
2012-06-12
A new simulation capability for modeling of individual fission events and chains and the transport of fission products in materials is presented. FREYA ( Fission Yield Event Yield Algorithm ) is a Monte Carlo code for generating fission events providing correlated kinematic information for prompt neutrons, gammas, and fragments. As a standalone code, FREYA calculates quantities such as multiplicity-energy, angular, and gamma-neutron energy sharing correlations. To study materials with multiplication, shielding effects, and detectors, we have integrated FREYA into the general purpose Monte Carlo code MCNP. This new tool will allow more accurate modeling of detector responses including correlations and the development of SNM detectors with increased sensitivity.
Energy Technology Data Exchange (ETDEWEB)
Gil, Alex Vergara [Centro de Proteccion e Higiene de las Radiaciones (CPHR), La Habana (Cuba); Perez, Marco A. Coca; Aroche, Leonel A. Torres, E-mail: mcoca@infomed.sld.cu, E-mail: leonel@infomed.sld.cu [Centro de Investigaciones Clinicas (CIC), La Habana (Cuba); Pacilio, Massimiliano, E-mail: mpacilio@scamilloforlanini.rm.it [Hospital S. Camillo Forlanini (AOSCF), Roma (Italy). Departmento de Fisica Medica
2013-07-01
The purpose of this paper is to present the MCID software, a tool for calculating specific absorbed dose of patients in nuclear medicine, based on Monte Carlo simulation. This paper evaluates new clinical cases and new phantoms whose results validate the methodology implemented in MCID, which has followed a process of incorporating new materials, image processing in DICOM and Analyze format, a module of regions of interest and improvements in user interface. Now it has a tool to calculate the patient-specific absorbed doses in nuclear medicine that can be applied in clinical practice.
Progress and status of the OpenMC Monte Carlo code
Energy Technology Data Exchange (ETDEWEB)
Romano, P. K.; Herman, B. R.; Horelik, N. E.; Forget, B.; Smith, K. [Massachusetts Institute of Technology, Department of Nuclear Science and Engineering, 77 Massachusetts Avenue, Cambridge, MA 02139 (United States); Siegel, A. R. [Argonne National Laboratory, Theory and Computing Sciences and Nuclear Engineering Division (United States)
2013-07-01
The present work describes the latest advances and progress in the development of the OpenMC Monte Carlo code, an open-source code originating from the Massachusetts Institute of Technology. First, an overview of the development workflow of OpenMC is given. Various enhancements to the code such as real-time XML input validation, state points, plotting, OpenMP threading, and coarse mesh finite difference acceleration are described. (authors)
DEFF Research Database (Denmark)
Taasti, Vicki Trier; Knudsen, Helge; Holzscheiter, Michael
2015-01-01
The Monte Carlo particle transport code SHIELD-HIT12A is designed to simulate therapeutic beams for cancer radiotherapy with fast ions. SHIELD-HIT12A allows creation of antiproton beam kernels for the treatment planning system TRiP98, but first it must be benchmarked against experimental data...
Neutral Particle Transport in Cylindrical Plasma Simulated by a Monte Carlo Code
Institute of Scientific and Technical Information of China (English)
YU Deliang; YAN Longwen; ZHONG Guangwu; LU Jie; YI Ping
2007-01-01
A Monte Carlo code (MCHGAS) has been developed to investigate the neutral particle transport.The code can calculate the radial profile and energy spectrum of neutral particles in cylindrical plasmas.The calculation time of the code is dramatically reduced when the Splitting and Roulette schemes are applied. The plasma model of an infinite cylinder is assumed in the code,which is very convenient in simulating neutral particle transports in small and middle-sized tokamaks.The design of the multi-channel neutral particle analyser (NPA) on HL-2A can be optimized by using this code.
Brown, Thomas A D; Alvarez, Diane; Matthews, Kenneth L; Ham, Kyungmin; 10.1118/1.4761870
2012-01-01
Ion chamber dosimetry is being used to calibrate dose for cell irradiations designed to investigate photoactivated Auger electron therapy at the Louisiana State University CAMD synchrotron facility. This study performed a dosimetry intercomparison for synchrotron-produced monochromatic x-ray beams at 25 and 35 keV. Ion chamber depth-dose measurements in a PMMA phantom were compared with the product of MCNP5 Monte Carlo calculations of dose per fluence and measured incident fluence. Monochromatic beams of 25 and 35 keV were generated on the tomography beamline at CAMD. A cylindrical, air-equivalent ion chamber was used to measure the ionization created in a 10x10x10-cm3 PMMA phantom for depths from 0.6 to 7.7 cm. The American Association of Physicists in Medicine TG-61 protocol was applied to convert measured ionization into dose. Photon fluence was determined using a NaI detector to make scattering measurements of the beam from a thin polyethylene target at angles 30 degrees to 60 degrees. Differential Compto...
Directory of Open Access Journals (Sweden)
Zeynab Fazli
2013-01-01
Full Text Available Purpose: For the treatment of nasopharnx carcinoma (NPC using brachytherapy methods and high-energy photon sources are common techniques. In the common three dimensional (3D treatments planning, all of the computed tomography images are assumed homogeneous. This study presents the results of Monte Carlo calculations for non-homogeneous nasopharynx phantom, MAGICA normoxic gel dosimetry and 3D treatment planning system (TPS. Materials and Methods: The head phantom was designed with Plexiglas cylinder, head bone, and nasopharynx brachytherapy silicon applicator. For the simulations, version 5 of the Monte Carlo N-particle transport code (MCNP5 was used. 3D treatment planning was performed in Flexiplan software. A normoxic radiosensitive polymer gel was fabricated under normal atmospheric conditions and poured into test tubes (for calibration curve and the head phantom. In addition, the head phantom was irradiated with Flexitron afterloader brachytherapy machine with 192 Ir source. To obtain calibration curves, 11 dosimeters were irradiated with dose range of 0-2000 cGy. Evaluations of dosimeters were performed on 1.5T scanner. Results: Two-dimensional iso-dose in coronal plan at distances of z = +0.3, –0.3 cm was calculated. There was a good accordance between 3D TPS and MCNP5 simulation and differences in various distances were between 2.4% and 6.1%. There was a predictable accordance between MAGICA gel dosimetry and MCNP5 simulation and differences in various distances were between 5.7% and 7.4%. Moreover, there was an acceptable accordance between MAGICA gel dosimetry and MCNP5 data and differences in various distances were between 5.2% and 9.4%. Conclusion: The sources of differences in this comparison are divided to calculations variation and practical errors that was added in experimental dosimetry. The result of quality assurance of nasopharynx high dose rate brachytherapy is consistent with international standards.
Fazli, Zeynab; Sadeghi, Mahdi; Zahmatkesh, M H; Mahdavi, Seied Rabei; Tenreiro, Claudio
2013-01-01
For the treatment of nasopharnx carcinoma (NPC) using brachytherapy methods and high-energy photon sources are common techniques. In the common three dimensional (3D) treatments planning, all of the computed tomography images are assumed homogeneous. This study presents the results of Monte Carlo calculations for non-homogeneous nasopharynx phantom, MAGICA normoxic gel dosimetry and 3D treatment planning system (TPS). The head phantom was designed with Plexiglas cylinder, head bone, and nasopharynx brachytherapy silicon applicator. For the simulations, version 5 of the Monte Carlo N-particle transport code (MCNP5) was used. 3D treatment planning was performed in Flexiplan software. A normoxic radiosensitive polymer gel was fabricated under normal atmospheric conditions and poured into test tubes (for calibration curve) and the head phantom. In addition, the head phantom was irradiated with Flexitron afterloader brachytherapy machine with (192)Ir source. To obtain calibration curves, 11 dosimeters were irradiated with dose range of 0-2000 cGy. Evaluations of dosimeters were performed on 1.5T scanner. Two-dimensional iso-dose in coronal plan at distances of z = +0.3, -0.3 cm was calculated. There was a good accordance between 3D TPS and MCNP5 simulation and differences in various distances were between 2.4% and 6.1%. There was a predictable accordance between MAGICA gel dosimetry and MCNP5 simulation and differences in various distances were between 5.7% and 7.4%. Moreover, there was an acceptable accordance between MAGICA gel dosimetry and MCNP5 data and differences in various distances were between 5.2% and 9.4%. The sources of differences in this comparison are divided to calculations variation and practical errors that was added in experimental dosimetry. The result of quality assurance of nasopharynx high dose rate brachytherapy is consistent with international standards.
Longitudinal development of extensive air showers: hybrid code SENECA and full Monte Carlo
Ortiz, J A; De Souza, V; Ortiz, Jeferson A.; Tanco, Gustavo Medina
2004-01-01
New experiments, exploring the ultra-high energy tail of the cosmic ray spectrum with unprecedented detail, are exerting a severe pressure on extensive air hower modeling. Detailed fast codes are in need in order to extract and understand the richness of information now available. Some hybrid simulation codes have been proposed recently to this effect (e.g., the combination of the traditional Monte Carlo scheme and system of cascade equations or pre-simulated air showers). In this context, we explore the potential of SENECA, an efficient hybrid tridimensional simulation code, as a valid practical alternative to full Monte Carlo simulations of extensive air showers generated by ultra-high energy cosmic rays. We extensively compare hybrid method with the traditional, but time consuming, full Monte Carlo code CORSIKA which is the de facto standard in the field. The hybrid scheme of the SENECA code is based on the simulation of each particle with the traditional Monte Carlo method at two steps of the shower devel...
Energy Technology Data Exchange (ETDEWEB)
Liang, Jingang; Wang, Kan; Qiu, Yishu [Dept. of Engineering Physics, LiuQing Building, Tsinghua University, Beijing (China); Chai, Xiao Ming; Qiang, Sheng Long [Science and Technology on Reactor System Design Technology Laboratory, Nuclear Power Institute of China, Chengdu (China)
2016-06-15
Because of prohibitive data storage requirements in large-scale simulations, the memory problem is an obstacle for Monte Carlo (MC) codes in accomplishing pin-wise three-dimensional (3D) full-core calculations, particularly for whole-core depletion analyses. Various kinds of data are evaluated and quantificational total memory requirements are analyzed based on the Reactor Monte Carlo (RMC) code, showing that tally data, material data, and isotope densities in depletion are three major parts of memory storage. The domain decomposition method is investigated as a means of saving memory, by dividing spatial geometry into domains that are simulated separately by parallel processors. For the validity of particle tracking during transport simulations, particles need to be communicated between domains. In consideration of efficiency, an asynchronous particle communication algorithm is designed and implemented. Furthermore, we couple the domain decomposition method with MC burnup process, under a strategy of utilizing consistent domain partition in both transport and depletion modules. A numerical test of 3D full-core burnup calculations is carried out, indicating that the RMC code, with the domain decomposition method, is capable of pin-wise full-core burnup calculations with millions of depletion regions.
Energy Technology Data Exchange (ETDEWEB)
Perfetti, Christopher M [ORNL; Martin, William R [University of Michigan; Rearden, Bradley T [ORNL; Williams, Mark L [ORNL
2012-01-01
Three methods for calculating continuous-energy eigenvalue sensitivity coefficients were developed and implemented into the SHIFT Monte Carlo code within the Scale code package. The methods were used for several simple test problems and were evaluated in terms of speed, accuracy, efficiency, and memory requirements. A promising new method for calculating eigenvalue sensitivity coefficients, known as the CLUTCH method, was developed and produced accurate sensitivity coefficients with figures of merit that were several orders of magnitude larger than those from existing methods.
Longitudinal development of extensive air showers: Hybrid code SENECA and full Monte Carlo
Ortiz, Jeferson A.; Medina-Tanco, Gustavo; de Souza, Vitor
2005-06-01
New experiments, exploring the ultra-high energy tail of the cosmic ray spectrum with unprecedented detail, are exerting a severe pressure on extensive air shower modelling. Detailed fast codes are in need in order to extract and understand the richness of information now available. Some hybrid simulation codes have been proposed recently to this effect (e.g., the combination of the traditional Monte Carlo scheme and system of cascade equations or pre-simulated air showers). In this context, we explore the potential of SENECA, an efficient hybrid tri-dimensional simulation code, as a valid practical alternative to full Monte Carlo simulations of extensive air showers generated by ultra-high energy cosmic rays. We extensively compare hybrid method with the traditional, but time consuming, full Monte Carlo code CORSIKA which is the de facto standard in the field. The hybrid scheme of the SENECA code is based on the simulation of each particle with the traditional Monte Carlo method at two steps of the shower development: the first step predicts the large fluctuations in the very first particle interactions at high energies while the second step provides a well detailed lateral distribution simulation of the final stages of the air shower. Both Monte Carlo simulation steps are connected by a cascade equation system which reproduces correctly the hadronic and electromagnetic longitudinal profile. We study the influence of this approach on the main longitudinal characteristics of proton, iron nucleus and gamma induced air showers and compare the predictions of the well known CORSIKA code using the QGSJET hadronic interaction model.
NRMC - A GPU code for N-Reverse Monte Carlo modeling of fluids in confined media
Sánchez-Gil, Vicente; Noya, Eva G.; Lomba, Enrique
2017-08-01
NRMC is a parallel code for performing N-Reverse Monte Carlo modeling of fluids in confined media [V. Sánchez-Gil, E.G. Noya, E. Lomba, J. Chem. Phys. 140 (2014) 024504]. This method is an extension of the usual Reverse Monte Carlo method to obtain structural models of confined fluids compatible with experimental diffraction patterns, specifically designed to overcome the problem of slow diffusion that can appear under conditions of tight confinement. Most of the computational time in N-Reverse Monte Carlo modeling is spent in the evaluation of the structure factor for each trial configuration, a calculation that can be easily parallelized. Implementation of the structure factor evaluation in NVIDIA® CUDA so that the code can be run on GPUs leads to a speed up of up to two orders of magnitude.
DEFF Research Database (Denmark)
Taasti, Vicki Trier; Knudsen, Helge; Holzscheiter, Michael
2015-01-01
The Monte Carlo particle transport code SHIELD-HIT12A is designed to simulate therapeutic beams for cancer radiotherapy with fast ions. SHIELD-HIT12A allows creation of antiproton beam kernels for the treatment planning system TRiP98, but first it must be benchmarked against experimental data. An...... conclude that more experimental cross section data are needed in the lower energy range in order to resolve this contradiction, ideally combined with more rigorous models for annihilation on compounds....
Homma, Yuto; Moriwaki, Hiroyuki; Ohki, Shigeo; Ikeda, Kazumi
2014-06-01
This paper deals with verification of three dimensional triangular prismatic discrete ordinates transport calculation code ENSEMBLE-TRIZ by comparison with multi-group Monte Carlo calculation code GMVP in a large fast breeder reactor. The reactor is a 750 MWe electric power sodium cooled reactor. Nuclear characteristics are calculated at beginning of cycle of an initial core and at beginning and end of cycle of equilibrium core. According to the calculations, the differences between the two methodologies are smaller than 0.0002 Δk in the multi-plication factor, relatively about 1% in the control rod reactivity, and 1% in the sodium void reactivity.
Energy Technology Data Exchange (ETDEWEB)
Descalle, M; Clouse, C; Pruet, J
2009-07-28
The authors have compared calculations of critical assembly activation ratios using 3 different Monte Carlo codes and one deterministic code. There is excellent agreement. Discrepancies between the different Monte Carlo codes are the 1-2% level. Notably, the deterministic calculations with 87 groups are also in good agreement with the continuous energy Monte Carlo results. The three codes underestimate the {sup 238}U(n,f) reaction, suggesting that there is room for improvement in the evaluation, or in the evaluations of other reactions influencing the spectrum in BigTen. Until statistical uncertainties are implemented in Mercury, they strongly advise long runs to guarantee sufficient convergence of the flux at high energies, and they strongly encourage comparing Mercury results to a well-developed and documented code such as MCNP5 and/or COG. It may be that ENDL2008 will be available for use in COG within a year. Finally, it may be worthwhile to add a 'standard' reaction rate tally similar to those implemented in COG and MCNP5, if the goal is to expand the central fission and activation ratios simulations to include isotopes that are not part of the specifications for the assembly material composition.
Update on the Development and Validation of MERCURY: A Modern, Monte Carlo Particle Transport Code
Energy Technology Data Exchange (ETDEWEB)
Procassini, R J; Taylor, J M; McKinley, M S; Greenman, G M; Cullen, D E; O' Brien, M J; Beck, B R; Hagmann, C A
2005-06-06
An update on the development and validation of the MERCURY Monte Carlo particle transport code is presented. MERCURY is a modern, parallel, general-purpose Monte Carlo code being developed at the Lawrence Livermore National Laboratory. During the past year, several major algorithm enhancements have been completed. These include the addition of particle trackers for 3-D combinatorial geometry (CG), 1-D radial meshes, 2-D quadrilateral unstructured meshes, as well as a feature known as templates for defining recursive, repeated structures in CG. New physics capabilities include an elastic-scattering neutron thermalization model, support for continuous energy cross sections and S ({alpha}, {beta}) molecular bound scattering. Each of these new physics features has been validated through code-to-code comparisons with another Monte Carlo transport code. Several important computer science features have been developed, including an extensible input-parameter parser based upon the XML data description language, and a dynamic load-balance methodology for efficient parallel calculations. This paper discusses the recent work in each of these areas, and describes a plan for future extensions that are required to meet the needs of our ever expanding user base.
MCNP: a general Monte Carlo code for neutron and photon transport
Energy Technology Data Exchange (ETDEWEB)
Forster, R.A.; Godfrey, T.N.K.
1985-01-01
MCNP is a very general Monte Carlo neutron photon transport code system with approximately 250 person years of Group X-6 code development invested. It is extremely portable, user-oriented, and a true production code as it is used about 60 Cray hours per month by about 150 Los Alamos users. It has as its data base the best cross-section evaluations available. MCNP contains state-of-the-art traditional and adaptive Monte Carlo techniques to be applied to the solution of an ever-increasing number of problems. Excellent user-oriented documentation is available for all facets of the MCNP code system. Many useful and important variants of MCNP exist for special applications. The Radiation Shielding Information Center (RSIC) in Oak Ridge, Tennessee is the contact point for worldwide MCNP code and documentation distribution. A much improved MCNP Version 3A will be available in the fall of 1985, along with new and improved documentation. Future directions in MCNP development will change the meaning of MCNP to Monte Carlo N Particle where N particle varieties will be transported.
Applications of FLUKA Monte Carlo code for nuclear and accelerator physics
Battistoni, Giuseppe; Brugger, Markus; Campanella, Mauro; Carboni, Massimo; Empl, Anton; Fasso, Alberto; Gadioli, Ettore; Cerutti, Francesco; Ferrari, Alfredo; Ferrari, Anna; Lantz, Matthias; Mairani, Andrea; Margiotta, M; Morone, Christina; Muraro, Silvia; Parodi, Katerina; Patera, Vincenzo; Pelliccioni, Maurizio; Pinsky, Lawrence; Ranft, Johannes; Roesler, Stefan; Rollet, Sofia; Sala, Paola R; Santana, Mario; Sarchiapone, Lucia; Sioli, Maximiliano; Smirnov, George; Sommerer, Florian; Theis, Christian; Trovati, Stefania; Villari, R; Vincke, Heinz; Vincke, Helmut; Vlachoudis, Vasilis; Vollaire, Joachim; Zapp, Neil
2011-01-01
FLUKA is a general purpose Monte Carlo code capable of handling all radiation components from thermal energies (for neutrons) or 1keV (for all other particles) to cosmic ray energies and can be applied in many different fields. Presently the code is maintained on Linux. The validity of the physical models implemented in FLUKA has been benchmarked against a variety of experimental data over a wide energy range, from accelerator data to cosmic ray showers in the Earth atmosphere. FLUKA is widely used for studies related both to basic research and to applications in particle accelerators, radiation protection and dosimetry, including the specific issue of radiation damage in space missions, radiobiology (including radiotherapy) and cosmic ray calculations. After a short description of the main features that make FLUKA valuable for these topics, the present paper summarizes some of the recent applications of the FLUKA Monte Carlo code in the nuclear as well high energy physics. In particular it addresses such top...
A new Monte Carlo code for absorption simulation of laser-skin tissue interaction
Institute of Scientific and Technical Information of China (English)
Afshan Shirkavand; Saeed Sarkar; Marjaneh Hejazi; Leila Ataie-Fashtami; Mohammad Reza Alinaghizadeh
2007-01-01
In laser clinical applications, the process of photon absorption and thermal energy diffusion in the target tissue and its surrounding tissue during laser irradiation are crucial. Such information allows the selection of proper operating parameters such as laser power, and exposure time for optimal therapeutic. The Monte Carlo method is a useful tool for studying laser-tissue interaction and simulation of energy absorption in tissue during laser irradiation. We use the principles of this technique and write a new code with MATLAB 6.5, and then validate it against Monte Carlo multi layer (MCML) code. The new code is proved to be with good accuracy. It can be used to calculate the total power bsorbed in the region of interest. This can be combined for heat modelling with other computerized programs.
ERSN-OpenMC, a Java-based GUI for OpenMC Monte Carlo code
Directory of Open Access Journals (Sweden)
Jaafar EL Bakkali
2016-07-01
Full Text Available OpenMC is a new Monte Carlo transport particle simulation code focused on solving two types of neutronic problems mainly the k-eigenvalue criticality fission source problems and external fixed fission source problems. OpenMC does not have any Graphical User Interface and the creation of one is provided by our java-based application named ERSN-OpenMC. The main feature of this application is to provide to the users an easy-to-use and flexible graphical interface to build better and faster simulations, with less effort and great reliability. Additionally, this graphical tool was developed with several features, as the ability to automate the building process of OpenMC code and related libraries as well as the users are given the freedom to customize their installation of this Monte Carlo code. A full description of the ERSN-OpenMC application is presented in this paper.
Lida Gholamkar; Mahdi Sadeghi; Ali Asghar Mowlavi; Mitra Athari
2016-01-01
Introduction One of the best methods in the diagnosis and control of breast cancer is mammography. The importance of mammography is directly related to its value in the detection of breast cancer in the early stages, which leads to a more effective treatment. The purpose of this article was to calculate the X-ray spectrum in a mammography system with Monte Carlo codes, including MCNPX and MCNP5. Materials and Methods The device, simulated using the MCNP code, was Planmed Nuance digital mammog...
Monte Carlo simulations of a D-T neutron generator shielding for landmine detection
Energy Technology Data Exchange (ETDEWEB)
Reda, A.M., E-mail: amreda2005@yahoo.com [College of Science, Shaqra University, Al-Dawadme, P.O. Box 1040 (Saudi Arabia)
2011-10-15
Shielding for a D-T sealed neutron generator has been designed using the MCNP5 Monte Carlo radiation transport code. The neutron generator will be used in field for the detection of explosives, landmines, drugs and other 'threat' materials. The optimization of the detection of buried objects was started by studying the signal-to-noise ratio for different geometric conditions. - Highlights: > A landmine detection system based on neutron fast/slow analysis has been designed. > Shielding for a D-T sealed neutron generator tube has been designed using Monte Carlo radiation transport code. > Detection of buried objects was started by studying the signal-to-noise ratio for different geometric conditions. > The signal-to-background ratio optimized at one position for all depths.
PyMercury: Interactive Python for the Mercury Monte Carlo Particle Transport Code
Energy Technology Data Exchange (ETDEWEB)
Iandola, F N; O' Brien, M J; Procassini, R J
2010-11-29
Monte Carlo particle transport applications are often written in low-level languages (C/C++) for optimal performance on clusters and supercomputers. However, this development approach often sacrifices straightforward usability and testing in the interest of fast application performance. To improve usability, some high-performance computing applications employ mixed-language programming with high-level and low-level languages. In this study, we consider the benefits of incorporating an interactive Python interface into a Monte Carlo application. With PyMercury, a new Python extension to the Mercury general-purpose Monte Carlo particle transport code, we improve application usability without diminishing performance. In two case studies, we illustrate how PyMercury improves usability and simplifies testing and validation in a Monte Carlo application. In short, PyMercury demonstrates the value of interactive Python for Monte Carlo particle transport applications. In the future, we expect interactive Python to play an increasingly significant role in Monte Carlo usage and testing.
Development of a space radiation Monte Carlo computer simulation based on the FLUKA and ROOT codes
Pinsky, L; Ferrari, A; Sala, P; Carminati, F; Brun, R
2001-01-01
This NASA funded project is proceeding to develop a Monte Carlo-based computer simulation of the radiation environment in space. With actual funding only initially in place at the end of May 2000, the study is still in the early stage of development. The general tasks have been identified and personnel have been selected. The code to be assembled will be based upon two major existing software packages. The radiation transport simulation will be accomplished by updating the FLUKA Monte Carlo program, and the user interface will employ the ROOT software being developed at CERN. The end-product will be a Monte Carlo-based code which will complement the existing analytic codes such as BRYNTRN/HZETRN presently used by NASA to evaluate the effects of radiation shielding in space. The planned code will possess the ability to evaluate the radiation environment for spacecraft and habitats in Earth orbit, in interplanetary space, on the lunar surface, or on a planetary surface such as Mars. Furthermore, it will be usef...
Srna - Monte Carlo codes for proton transport simulation in combined and voxelized geometries
Directory of Open Access Journals (Sweden)
Ilić Radovan D.
2002-01-01
Full Text Available This paper describes new Monte Carlo codes for proton transport simulations in complex geometrical forms and in materials of different composition. The SRNA codes were developed for three dimensional (3D dose distribution calculation in proton therapy and dosimetry. The model of these codes is based on the theory of proton multiple scattering and a simple model of compound nucleus decay. The developed package consists of two codes: SRNA-2KG and SRNA-VOX. The first code simulates proton transport in combined geometry that can be described by planes and second order surfaces. The second one uses the voxelized geometry of material zones and is specifically adopted for the application of patient computer tomography data. Transition probabilities for both codes are given by the SRNADAT program. In this paper, we will present the models and algorithms of our programs, as well as the results of the numerical experiments we have carried out applying them, along with the results of proton transport simulation obtained through the PETRA and GEANT programs. The simulation of the proton beam characterization by means of the Multi-Layer Faraday Cup and spatial distribution of positron emitters obtained by our program indicate the imminent application of Monte Carlo techniques in clinical practice.
A fast Monte Carlo code for proton transport in radiation therapy based on MCNPX.
Jabbari, Keyvan; Seuntjens, Jan
2014-07-01
An important requirement for proton therapy is a software for dose calculation. Monte Carlo is the most accurate method for dose calculation, but it is very slow. In this work, a method is developed to improve the speed of dose calculation. The method is based on pre-generated tracks for particle transport. The MCNPX code has been used for generation of tracks. A set of data including the track of the particle was produced in each particular material (water, air, lung tissue, bone, and soft tissue). This code can transport protons in wide range of energies (up to 200 MeV for proton). The validity of the fast Monte Carlo (MC) code is evaluated with data MCNPX as a reference code. While analytical pencil beam algorithm transport shows great errors (up to 10%) near small high density heterogeneities, there was less than 2% deviation of MCNPX results in our dose calculation and isodose distribution. In terms of speed, the code runs 200 times faster than MCNPX. In the Fast MC code which is developed in this work, it takes the system less than 2 minutes to calculate dose for 10(6) particles in an Intel Core 2 Duo 2.66 GHZ desktop computer.
A fast Monte Carlo code for proton transport in radiation therapy based on MCNPX
Directory of Open Access Journals (Sweden)
Keyvan Jabbari
2014-01-01
Full Text Available An important requirement for proton therapy is a software for dose calculation. Monte Carlo is the most accurate method for dose calculation, but it is very slow. In this work, a method is developed to improve the speed of dose calculation. The method is based on pre-generated tracks for particle transport. The MCNPX code has been used for generation of tracks. A set of data including the track of the particle was produced in each particular material (water, air, lung tissue, bone, and soft tissue. This code can transport protons in wide range of energies (up to 200 MeV for proton. The validity of the fast Monte Carlo (MC code is evaluated with data MCNPX as a reference code. While analytical pencil beam algorithm transport shows great errors (up to 10% near small high density heterogeneities, there was less than 2% deviation of MCNPX results in our dose calculation and isodose distribution. In terms of speed, the code runs 200 times faster than MCNPX. In the Fast MC code which is developed in this work, it takes the system less than 2 minutes to calculate dose for 10 6 particles in an Intel Core 2 Duo 2.66 GHZ desktop computer.
DgSMC-B code: A robust and autonomous direct simulation Monte Carlo code for arbitrary geometries
Kargaran, H.; Minuchehr, A.; Zolfaghari, A.
2016-07-01
In this paper, we describe the structure of a new Direct Simulation Monte Carlo (DSMC) code that takes advantage of combinatorial geometry (CG) to simulate any rarefied gas flows Medias. The developed code, called DgSMC-B, has been written in FORTRAN90 language with capability of parallel processing using OpenMP framework. The DgSMC-B is capable of handling 3-dimensional (3D) geometries, which is created with first-and second-order surfaces. It performs independent particle tracking for the complex geometry without the intervention of mesh. In addition, it resolves the computational domain boundary and volume computing in border grids using hexahedral mesh. The developed code is robust and self-governing code, which does not use any separate code such as mesh generators. The results of six test cases have been presented to indicate its ability to deal with wide range of benchmark problems with sophisticated geometries such as airfoil NACA 0012. The DgSMC-B code demonstrates its performance and accuracy in a variety of problems. The results are found to be in good agreement with references and experimental data.
Effects of physics change in Monte Carlo code on electron pencil beam dose distributions
Energy Technology Data Exchange (ETDEWEB)
Toutaoui, Abdelkader, E-mail: toutaoui.aek@gmail.com [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Khelassi-Toutaoui, Nadia, E-mail: nadiakhelassi@yahoo.fr [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Brahimi, Zakia, E-mail: zsbrahimi@yahoo.fr [Departement de Physique Medicale, Centre de Recherche Nucleaire d' Alger, 2 Bd Frantz Fanon BP399 Alger RP, Algiers (Algeria); Chami, Ahmed Chafik, E-mail: chafik_chami@yahoo.fr [Laboratoire de Sciences Nucleaires, Faculte de Physique, Universite des Sciences et de la Technologie Houari Boumedienne, BP 32 El Alia, Bab Ezzouar, Algiers (Algeria)
2012-01-15
Pencil beam algorithms used in computerized electron beam dose planning are usually described using the small angle multiple scattering theory. Alternatively, the pencil beams can be generated by Monte Carlo simulation of electron transport. In a previous work, the 4th version of the Electron Gamma Shower (EGS) Monte Carlo code was used to obtain dose distributions from monoenergetic electron pencil beam, with incident energy between 1 MeV and 50 MeV, interacting at the surface of a large cylindrical homogeneous water phantom. In 2000, a new version of this Monte Carlo code has been made available by the National Research Council of Canada (NRC), which includes various improvements in its electron-transport algorithms. In the present work, we were interested to see if the new physics in this version produces pencil beam dose distributions very different from those calculated with oldest one. The purpose of this study is to quantify as well as to understand these differences. We have compared a series of pencil beam dose distributions scored in cylindrical geometry, for electron energies between 1 MeV and 50 MeV calculated with two versions of the Electron Gamma Shower Monte Carlo Code. Data calculated and compared include isodose distributions, radial dose distributions and fractions of energy deposition. Our results for radial dose distributions show agreement within 10% between doses calculated by the two codes for voxels closer to the pencil beam central axis, while the differences are up to 30% for longer distances. For fractions of energy deposition, the results of the EGS4 are in good agreement (within 2%) with those calculated by EGSnrc at shallow depths for all energies, whereas a slightly worse agreement (15%) is observed at deeper distances. These differences may be mainly attributed to the different multiple scattering for electron transport adopted in these two codes and the inclusion of spin effect, which produces an increase of the effective range of
The Physical Models and Statistical Procedures Used in the RACER Monte Carlo Code
Energy Technology Data Exchange (ETDEWEB)
Sutton, T.M.; Brown, F.B.; Bischoff, F.G.; MacMillan, D.B.; Ellis, C.L.; Ward, J.T.; Ballinger, C.T.; Kelly, D.J.; Schindler, L.
1999-07-01
This report describes the MCV (Monte Carlo - Vectorized)Monte Carlo neutron transport code [Brown, 1982, 1983; Brown and Mendelson, 1984a]. MCV is a module in the RACER system of codes that is used for Monte Carlo reactor physics analysis. The MCV module contains all of the neutron transport and statistical analysis functions of the system, while other modules perform various input-related functions such as geometry description, material assignment, output edit specification, etc. MCV is very closely related to the 05R neutron Monte Carlo code [Irving et al., 1965] developed at Oak Ridge National Laboratory. 05R evolved into the 05RR module of the STEMB system, which was the forerunner of the RACER system. Much of the overall logic and physics treatment of 05RR has been retained and, indeed, the original verification of MCV was achieved through comparison with STEMB results. MCV has been designed to be very computationally efficient [Brown, 1981, Brown and Martin, 1984b; Brown, 1986]. It was originally programmed to make use of vector-computing architectures such as those of the CDC Cyber- 205 and Cray X-MP. MCV was the first full-scale production Monte Carlo code to effectively utilize vector-processing capabilities. Subsequently, MCV was modified to utilize both distributed-memory [Sutton and Brown, 1994] and shared memory parallelism. The code has been compiled and run on platforms ranging from 32-bit UNIX workstations to clusters of 64-bit vector-parallel supercomputers. The computational efficiency of the code allows the analyst to perform calculations using many more neutron histories than is practical with most other Monte Carlo codes, thereby yielding results with smaller statistical uncertainties. MCV also utilizes variance reduction techniques such as survival biasing, splitting, and rouletting to permit additional reduction in uncertainties. While a general-purpose neutron Monte Carlo code, MCV is optimized for reactor physics calculations. It has the
On the use of SERPENT Monte Carlo code to generate few group diffusion constants
Energy Technology Data Exchange (ETDEWEB)
Piovezan, Pamela, E-mail: pamela.piovezan@ctmsp.mar.mil.b [Centro Tecnologico da Marinha em Sao Paulo (CTMSP), Sao Paulo, SP (Brazil); Carluccio, Thiago; Domingos, Douglas Borges; Rossi, Pedro Russo; Mura, Luiz Felipe, E-mail: fermium@cietec.org.b, E-mail: thiagoc@ipen.b [Fermium Tecnologia Nuclear, Sao Paulo, SP (Brazil); Instituto de Pesquisas Energeticas e Nucleares (IPEN/CNEN-SP), Sao Paulo, SP (Brazil)
2011-07-01
The accuracy of diffusion reactor codes strongly depends on the quality of the groups constants processing. For many years, the generation of such constants was based on 1-D infinity cell transport calculations. Some developments using collision probability or the method of characteristics allow, nowadays, 2-D assembly group constants calculations. However, these 1-D and 2-D codes how some limitations as , for example, on complex geometries and in the neighborhood of heavy absorbers. On the other hand, since Monte Carlos (MC) codes provide accurate neutro flux distributions, the possibility of using these solutions to provide group constants to full-core reactor diffusion simulators has been recently investigated, especially for the cases in which the geometry and reactor types are beyond the capability of the conventional deterministic lattice codes. The two greatest difficulties on the use of MC codes to group constant generation are the computational costs and the methodological incompatibility between analog MC particle transport simulation and deterministic transport methods based in several approximations. The SERPENT code is a 3-D continuous energy MC transport code with built-in burnup capability that was specially optimized to generate these group constants. In this work, we present the preliminary results of using the SERPENT MC code to generate 3-D two-group diffusion constants for a PWR like assembly. These constants were used in the CITATION diffusion code to investigate the effects of the MC group constants determination on the neutron multiplication factor diffusion estimate. (author)
Françoise Benz
2006-01-01
2005-2006 ACADEMIC TRAINING PROGRAMME LECTURE SERIES 27, 28, 29 June 11:00-12:00 - TH Conference Room, bldg. 4 The use of Monte Carlo radiation transport codes in radiation physics and dosimetry F. Salvat Gavalda,Univ. de Barcelona, A. FERRARI, CERN-AB, M. SILARI, CERN-SC Lecture 1. Transport and interaction of electromagnetic radiation F. Salvat Gavalda,Univ. de Barcelona Interaction models and simulation schemes implemented in modern Monte Carlo codes for the simulation of coupled electron-photon transport will be briefly reviewed. Different schemes for simulating electron transport will be discussed. Condensed algorithms, which rely on multiple-scattering theories, are comparatively fast, but less accurate than mixed algorithms, in which hard interactions (with energy loss or angular deflection larger than certain cut-off values) are simulated individually. The reliability, and limitations, of electron-interaction models and multiple-scattering theories will be analyzed. Benchmark comparisons of simu...
Energy Technology Data Exchange (ETDEWEB)
T.J. Urbatsch; T.M. Evans
2006-02-15
We have released Version 2 of Milagro, an object-oriented, C++ code that performs radiative transfer using Fleck and Cummings' Implicit Monte Carlo method. Milagro, a part of the Jayenne program, is a stand-alone driver code used as a methods research vehicle and to verify its underlying classes. These underlying classes are used to construct Implicit Monte Carlo packages for external customers. Milagro-2 represents a design overhaul that allows better parallelism and extensibility. New features in Milagro-2 include verified momentum deposition, restart capability, graphics capability, exact energy conservation, and improved load balancing and parallel efficiency. A users' guide also describes how to configure, make, and run Milagro2.
Evaluation of CASMO-3 and HELIOS for Fuel Assembly Analysis from Monte Carlo Code
Energy Technology Data Exchange (ETDEWEB)
Shim, Hyung Jin; Song, Jae Seung; Lee, Chung Chan
2007-05-15
This report presents a study comparing deterministic lattice physics calculations with Monte Carlo calculations for LWR fuel pin and assembly problems. The study has focused on comparing results from the lattice physics code CASMO-3 and HELIOS against those from the continuous-energy Monte Carlo code McCARD. The comparisons include k{sub inf}, isotopic number densities, and pin power distributions. The CASMO-3 and HELIOS calculations for the k{sub inf}'s of the LWR fuel pin problems show good agreement with McCARD within 956pcm and 658pcm, respectively. For the assembly problems with Gadolinia burnable poison rods, the largest difference between the k{sub inf}'s is 1463pcm with CASMO-3 and 1141pcm with HELIOS. RMS errors for the pin power distributions of CASMO-3 and HELIOS are within 1.3% and 1.5%, respectively.
The Serpent Monte Carlo Code: Status, Development and Applications in 2013
Leppänen, Jaakko; Pusa, Maria; Viitanen, Tuomas; Valtavirta, Ville; Kaltiaisenaho, Toni
2014-06-01
The Serpent Monte Carlo reactor physics burnup calculation code has been developed at VTT Technical Research Centre of Finland since 2004, and is currently used in 100 universities and research organizations around the world. This paper presents the brief history of the project, together with the currently available methods and capabilities and plans for future work. Typical user applications are introduced in the form of a summary review on Serpent-related publications over the past few years.
Energy Technology Data Exchange (ETDEWEB)
Perfetti, C.; Martin, W. [Univ. of Michigan, Dept. of Nuclear Engineering and Radiological Sciences, 2355 Bonisteel Boulevard, Ann Arbor, MI 48109-2104 (United States); Rearden, B.; Williams, M. [Oak Ridge National Laboratory, Reactor and Nuclear Systems Div., Bldg. 5700, P.O. Box 2008, Oak Ridge, TN 37831-6170 (United States)
2012-07-01
Three methods for calculating continuous-energy eigenvalue sensitivity coefficients were developed and implemented into the Shift Monte Carlo code within the SCALE code package. The methods were used for two small-scale test problems and were evaluated in terms of speed, accuracy, efficiency, and memory requirements. A promising new method for calculating eigenvalue sensitivity coefficients, known as the CLUTCH method, was developed and produced accurate sensitivity coefficients with figures of merit that were several orders of magnitude larger than those from existing methods. (authors)
ASCOT: redesigned Monte Carlo code for simulations of minority species in tokamak plasmas
Hirvijoki, Eero; Koskela, Tuomas; Kurki-Suonio, Taina; Miettunen, Juho; Sipilä, Seppo; Snicker, Antti; Äkäslompolo, Simppa
2013-01-01
A comprehensive description of methods for Monte Carlo studies of fast ions and impurity species in tokamak plasmas is presented. The described methods include Hamiltonian orbit-following in particle and guiding center phase space, test particle or guiding center solution of the kinetic equation applying stochastic differential equations in the presence of Coulomb collisions, Neoclassical tearing modes and Alfv\\'en eigenmodes as electromagnetic perturbations relevant for fast ions, together with plasma flow and atomic reactions relevant for impurity studies. Applying the methods, a complete reimplementation of a well-established minority species code is carried out as a response both to the increase in computing power during the last twenty years and to the weakly structured growth of the previous code which has made implementation of additional models impractical. Also, a thorough benchmark between the previous code and the reimplementation is accomplished, showing good agreement between the codes.
Shchurovskaya, M. V.; Alferov, V. P.; Geraskin, N. I.; Radaev, A. I.
2017-01-01
The results of the validation of a research reactor calculation using Monte Carlo and deterministic codes against experimental data and based on code-to-code comparison are presented. The continuous energy Monte Carlo code MCU-PTR and the nodal diffusion-based deterministic code TIGRIS were used for full 3-D calculation of the IRT MEPhI research reactor. The validation included the investigations for the reactor with existing high enriched uranium (HEU, 90 w/o) fuel and low enriched uranium (LEU, 19.7 w/o, U-9%Mo) fuel.
Chetty, Indrin J.; Moran, Jean M.; Nurushev, Teamor S.; McShan, Daniel L.; Fraass, Benedick A.; Wilderman, Scott J.; Bielajew, Alex F.
2002-06-01
A comprehensive set of measurements and calculations has been conducted to investigate the accuracy of the Dose Planning Method (DPM) Monte Carlo code for electron beam dose calculations in heterogeneous media. Measurements were made using 10 MeV and 50 MeV minimally scattered, uncollimated electron beams from a racetrack microtron. Source distributions for the Monte Carlo calculations were reconstructed from in-air ion chamber scans and then benchmarked against measurements in a homogeneous water phantom. The in-air spatial distributions were found to have FWHM of 4.7 cm and 1.3 cm, at 100 cm from the source, for the 10 MeV and 50 MeV beams respectively. Energy spectra for the electron beams were determined by simulating the components of the microtron treatment head using the code MCNP4B. Profile measurements were made using an ion chamber in a water phantom with slabs of lung or bone-equivalent materials submerged at various depths. DPM calculations are, on average, within 2% agreement with measurement for all geometries except for the 50 MeV incident on a 6 cm lung-equivalent slab. Measurements using approximately monoenergetic, 50 MeV, 'pencil-beam'-type electrons in heterogeneous media provide conditions for maximum electronic disequilibrium and hence present a stringent test of the code's electron transport physics; the agreement noted between calculation and measurement illustrates that the DPM code is capable of accurate dose calculation even under such conditions.
Energy Technology Data Exchange (ETDEWEB)
Aleshin, Sergey S.; Gorodkov, Sergey S.; Shcherenko, Anna I. [National Research Centre ' Kurchatov Institute' , Moscow (Russian Federation)
2016-09-15
A burn-up calculation of large systems by Monte-Carlo code (MCU) is complex process and it requires large computational costs. Previously prepared isotopic compositions are proposed to be used for the Monte-Carlo code calculations of different system states with burnt fuel. Isotopic compositions are calculated by an approximation method. The approximation method is based on usage of a spectral functionality and reference isotopic compositions, that are calculated by the engineering codes (TVS-M, BIPR-7A and PERMAK-A). The multiplication factors and power distributions of FAs from a 3-D reactor core are calculated in this work by the Monte-Carlo code MCU using earlier prepared isotopic compositions. The separate conditions of the burnt core are observed. The results of MCU calculations were compared with those that were obtained by engineering codes.
Rabie, M.; Franck, C. M.
2016-06-01
We present a freely available MATLAB code for the simulation of electron transport in arbitrary gas mixtures in the presence of uniform electric fields. For steady-state electron transport, the program provides the transport coefficients, reaction rates and the electron energy distribution function. The program uses established Monte Carlo techniques and is compatible with the electron scattering cross section files from the open-access Plasma Data Exchange Project LXCat. The code is written in object-oriented design, allowing the tracing and visualization of the spatiotemporal evolution of electron swarms and the temporal development of the mean energy and the electron number due to attachment and/or ionization processes. We benchmark our code with well-known model gases as well as the real gases argon, N2, O2, CF4, SF6 and mixtures of N2 and O2.
A Parallel Monte Carlo Code for Simulating Collisional N-body Systems
Pattabiraman, Bharath; Liao, Wei-Keng; Choudhary, Alok; Kalogera, Vassiliki; Memik, Gokhan; Rasio, Frederic A
2012-01-01
We present a new parallel code for computing the dynamical evolution of collisional N-body systems with up to N~10^7 particles. Our code is based on the the H\\'enon Monte Carlo method for solving the Fokker-Planck equation, and makes assumptions of spherical symmetry and dynamical equilibrium. The principal algorithmic developments involve optimizing data structures, and the introduction of a parallel random number generation scheme, as well as a parallel sorting algorithm, required to find nearest neighbors for interactions and to compute the gravitational potential. The new algorithms we introduce along with our choice of decomposition scheme minimize communication costs and ensure optimal distribution of data and workload among the processing units. The implementation uses the Message Passing Interface (MPI) library for communication, which makes it portable to many different supercomputing architectures. We validate the code by calculating the evolution of clusters with initial Plummer distribution functi...
Comparison of Geant4-DNA simulation of S-values with other Monte Carlo codes
Energy Technology Data Exchange (ETDEWEB)
André, T. [Université Bordeaux 1, CENBG, UMR 5797, F-33170 Gradignan (France); CNRS, IN2P3, CENBG, UMR 5797, F-33170 Gradignan (France); Morini, F. [Research Group of Theoretical Chemistry and Molecular Modelling, Hasselt University, Agoralaan Gebouw D, B-3590 Diepenbeek (Belgium); Karamitros, M. [Université Bordeaux 1, CENBG, UMR 5797, F-33170 Gradignan (France); CNRS, IN2P3, CENBG, UMR 5797, F-33170 Gradignan (France); CNRS, INCIA, UMR 5287, F-33400 Talence (France); Delorme, R. [LPSC, Université Joseph Fourier Grenoble 1, CNRS/IN2P3, Grenoble INP, 38026 Grenoble (France); CEA, LIST, F-91191 Gif-sur-Yvette (France); Le Loirec, C. [CEA, LIST, F-91191 Gif-sur-Yvette (France); Campos, L. [Departamento de Física, Universidade Federal de Sergipe, São Cristóvão (Brazil); Champion, C. [Université Bordeaux 1, CENBG, UMR 5797, F-33170 Gradignan (France); CNRS, IN2P3, CENBG, UMR 5797, F-33170 Gradignan (France); Groetz, J.-E.; Fromm, M. [Université de Franche-Comté, Laboratoire Chrono-Environnement, UMR CNRS 6249, Besançon (France); Bordage, M.-C. [Laboratoire Plasmas et Conversion d’Énergie, UMR 5213 CNRS-INPT-UPS, Université Paul Sabatier, Toulouse (France); Perrot, Y. [Laboratoire de Physique Corpusculaire, UMR 6533, Aubière (France); Barberet, Ph. [Université Bordeaux 1, CENBG, UMR 5797, F-33170 Gradignan (France); CNRS, IN2P3, CENBG, UMR 5797, F-33170 Gradignan (France); and others
2014-01-15
Monte Carlo simulations of S-values have been carried out with the Geant4-DNA extension of the Geant4 toolkit. The S-values have been simulated for monoenergetic electrons with energies ranging from 0.1 keV up to 20 keV, in liquid water spheres (for four radii, chosen between 10 nm and 1 μm), and for electrons emitted by five isotopes of iodine (131, 132, 133, 134 and 135), in liquid water spheres of varying radius (from 15 μm up to 250 μm). The results have been compared to those obtained from other Monte Carlo codes and from other published data. The use of the Kolmogorov–Smirnov test has allowed confirming the statistical compatibility of all simulation results.
Neutron cross-section probability tables in TRIPOLI-3 Monte Carlo transport code
Energy Technology Data Exchange (ETDEWEB)
Zheng, S.H.; Vergnaud, T.; Nimal, J.C. [Commissariat a l`Energie Atomique, Gif-sur-Yvette (France). Lab. d`Etudes de Protection et de Probabilite
1998-03-01
Neutron transport calculations need an accurate treatment of cross sections. Two methods (multi-group and pointwise) are usually used. A third one, the probability table (PT) method, has been developed to produce a set of cross-section libraries, well adapted to describe the neutron interaction in the unresolved resonance energy range. Its advantage is to present properly the neutron cross-section fluctuation within a given energy group, allowing correct calculation of the self-shielding effect. Also, this PT cross-section representation is suitable for simulation of neutron propagation by the Monte Carlo method. The implementation of PTs in the TRIPOLI-3 three-dimensional general Monte Carlo transport code, developed at Commissariat a l`Energie Atomique, and several validation calculations are presented. The PT method is proved to be valid not only in the unresolved resonance range but also in all the other energy ranges.
ITS Version 6 : the integrated TIGER series of coupled electron/photon Monte Carlo transport codes.
Energy Technology Data Exchange (ETDEWEB)
Franke, Brian Claude; Kensek, Ronald Patrick; Laub, Thomas William
2008-04-01
ITS is a powerful and user-friendly software package permitting state-of-the-art Monte Carlo solution of lineartime-independent coupled electron/photon radiation transport problems, with or without the presence of macroscopic electric and magnetic fields of arbitrary spatial dependence. Our goal has been to simultaneously maximize operational simplicity and physical accuracy. Through a set of preprocessor directives, the user selects one of the many ITS codes. The ease with which the makefile system is applied combines with an input scheme based on order-independent descriptive keywords that makes maximum use of defaults and internal error checking to provide experimentalists and theorists alike with a method for the routine but rigorous solution of sophisticated radiation transport problems. Physical rigor is provided by employing accurate cross sections, sampling distributions, and physical models for describing the production and transport of the electron/photon cascade from 1.0 GeV down to 1.0 keV. The availability of source code permits the more sophisticated user to tailor the codes to specific applications and to extend the capabilities of the codes to more complex applications. Version 6, the latest version of ITS, contains (1) improvements to the ITS 5.0 codes, and (2) conversion to Fortran 90. The general user friendliness of the software has been enhanced through memory allocation to reduce the need for users to modify and recompile the code.
Srna-Monte Carlo codes for proton transport simulation in combined and voxelized geometries
Ilic, R D; Stankovic, S J
2002-01-01
This paper describes new Monte Carlo codes for proton transport simulations in complex geometrical forms and in materials of different composition. The SRNA codes were developed for three dimensional (3D) dose distribution calculation in proton therapy and dosimetry. The model of these codes is based on the theory of proton multiple scattering and a simple model of compound nucleus decay. The developed package consists of two codes: SRNA-2KG and SRNA-VOX. The first code simulates proton transport in combined geometry that can be described by planes and second order surfaces. The second one uses the voxelized geometry of material zones and is specifically adopted for the application of patient computer tomography data. Transition probabilities for both codes are given by the SRNADAT program. In this paper, we will present the models and algorithms of our programs, as well as the results of the numerical experiments we have carried out applying them, along with the results of proton transport simulation obtaine...
A Benchmarking Study of High Energy Carbon Ion Induced Neutron Using Several Monte Carlo Codes
Energy Technology Data Exchange (ETDEWEB)
Kim, D. H.; Oh, J. H.; Jung, N. S.; Lee, H. S. [Pohang Accelerator Laboratory, Pohang (Korea, Republic of); Shin, Y. S.; Kwon, D. Y.; Kim, Y. M. [Catholic Univ., Gyeongsan (Korea, Republic of); Oranj, L. Mokhtari [POSTECH, Pohang (Korea, Republic of)
2014-10-15
In this study, the benchmarking study was done for the representative particle interaction of the heavy ion accelerator, especially carbon-induced reaction. The secondary neutron is an important particle in the shielding analysis to define the source term and penetration ability of radiation fields. The performance of each Monte Carlo codes were verified for selected codes: MCNPX 2.7, PHITS 2.64 and FLUKA 2011.2b.6. For this benchmarking study, the experimental data of Kurosawa et al. in the SINBAD database of NEA was applied. The calculated results of the differential neutron yield produced from several materials irradiated by high energy carbon beam reproduced the experimental data well in small uncertainty. But the MCNPX results showed large discrepancy with experimental data, especially at the forward angle. The calculated results were lower a little than the experimental and it was clear in the cases of lower incident carbon energy, thinner target and forward angle. As expected, the influence of different model was found clearly at forward direction. In the shielding analysis, these characteristics of each Monte Carlo codes should be considered and utilized to determine the safety margin of a shield thickness.
OpenMC: A State-of-the-Art Monte Carlo Code for Research and Development
Romano, Paul K.; Horelik, Nicholas E.; Herman, Bryan R.; Nelson, Adam G.; Forget, Benoit; Smith, Kord
2014-06-01
This paper gives an overview of OpenMC, an open source Monte Carlo particle transport code recently developed at the Massachusetts Institute of Technology. OpenMC uses continuous-energy cross sections and a constructive solid geometry representation, enabling high-fidelity modeling of nuclear reactors and other systems. Modern, portable input/output file formats are used in OpenMC: XML for input, and HDF5 for output. High performance parallel algorithms in OpenMC have demonstrated near-linear scaling to over 100,000 processors on modern supercomputers. Other topics discussed in this paper include plotting, CMFD acceleration, variance reduction, eigenvalue calculations, and software development processes.
Energy Technology Data Exchange (ETDEWEB)
Both, J.P.; Nimal, J.C.; Vergnaud, T. (CEA Centre d' Etudes Nucleaires de Saclay, 91 - Gif-sur-Yvette (France). Service d' Etudes des Reacteurs et de Mathematiques Appliquees)
1990-01-01
We discuss an automated biasing procedure for generating the parameters necessary to achieve efficient Monte Carlo biasing shielding calculations. The biasing techniques considered here are exponential transform and collision biasing deriving from the concept of the biased game based on the importance function. We use a simple model of the importance function with exponential attenuation as the distance to the detector increases. This importance function is generated on a three-dimensional mesh including geometry and with graph theory algorithms. This scheme is currently being implemented in the third version of the neutron and gamma ray transport code TRIPOLI-3. (author).
A fast Monte Carlo code for proton transport in radiation therapy based on MCNPX
Keyvan Jabbari; Jan Seuntjens
2014-01-01
An important requirement for proton therapy is a software for dose calculation. Monte Carlo is the most accurate method for dose calculation, but it is very slow. In this work, a method is developed to improve the speed of dose calculation. The method is based on pre-generated tracks for particle transport. The MCNPX code has been used for generation of tracks. A set of data including the track of the particle was produced in each particular material (water, air, lung tissue, bone, and soft t...
Energy Technology Data Exchange (ETDEWEB)
Cullen, D E
1998-11-22
TART98 is a coupled neutron-photon, 3 Dimensional, combinatorial geometry, time dependent Monte Carlo radiation transport code. This code can run on any modern computer. It is a complete system to assist you with input preparation, running Monte Carlo calculations, and analysis of output results. TART98 is also incredibly FAST; if you have used similar codes, you will be amazed at how fast this code is compared to other similar codes. Use of the entire system can save you a great deal of time and energy. TART98 is distributed on CD. This CD contains on-line documentation for all codes included in the system, the codes configured to run on a variety of computers, and many example problems that you can use to familiarize yourself with the system. TART98 completely supersedes all older versions of TART, and it is strongly recommended that users only use the most recent version of TART98 and its data files.
Energy Technology Data Exchange (ETDEWEB)
Fayos Ferrer, F.; Antolin Sanmartin, E.; Simon de Blas, R.; Palazon Cano, I.; Bertomeu Padin, T.; Gutierrez Sarraga, J.; Rey Portoles, G.
2011-07-01
This paper is subjected to various tests including Monte Carlo dosimetric the code in the latest versions of Multi plan Accuracy planner. They compare their results and Ray-Tracing Algorithm (RT), present from the earliest versions, with the experimental results obtained by photographic dosimetry and ionization chamber measurements.
Lin, Yi-Chun; Huang, Tseng-Te; Liu, Yuan-Hao; Chen, Wei-Lin; Chen, Yen-Fu; Wu, Shu-Wei; Nievaart, Sander; Jiang, Shiang-Huei
2015-06-01
The paired ionization chambers (ICs) technique is commonly employed to determine neutron and photon doses in radiology or radiotherapy neutron beams, where neutron dose shows very strong dependence on the accuracy of accompanying high energy photon dose. During the dose derivation, it is an important issue to evaluate the photon and electron response functions of two commercially available ionization chambers, denoted as TE(TE) and Mg(Ar), used in our reactor based epithermal neutron beam. Nowadays, most perturbation corrections for accurate dose determination and many treatment planning systems are based on the Monte Carlo technique. We used general purposed Monte Carlo codes, MCNP5, EGSnrc, FLUKA or GEANT4 for benchmark verifications among them and carefully measured values for a precise estimation of chamber current from absorbed dose rate of cavity gas. Also, energy dependent response functions of two chambers were calculated in a parallel beam with mono-energies from 20 keV to 20 MeV photons and electrons by using the optimal simple spherical and detailed IC models. The measurements were performed in the well-defined (a) four primary M-80, M-100, M120 and M150 X-ray calibration fields, (b) primary 60Co calibration beam, (c) 6 MV and 10 MV photon, (d) 6 MeV and 18 MeV electron LINACs in hospital and (e) BNCT clinical trials neutron beam. For the TE(TE) chamber, all codes were almost identical over the whole photon energy range. In the Mg(Ar) chamber, MCNP5 showed lower response than other codes for photon energy region below 0.1 MeV and presented similar response above 0.2 MeV (agreed within 5% in the simple spherical model). With the increase of electron energy, the response difference between MCNP5 and other codes became larger in both chambers. Compared with the measured currents, MCNP5 had the difference from the measurement data within 5% for the 60Co, 6 MV, 10 MV, 6 MeV and 18 MeV LINACs beams. But for the Mg(Ar) chamber, the derivations reached 7
The use of Monte Carlo radiation transport codes in radiation physics and dosimetry
CERN. Geneva; Ferrari, Alfredo; Silari, Marco
2006-01-01
Transport and interaction of electromagnetic radiation Interaction models and simulation schemes implemented in modern Monte Carlo codes for the simulation of coupled electron-photon transport will be briefly reviewed. In these codes, photon transport is simulated by using the detailed scheme, i.e., interaction by interaction. Detailed simulation is easy to implement, and the reliability of the results is only limited by the accuracy of the adopted cross sections. Simulations of electron and positron transport are more difficult, because these particles undergo a large number of interactions in the course of their slowing down. Different schemes for simulating electron transport will be discussed. Condensed algorithms, which rely on multiple-scattering theories, are comparatively fast, but less accurate than mixed algorithms, in which hard interactions (with energy loss or angular deflection larger than certain cut-off values) are simulated individually. The reliability, and limitations, of electron-interacti...
Domain Decomposition of a Constructive Solid Geometry Monte Carlo Transport Code
Energy Technology Data Exchange (ETDEWEB)
O' Brien, M J; Joy, K I; Procassini, R J; Greenman, G M
2008-12-07
Domain decomposition has been implemented in a Constructive Solid Geometry (CSG) Monte Carlo neutron transport code. Previous methods to parallelize a CSG code relied entirely on particle parallelism; but in our approach we distribute the geometry as well as the particles across processors. This enables calculations whose geometric description is larger than what could fit in memory of a single processor, thus it must be distributed across processors. In addition to enabling very large calculations, we show that domain decomposition can speed up calculations compared to particle parallelism alone. We also show results of a calculation of the proposed Laser Inertial-Confinement Fusion-Fission Energy (LIFE) facility, which has 5.6 million CSG parts.
Generation of XS library for the reflector of VVER reactor core using Monte Carlo code Serpent
Usheva, K. I.; Kuten, S. A.; Khruschinsky, A. A.; Babichev, L. F.
2017-01-01
A physical model of the radial and axial reflector of VVER-1200-like reactor core has been developed. Five types of radial reflector with different material composition exist for the VVER reactor core and 1D and 2D models were developed for all of them. Axial top and bottom reflectors are described by the 1D model. A two-group XS library for diffusion code DYN3D has been generated for all types of reflectors by using Serpent 2 Monte Carlo code. Power distribution in the reactor core calculated in DYN3D is flattened in the core central region to more extent in the 2D model of the radial reflector than in its 1D model.
A portable, parallel, object-oriented Monte Carlo neutron transport code in C++
Energy Technology Data Exchange (ETDEWEB)
Lee, S.R.; Cummings, J.C. [Los Alamos National Lab., NM (United States); Nolen, S.D. [Texas A and M Univ., College Station, TX (United States)]|[Los Alamos National Lab., NM (United States)
1997-05-01
We have developed a multi-group Monte Carlo neutron transport code using C++ and the Parallel Object-Oriented Methods and Applications (POOMA) class library. This transport code, called MC++, currently computes k and {alpha}-eigenvalues and is portable to and runs parallel on a wide variety of platforms, including MPPs, clustered SMPs, and individual workstations. It contains appropriate classes and abstractions for particle transport and, through the use of POOMA, for portable parallelism. Current capabilities of MC++ are discussed, along with physics and performance results on a variety of hardware, including all Accelerated Strategic Computing Initiative (ASCI) hardware. Current parallel performance indicates the ability to compute {alpha}-eigenvalues in seconds to minutes rather than hours to days. Future plans and the implementation of a general transport physics framework are also discussed.
A 3DHZETRN Code in a Spherical Uniform Sphere with Monte Carlo Verification
Wilson, John W.; Slaba, Tony C.; Badavi, Francis F.; Reddell, Brandon D.; Bahadori, Amir A.
2014-01-01
The computationally efficient HZETRN code has been used in recent trade studies for lunar and Martian exploration and is currently being used in the engineering development of the next generation of space vehicles, habitats, and extra vehicular activity equipment. A new version (3DHZETRN) capable of transporting High charge (Z) and Energy (HZE) and light ions (including neutrons) under space-like boundary conditions with enhanced neutron and light ion propagation is under development. In the present report, new algorithms for light ion and neutron propagation with well-defined convergence criteria in 3D objects is developed and tested against Monte Carlo simulations to verify the solution methodology. The code will be available through the software system, OLTARIS, for shield design and validation and provides a basis for personal computer software capable of space shield analysis and optimization.
Coded aperture coherent scatter imaging for breast cancer detection: a Monte Carlo evaluation
Lakshmanan, Manu N.; Morris, Robert E.; Greenberg, Joel A.; Samei, Ehsan; Kapadia, Anuj J.
2016-03-01
It is known that conventional x-ray imaging provides a maximum contrast between cancerous and healthy fibroglandular breast tissues of 3% based on their linear x-ray attenuation coefficients at 17.5 keV, whereas coherent scatter signal provides a maximum contrast of 19% based on their differential coherent scatter cross sections. Therefore in order to exploit this potential contrast, we seek to evaluate the performance of a coded- aperture coherent scatter imaging system for breast cancer detection and investigate its accuracy using Monte Carlo simulations. In the simulations we modeled our experimental system, which consists of a raster-scanned pencil beam of x-rays, a bismuth-tin coded aperture mask comprised of a repeating slit pattern with 2-mm periodicity, and a linear-array of 128 detector pixels with 6.5-keV energy resolution. The breast tissue that was scanned comprised a 3-cm sample taken from a patient-based XCAT breast phantom containing a tomosynthesis- based realistic simulated lesion. The differential coherent scatter cross section was reconstructed at each pixel in the image using an iterative reconstruction algorithm. Each pixel in the reconstructed image was then classified as being either air or the type of breast tissue with which its normalized reconstructed differential coherent scatter cross section had the highest correlation coefficient. Comparison of the final tissue classification results with the ground truth image showed that the coded aperture imaging technique has a cancerous pixel detection sensitivity (correct identification of cancerous pixels), specificity (correctly ruling out healthy pixels as not being cancer) and accuracy of 92.4%, 91.9% and 92.0%, respectively. Our Monte Carlo evaluation of our experimental coded aperture coherent scatter imaging system shows that it is able to exploit the greater contrast available from coherently scattered x-rays to increase the accuracy of detecting cancerous regions within the breast.
Energy Technology Data Exchange (ETDEWEB)
Chetty, Indrin J. [Department of Radiation Oncology, University of Michigan, Ann Arbor, MI (United States)]. E-mail: indrin@med.umich.edu; Moran, Jean M.; Nurushev, Teamor S.; McShan, Daniel L.; Fraass, Benedick A. [Department of Radiation Oncology, University of Michigan, Ann Arbor, MI (United States); Wilderman, Scott J.; Bielajew, Alex F. [Department of Nuclear Engineering, University of Michigan, Ann Arbor, MI (United States)
2002-06-07
A comprehensive set of measurements and calculations has been conducted to investigate the accuracy of the Dose Planning Method (DPM) Monte Carlo code for electron beam dose calculations in heterogeneous media. Measurements were made using 10 MeV and 50 MeV minimally scattered, uncollimated electron beams from a racetrack microtron. Source distributions for the Monte Carlo calculations were reconstructed from in-air ion chamber scans and then benchmarked against measurements in a homogeneous water phantom. The in-air spatial distributions were found to have FWHM of 4.7 cm and 1.3 cm, at 100 cm from the source, for the 10 MeV and 50 MeV beams respectively. Energy spectra for the electron beams were determined by simulating the components of the microtron treatment head using the code MCNP4B. Profile measurements were made using an ion chamber in a water phantom with slabs of lung or bone-equivalent materials submerged at various depths. DPM calculations are, on average, within 2% agreement with measurement for all geometries except for the 50 MeV incident on a 6 cm lung-equivalent slab. Measurements using approximately monoenergetic, 50 MeV, 'pencil-beam'-type electrons in heterogeneous media provide conditions for maximum electronic disequilibrium and hence present a stringent test of the code's electron transport physics; the agreement noted between calculation and measurement illustrates that the DPM code is capable of accurate dose calculation even under such conditions. (author)
Implementation of the probability table method in a continuous-energy Monte Carlo code system
Energy Technology Data Exchange (ETDEWEB)
Sutton, T.M.; Brown, F.B. [Lockheed Martin Corp., Schenectady, NY (United States)
1998-10-01
RACER is a particle-transport Monte Carlo code that utilizes a continuous-energy treatment for neutrons and neutron cross section data. Until recently, neutron cross sections in the unresolved resonance range (URR) have been treated in RACER using smooth, dilute-average representations. This paper describes how RACER has been modified to use probability tables to treat cross sections in the URR, and the computer codes that have been developed to compute the tables from the unresolved resonance parameters contained in ENDF/B data files. A companion paper presents results of Monte Carlo calculations that demonstrate the effect of the use of probability tables versus the use of dilute-average cross sections for the URR. The next section provides a brief review of the probability table method as implemented in the RACER system. The production of the probability tables for use by RACER takes place in two steps. The first step is the generation of probability tables from the nuclear parameters contained in the ENDF/B data files. This step, and the code written to perform it, are described in Section 3. The tables produced are at energy points determined by the ENDF/B parameters and/or accuracy considerations. The tables actually used in the RACER calculations are obtained in the second step from those produced in the first. These tables are generated at energy points specific to the RACER calculation. Section 4 describes this step and the code written to implement it, as well as modifications made to RACER to enable it to use the tables. Finally, some results and conclusions are presented in Section 5.
Spread-out Bragg peak and monitor units calculation with the Monte Carlo code MCNPX.
Hérault, J; Iborra, N; Serrano, B; Chauvel, P
2007-02-01
The aim of this work was to study the dosimetric potential of the Monte Carlo code MCNPX applied to the protontherapy field. For series of clinical configurations a comparison between simulated and experimental data was carried out, using the proton beam line of the MEDICYC isochronous cyclotron installed in the Centre Antoine Lacassagne in Nice. The dosimetric quantities tested were depth-dose distributions, output factors, and monitor units. For each parameter, the simulation reproduced accurately the experiment, which attests the quality of the choices made both in the geometrical description and in the physics parameters for beam definition. These encouraging results enable us today to consider a simplification of quality control measurements in the future. Monitor Units calculation is planned to be carried out with preestablished Monte Carlo simulation data. The measurement, which was until now our main patient dose calibration system, will be progressively replaced by computation based on the MCNPX code. This determination of Monitor Units will be controlled by an independent semi-empirical calculation.
Load balancing in highly parallel processing of Monte Carlo code for particle transport
Energy Technology Data Exchange (ETDEWEB)
Higuchi, Kenji; Takemiya, Hiroshi [Japan Atomic Energy Research Inst., Tokyo (Japan); Kawasaki, Takuji [Fuji Research Institute Corporation, Tokyo (Japan)
2001-01-01
In parallel processing of Monte Carlo(MC) codes for neutron, photon and electron transport problems, particle histories are assigned to processors making use of independency of the calculation for each particle. Although we can easily parallelize main part of a MC code by this method, it is necessary and practically difficult to optimize the code concerning load balancing in order to attain high speedup ratio in highly parallel processing. In fact, the speedup ratio in the case of 128 processors remains in nearly one hundred times when using the test bed for the performance evaluation. Through the parallel processing of the MCNP code, which is widely used in the nuclear field, it is shown that it is difficult to attain high performance by static load balancing in especially neutron transport problems, and a load balancing method, which dynamically changes the number of assigned particles minimizing the sum of the computational and communication costs, overcomes the difficulty, resulting in nearly fifteen percentage of reduction for execution time. (author)
SU-E-T-578: MCEBRT, A Monte Carlo Code for External Beam Treatment Plan Verifications
Energy Technology Data Exchange (ETDEWEB)
Chibani, O; Ma, C [Fox Chase Cancer Center, Philadelphia, PA (United States); Eldib, A [Fox Chase Cancer Center, Philadelphia, PA (United States); Al-Azhar University, Cairo (Egypt)
2014-06-01
Purpose: Present a new Monte Carlo code (MCEBRT) for patient-specific dose calculations in external beam radiotherapy. The code MLC model is benchmarked and real patient plans are re-calculated using MCEBRT and compared with commercial TPS. Methods: MCEBRT is based on the GEPTS system (Med. Phys. 29 (2002) 835–846). Phase space data generated for Varian linac photon beams (6 – 15 MV) are used as source term. MCEBRT uses a realistic MLC model (tongue and groove, rounded ends). Patient CT and DICOM RT files are used to generate a 3D patient phantom and simulate the treatment configuration (gantry, collimator and couch angles; jaw positions; MLC sequences; MUs). MCEBRT dose distributions and DVHs are compared with those from TPS in absolute way (Gy). Results: Calculations based on the developed MLC model closely matches transmission measurements (pin-point ionization chamber at selected positions and film for lateral dose profile). See Fig.1. Dose calculations for two clinical cases (whole brain irradiation with opposed beams and lung case with eight fields) are carried out and outcomes are compared with the Eclipse AAA algorithm. Good agreement is observed for the brain case (Figs 2-3) except at the surface where MCEBRT dose can be higher by 20%. This is due to better modeling of electron contamination by MCEBRT. For the lung case an overall good agreement (91% gamma index passing rate with 3%/3mm DTA criterion) is observed (Fig.4) but dose in lung can be over-estimated by up to 10% by AAA (Fig.5). CTV and PTV DVHs from TPS and MCEBRT are nevertheless close (Fig.6). Conclusion: A new Monte Carlo code is developed for plan verification. Contrary to phantombased QA measurements, MCEBRT simulate the exact patient geometry and tissue composition. MCEBRT can be used as extra verification layer for plans where surface dose and tissue heterogeneity are an issue.
Energy Technology Data Exchange (ETDEWEB)
Zehtabian, M; Zaker, N; Sina, S [Shiraz University, Shiraz, Fars (Iran, Islamic Republic of); Meigooni, A Soleimani [Comprehensive Cancer Center of Nevada, Las Vegas, Nevada (United States)
2015-06-15
Purpose: Different versions of MCNP code are widely used for dosimetry purposes. The purpose of this study is to compare different versions of the MCNP codes in dosimetric evaluation of different brachytherapy sources. Methods: The TG-43 parameters such as dose rate constant, radial dose function, and anisotropy function of different brachytherapy sources, i.e. Pd-103, I-125, Ir-192, and Cs-137 were calculated in water phantom. The results obtained by three versions of Monte Carlo codes (MCNP4C, MCNPX, MCNP5) were compared for low and high energy brachytherapy sources. Then the cross section library of MCNP4C code was changed to ENDF/B-VI release 8 which is used in MCNP5 and MCNPX codes. Finally, the TG-43 parameters obtained using the MCNP4C-revised code, were compared with other codes. Results: The results of these investigations indicate that for high energy sources, the differences in TG-43 parameters between the codes are less than 1% for Ir-192 and less than 0.5% for Cs-137. However for low energy sources like I-125 and Pd-103, large discrepancies are observed in the g(r) values obtained by MCNP4C and the two other codes. The differences between g(r) values calculated using MCNP4C and MCNP5 at the distance of 6cm were found to be about 17% and 28% for I-125 and Pd-103 respectively. The results obtained with MCNP4C-revised and MCNPX were similar. However, the maximum difference between the results obtained with the MCNP5 and MCNP4C-revised codes was 2% at 6cm. Conclusion: The results indicate that using MCNP4C code for dosimetry of low energy brachytherapy sources can cause large errors in the results. Therefore it is recommended not to use this code for low energy sources, unless its cross section library is changed. Since the results obtained with MCNP4C-revised and MCNPX were similar, it is concluded that the difference between MCNP4C and MCNPX is their cross section libraries.
Energy Technology Data Exchange (ETDEWEB)
Kozier, K. S. [Atomic Energy of Canada Limited, Chalk River Laboratories, Chalk River, Ont. K0J 1J0 (Canada)
2006-07-01
This paper examines the sensitivity of MCNP5 k{sub eff} results to various deuterium data files for a simple benchmark problem consisting of an 8.4-cm radius sphere of uranium surrounded by an annulus of deuterium at the nuclide number density corresponding to heavy water. This study was performed to help clarify why {Delta}k{sub eff} values of about 10 mk are obtained when different ENDF/B deuterium data files are used in simulations of critical experiments involving solutions of high-enrichment uranyl fluoride in heavy water, while simulations of low-leakage, heterogeneous critical lattices of natural-uranium fuel rods in heavy water show differences of <1 mk. The benchmark calculations were performed as a function of deuterium reflector thickness for several uranium compositions using deuterium ACE files derived from ENDF/B-VII.b1 (release beta 1), ENDF/B-VI.4 and JENDL-3.3, which differ primarily in the energy/angle distributions for elastic scattering <3.2 MeV. Calculations were also performed using modified ACE files having equiprobable cosine bin values in the centre-of-mass reference frame in a progressive manner with increasing energy. It was found that the {Delta}k{sub eff} values increased with deuterium reflector thickness and uranium enrichment. The studies using modified ACE files indicate that most of the reactivity differences arise at energies <1 MeV; hence, this energy range should be given priority if new scattering distribution measurements are undertaken. (authors)
Mesh-based Monte Carlo code for fluorescence modeling in complex tissues with irregular boundaries
Wilson, Robert H.; Chen, Leng-Chun; Lloyd, William; Kuo, Shiuhyang; Marcelo, Cynthia; Feinberg, Stephen E.; Mycek, Mary-Ann
2011-07-01
There is a growing need for the development of computational models that can account for complex tissue morphology in simulations of photon propagation. We describe the development and validation of a user-friendly, MATLAB-based Monte Carlo code that uses analytically-defined surface meshes to model heterogeneous tissue geometry. The code can use information from non-linear optical microscopy images to discriminate the fluorescence photons (from endogenous or exogenous fluorophores) detected from different layers of complex turbid media. We present a specific application of modeling a layered human tissue-engineered construct (Ex Vivo Produced Oral Mucosa Equivalent, EVPOME) designed for use in repair of oral tissue following surgery. Second-harmonic generation microscopic imaging of an EVPOME construct (oral keratinocytes atop a scaffold coated with human type IV collagen) was employed to determine an approximate analytical expression for the complex shape of the interface between the two layers. This expression can then be inserted into the code to correct the simulated fluorescence for the effect of the irregular tissue geometry.
The FLUKA code for application of Monte Carlo methods to promote high precision ion beam therapy
Parodi, K; Cerutti, F; Ferrari, A; Mairani, A; Paganetti, H; Sommerer, F
2010-01-01
Monte Carlo (MC) methods are increasingly being utilized to support several aspects of commissioning and clinical operation of ion beam therapy facilities. In this contribution two emerging areas of MC applications are outlined. The value of MC modeling to promote accurate treatment planning is addressed via examples of application of the FLUKA code to proton and carbon ion therapy at the Heidelberg Ion Beam Therapy Center in Heidelberg, Germany, and at the Proton Therapy Center of Massachusetts General Hospital (MGH) Boston, USA. These include generation of basic data for input into the treatment planning system (TPS) and validation of the TPS analytical pencil-beam dose computations. Moreover, we review the implementation of PET/CT (Positron-Emission-Tomography / Computed- Tomography) imaging for in-vivo verification of proton therapy at MGH. Here, MC is used to calculate irradiation-induced positron-emitter production in tissue for comparison with the +-activity measurement in order to infer indirect infor...
Space applications of the MITS electron-photon Monte Carlo transport code system
Energy Technology Data Exchange (ETDEWEB)
Kensek, R.P.; Lorence, L.J.; Halbleib, J.A. [Sandia National Labs., Albuquerque, NM (United States); Morel, J.E. [Los Alamos National Lab., NM (United States)
1996-07-01
The MITS multigroup/continuous-energy electron-photon Monte Carlo transport code system has matured to the point that it is capable of addressing more realistic three-dimensional adjoint applications. It is first employed to efficiently predict point doses as a function of source energy for simple three-dimensional experimental geometries exposed to simulated uniform isotropic planar sources of monoenergetic electrons up to 4.0 MeV. Results are in very good agreement with experimental data. It is then used to efficiently simulate dose to a detector in a subsystem of a GPS satellite due to its natural electron environment, employing a relatively complex model of the satellite. The capability for survivability analysis of space systems is demonstrated, and results are obtained with and without variance reduction.
Premar-2: a Monte Carlo code for radiative transport simulation in atmospheric environments
Energy Technology Data Exchange (ETDEWEB)
Cupini, E. [ENEA, Centro Ricerche Ezio Clementel, Bologna, (Italy). Dipt. Innovazione
1999-07-01
The peculiarities of the PREMAR-2 code, aimed at radiation transport Monte Carlo simulation in atmospheric environments in the infrared-ultraviolet frequency range, are described. With respect to the previously developed PREMAR code, besides plane multilayers, spherical multilayers and finite sequences of vertical layers, each one with its own atmospheric behaviour, are foreseen in the new code, together with the refraction phenomenon, so that long range, highly slanted paths can now be more faithfully taken into account. A zenithal angular dependence of the albedo coefficient has moreover been introduced. Lidar systems, with spatially independent source and telescope, are allowed again to be simulated, and, in this latest version of the code, sensitivity analyses to be performed. According to this last feasibility, consequences on radiation transport of small perturbations in physical components of the atmospheric environment may be analyze and the related effects on searched results estimated. The availability of a library of physical data (reaction coefficients, phase functions and refraction indexes) is required by the code, providing the essential features of the environment of interest needed of the Monte Carlo simulation. Variance reducing techniques have been enhanced in the Premar-2 code, by introducing, for instance, a local forced collision technique, especially apt to be used in Lidar system simulations. Encouraging comparisons between code and experimental results carried out at the Brasimone Centre of ENEA, have so far been obtained, even if further checks of the code are to be performed. [Italian] Nel presente rapporto vengono descritte le principali caratteristiche del codice di calcolo PREMAR-2, che esegue la simulazione Montecarlo del trasporto della radiazione elettromagnetica nell'atmosfera, nell'intervallo di frequenza che va dall'infrarosso all'ultravioletto. Rispetto al codice PREMAR precedentemente sviluppato, il codice
Energy Technology Data Exchange (ETDEWEB)
Penna, Rodrigo [UNI-BH, Belo Horizonte, MG (Brazil). Dept. de Ciencias Biologicas, Ambientais e da Saude (DCBAS/DCET); Silva, Clemente Jose Gusmao Carneiro da [Universidade Estadual de Santa Cruz, UESC, Ilheus, BA (Brazil); Gomes, Paulo Mauricio Costa [Universidade FUMEC, Belo Horizonte, MG (Brazil)
2008-07-01
Viability of building a nuclear wood densimeter based on low energy photons Compton scattering was done using Monte Carlo code (MCNP- 4C). It is simulated a collimated 60 keV beam of gamma rays emitted by {sup 241}Am source reaching wood blocks. Backscattered radiation by these blocks was calculated. Photons scattered were correlated with blocks of different wood densities. Results showed a linear relationship on wood density and scattered photons, therefore the viability of this wood densimeter. (author)
egs_brachy: a versatile and fast Monte Carlo code for brachytherapy
Chamberland, Marc J. P.; Taylor, Randle E. P.; Rogers, D. W. O.; Thomson, Rowan M.
2016-12-01
egs_brachy is a versatile and fast Monte Carlo (MC) code for brachytherapy applications. It is based on the EGSnrc code system, enabling simulation of photons and electrons. Complex geometries are modelled using the EGSnrc C++ class library and egs_brachy includes a library of geometry models for many brachytherapy sources, in addition to eye plaques and applicators. Several simulation efficiency enhancing features are implemented in the code. egs_brachy is benchmarked by comparing TG-43 source parameters of three source models to previously published values. 3D dose distributions calculated with egs_brachy are also compared to ones obtained with the BrachyDose code. Well-defined simulations are used to characterize the effectiveness of many efficiency improving techniques, both as an indication of the usefulness of each technique and to find optimal strategies. Efficiencies and calculation times are characterized through single source simulations and simulations of idealized and typical treatments using various efficiency improving techniques. In general, egs_brachy shows agreement within uncertainties with previously published TG-43 source parameter values. 3D dose distributions from egs_brachy and BrachyDose agree at the sub-percent level. Efficiencies vary with radionuclide and source type, number of sources, phantom media, and voxel size. The combined effects of efficiency-improving techniques in egs_brachy lead to short calculation times: simulations approximating prostate and breast permanent implant (both with (2 mm)3 voxels) and eye plaque (with (1 mm)3 voxels) treatments take between 13 and 39 s, on a single 2.5 GHz Intel Xeon E5-2680 v3 processor core, to achieve 2% average statistical uncertainty on doses within the PTV. egs_brachy will be released as free and open source software to the research community.
Improvements in the Monte Carlo code for simulating 4πβ(PC)-γ coincidence system measurements
Dias, M. S.; Takeda, M. N.; Toledo, F.; Brancaccio, F.; Tongu, M. L. O.; Koskinas, M. F.
2013-01-01
A Monte Carlo simulation code known as ESQUEMA has been developed by the Nuclear Metrology Laboratory (Laboratório de Metrologia Nuclear-LMN) in the Nuclear and Energy Research Institute (Instituto de Pesquisas Energéticas e Nucleares-IPEN) to be used as a benchmark for radionuclide standardization. The early version of this code simulated only β-γ and ec-γ emitters with reasonably high electron and X-ray energies. To extend the code to include other radionuclides and enable the code to be applied to software coincidence counting systems, several improvements have been made and are presented in this work.
A user`s manual for MASH 1.0: A Monte Carlo Adjoint Shielding Code System
Energy Technology Data Exchange (ETDEWEB)
Johnson, J.O. [ed.
1992-03-01
The Monte Carlo Adjoint Shielding Code System, MASH, calculates neutron and gamma-ray environments and radiation protection factors for armored military vehicles, structures, trenches, and other shielding configurations by coupling a forward discrete ordinates air-over-ground transport calculation with an adjoint Monte Carlo treatment of the shielding geometry. Efficiency and optimum use of computer time are emphasized. The code system include the GRTUNCL and DORT codes for air-over-ground transport calculations, the MORSE code with the GIFT5 combinatorial geometry package for adjoint shielding calculations, and several peripheral codes that perform the required data preparations, transformations, and coupling functions. MASH is the successor to the Vehicle Code System (VCS) initially developed at Oak Ridge National Laboratory (ORNL). The discrete ordinates calculation determines the fluence on a coupling surface surrounding the shielding geometry due to an external neutron/gamma-ray source. The Monte Carlo calculation determines the effectiveness of the fluence at that surface in causing a response in a detector within the shielding geometry, i.e., the ``dose importance`` of the coupling surface fluence. A coupling code folds the fluence together with the dose importance, giving the desired dose response. The coupling code can determine the dose response a a function of the shielding geometry orientation relative to the source, distance from the source, and energy response of the detector. This user`s manual includes a short description of each code, the input required to execute the code along with some helpful input data notes, and a representative sample problem (input data and selected output edits) for each code.
A user's manual for MASH 1. 0: A Monte Carlo Adjoint Shielding Code System
Energy Technology Data Exchange (ETDEWEB)
Johnson, J.O. (ed.)
1992-03-01
The Monte Carlo Adjoint Shielding Code System, MASH, calculates neutron and gamma-ray environments and radiation protection factors for armored military vehicles, structures, trenches, and other shielding configurations by coupling a forward discrete ordinates air-over-ground transport calculation with an adjoint Monte Carlo treatment of the shielding geometry. Efficiency and optimum use of computer time are emphasized. The code system include the GRTUNCL and DORT codes for air-over-ground transport calculations, the MORSE code with the GIFT5 combinatorial geometry package for adjoint shielding calculations, and several peripheral codes that perform the required data preparations, transformations, and coupling functions. MASH is the successor to the Vehicle Code System (VCS) initially developed at Oak Ridge National Laboratory (ORNL). The discrete ordinates calculation determines the fluence on a coupling surface surrounding the shielding geometry due to an external neutron/gamma-ray source. The Monte Carlo calculation determines the effectiveness of the fluence at that surface in causing a response in a detector within the shielding geometry, i.e., the dose importance'' of the coupling surface fluence. A coupling code folds the fluence together with the dose importance, giving the desired dose response. The coupling code can determine the dose response a a function of the shielding geometry orientation relative to the source, distance from the source, and energy response of the detector. This user's manual includes a short description of each code, the input required to execute the code along with some helpful input data notes, and a representative sample problem (input data and selected output edits) for each code.
Energy Technology Data Exchange (ETDEWEB)
William Martin
2012-11-16
A new method to obtain Doppler broadened cross sections has been implemented into MCNP, removing the need to generate cross sections for isotopes at problem temperatures. Previous work had established the scientific feasibility of obtaining Doppler-broadened cross sections "on-the-fly" (OTF) during the random walk of the neutron. Thus, when a neutron of energy E enters a material region that is at some temperature T, the cross sections for that material at the exact temperature T are immediately obtained by interpolation using a high order functional expansion for the temperature dependence of the Doppler-broadened cross section for that isotope at the neutron energy E. A standalone Fortran code has been developed that generates the OTF library for any isotope that can be processed by NJOY. The OTF cross sections agree with the NJOY-based cross sections for all neutron energies and all temperatures in the range specified by the user, e.g., 250K - 3200K. The OTF methodology has been successfully implemented into the MCNP Monte Carlo code and has been tested on several test problems by comparing MCNP with conventional ACE cross sections versus MCNP with OTF cross sections. The test problems include the Doppler defect reactivity benchmark suite and two full-core VHTR configurations, including one with multiphysics coupling using RELAP5-3D/ATHENA for the thermal-hydraulic analysis. The comparison has been excellent, verifying that the OTF libraries can be used in place of the conventional ACE libraries generated at problem temperatures. In addition, it has been found that using OTF cross sections greatly reduces the complexity of the input for MCNP, especially for full-core temperature feedback calculations with many temperature regions. This results in an order of magnitude decrease in the number of input lines for full-core configurations, thus simplifying input preparation and reducing the potential for input errors. Finally, for full-core problems with multiphysics
The application of the Monte-Carlo neutron transport code MCNP to a small "nuclear battery" system
Puigdellívol Sadurní, Roger
2009-01-01
The project consist in calculate the keff to a small nuclear battery. The code Monte- Carlo neutron transport code MCNP is used to calculate the keff. The calculations are done at the beginning of life to know the capacity of the core becomes critical in different conditions. These conditions are the study parameters that determine the criticality of the core. These parameters are the uranium enrichment, the coated particles (TRISO) packing factor and the size of the core. More...
Criticality coefficient calculation for a small PWR using Monte Carlo Transport Code
Energy Technology Data Exchange (ETDEWEB)
Trombetta, Debora M.; Su, Jian, E-mail: dtrombetta@nuclear.ufrj.br, E-mail: sujian@nuclear.ufrj.br [Coordenacao dos Programas de Pos-Graduacao em Engenharia (COPPE/UFRJ), Rio de Janeiro, RJ (Brazil); Chirayath, Sunil S., E-mail: sunilsc@tamu.edu [Department of Nuclear Engineering and Nuclear Security Science and Policy Institute, Texas A and M University, TX (United States)
2015-07-01
Computational models of reactors are increasingly used to predict nuclear reactor physics parameters responsible for reactivity changes which could lead to accidents and losses. In this work, preliminary results for criticality coefficient calculation using the Monte Carlo transport code MCNPX were presented for a small PWR. The computational modeling developed consists of the core with fuel elements, radial reflectors, and control rods inside a pressure vessel. Three different geometries were simulated, a single fuel pin, a fuel assembly and the core, with the aim to compare the criticality coefficients among themselves.The criticality coefficients calculated were: Doppler Temperature Coefficient, Coolant Temperature Coefficient, Coolant Void Coefficient, Power Coefficient, and Control Rod Worth. The coefficient values calculated by the MCNP code were compared with literature results, showing good agreement with reference data, which validate the computational model developed and allow it to be used to perform more complex studies. Criticality Coefficient values for the three simulations done had little discrepancy for almost all coefficients investigated, the only exception was the Power Coefficient. Preliminary results presented show that simple modelling as a fuel assembly can describe changes at almost all the criticality coefficients, avoiding the need of a complex core simulation. (author)
Development of Monte Carlo code for coincidence prompt gamma-ray neutron activation analysis
Han, Xiaogang
Prompt Gamma-Ray Neutron Activation Analysis (PGNAA) offers a non-destructive, relatively rapid on-line method for determination of elemental composition of bulk and other samples. However, PGNAA has an inherently large background. These backgrounds are primarily due to the presence of the neutron excitation source. It also includes neutron activation of the detector and the prompt gamma rays from the structure materials of PGNAA devices. These large backgrounds limit the sensitivity and accuracy of PGNAA. Since most of the prompt gamma rays from the same element are emitted in coincidence, a possible approach for further improvement is to change the traditional PGNAA measurement technique and introduce the gamma-gamma coincidence technique. It is well known that the coincidence techniques can eliminate most of the interference backgrounds and improve the signal-to-noise ratio. A new Monte Carlo code, CEARCPG has been developed at CEAR to simulate gamma-gamma coincidence spectra in PGNAA experiment. Compared to the other existing Monte Carlo code CEARPGA I and CEARPGA II, a new algorithm of sampling the prompt gamma rays produced from neutron capture reaction and neutron inelastic scattering reaction, is developed in this work. All the prompt gamma rays are taken into account by using this new algorithm. Before this work, the commonly used method is to interpolate the prompt gamma rays from the pre-calculated gamma-ray table. This technique works fine for the single spectrum. However it limits the capability to simulate the coincidence spectrum. The new algorithm samples the prompt gamma rays from the nucleus excitation scheme. The primary nuclear data library used to sample the prompt gamma rays comes from ENSDF library. Three cases are simulated and the simulated results are benchmarked with experiments. The first case is the prototype for ETI PGNAA application. This case is designed to check the capability of CEARCPG for single spectrum simulation. The second
A User's Manual for MASH V1.5 - A Monte Carlo Adjoint Shielding Code System
Energy Technology Data Exchange (ETDEWEB)
C. O. Slater; J. M. Barnes; J. O. Johnson; J.D. Drischler
1998-10-01
The Monte Carlo ~djoint ~ielding Code System, MASH, calculates neutron and gamma- ray environments and radiation protection factors for armored military vehicles, structures, trenches, and other shielding configurations by coupling a forward discrete ordinates air- over-ground transport calculation with an adjoint Monte Carlo treatment of the shielding geometry. Efficiency and optimum use of computer time are emphasized. The code system includes the GRTUNCL and DORT codes for air-over-ground transport calculations, the MORSE code with the GIFT5 combinatorial geometry package for adjoint shielding calculations, and several peripheral codes that perform the required data preparations, transformations, and coupling functions. The current version, MASH v 1.5, is the successor to the original MASH v 1.0 code system initially developed at Oak Ridge National Laboratory (ORNL). The discrete ordinates calculation determines the fluence on a coupling surface surrounding the shielding geometry due to an external neutron/gamma-ray source. The Monte Carlo calculation determines the effectiveness of the fluence at that surface in causing a response in a detector within the shielding geometry, i.e., the "dose importance" of the coupling surface fluence. A coupling code folds the fluence together with the dose importance, giving the desired dose response. The coupling code can determine the dose response as a function of the shielding geometry orientation relative to the source, distance from the source, and energy response of the detector. This user's manual includes a short description of each code, the input required to execute the code along with some helpful input data notes, and a representative sample problem.
Monte Carlo calculation of energy deposition in ionization chambers for tritium measurements
Zhilin, Chen; Shuming, Peng; Dan, Meng; Yuehong, He; Heyi, Wang
2014-10-01
Energy deposition in ionization chambers for tritium measurements has been theoretically studied using Monte Carlo code MCNP 5. The influence of many factors, including carrier gas, chamber size, wall materials and gas pressure, has been evaluated in the simulations. It is found that β rays emitted by tritium deposit much more energy into chambers flowing through with argon than with deuterium in them, as much as 2.7 times higher at pressure 100 Pa. As chamber size gets smaller, energy deposition decreases sharply. For an ionization chamber of 1 mL, β rays deposit less than 1% of their energy at pressure 100 Pa and only 84% even if gas pressure is as high as 100 kPa. It also indicates that gold plated ionization chamber results in the highest deposition ratio while aluminum one leads to the lowest. In addition, simulations were validated by comparison with experimental data. Results show that simulations agree well with experimental data.
A Comparison Between GATE and MCNPX Monte Carlo Codes in Simulation of Medical Linear Accelerator
Sadoughi, Hamid-Reza; Nasseri, Shahrokh; Momennezhad, Mahdi; Sadeghi, Hamid-Reza; Bahreyni-Toosi, Mohammad-Hossein
2014-01-01
Radiotherapy dose calculations can be evaluated by Monte Carlo (MC) simulations with acceptable accuracy for dose prediction in complicated treatment plans. In this work, Standard, Livermore and Penelope electromagnetic (EM) physics packages of GEANT4 application for tomographic emission (GATE) 6.1 were compared versus Monte Carlo N-Particle eXtended (MCNPX) 2.6 in simulation of 6 MV photon Linac. To do this, similar geometry was used for the two codes. The reference values of percentage depth dose (PDD) and beam profiles were obtained using a 6 MV Elekta Compact linear accelerator, Scanditronix water phantom and diode detectors. No significant deviations were found in PDD, dose profile, energy spectrum, radial mean energy and photon radial distribution, which were calculated by Standard and Livermore EM models and MCNPX, respectively. Nevertheless, the Penelope model showed an extreme difference. Statistical uncertainty in all the simulations was MCNPX, Standard, Livermore and Penelope models, respectively. Differences between spectra in various regions, in radial mean energy and in photon radial distribution were due to different cross section and stopping power data and not the same simulation of physics processes of MCNPX and three EM models. For example, in the Standard model, the photoelectron direction was sampled from the Gavrila-Sauter distribution, but the photoelectron moved in the same direction of the incident photons in the photoelectric process of Livermore and Penelope models. Using the same primary electron beam, the Standard and Livermore EM models of GATE and MCNPX showed similar output, but re-tuning of primary electron beam is needed for the Penelope model. PMID:24696804
Full modelling of the MOSAIC animal PET system based on the GATE Monte Carlo simulation code
Merheb, C.; Petegnief, Y.; Talbot, J. N.
2007-02-01
Positron emission tomography (PET) systems dedicated to animal imaging are now widely used for biological studies. The scanner performance strongly depends on the design and the characteristics of the system. Many parameters must be optimized like the dimensions and type of crystals, geometry and field-of-view (FOV), sampling, electronics, lightguide, shielding, etc. Monte Carlo modelling is a powerful tool to study the effect of each of these parameters on the basis of realistic simulated data. Performance assessment in terms of spatial resolution, count rates, scatter fraction and sensitivity is an important prerequisite before the model can be used instead of real data for a reliable description of the system response function or for optimization of reconstruction algorithms. The aim of this study is to model the performance of the Philips Mosaic™ animal PET system using a comprehensive PET simulation code in order to understand and describe the origin of important factors that influence image quality. We use GATE, a Monte Carlo simulation toolkit for a realistic description of the ring PET model, the detectors, shielding, cap, electronic processing and dead times. We incorporate new features to adjust signal processing to the Anger logic underlying the Mosaic™ system. Special attention was paid to dead time and energy spectra descriptions. Sorting of simulated events in a list mode format similar to the system outputs was developed to compare experimental and simulated sensitivity and scatter fractions for different energy thresholds using various models of phantoms describing rat and mouse geometries. Count rates were compared for both cylindrical homogeneous phantoms. Simulated spatial resolution was fitted to experimental data for 18F point sources at different locations within the FOV with an analytical blurring function for electronic processing effects. Simulated and measured sensitivities differed by less than 3%, while scatter fractions agreed
M CNP5模拟γ射线反散射谱的影响因素%MCNP5 study of the Influence Factor on Gamma-Ray Backscattering Spectrum
Institute of Scientific and Technical Information of China (English)
肖明; 艾尔肯·阿不列木
2014-01-01
为了研究γ射线反散射峰与散射体的物质成分、厚度、入射射线能量和几何布置之间的关系。本论文基于蒙特卡罗方法，运用MCNP5程序模拟放射源137 Cs、60 Co发出的γ射线经过不同厚度的石蜡、玻璃、Al、Fe、Cu和Pb散射后反散射谱的变化，所得结果与实验谱符合较好。结果显示：散射体厚度与原子序数同时增加且原子序数大约到26以后，反散射峰值才随原子序数增加而减小；探测器与放射源的距离为10 mm时，137 Cs、60 Co发出的γ射线经Fe、Cu散射后，Fe、Cu的厚度分别为1．6 cm和2．4 cm时，铁的峰值高于铜；反散射峰值随源与探测器之间的距离增加而减小，与入射射线能量无关。试验结果对进一步开展反散射在工业，农业和医疗业的辐射屏蔽的研究有一定的指导作用。%This paper aims to study the correlation of the backscattering peak of gamma ray with the material composition, thickness of the scatter, incident (-ray energy and geometric arrangement.Based on the method of Monte Carlo, MCNP5 program was used to simulate the changes of the backscattering spectrum after gamma phone emitted by 137 Cs, 60 Co through different thickness of paraffin wax and glass, Al, Fe, Cu and Pb.The MCNP5 simulation result is in very good agreement with the experimental data.The results showed that the backscattering is increased with the thickness of scatter and with atomic number, until the atomic number about to 26 and later.When the distance between the detector and radiation source is 10mm the peak value of the 137 Cs and 60 Co scattered by Fe is higher than that of Cu with the thickness of 1.6 cm and 2.4 cm, respectively. When the distance between the source and the detector increases, the Backscattering peak is reduced and this has nothing to do with the incident ray energy.Our results are useful for applying the Backscattering method to the radiation shield of the industry
Energy Technology Data Exchange (ETDEWEB)
Galicia A, J.; Francois L, J. L.; Bastida O, G. E. [UNAM, Facultad de Ingenieria, Departamento de Sistemas Energeticos, Ciudad Universitaria, 04510 Ciudad de Mexico (Mexico); Esquivel E, J., E-mail: blink19871@hotmail.com [ININ, Carretera Mexico-Toluca s/n, 52750 Ocoyoacac, Estado de Mexico (Mexico)
2016-09-15
The development of the AZTLAN platform for the analysis and design of nuclear reactors is led by Instituto Nacional de Investigaciones Nucleares (ININ) and divided into four working groups, which have well-defined activities to achieve significant progress in this project individually and jointly. Within these working groups is the users group, whose main task is to use the codes that make up the AZTLAN platform to provide feedback to the developers, and in this way to make the final versions of the codes are efficient and at the same time reliable and easy to understand. In this paper we present the results provided by the AZNHEX v.1.0 code when simulating the core of a fast reactor cooled with sodium at steady state. The validation of these results is a fundamental part of the platform development and responsibility of the users group, so in this research the results obtained with AZNHEX are compared and analyzed with those provided by the Monte Carlo code MCNP-5, software worldwide used and recognized. A description of the methodology used with MCNP-5 is also presented for the calculation of the interest variables and the difference that is obtained with respect to the calculated with AZNHEX. (Author)
Directory of Open Access Journals (Sweden)
Hadad K
2015-03-01
Full Text Available Background: HDR brachytherapy is one of the commonest methods of nasopharyngeal cancer treatment. In this method, depending on how advanced one tumor is, 2 to 6 Gy dose as intracavitary brachytherapy is prescribed. Due to high dose rate and tumor location, accuracy evaluation of treatment planning system (TPS is particularly important. Common methods used in TPS dosimetry are based on computations in a homogeneous phantom. Heterogeneous phantoms, especially patient-specific voxel phantoms can increase dosimetric accuracy. Materials and Methods: In this study, using CT images taken from a patient and ctcreate-which is a part of the DOSXYZnrc computational code, patient-specific phantom was made. Dose distribution was plotted by DOSXYZnrc and compared with TPS one. Also, by extracting the voxels absorbed dose in treatment volume, dosevolume histograms (DVH was plotted and compared with Oncentra™ TPS DVHs. Results: The results from calculations were compared with data from Oncentra™ treatment planning system and it was observed that TPS calculation predicts lower dose in areas near the source, and higher dose in areas far from the source relative to MC code. Absorbed dose values in the voxels also showed that TPS reports D90 value is 40% higher than the Monte Carlo method. Conclusion: Today, most treatment planning systems use TG-43 protocol. This protocol may results in errors such as neglecting tissue heterogeneity, scattered radiation as well as applicator attenuation. Due to these errors, AAPM emphasized departing from TG-43 protocol and approaching new brachytherapy protocol TG-186 in which patient-specific phantom is used and heterogeneities are affected in dosimetry
Directory of Open Access Journals (Sweden)
Jingang Liang
2016-06-01
Full Text Available Because of prohibitive data storage requirements in large-scale simulations, the memory problem is an obstacle for Monte Carlo (MC codes in accomplishing pin-wise three-dimensional (3D full-core calculations, particularly for whole-core depletion analyses. Various kinds of data are evaluated and quantificational total memory requirements are analyzed based on the Reactor Monte Carlo (RMC code, showing that tally data, material data, and isotope densities in depletion are three major parts of memory storage. The domain decomposition method is investigated as a means of saving memory, by dividing spatial geometry into domains that are simulated separately by parallel processors. For the validity of particle tracking during transport simulations, particles need to be communicated between domains. In consideration of efficiency, an asynchronous particle communication algorithm is designed and implemented. Furthermore, we couple the domain decomposition method with MC burnup process, under a strategy of utilizing consistent domain partition in both transport and depletion modules. A numerical test of 3D full-core burnup calculations is carried out, indicating that the RMC code, with the domain decomposition method, is capable of pin-wise full-core burnup calculations with millions of depletion regions.
Energy Technology Data Exchange (ETDEWEB)
Blazy-Aubignac, L
2007-09-15
The treatment planning systems (T.P.S.) occupy a key position in the radiotherapy service: they realize the projected calculation of the dose distribution and the treatment duration. Traditionally, the quality control of the calculated distribution doses relies on their comparisons with dose distributions measured under the device of treatment. This thesis proposes to substitute these dosimetry measures to the profile of reference dosimetry calculations got by the Penelope Monte-Carlo code. The Monte-Carlo simulations give a broad choice of test configurations and allow to envisage a quality control of dosimetry aspects of T.P.S. without monopolizing the treatment devices. This quality control, based on the Monte-Carlo simulations has been tested on a clinical T.P.S. and has allowed to simplify the quality procedures of the T.P.S.. This quality control, in depth, more precise and simpler to implement could be generalized to every center of radiotherapy. (N.C.)
Modeling Monte Carlo of multileaf collimators using the code GEANT4
Energy Technology Data Exchange (ETDEWEB)
Oliveira, Alex C.H.; Lima, Fernando R.A., E-mail: oliveira.ach@yahoo.com, E-mail: falima@cnen.gov.br [Centro Regional de Ciencias Nucleares do Nordeste (CRCN-NE/CNEN-PE), Recife, PE (Brazil); Lima, Luciano S.; Vieira, Jose W., E-mail: lusoulima@yahoo.com.br [Instituto Federal de Educacao, Ciencia e Tecnologia de Pernambuco (IFPE), Recife, PE (Brazil)
2014-07-01
Radiotherapy uses various techniques and equipment for local treatment of cancer. The equipment most often used in radiotherapy to the patient irradiation is linear accelerator (Linac). Among the many algorithms developed for evaluation of dose distributions in radiotherapy planning, the algorithms based on Monte Carlo (MC) methods have proven to be very promising in terms of accuracy by providing more realistic results. The MC simulations for applications in radiotherapy are divided into two parts. In the first, the simulation of the production of the radiation beam by the Linac is performed and then the phase space is generated. The phase space contains information such as energy, position, direction, etc. of millions of particles (photons, electrons, positrons). In the second part the simulation of the transport of particles (sampled phase space) in certain configurations of irradiation field is performed to assess the dose distribution in the patient (or phantom). Accurate modeling of the Linac head is of particular interest in the calculation of dose distributions for intensity modulated radiation therapy (IMRT), where complex intensity distributions are delivered using a multileaf collimator (MLC). The objective of this work is to describe a methodology for modeling MC of MLCs using code Geant4. To exemplify this methodology, the Varian Millennium 120-leaf MLC was modeled, whose physical description is available in BEAMnrc Users Manual (20 11). The dosimetric characteristics (i.e., penumbra, leakage, and tongue-and-groove effect) of this MLC were evaluated. The results agreed with data published in the literature concerning the same MLC. (author)
Directory of Open Access Journals (Sweden)
Mona Zolfaghari
2015-07-01
Full Text Available Introduction Electron linear accelerator (LINAC can be used for neutron production in Boron Neutron Capture Therapy (BNCT. BNCT is an external radiotherapeutic method for the treatment of some cancers. In this study, Varian 2300 C/D LINAC was simulated as an electron accelerator-based photoneutron source to provide a suitable neutron flux for BNCT. Materials and Methods Photoneutron sources were simulated, using MCNPX Monte Carlo code. In this study, a 20 MeV LINAC was utilized for electron-photon reactions. After the evaluation of cross-sections and threshold energies, lead (Pb, uranium (U and beryllium deuteride (BeD2were selected as photoneutron sources. Results According to the simulation results, optimized photoneutron sources with a compact volume and photoneutron yields of 107, 108 and 109 (n.cm-2.s-1 were obtained for Pb, U and BeD2 composites. Also, photoneutrons increased by using enriched U (10-60% as an electron accelerator-based photoneutron source. Conclusion Optimized photoneutron sources were obtained with compact sizes of 107, 108 and 109 (n.cm-2.s-1, respectively. These fluxs can be applied for BNCT by decelerating fast neutrons and using a suitable beam-shaping assembly, surrounding electron-photon and photoneutron sources.
Directory of Open Access Journals (Sweden)
Mona Zolfaghari
2015-07-01
Full Text Available Introduction Electron linear accelerator (LINAC can be used for neutron production in Boron Neutron Capture Therapy (BNCT. BNCT is an external radiotherapeutic method for the treatment of some cancers. In this study, Varian 2300 C/D LINAC was simulated as an electron accelerator-based photoneutron source to provide a suitable neutron flux for BNCT. Materials and Methods Photoneutron sources were simulated, using MCNPX Monte Carlo code. In this study, a 20 MeV LINAC was utilized for electron-photon reactions. After the evaluation of cross-sections and threshold energies, lead (Pb, uranium (U and beryllium deuteride (BeD2were selected as photoneutron sources. Results According to the simulation results, optimized photoneutron sources with a compact volume and photoneutron yields of 107, 108 and 109 (n.cm-2.s-1 were obtained for Pb, U and BeD2 composites. Also, photoneutrons increased by using enriched U (10-60% as an electron accelerator-based photoneutron source. Conclusion Optimized photoneutron sources were obtained with compact sizes of 107, 108 and 109 (n.cm-2.s-1, respectively. These fluxs can be applied for BNCT by decelerating fast neutrons and using a suitable beam-shaping assembly, surrounding electron-photon and photoneutron sources.
Deep-penetration calculation for the ISIS target station shielding using the MARS Monte Carlo code
Nunomiya, T; Nakamura, T; Nakao, N
2002-01-01
A calculation of neutron penetration through a thick shield was performed with a three-dimensional multi-layer technique using the MARS14(02) Monte Carlo code to compare with the experimental shielding data in 1998 at the ISIS spallation neutron source facility. In this calculation, secondary particles from a tantalum target bombarded by 800-MeV protons were transmitted through a bulk shield of approximately 3-m-thick iron and 1-m-thick concrete. To accomplish this deep-penetration calculation with good statistics, the following three techniques were used in this study. First, the geometry of the bulk shield was three-dimensionally divided into several layers of about 50-cm thickness, and a step-by-step calculation was carried out to multiply the number of penetrated particles at the boundaries between the layers. Second, the source particles in the layers were divided into two parts to maintain the statistical balance on the spatial-flux distribution. Third, only high-energy particles above 20 MeV were trans...
Thyroid cell irradiation by radioiodines: a new Monte Carlo electron track-structure code
Energy Technology Data Exchange (ETDEWEB)
Champion, Christophe [Universite Paul Verlaine-Metz (France). Lab. de Physique Moleculaire et des Collisions]. E-mail: champion@univ-metz.fr; Elbast, Mouhamad; Colas-Linhart, Nicole [Universite Paris 7 (France). Faculte de Medecine. Lab. de Biophysique; Ting-Di Wu [INSERM U759, Orsay (France). Institut Curie Recherche. Imagerie Integrative
2007-09-15
The most significant impact of the Chernobyl accident is the increased incidence of thyroid cancer among children who were exposed to short-lived radioiodines and 131-iodine. In order to accurately estimate the radiation dose provided by these radioiodines, it is necessary to know where iodine is incorporated. To do that, the distribution at the cellular level of newly organified iodine in the immature rat thyroid was performed using secondary ion mass microscopy (NanoSIMS{sup 50}). Actual dosimetric models take only into account the averaged energy and range of beta particles of the radio-elements and may, therefore, imperfectly describe the real distribution of dose deposit at the microscopic level around the point sources. Our approach is radically different since based on a track-structure Monte Carlo code allowing following-up of electrons down to low energies ({approx}= 10 eV) what permits a nanometric description of the irradiation physics. The numerical simulations were then performed by modelling the complete disintegrations of the short-lived iodine isotopes as well as of {sup 131}I in new born rat thyroids in order to take into account accurate histological and biological data for the thyroid gland. (author)
HERMES: a Monte Carlo Code for the Propagation of Ultra-High Energy Nuclei
De Domenico, Manlio; Settimo, Mariangela
2013-01-01
Although the recent experimental efforts to improve the observation of Ultra-High Energy Cosmic Rays (UHECRs) above $10^{18}$ eV, the origin and the composition of such particles is still unknown. In this work, we present the novel Monte Carlo code (HERMES) simulating the propagation of UHE nuclei, in the energy range between $10^{16}$ and $10^{22}$ eV, accounting for propagation in the intervening extragalactic and Galactic magnetic fields and nuclear interactions with relic photons of the extragalactic background radiation. In order to show the potential applications of HERMES for astroparticle studies, we estimate the expected flux of UHE nuclei in different astrophysical scenarios, the GZK horizons and we show the expected arrival direction distributions in the presence of turbulent extragalactic magnetic fields. A stable version of HERMES will be released in the next future for public use together with libraries of already propagated nuclei to allow the community to perform mass composition and energy sp...
Energy Technology Data Exchange (ETDEWEB)
Prettyman, T.H.; Gardner, R.P.; Verghese, K. (North Carolina State Univ., Raleigh, NC (United States). Center for Engineering Applications and Radioisotopes)
1993-08-01
A new specific purpose Monte Carlo code called McENL for modeling the time response of epithermal neutron lifetime tools is described. The code was developed so that the Monte Carlo neophyte can easily use it. A minimum amount of input preparation is required and specified fixed values of the parameters used to control the code operation can be used. The weight windows technique, employing splitting and Russian Roulette, is used with an automated importance function based on the solution of an adjoint diffusion model to improve the code efficiency. Complete composition and density correlated sampling is also included in the code and can be used to study the effect on tool response of small variations in the formation, borehole, or logging tool composition and density. An illustration of the latter application is given here for the density of a thermal neutron filter. McENL was benchmarked against test-pit data for the Mobil pulsed neutron porosity (PNP) tool and found to be very accurate. Results of the experimental validation and details of code performance are presented.
Energy Technology Data Exchange (ETDEWEB)
Salome, Jean A.D.; Cardoso, Fabiano; Faria, Rochkhudson B.; Pereira, Claubia, E-mail: jadsalome@yahoo.com.br, E-mail: fabinuclear@yahoo.com.br, E-mail: rockdefaria@yahoo.com.br, E-mail: claubia@nuclear.ufmg.br [Universidade Federal de Minas Gerais (UFMG), Belo Horizonte, MG (Brazil). Departamento de Engenharia Nuclear
2015-07-01
The IPEN/MB-01 reactor, located in the city of Sao Paulo - Brazil, reached its first criticality on the year of 1988. The reactor is characterized by a low output power of 100 W only, even because its purpose is to produce knowledge about nuclear power plants on a smaller geometric scale without the requirement of an extremely complex cooling system. The use of devices such as this it is very interesting because it achieves the demands of nuclear engineering about the neutronic parameters needed in the design of large nuclear plants through relatively simple and inexpensive methods. In this paper, the computational mathematical code MCNP5 is used to perform the calculation of the thermal neutron flux in the core of the IPEN/MB-01 reactor. To do this is used an experiment from the LEU-COMP-THERM-077 benchmark that represents the standard rectangular configuration of the IPEN/MB-01 reactor. The thermal neutron flux is calculated at some axial planes of different heights and, after that, axial profiles of the thermal neutron flux are done and compared to experimental results issued previously. The experimental values used as reference refer to a cylindrical configuration of the core of the reactor. Finally, the pertinence and relevance of the results are checked. With this work is expected to produce more knowledge about the dynamics of neutron flux in the core of the IPEN/MB-01 reactor. (author)
Energy Technology Data Exchange (ETDEWEB)
Oleynik, D. S., E-mail: oleynik-ds@nrcki.ru [National Research Center Kurchatov Institute (Russian Federation)
2015-12-15
A new version of the tally module of the MCU software package is developed in which the approach for taking directly into account the uncertainty in initial data is implemented that is recommended by the international standard on estimating the uncertainty in results of measuring (ISO 13005). The new module makes it possible to evaluate the effect of uncertainty in initial data (caused by technological tolerances in fabrication of structural members of the core) on neutronic characteristics of the reactor. The developed software is adapted to parallel computing with the use of multiprocessor computers, which significantly reduces the computation time: the parallelization coefficient is almost equal to 1. Testing is performed by examples of solving the problem on criticality for the Godiva benchmark experiment and also for the infinite lattice of fuel assemblies of the VVER-440, VVER-1000, and VVER-1200. The results of calculations of the uncertainty in neutronic characteristics (effective multiplication factor, fission reaction rate), which is caused by uncertainties in initial data due to technological tolerances, are compared (in the first case) to the published results obtained using the precision MCNP5 code and (in the second case) to those obtained by means of the RADAR engineering program. A good agreement of results is achieved for all cases.
Oleynik, D. S.
2015-12-01
A new version of the tally module of the MCU software package is developed in which the approach for taking directly into account the uncertainty in initial data is implemented that is recommended by the international standard on estimating the uncertainty in results of measuring (ISO 13005). The new module makes it possible to evaluate the effect of uncertainty in initial data (caused by technological tolerances in fabrication of structural members of the core) on neutronic characteristics of the reactor. The developed software is adapted to parallel computing with the use of multiprocessor computers, which significantly reduces the computation time: the parallelization coefficient is almost equal to 1. Testing is performed by examples of solving the problem on criticality for the Godiva benchmark experiment and also for the infinite lattice of fuel assemblies of the VVER-440, VVER-1000, and VVER-1200. The results of calculations of the uncertainty in neutronic characteristics (effective multiplication factor, fission reaction rate), which is caused by uncertainties in initial data due to technological tolerances, are compared (in the first case) to the published results obtained using the precision MCNP5 code and (in the second case) to those obtained by means of the RADAR engineering program. A good agreement of results is achieved for all cases.
Energy Technology Data Exchange (ETDEWEB)
Pessanha, Paula Rocha; Queiroz Filho, Pedro Pacheco de; Santos, Denison de Souza, E-mail: pessanha.paular@gmail.com, E-mail: queiroz@ird.gov.br, E-mail: santosd@ird.gov.br [Instituto de Radioprotecao e Dosimetria (IRD/CNEN-RJ),Rio de Janeiro, RJ (Brazil)
2014-07-01
In this work, the implementation of a hand and forearm Geant4 phantom code, for further evaluation of occupational exposure of ends of the radionuclides decay manipulated during procedures involving the use of injection syringe. The simulation model offered by Geant4 includes a full set of features, with the reconstruction of trajectories, geometries and physical models. For this work, the values calculated in the simulation are compared with the measurements rates by thermoluminescent dosimeters (TLDs) in physical phantom REMAB®. From the analysis of the data obtained through simulation and experimentation, of the 14 points studied, there was a discrepancy of only 8.2% of kerma values found, and these figures are considered compatible. The geometric phantom implemented in Geant4 Monte Carlo code was validated and can be used later for the evaluation of doses at ends.
Comparative Dosimetric Estimates of a 25 keV Electron Micro-beam with three Monte Carlo Codes
Energy Technology Data Exchange (ETDEWEB)
Mainardi, Enrico; Donahue, Richard J.; Blakely, Eleanor A.
2002-09-11
The calculations presented compare the different performances of the three Monte Carlo codes PENELOPE-1999, MCNP-4C and PITS, for the evaluation of Dose profiles from a 25 keV electron micro-beam traversing individual cells. The overall model of a cell is a water cylinder equivalent for the three codes but with a different internal scoring geometry: hollow cylinders for PENELOPE and MCNP, whereas spheres are used for the PITS code. A cylindrical cell geometry with scoring volumes with the shape of hollow cylinders was initially selected for PENELOPE and MCNP because of its superior simulation of the actual shape and dimensions of a cell and for its improved computer-time efficiency if compared to spherical internal volumes. Some of the transfer points and energy transfer that constitute a radiation track may actually fall in the space between spheres, that would be outside the spherical scoring volume. This internal geometry, along with the PENELOPE algorithm, drastically reduced the computer time when using this code if comparing with event-by-event Monte Carlo codes like PITS. This preliminary work has been important to address dosimetric estimates at low electron energies. It demonstrates that codes like PENELOPE can be used for Dose evaluation, even with such small geometries and energies involved, which are far below the normal use for which the code was created. Further work (initiated in Summer 2002) is still needed however, to create a user-code for PENELOPE that allows uniform comparison of exact cell geometries, integral volumes and also microdosimetric scoring quantities, a field where track-structure codes like PITS, written for this purpose, are believed to be superior.
Energy Technology Data Exchange (ETDEWEB)
Gallardo, S.; Querol, A.; Ortiz, J.; Rodenas, J.; Verdu, G.
2014-07-01
In this paper the use of Monte Carlo code SWORD-GEANT is proposed to simulate an ultra pure germanium detector High Purity Germanium detector (HPGe) detector ORTEC specifically GMX40P4, coaxial geometry. (Author)
Energy Technology Data Exchange (ETDEWEB)
Valentine, T.; Perez, R. [Oak Ridge National Lab., TN (United States); Rugama, Y.; Munoz-Cobo, J.L. [Poly. Tech. Univ. of Valencia (Spain). Chemical and Nuclear Engineering Dept.
2001-07-01
The design of reactivity monitoring systems for accelerator-driven systems must be investigated to ensure that such systems remain subcritical during operation. The Monte Carlo codes LAHET and MCNP-DSP were combined together to facilitate the design of reactivity monitoring systems. The coupling of LAHET and MCNP-DSP provides a tool that can be used to simulate a variety of subcritical measurements such as the pulsed neutron, Rossi-{alpha}, or noise analysis measurements. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Valentine, T.E.; Rugama, Y. Munoz-Cobos, J.; Perez, R.
2000-10-23
The design of reactivity monitoring systems for accelerator-driven systems must be investigated to ensure that such systems remain subcritical during operation. The Monte Carlo codes LAHET and MCNP-DSP were combined together to facilitate the design of reactivity monitoring systems. The coupling of LAHET and MCNP-DSP provides a tool that can be used to simulate a variety of subcritical measurements such as the pulsed neutron, Rossi-{alpha}, or noise analysis measurements.
Energy Technology Data Exchange (ETDEWEB)
Vergnaud, T.; Nimal, J.C. (CEA Centre d' Etudes Nucleaires de Saclay, 91 - Gif-sur-Yvette (France))
1990-01-01
The three-dimensional polycinetic Monte Carlo particle transport code TRIPOLI has been under development in the French Shielding Laboratory at Saclay since 1965. TRIPOLI-1 began to run in 1970 and became TRIPOLI-2 in 1978: since then its capabilities have been improved and many studies have been performed. TRIPOLI can treat stationary or time dependent problems in shielding and in neutronics. Some examples of solved problems are presented to demonstrate the many possibilities of the system. (author).
EleCa: A Monte Carlo code for the propagation of extragalactic photons at ultra-high energy
Energy Technology Data Exchange (ETDEWEB)
Settimo, Mariangela [University of Siegen (Germany); De Domenico, Manlio [Laboratory of Complex Systems, Scuola Superiore di Catania and INFN (Italy); Lyberis, Haris [Federal University of Rio de Janeiro (Brazil)
2013-06-15
Ultra high energy photons, above 10{sup 17}–10{sup 18}eV, can interact with the extragalactic background radiation leading to the development of electromagnetic cascades. A Monte Carlo code to simulate the electromagnetic cascades initiated by high-energy photons and electrons is presented. Results from simulations and their impact on the predicted flux at Earth are discussed in different astrophysical scenarios.
Update on the Status of the FLUKA Monte Carlo Transport Code
Pinsky, L.; Anderson, V.; Empl, A.; Lee, K.; Smirnov, G.; Zapp, N; Ferrari, A.; Tsoulou, K.; Roesler, S.; Vlachoudis, V.; Battisoni, G.; Ceruti, F.; Gadioli, M. V.; Garzelli, M.; Muraro, S.; Rancati, T.; Sala, P.; Ballarini, R.; Ottolenghi, A.; Parini, V.; Scannicchio, D.; Pelliccioni, M.; Wilson, T. L.
2004-01-01
The FLUKA Monte Carlo transport code is a well-known simulation tool in High Energy Physics. FLUKA is a dynamic tool in the sense that it is being continually updated and improved by the authors. Here we review the progresses achieved in the last year on the physics models. From the point of view of hadronic physics, most of the effort is still in the field of nucleus--nucleus interactions. The currently available version of FLUKA already includes the internal capability to simulate inelastic nuclear interactions beginning with lab kinetic energies of 100 MeV/A up the the highest accessible energies by means of the DPMJET-II.5 event generator to handle the interactions for greater than 5 GeV/A and rQMD for energies below that. The new developments concern, at high energy, the embedding of the DPMJET-III generator, which represent a major change with respect to the DPMJET-II structure. This will also allow to achieve a better consistency between the nucleus-nucleus section with the original FLUKA model for hadron-nucleus collisions. Work is also in progress to implement a third event generator model based on the Master Boltzmann Equation approach, in order to extend the energy capability from 100 MeV/A down to the threshold for these reactions. In addition to these extended physics capabilities, structural changes to the programs input and scoring capabilities are continually being upgraded. In particular we want to mention the upgrades in the geometry packages, now capable of reaching higher levels of abstraction. Work is also proceeding to provide direct import into ROOT of the FLUKA output files for analysis and to deploy a user-friendly GUI input interface.
Antiproton annihilation physics in the Monte Carlo particle transport code SHIELD-HIT12A
Energy Technology Data Exchange (ETDEWEB)
Taasti, Vicki Trier; Knudsen, Helge [Dept. of Physics and Astronomy, Aarhus University (Denmark); Holzscheiter, Michael H. [Dept. of Physics and Astronomy, Aarhus University (Denmark); Dept. of Physics and Astronomy, University of New Mexico (United States); Sobolevsky, Nikolai [Institute for Nuclear Research of the Russian Academy of Sciences (INR), Moscow (Russian Federation); Moscow Institute of Physics and Technology (MIPT), Dolgoprudny (Russian Federation); Thomsen, Bjarne [Dept. of Physics and Astronomy, Aarhus University (Denmark); Bassler, Niels, E-mail: bassler@phys.au.dk [Dept. of Physics and Astronomy, Aarhus University (Denmark)
2015-03-15
The Monte Carlo particle transport code SHIELD-HIT12A is designed to simulate therapeutic beams for cancer radiotherapy with fast ions. SHIELD-HIT12A allows creation of antiproton beam kernels for the treatment planning system TRiP98, but first it must be benchmarked against experimental data. An experimental depth dose curve obtained by the AD-4/ACE collaboration was compared with an earlier version of SHIELD-HIT, but since then inelastic annihilation cross sections for antiprotons have been updated and a more detailed geometric model of the AD-4/ACE experiment was applied. Furthermore, the Fermi–Teller Z-law, which is implemented by default in SHIELD-HIT12A has been shown not to be a good approximation for the capture probability of negative projectiles by nuclei. We investigate other theories which have been developed, and give a better agreement with experimental findings. The consequence of these updates is tested by comparing simulated data with the antiproton depth dose curve in water. It is found that the implementation of these new capture probabilities results in an overestimation of the depth dose curve in the Bragg peak. This can be mitigated by scaling the antiproton collision cross sections, which restores the agreement, but some small deviations still remain. Best agreement is achieved by using the most recent antiproton collision cross sections and the Fermi–Teller Z-law, even if experimental data conclude that the Z-law is inadequately describing annihilation on compounds. We conclude that more experimental cross section data are needed in the lower energy range in order to resolve this contradiction, ideally combined with more rigorous models for annihilation on compounds.
Accurate simulation of ionization chamber response with the Monte Carlo code PENELOPE
Energy Technology Data Exchange (ETDEWEB)
Sempau, Josep [Technical University of Catalonia (Spain)
2010-07-01
Full text. Ionization chambers (IC) are routinely used in hospitals for the dosimetry of the photon and electron beams used for radiotherapy treatments. The determination of absorbed dose to water from the absorbed dose to the air filling the cavity requires the introduction of stopping power ratios and perturbation factors, which account for the disturbance caused by the presence of the chamber. Although this may seem a problem readily amenable to Monte Carlo simulation, the fact is that the accurate determination of IC response has been, during the last 20 years, one of the most important challenges of the simulation of electromagnetic showers. The main difficulty stems from the use of condensed history techniques for electron and positron transport. This approach, which involves grouping a large number of interactions into a single artificial event, is known to produce the so-called interface effects when particles travel across surfaces separating different media. These effects are extremely important when the electron step length is not negligible compared to the size of the region being crossed, as it is the case with the cavity of an IC. The artifact, which becomes apparent when the chamber response shows a marked dependence on the adopted step size, can be palliated with the use of sophisticated electron transport algorithms. These topics will be discussed in the context of the transport model implemented in the Penelope code. The degree of violation of the Fano theorem for a simple, planar geometry, will be used as a measure of the stability of the algorithm with respect to variations of the electron step length, thus assessing the 'quality' of its condensed history scheme. It will be shown that, with a suitable choice of transport parameters, Penelope can simulate IC response with an accuracy of the order of 0.1%. (author)
Update On the Status of the FLUKA Monte Carlo Transport Code*
Ferrari, A.; Lorenzo-Sentis, M.; Roesler, S.; Smirnov, G.; Sommerer, F.; Theis, C.; Vlachoudis, V.; Carboni, M.; Mostacci, A.; Pelliccioni, M.
2006-01-01
The FLUKA Monte Carlo transport code is a well-known simulation tool in High Energy Physics. FLUKA is a dynamic tool in the sense that it is being continually updated and improved by the authors. We review the progress achieved since the last CHEP Conference on the physics models, some technical improvements to the code and some recent applications. From the point of view of the physics, improvements have been made with the extension of PEANUT to higher energies for p, n, pi, pbar/nbar and for nbars down to the lowest energies, the addition of the online capability to evolve radioactive products and get subsequent dose rates, upgrading of the treatment of EM interactions with the elimination of the need to separately prepare preprocessed files. A new coherent photon scattering model, an updated treatment of the photo-electric effect, an improved pair production model, new photon cross sections from the LLNL Cullen database have been implemented. In the field of nucleus-- nucleus interactions the electromagnetic dissociation of heavy ions has been added along with the extension of the interaction models for some nuclide pairs to energies below 100 MeV/A using the BME approach, as well as the development of an improved QMD model for intermediate energies. Both DPMJET 2.53 and 3 remain available along with rQMD 2.4 for heavy ion interactions above 100 MeV/A. Technical improvements include the ability to use parentheses in setting up the combinatorial geometry, the introduction of pre-processor directives in the input stream. a new random number generator with full 64 bit randomness, new routines for mathematical special functions (adapted from SLATEC). Finally, work is progressing on the deployment of a user-friendly GUI input interface as well as a CAD-like geometry creation and visualization tool. On the application front, FLUKA has been used to extensively evaluate the potential space radiation effects on astronauts for future deep space missions, the activation
Monte Carlo simulation of MOSFET detectors for high-energy photon beams using the PENELOPE code
Energy Technology Data Exchange (ETDEWEB)
Panettieri, Vanessa [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Diagonal 647, 08028 Barcelona (Spain); Duch, Maria Amor [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Diagonal 647, 08028 Barcelona (Spain); Jornet, Nuria [Servei de RadiofIsica i Radioproteccio, Hospital de la Santa Creu i San Pau Sant Antoni Maria Claret 167, 08025 Barcelona (Spain); Ginjaume, Merce [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Diagonal 647, 08028 Barcelona (Spain); Carrasco, Pablo [Servei de RadiofIsica i Radioproteccio, Hospital de la Santa Creu i San Pau Sant Antoni Maria Claret 167, 08025 Barcelona (Spain); Badal, Andreu [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Diagonal 647, 08028 Barcelona (Spain); Ortega, Xavier [Institut de Tecniques Energetiques, Universitat Politecnica de Catalunya, Diagonal 647, 08028 Barcelona (Spain); Ribas, Montserrat [Servei de RadiofIsica i Radioproteccio, Hospital de la Santa Creu i San Pau Sant Antoni Maria Claret 167, 08025 Barcelona (Spain)
2007-01-07
The aim of this work was the Monte Carlo (MC) simulation of the response of commercially available dosimeters based on metal oxide semiconductor field effect transistors (MOSFETs) for radiotherapeutic photon beams using the PENELOPE code. The studied Thomson and Nielsen TN-502-RD MOSFETs have a very small sensitive area of 0.04 mm{sup 2} and a thickness of 0.5 {mu}m which is placed on a flat kapton base and covered by a rounded layer of black epoxy resin. The influence of different metallic and Plastic water(TM) build-up caps, together with the orientation of the detector have been investigated for the specific application of MOSFET detectors for entrance in vivo dosimetry. Additionally, the energy dependence of MOSFET detectors for different high-energy photon beams (with energy >1.25 MeV) has been calculated. Calculations were carried out for simulated 6 MV and 18 MV x-ray beams generated by a Varian Clinac 1800 linear accelerator, a Co-60 photon beam from a Theratron 780 unit, and monoenergetic photon beams ranging from 2 MeV to 10 MeV. The results of the validation of the simulated photon beams show that the average difference between MC results and reference data is negligible, within 0.3%. MC simulated results of the effect of the build-up caps on the MOSFET response are in good agreement with experimental measurements, within the uncertainties. In particular, for the 18 MV photon beam the response of the detectors under a tungsten cap is 48% higher than for a 2 cm Plastic water(TM) cap and approximately 26% higher when a brass cap is used. This effect is demonstrated to be caused by positron production in the build-up caps of higher atomic number. This work also shows that the MOSFET detectors produce a higher signal when their rounded side is facing the beam (up to 6%) and that there is a significant variation (up to 50%) in the response of the MOSFET for photon energies in the studied energy range. All the results have shown that the PENELOPE code system
Radiation Protection Studies for Medical Particle Accelerators using Fluka Monte Carlo Code.
Infantino, Angelo; Cicoria, Gianfranco; Lucconi, Giulia; Pancaldi, Davide; Vichi, Sara; Zagni, Federico; Mostacci, Domiziano; Marengo, Mario
2017-04-01
Radiation protection (RP) in the use of medical cyclotrons involves many aspects both in the routine use and for the decommissioning of a site. Guidelines for site planning and installation, as well as for RP assessment, are given in international documents; however, the latter typically offer analytic methods of calculation of shielding and materials activation, in approximate or idealised geometry set-ups. The availability of Monte Carlo (MC) codes with accurate up-to-date libraries for transport and interaction of neutrons and charged particles at energies below 250 MeV, together with the continuously increasing power of modern computers, makes the systematic use of simulations with realistic geometries possible, yielding equipment and site-specific evaluation of the source terms, shielding requirements and all quantities relevant to RP at the same time. In this work, the well-known FLUKA MC code was used to simulate different aspects of RP in the use of biomedical accelerators, particularly for the production of medical radioisotopes. In the context of the Young Professionals Award, held at the IRPA 14 conference, only a part of the complete work is presented. In particular, the simulation of the GE PETtrace cyclotron (16.5 MeV) installed at S. Orsola-Malpighi University Hospital evaluated the effective dose distribution around the equipment; the effective number of neutrons produced per incident proton and their spectral distribution; the activation of the structure of the cyclotron and the vault walls; the activation of the ambient air, in particular the production of 41Ar. The simulations were validated, in terms of physical and transport parameters to be used at the energy range of interest, through an extensive measurement campaign of the neutron environmental dose equivalent using a rem-counter and TLD dosemeters. The validated model was then used in the design and the licensing request of a new Positron Emission Tomography facility. © The Author 2016
Chiavassa, S; Aubineau-Lanièce, I; Bitar, A; Lisbona, A; Barbet, J; Franck, D; Jourdain, J R; Bardiès, M
2006-02-07
Dosimetric studies are necessary for all patients treated with targeted radiotherapy. In order to attain the precision required, we have developed Oedipe, a dosimetric tool based on the MCNPX Monte Carlo code. The anatomy of each patient is considered in the form of a voxel-based geometry created using computed tomography (CT) images or magnetic resonance imaging (MRI). Oedipe enables dosimetry studies to be carried out at the voxel scale. Validation of the results obtained by comparison with existing methods is complex because there are multiple sources of variation: calculation methods (different Monte Carlo codes, point kernel), patient representations (model or specific) and geometry definitions (mathematical or voxel-based). In this paper, we validate Oedipe by taking each of these parameters into account independently. Monte Carlo methodology requires long calculation times, particularly in the case of voxel-based geometries, and this is one of the limits of personalized dosimetric methods. However, our results show that the use of voxel-based geometry as opposed to a mathematically defined geometry decreases the calculation time two-fold, due to an optimization of the MCNPX2.5e code. It is therefore possible to envisage the use of Oedipe for personalized dosimetry in the clinical context of targeted radiotherapy.
Energy Technology Data Exchange (ETDEWEB)
Serikov, A., E-mail: arkady.serikov@kit.edu [Karlsruhe Institute of Technology KIT, Institute for Neutron Physics and Reactor Technology, Hermann-von-Helmholtz-Platz 1, 76344 Eggenstein-Leopoldshafen (Germany); Fischer, U.; Leichtle, D. [Karlsruhe Institute of Technology KIT, Institute for Neutron Physics and Reactor Technology, Hermann-von-Helmholtz-Platz 1, 76344 Eggenstein-Leopoldshafen (Germany); Pitcher, C.S. [ITER Organization, Route de Vinon sur Verdon, 13115, St. Paul lez Durance (France)
2012-08-15
Highlights: Black-Right-Pointing-Pointer Systematic neutronics analyses were conducted to assess the ITER Equatorial Port Plug radiation shielding performance. Black-Right-Pointing-Pointer Shielding optimization was achieved by parametric analyses of several design variants using the MCNP5, FISPACT-2007, and R2Smesh codes. Black-Right-Pointing-Pointer Dominant effect of radiation streaming along the port plug gaps was recognized. Black-Right-Pointing-Pointer Combination of the gap labyrinths and streaming stoppers or rails reduces shutdown doses by 2 orders of magnitude. Black-Right-Pointing-Pointer Using the proposed shielding, the shutdown dose in the ITER port interspace is less than the personnel access limit of 100 {mu}Sv/h. - Abstract: This paper addresses neutronics aspects of the design development of the Diagnostic Generic Equatorial Port Plug (EPP) in ITER. To secure the personnel access at the EPP back-end interspace, parametric neutronics analyses of the EPP radiation environment have been performed and practical shielding solutions have been found. Radiation transport was performed with the Monte Carlo MCNP5 code. Activation calculations were conducted with the FISPACT-2007 inventory code. The R2Smesh approach was applied to couple transport and activation calculations. Newly created EPP local MCNP5 model was devised by extracting the EPP and adjacent blanket modules from the ITER Alite-4.1 model with proper modification of the EPP geometry in accordance with recent 3D CAD CATIA model. The EPP local model reproduces the EPP neutronically important features and allows investigation of the EPP neutronics effects in isolation from all other ITER components. Thorough EPP parametric analyses revealed dominant effect of gaps around EPP and several EPP design improvements were implemented as the outcomes of the analyses. Gap labyrinths and streaming stoppers inserted into the gaps were shown are capable to reduce the shutdown dose rate which is below the 100
Energy Technology Data Exchange (ETDEWEB)
Serikov, A.; Fischer, U.; Grosse, D.; Leichtle, D.; Majerle, M., E-mail: arkady.serikov@kit.edu [Karlsruhe Institute of Technology (KIT), Eggenstein-Leopoldshafen (Germany)
2011-07-01
The Monte Carlo (MC) method is the most suitable computational technique of radiation transport for shielding applications in fusion neutronics. This paper is intended for sharing the results of long term experience of the fusion neutronics group at Karlsruhe Institute of Technology (KIT) in radiation shielding calculations with the MCNP5 code for the ITER fusion reactor with emphasizing on the use of several ITER project-driven computer programs developed at KIT. Two of them, McCad and R2S, seem to be the most useful in radiation shielding analyses. The McCad computer graphical tool allows to perform automatic conversion of the MCNP models from the underlying CAD (CATIA) data files, while the R2S activation interface couples the MCNP radiation transport with the FISPACT activation allowing to estimate nuclear responses such as dose rate and nuclear heating after the ITER reactor shutdown. The cell-based R2S scheme was applied in shutdown photon dose analysis for the designing of the In-Vessel Viewing System (IVVS) and the Glow Discharge Cleaning (GDC) unit in ITER. Newly developed at KIT mesh-based R2S feature was successfully tested on the shutdown dose rate calculations for the upper port in the Neutral Beam (NB) cell of ITER. The merits of McCad graphical program were broadly acknowledged by the neutronic analysts and its continuous improvement at KIT has introduced its stable and more convenient run with its Graphical User Interface. Detailed 3D ITER neutronic modeling with the MCNP Monte Carlo method requires a lot of computation resources, inevitably leading to parallel calculations on clusters. Performance assessments of the MCNP5 parallel runs on the JUROPA/HPC-FF supercomputer cluster permitted to find the optimal number of processors for ITER-type runs. (author)
Energy Technology Data Exchange (ETDEWEB)
Franke, Brian Claude; Kensek, Ronald Patrick; Laub, Thomas William
2004-06-01
ITS is a powerful and user-friendly software package permitting state of the art Monte Carlo solution of linear time-independent couple electron/photon radiation transport problems, with or without the presence of macroscopic electric and magnetic fields of arbitrary spatial dependence. Our goal has been to simultaneously maximize operational simplicity and physical accuracy. Through a set of preprocessor directives, the user selects one of the many ITS codes. The ease with which the makefile system is applied combines with an input scheme based on order-independent descriptive keywords that makes maximum use of defaults and internal error checking to provide experimentalists and theorists alike with a method for the routine but rigorous solution of sophisticated radiation transport problems. Physical rigor is provided by employing accurate cross sections, sampling distributions, and physical models for describing the production and transport of the electron/photon cascade from 1.0 GeV down to 1.0 keV. The availability of source code permits the more sophisticated user to tailor the codes to specific applications and to extend the capabilities of the codes to more complex applications. Version 5.0, the latest version of ITS, contains (1) improvements to the ITS 3.0 continuous-energy codes, (2)multigroup codes with adjoint transport capabilities, and (3) parallel implementations of all ITS codes. Moreover the general user friendliness of the software has been enhanced through increased internal error checking and improved code portability.
COOL: A code for Dynamic Monte Carlo Simulation of molecular dynamics
Barletta, Paolo
2012-02-01
Cool is a program to simulate evaporative and sympathetic cooling for a mixture of two gases co-trapped in an harmonic potential. The collisions involved are assumed to be exclusively elastic, and losses are due to evaporation from the trap. Each particle is followed individually in its trajectory, consequently properties such as spatial densities or energy distributions can be readily evaluated. The code can be used sequentially, by employing one output as input for another run. The code can be easily generalised to describe more complicated processes, such as the inclusion of inelastic collisions, or the possible presence of more than two species in the trap. New version program summaryProgram title: COOL Catalogue identifier: AEHJ_v2_0 Program summary URL:http://cpc.cs.qub.ac.uk/summaries/AEHJ_v2_0.html Program obtainable from: CPC Program Library, Queen's University, Belfast, N. Ireland Licensing provisions: Standard CPC licence, http://cpc.cs.qub.ac.uk/licence/licence.html No. of lines in distributed program, including test data, etc.: 1 097 733 No. of bytes in distributed program, including test data, etc.: 18 425 722 Distribution format: tar.gz Programming language: C++ Computer: Desktop Operating system: Linux RAM: 500 Mbytes Classification: 16.7, 23 Catalogue identifier of previous version: AEHJ_v1_0 Journal reference of previous version: Comput. Phys. Comm. 182 (2011) 388 Does the new version supersede the previous version?: Yes Nature of problem: Simulation of the sympathetic process occurring for two molecular gases co-trapped in a deep optical trap. Solution method: The Direct Simulation Monte Carlo method exploits the decoupling, over a short time period, of the inter-particle interaction from the trapping potential. The particle dynamics is thus exclusively driven by the external optical field. The rare inter-particle collisions are considered with an acceptance/rejection mechanism, that is, by comparing a random number to the collisional probability
Thyroid cell irradiation by radioiodines: a new Monte Carlo electron track-structure code
Directory of Open Access Journals (Sweden)
Christophe Champion
2007-09-01
Full Text Available The most significant impact of the Chernobyl accident is the increased incidence of thyroid cancer among children who were exposed to short-lived radioiodines and 131-iodine. In order to accurately estimate the radiation dose provided by these radioiodines, it is necessary to know where iodine is incorporated. To do that, the distribution at the cellular level of newly organified iodine in the immature rat thyroid was performed using secondary ion mass microscopy (NanoSIMS50. Actual dosimetric models take only into account the averaged energy and range of beta particles of the radio-elements and may, therefore, imperfectly describe the real distribution of dose deposit at the microscopic level around the point sources. Our approach is radically different since based on a track-structure Monte Carlo code allowing following-up of electrons down to low energies (~ 10eV what permits a nanometric description of the irradiation physics. The numerical simulations were then performed by modelling the complete disintegrations of the short-lived iodine isotopes as well as of 131I in new born rat thyroids in order to take into account accurate histological and biological data for the thyroid gland.O impacto mais significante do acidente de Chernobyl é o crescimento da incidência de câncer de tireóide em crianças que foram expostas a radioiodos de vida curta e ao Iodo-131. Na estimativa precisa da dose de radiação fornecida por esses radioiodos, é necessário conhecer onde o iodo está incorporado. Para obtermos esse resultado, a distribuição em nível celular de iodo recentemente organificado na tireóde de ratos imaturos foi realizada usando microscopia de massa iônica secundária (NanoSIMS50. Modelos dosimétricos atuais consideram apenas a energia média das partículas beta dos radioelementos e pode, imperfeitamente descrever a distribuição real de dose ao nível microscópico em torno dos pontos pesquisados. Nossa abordagem
De Geyter, Gert; Fritz, Jacopo; Camps, Peter
2012-01-01
We present FitSKIRT, a method to efficiently fit radiative transfer models to UV/optical images of dusty galaxies. These images have the advantage that they have better spatial resolution compared to FIR/submm data. FitSKIRT uses the GAlib genetic algorithm library to optimize the output of the SKIRT Monte Carlo radiative transfer code. Genetic algorithms prove to be a valuable tool in handling the multi- dimensional search space as well as the noise induced by the random nature of the Monte Carlo radiative transfer code. FitSKIRT is tested on artificial images of a simulated edge-on spiral galaxy, where we gradually increase the number of fitted parameters. We find that we can recover all model parameters, even if all 11 model parameters are left unconstrained. Finally, we apply the FitSKIRT code to a V-band image of the edge-on spiral galaxy NGC4013. This galaxy has been modeled previously by other authors using different combinations of radiative transfer codes and optimization methods. Given the different...
Development of an MCNP-tally based burnup code and validation through PWR benchmark exercises
Energy Technology Data Exchange (ETDEWEB)
El Bakkari, B. [ERSN-LMR, Department of physics, Faculty of Sciences P.O.Box 2121, Tetuan (Morocco)], E-mail: bakkari@gmail.com; El Bardouni, T.; Merroun, O.; El Younoussi, Ch.; Boulaich, Y. [ERSN-LMR, Department of physics, Faculty of Sciences P.O.Box 2121, Tetuan (Morocco); Chakir, E. [EPTN-LPMR, Faculty of Sciences Kenitra (Morocco)
2009-05-15
The aim of this study is to evaluate the capabilities of a newly developed burnup code called BUCAL1. The code provides the full capabilities of the Monte Carlo code MCNP5, through the use of the MCNP tally information. BUCAL1 uses the fourth order Runge Kutta method with the predictor-corrector approach as the integration method to determine the fuel composition at a desired burnup step. Validation of BUCAL1 was done by code vs. code comparison. Results of two different kinds of codes are employed. The first one is CASMO-4, a deterministic multi-group two-dimensional transport code. The second kind is MCODE and MOCUP, a link MCNP-ORIGEN codes. These codes use different burnup algorithms to solve the depletion equations system. Eigenvalue and isotope concentrations were compared for two PWR uranium and thorium benchmark exercises at cold (300 K) and hot (900 K) conditions, respectively. The eigenvalue comparison between BUCAL1 and the aforementioned two kinds of codes shows a good prediction of the systems'k-inf values during the entire burnup history, and the maximum difference is within 2%. The differences between the BUCAL1 isotope concentrations and the predictions of CASMO-4, MCODE and MOCUP are generally better, and only for a few sets of isotopes these differences exceed 10%.
Energy Technology Data Exchange (ETDEWEB)
Procassini, R.J. [Lawrence Livermore National lab., CA (United States)
1997-12-31
The fine-scale, multi-space resolution that is envisioned for accurate simulations of complex weapons systems in three spatial dimensions implies flop-rate and memory-storage requirements that will only be obtained in the near future through the use of parallel computational techniques. Since the Monte Carlo transport models in these simulations usually stress both of these computational resources, they are prime candidates for parallelization. The MONACO Monte Carlo transport package, which is currently under development at LLNL, will utilize two types of parallelism within the context of a multi-physics design code: decomposition of the spatial domain across processors (spatial parallelism) and distribution of particles in a given spatial subdomain across additional processors (particle parallelism). This implementation of the package will utilize explicit data communication between domains (message passing). Such a parallel implementation of a Monte Carlo transport model will result in non-deterministic communication patterns. The communication of particles between subdomains during a Monte Carlo time step may require a significant level of effort to achieve a high parallel efficiency.
Mairani, A; Valente, M; Battistoni, G; Botta, F; Pedroli, G; Ferrari, A; Cremonesi, M; Di Dia, A; Ferrari, M; Fasso, A
2011-01-01
Purpose: The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, FLUKA Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, FLUKA has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. Methods: FLUKA DPKS have been calculated in both water and compact bone for monoenergetic electrons (10-3 MeV) and for beta emitting isotopes commonly used for therapy ((89)Sr, (90)Y, (131)I, (153)Sm, (177)Lu, (186)Re, and (188)Re). Point isotropic...
Energy Technology Data Exchange (ETDEWEB)
Sanchez, R.A.; Fernandez V, J.M.; Salvat, F. [Servicio de Oncologia Radioterapica. Hospital Clinico de Barcelona. Villarroel 170 08036 Barcelona (Spain)
1998-12-31
In the present communication it is presented the results of the simulation utilizing the Penelope code (Penetration and Energy loss of Positrons and Electrons) in several applications of radiotherapy which can be the radioactive sources simulation: {sup 192} Ir, {sup 125} I, {sup 106} Ru or the electron beams simulation of a linear accelerator Siemens KDS. The simulations presented in this communication have been on computers of type Pentium PC of 100 throughout 300 MHz, and the times of execution were from some hours until several days depending of the complexity of the problem. It is concluded that Penelope is a very useful tool for the Monte Carlo calculations due to its great ability and its relative handling facilities. (Author)
Energy Technology Data Exchange (ETDEWEB)
Azorin V, C.G.; Rivera M, T. [CICATA-IPN, Legaria, Mexico D.F. (Mexico)]. e-mail: claudiaazorin@yahoo.com.mx
2007-07-01
Full text: At the present time many computer programs that simulate the radiation interaction with the matter using the Monte Carlo method. Presently work is carried out the comparative analysis of four of these codes (MCNPX, EGS4, GEANT, PENELOPE) for their later one use in the development of a simple algorithm that simulates the energy deposit when passing through the matter in patients subjected to radiotherapy. The results of the analysis show that the studied simulators model the interaction of almost all type of particles with the matter, although they have their variations among those the energy intervals that manage, the programming language in which are programmed, as well as the platform under which they are executed can be mentioned. (Author)
NEPHTIS: 2D/3D validation elements using MCNP4c and TRIPOLI4 Monte-Carlo codes
Energy Technology Data Exchange (ETDEWEB)
Courau, T.; Girardi, E. [EDF R and D/SINETICS, 1av du General de Gaulle, F92141 Clamart CEDEX (France); Damian, F.; Moiron-Groizard, M. [DEN/DM2S/SERMA/LCA, CEA Saclay, F91191 Gif-sur-Yvette CEDEX (France)
2006-07-01
High Temperature Reactors (HTRs) appear as a promising concept for the next generation of nuclear power applications. The CEA, in collaboration with AREVA-NP and EDF, is developing a core modeling tool dedicated to the prismatic block-type reactor. NEPHTIS (Neutronics Process for HTR Innovating System) is a deterministic codes system based on a standard two-steps Transport-Diffusion approach (APOLLO2/CRONOS2). Validation of such deterministic schemes usually relies on Monte-Carlo (MC) codes used as a reference. However, when dealing with large HTR cores the fission source stabilization is rather poor with MC codes. In spite of this, it is shown in this paper that MC simulations may be used as a reference for a wide range of configurations. The first part of the paper is devoted to 2D and 3D MC calculations of a HTR core with control devices. Comparisons between MCNP4c and TRIPOLI4 MC codes are performed and show very consistent results. Finally, the last part of the paper is devoted to the code to code validation of the NEPHTIS deterministic scheme. (authors)
A new Monte Carlo code for simulation of the effect of irregular surfaces on X-ray spectra
Energy Technology Data Exchange (ETDEWEB)
Brunetti, Antonio, E-mail: brunetti@uniss.it; Golosio, Bruno
2014-04-01
Generally, quantitative X-ray fluorescence (XRF) analysis estimates the content of chemical elements in a sample based on the areas of the fluorescence peaks in the energy spectrum. Besides the concentration of the elements, the peak areas depend also on the geometrical conditions. In fact, the estimate of the peak areas is simple if the sample surface is smooth and if the spectrum shows a good statistic (large-area peaks). For this reason often the sample is prepared as a pellet. However, this approach is not always feasible, for instance when cultural heritage or valuable samples must be analyzed. In this case, the sample surface cannot be smoothed. In order to address this problem, several works have been reported in the literature, based on experimental measurements on a few sets of specific samples or on Monte Carlo simulations. The results obtained with the first approach are limited by the specific class of samples analyzed, while the second approach cannot be applied to arbitrarily irregular surfaces. The present work describes a more general analysis tool based on a new fast Monte Carlo algorithm, which is virtually able to simulate any kind of surface. At the best of our knowledge, it is the first Monte Carlo code with this option. A study of the influence of surface irregularities on the measured spectrum is performed and some results reported. - Highlights: • We present a fast Monte Carlo code with the possibility to simulate any irregularly rough surfaces. • We show applications to multilayer measurements. • Real time simulations are available.
Chetty, Indrin J; Moran, Jean M; McShan, Daniel L; Fraass, Benedick A; Wilderman, Scott J; Bielajew, Alex F
2002-06-01
A comprehensive set of measurements and calculations has been conducted to investigate the accuracy of the Dose Planning Method (DPM) Monte Carlo code for dose calculations from 10 and 50 MeV scanned electron beams produced from a racetrack microtron. Central axis depth dose measurements and a series of profile scans at various depths were acquired in a water phantom using a Scanditronix type RK ion chamber. Source spatial distributions for the Monte Carlo calculations were reconstructed from in-air ion chamber measurements carried out across the two-dimensional beam profile at 100 cm downstream from the source. The in-air spatial distributions were found to have full width at half maximum of 4.7 and 1.3 cm, at 100 cm from the source, for the 10 and 50 MeV beams, respectively. Energy spectra for the 10 and 50 MeV beams were determined by simulating the components of the microtron treatment head using the code MCNP4B. DPM calculations are on average within +/- 2% agreement with measurement for all depth dose and profile comparisons conducted in this study. The accuracy of the DPM code illustrated in this work suggests that DPM may be used as a valuable tool for electron beam dose calculations.
Energy Technology Data Exchange (ETDEWEB)
Kurosu, K [Department of Radiation Oncology, Osaka University Graduate School of Medicine, Osaka (Japan); Department of Medical Physics ' Engineering, Osaka University Graduate School of Medicine, Osaka (Japan); Takashina, M; Koizumi, M [Department of Medical Physics ' Engineering, Osaka University Graduate School of Medicine, Osaka (Japan); Das, I; Moskvin, V [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN (United States)
2014-06-01
Purpose: Monte Carlo codes are becoming important tools for proton beam dosimetry. However, the relationships between the customizing parameters and percentage depth dose (PDD) of GATE and PHITS codes have not been reported which are studied for PDD and proton range compared to the FLUKA code and the experimental data. Methods: The beam delivery system of the Indiana University Health Proton Therapy Center was modeled for the uniform scanning beam in FLUKA and transferred identically into GATE and PHITS. This computational model was built from the blue print and validated with the commissioning data. Three parameters evaluated are the maximum step size, cut off energy and physical and transport model. The dependence of the PDDs on the customizing parameters was compared with the published results of previous studies. Results: The optimal parameters for the simulation of the whole beam delivery system were defined by referring to the calculation results obtained with each parameter. Although the PDDs from FLUKA and the experimental data show a good agreement, those of GATE and PHITS obtained with our optimal parameters show a minor discrepancy. The measured proton range R90 was 269.37 mm, compared to the calculated range of 269.63 mm, 268.96 mm, and 270.85 mm with FLUKA, GATE and PHITS, respectively. Conclusion: We evaluated the dependence of the results for PDDs obtained with GATE and PHITS Monte Carlo generalpurpose codes on the customizing parameters by using the whole computational model of the treatment nozzle. The optimal parameters for the simulation were then defined by referring to the calculation results. The physical model, particle transport mechanics and the different geometrybased descriptions need accurate customization in three simulation codes to agree with experimental data for artifact-free Monte Carlo simulation. This study was supported by Grants-in Aid for Cancer Research (H22-3rd Term Cancer Control-General-043) from the Ministry of Health
Energy Technology Data Exchange (ETDEWEB)
Sanchez G, R.; Cavalieri, T. A.; De Paiva, F.; Dalledone S, P. de T.; Yoriyaz, H. [Instituto de Pesquisas Energeticas e Nucleares, Centro de Engenharia Nuclear / CNEN, Av. Lineu Prestes 2242, Cidade Universitaria, 05508-000 Sao Paulo (Brazil); Rodrigues F, M. A. [Universidade Estadual Paulista, Faculdade de Medicina de Botucatu, Departamento de Dermatologia e Radioterapia, Av. Prof. Montenegro s/n, Rubiao Junior, 18601-970 Botucatu (Brazil); Vivolo, V., E-mail: chancez@hotmail.com [Instituto de Pesquisas Energeticas e Nucleares, Gerencia de Metrologia das Radiacoes / CNEN, Av. Lineu Prestes 2242, Cidade Universitaria, 05508-000 Sao Paulo (Brazil)
2014-08-15
The thermo luminescent dosimeter (TLD) is used as a radiation dosimeter and can be used as environmental and staff personnel monitoring. The TLD measures ionizing radiation exposure by a process in which the amount of radiation collected by the dosimeter is converted in visible light when the crystal is heated. The amount of emitted light is proportional to the radiation exposure, and then the response of the TLD must be the related to the real dose. In this work it was used twenty four TLD 700 in order to obtain eight values of doses from a diagnostic X-ray equipment. The TLD-700 is a LiF TLD enriched with {sup 7}Li isotope. One way to compare and study the response of TLD is by Monte Carlo method, which has been used as a computational tool to solve problems stochastically. This method can be applied to any geometry, even those where the boundary conditions are unknown, making the method particularly useful to solve problems a priori. In this work it was modeled the X-ray tube exactly as the one used to irradiate the TLD, after the simulation and the TLD irradiation the results of dose value from both were compared. (Author)
A Monte Carlo Code for Relativistic Radiation Transport Around Kerr Black Holes
Schnittman, Jeremy David; Krolik, Julian H.
2013-01-01
We present a new code for radiation transport around Kerr black holes, including arbitrary emission and absorption mechanisms, as well as electron scattering and polarization. The code is particularly useful for analyzing accretion flows made up of optically thick disks and optically thin coronae. We give a detailed description of the methods employed in the code and also present results from a number of numerical tests to assess its accuracy and convergence.
Energy Technology Data Exchange (ETDEWEB)
Ghoos, K., E-mail: kristel.ghoos@kuleuven.be [KU Leuven, Department of Mechanical Engineering, Celestijnenlaan 300A, 3001 Leuven (Belgium); Dekeyser, W. [KU Leuven, Department of Mechanical Engineering, Celestijnenlaan 300A, 3001 Leuven (Belgium); Samaey, G. [KU Leuven, Department of Computer Science, Celestijnenlaan 200A, 3001 Leuven (Belgium); Börner, P. [Institute of Energy and Climate Research (IEK-4), FZ Jülich GmbH, D-52425 Jülich (Germany); Baelmans, M. [KU Leuven, Department of Mechanical Engineering, Celestijnenlaan 300A, 3001 Leuven (Belgium)
2016-10-01
The plasma and neutral transport in the plasma edge of a nuclear fusion reactor is usually simulated using coupled finite volume (FV)/Monte Carlo (MC) codes. However, under conditions of future reactors like ITER and DEMO, convergence issues become apparent. This paper examines the convergence behaviour and the numerical error contributions with a simplified FV/MC model for three coupling techniques: Correlated Sampling, Random Noise and Robbins Monro. Also, practical procedures to estimate the errors in complex codes are proposed. Moreover, first results with more complex models show that an order of magnitude speedup can be achieved without any loss in accuracy by making use of averaging in the Random Noise coupling technique.
Energy Technology Data Exchange (ETDEWEB)
Walsh, J. A. [Department of Nuclear Science and Engineering, Massachusetts Institute of Technology, NW12-312 Albany, St. Cambridge, MA 02139 (United States); Palmer, T. S. [Department of Nuclear Engineering and Radiation Health Physics, Oregon State University, 116 Radiation Center, Corvallis, OR 97331 (United States); Urbatsch, T. J. [XTD-5: Air Force Systems, Los Alamos National Laboratory, Los Alamos, NM 87545 (United States)
2013-07-01
A new method for generating discrete scattering cross sections to be used in charged particle transport calculations is investigated. The method of data generation is presented and compared to current methods for obtaining discrete cross sections. The new, more generalized approach allows greater flexibility in choosing a cross section model from which to derive discrete values. Cross section data generated with the new method is verified through a comparison with discrete data obtained with an existing method. Additionally, a charged particle transport capability is demonstrated in the time-dependent Implicit Monte Carlo radiative transfer code package, Milagro. The implementation of this capability is verified using test problems with analytic solutions as well as a comparison of electron dose-depth profiles calculated with Milagro and an already-established electron transport code. An initial investigation of a preliminary integration of the discrete cross section generation method with the new charged particle transport capability in Milagro is also presented. (authors)
Ródenas, José; Abarca, Agustín; Gallardo, Sergio; Sollet, Eduardo
2010-07-01
Control rods are activated by neutron reactions into the reactor. The activation is produced mainly in stainless steel and its impurities. The dose produced by this activity is not important inside the reactor, but it has to be taken into account when the rod is withdrawn from it. The neutron activation has been modeled with the MCNP5 code based on the Monte Carlo method. The number of reactions obtained with the code can be converted into activity. In this work, a detailed model of the control rod has been developed considering all its components: handle, tubes, gain, and central core. On the other hand, the rod has been divided into 5 zones in order to consider the different axial exposition to neutron flux into the reactor. Results of the Monte Carlo simulation for the neutron activation constitute a gamma source in the control rod. With this source, applying again the Monte Carlo method, doses at certain distance of the rod have been calculated. Comparison of calculated doses with experimental measurements leads to the validation of the model developed.
Pölz, Stefan; Laubersheimer, Sven; Eberhardt, Jakob S; Harrendorf, Marco A; Keck, Thomas; Benzler, Andreas; Breustedt, Bastian
2013-08-21
The basic idea of Voxel2MCNP is to provide a framework supporting users in modeling radiation transport scenarios using voxel phantoms and other geometric models, generating corresponding input for the Monte Carlo code MCNPX, and evaluating simulation output. Applications at Karlsruhe Institute of Technology are primarily whole and partial body counter calibration and calculation of dose conversion coefficients. A new generic data model describing data related to radiation transport, including phantom and detector geometries and their properties, sources, tallies and materials, has been developed. It is modular and generally independent of the targeted Monte Carlo code. The data model has been implemented as an XML-based file format to facilitate data exchange, and integrated with Voxel2MCNP to provide a common interface for modeling, visualization, and evaluation of data. Also, extensions to allow compatibility with several file formats, such as ENSDF for nuclear structure properties and radioactive decay data, SimpleGeo for solid geometry modeling, ImageJ for voxel lattices, and MCNPX's MCTAL for simulation results have been added. The framework is presented and discussed in this paper and example workflows for body counter calibration and calculation of dose conversion coefficients is given to illustrate its application.
Elbast, M; Saudo, A; Franck, D; Petitot, F; Desbrée, A
2012-07-01
Microdosimetry using Monte Carlo simulation is a suitable technique to describe the stochastic nature of energy deposition by alpha particle at cellular level. Because of its short range, the energy imparted by this particle to the targets is highly non-uniform. Thus, to achieve accurate dosimetric results, the modelling of the geometry should be as realistic as possible. The objectives of the present study were to validate the use of the MCNPX and Geant4 Monte Carlo codes for microdosimetric studies using simple and three-dimensional voxelised geometry and to study their limit of validity in this last case. To that aim, the specific energy (z) deposited in the cell nucleus, the single-hit density of specific energy f(1)(z) and the mean-specific energy were calculated. Results show a good agreement when compared with the literature using simple geometry. The maximum percentage difference found is MCNPX for calculation time is 10 times higher with Geant4 than MCNPX code in the same conditions.
Zaker, Neda; Zehtabian, Mehdi; Sina, Sedigheh; Koontz, Craig; Meigooni, Ali S
2016-03-01
Monte Carlo simulations are widely used for calculation of the dosimetric parameters of brachytherapy sources. MCNP4C2, MCNP5, MCNPX, EGS4, EGSnrc, PTRAN, and GEANT4 are among the most commonly used codes in this field. Each of these codes utilizes a cross-sectional library for the purpose of simulating different elements and materials with complex chemical compositions. The accuracies of the final outcomes of these simulations are very sensitive to the accuracies of the cross-sectional libraries. Several investigators have shown that inaccuracies of some of the cross section files have led to errors in 125I and 103Pd parameters. The purpose of this study is to compare the dosimetric parameters of sample brachytherapy sources, calculated with three different versions of the MCNP code - MCNP4C, MCNP5, and MCNPX. In these simulations for each source type, the source and phantom geometries, as well as the number of the photons, were kept identical, thus eliminating the possible uncertainties. The results of these investigations indicate that for low-energy sources such as 125I and 103Pd there are discrepancies in gL(r) values. Discrepancies up to 21.7% and 28% are observed between MCNP4C and other codes at a distance of 6 cm for 103Pd and 10 cm for 125I from the source, respectively. However, for higher energy sources, the discrepancies in gL(r) values are less than 1.1% for 192Ir and less than 1.2% for 137Cs between the three codes. PACS number(s): 87.56.bg.
MC3: Multi-core Markov-chain Monte Carlo code
Cubillos, Patricio; Harrington, Joseph; Lust, Nate; Foster, AJ; Stemm, Madison; Loredo, Tom; Stevenson, Kevin; Campo, Chris; Hardin, Matt; Hardy, Ryan
2016-10-01
MC3 (Multi-core Markov-chain Monte Carlo) is a Bayesian statistics tool that can be executed from the shell prompt or interactively through the Python interpreter with single- or multiple-CPU parallel computing. It offers Markov-chain Monte Carlo (MCMC) posterior-distribution sampling for several algorithms, Levenberg-Marquardt least-squares optimization, and uniform non-informative, Jeffreys non-informative, or Gaussian-informative priors. MC3 can share the same value among multiple parameters and fix the value of parameters to constant values, and offers Gelman-Rubin convergence testing and correlated-noise estimation with time-averaging or wavelet-based likelihood estimation methods.
PENELOPE, and algorithm and computer code for Monte Carlo simulation of electron-photon showers
Energy Technology Data Exchange (ETDEWEB)
Salvat, F.; Fernandez-Varea, J.M.; Baro, J.; Sempau, J.
1996-10-01
The FORTRAN 77 subroutine package PENELOPE performs Monte Carlo simulation of electron-photon showers in arbitrary for a wide energy range, from similar{sub t}o 1 KeV to several hundred MeV. Photon transport is simulated by means of the standard, detailed simulation scheme. Electron and positron histories are generated on the basis of a mixed procedure, which combines detailed simulation of hard events with condensed simulation of soft interactions. A simple geometry package permits the generation of random electron-photon showers in material systems consisting of homogeneous bodies limited by quadric surfaces, i.e. planes, spheres cylinders, etc. This report is intended not only to serve as a manual of the simulation package, but also to provide the user with the necessary information to understand the details of the Monte Carlo algorithm.
PENELOPE, an algorithm and computer code for Monte Carlo simulation of electron-photon showers
Energy Technology Data Exchange (ETDEWEB)
Salvat, F.; Fernandez-Varea, J.M.; Baro, J.; Sempau, J.
1996-07-01
The FORTRAN 77 subroutine package PENELOPE performs Monte Carlo simulation of electron-photon showers in arbitrary for a wide energy range, from 1 keV to several hundred MeV. Photon transport is simulated by means of the standard, detailed simulation scheme. Electron and positron histories are generated on the basis of a mixed procedure, which combines detailed simulation of hard events with condensed simulation of soft interactions. A simple geometry package permits the generation of random electron-photon showers in material systems consisting of homogeneous bodies limited by quadric surfaces, i.e. planes, spheres, cylinders, etc. This report is intended not only to serve as a manual of the simulation package, but also to provide the user with the necessary information to understand the details of the Monte Carlo algorithm. (Author) 108 refs.
Monte Carlo Simulation of Siemens ONCOR Linear Accelerator with BEAMnrc and DOSXYZnrc Code.
Jabbari, Keyvan; Anvar, Hossein Saberi; Tavakoli, Mohammad Bagher; Amouheidari, Alireza
2013-07-01
The Monte Carlo method is the most accurate method for simulation of radiation therapy equipment. The linear accelerators (linac) are currently the most widely used machines in radiation therapy centers. In this work, a Monte Carlo modeling of the Siemens ONCOR linear accelerator in 6 MV and 18 MV beams was performed. The results of simulation were validated by measurements in water by ionization chamber and extended dose range (EDR2) film in solid water. The linac's X-ray particular are so sensitive to the properties of primary electron beam. Square field size of 10 cm × 10 cm produced by the jaws was compared with ionization chamber and film measurements. Head simulation was performed with BEAMnrc and dose calculation with DOSXYZnrc for film measurements and 3ddose file produced by DOSXYZnrc analyzed used homemade MATLAB program. At 6 MV, the agreement between dose calculated by Monte Carlo modeling and direct measurement was obtained to the least restrictive of 1%, even in the build-up region. At 18 MV, the agreement was obtained 1%, except for in the build-up region. In the build-up region, the difference was 1% at 6 MV and 2% at 18 MV. The mean difference between measurements and Monte Carlo simulation is very small in both of ONCOR X-ray energy. The results are highly accurate and can be used for many applications such as patient dose calculation in treatment planning and in studies that model this linac with small field size like intensity-modulated radiation therapy technique.
Energy Technology Data Exchange (ETDEWEB)
Habib, B.; Poumarede, B.; Tola, F.; Barthe, J. [CEA, LIST, Dept Technol Capteur et Signal, F-91191 Gif Sur Yvette, (France)
2010-07-01
The aim of the present study is to demonstrate the potential of accelerated dose calculations, using the fast Monte Carlo (MC) code referred to as PENFAST, rather than the conventional MC code PENELOPE, without losing accuracy in the computed dose. For this purpose, experimental measurements of dose distributions in homogeneous and inhomogeneous phantoms were compared with simulated results using both PENELOPE and PENFAST. The simulations and experiments were performed using a Saturne 43 linac operated at 12 MV (photons), and at 18 MeV (electrons). Pre-calculated phase space files (PSFs) were used as input data to both the PENELOPE and PENFAST dose simulations. Since depth-dose and dose profile comparisons between simulations and measurements in water were found to be in good agreement (within {+-} 1% to 1 mm), the PSF calculation is considered to have been validated. In addition, measured dose distributions were compared to simulated results in a set of clinically relevant, inhomogeneous phantoms, consisting of lung and bone heterogeneities in a water tank. In general, the PENFAST results agree to within a 1% to 1 mm difference with those produced by PENELOPE, and to within a 2% to 2 mm difference with measured values. Our study thus provides a pre-clinical validation of the PENFAST code. It also demonstrates that PENFAST provides accurate results for both photon and electron beams, equivalent to those obtained with PENELOPE. CPU time comparisons between both MC codes show that PENFAST is generally about 9-21 times faster than PENELOPE. (authors)
Energy Technology Data Exchange (ETDEWEB)
Palomba, M. E-mail: maurizio.palomba@ba.infn.it; D' Erasmo, G.; Pantaleo, A
2003-02-11
The CSSE code, a GEANT3-based Monte Carlo simulation program, has been developed in the framework of the EXPLODET project (Nucl. Instr. and Meth. A 422 (1999) 918) with the aim to simulate experimental set-ups employed in Thermal Neutron Analysis (TNA) for the landmines detection. Such a simulation code appears to be useful for studying the background in the {gamma}-ray spectra obtained with this technique, especially in the region where one expects to find the explosive signature (the {gamma}-ray peak at 10.83 MeV coming from neutron capture by nitrogen). The main features of the CSSE code are introduced and original innovations emphasized. Among the latter, an algorithm simulating the time correlation between primary particles, according with their time distributions is presented. Such a correlation is not usually achievable within standard GEANT-based codes and allows to reproduce some important phenomena, as the pulse pile-up inside the NaI(Tl) {gamma}-ray detector employed, producing a more realistic detector response simulation. CSSE has been successfully tested by reproducing a real nuclear sensor prototype assembled at the Physics Department of Bari University.
Palomba, M.; D'Erasmo, G.; Pantaleo, A.
2003-02-01
The CSSE code, a GEANT3-based Monte Carlo simulation program, has been developed in the framework of the EXPLODET project (Nucl. Instr. and Meth. A 422 (1999) 918) with the aim to simulate experimental set-ups employed in Thermal Neutron Analysis (TNA) for the landmines detection. Such a simulation code appears to be useful for studying the background in the γ-ray spectra obtained with this technique, especially in the region where one expects to find the explosive signature (the γ-ray peak at 10.83 MeV coming from neutron capture by nitrogen). The main features of the CSSE code are introduced and original innovations emphasized. Among the latter, an algorithm simulating the time correlation between primary particles, according with their time distributions is presented. Such a correlation is not usually achievable within standard GEANT-based codes and allows to reproduce some important phenomena, as the pulse pile-up inside the NaI(Tl) γ-ray detector employed, producing a more realistic detector response simulation. CSSE has been successfully tested by reproducing a real nuclear sensor prototype assembled at the Physics Department of Bari University.
Proton Dose Assessment to the Human Eye Using Monte Carlo N-Particle Transport Code (MCNPX)
2006-08-01
objective of this project was to develop a simple MCNPX model of the human eye to approximate dose delivered from proton therapy. The calculated dose...computer code MCNPX that approximates dose delivered during proton therapy. The calculations considered proton interactions and secondary interactions...Volume Calculation The MCNPX code has limited ability to compute the volumes of defined cells. The dosimetric volumes in the outer wall of the eye are
A Complex-Geometry Validation Experiment for Advanced Neutron Transport Codes
Energy Technology Data Exchange (ETDEWEB)
David W. Nigg; Anthony W. LaPorta; Joseph W. Nielsen; James Parry; Mark D. DeHart; Samuel E. Bays; William F. Skerjanc
2013-11-01
The Idaho National Laboratory (INL) has initiated a focused effort to upgrade legacy computational reactor physics software tools and protocols used for support of core fuel management and experiment management in the Advanced Test Reactor (ATR) and its companion critical facility (ATRC) at the INL.. This will be accomplished through the introduction of modern high-fidelity computational software and protocols, with appropriate new Verification and Validation (V&V) protocols, over the next 12-18 months. Stochastic and deterministic transport theory based reactor physics codes and nuclear data packages that support this effort include MCNP5[1], SCALE/KENO6[2], HELIOS[3], SCALE/NEWT[2], and ATTILA[4]. Furthermore, a capability for sensitivity analysis and uncertainty quantification based on the TSUNAMI[5] system has also been implemented. Finally, we are also evaluating the Serpent[6] and MC21[7] codes, as additional verification tools in the near term as well as for possible applications to full three-dimensional Monte Carlo based fuel management modeling in the longer term. On the experimental side, several new benchmark-quality code validation measurements based on neutron activation spectrometry have been conducted using the ATRC. Results for the first four experiments, focused on neutron spectrum measurements within the Northwest Large In-Pile Tube (NW LIPT) and in the core fuel elements surrounding the NW LIPT and the diametrically opposite Southeast IPT have been reported [8,9]. A fifth, very recent, experiment focused on detailed measurements of the element-to-element core power distribution is summarized here and examples of the use of the measured data for validation of corresponding MCNP5, HELIOS, NEWT, and Serpent computational models using modern least-square adjustment methods are provided.
Directory of Open Access Journals (Sweden)
Diego Ferraro
2011-01-01
Full Text Available Monte Carlo neutron transport codes are usually used to perform criticality calculations and to solve shielding problems due to their capability to model complex systems without major approximations. However, these codes demand high computational resources. The improvement in computer capabilities leads to several new applications of Monte Carlo neutron transport codes. An interesting one is to use this method to perform cell-level fuel assembly calculations in order to obtain few group constants to be used on core calculations. In the present work the VTT recently developed Serpent v.1.1.7 cell-oriented neutronic calculation code is used to perform cell calculations of a theoretical BWR lattice benchmark with burnable poisons, and the main results are compared to reported ones and with calculations performed with Condor v.2.61, the INVAP's neutronic collision probability cell code.
Patni, H K; Nadar, M Y; Akar, D K; Bhati, S; Sarkar, P K
2011-11-01
The adult reference male and female computational voxel phantoms recommended by ICRP are adapted into the Monte Carlo transport code FLUKA. The FLUKA code is then utilised for computation of dose conversion coefficients (DCCs) expressed in absorbed dose per air kerma free-in-air for colon, lungs, stomach wall, breast, gonads, urinary bladder, oesophagus, liver and thyroid due to a broad parallel beam of mono-energetic photons impinging in anterior-posterior and posterior-anterior directions in the energy range of 15 keV-10 MeV. The computed DCCs of colon, lungs, stomach wall and breast are found to be in good agreement with the results published in ICRP publication 110. The present work thus validates the use of FLUKA code in computation of organ DCCs for photons using ICRP adult voxel phantoms. Further, the DCCs for gonads, urinary bladder, oesophagus, liver and thyroid are evaluated and compared with results published in ICRP 74 in the above-mentioned energy range and geometries. Significant differences in DCCs are observed for breast, testis and thyroid above 1 MeV, and for most of the organs at energies below 60 keV in comparison with the results published in ICRP 74. The DCCs of female voxel phantom were found to be higher in comparison with male phantom for almost all organs in both the geometries.
Comparative Dosimetric Estimates of a 25 keV Electron Micro-beam with three Monte Carlo Codes
Mainardi, E; Donahue, R J
2002-01-01
The calculations presented compare the different performances of the three Monte Carlo codes PENELOPE-1999, MCNP-4C and PITS, for the evaluation of Dose profiles from a 25 keV electron micro-beam traversing individual cells. The overall model of a cell is a water cylinder equivalent for the three codes but with a different internal scoring geometry: hollow cylinders for PENELOPE and MCNP, whereas spheres are used for the PITS code. A cylindrical cell geometry with scoring volumes with the shape of hollow cylinders was initially selected for PENELOPE and MCNP because of its superior simulation of the actual shape and dimensions of a cell and for its improved computer-time efficiency if compared to spherical internal volumes. Some of the transfer points and energy transfer that constitute a radiation track may actually fall in the space between spheres, that would be outside the spherical scoring volume. This internal geometry, along with the PENELOPE algorithm, drastically reduced the computer time when using ...
Energy Technology Data Exchange (ETDEWEB)
Vergnaud, Th.; Nimal, J.C.; Chiron, M
2001-07-01
The TRIPOLI-3 code applies the Monte Carlo method to neutron, gamma-ray and coupled neutron and gamma-ray transport calculations in three-dimensional geometries, either in steady-state conditions or having a time dependence. It can be used to study problems where there is a high flux attenuation between the source zone and the result zone (studies of shielding configurations or source driven sub-critical systems, with fission being taken into account), as well as problems where there is a low flux attenuation (neutronic calculations -- in a fuel lattice cell, for example -- where fission is taken into account, usually with the calculation on the effective multiplication factor, fine structure studies, numerical experiments to investigate methods approximations, etc). TRIPOLI-3 has been operational since 1995 and is the version of the TRIPOLI code that follows on from TRIPOLI-2; it can be used on SUN, RISC600 and HP workstations and on PC using the Linux or Windows/NT operating systems. The code uses nuclear data libraries generated using the THEMIS/NJOY system. The current libraries were derived from ENDF/B6 and JEF2. There is also a response function library based on a number of evaluations, notably the dosimetry libraries IRDF/85, IRDF/90 and also evaluations from JEF2. The treatment of particle transport is the same in version 3.5 as in version 3.4 of the TRIPOLI code; but the version 3.5 is more convenient for preparing the input data and for reading the output. The french version of the user's manual exists. (authors)
Shinn, Judy L.; Wilson, John W.; Nealy, John E.; Cucinotta, Francis A.
1990-01-01
Continuing efforts toward validating the buildup factor method and the BRYNTRN code, which use the deterministic approach in solving radiation transport problems and are the candidate engineering tools in space radiation shielding analyses, are presented. A simplified theory of proton buildup factors assuming no neutron coupling is derived to verify a previously chosen form for parameterizing the dose conversion factor that includes the secondary particle buildup effect. Estimates of dose in tissue made by the two deterministic approaches and the Monte Carlo method are intercompared for cases with various thicknesses of shields and various types of proton spectra. The results are found to be in reasonable agreement but with some overestimation by the buildup factor method when the effect of neutron production in the shield is significant. Future improvement to include neutron coupling in the buildup factor theory is suggested to alleviate this shortcoming. Impressive agreement for individual components of doses, such as those from the secondaries and heavy particle recoils, are obtained between BRYNTRN and Monte Carlo results.
Kum, Oyeon; Han, Youngyih; Jeong, Hae Sun
2012-05-01
Minimizing the differences between dose distributions calculated at the treatment planning stage and those delivered to the patient is an essential requirement for successful radiotheraphy. Accurate calculation of dose distributions in the treatment planning process is important and can be done only by using a Monte Carlo calculation of particle transport. In this paper, we perform a further validation of our previously developed parallel Monte Carlo electron and photon transport (PMCEPT) code [Kum and Lee, J. Korean Phys. Soc. 47, 716 (2005) and Kim and Kum, J. Korean Phys. Soc. 49, 1640 (2006)] for applications to clinical radiation problems. A linear accelerator, Siemens' Primus 6 MV, was modeled and commissioned. A thorough validation includes both small fields, closely related to the intensity modulated radiation treatment (IMRT), and large fields. Two-dimensional comparisons with film measurements were also performed. The PMCEPT results, in general, agreed well with the measured data within a maximum error of about 2%. However, considering the experimental errors, the PMCEPT results can provide the gold standard of dose distributions for radiotherapy. The computing time was also much faster, compared to that needed for experiments, although it is still a bottleneck for direct applications to the daily routine treatment planning procedure.
A Monte Carlo transport code study of the space radiation environment using FLUKA and ROOT
Wilson, T; Carminati, F; Brun, R; Ferrari, A; Sala, P; Empl, A; MacGibbon, J
2001-01-01
We report on the progress of a current study aimed at developing a state-of-the-art Monte-Carlo computer simulation of the space radiation environment using advanced computer software techniques recently available at CERN, the European Laboratory for Particle Physics in Geneva, Switzerland. By taking the next-generation computer software appearing at CERN and adapting it to known problems in the implementation of space exploration strategies, this research is identifying changes necessary to bring these two advanced technologies together. The radiation transport tool being developed is tailored to the problem of taking measured space radiation fluxes impinging on the geometry of any particular spacecraft or planetary habitat and simulating the evolution of that flux through an accurate model of the spacecraft material. The simulation uses the latest known results in low-energy and high-energy physics. The output is a prediction of the detailed nature of the radiation environment experienced in space as well a...
Characterisation of the TRIUMF neutron facility using a Monte Carlo simulation code.
Monk, S D; Abram, T; Joyce, M J
2015-04-01
Here, the characterisation of the high-energy neutron field at TRIUMF (The Tri Universities Meson Facility, Vancouver, British Columbia) with Monte Carlo simulation software is described. The package used is MCNPX version 2.6.0, with the neutron fluence rate determined at three locations within the TRIUMF Thermal Neutron Facility (TNF), including the exit of the neutron channel where users of the facility can test devices that may be susceptible to the effects of this form of radiation. The facility is often used to roughly emulate the field likely to be encountered at high altitudes due to radiation of galactic origin and thus the simulated information is compared with the energy spectrum calculated to be due to neutron radiation of cosmic origin at typical aircraft altitudes. The calculated values were also compared with neutron flux measurements that were estimated using the activation of various foils by the staff of the facility, showing agreement within an order of magnitude.
Towards scalable parellelism in Monte Carlo particle transport codes using remote memory access
Energy Technology Data Exchange (ETDEWEB)
Romano, Paul K [Los Alamos National Laboratory; Brown, Forrest B [Los Alamos National Laboratory; Forget, Benoit [MIT
2010-01-01
One forthcoming challenge in the area of high-performance computing is having the ability to run large-scale problems while coping with less memory per compute node. In this work, they investigate a novel data decomposition method that would allow Monte Carlo transport calculations to be performed on systems with limited memory per compute node. In this method, each compute node remotely retrieves a small set of geometry and cross-section data as needed and remotely accumulates local tallies when crossing the boundary of the local spatial domain. initial results demonstrate that while the method does allow large problems to be run in a memory-limited environment, achieving scalability may be difficult due to inefficiencies in the current implementation of RMA operations.
Directory of Open Access Journals (Sweden)
MEHMET E. KORKMAZ
2014-06-01
Full Text Available In this research, we investigated the burnup characteristics and the conversion of fertile 232Th into fissile 233U in the core of a Sodium-Cooled Fast Reactor (SFR. The SFR fuel assemblies were designed for burning 232Th fuel (fuel pin 1 and 233U fuel (fuel pin 2 and include mixed minor actinide compositions. Monte Carlo simulations were performed using Serpent Code1.1.19 to compare with CRAM (Chebyshev Rational Approximation Method and TTA (Transmutation Trajectory Analysis method in the burnup calculation mode. The total heating power generated in the system was assumed to be 2000 MWth. During the reactor operation period of 600 days, the effective multiplication factor (keff was between 0.964 and 0.954 and peaking factor is 1.88867.
Evaluation of a 50-MV photon therapy beam from a racetrack microtron using MCNP4B Monte Carlo code
Energy Technology Data Exchange (ETDEWEB)
Gudowska, I.; Svensson, R. [Karolinska Inst. (Sweden). Dept. of Medical Radiation Physics]|[Huddinge Univ. Hospital, Stockholm (Sweden). Dept. of Medical Physics; Sorcini, B. [Karolinska Inst. (Sweden). Dept. of Medical Radiation Physics]|[Stockholm Univ. (Sweden)
2001-07-01
High energy photon therapy beam from the 50 MV racetrack microtron has been evaluated using the Monte Carlo code MCNP4B. The spatial and energy distribution of photons, radial and depth dose distributions in the phantom are calculated for the stationary and scanned photon beams from different targets. The calculated dose distributions are compared to the experimental data using a silicon diode detector. Measured and calculated depth-dose distributions are in fairly good agreement, within 2-3% for the positions in the range 2-30 cm in the phantom, whereas the larger discrepancies up to 10% are observed in the dose build-up region. For the stationary beams the differences in the calculated and measured radial dose distributions are about 2-10%. (orig.)
Energy Technology Data Exchange (ETDEWEB)
Korkmaz, Mehmet E.; Agar, Osman [Karamanoglu Mehmetbey University, Faculty of Kamil Oezdag Science, Karaman (Turkmenistan)
2014-06-15
In this research, we investigated the burnup characteristics and the conversion of fertile {sup 232}Th into fissile {sup 233}U in the core of a Sodium-Cooled Fast Reactor (SFR). The SFR fuel assemblies were designed for burning {sup 232}Th fuel (fuel pin 1) and {sup 233}U fuel (fuel pin 2) and include mixed minor actinide compositions. Monte Carlo simulations were performed using Serpent Code1.1.19 to compare with CRAM (Chebyshev Rational Approximation Method) and TTA (Transmutation Trajectory Analysis) method in the burnup calculation mode. The total heating power generated in the system was assumed to be 2000 MWth. During the reactor operation period of 600 days, the effective multiplication factor (keff) was between 0.964 and 0.954 and peaking factor is 1.88867.
Dos Santos, M; Clairand, I; Gruel, G; Barquinero, J F; Incerti, S; Villagrasa, C
2014-10-01
The purpose of this work is to evaluate the influence of the chromatin condensation on the number of direct double-strand break (DSB) damages induced by ions. Two geometries of chromosome territories containing either condensed or decondensed chromatin were implemented as biological targets in the Geant4 Monte Carlo simulation code and proton and alpha irradiation was simulated using the Geant4-DNA processes. A DBSCAN algorithm was used in order to detect energy deposition clusters that could give rise to single-strand breaks or DSBs on the DNA molecule. The results of this study show an increase in the number and complexity of DNA DSBs in condensed chromatin when compared with decondensed chromatin.
Indian Academy of Sciences (India)
MOHAMED M OULD; DIB A S A; BELBACHIR A H
2016-07-01
Cosmic rays cause significant damage to the electronic equipments of the aircrafts. In this paper, we have investigated the accumulation of the deposited energy of cosmic rays on the Earth’s atmosphere, especially in the aircraft area. In fact, if a high-energy neutron or proton interacts with a nanodevice having only a few atoms, this neutron or proton particle can change the nature of this device and destroy it. Our simulation based on Monte Carlo using Geant4 code shows that the deposited energy of neutron particles ranging between 200MeV and 5 GeV are strongly concentrated in the region between 10 and 15 km from the sea level which is exactly the avionic area. However, the Bragg peak energy of proton particle is slightly localized above the avionic area.
Initial validation of 4D-model for a clinical PET scanner using the Monte Carlo code gate
Energy Technology Data Exchange (ETDEWEB)
Vieira, Igor F.; Lima, Fernando R.A.; Gomes, Marcelo S., E-mail: falima@cnen.gov.b [Centro Regional de Ciencias Nucleares do Nordeste (CRCN-NE/CNEN-PE), Recife, PE (Brazil); Vieira, Jose W.; Pacheco, Ludimila M. [Instituto Federal de Educacao, Ciencia e Tecnologia (IFPE), Recife, PE (Brazil); Chaves, Rosa M. [Instituto de Radium e Supervoltagem Ivo Roesler, Recife, PE (Brazil)
2011-07-01
Building exposure computational models (ECM) of emission tomography (PET and SPECT) currently has several dedicated computing tools based on Monte Carlo techniques (SimSET, SORTEO, SIMIND, GATE). This paper is divided into two steps: (1) using the dedicated code GATE (Geant4 Application for Tomographic Emission) to build a 4D model (where the fourth dimension is the time) of a clinical PET scanner from General Electric, GE ADVANCE, simulating the geometric and electronic structures suitable for this scanner, as well as some phenomena 4D, for example, rotating gantry; (2) the next step is to evaluate the performance of the model built here in the reproduction of test noise equivalent count rate (NEC) based on the NEMA Standards Publication NU protocols 2-2007 for this tomography. The results for steps (1) and (2) will be compared with experimental and theoretical values of the literature showing actual state of art of validation. (author)
Energy Technology Data Exchange (ETDEWEB)
Infantino, Angelo, E-mail: angelo.infantino@unibo.it [Department of Industrial Engineering, Montecuccolino Laboratory, University of Bologna, Via dei Colli 16, 40136 Bologna (Italy); Oehlke, Elisabeth [TRIUMF, 4004 Wesbrook Mall, V6T 2A3 Vancouver, BC (Canada); Department of Radiation Science & Technology, Delft University of Technology, Postbus 5, 2600 AA Delft (Netherlands); Mostacci, Domiziano [Department of Industrial Engineering, Montecuccolino Laboratory, University of Bologna, Via dei Colli 16, 40136 Bologna (Italy); Schaffer, Paul; Trinczek, Michael; Hoehr, Cornelia [TRIUMF, 4004 Wesbrook Mall, V6T 2A3 Vancouver, BC (Canada)
2016-01-01
The Monte Carlo code FLUKA is used to simulate the production of a number of positron emitting radionuclides, {sup 18}F, {sup 13}N, {sup 94}Tc, {sup 44}Sc, {sup 68}Ga, {sup 86}Y, {sup 89}Zr, {sup 52}Mn, {sup 61}Cu and {sup 55}Co, on a small medical cyclotron with a proton beam energy of 13 MeV. Experimental data collected at the TR13 cyclotron at TRIUMF agree within a factor of 0.6 ± 0.4 with the directly simulated data, except for the production of {sup 55}Co, where the simulation underestimates the experiment by a factor of 3.4 ± 0.4. The experimental data also agree within a factor of 0.8 ± 0.6 with the convolution of simulated proton fluence and cross sections from literature. Overall, this confirms the applicability of FLUKA to simulate radionuclide production at 13 MeV proton beam energy.
Energy Technology Data Exchange (ETDEWEB)
Liu, T; Lin, H; Xu, X [Rensselaer Polytechnic Institute, Troy, NY (United States); Stabin, M [Vanderbilt Univ Medical Ctr, Nashville, TN (United States)
2015-06-15
Purpose: To develop a nuclear medicine dosimetry module for the GPU-based Monte Carlo code ARCHER. Methods: We have developed a nuclear medicine dosimetry module for the fast Monte Carlo code ARCHER. The coupled electron-photon Monte Carlo transport kernel included in ARCHER is built upon the Dose Planning Method code (DPM). The developed module manages the radioactive decay simulation by consecutively tracking several types of radiation on a per disintegration basis using the statistical sampling method. Optimization techniques such as persistent threads and prefetching are studied and implemented. The developed module is verified against the VIDA code, which is based on Geant4 toolkit and has previously been verified against OLINDA/EXM. A voxelized geometry is used in the preliminary test: a sphere made of ICRP soft tissue is surrounded by a box filled with water. Uniform activity distribution of I-131 is assumed in the sphere. Results: The self-absorption dose factors (mGy/MBqs) of the sphere with varying diameters are calculated by ARCHER and VIDA respectively. ARCHER’s result is in agreement with VIDA’s that are obtained from a previous publication. VIDA takes hours of CPU time to finish the computation, while it takes ARCHER 4.31 seconds for the 12.4-cm uniform activity sphere case. For a fairer CPU-GPU comparison, more effort will be made to eliminate the algorithmic differences. Conclusion: The coupled electron-photon Monte Carlo code ARCHER has been extended to radioactive decay simulation for nuclear medicine dosimetry. The developed code exhibits good performance in our preliminary test. The GPU-based Monte Carlo code is developed with grant support from the National Institute of Biomedical Imaging and Bioengineering through an R01 grant (R01EB015478)
Energy Technology Data Exchange (ETDEWEB)
Fonseca, T.C.F.; Bastos, F.M.; Figueiredo, M.T.T.; Souza, L.S.; Guimaraes, M.C.; Silva, C.R.E.; Mello, O.A.; Castelo e Silva, L.A.; Paixao, L., E-mail: tcff01@gmail.com [Centro de Desenvolvimento da Tecnologia Nuclear (CDTN/CNEN-MG), Belo Horizonte, MG (Brazil); Benavente, J.A.; Paiva, F.G. [Universidade Federal de Minas Gerais (PCTN/UFMG), Belo Horizonte, MG (Brazil). Curso de Pos-Graduacao em Ciencias e Tecnicas Nucleares
2015-07-01
Computational Monte Carlo (MC) codes have been used for simulation of nuclear installations mainly for internal monitoring of workers, the well known as Whole Body Counters (WBC). The main goal of this project was the modeling and simulation of the counting efficiency (CE) of a WBC system using three different MC codes: MCNPX, EGSnrc and VMC in-vivo. The simulations were performed for three different groups of analysts. The results shown differences between the three codes, as well as in the results obtained by the same code and modeled by different analysts. Moreover, all the results were also compared to the experimental results obtained in laboratory for meaning of validation and final comparison. In conclusion, it was possible to detect the influence on the results when the system is modeled by different analysts using the same MC code and in which MC code the results were best suited, when comparing to the experimental data result. (author)
The Current Status and Future Plans for the Monte Carlo Codes MONK and MCBEND
Smith, N.; Shuttleworth, T.; Grimstone, M.; Hutton, L.; Armishaw, M.; Bird, A.; France, N.; Connolly, S.
MONK and MCBEND are software tools used for criticality/reactor physics and shielding/dosimetry/general radiation transport applications respectively. The codes are developed by a collaboration comprising AEA Technology and BNFL, and AEA Technology's ANSWERS Software Service supplies them to customers throughout the world on a commercial basis. In keeping with this commercial nature it is vital that the codes' development programmes evolve to meet the diverse expectations of the current and potential customer base. This involves striving to maintain an acceptable balance in the development of the various components that comprise a modern software package. This paper summarises the current status of MONK and MCBEND by indicating how the task of trying to achieve this difficult balance has been addressed in the recent past and is being addressed for the future.
Institute of Scientific and Technical Information of China (English)
朱贵凤; 邹杨; 李明海; 严睿; 彭红花; 徐洪杰
2015-01-01
The burnup calculation code PBRE coupling MCNP5 and ORIGEN2 was developed for pebble‐bed high temperature reactor at equilibrium state ,and it can be used to analyze the neutronic performance of equilibrium core .The iteration method was optimized in order to save Monte Carlo calculation time ,and the convergence can be reached in 10 iterative steps .The average discharged burnup for HTR‐10 is consistent with literature ,and it indicates that the PBRE is suitable to analyze the burnup for pebble‐bed reactor at equilibrium state .%基于MCNP5和ORIGEN2耦合方法，开发了平衡态下球床高温堆的燃耗计算程序PBRE ，用于堆的性能价值分析。为节省蒙特卡罗计算时间，对迭代收敛的方法进行优化，使之可在10个迭代步内收敛。使用PBRE对清华大学H T R‐10进行建模计算，得到的平均卸料燃耗深度与文献报道值一致，表明PBRE程序适用于球床堆平衡态的燃耗分析。
Burnup simulations of different fuel grades using the MCNPX Monte Carlo code
Directory of Open Access Journals (Sweden)
Asah-Opoku Fiifi
2014-01-01
Full Text Available Global energy problems range from the increasing cost of fuel to the unequal distribution of energy resources and the potential climate change resulting from the burning of fossil fuels. A sustainable nuclear energy would augment the current world energy supply and serve as a reliable future energy source. This research focuses on Monte Carlo simulations of pressurized water reactor systems. Three different fuel grades - mixed oxide fuel (MOX, uranium oxide fuel (UOX, and commercially enriched uranium or uranium metal (CEU - are used in this simulation and their impact on the effective multiplication factor (Keff and, hence, criticality and total radioactivity of the reactor core after fuel burnup analyzed. The effect of different clad materials on Keff is also studied. Burnup calculation results indicate a buildup of plutonium isotopes in UOX and CEU, as opposed to a decline in plutonium radioisotopes for MOX fuel burnup time. For MOX fuel, a decrease of 31.9% of the fissile plutonium isotope is observed, while for UOX and CEU, fissile plutonium isotopes increased by 82.3% and 83.8%, respectively. Keff results show zircaloy as a much more effective clad material in comparison to zirconium and stainless steel.
Monte Carlo simulation of a multi-leaf collimator design for telecobalt machine using BEAMnrc code
Directory of Open Access Journals (Sweden)
Ayyangar Komanduri
2010-01-01
Full Text Available This investigation aims to design a practical multi-leaf collimator (MLC system for the cobalt teletherapy machine and check its radiation properties using the Monte Carlo (MC method. The cobalt machine was modeled using the BEAMnrc Omega-Beam MC system, which could be freely downloaded from the website of the National Research Council (NRC, Canada. Comparison with standard depth dose data tables and the theoretically modeled beam showed good agreement within 2%. An MLC design with low melting point alloy (LMPA was tested for leakage properties of leaves. The LMPA leaves with a width of 7 mm and height of 6 cm, with tongue and groove of size 2 mm wide by 4 cm height, produced only 4% extra leakage compared to 10 cm height tungsten leaves. With finite 60 Co source size, the interleaf leakage was insignificant. This analysis helped to design a prototype MLC as an accessory mount on a cobalt machine. The complete details of the simulation process and analysis of results are discussed.
Griesheimer, D. P.; Gill, D. F.; Nease, B. R.; Sutton, T. M.; Stedry, M. H.; Dobreff, P. S.; Carpenter, D. C.; Trumbull, T. H.; Caro, E.; Joo, H.; Millman, D. L.
2014-06-01
MC21 is a continuous-energy Monte Carlo radiation transport code for the calculation of the steady-state spatial distributions of reaction rates in three-dimensional models. The code supports neutron and photon transport in fixed source problems, as well as iterated-fission-source (eigenvalue) neutron transport problems. MC21 has been designed and optimized to support large-scale problems in reactor physics, shielding, and criticality analysis applications. The code also supports many in-line reactor feedback effects, including depletion, thermal feedback, xenon feedback, eigenvalue search, and neutron and photon heating. MC21 uses continuous-energy neutron/nucleus interaction physics over the range from 10-5 eV to 20 MeV. The code treats all common neutron scattering mechanisms, including fast-range elastic and non-elastic scattering, and thermal- and epithermal-range scattering from molecules and crystalline materials. For photon transport, MC21 uses continuous-energy interaction physics over the energy range from 1 keV to 100 GeV. The code treats all common photon interaction mechanisms, including Compton scattering, pair production, and photoelectric interactions. All of the nuclear data required by MC21 is provided by the NDEX system of codes, which extracts and processes data from EPDL-, ENDF-, and ACE-formatted source files. For geometry representation, MC21 employs a flexible constructive solid geometry system that allows users to create spatial cells from first- and second-order surfaces. The system also allows models to be built up as hierarchical collections of previously defined spatial cells, with interior detail provided by grids and template overlays. Results are collected by a generalized tally capability which allows users to edit integral flux and reaction rate information. Results can be collected over the entire problem or within specific regions of interest through the use of phase filters that control which particles are allowed to score each
Energy Technology Data Exchange (ETDEWEB)
Lin, Yi-Chun [Health Physics Division, Institute of Nuclear Energy Research, Taoyuan County, Taiwan (China); Huang, Tseng-Te, E-mail: huangtt@iner.gov.tw [Health Physics Division, Institute of Nuclear Energy Research, Taoyuan County, Taiwan (China); Liu, Yuan-Hao [Nuclear Science and Technology Development Center, National Tsing Hua University, Hsinchu City, Taiwan (China); Chen, Wei-Lin [Institute of Nuclear Engineering and Science, National Tsing Hua University, Hsinchu City, Taiwan (China); Chen, Yen-Fu [Atomic Energy Council, New Taipei City, Taiwan (China); Wu, Shu-Wei [Dept. of Biomedical Engineering and Environmental Sciences, National Tsing Hua University, Hsinchu, Taiwan (China); Nievaart, Sander [Institute for Energy, Joint Research Centre, European Commission, Petten (Netherlands); Jiang, Shiang-Huei [Dept. of Engineering and System Science, National Tsing Hua University, Hsinchu, Taiwan (China)
2015-06-01
The paired ionization chambers (ICs) technique is commonly employed to determine neutron and photon doses in radiology or radiotherapy neutron beams, where neutron dose shows very strong dependence on the accuracy of accompanying high energy photon dose. During the dose derivation, it is an important issue to evaluate the photon and electron response functions of two commercially available ionization chambers, denoted as TE(TE) and Mg(Ar), used in our reactor based epithermal neutron beam. Nowadays, most perturbation corrections for accurate dose determination and many treatment planning systems are based on the Monte Carlo technique. We used general purposed Monte Carlo codes, MCNP5, EGSnrc, FLUKA or GEANT4 for benchmark verifications among them and carefully measured values for a precise estimation of chamber current from absorbed dose rate of cavity gas. Also, energy dependent response functions of two chambers were calculated in a parallel beam with mono-energies from 20 keV to 20 MeV photons and electrons by using the optimal simple spherical and detailed IC models. The measurements were performed in the well-defined (a) four primary M-80, M-100, M120 and M150 X-ray calibration fields, (b) primary {sup 60}Co calibration beam, (c) 6 MV and 10 MV photon, (d) 6 MeV and 18 MeV electron LINACs in hospital and (e) BNCT clinical trials neutron beam. For the TE(TE) chamber, all codes were almost identical over the whole photon energy range. In the Mg(Ar) chamber, MCNP5 showed lower response than other codes for photon energy region below 0.1 MeV and presented similar response above 0.2 MeV (agreed within 5% in the simple spherical model). With the increase of electron energy, the response difference between MCNP5 and other codes became larger in both chambers. Compared with the measured currents, MCNP5 had the difference from the measurement data within 5% for the {sup 60}Co, 6 MV, 10 MV, 6 MeV and 18 MeV LINACs beams. But for the Mg(Ar) chamber, the derivations
Hybrid parallel code acceleration methods in full-core reactor physics calculations
Energy Technology Data Exchange (ETDEWEB)
Courau, T.; Plagne, L.; Ponicot, A. [EDF R and D, 1, Avenue du General de Gaulle, 92141 Clamart Cedex (France); Sjoden, G. [Nuclear and Radiological Engineering, Georgia Inst. of Technology, Atlanta, GA 30332 (United States)
2012-07-01
When dealing with nuclear reactor calculation schemes, the need for three dimensional (3D) transport-based reference solutions is essential for both validation and optimization purposes. Considering a benchmark problem, this work investigates the potential of discrete ordinates (Sn) transport methods applied to 3D pressurized water reactor (PWR) full-core calculations. First, the benchmark problem is described. It involves a pin-by-pin description of a 3D PWR first core, and uses a 8-group cross-section library prepared with the DRAGON cell code. Then, a convergence analysis is performed using the PENTRAN parallel Sn Cartesian code. It discusses the spatial refinement and the associated angular quadrature required to properly describe the problem physics. It also shows that initializing the Sn solution with the EDF SPN solver COCAGNE reduces the number of iterations required to converge by nearly a factor of 6. Using a best estimate model, PENTRAN results are then compared to multigroup Monte Carlo results obtained with the MCNP5 code. Good consistency is observed between the two methods (Sn and Monte Carlo), with discrepancies that are less than 25 pcm for the k{sub eff}, and less than 2.1% and 1.6% for the flux at the pin-cell level and for the pin-power distribution, respectively. (authors)
Lee, Y.-K.; Brun, E.
2014-04-01
The Sodium-cooled fast neutron reactor ASTRID is currently under design and development in France. Traditional ECCO/ERANOS fast reactor code system used for ASTRID core design calculations relies on multi-group JEFF-3.1.1 data library. To gauge the use of ENDF/B-VII.0 and JEFF-3.1.1 nuclear data libraries in the fast reactor applications, two recent OECD/NEA computational benchmarks specified by Argonne National Laboratory were calculated. Using the continuous-energy TRIPOLI-4 Monte Carlo transport code, both ABR-1000 MWth MOX core and metallic (U-Pu) core were investigated. Under two different fast neutron spectra and two data libraries, ENDF/B-VII.0 and JEFF-3.1.1, reactivity impact studies were performed. Using JEFF-3.1.1 library under the BOEC (Beginning of equilibrium cycle) condition, high reactivity effects of 808 ± 17 pcm and 1208 ± 17 pcm were observed for ABR-1000 MOX core and metallic core respectively. To analyze the causes of these differences in reactivity, several TRIPOLI-4 runs using mixed data libraries feature allow us to identify the nuclides and the nuclear data accounting for the major part of the observed reactivity discrepancies.
Mairani, A; Kraemer, M; Sommerer, F; Parodi, K; Scholz, M; Cerutti, F; Ferrari, A; Fasso, A
2010-01-01
Clinical Monte Carlo (MC) calculations for carbon ion therapy have to provide absorbed and RBE-weighted dose. The latter is defined as the product of the dose and the relative biological effectiveness (RBE). At the GSI Helmholtzzentrum fur Schwerionenforschung as well as at the Heidelberg Ion Therapy Center (HIT), the RBE values are calculated according to the local effect model (LEM). In this paper, we describe the approach followed for coupling the FLUKA MC code with the LEM and its application to dose and RBE-weighted dose calculations for a superimposition of two opposed C-12 ion fields as applied in therapeutic irradiations. The obtained results are compared with the available experimental data of CHO (Chinese hamster ovary) cell survival and the outcomes of the GSI analytical treatment planning code TRiP98. Some discrepancies have been observed between the analytical and MC calculations of absorbed physical dose profiles, which can be explained by the differences between the laterally integrated depth-d...
Chen, Xuhui; Liang, Edison; Boettcher, Markus
2011-01-01
(abridged) We present a new time-dependent multi-zone radiative transfer code and its application to study the SSC emission of Mrk 421. The code couples Fokker-Planck and Monte Carlo methods, in a 2D geometry. For the first time all the light travel time effects (LCTE) are fully considered, along with a proper treatment of Compton cooling, which depends on them. We study a set of simple scenarios where the variability is produced by injection of relativistic electrons as a `shock front' crosses the emission region. We consider emission from two components, with the second one either being pre-existing and co-spatial and participating in the evolution of the active region, or spatially separated and independent, only diluting the observed variability. Temporal and spectral results of the simulation are compared to the multiwavelength observations of Mrk 421 in March 2001. We find parameters that can adequately fit the observed SEDs and multiwavelength light curves and correlations. There remain however a few o...
Magro, G.; Dahle, T. J.; Molinelli, S.; Ciocca, M.; Fossati, P.; Ferrari, A.; Inaniwa, T.; Matsufuji, N.; Ytre-Hauge, K. S.; Mairani, A.
2017-05-01
Particle therapy facilities often require Monte Carlo (MC) simulations to overcome intrinsic limitations of analytical treatment planning systems (TPS) related to the description of the mixed radiation field and beam interaction with tissue inhomogeneities. Some of these uncertainties may affect the computation of effective dose distributions; therefore, particle therapy dedicated MC codes should provide both absorbed and biological doses. Two biophysical models are currently applied clinically in particle therapy: the local effect model (LEM) and the microdosimetric kinetic model (MKM). In this paper, we describe the coupling of the NIRS (National Institute for Radiological Sciences, Japan) clinical dose to the FLUKA MC code. We moved from the implementation of the model itself to its application in clinical cases, according to the NIRS approach, where a scaling factor is introduced to rescale the (carbon-equivalent) biological dose to a clinical dose level. A high level of agreement was found with published data by exploring a range of values for the MKM input parameters, while some differences were registered in forward recalculations of NIRS patient plans, mainly attributable to differences with the analytical TPS dose engine (taken as reference) in describing the mixed radiation field (lateral spread and fragmentation). We presented a tool which is being used at the Italian National Center for Oncological Hadrontherapy to support the comparison study between the NIRS clinical dose level and the LEM dose specification.
Energy Technology Data Exchange (ETDEWEB)
Birikorang, S.A., E-mail: anddydat@yahoo.com [Department of Nuclear Engineering and Material Science, School of Nuclear and Allied Sciences (SNAS), University of Ghana, P.O. Box AE 1, Atomic Energy, Accra (Ghana); Akaho, E.H.K.; Nyarko, B.J.B. [National Nuclear Research Institute, Ghana Atomic Energy Commission, P.O. Box LG 80, Legon, Accra-Ghana (Ghana); Ampomah-Amoako, E.; Seth, Debrah K.; Gyabour, R.A.; Sogbgaji, R.B.M. [Department of Nuclear Engineering and Material Science, School of Nuclear and Allied Sciences (SNAS), University of Ghana, P.O. Box AE 1, Atomic Energy, Accra (Ghana)
2011-07-15
Highlights: > The photo-neutron source was investigated within Ghana MNSR irradiation channels. > Irradiation channels under study were inner, outer and the fission chamber. > Thermal rated power at sub-critical state was estimated. > Neutron flux variation was investigated within the channels. > MCNP code has been used to investigate the flux variation. - Abstract: Computer simulation was carried out for photo-neutron source variation in outer irradiation channel, inner irradiation channels and the fission channel of a tank-in-pool reactor, a Miniature Neutron Source Reactor (MNSR) in sub-critical condition. Evaluation of the photo-neutron was done after the reactor has been in sub-critical condition for three month period using Monte Carlo Neutron Particle (MCNP) code. Neutron flux monitoring from the Micro Computer Control Loop System (MCCLS) was also investigated at sub-critical condition. The recorded neutron fluxes from the MCCLS after investigations were used to calculate the power of the reactor at sub-critical state. The computed power at sub-critical state was used to normalize the un-normalized results from the MCNP.
Simulation of Electron Trajectories in the Multicusp Ion Source Using Geantn4 Monte Carlo Code
Khodadadi Azadboni, Fatemeh; Sedaghatizade, Mahmood
2010-04-01
To optimize the multicusp ion source, understanding of transport properties of electrons is indispensable. Since the transport of electrons in the multicusp ion source is a three-dimensional problem, we use the 3D computer code Geant4, to model the particle trajectories. The goal is to study the effect of electron injection into a cylindrical gas chamber and the electron trajectories. The role of the magnetic filter in contemporary negative ion sources is analyzed. The conditions in the magnetic filter adjacent to the plasma electrode optimum for the generation, formation, and extraction of an H- ion beam are found. The simulation results are in good agreement with the experimental data.
Directory of Open Access Journals (Sweden)
Nilseia Aparecida Barbosa
2014-08-01
Full Text Available Purpose: Melanoma at the choroid region is the most common primary cancer that affects the eye in adult patients. Concave ophthalmic applicators with 106Ru/106Rh beta sources are the more used for treatment of these eye lesions, mainly lesions with small and medium dimensions. The available treatment planning system for 106Ru applicators is based on dose distributions on a homogeneous water sphere eye model, resulting in a lack of data in the literature of dose distributions in the eye radiosensitive structures, information that may be crucial to improve the treatment planning process, aiming the maintenance of visual acuity. Methods: The Monte Carlo code MCNPX was used to calculate the dose distribution in a complete mathematical model of the human eye containing a choroid melanoma; considering the eye actual dimensions and its various component structures, due to an ophthalmic brachytherapy treatment, using 106Ru/106Rh beta-ray sources. Two possibilities were analyzed; a simple water eye and a heterogeneous eye considering all its structures. Two concave applicators, CCA and CCB manufactured by BEBIG and a complete mathematical model of the human eye were modeled using the MCNPX code. Results and Conclusion: For both eye models, namely water model and heterogeneous model, mean dose values simulated for the same eye regions are, in general, very similar, excepting for regions very distant from the applicator, where mean dose values are very low, uncertainties are higher and relative differences may reach 20.4%. For the tumor base and the eye structures closest to the applicator, such as sclera, choroid and retina, the maximum difference observed was 4%, presenting the heterogeneous model higher mean dose values. For the other eye regions, the higher doses were obtained when the homogeneous water eye model is taken into consideration. Mean dose distributions determined for the homogeneous water eye model are similar to those obtained for the
Botta, F; Mairani, A; Battistoni, G; Cremonesi, M; Di Dia, A; Fassò, A; Ferrari, A; Ferrari, M; Paganelli, G; Pedroli, G; Valente, M
2011-07-01
The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, FLUKA Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, FLUKA has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. FLUKA DPKS have been calculated in both water and compact bone for monoenergetic electrons (10-3 MeV) and for beta emitting isotopes commonly used for therapy (89Sr, 90Y, 131I 153Sm, 177Lu, 186Re, and 188Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. FLUKA outcomes have been compared to PENELOPE v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (ETRAN, GEANT4, MCNPX) has been done. Maximum percentage differences within 0.8.RCSDA and 0.9.RCSDA for monoenergetic electrons (RCSDA being the continuous slowing down approximation range) and within 0.8.X90 and 0.9.X90 for isotopes (X90 being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9.RCSDA and 0.9.X90 for electrons and isotopes, respectively. Concerning monoenergetic electrons, within 0.8.RCSDA (where 90%-97% of the particle energy is deposed), FLUKA and PENELOPE agree mostly within 7%, except for 10 and 20 keV electrons (12% in water, 8.3% in bone). The
Energy Technology Data Exchange (ETDEWEB)
Botta, F; Di Dia, A; Pedroli, G; Mairani, A; Battistoni, G; Fasso, A; Ferrari, A; Ferrari, M; Paganelli, G
2011-06-01
The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, fluka Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, fluka has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one.Methods: fluka DPKs have been calculated in both water and compact bone for monoenergetic electrons (10–3 MeV) and for beta emitting isotopes commonly used for therapy (89Sr, 90Y, 131I, 153Sm, 177Lu, 186Re, and 188Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. fluka outcomes have been compared to penelope v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (etran, geant4, mcnpx) has been done. Maximum percentage differences within 0.8·RCSDA and 0.9·RCSDA for monoenergetic electrons (RCSDA being the continuous slowing down approximation range) and within 0.8·X90 and 0.9·X90 for isotopes (X90 being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9·RCSDA and 0.9·X90 for electrons and isotopes, respectively.Results: Concerning monoenergetic electrons, within 0.8·RCSDA (where 90%–97% of the particle energy is deposed), fluka and penelope agree mostly within 7%, except for 10 and 20 keV electrons (12% in water, 8
Energy Technology Data Exchange (ETDEWEB)
Botta, F.; Mairani, A.; Battistoni, G.; Cremonesi, M.; Di Dia, A.; Fasso, A.; Ferrari, A.; Ferrari, M.; Paganelli, G.; Pedroli, G.; Valente, M. [Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); Istituto Nazionale di Fisica Nucleare (I.N.F.N.), Via Celoria 16, 20133 Milan (Italy); Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); Jefferson Lab, 12000 Jefferson Avenue, Newport News, Virginia 23606 (United States); CERN, 1211 Geneva 23 (Switzerland); Medical Physics Department, European Institute of Oncology, Milan (Italy); Nuclear Medicine Department, European Institute of Oncology, Via Ripamonti 435, 2014 Milan (Italy); Medical Physics Department, European Institute of Oncology, Via Ripamonti 435, 20141 Milan (Italy); FaMAF, Universidad Nacional de Cordoba and CONICET, Cordoba, Argentina C.P. 5000 (Argentina)
2011-07-15
Purpose: The calculation of patient-specific dose distribution can be achieved by Monte Carlo simulations or by analytical methods. In this study, fluka Monte Carlo code has been considered for use in nuclear medicine dosimetry. Up to now, fluka has mainly been dedicated to other fields, namely high energy physics, radiation protection, and hadrontherapy. When first employing a Monte Carlo code for nuclear medicine dosimetry, its results concerning electron transport at energies typical of nuclear medicine applications need to be verified. This is commonly achieved by means of calculation of a representative parameter and comparison with reference data. Dose point kernel (DPK), quantifying the energy deposition all around a point isotropic source, is often the one. Methods: fluka DPKs have been calculated in both water and compact bone for monoenergetic electrons (10{sup -3} MeV) and for beta emitting isotopes commonly used for therapy ({sup 89}Sr, {sup 90}Y, {sup 131}I, {sup 153}Sm, {sup 177}Lu, {sup 186}Re, and {sup 188}Re). Point isotropic sources have been simulated at the center of a water (bone) sphere, and deposed energy has been tallied in concentric shells. fluka outcomes have been compared to penelope v.2008 results, calculated in this study as well. Moreover, in case of monoenergetic electrons in water, comparison with the data from the literature (etran, geant4, mcnpx) has been done. Maximum percentage differences within 0.8{center_dot}R{sub CSDA} and 0.9{center_dot}R{sub CSDA} for monoenergetic electrons (R{sub CSDA} being the continuous slowing down approximation range) and within 0.8{center_dot}X{sub 90} and 0.9{center_dot}X{sub 90} for isotopes (X{sub 90} being the radius of the sphere in which 90% of the emitted energy is absorbed) have been computed, together with the average percentage difference within 0.9{center_dot}R{sub CSDA} and 0.9{center_dot}X{sub 90} for electrons and isotopes, respectively. Results: Concerning monoenergetic electrons
Development of Momentum Conserving Monte Carlo Simulation Code for ECCD Study in Helical Plasmas
Directory of Open Access Journals (Sweden)
Murakami S.
2015-01-01
Full Text Available Parallel momentum conserving collision model is developed for GNET code, in which a linearized drift kinetic equation is solved in the five dimensional phase-space to study the electron cyclotron current drive (ECCD in helical plasmas. In order to conserve the parallel momentum, we introduce a field particle collision term in addition to the test particle collision term. Two types of the field particle collision term are considered. One is the high speed limit model, where the momentum conserving term does not depend on the velocity of the background plasma and can be expressed in a simple form. The other is the velocity dependent model, which is derived from the Fokker–Planck collision term directly. In the velocity dependent model the field particle operator can be expressed using Legendre polynominals and, introducing the Rosenbluth potential, we derive the field particle term for each Legendre polynominals. In the GNET code, we introduce an iterative process to implement the momentum conserving collision operator. The high speed limit model is applied to the ECCD simulation of the heliotron-J plasma. The simulation results show a good conservation of the momentum with the iterative scheme.
Development of Momentum Conserving Monte Carlo Simulation Code for ECCD Study in Helical Plasmas
Murakami, S.; Hasegawa, S.; Moriya, Y.
2015-03-01
Parallel momentum conserving collision model is developed for GNET code, in which a linearized drift kinetic equation is solved in the five dimensional phase-space to study the electron cyclotron current drive (ECCD) in helical plasmas. In order to conserve the parallel momentum, we introduce a field particle collision term in addition to the test particle collision term. Two types of the field particle collision term are considered. One is the high speed limit model, where the momentum conserving term does not depend on the velocity of the background plasma and can be expressed in a simple form. The other is the velocity dependent model, which is derived from the Fokker-Planck collision term directly. In the velocity dependent model the field particle operator can be expressed using Legendre polynominals and, introducing the Rosenbluth potential, we derive the field particle term for each Legendre polynominals. In the GNET code, we introduce an iterative process to implement the momentum conserving collision operator. The high speed limit model is applied to the ECCD simulation of the heliotron-J plasma. The simulation results show a good conservation of the momentum with the iterative scheme.
Performance Analysis of Korean Liquid metal type TBM based on Monte Carlo code
Energy Technology Data Exchange (ETDEWEB)
Kim, C. H.; Han, B. S.; Park, H. J.; Park, D. K. [Seoul National Univ., Seoul (Korea, Republic of)
2007-01-15
The objective of this project is to analyze a nuclear performance of the Korean HCML(Helium Cooled Molten Lithium) TBM(Test Blanket Module) which will be installed in ITER(International Thermonuclear Experimental Reactor). This project is intended to analyze a neutronic design and nuclear performances of the Korean HCML ITER TBM through the transport calculation of MCCARD. In detail, we will conduct numerical experiments for analyzing the neutronic design of the Korean HCML TBM and the DEMO fusion blanket, and improving the nuclear performances. The results of the numerical experiments performed in this project will be utilized further for a design optimization of the Korean HCML TBM. In this project, Monte Carlo transport calculations for evaluating TBR (Tritium Breeding Ratio) and EMF (Energy Multiplication factor) were conducted to analyze a nuclear performance of the Korean HCML TBM. The activation characteristics and shielding performances for the Korean HCML TBM were analyzed using ORIGEN and MCCARD. We proposed the neutronic methodologies for analyzing the nuclear characteristics of the fusion blanket, which was applied to the blanket analysis of a DEMO fusion reactor. In the results, the TBR of the Korean HCML ITER TBM is 0.1352 and the EMF is 1.362. Taking into account a limitation for the Li amount in ITER TBM, it is expected that tritium self-sufficiency condition can be satisfied through a change of the Li quantity and enrichment. In the results of activation and shielding analysis, the activity drops to 1.5% of the initial value and the decay heat drops to 0.02% of the initial amount after 10 years from plasma shutdown.
Thermal neutron response of a boron-coated GEM detector via GEANT4 Monte Carlo code.
Jamil, M; Rhee, J T; Kim, H G; Ahmad, Farzana; Jeon, Y J
2014-10-22
In this work, we report the design configuration and the performance of the hybrid Gas Electron Multiplier (GEM) detector. In order to make the detector sensitive to thermal neutrons, the forward electrode of the GEM has been coated with the enriched boron-10 material, which works as a neutron converter. A total of 5×5cm(2) configuration of GEM has been used for thermal neutron studies. The response of the detector has been estimated via using GEANT4 MC code with two different physics lists. Using the QGSP_BIC_HP physics list, the neutron detection efficiency was determined to be about 3%, while with QGSP_BERT_HP physics list the efficiency was around 2.5%, at the incident thermal neutron energies of 25meV. The higher response of the detector proves that GEM-coated with boron converter improves the efficiency for thermal neutrons detection.
Blind Decoding of Multiple Description Codes over OFDM Systems via Sequential Monte Carlo
Directory of Open Access Journals (Sweden)
Guo Dong
2005-01-01
Full Text Available We consider the problem of transmitting a continuous source through an OFDM system. Multiple description scalar quantization (MDSQ is applied to the source signal, resulting in two correlated source descriptions. The two descriptions are then OFDM modulated and transmitted through two parallel frequency-selective fading channels. At the receiver, a blind turbo receiver is developed for joint OFDM demodulation and MDSQ decoding. Transformation of the extrinsic information of the two descriptions are exchanged between each other to improve system performance. A blind soft-input soft-output OFDM detector is developed, which is based on the techniques of importance sampling and resampling. Such a detector is capable of exchanging the so-called extrinsic information with the other component in the above turbo receiver, and successively improving the overall receiver performance. Finally, we also treat channel-coded systems, and a novel blind turbo receiver is developed for joint demodulation, channel decoding, and MDSQ source decoding.
Validation of a GPU-based Monte Carlo code (gPMC) for proton radiation therapy: clinical cases study
Giantsoudi, Drosoula; Schuemann, Jan; Jia, Xun; Dowdell, Stephen; Jiang, Steve; Paganetti, Harald
2015-03-01
Monte Carlo (MC) methods are recognized as the gold-standard for dose calculation, however they have not replaced analytical methods up to now due to their lengthy calculation times. GPU-based applications allow MC dose calculations to be performed on time scales comparable to conventional analytical algorithms. This study focuses on validating our GPU-based MC code for proton dose calculation (gPMC) using an experimentally validated multi-purpose MC code (TOPAS) and compare their performance for clinical patient cases. Clinical cases from five treatment sites were selected covering the full range from very homogeneous patient geometries (liver) to patients with high geometrical complexity (air cavities and density heterogeneities in head-and-neck and lung patients) and from short beam range (breast) to large beam range (prostate). Both gPMC and TOPAS were used to calculate 3D dose distributions for all patients. Comparisons were performed based on target coverage indices (mean dose, V95, D98, D50, D02) and gamma index distributions. Dosimetric indices differed less than 2% between TOPAS and gPMC dose distributions for most cases. Gamma index analysis with 1%/1 mm criterion resulted in a passing rate of more than 94% of all patient voxels receiving more than 10% of the mean target dose, for all patients except for prostate cases. Although clinically insignificant, gPMC resulted in systematic underestimation of target dose for prostate cases by 1-2% compared to TOPAS. Correspondingly the gamma index analysis with 1%/1 mm criterion failed for most beams for this site, while for 2%/1 mm criterion passing rates of more than 94.6% of all patient voxels were observed. For the same initial number of simulated particles, calculation time for a single beam for a typical head and neck patient plan decreased from 4 CPU hours per million particles (2.8-2.9 GHz Intel X5600) for TOPAS to 2.4 s per million particles (NVIDIA TESLA C2075) for gPMC. Excellent agreement was
Energy Technology Data Exchange (ETDEWEB)
O' Brien, M J; Procassini, R J; Joy, K I
2009-03-09
Validation of the problem definition and analysis of the results (tallies) produced during a Monte Carlo particle transport calculation can be a complicated, time-intensive processes. The time required for a person to create an accurate, validated combinatorial geometry (CG) or mesh-based representation of a complex problem, free of common errors such as gaps and overlapping cells, can range from days to weeks. The ability to interrogate the internal structure of a complex, three-dimensional (3-D) geometry, prior to running the transport calculation, can improve the user's confidence in the validity of the problem definition. With regard to the analysis of results, the process of extracting tally data from printed tables within a file is laborious and not an intuitive approach to understanding the results. The ability to display tally information overlaid on top of the problem geometry can decrease the time required for analysis and increase the user's understanding of the results. To this end, our team has integrated VisIt, a parallel, production-quality visualization and data analysis tool into Mercury, a massively-parallel Monte Carlo particle transport code. VisIt provides an API for real time visualization of a simulation as it is running. The user may select which plots to display from the VisIt GUI, or by sending VisIt a Python script from Mercury. The frequency at which plots are updated can be set and the user can visualize the simulation results as it is running.
Energy Technology Data Exchange (ETDEWEB)
Parreno Z, F.; Paucar J, R.; Picon C, C. [Instituto Peruano de Energia Nuclear, Av. Canada 1470, San Borja, Lima 41 (Peru)
1998-12-31
The simulation by Monte Carlo is tool which Medical Physics counts with it for the development of its research, the interest by this tool is growing, as we may observe in the main scientific journals for the years 1995-1997 where more than 27 % of the papers treat over Monte Carlo and/or its applications in the radiation transport.In the Peruvian Institute of Nuclear Energy we are implementing and making use of the MCNP4 and EGS4 codes. In this work are presented the general features of the Monte Carlo method and its more useful applications in Medical Physics. Likewise, it is made a simulation of the calculation of isodose curves in an interstitial treatment with Ir-192 wires in a mammary gland carcinoma. (Author)
Energy Technology Data Exchange (ETDEWEB)
Carvajal, M A; Palma, A J [Departamento de Electronica y Tecnologia de Computadores, Universidad de Granada, E-18071 Granada (Spain); Garcia-Pareja, S [Servicio de Radiofisica Hospitalaria, Hospital Regional Universitario ' Carlos Haya' , Avda Carlos Haya, s/n, E-29010 Malaga (Spain); Guirado, D [Servicio de RadiofIsica, Hospital Universitario ' San Cecilio' , Avda Dr Oloriz, 16, E-18012 Granada (Spain); Vilches, M [Servicio de Fisica y Proteccion Radiologica, Hospital Regional Universitario ' Virgen de las Nieves' , Avda Fuerzas Armadas, 2, E-18014 Granada (Spain); Anguiano, M; Lallena, A M [Departamento de Fisica Atomica, Molecular y Nuclear, Universidad de Granada, E-18071 Granada (Spain)], E-mail: carvajal@ugr.es, E-mail: garciapareja@gmail.com, E-mail: dguirado@ugr.es, E-mail: mvilches@ugr.es, E-mail: mangui@ugr.es, E-mail: ajpalma@ugr.es, E-mail: lallena@ugr.es
2009-10-21
In this work we have developed a simulation tool, based on the PENELOPE code, to study the response of MOSFET devices to irradiation with high-energy photons. The energy deposited in the extremely thin silicon dioxide layer has been calculated. To reduce the statistical uncertainties, an ant colony algorithm has been implemented to drive the application of splitting and Russian roulette as variance reduction techniques. In this way, the uncertainty has been reduced by a factor of {approx}5, while the efficiency is increased by a factor of above 20. As an application, we have studied the dependence of the response of the pMOS transistor 3N163, used as a dosimeter, with the incidence angle of the radiation for three common photons sources used in radiotherapy: a {sup 60}Co Theratron-780 and the 6 and 18 MV beams produced by a Mevatron KDS LINAC. Experimental and simulated results have been obtained for gantry angles of 0 deg., 15 deg., 30 deg., 45 deg., 60 deg. and 75 deg. The agreement obtained has permitted validation of the simulation tool. We have studied how to reduce the angular dependence of the MOSFET response by using an additional encapsulation made of brass in the case of the two LINAC qualities considered.
Implementation of tetrahedral-mesh geometry in Monte Carlo radiation transport code PHITS
Furuta, Takuya; Sato, Tatsuhiko; Han, Min Cheol; Yeom, Yeon Soo; Kim, Chan Hyeong; Brown, Justin L.; Bolch, Wesley E.
2017-06-01
A new function to treat tetrahedral-mesh geometry was implemented in the particle and heavy ion transport code systems. To accelerate the computational speed in the transport process, an original algorithm was introduced to initially prepare decomposition maps for the container box of the tetrahedral-mesh geometry. The computational performance was tested by conducting radiation transport simulations of 100 MeV protons and 1 MeV photons in a water phantom represented by tetrahedral mesh. The simulation was repeated with varying number of meshes and the required computational times were then compared with those of the conventional voxel representation. Our results show that the computational costs for each boundary crossing of the region mesh are essentially equivalent for both representations. This study suggests that the tetrahedral-mesh representation offers not only a flexible description of the transport geometry but also improvement of computational efficiency for the radiation transport. Due to the adaptability of tetrahedrons in both size and shape, dosimetrically equivalent objects can be represented by tetrahedrons with a much fewer number of meshes as compared its voxelized representation. Our study additionally included dosimetric calculations using a computational human phantom. A significant acceleration of the computational speed, about 4 times, was confirmed by the adoption of a tetrahedral mesh over the traditional voxel mesh geometry.
Tessonnier, T.; Mairani, A.; Brons, S.; Sala, P.; Cerutti, F.; Ferrari, A.; Haberer, T.; Debus, J.; Parodi, K.
2017-08-01
In the field of particle therapy helium ion beams could offer an alternative for radiotherapy treatments, owing to their interesting physical and biological properties intermediate between protons and carbon ions. We present in this work the comparisons and validations of the Monte Carlo FLUKA code against in-depth dosimetric measurements acquired at the Heidelberg Ion Beam Therapy Center (HIT). Depth dose distributions in water with and without ripple filter, lateral profiles at different depths in water and a spread-out Bragg peak were investigated. After experimentally-driven tuning of the less known initial beam characteristics in vacuum (beam lateral size and momentum spread) and simulation parameters (water ionization potential), comparisons of depth dose distributions were performed between simulations and measurements, which showed overall good agreement with range differences below 0.1 mm and dose-weighted average dose-differences below 2.3% throughout the entire energy range. Comparisons of lateral dose profiles showed differences in full-width-half-maximum lower than 0.7 mm. Measurements of the spread-out Bragg peak indicated differences with simulations below 1% in the high dose regions and 3% in all other regions, with a range difference less than 0.5 mm. Despite the promising results, some discrepancies between simulations and measurements were observed, particularly at high energies. These differences were attributed to an underestimation of dose contributions from secondary particles at large angles, as seen in a triple Gaussian parametrization of the lateral profiles along the depth. However, the results allowed us to validate FLUKA simulations against measurements, confirming its suitability for 4He ion beam modeling in preparation of clinical establishment at HIT. Future activities building on this work will include treatment plan comparisons using validated biological models between proton and helium ions, either within a Monte Carlo
Kahraman, A.; Kaya, S.; Jaksic, A.; Yilmaz, E.
2015-05-01
Radiation-sensing Field Effect Transistors (RadFETs or MOSFET dosimeters) with SiO2 gate dielectric have found applications in space, radiotherapy clinics, and high-energy physics laboratories. More sensitive RadFETs, which require modifications in device design, including gate dielectric, are being considered for personal dosimetry applications. This paper presents results of a detailed study of the RadFET energy response simulated with PENELOPE Monte Carlo code. Alternative materials to SiO2 were investigated to develop high-efficiency new radiation sensors. Namely, in addition to SiO2, Al2O3 and HfO2 were simulated as gate material and deposited energy amounts in these layers were determined for photon irradiation with energies between 20 keV and 5 MeV. The simulations were performed for capped and uncapped configurations of devices irradiated by point and extended sources, the surface area of which is the same with that of the RadFETs. Energy distributions of transmitted and backscattered photons were estimated using impact detectors to provide information about particle fluxes within the geometrical structures. The absorbed energy values in the RadFETs material zones were recorded. For photons with low and medium energies, the physical processes that affect the absorbed energy values in different gate materials are discussed on the basis of modelling results. The results show that HfO2 is the most promising of the simulated gate materials.
Shielding calculations for industrial 5/7.5MeV electron accelerators using the MCNP Monte Carlo Code
Peri, Eyal; Orion, Itzhak
2017-09-01
High energy X-rays from accelerators are used to irradiate food ingredients to prevent growth and development of unwanted biological organisms in food, and by that extend the shelf life of the products. The production of X-rays is done by accelerating 5 MeV electrons and bombarding them into a heavy target (high Z). Since 2004, the FDA has approved using 7.5 MeV energy, providing higher production rates with lower treatments costs. In this study we calculated all the essential data needed for a straightforward concrete shielding design of typical food accelerator rooms. The following evaluation is done using the MCNP Monte Carlo code system: (1) Angular dependence (0-180°) of photon dose rate for 5 MeV and 7.5 MeV electron beams bombarding iron, aluminum, gold, tantalum, and tungsten targets. (2) Angular dependence (0-180°) spectral distribution simulations of bremsstrahlung for gold, tantalum, and tungsten bombarded by 5 MeV and 7.5 MeV electron beams. (3) Concrete attenuation calculations in several photon emission angles for the 5 MeV and 7.5 MeV electron beams bombarding a tantalum target. Based on the simulation, we calculated the expected increase in dose rate for facilities intending to increase the energy from 5 MeV to 7.5 MeV, and the concrete width needed to be added in order to keep the existing dose rate unchanged.
Naeem, Hamza; Zheng, Huaqing; Cao, Ruifen; Pei, Xi; Hu, Liqin; Wu, Yican
2016-01-01
The $^{192}$Ir sources are widely used for high dose rate (HDR) brachytherapy treatments. The aim of this study is to simulate $^{192}$Ir MicroSelectron v2 HDR brachytherapy source and calculate the air kerma strength, dose rate constant, radial dose function and anisotropy function established in the updated AAPM Task Group 43 protocol. The EGSnrc Monte Carlo (MC) code package is used to calculate these dosimetric parameters, including dose contribution from secondary electron source and also contribution of bremsstrahlung photons to air kerma strength. The Air kerma strength, dose rate constant and radial dose function while anisotropy functions for the distance greater than 0.5 cm away from the source center are in good agreement with previous published studies. Obtained value from MC simulation for air kerma strength is $9.762\\times 10^{-8} \\textrm{UBq}^{-1}$and dose rate constant is $1.108\\pm 0.13\\%\\textrm{cGyh}^{-1} \\textrm{U}^{-1}$.
EleCa: a Monte Carlo code for the propagation of extragalactic photons at ultra-high energy
Settimo, Mariangela
2013-01-01
Ultra high energy photons play an important role as an independent probe of the photo-pion production mechanism by UHE cosmic rays. Their observation, or non-observation, may constrain astrophysical scenarios for the origin of UHECRs and help to understand the nature of the flux suppression observed by several experiments at energies above 10$^{19.5}$ eV. Whereas the interaction length of UHE photons above 10$^{17}$ eV is only of a few hundred kpc up to tenths of Mpc, photons can interact with the extragalactic background radiation leading to the development of electromagnetic cascades which affect the fluxes of photons observed at Earth. The interpretation of the current experimental results rely on the simulations of the UHE photon propagation. In this contribution, we present the novel Monte Carlo code "EleCa" to simulate the \\emph{Ele}ctromagnetic \\emph{Ca}scading initiated by high-energy photons and electrons. The distance within which we expect to observe UHE photons is discussed and the flux of GZK pho...
Directory of Open Access Journals (Sweden)
Lida Gholamkar
2016-09-01
Full Text Available Introduction One of the best methods in the diagnosis and control of breast cancer is mammography. The importance of mammography is directly related to its value in the detection of breast cancer in the early stages, which leads to a more effective treatment. The purpose of this article was to calculate the X-ray spectrum in a mammography system with Monte Carlo codes, including MCNPX and MCNP5. Materials and Methods The device, simulated using the MCNP code, was Planmed Nuance digital mammography device (Planmed Oy, Finland, equipped with an amorphous selenium detector. Different anode/filter materials, such as molybdenum-rhodium (Mo-Rh, molybdenum-molybdenum (Mo-Mo, tungsten-tin (W-Sn, tungsten-silver (W-Ag, tungsten-palladium (W-Pd, tungsten-aluminum (W-Al, tungsten-molybdenum (W-Mo, molybdenum-aluminum (Mo-Al, tungsten-rhodium (W-Rh, rhodium-aluminum (Rh-Al, and rhodium-rhodium (Rh-Rh, were simulated in this study. The voltage range of the X-ray tube was between 24 and 34 kV with a 2 kV interval. Results The charts of changing photon flux versus energy were plotted for different types of anode-filter combinations. The comparison with the findings reported by others indicated acceptable consistency. Also, the X-ray spectra, obtained from MCNP5 and MCNPX codes for W-Ag and W-Rh combinations, were compared. We compared the present results with the reported data of MCNP4C and IPEM report No. 78 for Mo-Mo, Mo-Rh, and W-Al combinations. Conclusion The MCNPX calculation outcomes showed acceptable results in a low-energy X-ray beam range (10-35 keV. The obtained simulated spectra for different anode/filter combinations were in good conformity with the finding of previous research.
Schiavi, A.; Senzacqua, M.; Pioli, S.; Mairani, A.; Magro, G.; Molinelli, S.; Ciocca, M.; Battistoni, G.; Patera, V.
2017-09-01
Ion beam therapy is a rapidly growing technique for tumor radiation therapy. Ions allow for a high dose deposition in the tumor region, while sparing the surrounding healthy tissue. For this reason, the highest possible accuracy in the calculation of dose and its spatial distribution is required in treatment planning. On one hand, commonly used treatment planning software solutions adopt a simplified beam–body interaction model by remapping pre-calculated dose distributions into a 3D water-equivalent representation of the patient morphology. On the other hand, Monte Carlo (MC) simulations, which explicitly take into account all the details in the interaction of particles with human tissues, are considered to be the most reliable tool to address the complexity of mixed field irradiation in a heterogeneous environment. However, full MC calculations are not routinely used in clinical practice because they typically demand substantial computational resources. Therefore MC simulations are usually only used to check treatment plans for a restricted number of difficult cases. The advent of general-purpose programming GPU cards prompted the development of trimmed-down MC-based dose engines which can significantly reduce the time needed to recalculate a treatment plan with respect to standard MC codes in CPU hardware. In this work, we report on the development of fred, a new MC simulation platform for treatment planning in ion beam therapy. The code can transport particles through a 3D voxel grid using a class II MC algorithm. Both primary and secondary particles are tracked and their energy deposition is scored along the trajectory. Effective models for particle–medium interaction have been implemented, balancing accuracy in dose deposition with computational cost. Currently, the most refined module is the transport of proton beams in water: single pencil beam dose–depth distributions obtained with fred agree with those produced by standard MC codes within 1–2% of
Energy Technology Data Exchange (ETDEWEB)
Campioni, Guillaume; Mounier, Claude [Commissariat a l' Energie Atomique, CEA, 31-33, rue de la Federation, 75752 Paris cedex (France)
2006-07-01
The main goal of the thesis about studies of cold neutrons sources (CNS) in research reactors was to create a complete set of tools to design efficiently CNS. The work raises the problem to run accurate simulations of experimental devices inside reactor reflector valid for parametric studies. On one hand, deterministic codes have reasonable computation times but introduce problems for geometrical description. On the other hand, Monte Carlo codes give the possibility to compute on precise geometry, but need computation times so important that parametric studies are impossible. To decrease this computation time, several developments were made in the Monte Carlo code TRIPOLI-4.4. An uncoupling technique is used to isolate a study zone in the complete reactor geometry. By recording boundary conditions (incoming flux), further simulations can be launched for parametric studies with a computation time reduced by a factor 60 (case of the cold neutron source of the Orphee reactor). The short response time allows to lead parametric studies using Monte Carlo code. Moreover, using biasing methods, the flux can be recorded on the surface of neutrons guides entries (low solid angle) with a further gain of running time. Finally, the implementation of a coupling module between TRIPOLI- 4.4 and the Monte Carlo code McStas for research in condensed matter field gives the possibility to obtain fluxes after transmission through neutrons guides, thus to have the neutron flux received by samples studied by scientists of condensed matter. This set of developments, involving TRIPOLI-4.4 and McStas, represent a complete computation scheme for research reactors: from nuclear core, where neutrons are created, to the exit of neutrons guides, on samples of matter. This complete calculation scheme is tested against ILL4 measurements of flux in cold neutron guides. (authors)
Energy Technology Data Exchange (ETDEWEB)
Kurosu, Keita [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN 46202 (United States); Department of Radiation Oncology, Osaka University Graduate School of Medicine, Suita, Osaka 565-0871 (Japan); Department of Radiology, Osaka University Hospital, Suita, Osaka 565-0871 (Japan); Das, Indra J. [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN 46202 (United States); Moskvin, Vadim P. [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN 46202 (United States); Department of Radiation Oncology, St. Jude Children’s Research Hospital, Memphis, TN 38105 (United States)
2016-01-15
Spot scanning, owing to its superior dose-shaping capability, provides unsurpassed dose conformity, in particular for complex targets. However, the robustness of the delivered dose distribution and prescription has to be verified. Monte Carlo (MC) simulation has the potential to generate significant advantages for high-precise particle therapy, especially for medium containing inhomogeneities. However, the inherent choice of computational parameters in MC simulation codes of GATE, PHITS and FLUKA that is observed for uniform scanning proton beam needs to be evaluated. This means that the relationship between the effect of input parameters and the calculation results should be carefully scrutinized. The objective of this study was, therefore, to determine the optimal parameters for the spot scanning proton beam for both GATE and PHITS codes by using data from FLUKA simulation as a reference. The proton beam scanning system of the Indiana University Health Proton Therapy Center was modeled in FLUKA, and the geometry was subsequently and identically transferred to GATE and PHITS. Although the beam transport is managed by spot scanning system, the spot location is always set at the center of a water phantom of 600 × 600 × 300 mm{sup 3}, which is placed after the treatment nozzle. The percentage depth dose (PDD) is computed along the central axis using 0.5 × 0.5 × 0.5 mm{sup 3} voxels in the water phantom. The PDDs and the proton ranges obtained with several computational parameters are then compared to those of FLUKA, and optimal parameters are determined from the accuracy of the proton range, suppressed dose deviation, and computational time minimization. Our results indicate that the optimized parameters are different from those for uniform scanning, suggesting that the gold standard for setting computational parameters for any proton therapy application cannot be determined consistently since the impact of setting parameters depends on the proton irradiation
Lázaro, Ignacio; Ródenas, José; Marques, José G.; Gallardo, Sergio
2014-06-01
Materials in a nuclear reactor are activated by neutron irradiation. When they are withdrawn from the reactor and placed in some storage, the potential dose received by workers in the surrounding area must be taken into account. In previous papers, activation of control rods in a NPP with BWR and dose rates around the storage pool have been estimated using the MCNP5 code based on the Monte Carlo method. Models were validated comparing simulation results with experimental measurements. As the activation is mostly produced in stainless steel components of control rods the activation model can be also validated by means of experimental measurements on a stainless steel sample after being irradiated in a reactor. This has been done in the Portuguese Research Reactor at Instituto Tecnológico e Nuclear. The neutron activation has been calculated by two different methods, Monte Carlo and CINDER'90, and results have been compared. After irradiation, dose rates at the water surface of the reactor pool were measured, with the irradiated stainless steel sample submerged at different positions under water. Experimental measurements have been compared with simulation results using Monte Carlo. The comparison shows a good agreement confirming the validation of models.
A study of the earth radiation budget using a 3D Monte-Carlo radiative transer code
Okata, M.; Nakajima, T.; Sato, Y.; Inoue, T.; Donovan, D. P.
2013-12-01
The purpose of this study is to evaluate the earth's radiation budget when data are available from satellite-borne active sensors, i.e. cloud profiling radar (CPR) and lidar, and a multi-spectral imager (MSI) in the project of the Earth Explorer/EarthCARE mission. For this purpose, we first developed forward and backward 3D Monte Carlo radiative transfer codes that can treat a broadband solar flux calculation including thermal infrared emission calculation by k-distribution parameters of Sekiguchi and Nakajima (2008). In order to construct the 3D cloud field, we tried the following three methods: 1) stochastic cloud generated by randomized optical thickness each layer distribution and regularly-distributed tilted clouds, 2) numerical simulations by a non-hydrostatic model with bin cloud microphysics model and 3) Minimum cloud Information Deviation Profiling Method (MIDPM) as explained later. As for the method-2 (numerical modeling method), we employed numerical simulation results of Californian summer stratus clouds simulated by a non-hydrostatic atmospheric model with a bin-type cloud microphysics model based on the JMA NHM model (Iguchi et al., 2008; Sato et al., 2009, 2012) with horizontal (vertical) grid spacing of 100m (20m) and 300m (20m) in a domain of 30km (x), 30km (y), 1.5km (z) and with a horizontally periodic lateral boundary condition. Two different cell systems were simulated depending on the cloud condensation nuclei (CCN) concentration. In the case of horizontal resolution of 100m, regionally averaged cloud optical thickness, , and standard deviation of COT, were 3.0 and 4.3 for pristine case and 8.5 and 7.4 for polluted case, respectively. In the MIDPM method, we first construct a library of pair of observed vertical profiles from active sensors and collocated imager products at the nadir footprint, i.e. spectral imager radiances, cloud optical thickness (COT), effective particle radius (RE) and cloud top temperature (Tc). We then select a best
Absorbed dose estimations of 131I for critical organs using the GEANT4 Monte Carlo simulation code
Institute of Scientific and Technical Information of China (English)
Ziaur Rahman; Shakeel ur Rehman; Waheed Arshed; Nasir M Mirza; Abdul Rashid; Jahan Zeb
2012-01-01
The aim of this study is to compare the absorbed doses of critical organs of 131I using the MIRD (Medical Internal Radiation Dose) with the corresponding predictions made by GEANT4 simulations.S-values (mean absorbed dose rate per unit activity) and energy deposition per decay for critical organs of 131I for various ages,using standard cylindrical phantom comprising water and ICRP soft-tissue material,have also been estimated.In this study the effect of volume reduction of thyroid,during radiation therapy,on the calculation of absorbed dose is also being estimated using GEANT4.Photon specific energy deposition in the other organs of the neck,due to 131I decay in the thyroid organ,has also been estimated.The maximum relative difference of MIRD with the GEANT4 simulated results is 5.64％ for an adult's critical organs of 131I.Excellent agreement was found between the results of water and ICRP soft tissue using the cylindrical model.S-values are tabulated for critical organs of 131I,using 1,5,10,15 and 18 years (adults) individuals.S-values for a cylindrical thyroid of different sizes,having 3.07％ relative differences of GEANT4 with Siegel & Stabin results.Comparison of the experimentally measured values at 0.5 and 1 m away from neck of the ionization chamber with GEANT4 based Monte Carlo simulations results show good agreement.This study shows that GEANT4 code is an important tool for the internal dosimetry calculations.
Energy Technology Data Exchange (ETDEWEB)
Abramov, B. M. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Alekseev, P. N. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Borodin, Yu. A. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Bulychjov, S. A. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Dukhovskoy, I. A. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Krutenkova, A. P. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Martemianov, M. A. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Matsyuk, M. A. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Turdakina, E. N. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Khanov, A. I. [Inst. of Theoretical and Experimental Physics (ITEP), Moscow (Russian Federation); Mashnik, Stepan Georgievich [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2015-02-03
Momentum spectra of hydrogen isotopes have been measured at 3.5° from ^{12}C fragmentation on a Be target. Momentum spectra cover both the region of fragmentation maximum and the cumulative region. Differential cross sections span five orders of magnitude. The data are compared to predictions of four Monte Carlo codes: QMD, LAQGSM, BC, and INCL++. There are large differences between the data and predictions of some models in the high momentum region. The INCL++ code gives the best and almost perfect description of the data.
TARTNP: a coupled neutron--photon Monte Carlo transport code. [10-/sup 9/ to 20 MeV; in LLL FORTRAN
Energy Technology Data Exchange (ETDEWEB)
Plechaty, E.F.; Kimlinger, J.R.
1976-07-04
A Monte Carlo code was written that calculates the transport of neutrons, photons, and neutron-induced photons. The cross sections of these particles are derived from TARTNP's data base, the Evaluated Nuclear Data Library. The energy range of the neutron data in the Library is 10/sup -9/ MeV to 20 MeV; the photon energy range is 1 keV to 20 MeV. One of the chief advantages of the code is its flexibility: it allows up to 17 different kinds of output to be evaluated in the same problem.
Abramov, B M; Borodin, Yu A; Bulychjov, S A; Dukhovskoy, I A; Krutenkova, A P; Kulikov, V V; Martemianov, M A; Matsyuk, M A; Turdakina, E N; Khanov, A I; Mashnik, S G
2015-01-01
Momentum spectra of hydrogen isotopes have been measured at 3.5 deg from C12 fragmentation on a Be target. Momentum spectra cover both the region of fragmentation maximum and the cumulative region. Differential cross sections span five orders of magnitude. The data are compared to predictions of four Monte Carlo codes: QMD, LAQGSM, BC, and INCL++. There are large differences between the data and predictions of some models in the high momentum region. The INCL++ code gives the best and almost perfect description of the data.
Energy Technology Data Exchange (ETDEWEB)
Ford, R.L.; Nelson, W.R.
1978-06-01
A code to simulate almost any electron--photon transport problem conceivable is described. The report begins with a lengthy historical introduction and a description of the shower generation process. Then the detailed physics of the shower processes and the methods used to simulate them are presented. Ideas of sampling theory, transport techniques, particle interactions in general, and programing details are discussed. Next, EGS calculations and various experiments and other Monte Carlo results are compared. The remainder of the report consists of user manuals for EGS, PEGS, and TESTSR codes; options, input specifications, and typical output are included. 38 figures, 12 tables. (RWR)
Tani, K.; Shinohara, K.; Oikawa, T.; Tsutsui, H.; McClements, K. G.; Akers, R. J.; Liu, Y. Q.; Suzuki, M.; Ide, S.; Kusama, Y.; Tsuji-Iio, S.
2016-11-01
As part of the verification and validation of a newly developed non-steady-state orbit-following Monte-Carlo code, application studies of time dependent neutron rates have been made for a specific shot in the Mega Amp Spherical Tokamak (MAST) using 3D fields representing vacuum resonant magnetic perturbations (RMPs) and toroidal field (TF) ripples. The time evolution of density, temperature and rotation rate in the application of the code to MAST are taken directly from experiment. The calculation results approximately agree with the experimental data. It is also found that a full orbit-following scheme is essential to reproduce the neutron rates in MAST.
Directory of Open Access Journals (Sweden)
Huseyin Ozan Tekin
2016-01-01
Full Text Available Gamma-ray measurements in various research fields require efficient detectors. One of these research fields is mass attenuation coefficients of different materials. Apart from experimental studies, the Monte Carlo (MC method has become one of the most popular tools in detector studies. An NaI(Tl detector has been modeled, and, for a validation study of the modeled NaI(Tl detector, the absolute efficiency of 3 × 3 inch cylindrical NaI(Tl detector has been calculated by using the general purpose Monte Carlo code MCNP-X (version 2.4.0 and compared with previous studies in literature in the range of 661–2620 keV. In the present work, the applicability of MCNP-X Monte Carlo code for mass attenuation of concrete sample material as building material at photon energies 59.5 keV, 80 keV, 356 keV, 661.6 keV, 1173.2 keV, and 1332.5 keV has been tested by using validated NaI(Tl detector. The mass attenuation coefficients of concrete sample have been calculated. The calculated results agreed well with experimental and some other theoretical results. The results specify that this process can be followed to determine the data on the attenuation of gamma-rays with other required energies in other materials or in new complex materials. It can be concluded that data from Monte Carlo is a strong tool not only for efficiency studies but also for mass attenuation coefficients calculations.
Energy Technology Data Exchange (ETDEWEB)
Both, J.P.; Mazzolo, A.; Peneliau, Y.; Petit, O.; Roesslinger, B
2003-07-01
This manual relates to Version 4.3 TRIPOLI-4 code. TRIPOLI-4 is a computer code simulating the transport of neutrons, photons, electrons and positrons. It can be used for radiation shielding calculations (long-distance propagation with flux attenuation in non-multiplying media) and neutronic calculations (fissile medium, criticality or sub-criticality basis). This makes it possible to calculate k{sub eff} (for criticality), flux, currents, reaction rates and multi-group cross-sections. TRIPOLI-4 is a three-dimensional code that uses the Monte-Carlo method. It allows for point-wise description in terms of energy of cross-sections and multi-group homogenized cross-sections and features two modes of geometrical representation: surface and combinatorial. The code uses cross-section libraries in ENDF/B format (such as JEF2-2, ENDF/B-VI and JENDL) for point-wise description cross-sections in APOTRIM format (from the APOLLO2 code) or a format specific to TRIPOLI-4 for multi-group description. (authors)
Energy Technology Data Exchange (ETDEWEB)
Rojas C, E.L.; Varon T, C.F.; Pedraza N, R. [ININ, 52750 La Marquesa, Estado de Mexico (Mexico)]. e-mail: elrc@nuclear.inin.mx
2007-07-01
The treatment of the breast cancer at early stages is of vital importance. For that, most of the investigations are dedicated to the early detection of the suffering and their treatment. As investigation consequence and clinical practice, in 2002 it was developed in U.S.A. an irradiation system of high dose rate known as Mammosite. In this work we carry out dose calculations for a simplified Mammosite system with the Monte Carlo Penelope simulation code and MCNPX, varying the concentration of the contrast material that it is used in the one. (Author)
Energy Technology Data Exchange (ETDEWEB)
Albuquerque, M.A.G.; David, M.G.; Almeida, C.E. de; Magalhaes, L.A.G., E-mail: malbuqueque@hotmail.com [Universidade do Estado do Rio de Janeiro (UERJ), Rio de Janeiro, RJ (Brazil). Lab. de Ciencias Radiologicas; Bernal, M. [Universidade Estadual de Campinas (UNICAMP), SP (Brazil); Braz, D. [Universidade Federal do Rio de Janeiro (UFRJ), RJ (Brazil)
2015-07-01
Breast cancer is the most common type of cancer among women. The main strategy to increase the long-term survival of patients with this disease is the early detection of the tumor, and mammography is the most appropriate method for this purpose. Despite the reduction of cancer deaths, there is a big concern about the damage caused by the ionizing radiation to the breast tissue. To evaluate these measures it was modeled a mammography equipment, and obtained the depth spectra using the Monte Carlo method - PENELOPE code. The average energies of the spectra in depth and the half value layer of the mammography output spectrum. (author)
Energy Technology Data Exchange (ETDEWEB)
Mazrou, Hakim, E-mail: mazrou_h@crna.d [Centre de Recherche Nucleaire d' Alger (CRNA), 02 Boulevard Frantz, Fanon, B.P. 399, Alger-RP 16000 (Algeria); Sidahmed, Tassadit [Centre de Recherche Nucleaire d' Alger (CRNA), 02 Boulevard Frantz, Fanon, B.P. 399, Alger-RP 16000 (Algeria); Allab, Malika [Faculte de Physique, Universite des Sciences et de la Technologie de Houari-Boumediene (USTHB), 16111, Alger (Algeria)
2010-10-15
An irradiation system has been acquired by the Nuclear Research Center of Algiers (CRNA) to provide neutron references for metrology and dosimetry purposes. It consists of an {sup 241}Am-Be radionuclide source of 185 GBq (5 Ci) activity inside a cylindrical steel-enveloped polyethylene container with radially positioned beam channel. Because of its composition, filled with hydrogenous material, which is not recommended by ISO standards, we expect large changes in the physical quantities of primary importance of the source compared to a free-field situation. Thus, the main goal of the present work is to fully characterize neutron field of such special delivered set-up. This was conducted by both extensive Monte-Carlo calculations and experimental measurements obtained by using BF{sub 3} and {sup 3}He based neutron area dosimeters. Effects of each component present in the bunker facility of the Algerian Secondary Standard Dosimetry Laboratory (SSDL) on the energy neutron spectrum have been investigated by simulating four irradiation configurations and comparison to the ISO spectrum has been performed. The ambient dose equivalent rate was determined based upon a correct estimate of the mean fluence to ambient dose equivalent conversion factors at different irradiations positions by means of a 3-D transport code MCNP5. Finally, according to practical requirements established for calibration purposes an optimal irradiation position has been suggested to the SSDL staff to perform, in appropriate manner, their routine calibrations.
Energy Technology Data Exchange (ETDEWEB)
Morris, R [Durham, NC (United States); Lakshmanan, M; Fong, G; Kapadia, A [Carl E Ravin Advanced Imaging Laboratories, Durham, NC (United States); Greenberg, J [Duke University, Durham, NC (United States)
2016-06-15
Purpose: Coherent scatter based imaging has shown improved contrast and molecular specificity over conventional digital mammography however the biological risks have not been quantified due to a lack of accurate information on absorbed dose. This study intends to characterize the dose distribution and average glandular dose from coded aperture coherent scatter spectral imaging of the breast. The dose deposited in the breast from this new diagnostic imaging modality has not yet been quantitatively evaluated. Here, various digitized anthropomorphic phantoms are tested in a Monte Carlo simulation to evaluate the absorbed dose distribution and average glandular dose using clinically feasible scan protocols. Methods: Geant4 Monte Carlo radiation transport simulation software is used to replicate the coded aperture coherent scatter spectral imaging system. Energy sensitive, photon counting detectors are used to characterize the x-ray beam spectra for various imaging protocols. This input spectra is cross-validated with the results from XSPECT, a commercially available application that yields x-ray tube specific spectra for the operating parameters employed. XSPECT is also used to determine the appropriate number of photons emitted per mAs of tube current at a given kVp tube potential. With the implementation of the XCAT digital anthropomorphic breast phantom library, a variety of breast sizes with differing anatomical structure are evaluated. Simulations were performed with and without compression of the breast for dose comparison. Results: Through the Monte Carlo evaluation of a diverse population of breast types imaged under real-world scan conditions, a clinically relevant average glandular dose for this new imaging modality is extrapolated. Conclusion: With access to the physical coherent scatter imaging system used in the simulation, the results of this Monte Carlo study may be used to directly influence the future development of the modality to keep breast dose to
Mosleh-Shirazi, Mohammad Amin; Zarrini-Monfared, Zinat; Karbasi, Sareh; Zamani, Ali
2014-01-01
Two-dimensional (2D) arrays of thick segmented scintillators are of interest as X-ray detectors for both 2D and 3D image-guided radiotherapy (IGRT). Their detection process involves ionizing radiation energy deposition followed by production and transport of optical photons. Only a very limited number of optical Monte Carlo simulation models exist, which has limited the number of modeling studies that have considered both stages of the detection process. We present ScintSim1, an in-house optical Monte Carlo simulation code for 2D arrays of scintillation crystals, developed in the MATLAB programming environment. The code was rewritten and revised based on an existing program for single-element detectors, with the additional capability to model 2D arrays of elements with configurable dimensions, material, etc., The code generates and follows each optical photon history through the detector element (and, in case of cross-talk, the surrounding ones) until it reaches a configurable receptor, or is attenuated. The new model was verified by testing against relevant theoretically known behaviors or quantities and the results of a validated single-element model. For both sets of comparisons, the discrepancies in the calculated quantities were all detector optimization.
Feng, Sheng; Fang, Ye; Tam, Ka-Ming; Thakur, Bhupender; Yun, Zhifeng; Tomko, Karen; Moreno, Juana; Ramanujam, Jagannathan; Jarrell, Mark
2013-03-01
The Edwards Anderson model is a typical example of random frustrated system. It has been a long standing problem in computational physics due to its long relaxation time. Some important properties of the low temperature spin glass phase are still poorly understood after decades of study. The recent advances of GPU computing provide a new opportunity to substantially improve the simulations. We developed an MPI-CUDA hybrid code with multi-spin coding for parallel tempering Monte Carlo simulation of Edwards Anderson model. Since the system size is relatively small, and a large number of parallel replicas and Monte Carlo moves are required, the problem suits well for modern GPUs with CUDA architecture. We use the code to perform an extensive simulation on the three-dimensional Edwards Anderson model with an external field. This work is funded by the NSF EPSCoR LA-SiGMA project under award number EPS-1003897. This work is partly done on the machines of Ohio Supercomputer Center.
Energy Technology Data Exchange (ETDEWEB)
Mazurier, J
1999-05-28
This thesis has been performed in the framework of national reference setting-up for absorbed dose in water and high energy photon beam provided with the SATURNE-43 medical accelerator of the BNM-LPRI (acronym for National Bureau of Metrology and Primary standard laboratory of ionising radiation). The aim of this work has been to develop and validate different user codes, based on PENELOPE Monte Carlo code system, to determine the photon beam characteristics and calculate the correction factors of reference dosimeters such as Fricke dosimeters and graphite calorimeter. In the first step, the developed user codes have permitted the influence study of different components constituting the irradiation head. Variance reduction techniques have been used to reduce the calculation time. The phase space has been calculated for 6, 12 and 25 MV at the output surface level of the accelerator head, then used for calculating energy spectra and dose distributions in the reference water phantom. Results obtained have been compared with experimental measurements. The second step has been devoted to develop an user code allowing calculation correction factors associated with both BNM-LPRI's graphite and Fricke dosimeters thanks to a correlated sampling method starting with energy spectra obtained in the first step. Then the calculated correction factors have been compared with experimental and calculated results obtained with the Monte Carlo EGS4 code system. The good agreement, between experimental and calculated results, leads to validate simulations performed with the PENELOPE code system. (author)
Leclaire, N.; Cochet, B.; Le Dauphin, F. X.; Haeck, W.; Jacquet, O.
2014-06-01
The present paper aims at providing experimental validation for the use of the MORET 5 code for advanced concepts of reactor involving thorium and heavy water. It therefore constitutes an opportunity to test and improve the thermal-scattering data of heavy water and also to test the recent implementation of probability tables in the MORET 5 code.
Energy Technology Data Exchange (ETDEWEB)
Champion, C. [Univ Metz, Lab Phys Mol et Collis, Inst Phys, F-57078 Metz 3 (France); Zanotti-Fregonara, P. [Commissariat Energie Atom, DSV, I2BM, SHFJ, LIME, Orsay (France); Hindie, E [Hop St Louis, AP-HP, Paris (France); Hindie, E. [Imagerie Mol Diagnost et Ciblage Therapeut, Ecole Doctorale B2T, IUH, Paris, Univ Paris 07 (France)
2008-07-01
Monte Carlo simulation can be particularly suitable for modeling the microscopic distribution of energy received by normal tissues or cancer cells and for evaluating the relative merits of different radiopharmaceuticals. We used a new code, CELLDOSE, to assess electron dose for isolated spheres with radii varying from 2,500 {mu}m down to 0.05 {mu}m, in which {sup 131}I is homogeneously distributed. Methods: All electron emissions of {sup 131}I were considered,including the whole {beta}{sup -} {sup 131}I spectrum, 108 internal conversion electrons, and 21 Auger electrons. The Monte Carlo track-structure code used follows all electrons down to an energy threshold E-cutoff 7.4 eV. Results: Calculated S values were in good agreement with published analytic methods, lying in between reported results for all experimental points. Our S values were also close to other published data using a Monte Carlo code. Contrary to the latter published results, our results show that dose distribution inside spheres is not homogeneous, with the dose at the outmost layer being approximately half that at the center. The fraction of electron energy retained within the spheres decreased with decreasing radius (r): 87.1 % for r 2,500 {mu}m, 8.73% for r 50 {mu}m, and 1.18% for r 5 {mu}m. Thus, a radioiodine concentration that delivers a dose of 100 Gy to a micro-metastasis of 2,500 {mu}m radius would deliver 10 Gy in a cluster of 50 {mu}m and only 1.4 Gy in an isolated cell. The specific contribution from Auger electrons varied from 0.25% for the largest sphere up to 76.8% for the smallest sphere. Conclusion: The dose to a tumor cell will depend on its position in a metastasis. For the treatment of very small metastases, {sup 131}I may not be the isotope of choice. When trying to kill isolated cells or a small cluster of cells with {sup 131}I, it is important to get the iodine as close as possible to the nucleus to get the enhancement factor from Auger electrons. The Monte Carlo code
Energy Technology Data Exchange (ETDEWEB)
Franke, Brian Claude; Kensek, Ronald Patrick; Laub, Thomas William
2005-09-01
ITS is a powerful and user-friendly software package permitting state-of-the-art Monte Carlo solution of linear time-independent coupled electron/photon radiation transport problems, with or without the presence of macroscopic electric and magnetic fields of arbitrary spatial dependence. Our goal has been to simultaneously maximize operational simplicity and physical accuracy. Through a set of preprocessor directives, the user selects one of the many ITS codes. The ease with which the makefile system is applied combines with an input scheme based on order-independent descriptive keywords that makes maximum use of defaults and internal error checking to provide experimentalists and theorists alike with a method for the routine but rigorous solution of sophisticated radiation transport problems. Physical rigor is provided by employing accurate cross sections, sampling distributions, and physical models for describing the production and transport of the electron/photon cascade from 1.0 GeV down to 1.0 keV. The availability of source code permits the more sophisticated user to tailor the codes to specific applications and to extend the capabilities of the codes to more complex applications. Version 5.0, the latest version of ITS, contains (1) improvements to the ITS 3.0 continuous-energy codes, (2) multigroup codes with adjoint transport capabilities, (3) parallel implementations of all ITS codes, (4) a general purpose geometry engine for linking with CAD or other geometry formats, and (5) the Cholla facet geometry library. Moreover, the general user friendliness of the software has been enhanced through increased internal error checking and improved code portability.
Energy Technology Data Exchange (ETDEWEB)
Vilches, M. [Servicio de Fisica y Proteccion Radiologica, Hospital Regional Universitario ' Virgen de las Nieves' , Avda. de las Fuerzas Armadas, 2, E-18014 Granada (Spain)], E-mail: mvilches@ugr.es; Garcia-Pareja, S. [Servicio de Radiofisica Hospitalaria, Hospital Regional Universitario ' Carlos Haya' , Avda. Carlos Haya, s/n, E-29010 Malaga (Spain); Guerrero, R. [Servicio de Radiofisica, Hospital Universitario ' San Cecilio' , Avda. Dr. Oloriz, 16, E-18012 Granada (Spain); Anguiano, M.; Lallena, A.M. [Departamento de Fisica Atomica, Molecular y Nuclear, Universidad de Granada, E-18071 Granada (Spain)
2007-09-21
When a therapeutic electron linear accelerator is simulated using a Monte Carlo (MC) code, the tuning of the initial spectra and the renormalization of dose (e.g., to maximum axial dose) constitute a common practice. As a result, very similar depth dose curves are obtained for different MC codes. However, if renormalization is turned off, the results obtained with the various codes disagree noticeably. The aim of this work is to investigate in detail the reasons of this disagreement. We have found that the observed differences are due to non-negligible differences in the angular scattering of the electron beam in very thin slabs of dense material (primary foil) and thick slabs of very low density material (air). To gain insight, the effects of the angular scattering models considered in various MC codes on the dose distribution in a water phantom are discussed using very simple geometrical configurations for the LINAC. The MC codes PENELOPE 2003, PENELOPE 2005, GEANT4, GEANT3, EGSnrc and MCNPX have been used.
White, M C
2000-01-01
The fundamental motivation for the research presented in this dissertation was the need to development a more accurate prediction method for characterization of mixed radiation fields around medical electron accelerators (MEAs). Specifically, a model is developed for simulation of neutron and other particle production from photonuclear reactions and incorporated in the Monte Carlo N-Particle (MCNP) radiation transport code. This extension of the capability within the MCNP code provides for the more accurate assessment of the mixed radiation fields. The Nuclear Theory and Applications group of the Los Alamos National Laboratory has recently provided first-of-a-kind evaluated photonuclear data for a select group of isotopes. These data provide the reaction probabilities as functions of incident photon energy with angular and energy distribution information for all reaction products. The availability of these data is the cornerstone of the new methodology for state-of-the-art mutually coupled photon-neutron tran...
Energy Technology Data Exchange (ETDEWEB)
Kramer, R.; Khoury, H. J. [Pernambuco Univ. (UFPE), Recife, PE (Brazil). Dept. de Energia Nuclear; Yoriyaz, H. [Instituto de Pesquisas Energeticas e Nucleares (IPEN-CNEN/SP), Sao Paulo, SP (Brazil); Lima, F.R.A. [Centro Regional de Ciencias Nucleares (CRCN-NE/CNEN), Recife, PE (Brazil); Faculdade Boa Viagem (FBV), Recife, PE (Brazil)]. E-mail: rkramer@uol.com.br
2004-07-01
Dose coefficients for intakes of radionuclides published by the International Commission on Radiological Protection (ICRP) are based on specific absorbed fractions (SAFs), which have been calculated in the mathematical MIRD phantoms. The replacement of the MIRD phantoms by voxel phantoms proposed by the ICRP raises the question about the changes to be expected for the SAFs, and consequently also for the dose coefficients. In order to investigate the dosimetric consequences of this replacement, SAFs have been calculated in the recently introduced MAX (Male Adult voXel) and FAXht (Female Adult voXel) head + trunk phantoms for photon energies between 10 keV and 4 MeV. For this purpose the phantoms have been connected to the EGS4 as well as to the MCNP4 code, which at present are probably the most used general-purpose Monte Carlo codes. Thereby it was possible to assess the impact on the SAFs, if different radiation transport methods are used. The mathematical MIRD phantoms have also been connected to the EGS4 code, and their elemental compositions of body tissues were replaced by those used in the voxel phantoms. In this manner it was possible to compare the SAFs of the MIRD phantoms on the one hand and the MAX and FAX phantom on the other hand as a function of the geometrical anatomy only, i.e. the volume, the shape and the position of organs at risk. (author)
Directory of Open Access Journals (Sweden)
Malouch Fadhel
2016-01-01
Full Text Available An irradiation program DV50 was carried out from 2002 to 2006 in the OSIRIS material testing reactor (CEA-Saclay center to assess the pressure vessel steel toughness curve for a fast neutron fluence (E > 1 MeV equivalent to a French 900-MWe PWR lifetime of 50 years. This program allowed the irradiation of 120 specimens out of vessel steel, subdivided in two successive irradiations DV50 n∘1 and DV50 n∘2. To measure the fast neutron fluence (E > 1 MeV received by specimens after each irradiation, sample holders were equipped with activation foils that were withdrawn at the end of irradiation for activity counting and processing. The fast effective cross-sections used in the dosimeter processing were determined with a specific calculation scheme based on the Monte-Carlo code TRIPOLI-3 (and the nuclear data ENDF/B-VI and IRDF-90. In order to put vessel-steel experiments at the same standard, a new dosimetric interpretation of the DV50 experiment has been performed by using the Monte-Carlo code TRIPOLI-4 and more recent nuclear data (JEFF3.1.1 and IRDF-2002. This paper presents a comparison of previous and recent calculations performed for the DV50 vessel-steel experiment to assess the impact on the dosimetric interpretation.
Wood, Kenneth; Whitney, Barbara A.; Robitaille, Thomas; Draine, Bruce T.
2008-12-01
We have modeled optical to far-infrared images, photometry, and spectroscopy of the object known as Gomez's Hamburger. We reproduce the images and spectrum with an edge-on disk of mass 0.3 M⊙ and radius 1600 AU, surrounding an A0 III star at a distance of 280 pc. Our mass estimate is in excellent agreement with recent CO observations. However, our distance determination is more than an order of magnitude smaller than previous analyses, which inaccurately interpreted the optical spectrum. To accurately model the infrared spectrum we have extended our Monte Carlo radiation transfer codes to include emission from polycyclic aromatic hydrocarbon (PAH) molecules and very small grains (VSG). We do this using precomputed PAH/VSG emissivity files for a wide range of values of the mean intensity of the exciting radiation field. When Monte Carlo energy packets are absorbed by PAHs/VSGs, we reprocess them to other wavelengths by sampling from the emissivity files, thus simulating the absorption and reemission process without reproducing lengthy computations of statistical equilibrium, excitation, and de-excitation in the complex many-level molecules. Using emissivity lookup tables in our Monte Carlo codes gives us the flexibility to use the latest grain physics calculations of PAH/VSG emissivity and opacity that are being continually updated in the light of higher resolution infrared spectra. We find our approach gives a good representation of the observed PAH spectrum from the disk of Gomez's Hamburger. Our models also indicate that the PAHs/VSGs in the disk have a larger scale height than larger radiative equilibrium grains, providing evidence for dust coagulation and settling to the midplane.
Gallardo, Sergio; Pozuelo, Fausto; Querol, Andrea; Ródenas, José; Verdú, Gumersindo
2014-06-01
An accurate knowledge of the photon spectra emitted by X-ray tubes in radiodiagnostic is essential to better estimate the imparted dose to patients and to improve the quality image obtained with these devices. In this work, it is proposed the use of a flat panel detector together with a PMMA wedge to estimate the actual X-ray spectrum using the Monte Carlo method and unfolding techniques. The MCNP5 code has been used to model different flat panels (based on indirect and direct methods to produce charge carriers from absorbed X-rays) and to obtain the dose curves and system response functions. Most of the actual flat panel devices use scintillator materials that present K-edge discontinuities in the mass energy-absorption coefficient, which strongly affect the response matrix. In this paper, the applicability of different flat panels for reconstructing X-ray spectra is studied. The effect of the mass energy-absorption coefficient of the scintillator material has been studied on the response matrix and consequently, in the reconstructed spectra. Different unfolding methods are tested to reconstruct the actual X-ray spectrum knowing the dose curve and the response function. It has been concluded that the regularization method MTSVD is appropriate to unfold X-ray spectra in all the scintillators studied.
Karimi Jashni, Hojatollah; Safigholi, Habib; Meigooni, Ali S
2014-10-23
Dose calculations in current brachytherapy treatment planning systems (TPS) are commonly based on TG-43U1 formalism. These TPS are obtained by superposition principle of single-source dosimetric parameters in liquid water, neglecting the effects of tissue heterogeneity. In this work, the sensitivity of the TG-43U1 based radial dose function (g(r)) of Miniature Electronic Brachytherapy X-ray Sources (MEBXS) to bone-heterogeneity was examined. To quantify the heterogeneity effects for g(r), a series of Monte Carlo (MC) based radiation transport simulations at the center of homogeneous and heterogeneous spherical phantoms were performed using the MCNP5 code. The ratio of the g(r) in the heterogeneius phantom to the uniform soft tisuue phantom as a function of the bone thickness was determined. These results indicated that for 40keV beam, the maximum ratios for thicknesses of 1cm and 2cm were 3.36 and 3.27, respectively. These values changed to 4.28 and 4.06, for 60keV beam, respectively. Introduction of 0.5cm or 1cm red marrow, into the interior of the cortical bone changed the maximum variations to, 3.54, and 3.57 for 40keV, and 4.28, and 4.25, for 60keV, respectively. Copyright © 2014 Elsevier Ltd. All rights reserved.
Mohammadyari, Parvin; Faghihi, Reza; Mosleh-Shirazi, Mohammad Amin; Lotfi, Mehrzad; Rahim Hematiyan, Mohammad; Koontz, Craig; Meigooni, Ali S.
2015-12-01
Compression is a technique to immobilize the target or improve the dose distribution within the treatment volume during different irradiation techniques such as AccuBoost® brachytherapy. However, there is no systematic method for determination of dose distribution for uncompressed tissue after irradiation under compression. In this study, the mechanical behavior of breast tissue between compressed and uncompressed states was investigated. With that, a novel method was developed to determine the dose distribution in uncompressed tissue after irradiation of compressed breast tissue. Dosimetry was performed using two different methods, namely, Monte Carlo simulations using the MCNP5 code and measurements using thermoluminescent dosimeters (TLD). The displacement of the breast elements was simulated using a finite element model and calculated using ABAQUS software. From these results, the 3D dose distribution in uncompressed tissue was determined. The geometry of the model was constructed from magnetic resonance images of six different women volunteers. The mechanical properties were modeled by using the Mooney-Rivlin hyperelastic material model. Experimental dosimetry was performed by placing the TLD chips into the polyvinyl alcohol breast equivalent phantom. The results determined that the nodal displacements, due to the gravitational force and the 60 Newton compression forces (with 43% contraction in the loading direction and 37% expansion in the orthogonal direction) were determined. Finally, a comparison of the experimental data and the simulated data showed agreement within 11.5% ± 5.9%.
Directory of Open Access Journals (Sweden)
Khan Hamda
2017-01-01
Full Text Available This paper carries out a Monte Carlo simulation of a landmine detection system, using the MCNP5 code, for the detection of concealed explosives such as trinitrotoluene and cyclonite. In portable field detectors, the signal strength of backscattered neutrons and gamma rays from thermal neutron activation is sensitive to a number of parameters such as the mass of explosive, depth of concealment, neutron moderation, background soil composition, soil porosity, soil moisture, multiple scattering in the background material, and configuration of the detection system. In this work, a detection system, with BF3 detectors for neutrons and sodium iodide scintillator for g-rays, is modeled to investigate the neutron signal-to-noise ratio and to obtain an empirical formula for the photon production rate Ri(n,γ= SfGfMf(d,m from radiative capture reactions in constituent nuclides of trinitrotoluene. This formula can be used for the efficient landmine detection of explosives in quantities as small as ~200 g of trinitrotoluene concealed at depths down to about 15 cm. The empirical formula can be embedded in a field programmable gate array on a field-portable explosives' sensor for efficient online detection.
Tanny, Sean
The advent of high-energy linear accelerators for dedicated medical use in the 1950's by Henry Kaplan and the Stanford University physics department began a revolution in radiation oncology. Today, linear accelerators are the standard of care for modern radiation therapy and can generate high-energy beams that can produce tens of Gy per minute at isocenter. This creates a need for a large amount of shielding material to properly protect members of the public and hospital staff. Standardized vault designs and guidance on shielding properties of various materials are provided by the National Council on Radiation Protection (NCRP) Report 151. However, physicists are seeking ways to minimize the footprint and volume of shielding material needed which leads to the use of non-standard vault configurations and less-studied materials, such as high-density concrete. The University of Toledo Dana Cancer Center has utilized both of these methods to minimize the cost and spatial footprint of the requisite radiation shielding. To ensure a safe work environment, computer simulations were performed to verify the attenuation properties and shielding workloads produced by a variety of situations where standard recommendations and guidance documents were insufficient. This project studies two areas of concern that are not addressed by NCRP 151, the radiation shielding workload for the vault door with a non-standard design, and the attenuation properties of high-density concrete for both photon and neutron radiation. Simulations have been performed using a Monte-Carlo code produced by the Los Alamos National Lab (LANL), Monte Carlo Neutrons, Photons 5 (MCNP5). Measurements have been performed using a shielding test port designed into the maze of the Varian Edge treatment vault.
Energy Technology Data Exchange (ETDEWEB)
Pacilio, M.; Lanconelli, N.; Lo Meo, S.; Betti, M.; Montani, L.; Torres Aroche, L. A.; Coca Perez, M. A. [Department of Medical Physics, Azienda Ospedaliera S. Camillo Forlanini, Piazza Forlanini 1, Rome 00151 (Italy); Department of Physics, Alma Mater Studiorum University of Bologna, Viale Berti-Pichat 6/2, Bologna 40127 (Italy); Department of Medical Physics, Azienda Ospedaliera S. Camillo Forlanini, Piazza Forlanini 1, Rome 00151 (Italy); Department of Medical Physics, Azienda Ospedaliera Sant' Andrea, Via di Grotarossa 1035, Rome 00189 (Italy); Department of Medical Physics, Center for Clinical Researches, Calle 34 North 4501, Havana 11300 (Cuba)
2009-05-15
Several updated Monte Carlo (MC) codes are available to perform calculations of voxel S values for radionuclide targeted therapy. The aim of this work is to analyze the differences in the calculations obtained by different MC codes and their impact on absorbed dose evaluations performed by voxel dosimetry. Voxel S values for monoenergetic sources (electrons and photons) and different radionuclides ({sup 90}Y, {sup 131}I, and {sup 188}Re) were calculated. Simulations were performed in soft tissue. Three general-purpose MC codes were employed for simulating radiation transport: MCNP4C, EGSnrc, and GEANT4. The data published by the MIRD Committee in Pamphlet No. 17, obtained with the EGS4 MC code, were also included in the comparisons. The impact of the differences (in terms of voxel S values) among the MC codes was also studied by convolution calculations of the absorbed dose in a volume of interest. For uniform activity distribution of a given radionuclide, dose calculations were performed on spherical and elliptical volumes, varying the mass from 1 to 500 g. For simulations with monochromatic sources, differences for self-irradiation voxel S values were mostly confined within 10% for both photons and electrons, but with electron energy less than 500 keV, the voxel S values referred to the first neighbor voxels showed large differences (up to 130%, with respect to EGSnrc) among the updated MC codes. For radionuclide simulations, noticeable differences arose in voxel S values, especially in the bremsstrahlung tails, or when a high contribution from electrons with energy of less than 500 keV is involved. In particular, for {sup 90}Y the updated codes showed a remarkable divergence in the bremsstrahlung region (up to about 90% in terms of voxel S values) with respect to the EGS4 code. Further, variations were observed up to about 30%, for small source-target voxel distances, when low-energy electrons cover an important part of the emission spectrum of the radionuclide
Energy Technology Data Exchange (ETDEWEB)
Schach von Wittenau, A.E.; Cox, L.J.; Bergstrom, P.M. Jr.; Hornstein, S.M. [Lawrence Livermore National Lab., CA (United States); Mohan, R.; Libby, B.; Wu, Q. [Medical Coll. of Virginia, Richmond, VA (United States); Lovelock, D.M.J. [Memorial Sloan-Kettering Cancer Center, New York, NY (United States)
1997-03-01
The goal of the PEREGRINE Monte Carlo Dose Calculation Project is to deliver a Monte Carlo package that is both accurate and sufficiently fast for routine clinical use. One of the operational requirements for photon-treatment plans is a fast, accurate method of describing the photon phase-space distribution at the surface of the patient. The open-field case is computationally the most tractable; we know, a priori, for a given machine and energy, the locations and compositions of the relevant accelerator components (i.e., target, primary collimator, flattening filter, and monitor chamber). Therefore, we can precalculate and store the expected photon distributions. For any open-field treatment plan, we then evaluate these existing photon phase-space distributions at the patient`s surface, and pass the obtained photons to the dose calculation routines within PEREGRINE. We neglect any effect of the intervening air column, including attenuation of the photons and production of contaminant electrons. In principle, for treatment plans requiring jaws, blocks, and wedges, we could precalculate and store photon phase-space distributions for various combinations of field sizes and wedges. This has the disadvantage that we would have to anticipate those combinations and that subsequently PEREGRINE would not be able to treat other plans. Therefore, PEREGRINE tracks photons through the patient-dependent beam modifiers. The geometric and physics methods used to do this are described here. 4 refs., 8 figs.
Energy Technology Data Exchange (ETDEWEB)
Kim, Kyung-O; Roh, Gyuhong; Lee, Byungchul [Korea Atomic Energy Research Institute, Daejeon (Korea, Republic of)
2015-10-15
As a result, there was a difference within about 300-400 pcm between keff values at each enrichment due to the difference of codes and nuclear data used in the evaluations. The FTC was changed to be less negative with the increase of uranium enrichment, and it followed the form of asymptotic curve. However, it is necessary to perform additional study for investigating what factor causes the differences more than two standard deviation (2σ) among the FTCs at partial enrichment region. The interaction probability of incident neutron with nuclear fuel is depended on the relative velocity between the neutron and the target nuclei. The Fuel Temperature Coefficient (FTC) is defined as the change of Doppler effect with respect to the change in fuel temperature without any other change such as moderator temperature, moderator density, etc. In this study, the FTCs for UO{sub 2} fuel were evaluated by using MCNP6.1 and KENO6 codes based on a Monte Carlo method. In addition, the latest neutron cross-sections (ENDF/B-VI and VII) were applied to analyze the effect of these data on the evaluation of FTC, and nuclear data used in MCNP calculations were generated from the makxsf code. An evaluation of the Doppler effect and FTC for UO{sub 2} fuel widely used in PWR was conducted using MCNP6.1 and KENO6 codes. The ENDF/B-VI and VII were also applied to analyze what effect these data has on those evaluations. All cross-sections needed for MCNP calculation were produced using makxsf code. The calculation models used in the evaluations were based on the typical PWR UO{sub 2} lattice.
Energy Technology Data Exchange (ETDEWEB)
Kurosu, Keita [Department of Medical Physics and Engineering, Osaka University Graduate School of Medicine, Suita, Osaka 565-0871 (Japan); Department of Radiation Oncology, Osaka University Graduate School of Medicine, Suita, Osaka 565-0871 (Japan); Takashina, Masaaki; Koizumi, Masahiko [Department of Medical Physics and Engineering, Osaka University Graduate School of Medicine, Suita, Osaka 565-0871 (Japan); Das, Indra J. [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN 46202 (United States); Moskvin, Vadim P., E-mail: vadim.p.moskvin@gmail.com [Department of Radiation Oncology, Indiana University School of Medicine, Indianapolis, IN 46202 (United States)
2014-10-01
Although three general-purpose Monte Carlo (MC) simulation tools: Geant4, FLUKA and PHITS have been used extensively, differences in calculation results have been reported. The major causes are the implementation of the physical model, preset value of the ionization potential or definition of the maximum step size. In order to achieve artifact free MC simulation, an optimized parameters list for each simulation system is required. Several authors have already proposed the optimized lists, but those studies were performed with a simple system such as only a water phantom. Since particle beams have a transport, interaction and electromagnetic processes during beam delivery, establishment of an optimized parameters-list for whole beam delivery system is therefore of major importance. The purpose of this study was to determine the optimized parameters list for GATE and PHITS using proton treatment nozzle computational model. The simulation was performed with the broad scanning proton beam. The influences of the customizing parameters on the percentage depth dose (PDD) profile and the proton range were investigated by comparison with the result of FLUKA, and then the optimal parameters were determined. The PDD profile and the proton range obtained from our optimized parameters list showed different characteristics from the results obtained with simple system. This led to the conclusion that the physical model, particle transport mechanics and different geometry-based descriptions need accurate customization in planning computational experiments for artifact-free MC simulation.
Wysocka-Rabin, A
2013-01-01
The introductory chapter of this monograph, which follows this Preface, provides an overview of radiotherapy and treatment planning. The main chapters that follow describe in detail three significant aspects of radiotherapy on which the author has focused her research efforts. Chapter 2 presents studies the author worked on at the German National Cancer Institute (DKFZ) in Heidelberg. These studies applied the Monte Carlo technique to investigate the feasibility of performing Intensity Modulated Radiotherapy (IMRT) by scanning with a narrow photon beam. This approach represents an alternative to techniques that generate beam modulation by absorption, such as MLC, individually-manufactured compensators, and special tomotherapy modulators. The technical realization of this concept required investigation of the influence of various design parameters on the final small photon beam. The photon beam to be scanned should have a diameter of approximately 5 mm at Source Surface Distance (SSD) distance, and the penumbr...
Stepanek, J; Laissue, J A; Lyubimova, N; Di Michiel, F; Slatkin, D N
2000-01-01
Microbeam radiation therapy (MRT) is a currently experimental method of radiotherapy which is mediated by an array of parallel microbeams of synchrotron-wiggler-generated X-rays. Suitably selected, nominally supralethal doses of X-rays delivered to parallel microslices of tumor-bearing tissues in rats can be either palliative or curative while causing little or no serious damage to contiguous normal tissues. Although the pathogenesis of MRT-mediated tumor regression is not understood, as in all radiotherapy such understanding will be based ultimately on our understanding of the relationships among the following three factors: (1) microdosimetry, (2) damage to normal tissues, and (3) therapeutic efficacy. Although physical microdosimetry is feasible, published information on MRT microdosimetry to date is computational. This report describes Monte Carlo-based computational MRT microdosimetry using photon and/or electron scattering and photoionization cross-section data in the 1 e V through 100 GeV range distrib...
Sarrut, David; Bardiès, Manuel; Boussion, Nicolas; Freud, Nicolas; Jan, Sébastien; Létang, Jean-Michel; Loudos, George; Maigne, Lydia; Marcatili, Sara; Mauxion, Thibault; Papadimitroulas, Panagiotis; Perrot, Yann; Pietrzyk, Uwe; Robert, Charlotte; Schaart, Dennis R; Visvikis, Dimitris; Buvat, Irène
2014-06-01
In this paper, the authors' review the applicability of the open-source GATE Monte Carlo simulation platform based on the GEANT4 toolkit for radiation therapy and dosimetry applications. The many applications of GATE for state-of-the-art radiotherapy simulations are described including external beam radiotherapy, brachytherapy, intraoperative radiotherapy, hadrontherapy, molecular radiotherapy, and in vivo dose monitoring. Investigations that have been performed using GEANT4 only are also mentioned to illustrate the potential of GATE. The very practical feature of GATE making it easy to model both a treatment and an imaging acquisition within the same framework is emphasized. The computational times associated with several applications are provided to illustrate the practical feasibility of the simulations using current computing facilities.
Energy Technology Data Exchange (ETDEWEB)
Sarrut, David, E-mail: david.sarrut@creatis.insa-lyon.fr [Université de Lyon, CREATIS, CNRS UMR5220, Inserm U1044, INSA-Lyon (France); Université Lyon 1 (France); Centre Léon Bérard (France); Bardiès, Manuel; Marcatili, Sara; Mauxion, Thibault [Inserm, UMR1037 CRCT, F-31000 Toulouse, France and Université Toulouse III-Paul Sabatier, UMR1037 CRCT, F-31000 Toulouse (France); Boussion, Nicolas [INSERM, UMR 1101, LaTIM, CHU Morvan, 29609 Brest (France); Freud, Nicolas; Létang, Jean-Michel [Université de Lyon, CREATIS, CNRS UMR5220, Inserm U1044, INSA-Lyon, Université Lyon 1, Centre Léon Bérard, 69008 Lyon (France); Jan, Sébastien [CEA/DSV/I2BM/SHFJ, Orsay 91401 (France); Loudos, George [Department of Medical Instruments Technology, Technological Educational Institute of Athens, Athens 12210 (Greece); Maigne, Lydia; Perrot, Yann [UMR 6533 CNRS/IN2P3, Université Blaise Pascal, 63171 Aubière (France); Papadimitroulas, Panagiotis [Department of Biomedical Engineering, Technological Educational Institute of Athens, 12210, Athens (Greece); Pietrzyk, Uwe [Institut für Neurowissenschaften und Medizin, Forschungszentrum Jülich GmbH, 52425 Jülich, Germany and Fachbereich für Mathematik und Naturwissenschaften, Bergische Universität Wuppertal, 42097 Wuppertal (Germany); Robert, Charlotte [IMNC, UMR 8165 CNRS, Universités Paris 7 et Paris 11, Orsay 91406 (France); and others
2014-06-15
In this paper, the authors' review the applicability of the open-source GATE Monte Carlo simulation platform based on the GEANT4 toolkit for radiation therapy and dosimetry applications. The many applications of GATE for state-of-the-art radiotherapy simulations are described including external beam radiotherapy, brachytherapy, intraoperative radiotherapy, hadrontherapy, molecular radiotherapy, and in vivo dose monitoring. Investigations that have been performed using GEANT4 only are also mentioned to illustrate the potential of GATE. The very practical feature of GATE making it easy to model both a treatment and an imaging acquisition within the same frameworkis emphasized. The computational times associated with several applications are provided to illustrate the practical feasibility of the simulations using current computing facilities.
Truchet, G.; Leconte, P.; Peneliau, Y.; Santamarina, A.; Malvagi, F.
2014-06-01
Pile-oscillation experiments are performed in the MINERVE reactor at the CEA Cadarache to improve nuclear data accuracy. In order to precisely calculate small reactivity variations (experiments, a reference calculation need to be achieved. This calculation may be accomplished using the continuous-energy Monte Carlo code TRIPOLI-4® by using the eigenvalue difference method. This "direct" method has shown limitations in the evaluation of very small reactivity effects because it needs to reach a very small variance associated to the reactivity in both states. To answer this problem, it has been decided to implement the exact perturbation theory in TRIPOLI-4® and, consequently, to calculate a continuous-energy adjoint flux. The Iterated Fission Probability (IFP) method was chosen because it has shown great results in some other Monte Carlo codes. The IFP method uses a forward calculation to compute the adjoint flux, and consequently, it does not rely on complex code modifications but on the physical definition of the adjoint flux as a phase-space neutron importance. In the first part of this paper, the IFP method implemented in TRIPOLI-4® is described. To illustrate the effciency of the method, several adjoint fluxes are calculated and compared with their equivalent obtained by the deterministic code APOLLO-2. The new implementation can calculate angular adjoint flux. In the second part, a procedure to carry out an exact perturbation calculation is described. A single cell benchmark has been used to test the accuracy of the method, compared with the "direct" estimation of the perturbation. Once again the method based on the IFP shows good agreement for a calculation time far more inferior to the "direct" method. The main advantage of the method is that the relative accuracy of the reactivity variation does not depend on the magnitude of the variation itself, which allows us to calculate very small reactivity perturbations with high precision. Other applications of
Iwamoto, Yosuke; Ogawa, Tatsuhiko
2017-04-01
Because primary knock-on atoms (PKAs) create point defects and clusters in materials that are irradiated with neutrons, it is important to validate the calculations of recoil cross section spectra that are used to estimate radiation damage in materials. Here, the recoil cross section spectra of fission- and fusion-relevant materials were calculated using the Event Generator Mode (EGM) of the Particle and Heavy Ion Transport code System (PHITS) and also using the data processing code NJOY2012 with the nuclear data libraries TENDL2015, ENDF/BVII.1, and JEFF3.2. The heating number, which is the integral of the recoil cross section spectra, was also calculated using PHITS-EGM and compared with data extracted from the ACE files of TENDL2015, ENDF/BVII.1, and JENDL4.0. In general, only a small difference was found between the PKA spectra of PHITS + TENDL2015 and NJOY + TENDL2015. From analyzing the recoil cross section spectra extracted from the nuclear data libraries using NJOY2012, we found that the recoil cross section spectra were incorrect for 72Ge, 75As, 89Y, and 109Ag in the ENDF/B-VII.1 library, and for 90Zr and 55Mn in the JEFF3.2 library. From analyzing the heating number, we found that the data extracted from the ACE file of TENDL2015 for all nuclides were problematic in the neutron capture region because of incorrect data regarding the emitted gamma energy. However, PHITS + TENDL2015 can calculate PKA spectra and heating numbers correctly.
Energy Technology Data Exchange (ETDEWEB)
Iwamoto, Yosuke, E-mail: iwamoto.yosuke@jaea.go.jp; Ogawa, Tatsuhiko
2017-04-01
Because primary knock-on atoms (PKAs) create point defects and clusters in materials that are irradiated with neutrons, it is important to validate the calculations of recoil cross section spectra that are used to estimate radiation damage in materials. Here, the recoil cross section spectra of fission- and fusion-relevant materials were calculated using the Event Generator Mode (EGM) of the Particle and Heavy Ion Transport code System (PHITS) and also using the data processing code NJOY2012 with the nuclear data libraries TENDL2015, ENDF/BVII.1, and JEFF3.2. The heating number, which is the integral of the recoil cross section spectra, was also calculated using PHITS-EGM and compared with data extracted from the ACE files of TENDL2015, ENDF/BVII.1, and JENDL4.0. In general, only a small difference was found between the PKA spectra of PHITS + TENDL2015 and NJOY + TENDL2015. From analyzing the recoil cross section spectra extracted from the nuclear data libraries using NJOY2012, we found that the recoil cross section spectra were incorrect for {sup 72}Ge, {sup 75}As, {sup 89}Y, and {sup 109}Ag in the ENDF/B-VII.1 library, and for {sup 90}Zr and {sup 55}Mn in the JEFF3.2 library. From analyzing the heating number, we found that the data extracted from the ACE file of TENDL2015 for all nuclides were problematic in the neutron capture region because of incorrect data regarding the emitted gamma energy. However, PHITS + TENDL2015 can calculate PKA spectra and heating numbers correctly.
Energy Technology Data Exchange (ETDEWEB)
Jo, Yu Gwon; Yun, Sung Hwan; Cho, Nam Zin [KAIST, Daejeon (Korea, Republic of)
2011-10-15
Since the Monte Carlo method overcomes limitations in multi-group approximation and geometry description, it is gaining increasing use in reactor physics problems. Recently, a new leakage-corrected method was suggested by the authors, in which the critical spectrum is obtained by albedo-based leakage correction in the Monte Carlo method. In this paper, the critical spectrum based on the albedo-based leakage-corrected method will be compared with the critical spectrum by conventional B1 method (with condensed cross sections) and reference critical whole-core assembly spectrum. These two methods are implemented in our local MCNP5 (version 1.50)
Villoing, Daphnée; Marcatili, Sara; Garcia, Marie-Paule; Bardiès, Manuel
2017-03-01
The purpose of this work was to validate GATE-based clinical scale absorbed dose calculations in nuclear medicine dosimetry. GATE (version 6.2) and MCNPX (version 2.7.a) were used to derive dosimetric parameters (absorbed fractions, specific absorbed fractions and S-values) for the reference female computational model proposed by the International Commission on Radiological Protection in ICRP report 110. Monoenergetic photons and electrons (from 50 keV to 2 MeV) and four isotopes currently used in nuclear medicine (fluorine-18, lutetium-177, iodine-131 and yttrium-90) were investigated. Absorbed fractions, specific absorbed fractions and S-values were generated with GATE and MCNPX for 12 regions of interest in the ICRP 110 female computational model, thereby leading to 144 source/target pair configurations. Relative differences between GATE and MCNPX obtained in specific configurations (self-irradiation or cross-irradiation) are presented. Relative differences in absorbed fractions, specific absorbed fractions or S-values are below 10%, and in most cases less than 5%. Dosimetric results generated with GATE for the 12 volumes of interest are available as supplemental data. GATE can be safely used for radiopharmaceutical dosimetry at the clinical scale. This makes GATE a viable option for Monte Carlo modelling of both imaging and absorbed dose in nuclear medicine.
Lapins, Janis; Guilliard, Nicole; Bernnat, Wolfgang; Buck, Arnulf
2017-09-01
During heavy ion irradiation therapy the patient has to be located exactly at the right position to make sure that the Bragg peak occurs in the tumour. The patient has to be moved in the range of millimetres to scan the ill tissue. For that reason a special table was developed which allows exact positioning. The electronic control can be located outside the surgery. But that has some disadvantage for the construction. To keep the system compact it would be much more comfortable to put the electronic control inside the surgery. As a lot of high energetic secondary particles are produced during the therapy causing a high dose in the room it is important to find positions with low dose rates. Therefore, investigations are needed where the electronic devices should be located to obtain a minimum of radiation, help to prevent the failure of sensitive devices. The dose rate was calculated for carbon ions with different initial energy and protons over the entire therapy room with Monte Carlo particle tracking using MCNP6. The types of secondary particles were identified and the dose rate for a thin silicon layer and an electronic mixture material was determined. In addition, the shielding effect of several selected material layers was calculated using MCNP6.
Energy Technology Data Exchange (ETDEWEB)
Ignatova, V.A. E-mail: velislav@uia.ua.ac.be; Chakarov, I.R.; Katardjiev, I.V
2003-04-01
The redistribution of the elements as a result of atomic relocations produced by the ions and the recoils due to the ballistic and transport processes is investigated by making use of a dynamic Monte Carlo code. Phenomena, such as radiation-enhanced diffusion (RED) and bombardment-induced segregation (BIS) triggered by the ion bombardment may also contribute to the migration of atoms within the target. In order to include both RED and BIS in the code, we suggest an approach which is considered as an extension of the binary collision approximation, i.e. it takes place 'simultaneously' with the cascade and acts as a correction to the particle redistribution for low energies. Both RED and BIS models are based on the common approach to treat the transport processes as a result of a random migration of point defects (vacancies and interstitials) according to a probability given by a pre-defined Gaussian. The models are tested and the influence of the diffusion and segregation is illustrated in the cases of 12 keV {sup 121}Sb{sup +} implantation at low fluence in SiO{sub 2}/Si substrate and of self-sputtering of Ga{sup +} ions during profiling of SiO{sub 2}/Si interfaces.
Ignatova, V A; Katardjiev, I V
2003-01-01
The redistribution of the elements as a result of atomic relocations produced by the ions and the recoils due to the ballistic and transport processes is investigated by making use of a dynamic Monte Carlo code. Phenomena, such as radiation-enhanced diffusion (RED) and bombardment-induced segregation (BIS) triggered by the ion bombardment may also contribute to the migration of atoms within the target. In order to include both RED and BIS in the code, we suggest an approach which is considered as an extension of the binary collision approximation, i.e. it takes place 'simultaneously' with the cascade and acts as a correction to the particle redistribution for low energies. Both RED and BIS models are based on the common approach to treat the transport processes as a result of a random migration of point defects (vacancies and interstitials) according to a probability given by a pre-defined Gaussian. The models are tested and the influence of the diffusion and segregation is illustrated in the cases of 12 keV ...
Icarus: A 2D direct simulation Monte Carlo (DSMC) code for parallel computers. User`s manual - V.3.0
Energy Technology Data Exchange (ETDEWEB)
Bartel, T.; Plimpton, S.; Johannes, J.; Payne, J.
1996-10-01
Icarus is a 2D Direct Simulation Monte Carlo (DSMC) code which has been optimized for the parallel computing environment. The code is based on the DSMC method of Bird and models from free-molecular to continuum flowfields in either cartesian (x, y) or axisymmetric (z, r) coordinates. Computational particles, representing a given number of molecules or atoms, are tracked as they have collisions with other particles or surfaces. Multiple species, internal energy modes (rotation and vibration), chemistry, and ion transport are modelled. A new trace species methodology for collisions and chemistry is used to obtain statistics for small species concentrations. Gas phase chemistry is modelled using steric factors derived from Arrhenius reaction rates. Surface chemistry is modelled with surface reaction probabilities. The electron number density is either a fixed external generated field or determined using a local charge neutrality assumption. Ion chemistry is modelled with electron impact chemistry rates and charge exchange reactions. Coulomb collision cross-sections are used instead of Variable Hard Sphere values for ion-ion interactions. The electrostatic fields can either be externally input or internally generated using a Langmuir-Tonks model. The Icarus software package includes the grid generation, parallel processor decomposition, postprocessing, and restart software. The commercial graphics package, Tecplot, is used for graphics display. The majority of the software packages are written in standard Fortran.
Energy Technology Data Exchange (ETDEWEB)
Karalidi, Theodora; Apai, Dániel; Schneider, Glenn; Hanson, Jake R. [Steward Observatory, Department of Astronomy, University of Arizona, 933 N. Cherry Avenue, Tucson, AZ 85721 (United States); Pasachoff, Jay M., E-mail: tkaralidi@email.arizona.edu [Hopkins Observatory, Williams College, 33 Lab Campus Drive, Williamstown, MA 01267 (United States)
2015-11-20
Deducing the cloud cover and its temporal evolution from the observed planetary spectra and phase curves can give us major insight into the atmospheric dynamics. In this paper, we present Aeolus, a Markov chain Monte Carlo code that maps the structure of brown dwarf and other ultracool atmospheres. We validated Aeolus on a set of unique Jupiter Hubble Space Telescope (HST) light curves. Aeolus accurately retrieves the properties of the major features of the Jovian atmosphere, such as the Great Red Spot and a major 5 μm hot spot. Aeolus is the first mapping code validated on actual observations of a giant planet over a full rotational period. For this study, we applied Aeolus to J- and H-band HST light curves of 2MASS J21392676+0220226 and 2MASS J0136565+093347. Aeolus retrieves three spots at the top of the atmosphere (per observational wavelength) of these two brown dwarfs, with a surface coverage of 21% ± 3% and 20.3% ± 1.5%, respectively. The Jupiter HST light curves will be publicly available via ADS/VIZIR.
Directory of Open Access Journals (Sweden)
M. Abbasian Motlagh
2014-04-01
Full Text Available For their appropriate temporal resolution, scintillator detectors are used in the Alborz observatory. In this work, the behavior of the scintillation detectors for the passage of electrons with different energies and directions were studied using the simulation code GEANT4. Pulse shapes of scintillation light, and such characteristics as the total number of photons, the rise time and the falling time for the optical pulses were computed for the passage of electrons with energies of 10, 100 and 1000 MeV. Variations of the characteristics of optical pulse of scintillation with incident angle and the location of electrons were also investigated
Energy Technology Data Exchange (ETDEWEB)
White, Morgan C. [Univ. of Florida, Gainesville, FL (United States)
2000-07-01
The fundamental motivation for the research presented in this dissertation was the need to development a more accurate prediction method for characterization of mixed radiation fields around medical electron accelerators (MEAs). Specifically, a model is developed for simulation of neutron and other particle production from photonuclear reactions and incorporated in the Monte Carlo N-Particle (MCNP) radiation transport code. This extension of the capability within the MCNP code provides for the more accurate assessment of the mixed radiation fields. The Nuclear Theory and Applications group of the Los Alamos National Laboratory has recently provided first-of-a-kind evaluated photonuclear data for a select group of isotopes. These data provide the reaction probabilities as functions of incident photon energy with angular and energy distribution information for all reaction products. The availability of these data is the cornerstone of the new methodology for state-of-the-art mutually coupled photon-neutron transport simulations. The dissertation includes details of the model development and implementation necessary to use the new photonuclear data within MCNP simulations. A new data format has been developed to include tabular photonuclear data. Data are processed from the Evaluated Nuclear Data Format (ENDF) to the new class ''u'' A Compact ENDF (ACE) format using a standalone processing code. MCNP modifications have been completed to enable Monte Carlo sampling of photonuclear reactions. Note that both neutron and gamma production are included in the present model. The new capability has been subjected to extensive verification and validation (V&V) testing. Verification testing has established the expected basic functionality. Two validation projects were undertaken. First, comparisons were made to benchmark data from literature. These calculations demonstrate the accuracy of the new data and transport routines to better than 25 percent. Second
Energy Technology Data Exchange (ETDEWEB)
Amharrak, H.; Reynard-Carette, C.; Carette, M. [Aix Marseille Universite, CNRS, Universite de Toulon, IM2NP UMR 7334, 13397, Marseille (France); Lemaire, M.; Vaglio-Gaudard, C. [CEA, DEN, DER, SPRC, LPN, Cadarache, F-13108 Saint Paul Lez Durance (France); Fourmentel, D.; Lyoussi, A. [CEA, DEN, Departement d' Etudes des Reacteurs, Service de Physique Experimentale, Laboratoire Dosimetrie Capteurs Instrumentation, 13108 Saint-Paul-lez-Durance (France)
2015-07-01
calorimeter were carried out. A preliminary analysis shows that the numerical results overestimate the measurements by about 20 %. A new approach has been developed in order to estimate the nuclear heating by two methods (energy deposition or KERMA) by considering the whole complete geometry of the sensor. This new approach will contribute to the interpretation of the irradiation campaign and will be useful to improve the out-of-pile calibration procedure of the sensor and its thermal response during irradiations. The aim of this paper is to present simulations made by using MCNP5 Monte-Carlo transport code (using ENDF/B-VI nuclear data library) for the nuclear heating inside the different parts of the calorimeter (head, rod and base). Calculations into two steps will be realized. We will use as an input source in the model new spectra (neutrons, prompt-photons and delayed-photons) calculated with the Monte Carlo code TRIPOLI-4{sup R} inside different experimental channels (water) located into the OSIRIS periphery and used during the CARMEN-1P irradiation campaign. We will consider Neutrons- Photons-Electrons and Photons-Electrons modes. We will begin by a brief description of the differential-calorimeter device geometry. Then the MCNP5 model used for the calculations of nuclear heating inside the calorimeter elements will be introduced. The energy deposition due to the prompt-gamma, delayed-gamma and neutrons, the neutron-activation of the device will be considered. The different components of the nuclear heating inside the different parts of the calorimeter will be detailed. Moreover, a comparison between KERMA and nuclear energy deposition estimations will be given. Finally, a comparison between this total nuclear heating Calculation and Experiment in graphite sample will be determined. (authors)
Energy Technology Data Exchange (ETDEWEB)
Reis, Lucas P.; Santos, Adriano M.; Grynberg, Suely E., E-mail: lpr@cdtn.b [Centro de Desenvolvimento da Tecnologia Nuclear (CDTN/CNEN-MG), Belo Horizonte, MG (Brazil). Lab. de Dosimetria e Simulacao Computacional; Facure, Alessandro, E-mail: facure@cnen.gov.b [Comissao Nacional de Energia Nuclear (CNEN), Rio de Janeiro, RJ (Brazil)
2011-07-01
For prostate cancer treatments, there is an increasing interest in the permanent radioactive seeds implant technique. Currently, in Brazil, the seeds are imported at high prices, which prohibit their use in public hospitals. One of the seed models that have been developed at CDTN has a ceramic matrix as a radioisotope carrier and a radiographic marker; the seed is encapsulated with biocompatible polymer. In this work, Monte Carlo simulations were performed in order to assess the dose distributions generated by the prototype seed model. The obtained data was assessed as described in the TG-43U1 report by the AAPM. The dosimetric parameters dose rate constant, {Lambda}, radial dose function, g{sub L}(r), and anisotropy function, F(r,{theta}), were derived from simulations using the MCNP5 code. The function g(r) shows that the seed has a lower decrease in dose rate on its transverse axis when compared to the 6711 model (one of the most used seeds in permanent prostate implants). F(r,{theta}) shows that CDTN's seed anisotropy curves are smoother than the 6711 model curves for {theta}{<=}20 deg and 0.25{<=}r{<=}1 cm. As well, the {Lambda} value is 15% lower than the {Lambda} value of 6711. The results show that CDTN's seed model can deposit a more isotropic dose. Because of the model's characteristics, the seeds can be impregnated with iodine of lower specific activity which would help reducing costs. (author)
Energy Technology Data Exchange (ETDEWEB)
Caribe, Paulo Rauli Rafeson Vasconcelos, E-mail: raulycaribe@hotmail.com [Universidade Federal Rural de Pernambuco (UFRPE), Recife, PE (Brazil). Fac. de Fisica; Cassola, Vagner Ferreira; Kramer, Richard; Khoury, Helen Jamil [Universidade Federal de Pernambuco (UFPE), Recife, PE (Brazil). Departamento de Energia Nuclear
2013-07-01
The use of three-dimensional models described by polygonal meshes in numerical dosimetry enables more accurate modeling of complex objects than the use of simple solid. The objectives of this work were validate the coupling of mesh models to the Monte Carlo code GEANT4 and evaluate the influence of the number of vertices in the simulations to obtain absorbed fractions of energy (AFEs). Validation of the coupling was performed to internal sources of photons with energies between 10 keV and 1 MeV for spherical geometries described by the GEANT4 and three-dimensional models with different number of vertices and triangular or quadrilateral faces modeled using Blender program. As a result it was found that there were no significant differences between AFEs for objects described by mesh models and objects described using solid volumes of GEANT4. Since that maintained the shape and the volume the decrease in the number of vertices to describe an object does not influence so meant dosimetric data, but significantly decreases the time required to achieve the dosimetric calculations, especially for energies less than 100 keV.
Energy Technology Data Exchange (ETDEWEB)
Vieira, Jose Wilson
2004-07-15
The MAX phantom has been developed from existing segmented images of a male adult body, in order to achieve a representation as close as possible to the anatomical properties of the reference adult male specified by the ICRP. In computational dosimetry, MAX can simulate the geometry of a human body under exposure to ionizing radiations, internal or external, with the objective of calculating the equivalent dose in organs and tissues for occupational, medical or environmental purposes of the radiation protection. This study presents a methodology used to build a new computational exposure model MAX/EGS4: the geometric construction of the phantom; the development of the algorithm of one-directional, divergent, and isotropic radioactive sources; new methods for calculating the equivalent dose in the red bone marrow and in the skin, and the coupling of the MAX phantom with the EGS4 Monte Carlo code. Finally, some results of radiation protection, in the form of conversion coefficients between equivalent dose (or effective dose) and free air-kerma for external photon irradiation are presented and discussed. Comparing the results presented with similar data from other human phantoms it is possible to conclude that the coupling MAX/EGS4 is satisfactory for the calculation of the equivalent dose in radiation protection. (author)
Mashnik, Stepan G.; Kerby, Leslie M.; Gudima, Konstantin K.; Sierk, Arnold J.; Bull, Jeffrey S.; James, Michael R.
2017-03-01
We extend the cascade-exciton model (CEM), and the Los Alamos version of the quark-gluon string model (LAQGSM), event generators of the Monte Carlo N -particle transport code version 6 (MCNP6), to describe production of energetic light fragments (LF) heavier than 4He from various nuclear reactions induced by particles and nuclei at energies up to about 1 TeV/nucleon. In these models, energetic LF can be produced via Fermi breakup, preequilibrium emission, and coalescence of cascade particles. Initially, we study several variations of the Fermi breakup model and choose the best option for these models. Then, we extend the modified exciton model (MEM) used by these codes to account for a possibility of multiple emission of up to 66 types of particles and LF (up to 28Mg) at the preequilibrium stage of reactions. Then, we expand the coalescence model to allow coalescence of LF from nucleons emitted at the intranuclear cascade stage of reactions and from lighter clusters, up to fragments with mass numbers A ≤7 , in the case of CEM, and A ≤12 , in the case of LAQGSM. Next, we modify MCNP6 to allow calculating and outputting spectra of LF and heavier products with arbitrary mass and charge numbers. The improved version of CEM is implemented into MCNP6. Finally, we test the improved versions of CEM, LAQGSM, and MCNP6 on a variety of measured nuclear reactions. The modified codes give an improved description of energetic LF from particle- and nucleus-induced reactions; showing a good agreement with a variety of available experimental data. They have an improved predictive power compared to the previous versions and can be used as reliable tools in simulating applications involving such types of reactions.
Gifford, Kent A; Wareing, Todd A; Failla, Gregory; Horton, John L; Eifel, Patricia J; Mourtada, Firas
2009-12-03
A patient dose distribution was calculated by a 3D multi-group S N particle transport code for intracavitary brachytherapy of the cervix uteri and compared to previously published Monte Carlo results. A Cs-137 LDR intracavitary brachytherapy CT data set was chosen from our clinical database. MCNPX version 2.5.c, was used to calculate the dose distribution. A 3D multi-group S N particle transport code, Attila version 6.1.1 was used to simulate the same patient. Each patient applicator was built in SolidWorks, a mechanical design package, and then assembled with a coordinate transformation and rotation for the patient. The SolidWorks exported applicator geometry was imported into Attila for calculation. Dose matrices were overlaid on the patient CT data set. Dose volume histograms and point doses were compared. The MCNPX calculation required 14.8 hours, whereas the Attila calculation required 22.2 minutes on a 1.8 GHz AMD Opteron CPU. Agreement between Attila and MCNPX dose calculations at the ICRU 38 points was within +/- 3%. Calculated doses to the 2 cc and 5 cc volumes of highest dose differed by not more than +/- 1.1% between the two codes. Dose and DVH overlays agreed well qualitatively. Attila can calculate dose accurately and efficiently for this Cs-137 CT-based patient geometry. Our data showed that a three-group cross-section set is adequate for Cs-137 computations. Future work is aimed at implementing an optimized version of Attila for radiotherapy calculations.
Energy Technology Data Exchange (ETDEWEB)
Junior, Reginaldo G., E-mail: reginaldo.junior@ifmg.edu.br [Instituto Federal de Minas Gerais (IFMG), Formiga, MG (Brazil). Departamento de Engenharia Eletrica; Oliveira, Arno H. de; Sousa, Romulo V., E-mail: arnoheeren@gmail.com, E-mail: romuloverdolin@yahoo.com.br [Universidade Federal de Minas Gerais (UFMG), Belo Horizonte, MG (Brazil). Departamento de Engenharia Nuclear; Mourao, Arnaldo P., E-mail: apratabhz@gmail.com [Centro Federal de Educacao Tecnologica de Minas Gerais, Belo Horizonte, MG (Brazil)
2015-07-01
This paper reports the modeling of a linear accelerator Clinac 600 CD with BEAMnrc application, derived from EGSnrc radiation transport code, indicating relevant details of modeling that traditionally involve difficulties imposed on the process. This accelerator was commissioned by the confrontation of experimental dosimetric data with the computer data obtained by DOSXYZnrc application. The information compared in dosimetry process were: field profiles and dose percentage curves obtained in a water phantom with cubic edge of 30 cm. In all comparisons made, the computational data showed satisfactory precision and discrepancies with the experimental data did not exceed 3%, proving the electiveness of the model. Both the accelerator model and the computational dosimetry methodology, revealed the need for adjustments that probably will allow obtaining more accurate data than those obtained in the simulations presented here. These adjustments are mainly associated to improve the resolution of the eld profiles, the voxelization in phantom and optimization of computing time. (author)
Dunn, William L
2012-01-01
Exploring Monte Carlo Methods is a basic text that describes the numerical methods that have come to be known as "Monte Carlo." The book treats the subject generically through the first eight chapters and, thus, should be of use to anyone who wants to learn to use Monte Carlo. The next two chapters focus on applications in nuclear engineering, which are illustrative of uses in other fields. Five appendices are included, which provide useful information on probability distributions, general-purpose Monte Carlo codes for radiation transport, and other matters. The famous "Buffon's needle proble
Del Lama, L. S.; Cunha, D. M.; Poletti, M. E.
2017-08-01
The presence and morphology of microcalcification clusters are the main point to provide early indications of breast carcinomas. However, the visualization of those structures may be jeopardized due to overlapping tissues even for digital mammography systems. Although digital mammography is the current standard for breast cancer diagnosis, further improvements should be achieved in order to address some of those physical limitations. One possible solution for such issues is the application of the dual-energy technique (DE), which is able to highlight specific lesions or cancel out the tissue background. In this sense, this work aimed to evaluate several quantities of interest in radiation applications and compare those values with works present in the literature to validate a modified PENELOPE code for digital mammography applications. For instance, the scatter-to-primary ratio (SPR), the scatter fraction (SF) and the normalized mean glandular dose (DgN) were evaluated by simulations and the resulting values were compared to those found in earlier studies. Our results present a good correlation for the evaluated quantities, showing agreement equal or better than 5% for the scatter and dosimetric-related quantities when compared to the literature. Finally, a DE imaging chain was simulated and the visualization of microcalcifications was investigated.
Energy Technology Data Exchange (ETDEWEB)
Hindie, Elif; Moretti, Jean-Luc [Hopital Saint-Louis, Service de Medecine Nucleaire, Paris (France)]|[Universite Paris 7, Imagerie Moleculaire Diagnostique et Ciblage Therapeutique, Paris (France); Champion, Christophe [Universite Paul Verlaine, Laboratoire de Physique Moleculaire et des Collisions, Metz Institut de Physique, Metz (France); Zanotti-Fregonara, Paolo; Ravasi, Laura [Commissariat a l' Energie Atomique, DSV/I2BM/SHFJ/LIME, Orsay (France); Rubello, Domenico [Instituto Oncologico Veneto (IOV) - IRCCS, Department of Nuclear Medicine - PET Centre, Rovigo (Italy); Colas-Linhart, Nicole [Faculte de Medecine Xavier Bichat, Laboratoire de Biophysique, Paris (France)
2009-01-15
We used the Monte Carlo code ''CELLDOSE'' to assess the dose received by specific target cells from electron emissions in a complex environment. {sup 131}I in a simulated thyroid was used as a model. Thyroid follicles were represented by 170{mu}m diameter spherical units made of a lumen of 150{mu}m diameter containing colloidal matter and a peripheral layer of 10{mu}m thick thyroid cells. Neighbouring follicles are 4{mu}m apart. {sup 131}I was assumed to be homogeneously distributed in the lumen and absent in cells. We firstly assessed electron dose distribution in a single follicle. Then, we expanded the simulation by progressively adding neighbouring layers of follicles, so to reassess the electron dose to this single follicle implemented with the contribution of the added layers. Electron dose gradient around a point source showed that the {sup 131}I electron dose is close to zero after 2,100{mu}m. Therefore, we studied all contributions to the central follicle deriving from follicles within 12 orders of neighbourhood (15,624 follicles surrounding the central follicle). The dose to colloid of the single follicle was twice as high as the dose to thyroid cells. Even when all neighbours were taken into account, the dose in the central follicle remained heterogeneous. For a 1-Gy average dose to tissue, the dose to colloidal matter was 1.168 Gy, the dose to thyroid cells was 0.982 Gy, and the dose to the inter-follicular tissue was 0.895 Gy. Analysis of the different contributions to thyroid cell dose showed that 17.3% of the dose derived from the colloidal matter of their own follicle, while the remaining 82.7% was delivered by the surrounding follicles. On the basis of these data, it is shown that when different follicles contain different concentrations of {sup 131}I, the impact in terms of cell dose heterogeneity can be important. By means of {sup 131}I in the thyroid as a theoretical model, we showed how a Monte Carlo code can be used to map
Energy Technology Data Exchange (ETDEWEB)
Lara, Rafael G.; Maiorino, Jose R., E-mail: rafael.lara@aluno.ufabc.edu.br, E-mail: joserubens.maiorino@ufabc.edu.br [Universidade Federal do ABC (UFABC), Santo Andre, SP (Brazil). Centro de Engenharia, Modelagem e Ciencias Sociais Aplicadas
2013-07-01
This work aimed at the implementation and qualification of MCNP code in a supercomputer of the Universidade Federal do ABC, so that may be available a next-generation simulation tool for precise calculations of nuclear reactors and systems subject to radiation. The implementation of this tool will have multidisciplinary applications, covering various areas of engineering (nuclear, aerospace, biomedical), radiation physics and others.
Energy Technology Data Exchange (ETDEWEB)
Brown, F.B.; Sutton, T.M.
1996-02-01
This report is composed of the lecture notes from the first half of a 32-hour graduate-level course on Monte Carlo methods offered at KAPL. These notes, prepared by two of the principle developers of KAPL`s RACER Monte Carlo code, cover the fundamental theory, concepts, and practices for Monte Carlo analysis. In particular, a thorough grounding in the basic fundamentals of Monte Carlo methods is presented, including random number generation, random sampling, the Monte Carlo approach to solving transport problems, computational geometry, collision physics, tallies, and eigenvalue calculations. Furthermore, modern computational algorithms for vector and parallel approaches to Monte Carlo calculations are covered in detail, including fundamental parallel and vector concepts, the event-based algorithm, master/slave schemes, parallel scaling laws, and portability issues.
Pietrzak, Robert; Konefał, Adam; Sokół, Maria; Orlef, Andrzej
2016-08-01
The success of proton therapy depends strongly on the precision of treatment planning. Dose distribution in biological tissue may be obtained from Monte Carlo simulations using various scientific codes making it possible to perform very accurate calculations. However, there are many factors affecting the accuracy of modeling. One of them is a structure of objects called bins registering a dose. In this work the influence of bin structure on the dose distributions was examined. The MCNPX code calculations of Bragg curve for the 60 MeV proton beam were done in two ways: using simple logical detectors being the volumes determined in water, and using a precise model of ionization chamber used in clinical dosimetry. The results of the simulations were verified experimentally in the water phantom with Marcus ionization chamber. The average local dose difference between the measured relative doses in the water phantom and those calculated by means of the logical detectors was 1.4% at first 25 mm, whereas in the full depth range this difference was 1.6% for the maximum uncertainty in the calculations less than 2.4% and for the maximum measuring error of 1%. In case of the relative doses calculated with the use of the ionization chamber model this average difference was somewhat greater, being 2.3% at depths up to 25 mm and 2.4% in the full range of depths for the maximum uncertainty in the calculations of 3%. In the dose calculations the ionization chamber model does not offer any additional advantages over the logical detectors. The results provided by both models are similar and in good agreement with the measurements, however, the logical detector approach is a more time-effective method.
Bobin, C; Thiam, C; Bouchard, J
2016-03-01
At LNE-LNHB, a liquid scintillation (LS) detection setup designed for Triple to Double Coincidence Ratio (TDCR) measurements is also used in the β-channel of a 4π(LS)β-γ coincidence system. This LS counter based on 3 photomultipliers was first modeled using the Monte Carlo code Geant4 to enable the simulation of optical photons produced by scintillation and Cerenkov effects. This stochastic modeling was especially designed for the calculation of double and triple coincidences between photomultipliers in TDCR measurements. In the present paper, this TDCR-Geant4 model is extended to 4π(LS)β-γ coincidence counting to enable the simulation of the efficiency-extrapolation technique by the addition of a γ-channel. This simulation tool aims at the prediction of systematic biases in activity determination due to eventual non-linearity of efficiency-extrapolation curves. First results are described in the case of the standardization (59)Fe. The variation of the γ-efficiency in the β-channel due to the Cerenkov emission is investigated in the case of the activity measurements of (54)Mn. The problem of the non-linearity between β-efficiencies is featured in the case of the efficiency tracing technique for the activity measurements of (14)C using (60)Co as a tracer.
Institute of Scientific and Technical Information of China (English)
马永红; 葛良全; 张庆贤; 谷懿
2012-01-01
为了研究航空γ能谱低能谱段与地质体岩性的相关关系,基于蒙特卡罗方法,应用MCNP5程序模拟天然γ射线穿过不同地质体(辉绿岩、砂岩、白云岩、玄武岩、黑云母花岗岩、花岗斑岩),且经过100 m高度空气介质散射后γ能谱的变化.模拟结果显示:地质体的有效原子序数对航空γ能谱低能谱段能量＜ 150 keV的能量段影响较大,低能散射峰峰位、不同能量段峰面积之比与原子序数均有一定的相关性.地质体的密度不会改变低能段散射峰峰位.%Hie paper simulated the changes of low energy airborne gamma - ray spectrometry after gamma photons eitted by natural source through mediums,such as diabase、sandstone、doloinite、basah、biotite granite 、granite porphyry and so on. The result showed that the effective atomic number of medium has great influence on low energy spectrum( < 150 keV). The position of peak ,as well as the peak area ratio of different energy,has a certain relationship with effective atomic number of medium, the Medium density has little influence on peak situation of low energy spectrum.
MCDB Monte Carlo dosimetry code system and its applications%MCDB蒙特卡罗剂量计算系统及应用
Institute of Scientific and Technical Information of China (English)
邓力; 李刚; 陈朝斌; 叶涛
2012-01-01
硼中子俘获治疗(BNCT)蒙特卡罗剂量计算软件系统MCDB(Monte Carlo dosimetry code for brain)已经开发成功.它包括医学前处理、剂量计算和后处理.前处理把CT、MRI图像数据自动转化为剂量计算的输入文件,剂量计算基于蒙特卡罗(MC)方法,后处理是确定照射方向和照射时间.为了提高剂量计算的精度和缩短计算时间,MCDB发展了针对体素模型的快速粒子径迹算法,构造材料矩阵和计数矩阵,程序实现了MPI并行化.通过一个病例,MCDB完成了从CT、MRI提取数据、剂量计算和后处理的全过程.计算取得了与MCNP程序一致的结果,串行计算速度较MCNP提高3倍以上,并行效率可以达到90％,完全满足临床对计算精度和计算时间的要求.%MCDB is developed for boron neutron capture therapy ( BNCT). This system consists of a medical pre-processor, a dose computation and a post-processor. MCDB automatically produces the input file from CT and MRI image data. In Monte Carlo dose calculation, several accelerated measures, such as the fast track technique , mesh tally matrix and material matrix, are developed. In this paper, we proposed a real model simulated by MCNP and MCDB, respectively. The almost same results as MCNP are achieved. MCDB is faster in computational speed than MCNP.
Energy Technology Data Exchange (ETDEWEB)
Guan, Fada; Peeler, Christopher; Taleei, Reza; Randeniya, Sharmalee; Ge, Shuaiping; Mirkovic, Dragan; Mohan, Radhe; Titt, Uwe, E-mail: UTitt@mdanderson.org [Department of Radiation Physics, The University of Texas MD Anderson Cancer Center, 1515 Holcombe Boulevard, Houston, Texas 77030 (United States); Bronk, Lawrence [Department of Experimental Radiation Oncology, The University of Texas MD Anderson Cancer Center, 1515 Holcombe Boulevard, Houston, Texas 77030 (United States); Geng, Changran [Department of Nuclear Science and Engineering, Nanjing University of Aeronautics and Astronautics, Nanjing 210016, China and Department of Radiation Oncology, Massachusetts General Hospital and Harvard Medical School, Boston, Massachusetts 02114 (United States); Grosshans, David [Department of Experimental Radiation Oncology, The University of Texas MD Anderson Cancer Center, 1515 Holcombe Boulevard, Houston, Texas 77030 and Department of Radiation Oncology, The University of Texas MD Anderson Cancer Center, 1515 Holcombe Boulevard, Houston, Texas 77030 (United States)
2015-11-15
Purpose: The motivation of this study was to find and eliminate the cause of errors in dose-averaged linear energy transfer (LET) calculations from therapeutic protons in small targets, such as biological cell layers, calculated using the GEANT 4 Monte Carlo code. Furthermore, the purpose was also to provide a recommendation to select an appropriate LET quantity from GEANT 4 simulations to correlate with biological effectiveness of therapeutic protons. Methods: The authors developed a particle tracking step based strategy to calculate the average LET quantities (track-averaged LET, LET{sub t} and dose-averaged LET, LET{sub d}) using GEANT 4 for different tracking step size limits. A step size limit refers to the maximally allowable tracking step length. The authors investigated how the tracking step size limit influenced the calculated LET{sub t} and LET{sub d} of protons with six different step limits ranging from 1 to 500 μm in a water phantom irradiated by a 79.7-MeV clinical proton beam. In addition, the authors analyzed the detailed stochastic energy deposition information including fluence spectra and dose spectra of the energy-deposition-per-step of protons. As a reference, the authors also calculated the averaged LET and analyzed the LET spectra combining the Monte Carlo method and the deterministic method. Relative biological effectiveness (RBE) calculations were performed to illustrate the impact of different LET calculation methods on the RBE-weighted dose. Results: Simulation results showed that the step limit effect was small for LET{sub t} but significant for LET{sub d}. This resulted from differences in the energy-deposition-per-step between the fluence spectra and dose spectra at different depths in the phantom. Using the Monte Carlo particle tracking method in GEANT 4 can result in incorrect LET{sub d} calculation results in the dose plateau region for small step limits. The erroneous LET{sub d} results can be attributed to the algorithm to
Sangeetha, S.; Sureka, C. S.
2017-06-01
The present study is focused to compare the characteristics of Varian Clinac 600 C/D flattened and unflattened 6 MV photon beams for small field dosimetry using EGSnrc Monte Carlo Simulation since the small field dosimetry is considered to be the most crucial and provoking task in the field of radiation dosimetry. A 6 MV photon beam of a Varian Clinac 600 C/D medical linear accelerator operates with Flattening Filter (FF) and Flattening-Filter-Free (FFF) mode for small field dosimetry were performed using EGSnrc Monte Carlo user codes (BEAMnrc and DOSXYZnrc) in order to calculate the beam characteristics using Educated-trial and error method. These includes: Percentage depth dose, lateral beam profile, dose rate delivery, photon energy spectra, photon beam uniformity, out-of-field dose, surface dose, penumbral dose and output factor for small field dosimetry (0.5×0.5 cm2 to 4×4 cm2) and are compared with magna-field sizes (5×5 cm2 to 40×40 cm2) at various depths. The results obtained showed that the optimized beam energy and Full-width-half maximum value for small field dosimetry and magna-field dosimetry was found to be 5.7 MeV and 0.13 cm for both FF and FFF beams. The depth of dose maxima for small field size deviates minimally for both FF and FFF beams similar to magna-fields. The depths greater than dmax depicts a steeper dose fall off in the exponential region for FFF beams comparing FF beams where its deviations gets increased with the increase in field size. The shape of the lateral beam profiles of FF and FFF beams varies remains similar for the small field sizes less than 4×4 cm2 whereas it varies in the case of magna-fields. Dose rate delivery for FFF beams shows an eminent increase with a two-fold factor for both small field dosimetry and magna-field sizes. The surface dose measurements of FFF beams for small field size were found to be higher whereas it gets lower for magna-fields than FF beam. The amount of out-of-field dose reduction gets
Self-Shielding Treatment to Perform Cell Calculation for Seed Furl In Th/U Pwr Using Dragon Code
Directory of Open Access Journals (Sweden)
Ahmed Amin El Said Abd El Hameed
2015-08-01
Full Text Available Time and precision of the results are the most important factors in any code used for nuclear calculations. Despite of the high accuracy of Monte Carlo codes, MCNP and Serpent, in many cases their relatively long computational time leads to difficulties in using any of them as the main calculation code. Usually, Monte Carlo codes are used only to benchmark the results. The deterministic codes, which are usually used in nuclear reactor’s calculations, have limited precision, due to the approximations in the methods used to solve the multi-group transport equation. Self- Shielding treatment, an algorithm that produces an average cross-section defined over the complete energy domain of the neutrons in a nuclear reactor, is responsible for the biggest error in any deterministic codes. There are mainly two resonance self-shielding models commonly applied: models based on equivalence and dilution and models based on subgroup approach. The fundamental problem with any self-shielding method is that it treats any isotope as there are no other isotopes with resonance present in the reactor. The most practical way to solve this problem is to use multi-energy groups (50-200 that are chosen in a way that allows us to use all major resonances without self-shielding. In this paper, we perform cell calculations, for a fresh seed fuel pin which is used in thorium/uranium reactors, by solving 172 energy group transport equation using the deterministic DRAGON code, for the two types of self-shielding models (equivalence and dilution models and subgroup models Using WIMS-D5 and DRAGON data libraries. The results are then tested by comparing it with the stochastic MCNP5 code. We also tested the sensitivity of the results to a specific change in self-shielding method implemented, for example the effect of applying Livolant-Jeanpierre Normalization scheme and Rimman Integration improvement on the equivalence and dilution method, and the effect of using Ribbon
Energy Technology Data Exchange (ETDEWEB)
Chibani, Omar, E-mail: omar.chibani@fccc.edu; C-M Ma, Charlie [Fox Chase Cancer Center, Philadelphia, Pennsylvania 19111 (United States)
2014-05-15
Purpose: To present a new accelerated Monte Carlo code for CT-based dose calculations in high dose rate (HDR) brachytherapy. The new code (HDRMC) accounts for both tissue and nontissue heterogeneities (applicator and contrast medium). Methods: HDRMC uses a fast ray-tracing technique and detailed physics algorithms to transport photons through a 3D mesh of voxels representing the patient anatomy with applicator and contrast medium included. A precalculated phase space file for the{sup 192}Ir source is used as source term. HDRM is calibrated to calculated absolute dose for real plans. A postprocessing technique is used to include the exact density and composition of nontissue heterogeneities in the 3D phantom. Dwell positions and angular orientations of the source are reconstructed using data from the treatment planning system (TPS). Structure contours are also imported from the TPS to recalculate dose-volume histograms. Results: HDRMC was first benchmarked against the MCNP5 code for a single source in homogenous water and for a loaded gynecologic applicator in water. The accuracy of the voxel-based applicator model used in HDRMC was also verified by comparing 3D dose distributions and dose-volume parameters obtained using 1-mm{sup 3} versus 2-mm{sup 3} phantom resolutions. HDRMC can calculate the 3D dose distribution for a typical HDR cervix case with 2-mm resolution in 5 min on a single CPU. Examples of heterogeneity effects for two clinical cases (cervix and esophagus) were demonstrated using HDRMC. The neglect of tissue heterogeneity for the esophageal case leads to the overestimate of CTV D90, CTV D100, and spinal cord maximum dose by 3.2%, 3.9%, and 3.6%, respectively. Conclusions: A fast Monte Carlo code for CT-based dose calculations which does not require a prebuilt applicator model is developed for those HDR brachytherapy treatments that use CT-compatible applicators. Tissue and nontissue heterogeneities should be taken into account in modern HDR
Institute of Scientific and Technical Information of China (English)
邓力; 刘杰; 张文勇
2003-01-01
The particle transport Monte Carlo code MCNP had been realized the paral-lelization in MPI (Message Passing Interface) in 1999. But due to adopting the leap random number producer, some differences were existed between the parallel result and the serial result. Now the same results have been achieved by using the segment random number. The speedup of the applied problem is the liner ups to 53 in 64-Processors and the parallel efficiencv is up to 83% in 64-Processors.
Ferretti, A; Martignano, A; Simonato, F; Paiusco, M
2014-02-01
The aim of the present work was the validation of the VMC(++) Monte Carlo (MC) engine implemented in the Oncentra Masterplan (OMTPS) and used to calculate the dose distribution produced by the electron beams (energy 5-12 MeV) generated by the linear accelerator (linac) Primus (Siemens), shaped by a digital variable applicator (DEVA). The BEAMnrc/DOSXYZnrc (EGSnrc package) MC model of the linac head was used as a benchmark. Commissioning results for both MC codes were evaluated by means of 1D Gamma Analysis (2%, 2 mm), calculated with a home-made Matlab (The MathWorks) program, comparing the calculations with the measured profiles. The results of the commissioning of OMTPS were good [average gamma index (γ) > 97%]; some mismatches were found with large beams (size ≥ 15 cm). The optimization of the BEAMnrc model required to increase the beam exit window to match the calculated and measured profiles (final average γ > 98%). Then OMTPS dose distribution maps were compared with DOSXYZnrc with a 2D Gamma Analysis (3%, 3 mm), in 3 virtual water phantoms: (a) with an air step, (b) with an air insert, and (c) with a bone insert. The OMTPD and EGSnrc dose distributions with the air-water step phantom were in very high agreement (γ ∼ 99%), while for heterogeneous phantoms there were differences of about 9% in the air insert and of about 10-15% in the bone region. This is due to the Masterplan implementation of VMC(++) which reports the dose as "dose to water", instead of "dose to medium".
Energy Technology Data Exchange (ETDEWEB)
Pietrzak, Robert [Department of Nuclear Physics and Its Applications, Institute of Physics, University of Silesia, Katowice (Poland); Konefał, Adam, E-mail: adam.konefal@us.edu.pl [Department of Nuclear Physics and Its Applications, Institute of Physics, University of Silesia, Katowice (Poland); Sokół, Maria; Orlef, Andrzej [Department of Medical Physics, Maria Sklodowska-Curie Memorial Cancer Center, Institute of Oncology, Gliwice (Poland)
2016-08-01
The success of proton therapy depends strongly on the precision of treatment planning. Dose distribution in biological tissue may be obtained from Monte Carlo simulations using various scientific codes making it possible to perform very accurate calculations. However, there are many factors affecting the accuracy of modeling. One of them is a structure of objects called bins registering a dose. In this work the influence of bin structure on the dose distributions was examined. The MCNPX code calculations of Bragg curve for the 60 MeV proton beam were done in two ways: using simple logical detectors being the volumes determined in water, and using a precise model of ionization chamber used in clinical dosimetry. The results of the simulations were verified experimentally in the water phantom with Marcus ionization chamber. The average local dose difference between the measured relative doses in the water phantom and those calculated by means of the logical detectors was 1.4% at first 25 mm, whereas in the full depth range this difference was 1.6% for the maximum uncertainty in the calculations less than 2.4% and for the maximum measuring error of 1%. In case of the relative doses calculated with the use of the ionization chamber model this average difference was somewhat greater, being 2.3% at depths up to 25 mm and 2.4% in the full range of depths for the maximum uncertainty in the calculations of 3%. In the dose calculations the ionization chamber model does not offer any additional advantages over the logical detectors. The results provided by both models are similar and in good agreement with the measurements, however, the logical detector approach is a more time-effective method. - Highlights: • Influence of the bin structure on the proton dose distributions was examined for the MC simulations. • The considered relative proton dose distributions in water correspond to the clinical application. • MC simulations performed with the logical detectors and the
Ficaro, Edward Patrick
The ^{252}Cf -source-driven noise analysis (CSDNA) requires the measurement of the cross power spectral density (CPSD) G_ {23}(omega), between a pair of neutron detectors (subscripts 2 and 3) located in or near the fissile assembly, and the CPSDs, G_{12}( omega) and G_{13}( omega), between the neutron detectors and an ionization chamber 1 containing ^{252}Cf also located in or near the fissile assembly. The key advantage of this method is that the subcriticality of the assembly can be obtained from the ratio of spectral densities,{G _sp{12}{*}(omega)G_ {13}(omega)over G_{11 }(omega)G_{23}(omega) },using a point kinetic model formulation which is independent of the detector's properties and a reference measurement. The multigroup, Monte Carlo code, KENO-NR, was developed to eliminate the dependence of the measurement on the point kinetic formulation. This code utilizes time dependent, analog neutron tracking to simulate the experimental method, in addition to the underlying nuclear physics, as closely as possible. From a direct comparison of simulated and measured data, the calculational model and cross sections are validated for the calculation, and KENO-NR can then be rerun to provide a distributed source k_ {eff} calculation. Depending on the fissile assembly, a few hours to a couple of days of computation time are needed for a typical simulation executed on a desktop workstation. In this work, KENO-NR demonstrated the ability to accurately estimate the measured ratio of spectral densities from experiments using capture detectors performed on uranium metal cylinders, a cylindrical tank filled with aqueous uranyl nitrate, and arrays of safe storage bottles filled with uranyl nitrate. Good agreement was also seen between simulated and measured values of the prompt neutron decay constant from the fitted CPSDs. Poor agreement was seen between simulated and measured results using composite ^6Li-glass-plastic scintillators at large subcriticalities for the tank of
Hajizadeh-Safar, M; Ghorbani, M; Khoshkharam, S; Ashrafi, Z
2014-07-01
Gamma camera is an important apparatus in nuclear medicine imaging. Its detection part is consists of a scintillation detector with a heavy collimator. Substitution of semiconductor detectors instead of scintillator in these cameras has been effectively studied. In this study, it is aimed to introduce a new design of P-N semiconductor detector array for nuclear medicine imaging. A P-N semiconductor detector composed of N-SnO2 :F, and P-NiO:Li, has been introduced through simulating with MCNPX monte carlo codes. Its sensitivity with different factors such as thickness, dimension, and direction of emission photons were investigated. It is then used to configure a new design of an array in one-dimension and study its spatial resolution for nuclear medicine imaging. One-dimension array with 39 detectors was simulated to measure a predefined linear distribution of Tc(99_m) activity and its spatial resolution. The activity distribution was calculated from detector responses through mathematical linear optimization using LINPROG code on MATLAB software. Three different configurations of one-dimension detector array, horizontal, vertical one sided, and vertical double-sided were simulated. In all of these configurations, the energy windows of the photopeak were ± 1%. The results show that the detector response increases with an increase of dimension and thickness of the detector with the highest sensitivity for emission photons 15-30° above the surface. Horizontal configuration array of detectors is not suitable for imaging of line activity sources. The measured activity distribution with vertical configuration array, double-side detectors, has no similarity with emission sources and hence is not suitable for imaging purposes. Measured activity distribution using vertical configuration array, single side detectors has a good similarity with sources. Therefore, it could be introduced as a suitable configuration for nuclear medicine imaging. It has been shown that using
Development of Burnup Calculation Function in Reactor Monte Carlo Code RMC%堆用蒙卡程序燃耗计算功能开发
Institute of Scientific and Technical Information of China (English)
佘顶; 王侃; 余纲林
2012-01-01
This paper presents the burnup calculation capability of RMC, which is a new Monte Carlo (MC) neutron transport code developed by Reactor Engineering Analysis Laboratory (REAL) in Tsinghua university of China. Unlike most of existing MC depletion codes which explicitly couple the depletion module, RMC incorporates ORIGEN 2.1 in an implicit way. Different burn step strategies, including the middle-of-step approximation and the predictor-corrector method, are adopted by RMC to assure the accuracy under large burnup step size. RMC employs a spectrum-based method of tallying one-group cross section, which can considerably saves computational time with negligible accuracy loss. According to the validation results of benchmarks and examples, it is proved that the burnup function of RMC performs quite well in accuracy and efficiency.%堆用蒙卡程序(RMC)是由清华大学工程物理系REAL实验室自主开发的用于反应堆物理分析的中子输运蒙卡程序,本文主要介绍其燃耗计算功能的开发与验证.RMC的燃耗计算功能具有的特点:内部耦合ORIGEN,相比于外耦合方式,更加灵活和高效；使用基于能谱的单群截面统计方法,可在保证精度的前提下,显著提高计算效率；采取预估修正和中点近似等多种燃耗步策略,减小大燃耗步长时的计算误差.通过计算压水堆栅元、沸水堆组件、快堆等一系列基准题和算例,验证了RMC燃耗计算的正确性和速度优势.
Energy Technology Data Exchange (ETDEWEB)
Hashimoto, M.; Saito, K.; Ando, H. [Power Reactor and Nuclear Fuel Development Corp., Oarai, Ibaraki (Japan). Oarai Engineering Center
1998-05-01
The method to calculate the response function of spherical BF{sub 3} proportional counter, which is commonly used as neutron dose rate meter and neutron spectrometer with multi moderator system, is developed. As the calculation code for evaluating the response function, the existing code series NRESP, the Monte Carlo code for the calculation of response function of neutron detectors, is selected. However, the application scope of the existing NRESP is restricted, the NRESP98 is tuned as generally applicable code, with expansion of the geometrical condition, the applicable element, etc. The NRESP98 is tested with the response function of the spherical BF{sub 3} proportional counter. Including the effect of the distribution of amplification factor, the detailed evaluation of the charged particle transportation and the effect of the statistical distribution, the result of NRESP98 calculation fit the experience within {+-}10%. (author)
Saizu, Mirela Angela
2016-09-01
The developments of high-purity germanium detectors match very well the requirements of the in-vivo human body measurements regarding the gamma energy ranges of the radionuclides intended to be measured, the shape of the extended radioactive sources, and the measurement geometries. The Whole Body Counter (WBC) from IFIN-HH is based on an “over-square” high-purity germanium detector (HPGe) to perform accurate measurements of the incorporated radionuclides emitting X and gamma rays in the energy range of 10 keV-1500 keV, under conditions of good shielding, suitable collimation, and calibration. As an alternative to the experimental efficiency calibration method consisting of using reference calibration sources with gamma energy lines that cover all the considered energy range, it is proposed to use the Monte Carlo method for the efficiency calibration of the WBC using the radiation transport code MCNP5. The HPGe detector was modelled and the gamma energy lines of 241Am, 57Co, 133Ba, 137Cs, 60Co, and 152Eu were simulated in order to obtain the virtual efficiency calibration curve of the WBC. The Monte Carlo method was validated by comparing the simulated results with the experimental measurements using point-like sources. For their optimum matching, the impact of the variation of the front dead layer thickness and of the detector photon absorbing layers materials on the HPGe detector efficiency was studied, and the detector’s model was refined. In order to perform the WBC efficiency calibration for realistic people monitoring, more numerical calculations were generated simulating extended sources of specific shape according to the standard man characteristics.
Development of a simple detector response function generation program: the CEARDRFs code.
Wang, Jiaxin; Wang, Zhijian; Peeples, Johanna; Yu, Huawei; Gardner, Robin P
2012-07-01
A simple Monte Carlo program named CEARDRFs has been developed to generate very accurate detector response functions (DRFs) for scintillation detectors. It utilizes relatively rigorous gamma-ray transport with simple electron transport, and accounts for two phenomena that have rarely been treated: scintillator non-linearity and the variable flat continuum part of the DRF. It has been proven that these physics and treatments work well for 3×3″ and 6×6″ cylindrical NaI detector in CEAR's previous work. Now this approach has been expanded to cover more scintillation detectors with various common shapes and sizes. Benchmark experiments of 2×2″ cylindrical BGO detector and 2×4×16″ rectangular NaI detector have been carried out at CEAR with various radiactive sources. The simulation results of CEARDRFs have also been compared with MCNP5 calculations. The benchmark and comparison show that CEARDRFs can generate very accurate DRFs (more accurate than MCNP5) at a very fast speed (hundred times faster than MCNP5). The use of this program can significantly increase the accuracy of applications relying on detector spectroscopy like prompt gamma-ray neutron activation analysis, X-ray fluorescence analysis, oil well logging and homeland security. Copyright © 2011 Elsevier Ltd. All rights reserved.
Monte Carlo transition probabilities
Lucy, L. B.
2001-01-01
Transition probabilities governing the interaction of energy packets and matter are derived that allow Monte Carlo NLTE transfer codes to be constructed without simplifying the treatment of line formation. These probabilities are such that the Monte Carlo calculation asymptotically recovers the local emissivity of a gas in statistical equilibrium. Numerical experiments with one-point statistical equilibrium problems for Fe II and Hydrogen confirm this asymptotic behaviour. In addition, the re...
Development of a simple detector response function generation program: The CEARDRFs code
Energy Technology Data Exchange (ETDEWEB)
Wang Jiaxin, E-mail: jwang3@ncsu.edu [Center for Engineering Applications of Radioisotopes (CEAR), Department of Nuclear Engineering, North Carolina State University, Raleigh, NC 27695 (United States); Wang Zhijian; Peeples, Johanna [Center for Engineering Applications of Radioisotopes (CEAR), Department of Nuclear Engineering, North Carolina State University, Raleigh, NC 27695 (United States); Yu Huawei [Center for Engineering Applications of Radioisotopes (CEAR), Department of Nuclear Engineering, North Carolina State University, Raleigh, NC 27695 (United States); College of Geo-Resources and Information, China University of Petroleum, Qingdao, Shandong 266555 (China); Gardner, Robin P. [Center for Engineering Applications of Radioisotopes (CEAR), Department of Nuclear Engineering, North Carolina State University, Raleigh, NC 27695 (United States)
2012-07-15
A simple Monte Carlo program named CEARDRFs has been developed to generate very accurate detector response functions (DRFs) for scintillation detectors. It utilizes relatively rigorous gamma-ray transport with simple electron transport, and accounts for two phenomena that have rarely been treated: scintillator non-linearity and the variable flat continuum part of the DRF. It has been proven that these physics and treatments work well for 3 Multiplication-Sign 3 Double-Prime and 6 Multiplication-Sign 6 Double-Prime cylindrical NaI detector in CEAR's previous work. Now this approach has been expanded to cover more scintillation detectors with various common shapes and sizes. Benchmark experiments of 2 Multiplication-Sign 2 Double-Prime cylindrical BGO detector and 2 Multiplication-Sign 4 Multiplication-Sign 16 Double-Prime rectangular NaI detector have been carried out at CEAR with various radiactive sources. The simulation results of CEARDRFs have also been compared with MCNP5 calculations. The benchmark and comparison show that CEARDRFs can generate very accurate DRFs (more accurate than MCNP5) at a very fast speed (hundred times faster than MCNP5). The use of this program can significantly increase the accuracy of applications relying on detector spectroscopy like prompt gamma-ray neutron activation analysis, X-ray fluorescence analysis, oil well logging and homeland security. - Highlights: Black-Right-Pointing-Pointer CEARDRF has been developed to generate detector response functions (DRFs) for scintillation detectors a. Black-Right-Pointing-Pointer Generated DRFs are very accurate. Black-Right-Pointing-Pointer Simulation speed is hundreds of times faster than MCNP5. Black-Right-Pointing-Pointer It utilizes rigorous gamma-ray transport with simple electron transport. Black-Right-Pointing-Pointer It also accounts for scintillator non-linearity and the variable flat continuum part.
Energy Technology Data Exchange (ETDEWEB)
Mukhopadhyay, Nitai D. [Department of Biostatistics, Virginia Commonwealth University, Richmond, VA 23298 (United States); Sampson, Andrew J. [Department of Radiation Oncology, Virginia Commonwealth University, Richmond, VA 23298 (United States); Deniz, Daniel; Alm Carlsson, Gudrun [Department of Radiation Physics, Faculty of Health Sciences, Linkoeping University, SE 581 85 (Sweden); Williamson, Jeffrey [Department of Radiation Oncology, Virginia Commonwealth University, Richmond, VA 23298 (United States); Malusek, Alexandr, E-mail: malusek@ujf.cas.cz [Department of Radiation Physics, Faculty of Health Sciences, Linkoeping University, SE 581 85 (Sweden); Department of Radiation Dosimetry, Nuclear Physics Institute AS CR v.v.i., Na Truhlarce 39/64, 180 86 Prague (Czech Republic)
2012-01-15
Correlated sampling Monte Carlo methods can shorten computing times in brachytherapy treatment planning. Monte Carlo efficiency is typically estimated via efficiency gain, defined as the reduction in computing time by correlated sampling relative to conventional Monte Carlo methods when equal statistical uncertainties have been achieved. The determination of the efficiency gain uncertainty arising from random effects, however, is not a straightforward task specially when the error distribution is non-normal. The purpose of this study is to evaluate the applicability of the F distribution and standardized uncertainty propagation methods (widely used in metrology to estimate uncertainty of physical measurements) for predicting confidence intervals about efficiency gain estimates derived from single Monte Carlo runs using fixed-collision correlated sampling in a simplified brachytherapy geometry. A bootstrap based algorithm was used to simulate the probability distribution of the efficiency gain estimates and the shortest 95% confidence interval was estimated from this distribution. It was found that the corresponding relative uncertainty was as large as 37% for this particular problem. The uncertainty propagation framework predicted confidence intervals reasonably well; however its main disadvantage was that uncertainties of input quantities had to be calculated in a separate run via a Monte Carlo method. The F distribution noticeably underestimated the confidence interval. These discrepancies were influenced by several photons with large statistical weights which made extremely large contributions to the scored absorbed dose difference. The mechanism of acquiring high statistical weights in the fixed-collision correlated sampling method was explained and a mitigation strategy was proposed.
Mukhopadhyay, Nitai D; Sampson, Andrew J; Deniz, Daniel; Alm Carlsson, Gudrun; Williamson, Jeffrey; Malusek, Alexandr
2012-01-01
Correlated sampling Monte Carlo methods can shorten computing times in brachytherapy treatment planning. Monte Carlo efficiency is typically estimated via efficiency gain, defined as the reduction in computing time by correlated sampling relative to conventional Monte Carlo methods when equal statistical uncertainties have been achieved. The determination of the efficiency gain uncertainty arising from random effects, however, is not a straightforward task specially when the error distribution is non-normal. The purpose of this study is to evaluate the applicability of the F distribution and standardized uncertainty propagation methods (widely used in metrology to estimate uncertainty of physical measurements) for predicting confidence intervals about efficiency gain estimates derived from single Monte Carlo runs using fixed-collision correlated sampling in a simplified brachytherapy geometry. A bootstrap based algorithm was used to simulate the probability distribution of the efficiency gain estimates and the shortest 95% confidence interval was estimated from this distribution. It was found that the corresponding relative uncertainty was as large as 37% for this particular problem. The uncertainty propagation framework predicted confidence intervals reasonably well; however its main disadvantage was that uncertainties of input quantities had to be calculated in a separate run via a Monte Carlo method. The F distribution noticeably underestimated the confidence interval. These discrepancies were influenced by several photons with large statistical weights which made extremely large contributions to the scored absorbed dose difference. The mechanism of acquiring high statistical weights in the fixed-collision correlated sampling method was explained and a mitigation strategy was proposed.
Energy Technology Data Exchange (ETDEWEB)
Casanovas, R., E-mail: ramon.casanovas@urv.cat [Unitat de Fisica Medica, Facultat de Medicina i Ciencies de la Salut, Universitat Rovira i Virgili, ES-43201 Reus (Tarragona) (Spain); Morant, J.J. [Servei de Proteccio Radiologica, Facultat de Medicina i Ciencies de la Salut, Universitat Rovira i Virgili, ES-43201 Reus (Tarragona) (Spain); Salvado, M. [Unitat de Fisica Medica, Facultat de Medicina i Ciencies de la Salut, Universitat Rovira i Virgili, ES-43201 Reus (Tarragona) (Spain)
2012-05-21
The radiation detectors yield the optimal performance if they are accurately calibrated. This paper presents the energy, resolution and efficiency calibrations for two scintillation detectors, NaI(Tl) and LaBr{sub 3}(Ce). For the two former calibrations, several fitting functions were tested. To perform the efficiency calculations, a Monte Carlo user code for the EGS5 code system was developed with several important implementations. The correct performance of the simulations was validated by comparing the simulated spectra with the experimental spectra and reproducing a number of efficiency and activity calculations. - Highlights: Black-Right-Pointing-Pointer NaI(Tl) and LaBr{sub 3}(Ce) scintillation detectors are used for gamma-ray spectrometry. Black-Right-Pointing-Pointer Energy, resolution and efficiency calibrations are discussed for both detectors. Black-Right-Pointing-Pointer For the two former calibrations, several fitting functions are tested. Black-Right-Pointing-Pointer A Monte Carlo user code for EGS5 was developed for the efficiency calculations. Black-Right-Pointing-Pointer The code was validated reproducing some efficiency and activity calculations.
Chabert, I.; Barat, E.; Dautremer, T.; Montagu, T.; Agelou, M.; Croc de Suray, A.; Garcia-Hernandez, J. C.; Gempp, S.; Benkreira, M.; de Carlan, L.; Lazaro, D.
2016-07-01
This work aims at developing a generic virtual source model (VSM) preserving all existing correlations between variables stored in a Monte Carlo pre-computed phase space (PS) file, for dose calculation and high-resolution portal image prediction. The reference PS file was calculated using the PENELOPE code, after the flattening filter (FF) of an Elekta Synergy 6 MV photon beam. Each particle was represented in a mobile coordinate system by its radial position (r s ) in the PS plane, its energy (E), and its polar and azimuthal angles (φ d and θ d ), describing the particle deviation compared to its initial direction after bremsstrahlung, and the deviation orientation. Three sub-sources were created by sorting out particles according to their last interaction location (target, primary collimator or FF). For each sub-source, 4D correlated-histograms were built by storing E, r s , φ d and θ d values. Five different adaptive binning schemes were studied to construct 4D histograms of the VSMs, to ensure histogram efficient handling as well as an accurate reproduction of E, r s , φ d and θ d distribution details. The five resulting VSMs were then implemented in PENELOPE. Their accuracy was first assessed in the PS plane, by comparing E, r s , φ d and θ d distributions with those obtained from the reference PS file. Second, dose distributions computed in water, using the VSMs and the reference PS file located below the FF, and also after collimation in both water and heterogeneous phantom, were compared using a 1.5%-0 mm and a 2%-0 mm global gamma index, respectively. Finally, portal images were calculated without and with phantoms in the beam. The model was then evaluated using a 1%-0 mm global gamma index. Performance of a mono-source VSM was also investigated and led, as with the multi-source model, to excellent results when combined with an adaptive binning scheme.
Chabert, I; Barat, E; Dautremer, T; Montagu, T; Agelou, M; Croc de Suray, A; Garcia-Hernandez, J C; Gempp, S; Benkreira, M; de Carlan, L; Lazaro, D
2016-07-21
This work aims at developing a generic virtual source model (VSM) preserving all existing correlations between variables stored in a Monte Carlo pre-computed phase space (PS) file, for dose calculation and high-resolution portal image prediction. The reference PS file was calculated using the PENELOPE code, after the flattening filter (FF) of an Elekta Synergy 6 MV photon beam. Each particle was represented in a mobile coordinate system by its radial position (r s ) in the PS plane, its energy (E), and its polar and azimuthal angles (φ d and θ d ), describing the particle deviation compared to its initial direction after bremsstrahlung, and the deviation orientation. Three sub-sources were created by sorting out particles according to their last interaction location (target, primary collimator or FF). For each sub-source, 4D correlated-histograms were built by storing E, r s , φ d and θ d values. Five different adaptive binning schemes were studied to construct 4D histograms of the VSMs, to ensure histogram efficient handling as well as an accurate reproduction of E, r s , φ d and θ d distribution details. The five resulting VSMs were then implemented in PENELOPE. Their accuracy was first assessed in the PS plane, by comparing E, r s , φ d and θ d distributions with those obtained from the reference PS file. Second, dose distributions computed in water, using the VSMs and the reference PS file located below the FF, and also after collimation in both water and heterogeneous phantom, were compared using a 1.5%-0 mm and a 2%-0 mm global gamma index, respectively. Finally, portal images were calculated without and with phantoms in the beam. The model was then evaluated using a 1%-0 mm global gamma index. Performance of a mono-source VSM was also investigated and led, as with the multi-source model, to excellent results when combined with an adaptive binning scheme.
Energy Technology Data Exchange (ETDEWEB)
Dimitriadis, A; Gialousis, G; Karlatira, M; Karaiskos, P; Georgiou, E; Yakoumakis, E [Medical Physics Department, Medical School, University of Athens, 75 Mikras Asias Str., Goudi 11527, Athens (Greece); Makri, T; Papaodysseas, S, E-mail: anestisdim@yahoo.com [Radiological Imaging Department, Ag. Sofia Hospital, Lebadias and Thibon, Goudi 11527, Athens (Greece)
2011-01-21
Organ doses are important quantities in assessing the radiation risk. In the case of children, estimation of this risk is of particular concern due to their significant radiosensitivity and the greater health detriment. The purpose of this study is to estimate the organ doses to paediatric patients undergoing barium meal and micturating cystourethrography examinations by clinical measurements and Monte Carlo simulation. In clinical measurements, dose-area products (DAPs) were assessed during examination of 50 patients undergoing barium meal and 90 patients undergoing cystourethrography examinations, separated equally within three age categories: namely newborn, 1 year and 5 years old. Monte Carlo simulation of photon transport in male and female mathematical phantoms was applied using the MCNP5 code in order to estimate the equivalent organ doses. Regarding the micturating cystourethrography examinations, the organs receiving considerable amounts of radiation doses were the urinary bladder (1.87, 2.43 and 4.7 mSv, the first, second and third value in the parentheses corresponds to neonatal, 1 year old and 5 year old patients, respectively), the large intestines (1.54, 1.8, 3.1 mSv), the small intestines (1.34, 1.56, 2.78 mSv), the stomach (1.46, 1.02, 2.01 mSv) and the gall bladder (1.46, 1.66, 2.18 mSv), depending upon the age of the child. Organs receiving considerable amounts of radiation during barium meal examinations were the stomach (9.81, 9.92, 11.5 mSv), the gall bladder (3.05, 5.74, 7.15 mSv), the rib bones (9.82, 10.1, 11.1 mSv) and the pancreas (5.8, 5.93, 6.65 mSv), depending upon the age of the child. DAPs to organ/effective doses conversion factors were derived for each age and examination in order to be compared with other studies.
Energy Technology Data Exchange (ETDEWEB)
Marcus, Ryan C. [Los Alamos National Laboratory
2012-07-25
MCMini is a proof of concept that demonstrates the possibility for Monte Carlo neutron transport using OpenCL with a focus on performance. This implementation, written in C, shows that tracing particles and calculating reactions on a 3D mesh can be done in a highly scalable fashion. These results demonstrate a potential path forward for MCNP or other Monte Carlo codes.
Energy Technology Data Exchange (ETDEWEB)
Bastida O, G. E.; Vallejo Q, J. A.; Galicia A, J.; Francois L, J. L. [UNAM, Facultad de Ingenieria, Departamento de Sistemas Energeticos, Paseo Cuauhnahuac 8532, 62550 Jiutepec, Morelos (Mexico); Xolocostli M, J. V.; Rodriguez H, A.; Gomez T, A. M., E-mail: gbo729@yahoo.com.mx [ININ, Carretera Mexico-Toluca s/n, 52750 Ocoyoacac, Estado de Mexico (Mexico)
2016-09-15
This paper presents an analysis of results obtained from simulations performed with the neutron transport code AZTRAN and the kinetic code of neutron diffusion AZKIND, based on comparisons with models corresponding to a typical BWR, in order to verify the behavior and reliability of the values obtained with said code for its current development. For this, simulations of different geometries were made using validated nuclear codes, such as CASMO, MCNP5 and Serpent. The results obtained are considered adequate since they are comparable with those obtained and reported with other codes, based mainly on the neutron multiplication factor and the power distribution of the same. (Author)
Energy Technology Data Exchange (ETDEWEB)
Handley, G. R.; Masters, L. C.; Stachowiak, R. V.
1981-04-10
Validation of the Monte Carlo criticality code, KENO IV, and the Hansen-Roach sixteen-energy-group cross sections was accomplished by calculating the effective neutron multiplication constant, k/sub eff/, of 29 experimentally critical assemblies which had uranium enrichments of 92.6% or higher in the uranium-235 isotope. The experiments were chosen so that a large variety of geometries and of neutron energy spectra were covered. Problems, calculating the k/sub eff/ of systems with high-uranium-concentration uranyl nitrate solution that were minimally reflected or unreflected, resulted in the separate examination of five cases.
(U) Introduction to Monte Carlo Methods
Energy Technology Data Exchange (ETDEWEB)
Hungerford, Aimee L. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2017-03-20
Monte Carlo methods are very valuable for representing solutions to particle transport problems. Here we describe a “cook book” approach to handling the terms in a transport equation using Monte Carlo methods. Focus is on the mechanics of a numerical Monte Carlo code, rather than the mathematical foundations of the method.
Energy Technology Data Exchange (ETDEWEB)
Castelo e Silva, L.A., E-mail: castelo@ifsp.edu.br [Instituto Federal de Sao Paulo (IFSP), SP (Brazil); Mendes, M.B.; Goncalves, B.R.; Santos, D.M.M.; Vieira, M.V.; Fonseca, R.L.M.; Zenobio, M.A.F.; Fonseca, T.C.F. [Centro de Desenvolvimento da Tecnologia Nuclear (CDTN/CNEN-MG), Belo Horizonte, MG (Brazil); Paixao, L. [Universidade Federal de Minas Gerais (UFMG), Belo Horizonte, MG (Brazil)
2016-07-01
The main goal of this work is to publish the results of an inter-comparison simulation exercise of a clinical 10 x 10 cm{sup 2} beam model of a 6 MV LINAC using two different Monte Carlo codes: the MCNPX and EGSnrc. Results obtained for the dosimetric parameters PDD{sub 20,10} and TPR{sub 20,10} were compared with experimental data obtained in Radiotherapy and Megavoltage Institute of Minas Gerais. The main challenges on the computational modeling of this system are reported and discussed for didactic purposes in the area of modeling and simulation. (author)
Energy Technology Data Exchange (ETDEWEB)
Biondo, Elliott D [ORNL; Ibrahim, Ahmad M [ORNL; Mosher, Scott W [ORNL; Grove, Robert E [ORNL
2015-01-01
Detailed radiation transport calculations are necessary for many aspects of the design of fusion energy systems (FES) such as ensuring occupational safety, assessing the activation of system components for waste disposal, and maintaining cryogenic temperatures within superconducting magnets. Hybrid Monte Carlo (MC)/deterministic techniques are necessary for this analysis because FES are large, heavily shielded, and contain streaming paths that can only be resolved with MC. The tremendous complexity of FES necessitates the use of CAD geometry for design and analysis. Previous ITER analysis has required the translation of CAD geometry to MCNP5 form in order to use the AutomateD VAriaNce reducTion Generator (ADVANTG) for hybrid MC/deterministic transport. In this work, ADVANTG was modified to support CAD geometry, allowing hybrid (MC)/deterministic transport to be done automatically and eliminating the need for this translation step. This was done by adding a new ray tracing routine to ADVANTG for CAD geometries using the Direct Accelerated Geometry Monte Carlo (DAGMC) software library. This new capability is demonstrated with a prompt dose rate calculation for an ITER computational benchmark problem using both the Consistent Adjoint Driven Importance Sampling (CADIS) method an the Forward Weighted (FW)-CADIS method. The variance reduction parameters produced by ADVANTG are shown to be the same using CAD geometry and standard MCNP5 geometry. Significant speedups were observed for both neutrons (as high as a factor of 7.1) and photons (as high as a factor of 59.6).
Energy Technology Data Exchange (ETDEWEB)
Cupini, E. [ENEA, Centro Ricerche `Ezio Clementel`, Bologna (Italy). Dipt. Innovazione; Borgia, M.G. [ENEA, Centro Ricerche `Ezio Clementel`, Bologna (Italy). Dipt. Energia; Premuda, M. [Consiglio Nazionale delle Ricerche, Bologna (Italy). Ist. FISBAT
1997-03-01
The Montecarlo code PREMAR is described, which allows the user to simulate the radiation transport in the atmosphere, in the ultraviolet-infrared frequency interval. A plan multilayer geometry is at present foreseen by the code, witch albedo possibility at the lower boundary surface. For a given monochromatic point source, the main quantities computed by the code are the absorption spatial distributions of aerosol and molecules, together with the related atmospheric transmittances. Moreover, simulation of of Lidar experiments are foreseen by the code, the source and telescope fields of view being assigned. To build-up the appropriate probability distributions, an input data library is assumed to be read by the code. For this purpose the radiance-transmittance LOWTRAN-7 code has been conveniently adapted as a source of the library so as to exploit the richness of information of the code for a large variety of atmospheric simulations. Results of applications of the PREMAR code are finally presented, with special reference to simulations of Lidar system and radiometer experiments carried out at the Brasimone ENEA Centre by the Environment Department.
Hrivnacova, I; Berejnov, V V; Brun, R; Carminati, F; Fassò, A; Futo, E; Gheata, A; Caballero, I G; Morsch, Andreas
2003-01-01
The concept of Virtual Monte Carlo (VMC) has been developed by the ALICE Software Project to allow different Monte Carlo simulation programs to run without changing the user code, such as the geometry definition, the detector response simulation or input and output formats. Recently, the VMC classes have been integrated into the ROOT framework, and the other relevant packages have been separated from the AliRoot framework and can be used individually by any other HEP project. The general concept of the VMC and its set of base classes provided in ROOT will be presented. Existing implementations for Geant3, Geant4 and FLUKA and simple examples of usage will be described.
Energy Technology Data Exchange (ETDEWEB)
Jones, Bernard L; Cho, Sang Hyun, E-mail: scho@gatech.edu [Nuclear/Radiological Engineering and Medical Physics Programs, Woodruff School of Mechanical Engineering, Georgia Institute of Technology, Atlanta, GA 30332-0405 (United States)
2011-06-21
A recent study investigated the feasibility to develop a bench-top x-ray fluorescence computed tomography (XFCT) system capable of determining the spatial distribution and concentration of gold nanoparticles (GNPs) in vivo using a diagnostic energy range polychromatic (i.e. 110 kVp) pencil-beam source. In this follow-up study, we examined the feasibility of a polychromatic cone-beam implementation of XFCT by Monte Carlo (MC) simulations using the MCNP5 code. In the current MC model, cylindrical columns with various sizes (5-10 mm in diameter) containing water loaded with GNPs (0.1-2% gold by weight) were inserted into a 5 cm diameter cylindrical polymethyl methacrylate phantom. The phantom was then irradiated by a lead-filtered 110 kVp x-ray source, and the resulting gold fluorescence and Compton-scattered photons were collected by a series of energy-sensitive tallies after passing through lead parallel-hole collimators. A maximum-likelihood iterative reconstruction algorithm was implemented to reconstruct the image of GNP-loaded objects within the phantom. The effects of attenuation of both the primary beam through the phantom and the gold fluorescence photons en route to the detector were corrected during the image reconstruction. Accurate images of the GNP-containing phantom were successfully reconstructed for three different phantom configurations, with both spatial distribution and relative concentration of GNPs well identified. The pixel intensity of regions containing GNPs was linearly proportional to the gold concentration. The current MC study strongly suggests the possibility of developing a bench-top, polychromatic, cone-beam XFCT system for in vivo imaging.
Jones, Bernard L.; Cho, Sang Hyun
2011-06-01
A recent study investigated the feasibility to develop a bench-top x-ray fluorescence computed tomography (XFCT) system capable of determining the spatial distribution and concentration of gold nanoparticles (GNPs) in vivo using a diagnostic energy range polychromatic (i.e. 110 kVp) pencil-beam source. In this follow-up study, we examined the feasibility of a polychromatic cone-beam implementation of XFCT by Monte Carlo (MC) simulations using the MCNP5 code. In the current MC model, cylindrical columns with various sizes (5-10 mm in diameter) containing water loaded with GNPs (0.1-2% gold by weight) were inserted into a 5 cm diameter cylindrical polymethyl methacrylate phantom. The phantom was then irradiated by a lead-filtered 110 kVp x-ray source, and the resulting gold fluorescence and Compton-scattered photons were collected by a series of energy-sensitive tallies after passing through lead parallel-hole collimators. A maximum-likelihood iterative reconstruction algorithm was implemented to reconstruct the image of GNP-loaded objects within the phantom. The effects of attenuation of both the primary beam through the phantom and the gold fluorescence photons en route to the detector were corrected during the image reconstruction. Accurate images of the GNP-containing phantom were successfully reconstructed for three different phantom configurations, with both spatial distribution and relative concentration of GNPs well identified. The pixel intensity of regions containing GNPs was linearly proportional to the gold concentration. The current MC study strongly suggests the possibility of developing a bench-top, polychromatic, cone-beam XFCT system for in vivo imaging.
Miklaszewski, R.; Wiącek, U.; Dworak, D.; Drozdowicz, K.; Gribkov, V.
2012-07-01
Recent progress in the development of a Nanosecond Impulse Neutron Investigation System (NINIS) intended for interrogation of hidden objects (explosives and other illicit materials) by means of measuring elastically and non-elastically scattered neutrons is presented. The method uses very bright neutron pulses having durations of the order of few nanoseconds, generated by a dense plasma focus (DPF) devices filled with pure deuterium or a deuterium-tritium mixture as a working gas. A very short duration of the neutron pulse, as well as its high brightness and mono-chromaticity allows using time-of-flight methods with bases of about few meters to distinguish signals from neutrons scattered by different elements. Results of the Monte Carlo simulations of the scattered neutron field from several compounds (explosives and everyday use materials) are presented. The MCNP5 code has been used to get information on the angular and energy distributions of neutrons scattered by the above mentioned compounds assuming the initial neutron energies to be equal to 2.45 MeV (DD) and 14 MeV (DT). A new input has been elaborated that allows modeling not only a spectrum of the neutrons scattered at different angles but also their time history from the moment of generation up to the detection. Such an approach allows getting approximate signals registered by hypothetic scintillator + photomultipler probes placed at various distances from the scattering object, demonstrating principal capability of the method to identify an elemental content of the inspected objects. The extensive computations reveled also several limitations of the proposed method, namely: low number of neutrons reaching detector system, distortions and interferences of scattered neutron signals etc. Further more, preliminary results of the MCNP modeling of the hidden fissile materials detection process are presented.
Energy Technology Data Exchange (ETDEWEB)
Rivard, Mark J.; Granero, Domingo; Perez-Calatayud, Jose; Ballester, Facundo [Department of Radiation Oncology, Tufts University School of Medicine, Boston, Massachusetts 02111 (United States); Department of Radiation Oncology, ERESA, Hospital General Universitario, E-46014 Valencia (Spain); Department of Radiation Oncology, La Fe University Hospital, E-46009 Valencia (Spain); Department of Atomic, Molecular, and Nuclear Physics, University of Valencia, E-46100 Burjassot, Spain and IFIC, CSIC-University of Valencia, E-46100 Burjassot (Spain)
2010-02-15
Purpose: For a given radionuclide, there are several photon spectrum choices available to dosimetry investigators for simulating the radiation emissions from brachytherapy sources. This study examines the dosimetric influence of selecting the spectra for {sup 192}Ir, {sup 125}I, and {sup 103}Pd on the final estimations of kerma and dose. Methods: For {sup 192}Ir, {sup 125}I, and {sup 103}Pd, the authors considered from two to five published spectra. Spherical sources approximating common brachytherapy sources were assessed. Kerma and dose results from GEANT4, MCNP5, and PENELOPE-2008 were compared for water and air. The dosimetric influence of {sup 192}Ir, {sup 125}I, and {sup 103}Pd spectral choice was determined. Results: For the spectra considered, there were no statistically significant differences between kerma or dose results based on Monte Carlo code choice when using the same spectrum. Water-kerma differences of about 2%, 2%, and 0.7% were observed due to spectrum choice for {sup 192}Ir, {sup 125}I, and {sup 103}Pd, respectively (independent of radial distance), when accounting for photon yield per Bq. Similar differences were observed for air-kerma rate. However, their ratio (as used in the dose-rate constant) did not significantly change when the various photon spectra were selected because the differences compensated each other when dividing dose rate by air-kerma strength. Conclusions: Given the standardization of radionuclide data available from the National Nuclear Data Center (NNDC) and the rigorous infrastructure for performing and maintaining the data set evaluations, NNDC spectra are suggested for brachytherapy simulations in medical physics applications.
2014-03-27
want to express my sincere love, respect, and admiration for my wife, who motivated and supported me throughout this long endeavor; this document ...widely utilized radiation transport code is MCNP. First created at Los Alamos National Laboratory ( LANL ) in 1957, the code simulated neutral...explanation of the current capabilities of MCNP will occur within the next chapter of this document ; however, it is important to note that MCNP
Energy Technology Data Exchange (ETDEWEB)
Rojas C, E. L. [ININ, Carretera Mexico-Toluca s/n, Ocoyoacac 52750, Estado de Mexico (Mexico)
2008-07-01
The objective of this study is to investigate the changes observed in the absorbed doses in mammary gland tissue when irradiated with a equipment of high dose rate known as Mammosite and introducing material resources contrary to the tissue that constitutes the mammary gland. The modeling study is performed with the code MCNPX, 2005 version, the equipment and the mammary gland and calculating the absorbed doses in tissue when introduced small volumes of air or calcium in the system. (Author)
Energy Technology Data Exchange (ETDEWEB)
Meireles, Ramiro Conceicao
2016-07-01
The shielding calculation methodology for radiotherapy services adopted in Brazil and in several countries is that described in publication 151 of the National Council on Radiation Protection and Measurements (NCRP 151). This methodology however, markedly employs several approaches that can impact both in the construction cost and in the radiological safety of the facility. Although this methodology is currently well established by the high level of use, some parameters employed in the calculation methodology did not undergo to a detailed assessment to evaluate the impact of the various approaches considered. In this work the MCNP5 Monte Carlo code was used with the purpose of evaluating the above mentioned approaches. TVLs values were obtained for photons in conventional concrete (2.35g / cm{sup 3}), considering the energies of 6, 10 and 25 MeV, respectively, first considering an isotropic radiation source impinging perpendicular to the barriers, and subsequently a lead head shielding emitting a shaped beam, in the format of a pyramid trunk. Primary barriers safety margins, taking in account the head shielding emitting photon beam pyramid-shaped in the energies of 6, 10, 15 and 18 MeV were assessed. A study was conducted considering the attenuation provided by the patient's body in the energies of 6,10, 15 and 18 MeV, leading to new attenuation factors. Experimental measurements were performed in a real radiotherapy room, in order to map the leakage radiation emitted by the accelerator head shielding and the results obtained were employed in the Monte Carlo simulation, as well as to validate the entire study. The study results indicate that the TVLs values provided by (NCRP, 2005) show discrepancies in comparison with the values obtained by simulation and that there may be some barriers that are calculated with insufficient thickness. Furthermore, the simulation results show that the additional safety margins considered when calculating the width of the
Rühm, W; Pioch, C; Agosteo, S; Endo, A; Ferrarini, M; Rakhno, I; Rollet, S; Satoh, D; Vincke, H
2014-01-01
Bonner Spheres Spectrometry in its high-energy extended version is an established method to quantify neutrons at a wide energy range from several meV up to more than 1 GeV. In order to allow for quantitative measurements, the responses of the various spheres used in a Bonner Sphere Spectrometer (BSS) are usually simulated by Monte Carlo (MC) codes over the neutron energy range of interest. Because above 20 MeV experimental cross section data are scarce, intra-nuclear cascade (INC) and evaporation models are applied in these MC codes. It was suspected that this lack of data above 20 MeV may translate to differences in simulated BSS response functions depending on the MC code and nuclear models used, which in turn may add to the uncertainty involved in Bonner Sphere Spectrometry, in particular for neutron energies above 20 MeV. In order to investigate this issue in a systematic way, EURADOS (European Radiation Dosimetry Group) initiated an exercise where six groups having experience in neutron transport calcula...
Nelson, Benjamin E; Payne, Matthew J
2013-01-01
In the 20+ years of Doppler observations of stars, scientists have uncovered a diverse population of extrasolar multi-planet systems. A common technique for characterizing the orbital elements of these planets is Markov chain Monte Carlo (MCMC), using a Keplerian model with random walk proposals and paired with the Metropolis-Hastings algorithm. For approximately a couple of dozen planetary systems with Doppler observations, there are strong planet-planet interactions due to the system being in or near a mean-motion resonance (MMR). An N-body model is often required to accurately describe these systems. Further computational difficulties arise from exploring a high-dimensional parameter space ($\\sim$7 x number of planets) that can have complex parameter correlations. To surmount these challenges, we introduce a differential evolution MCMC (DEMCMC) applied to radial velocity data while incorporating self-consistent N-body integrations. Our Radial velocity Using N-body DEMCMC (RUN DMC) algorithm improves upon t...
Hermann, M; Vandoni, G; Kersevan, R
2013-01-01
The existing ISOLDE radio frequency quadrupole cooler and buncher (RFQCB) will be upgraded in the framework of the HIE-ISOLDE design study. In order to improve beam properties, the upgrade includes vacuum optimization with the aim of tayloring the overall pressure profile: increasing gas pressure at the injection to enhance cooling and reducing it at the extraction to avoid emittance blow up while the beam is being bunched. This paper describes the vacuum modelling of the present RFQCB using Test Particle Monte Carlo (Molflow+). In order to benchmark the simulation results, real pressure profiles along the existing RFQCB are measured using variable helium flux in the cooling section and compared with the pressure profiles obtained with Molflow+. Vacuum conditions of the improved future RFQCB can then be simulated to validate its design. (C) 2013 Elsevier B.V. All rights reserved.
Reddell, Brandon
2015-01-01
Designing hardware to operate in the space radiation environment is a very difficult and costly activity. Ground based particle accelerators can be used to test for exposure to the radiation environment, one species at a time, however, the actual space environment cannot be duplicated because of the range of energies and isotropic nature of space radiation. The FLUKA Monte Carlo code is an integrated physics package based at CERN that has been under development for the last 40+ years and includes the most up-to-date fundamental physics theory and particle physics data. This work presents an overview of FLUKA and how it has been used in conjunction with ground based radiation testing for NASA and improve our understanding of secondary particle environments resulting from the interaction of space radiation with matter.
Copeland, Kyle; Parker, Donald E; Friedberg, Wallace
2010-03-01
Conversion coefficients have been calculated for fluence-to-absorbed dose, fluence-to-effective dose and fluence-to-gray equivalent for isotropic exposure of an adult male and an adult female to (56)Fe(26+) in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). The coefficients were calculated using Monte Carlo transport code MCNPX 2.7.A and BodyBuilder 1.3 anthropomorphic phantoms modified to allow calculation of effective dose using tissues and tissue weighting factors from either the 1990 or 2007 recommendations of the International Commission on Radiological Protection (ICRP) and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. Calculations using ICRP 2007 recommendations result in fluence-to-effective dose conversion coefficients that are almost identical at most energies to those calculated using ICRP 1990 recommendations.
Copeland, Kyle; Parker, Donald E; Friedberg, Wallace
2010-03-01
Conversion coefficients have been calculated for fluence to absorbed dose, fluence to effective dose and fluence to gray equivalent, for isotropic exposure to alpha particles in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). The coefficients were calculated using Monte Carlo transport code MCNPX 2.7.A and BodyBuilder 1.3 anthropomorphic phantoms modified to allow calculation of effective dose to a Reference Person using tissues and tissue weighting factors from 1990 and 2007 recommendations of the International Commission on Radiological Protection (ICRP) and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. Coefficients for effective dose are within 30 % of those calculated using ICRP 1990 recommendations.
Energy Technology Data Exchange (ETDEWEB)
Rhee, J. T.; Jo, H. Y.; Jamil, M.; Jeon, Y. J. [Konkuk University, Seoul (Korea, Republic of)
2012-04-15
This article reports the simulated response to fast neutrons of a multi-gap resistive plate chamber (MRPC) by using the GEANT4 MC code. In this study, a thin polyethylene layer, which acted as the converter material for the detection of fast neutrons, was coated on the surface of the MRPC, which acts as the converter material for the detection of fast neutrons. The converter based on the polyethylene material improved the chamber's ability to detect fast neutrons. By employing the GEANT4 MC code, fast neutrons were inserted into the converter-based MRPC chamber in the energy range of 1.0 - 20.0 MeV. The response of the polyethylene-coated MRPC were evaluated as a function of the neutron energy by using the QGSP{sub B}ERT{sub H}P and the QGSP{sub B}IC{sub H}P physics list with the GEANT4 code. For a 0.13-mm converter thickness, a detection efficiency of 6.4x10{sup -3} were found for fast neutrons with an energy of E{sub n} = 6.0 by the QGSP{sub B}ERT{sub H}P physics list. The simulation test further confirmed that a higher response of the fast neutrons could be achieved if the converter thickness were to be increased. A detailed outline of the simulation test and the obtained results are presented.
Yan, Qiang; Shao, Lin
2017-03-01
Current popular Monte Carlo simulation codes for simulating electron bombardment in solids focus primarily on electron trajectories, instead of electron-induced displacements. Here we report a Monte Carol simulation code, DEEPER (damage creation and particle transport in matter), developed for calculating 3-D distributions of displacements produced by electrons of incident energies up to 900 MeV. Electron elastic scattering is calculated by using full-Mott cross sections for high accuracy, and primary-knock-on-atoms (PKAs)-induced damage cascades are modeled using ZBL potential. We compare and show large differences in 3-D distributions of displacements and electrons in electron-irradiated Fe. The distributions of total displacements are similar to that of PKAs at low electron energies. But they are substantially different for higher energy electrons due to the shifting of PKA energy spectra towards higher energies. The study is important to evaluate electron-induced radiation damage, for the applications using high flux electron beams to intentionally introduce defects and using an electron analysis beam for microstructural characterization of nuclear materials.
STUDI PEMODELAN DAN PERHITUNGAN TRANSPORT MONTE CARLO DALAM TERAS HTR PEBBLE BED
Directory of Open Access Journals (Sweden)
Zuhair .
2013-01-01
Full Text Available Konsep sistem energi VHTR baik yang berbahan bakar pebble (VHTR pebble bed maupun blok prismatik (VHTR prismatik menarik perhatian fisikawan reaktor nuklir. Salah satu kelebihan teknologi bahan bakar bola adalah menawarkan terobosan teknologi pengisian bahan bakar tanpa harus menghentikan produksi listrik. Selain itu, partikel bahan bakar pebble dengan kernel uranium oksida (UO2 atau uranium oksikarbida (UCO yang dibalut TRISO dan pelapisan silikon karbida (SiC dianggap sebagai opsi utama dengan pertimbangan performa tinggi pada burn-up bahan bakar dan temperatur tinggi. Makalah ini mendiskusikan pemodelan dan perhitungan transport Monte Carlo dalam teras HTR pebble bed. HTR pebble bed adalah reaktor berpendingin gas temperatur tinggi dan bermoderator grafit dengan kemampuan kogenerasi. Perhitungan dikerjakan dengan program MCNP5 pada temperatur 1200 K. Pustaka data nuklir energi kontinu ENDF/B-V dan ENDF/B-VI dimanfaatkan untuk melengkapi analisis. Hasil perhitungan secara keseluruhan menunjukkan konsistensi dengan nilai keff yang hampir sama untuk pustaka data nuklir yang digunakan. Pustaka ENDF/B-VI (66c selalu memproduksi keff lebih besar dibandingkan ENDF/B-V (50c maupun ENDF/B-VI (60c dengan bias kurang dari 0,25%. Kisi BCC memprediksi keff hampir selalu lebih kecil daripada kisi lainnya, khususnya FCC. Nilai keff kisi BCC lebih dekat dengan kisi FCC dengan bias kurang dari 0,19% sedangkan dengan kisi SH bias perhitungannya kurang dari 0,22%. Fraksi packing yang sedikit berbeda (BCC= 61%, SH= 60,459% tidak membuat bias perhitungan menjadi berbeda jauh. Estimasi keff ketiga model kisi menyimpulkan bahwa model BCC lebih bisa diadopsi dalam perhitungan HTR pebble bed dibandingkan model FCC dan SH. Verifikasi hasil estimasi ini perlu dilakukan dengan simulasi Monte Carlo atau bahkan program deterministik lainnya guna optimisasi perhitungan teras reaktor temperatur tinggi. Kata-kunci: kernel, TRISO, bahan bakar pebble, HTR pebble bed
Institute of Scientific and Technical Information of China (English)
刘丽鹃; 李圆圆; 李建
2013-01-01
在反应堆上进行中子残余应力谱仪(RSND)的建设工程中,生物屏蔽是降低谱仪测量本底,保证工作人员安全的主要技术手段.为了对屏蔽系统进行优化设计,采用蒙特卡罗方法对其进行模拟计算.采用McStas和MCNP5两种计算程序,理论分析提高计算效率的方法和技巧,并分三步进行谱仪辐射场的计算.首先用McStas程序模拟热中子在导管中的输运过程,然后用MCNP程序计算了屏蔽块之间接合处的拼缝对谱仪辐射场的影响,最后建立整个谱仪屏蔽系统的MCNP模型,并进行计算.将经过用影响因素修正后的计算结果与现场剂量仪实测值进行比较,结果表明二者在允许的误差范围内基本一致.
Institute of Scientific and Technical Information of China (English)
武祥; 若夕子; 于涛; 谢金森; 陈昊威
2014-01-01
TRITON couples multi group Monte Carlo Transport code KENO V. a and point-burnup code ORIGEN-S. It features adaptability on complex geometries,flexible processing ability on cross section and rapid calculating speed. Based on the thorium-based fuel cell benchmark of Idaho National Laboratory ( INL) ,the verification on TRITON burnup calcu-lation was performed,which showed good coincidence with the result of MOCUP code by INL. Furthermore, the results of burnup isotopes selection schemes in TRITON showed that,for thorium based fuel,only important nuclides on Th-U cycle was included,correct results can be obtained by TRITON. Conclusions in the present paper will support further applications of TRITON.%TRITON程序系统耦合了多群蒙特卡罗输运程序KENO V. a与点燃耗程序ORIGEN-S,具有几何适应性强、截面处理能力灵活、计算速度快等显著特点.本文基于爱达荷国家实验室( INL)钍基燃料元件燃耗基准题,开展了TRITON程序燃耗功能的验证,结果与INL采用MOCUP程序给出的结果吻合很好.同时,燃耗核素选取对TRITON计算结果的影响分析表明对于钍基燃料,只有在考虑Th-U循环重要核素的前提下,TRITON才能给出正确结果.上述结论为TRITON程序的应用奠定了基础.
Energy Technology Data Exchange (ETDEWEB)
Moskvin, V; Tsiamas, P; Axente, M; Farr, J [St. Jude Children’s Research Hospital, Memphis, TN (United States); Stewart, R [University of Washington, Seattle, WA. (United States)
2015-06-15
Purpose: One of the more critical initiating events for reproductive cell death is the creation of a DNA double strand break (DSB). In this study, we present a computationally efficient way to determine spatial variations in the relative biological effectiveness (RBE) of proton therapy beams within the FLUKA Monte Carlo (MC) code. Methods: We used the independently tested Monte Carlo Damage Simulation (MCDS) developed by Stewart and colleagues (Radiat. Res. 176, 587–602 2011) to estimate the RBE for DSB induction of monoenergetic protons, tritium, deuterium, hellium-3, hellium-4 ions and delta-electrons. The dose-weighted (RBE) coefficients were incorporated into FLUKA to determine the equivalent {sup 6}°60Co γ-ray dose for representative proton beams incident on cells in an aerobic and anoxic environment. Results: We found that the proton beam RBE for DSB induction at the tip of the Bragg peak, including primary and secondary particles, is close to 1.2. Furthermore, the RBE increases laterally to the beam axis at the area of Bragg peak. At the distal edge, the RBE is in the range from 1.3–1.4 for cells irradiated under aerobic conditions and may be as large as 1.5–1.8 for cells irradiated under anoxic conditions. Across the plateau region, the recorded RBE for DSB induction is 1.02 for aerobic cells and 1.05 for cells irradiated under anoxic conditions. The contribution to total effective dose from secondary heavy ions decreases with depth and is higher at shallow depths (e.g., at the surface of the skin). Conclusion: Multiscale simulation of the RBE for DSB induction provides useful insights into spatial variations in proton RBE within pristine Bragg peaks. This methodology is potentially useful for the biological optimization of proton therapy for the treatment of cancer. The study highlights the need to incorporate spatial variations in proton RBE into proton therapy treatment plans.
Directory of Open Access Journals (Sweden)
Asghar Mesbahi
2015-09-01
Full Text Available Introduction Radiotherapy with small fields is used widely in newly developed techniques. Additionally, dose calculation accuracy of treatment planning systems in small fields plays a crucial role in treatment outcome. In the present study, dose calculation accuracy of two commercial treatment planning systems was evaluated against Monte Carlo method. Materials and Methods Siemens Once or linear accelerator was simulated, using MCNPX Monte Carlo code, according to manufacturer’s instructions. Three analytical algorithms for dose calculation including full scatter convolution (FSC in TiGRT, along with convolution and superposition in XiO system were evaluated for a small solid liver tumor. This solid tumor with a diameter of 1.8 cm was evaluated in a thorax phantom, and calculations were performed for different field sizes (1×1, 2×2, 3×3 and4×4 cm2. The results obtained in these treatment planning systems were compared with calculations by MC method (regarded as the most reliable method. Results For FSC and convolution algorithm, comparison with MC calculations indicated dose overestimations of up to 120%and 25% inside the lung and tumor, respectively in 1×1 cm2field size, using an 18 MV photon beam. Regarding superposition, a close agreement was seen with MC simulation in all studied field sizes. Conclusion The obtained results showed that FSC and convolution algorithm significantly overestimated doses of the lung and solid tumor; therefore, significant errors could arise in treatment plans of lung region, thus affecting the treatment outcomes. Therefore, use of MC-based methods and super position is recommended for lung treatments, using small fields and beamlets.
Monte Carlo Particle Lists: MCPL
Kittelmann, Thomas; Knudsen, Erik B; Willendrup, Peter; Cai, Xiao Xiao; Kanaki, Kalliopi
2016-01-01
A binary format with lists of particle state information, for interchanging particles between various Monte Carlo simulation applications, is presented. Portable C code for file manipulation is made available to the scientific community, along with converters and plugins for several popular simulation packages.
Energy Technology Data Exchange (ETDEWEB)
Murata, Isao; Miyamaru, Hiroyuki [Division of Electrical, Electronic and Information Engineering, Osaka University, Yamada-oka 2-1, Suita, Osaka, 565-0871 (Japan)
2008-07-01
Spherical elements have remarkable features in various applications in the nuclear engineering field. In 1990's, by the project of HTR-PROTEUS at PSI various pebble bed reactor experiments were conducted including cores with a lot of spherical fuel elements loaded randomly. In this study, criticality experiments of the random-loading HTR-PROTEUS cores were analyzed by MCNP-BALL, which could deal with a random arrangement of spherical fuel elements exactly with a statistical geometry model. As a result of analysis, the calculated effective multiplication factors were in fairly good agreement with the measurements within about 0.5%DELTAk/k. In comparison with other numerical analysis, our effective multiplication factors were between the experimental values and the VSOP calculations. To investigate the discrepancy of the effective multiplication factors between the experiments and calculations, sensitivity analyses were performed. As the result, the sensitivity of impurity boron concentration was fairly large. The reason of the present slight overestimation was not made clear at present. However, the presently existing difference was thought to be related to the impurity boron concentration, not to the modelling of the reactor and the used nuclear data. From the present study, it was confirmed that MCNP-BALL would have an advantage to conventional transport codes by comparing with their numerical results and the experimental values. As for the criticality experiment of PROTEUS, we would conclude that the two cores of Core 4.2 and 4.3 could be regarded as an equivalent experiment of a reference critical core, which was packed in the packing fraction of RLP. (authors)
Energy Technology Data Exchange (ETDEWEB)
Pazianotto, Mauricio Tizziani; Goncalez, Odair Lelis; Federico, Claudio Antonio [Centro Tecnico Aeroespacial (IEAv/CTA), Sao Jose dos Campos, SP (Brazil). Inst. de Estudos Avancados; Carlson, Brett Vern [Centro Tecnico Aeroespacial (ITA/CTA), Sao Jose dos Campos, SP (Brazil). Inst. Tecnologico de Aeronautica
2010-07-01
Full text: The Institute for Advanced Studies (IEAv) is developing activities to study the dose levels of ionizing radiation from cosmic rays (CR) received by aircraft crews, sensitive equipment (on-board computers, for example) and embedded electronics in Brazilian airspace. Neutrons generated by the interaction of CR with the atmosphere are the dominant particles in the dose accumulation in electronic circuits and aircraft crews at flight altitude. Their production has a very broad energy spectrum, ranging from thermal neutrons (0.025eV ) to neutrons of several hundreds of MeV , making their detection a very difficult process. To observe the temporal variation in flow during the measurements, a detector of the Long Counter (LC) type is being used. This detector is designed to measure the one-way flow of neutrons with constant response over a wide energy range (thermal to 20 MeV ). However, to measure cosmic rays, the flow of which is non-directional, the dependence of the response on the angle of incidence, as well as energy, should be properly investigated. The objective of this study is to assess the angular response of the neutron detector (Long Counter) using the code MCNP5 (Monte Carlo N-Particle) and to compare it with the experimental data previously obtained with a {sup 241}Am-Be source at a distance of 1.66 m from the geometric center of the detector, varying the angle of incidence from 00 to 3600 in intervals of 150. The simulation was performed by modeling in detail the structure and materials of the LC, as well as the experimental arrangement for irradiation. The results of the simulation present reasonable agreement with the experimental data. This agreement shows that the modeling of the geometry of the source-detector system is adequate. The next step is to develop a model of neutron detection for the higher energy present in cosmic radiation fields, for which the experimental calibration is not so easily achievable. (author)
Aygun, Bünyamin; Korkut, Turgay; Karabulut, Abdulhalik
2016-05-01
Despite the possibility of depletion of fossil fuels increasing energy needs the use of radiation tends to increase. Recently the security-focused debate about planned nuclear power plants still continues. The objective of this thesis is to prevent the radiation spread from nuclear reactors into the environment. In order to do this, we produced higher performanced of new shielding materials which are high radiation holders in reactors operation. Some additives used in new shielding materials; some of iron (Fe), rhenium (Re), nickel (Ni), chromium (Cr), boron (B), copper (Cu), tungsten (W), tantalum (Ta), boron carbide (B4C). The results of this experiments indicated that these materials are good shields against gamma and neutrons. The powder metallurgy technique was used to produce new shielding materials. CERN - FLUKA Geant4 Monte Carlo simulation code and WinXCom were used for determination of the percentages of high temperature resistant and high-level fast neutron and gamma shielding materials participated components. Super alloys was produced and then the experimental fast neutron dose equivalent measurements and gamma radiation absorpsion of the new shielding materials were carried out. The produced products to be used safely reactors not only in nuclear medicine, in the treatment room, for the storage of nuclear waste, nuclear research laboratories, against cosmic radiation in space vehicles and has the qualities.
Copeland, Kyle; Parker, Donald E; Friedberg, Wallace
2011-01-01
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent for isotropic exposure of an adult female and an adult male to deuterons ((2)H(+)) in the energy range 10 MeV-1 TeV (0.01-1000 GeV). Coefficients were calculated using the Monte Carlo transport code MCNPX 2.7.C and BodyBuilder™ 1.3 anthropomorphic phantoms. Phantoms were modified to allow calculation of the effective dose to a Reference Person using tissues and tissue weighting factors from 1990 and 2007 recommendations of the International Commission on Radiological Protection (ICRP) and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. Coefficients for the equivalent and effective dose incorporated a radiation weighting factor of 2. At 15 of 19 energies for which coefficients for the effective dose were calculated, coefficients based on ICRP 1990 and 2007 recommendations differed by <3%. The greatest difference, 47%, occurred at 30 MeV.
Copeland, Kyle; Parker, Donald E; Friedberg, Wallace
2010-12-01
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent for isotropic exposure of an adult female and an adult male to tritons ((3)H(+)) in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). Coefficients were calculated using Monte Carlo transport code MCNPX 2.7.C and BodyBuilder™ 1.3 anthropomorphic phantoms. Phantoms were modified to allow calculation of effective dose to a Reference Person using tissues and tissue weighting factors from 1990 and 2007 recommendations of the International Commission on Radiological Protection (ICRP) and calculation of gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. At 15 of the 19 energies for which coefficients for effective dose were calculated, coefficients based on ICRP 2007 and 1990 recommendations differed by less than 3%. The greatest difference, 43%, occurred at 30 MeV.
Copeland, Kyle; Parker, Donald E; Friedberg, Wallace
2010-12-01
Conversion coefficients were calculated for fluence-to-absorbed dose, fluence-to-equivalent dose, fluence-to-effective dose and fluence-to-gray equivalent, for isotropic exposure of an adult male and an adult female to helions ((3)He(2+)) in the energy range of 10 MeV to 1 TeV (0.01-1000 GeV). Calculations were performed using Monte Carlo transport code MCNPX 2.7.C and BodyBuilder™ 1.3 anthropomorphic phantoms modified to allow calculation of effective dose using tissues and tissue weighting factors from either the 1990 or 2007 recommendations of the International Commission on Radiological Protection (ICRP), and gray equivalent to selected tissues as recommended by the National Council on Radiation Protection and Measurements. At 15 of the 19 energies for which coefficients for effective dose were calculated, coefficients based on ICRP 2007 and 1990 recommendations differed by less than 2%. The greatest difference, 62%, occurred at 100 MeV.
Fast quantum Monte Carlo on a GPU
Lutsyshyn, Y
2013-01-01
We present a scheme for the parallelization of quantum Monte Carlo on graphical processing units, focusing on bosonic systems and variational Monte Carlo. We use asynchronous execution schemes with shared memory persistence, and obtain an excellent acceleration. Comparing with single core execution, GPU-accelerated code runs over x100 faster. The CUDA code is provided along with the package that is necessary to execute variational Monte Carlo for a system representing liquid helium-4. The program was benchmarked on several models of Nvidia GPU, including Fermi GTX560 and M2090, and the latest Kepler architecture K20 GPU. Kepler-specific optimization is discussed.
Energy Technology Data Exchange (ETDEWEB)
Barbosa, Nilseia A.; Rosa, Luiz A. Ribeiro da, E-mail: nilseia@ird.gov.br, E-mail: lrosa@ird.gov.br [Instituto de Radioprotecao e Dosimetria (IRD/CNEN-RJ),Rio de Janeiro, RJ (Brazil); Braz, Delson, E-mail: delson@nuclear.ufrj.br [Coordenacao dos programas de Pos-Graduacao em Engenharia (PEN/COPPE/UFRJ), Rio de Janeiro, RJ (Brazil). Programa de Engenharia Nuclear
2014-07-01
The COC ophthalmic applicators using beta radiation source of {sup 106}Ru/{sup 106}Rh are used in the treatment of intraocular tumors near the optic nerve. In this type of treatment is very important to know the dose distribution in order to provide the best possible delivery of prescribed dose to the tumor, preserves the optic nerve region extremely critical, that if damaged, can compromise the patient's visual acuity, and cause brain sequelae. These dose distributions are complex and doctors, who will have the responsibility on the therapy, only have the source calibration certificate provided by the manufacturer Eckert and Ziegler BEBIG GmbH. These certificates provide 10 absorbed dose values at water depth along the central axis applicator with the uncertainties of the order of 20% isodose and in a plane located 1 mm from the applicator surface. Thus, it is important to know with more detail and precision the dose distributions in water generated by such applicators. To this end, the Monte Carlo simulation was used using MCNPX code. Initially, was validated the simulation by comparing the obtained results to the central axis of the applicator with those provided by the certificate. The different percentages were lower than 5%, validating the used method. Lateral dose profile was calculated for 6 different depths in intervals of 1 mm and the dose rates in mGy.min{sup -1} for the same depths.
Energy Technology Data Exchange (ETDEWEB)
Noelle, P
2006-12-15
In vivo lung counting, one of the preferred methods for monitoring people exposed to the risk of actinide inhalation, is nevertheless limited by the use of physical calibration phantoms which, for technical reasons, can only provide a rough representation of human tissue. A new approach to in vivo measurements has been developed to take advantage of advances in medical imaging and computing; this consists of numerical phantoms based on tomographic images (CT) or magnetic resonance images (R.M.I.) combined with Monte Carlo computing techniques. Under laboratory implementation of this innovative method using specific software called O.E.D.I.P.E., the main thrust of this thesis was to provide answers to the following question: what do numerical phantoms and new techniques like O.E.D.I.P.E. contribute to the improvement in calibration of low-energy in vivo counting systems? After a few developments of the O.E.D.I.P.E. interface, the numerical method was validated for systems composed of four germanium detectors, the most widespread configuration in radio bioassay laboratories (a good match was found, with less than 10% variation). This study represents the first step towards a person-specific numerical calibration of counting systems, which will improve assessment of the activity retained. A second stage focusing on an exhaustive evaluation of uncertainties encountered in in vivo lung counting was possible thanks to the approach offered by the previously-validated O.E.D.I.P.E. software. It was shown that the uncertainties suggested by experiments in a previous study were underestimated, notably morphological differences between the physical phantom and the measured person. Some improvements in the measurement procedure were then proposed, particularly new bio-metric equations specific to French measurement configurations that allow a more sensible choice of the calibration phantom, directly assessing the thickness of the torso plate to be added to the Livermore phantom
Institute of Scientific and Technical Information of China (English)
张坤明; 张雄杰; 瞿金辉; 汤彬
2015-01-01
利用MCNP程序模拟研究脉冲中子－裂变中子探测铀黄饼，采用脉冲式中子源，利用氦三管中子探测器记录裂变中子，得到铀黄饼中的铀含量信息。通过对14 MeV脉冲中子源和产生的裂变中子在不同铀含量模型中的输运计算，分析了裂变中子与铀含量的关系。结果表明：利用裂变超热中子衰减时间谱，可以确定铀黄饼中的铀含量；通过对热中子衰减时间谱进行校正，可以提高铀黄饼中铀含量计算结果的准确度。%The Monte Carlo N particle transport code ( MCNP ) is used to simulate how to explore the uranium yel⁃lowcake by using the pulsed neutron⁃fission neutron ( PNFN) method. In order to obtain uranium yellowcake quan⁃titation, pulsed neutron source was used, prompt fission neutrons were detected by using the neutron detector. Un⁃der the condition of different uranium quantitation models, the transport of the 14 MeV pulsed neutron source and the released fission neutron were calculated. On the basis of these, the relationship between fission neutron and ura⁃nium quantitation was studied. The results show that using the epithermal neutron time decay spectrum, the urani⁃um yellowcake quantitation can be determined; the precision of the uranium yellowcake quantitation could be in⁃creased by the correction of thermal neutron time decay spectrum.
Institute of Scientific and Technical Information of China (English)
贾清刚; 张天奎; 张凤娜; 胡华四
2013-01-01
开发了基于Geant4的Z箍缩中子编码成像系统模拟平台,实现聚变中子编码成像诊断系统各关键部件的完整模拟.获得了低中子产额(约1010量级)下,中子经编码孔编码后在闪烁体阵列中形成的发光分布图像.利用维纳滤波、Richardson-Lucy(RL)及遗传算法(GA)对低中子产额下获得的极低信噪比图像进行重建,并对信噪比、中子产额及重建效果进行了对比研究,结果表明:遗传算法对低信噪比中子编码图像的重建具有很强的鲁棒性;中子编码图像的信噪比与遗传算法重建结果的准确性呈正比.%The model of Z-pinch driven fusion imaging diagnosis system was set up by a Monte Carlo code based on the Geant4 simulation toolkit. All physical processes that the reality involves are taken into consideration in simulation. The light image of low neutron yield (about 1010) pill was obtained. Three types of image reconstruction algorithm, i. e. Richardson-Lucy, Wiener filtering and genetic algorithm were employed to reconstruct the neutron image with a low signal to noise ratio (SNR) and yield. The effects of neutron yields and the SNR on reconstruction performance were discussed. The results show that genetic algorithm is very robust for reconstructing neutron images with a low SNR. And the index of reconstruction performance and the image correlation coefficient using genetic algorithm, are proportional to the SNR of the neutron coded image.
Energy Technology Data Exchange (ETDEWEB)
De Carlan, L.; Bordy, J.M.; Gouriou, J. [CEA Saclay, LIST, Laboratoire National Henri Becquerel, Laboratoire de Metrologie de la Dose 91 - Gif-sur-Yvette (France)
2010-07-01
Within the frame of the CONRAD European project (Coordination Network for Radiation Dosimetry), and more precisely within a work group paying attention to uncertainty assessment in computational dosimetry and aiming at comparing different approaches, the authors report the simulation of an irradiator containing a caesium 137 source to calculate the kerma in air as well as its uncertainty due to different parameters. They present the problem geometry, recall the studied issues (kerma uncertainty, influence of capsule source, influence of the collimator, influence of the air volume surrounding the source). They indicate the codes which have been used (MNCP, Fluka, Penelope, etc.) and discuss the obtained results for the first issue
The impact of Monte Carlo simulation: a scientometric analysis of scholarly literature
Pia, Maria Grazia; Bell, Zane W; Dressendorfer, Paul V
2010-01-01
A scientometric analysis of Monte Carlo simulation and Monte Carlo codes has been performed over a set of representative scholarly journals related to radiation physics. The results of this study are reported and discussed. They document and quantitatively appraise the role of Monte Carlo methods and codes in scientific research and engineering applications.
Institute of Scientific and Technical Information of China (English)
汪量子; 姚栋; 王侃
2012-01-01
A method of using iterated fission probability to estimate the adjoint fluence during particles simulation, and using it as the weighting function to calculate kinetics parameters βell and A in Monte Carlo codes, was introduced in this paper. Implements of this method in continuous energy Monte Carlo code MCNP and multi-group Monte Carlo code MCMG are both elaborated. Verification results show that, with regardless additional computing cost, using this method, the adjoint fluence accounted by MCMG matches well with the result computed by ANISN, and the kinetics parameters calculated by MCNP agree very well with benchmarks. This method is proved to be reliable, and the function of calculating kinetics parameters in Monte Carlo codes is carried out effectively, which could be the basement for Monte Carlo codes' utility in the analysis of nuclear reactors' transient behavior.%文章介绍了在蒙特卡罗程序中,使用反复裂变几率的统计结果作为共轭通量的估计,并作为权重函数计算动力学参数βeff和Λ的方法,阐释了在连续能量蒙特卡罗程序MCNP和多群蒙特卡罗程序MCMG中实现这种方法的过程.数值校验结果表明:在几乎不带来附加计算量的同时,在MCMG中使用该方法统计得到的共轭通量与ANISN的共轭通量计算结果符合较好,在MCNP中使用该方法计算得到的中子动力学参数与基准测量结果符合较好.在蒙特卡罗程序中实现了高效率计算中子动力学参数的功能,为蒙特卡罗程序进一步用于反应堆动态行为的分析奠定了基础.
Ramamoorthy, Karthikeyan
and predominantly scattering isotopes. When the concentration of resonant isotopes is small, its presence does not affect the flux shape which is smooth. But when the concentration becomes high, there will be dips in the flux where resonances of the isotopes occur. This will affect the reaction rate, which is a product of cross section and flux. The reaction rate will thus be lower than that when one does not consider the flux dip. This is the phenomenon of self shielding. Self shielding treatment is thus a very important aspect of reactor lattice analysis code. This needs to be correctly modelled to obtain a physically sound and acceptable solution. In this research we will be looking into behaviour of the advanced self shielding models that have been incorporated in the code DRAGON Version4. The self shielding models are primarily classified into two broad groups, which are based on "equivalence in dilution" and "subgroup approach". These self shielding models will be tested against a variety of lattices which include Canada Deuterium Uranium (CANDU-6), CANDU-New Generation (CANDU-NG), Light Water Reactor (LWR), and High Conversion Light Water Reactor (HCLWR). The fuel composition will vary from natural uranium oxide to enriched uranium oxide and plutonium-uranium mixed oxide (MOX). We will also consider the presence of strong neutron absorbers like gadolinium and dysprosium in the lattice. The coolant/moderator chosen for the analysis will be light water/heavy water or a combination. The lattice geometry will vary from square, hexagonal and annular. Thus a broad spectrum of lattices will be analysed to assess the behaviour of advanced self shielding models. The results obtained using DRAGON will be validated against that obtained using Monte Carlo code MCNP5. The reference solutions for all situations will be provided by MCNP5. The depletion behaviour of any lattice will depend on the power or flux normalization that is considered. In general the flux in various
Directory of Open Access Journals (Sweden)
Hammam Oktajianto
2014-12-01
Full Text Available Gas-cooled nuclear reactor is a Generation IV reactor which has been receiving significant attention due to many desired characteristics such as inherent safety, modularity, relatively low cost, short construction period, and easy financing. High temperature reactor (HTR pebble-bed as one of type of gas-cooled reactor concept is getting attention. In HTR pebble-bed design, radius and enrichment of the fuel kernel are the key parameter that can be chosen freely to determine the desired value of criticality. This paper models HTR pebble-bed 10 MW and determines an effective of enrichment and radius of the fuel (Kernel to get criticality value of reactor. The TRISO particle coated fuel particle which was modelled explicitly and distributed in the fuelled region of the fuel pebbles using a Simple-Cubic (SC lattice. The pebble-bed balls and moderator balls distributed in the core zone using a Body-Centred Cubic lattice with assumption of a fresh fuel by the fuel enrichment was 7-17% at 1% range and the size of the fuel radius was 175-300 µm at 25 µm ranges. The geometrical model of the full reactor is obtained by using lattice and universe facilities provided by MCNP4C. The details of model are discussed with necessary simplifications. Criticality calculations were conducted by Monte Carlo transport code MCNP4C and continuous energy nuclear data library ENDF/B-VI. From calculation results can be concluded that an effective of enrichment and radius of fuel (Kernel to achieve a critical condition was the enrichment of 15-17% at a radius of 200 µm, the enrichment of 13-17% at a radius of 225 µm, the enrichments of 12-15% at radius of 250 µm, the enrichments of 11-14% at a radius of 275 µm and the enrichment of 10-13% at a radius of 300 µm, so that the effective of enrichments and radii of fuel (Kernel can be considered in the HTR 10 MW. Keywords—MCNP4C, HTR, enrichment, radius, criticality
Directory of Open Access Journals (Sweden)
V.P. Bereznev
2015-10-01
An iterative solution process is used, including external iterations for the fission source and internal iterations for the scattering source. The paper presents the results of a cross-verification against the Monte Carlo MMK code [3] and on a model of the BN-800 reactor core.
Energy Technology Data Exchange (ETDEWEB)
Gilles, D
2005-07-01
This report is devoted to illustrate the power of a Monte Carlo (MC) simulation code to study the thermodynamical properties of a plasma, composed of classical point particles at thermodynamical equilibrium. Such simulations can help us to manage successfully the challenge of taking into account 'exactly' all classical correlations between particles due to density effects, unlike analytical or semi-analytical approaches, often restricted to low dense plasmas. MC simulations results allow to cover, for laser or astrophysical applications, a wide range of thermodynamical conditions from more dense (and correlated) to less dense ones (where potentials are long ranged type). Therefore Yukawa potentials, with a Thomas-Fermi temperature- and density-dependent screening length, are used to describe the effective ion-ion potentials. In this report we present two MC codes ('PDE' and 'PUCE') and applications performed with these codes in different fields (spectroscopy, opacity, equation of state). Some examples of them are discussed and illustrated at the end of the report. (author)
Vectorized Monte Carlo methods for reactor lattice analysis
Brown, F. B.
1984-01-01
Some of the new computational methods and equivalent mathematical representations of physics models used in the MCV code, a vectorized continuous-enery Monte Carlo code for use on the CYBER-205 computer are discussed. While the principal application of MCV is the neutronics analysis of repeating reactor lattices, the new methods used in MCV should be generally useful for vectorizing Monte Carlo for other applications. For background, a brief overview of the vector processing features of the CYBER-205 is included, followed by a discussion of the fundamentals of Monte Carlo vectorization. The physics models used in the MCV vectorized Monte Carlo code are then summarized. The new methods used in scattering analysis are presented along with details of several key, highly specialized computational routines. Finally, speedups relative to CDC-7600 scalar Monte Carlo are discussed.
Iba, Yukito
2000-01-01
``Extended Ensemble Monte Carlo''is a generic term that indicates a set of algorithms which are now popular in a variety of fields in physics and statistical information processing. Exchange Monte Carlo (Metropolis-Coupled Chain, Parallel Tempering), Simulated Tempering (Expanded Ensemble Monte Carlo), and Multicanonical Monte Carlo (Adaptive Umbrella Sampling) are typical members of this family. Here we give a cross-disciplinary survey of these algorithms with special emphasis on the great f...
Quantum Monte Carlo approaches for correlated systems
Becca, Federico
2017-01-01
Over the past several decades, computational approaches to studying strongly-interacting systems have become increasingly varied and sophisticated. This book provides a comprehensive introduction to state-of-the-art quantum Monte Carlo techniques relevant for applications in correlated systems. Providing a clear overview of variational wave functions, and featuring a detailed presentation of stochastic samplings including Markov chains and Langevin dynamics, which are developed into a discussion of Monte Carlo methods. The variational technique is described, from foundations to a detailed description of its algorithms. Further topics discussed include optimisation techniques, real-time dynamics and projection methods, including Green's function, reptation and auxiliary-field Monte Carlo, from basic definitions to advanced algorithms for efficient codes, and the book concludes with recent developments on the continuum space. Quantum Monte Carlo Approaches for Correlated Systems provides an extensive reference ...
Monte-Carlo Simulation for Shielding of Cold Neutron Triple-axis Spectrometer%冷中子三轴谱仪屏蔽体的蒙特卡罗模拟
Institute of Scientific and Technical Information of China (English)
宋建明; 尹延朋; 李建; 刘丽鹃; 王虹; 孙光爱
2013-01-01
The shielding of Cold neutron Triple-Axis Spectrometer( CTAS) is important for radiation safety of workers, and reduce the background of scattering hall as well as enhancing the ratio of signal-to-noise. In this study, Monte-Carlo simulation was performed to conduct the calculation on the shielding of CTAS. To increase the calculation efficiency, neutron source was obtained by using Mcstas code. The results indicate that, in the case of heavy concrete (density 4.6 g/cm3) with thickness of 350 mm for the shielding behind the monochromater, and heavy concrete (density 3.6 g/cm3) with thickness of 300 mm for the other monochromater shielding, as well as the heavy concrete (density 3.6 g/cm3) with thickness of 150 mm for biological shielding, the dose rate outside shielding may meet the requirement of national standard of China.%冷中子三轴谱仪(CTAS)的屏蔽体对于保障工作人员安全、降低散射大厅本底及提高信噪比具有重要的意义.采用蒙特卡罗程序MCNP5对谱仪各部分屏蔽体进行了计算,并结合Mcstas程序确定了CTAS入口处的中子源,大大提高了计算效率.经过模拟计算和优化表明:单色器后端使用厚350mm、密度4.6g/cm3的重混凝土,衔接屏蔽体使用厚300 mm、密度3.6 g/cm3的重混凝土,生物屏蔽采用厚150 mm、密度3.6g/cm3的重混凝土可保证屏蔽体外表面的剂量率满足散射大厅的剂量要求.
A Study of Neutronics Effects of the Spacer Grids in a Typical PWR via Monte Carlo Calculation
Directory of Open Access Journals (Sweden)
Xuan Bach Tran
2016-02-01
Full Text Available Spacer grids play an important role in maintaining the proper form of the fuel assembly structure and ensuring the safety of reactor core design. This study applies the Monte Carlo method to the analysis of the neutronics effects of spacer grids in a typical pressurized water reactor (PWR. The core problem used to analyze the neutronics effects of spacer grids is a modified version of Korea Advanced Institute of Science and Technology benchmark problem 1B, based on an Advanced Power Reactor 1400 (APR1400 core model. The spacer grids are modeled and added to this test problem in various ways. Then, by running MCNP5 for all cases of spacer grid modeling, some important numerical results, such as the effective multiplication factor, the spatial distributions of neutron flux, and its energy spectrum are obtained. The numerical results of each case of spacer grid modeling are analyzed and compared to assess which type has more advantages in accuracy of numerical results and effectiveness in terms of geometry building. The conclusion is that the most realistic modeling for Monte Carlo calculation is the “volume-preserving” streamlined heterogeneous spacer grids, but the “banded” dissolution spacer grids modeling is a more practical yet accurate model for routine (deterministic analysis.
Energy Technology Data Exchange (ETDEWEB)
N.V. Mokhov
2003-04-09
Status and recent developments of the MARS 14 Monte Carlo code system for simulation of hadronic and electromagnetic cascades in shielding, accelerator and detector components in the energy range from a fraction of an electronvolt up to 100 TeV are described. these include physics models both in strong and electromagnetic interaction sectors, variance reduction techniques, residual dose, geometry, tracking, histograming. MAD-MARS Beam Line Build and Graphical-User Interface.
Bardenet, R.
2012-01-01
ISBN:978-2-7598-1032-1; International audience; Bayesian inference often requires integrating some function with respect to a posterior distribution. Monte Carlo methods are sampling algorithms that allow to compute these integrals numerically when they are not analytically tractable. We review here the basic principles and the most common Monte Carlo algorithms, among which rejection sampling, importance sampling and Monte Carlo Markov chain (MCMC) methods. We give intuition on the theoretic...
State-of-the-art Monte Carlo 1988
Energy Technology Data Exchange (ETDEWEB)
Soran, P.D.
1988-06-28
Particle transport calculations in highly dimensional and physically complex geometries, such as detector calibration, radiation shielding, space reactors, and oil-well logging, generally require Monte Carlo transport techniques. Monte Carlo particle transport can be performed on a variety of computers ranging from APOLLOs to VAXs. Some of the hardware and software developments, which now permit Monte Carlo methods to be routinely used, are reviewed in this paper. The development of inexpensive, large, fast computer memory, coupled with fast central processing units, permits Monte Carlo calculations to be performed on workstations, minicomputers, and supercomputers. The Monte Carlo renaissance is further aided by innovations in computer architecture and software development. Advances in vectorization and parallelization architecture have resulted in the development of new algorithms which have greatly reduced processing times. Finally, the renewed interest in Monte Carlo has spawned new variance reduction techniques which are being implemented in large computer codes. 45 refs.
Monte Carlo simulation of neutron scattering instruments
Energy Technology Data Exchange (ETDEWEB)
Seeger, P.A.; Daemen, L.L.; Hjelm, R.P. Jr.
1998-12-01
A code package consisting of the Monte Carlo Library MCLIB, the executing code MC{_}RUN, the web application MC{_}Web, and various ancillary codes is proposed as an open standard for simulation of neutron scattering instruments. The architecture of the package includes structures to define surfaces, regions, and optical elements contained in regions. A particle is defined by its vector position and velocity, its time of flight, its mass and charge, and a polarization vector. The MC{_}RUN code handles neutron transport and bookkeeping, while the action on the neutron within any region is computed using algorithms that may be deterministic, probabilistic, or a combination. Complete versatility is possible because the existing library may be supplemented by any procedures a user is able to code. Some examples are shown.
Guideline of Monte Carlo calculation. Neutron/gamma ray transport simulation by Monte Carlo method
2002-01-01
This report condenses basic theories and advanced applications of neutron/gamma ray transport calculations in many fields of nuclear energy research. Chapters 1 through 5 treat historical progress of Monte Carlo methods, general issues of variance reduction technique, cross section libraries used in continuous energy Monte Carlo codes. In chapter 6, the following issues are discussed: fusion benchmark experiments, design of ITER, experiment analyses of fast critical assembly, core analyses of JMTR, simulation of pulsed neutron experiment, core analyses of HTTR, duct streaming calculations, bulk shielding calculations, neutron/gamma ray transport calculations of the Hiroshima atomic bomb. Chapters 8 and 9 treat function enhancements of MCNP and MVP codes, and a parallel processing of Monte Carlo calculation, respectively. An important references are attached at the end of this report.
ROESSEL, ROBERT A., JR.
THE FIRST SECTION OF THIS BOOK COVERS THE HISTORICAL AND CULTURAL BACKGROUND OF THE SAN CARLOS APACHE INDIANS, AS WELL AS AN HISTORICAL SKETCH OF THE DEVELOPMENT OF THEIR FORMAL EDUCATIONAL SYSTEM. THE SECOND SECTION IS DEVOTED TO THE PROBLEMS OF TEACHERS OF THE INDIAN CHILDREN IN GLOBE AND SAN CARLOS, ARIZONA. IT IS DIVIDED INTO THREE PARTS--(1)…
DEFF Research Database (Denmark)
Holm, Bent
2005-01-01
En indplacering af operahuset San Carlo i en kulturhistorisk repræsentationskontekst med særligt henblik på begrebet napolalità.......En indplacering af operahuset San Carlo i en kulturhistorisk repræsentationskontekst med særligt henblik på begrebet napolalità....
Energy Technology Data Exchange (ETDEWEB)
Nerio, U. [Universidad de Cordoba, Monteria (Colombia); Instituto Nacional de Cancerologia, Bogota (Colombia); Chica, L. [Universidad Nacional de Colombia, Bogota (Colombia); Paul, A. [Universite de la Mediterranee, Marseille (France)
2004-07-01
The Monte Carlo method is applied to find the dose rates distribution in tissue around {sup 125} I seeds model 6711 as a function of the silver cylinder radius, R{sub sc} (0.017, 0.021, 0.025, 0.029 and 0.033) cm are used as radius values. It is found here that the dose rate at any point within the tissue decreases as R{sub sc} increases. The relative difference of dose rate that produced by the standard R{sub sc} seed, is less than 5%, for seeds with Rsc between 0.017 and 0.033 cm. (author)
Monte Carlo tests of the ELIPGRID-PC algorithm
Energy Technology Data Exchange (ETDEWEB)
Davidson, J.R.
1995-04-01
The standard tool for calculating the probability of detecting pockets of contamination called hot spots has been the ELIPGRID computer code of Singer and Wickman. The ELIPGRID-PC program has recently made this algorithm available for an IBM{reg_sign} PC. However, no known independent validation of the ELIPGRID algorithm exists. This document describes a Monte Carlo simulation-based validation of a modified version of the ELIPGRID-PC code. The modified ELIPGRID-PC code is shown to match Monte Carlo-calculated hot-spot detection probabilities to within {plus_minus}0.5% for 319 out of 320 test cases. The one exception, a very thin elliptical hot spot located within a rectangular sampling grid, differed from the Monte Carlo-calculated probability by about 1%. These results provide confidence in the ability of the modified ELIPGRID-PC code to accurately predict hot-spot detection probabilities within an acceptable range of error.
Research on GPU Acceleration for Monte Carlo Criticality Calculation
Xu, Qi; Yu, Ganglin; Wang, Kan
2014-06-01
The Monte Carlo neutron transport method can be naturally parallelized by multi-core architectures due to the dependency between particles during the simulation. The GPU+CPU heterogeneous parallel mode has become an increasingly popular way of parallelism in the field of scientific supercomputing. Thus, this work focuses on the GPU acceleration method for the Monte Carlo criticality simulation, as well as the computational efficiency that GPUs can bring. The "neutron transport step" is introduced to increase the GPU thread occupancy. In order to test the sensitivity of the MC code's complexity, a 1D one-group code and a 3D multi-group general purpose code are respectively transplanted to GPUs, and the acceleration effects are compared. The result of numerical experiments shows considerable acceleration effect of the "neutron transport step" strategy. However, the performance comparison between the 1D code and the 3D code indicates the poor scalability of MC codes on GPUs.
Plante, Ianik
2011-08-01
The importance of the radiolysis of water in irradiation of biological systems has motivated considerable theoretical and experimental work in the radiation chemistry of water and aqueous solutions. In particular, Monte-Carlo simulations of radiation track structure and non-homogeneous chemistry have greatly contributed to the understanding of experimental results in radiation chemistry of heavy ions. Actually, most simulations of the non-homogeneous chemistry are done using the Independent Reaction Time (IRT) method, a very fast technique. The main limitation of the IRT method is that the positions of the radiolytic species are not calculated as a function of time, which is needed to simulate the irradiation of more complex systems. Step-by-step (SBS) methods, which are able to provide such information, have been used only sparsely because these are time consuming in terms of calculation. Recent improvements in computer performance now allow the regular use of the SBS method in radiation chemistry. In the present paper, the first of a series of two, the SBS method is reviewed in detail. To these ends, simulation of diffusion of particles and chemical reactions in aqueous solutions is reviewed, and implementation of the program is discussed. Simulation of model systems is then performed to validate the adequacy of stepwise diffusion and reaction schemes. In the second paper, radiochemical yields of simulated radiation tracks calculated by the SBS program in different conditions of LET, pH, and temperature are compared with results from the IRT program and experimental data.
Sharma, Diksha; Badano, Aldo
2013-03-01
hybridMANTIS is a Monte Carlo package for modeling indirect x-ray imagers using columnar geometry based on a hybrid concept that maximizes the utilization of available CPU and graphics processing unit processors in a workstation. The authors compare hybridMANTIS x-ray response simulations to previously published MANTIS and experimental data for four cesium iodide scintillator screens. These screens have a variety of reflective and absorptive surfaces with different thicknesses. The authors analyze hybridMANTIS results in terms of modulation transfer function and calculate the root mean square difference and Swank factors from simulated and experimental results. The comparison suggests that hybridMANTIS better matches the experimental data as compared to MANTIS, especially at high spatial frequencies and for the thicker screens. hybridMANTIS simulations are much faster than MANTIS with speed-ups up to 5260. hybridMANTIS is a useful tool for improved description and optimization of image acquisition stages in medical imaging systems and for modeling the forward problem in iterative reconstruction algorithms.
Implict Monte Carlo Radiation Transport Simulations of Four Test Problems
Energy Technology Data Exchange (ETDEWEB)
Gentile, N
2007-08-01
Radiation transport codes, like almost all codes, are difficult to develop and debug. It is helpful to have small, easy to run test problems with known answers to use in development and debugging. It is also prudent to re-run test problems periodically during development to ensure that previous code capabilities have not been lost. We describe four radiation transport test problems with analytic or approximate analytic answers. These test problems are suitable for use in debugging and testing radiation transport codes. We also give results of simulations of these test problems performed with an Implicit Monte Carlo photonics code.
Quantum Monte Carlo simulation
Wang, Yazhen
2011-01-01
Contemporary scientific studies often rely on the understanding of complex quantum systems via computer simulation. This paper initiates the statistical study of quantum simulation and proposes a Monte Carlo method for estimating analytically intractable quantities. We derive the bias and variance for the proposed Monte Carlo quantum simulation estimator and establish the asymptotic theory for the estimator. The theory is used to design a computational scheme for minimizing the mean square er...
SKIRT: the design of a suite of input models for Monte Carlo radiative transfer simulations
Baes, Maarten
2015-01-01
The Monte Carlo method is the most popular technique to perform radiative transfer simulations in a general 3D geometry. The algorithms behind and acceleration techniques for Monte Carlo radiative transfer are discussed extensively in the literature, and many different Monte Carlo codes are publicly available. On the contrary, the design of a suite of components that can be used for the distribution of sources and sinks in radiative transfer codes has received very little attention. The availability of such models, with different degrees of complexity, has many benefits. For example, they can serve as toy models to test new physical ingredients, or as parameterised models for inverse radiative transfer fitting. For 3D Monte Carlo codes, this requires algorithms to efficiently generate random positions from 3D density distributions. We describe the design of a flexible suite of components for the Monte Carlo radiative transfer code SKIRT. The design is based on a combination of basic building blocks (which can...
Radiative Equilibrium and Temperature Correction in Monte Carlo Radiation Transfer
Bjorkman, J. E.; Wood, Kenneth
2001-01-01
We describe a general radiative equilibrium and temperature correction procedure for use in Monte Carlo radiation transfer codes with sources of temperature-independent opacity, such as astrophysical dust. The technique utilizes the fact that Monte Carlo simulations track individual photon packets, so we may easily determine where their energy is absorbed. When a packet is absorbed, it heats a particular cell within the envelope, raising its temperature. To enforce radiative equilibrium, the ...
MONTE CARLO SIMULATION OF CHARGED PARTICLE IN AN ELECTRONEGATIVE PLASMA
Directory of Open Access Journals (Sweden)
L SETTAOUTI
2003-12-01
Full Text Available Interest in radio frequency (rf discharges has grown tremendously in recent years due to their importance in microelectronic technologies. Especially interesting are the properties of discharges in electronegative gases which are most frequently used for technological applications. Monte Carlo simulation have become increasingly important as a simulation tool particularly in the area of plasma physics. In this work, we present some detailed properties of rf plasmas obtained by Monte Carlo simulation code, in SF6
The impact of advances in computer technology on particle transport Monte Carlo
Energy Technology Data Exchange (ETDEWEB)
Martin, W.R. [Michigan Univ., Ann Arbor, MI (United States). Dept. of Nuclear Engineering; Rathkopf, J.A. [Lawrence Livermore National Lab., CA (United States); Brown, F.B. [Knolls Atomic Power Lab., Schenectady, NY (United States)
1992-01-21
Advances in computer technology, including hardware, architectural, and software advances, have led to dramatic gains in computer performance over the past decade. We summarize these performance trends and discuss the extent to which particle transport Monte Carlo codes have been able to take advantage of these performance gains. We consider MIMD, SIMD, and parallel distributed computer configurations for particle transport Monte Carlo applications. Some specific experience with vectorization and parallelization of production Monte Carlo codes is included. The topic of parallel random number generation is discussed in some detail. Finally, some software issues that hinder the implementation of Monte Carlo methods on parallel processors are addressed.
Monte Carlo systems used for treatment planning and dose verification
Energy Technology Data Exchange (ETDEWEB)
Brualla, Lorenzo [Universitaetsklinikum Essen, NCTeam, Strahlenklinik, Essen (Germany); Rodriguez, Miguel [Centro Medico Paitilla, Balboa (Panama); Lallena, Antonio M. [Universidad de Granada, Departamento de Fisica Atomica, Molecular y Nuclear, Granada (Spain)
2017-04-15
General-purpose radiation transport Monte Carlo codes have been used for estimation of the absorbed dose distribution in external photon and electron beam radiotherapy patients since several decades. Results obtained with these codes are usually more accurate than those provided by treatment planning systems based on non-stochastic methods. Traditionally, absorbed dose computations based on general-purpose Monte Carlo codes have been used only for research, owing to the difficulties associated with setting up a simulation and the long computation time required. To take advantage of radiation transport Monte Carlo codes applied to routine clinical practice, researchers and private companies have developed treatment planning and dose verification systems that are partly or fully based on fast Monte Carlo algorithms. This review presents a comprehensive list of the currently existing Monte Carlo systems that can be used to calculate or verify an external photon and electron beam radiotherapy treatment plan. Particular attention is given to those systems that are distributed, either freely or commercially, and that do not require programming tasks from the end user. These systems are compared in terms of features and the simulation time required to compute a set of benchmark calculations. (orig.) [German] Seit mehreren Jahrzehnten werden allgemein anwendbare Monte-Carlo-Codes zur Simulation des Strahlungstransports benutzt, um die Verteilung der absorbierten Dosis in der perkutanen Strahlentherapie mit Photonen und Elektronen zu evaluieren. Die damit erzielten Ergebnisse sind meist akkurater als solche, die mit nichtstochastischen Methoden herkoemmlicher Bestrahlungsplanungssysteme erzielt werden koennen. Wegen des damit verbundenen Arbeitsaufwands und der langen Dauer der Berechnungen wurden Monte-Carlo-Simulationen von Dosisverteilungen in der konventionellen Strahlentherapie in der Vergangenheit im Wesentlichen in der Forschung eingesetzt. Im Bemuehen, Monte-Carlo-Codes
Kuipers, J; Vermaseren, J A M
2013-01-01
We describe the implementation of output code optimization in the open source computer algebra system FORM. This implementation is based on recently discovered techniques of Monte Carlo tree search to find efficient multivariate Horner schemes, in combination with other optimization algorithms, such as common subexpression elimination. For systems for which no specific knowledge is provided it performs significantly better than other methods we could compare with. Because the method has a number of free parameters, we also show some methods by which to tune them to different types of problems.
2009-01-01
Carlo Rubbia turned 75 on March 31, and CERN held a symposium to mark his birthday and pay tribute to his impressive contribution to both CERN and science. Carlo Rubbia, 4th from right, together with the speakers at the symposium.On 7 April CERN hosted a celebration marking Carlo Rubbia’s 75th birthday and 25 years since he was awarded the Nobel Prize for Physics. "Today we will celebrate 100 years of Carlo Rubbia" joked CERN’s Director-General, Rolf Heuer in his opening speech, "75 years of his age and 25 years of the Nobel Prize." Rubbia received the Nobel Prize along with Simon van der Meer for contributions to the discovery of the W and Z bosons, carriers of the weak interaction. During the symposium, which was held in the Main Auditorium, several eminent speakers gave lectures on areas of science to which Carlo Rubbia made decisive contributions. Among those who spoke were Michel Spiro, Director of the French National Insti...
Directory of Open Access Journals (Sweden)
Fabio Burderi
2007-05-01
Full Text Available Motivated by the study of decipherability conditions for codes weaker than Unique Decipherability (UD, we introduce the notion of coding partition. Such a notion generalizes that of UD code and, for codes that are not UD, allows to recover the ``unique decipherability" at the level of the classes of the partition. By tacking into account the natural order between the partitions, we define the characteristic partition of a code X as the finest coding partition of X. This leads to introduce the canonical decomposition of a code in at most one unambiguouscomponent and other (if any totally ambiguouscomponents. In the case the code is finite, we give an algorithm for computing its canonical partition. This, in particular, allows to decide whether a given partition of a finite code X is a coding partition. This last problem is then approached in the case the code is a rational set. We prove its decidability under the hypothesis that the partition contains a finite number of classes and each class is a rational set. Moreover we conjecture that the canonical partition satisfies such a hypothesis. Finally we consider also some relationships between coding partitions and varieties of codes.
Recent developments in the Los Alamos radiation transport code system
Energy Technology Data Exchange (ETDEWEB)
Forster, R.A.; Parsons, K. [Los Alamos National Lab., NM (United States)
1997-06-01
A brief progress report on updates to the Los Alamos Radiation Transport Code System (LARTCS) for solving criticality and fixed-source problems is provided. LARTCS integrates the Diffusion Accelerated Neutral Transport (DANT) discrete ordinates codes with the Monte Carlo N-Particle (MCNP) code. The LARCTS code is being developed with a graphical user interface for problem setup and analysis. Progress in the DANT system for criticality applications include a two-dimensional module which can be linked to a mesh-generation code and a faster iteration scheme. Updates to MCNP Version 4A allow statistical checks of calculated Monte Carlo results.
Baräo, Fernando; Nakagawa, Masayuki; Távora, Luis; Vaz, Pedro
2001-01-01
This book focusses on the state of the art of Monte Carlo methods in radiation physics and particle transport simulation and applications, the latter involving in particular, the use and development of electron--gamma, neutron--gamma and hadronic codes. Besides the basic theory and the methods employed, special attention is paid to algorithm development for modeling, and the analysis of experiments and measurements in a variety of fields ranging from particle to medical physics.
Leonardo Rossi
Carlo Caso (1940 - 2007) Our friend and colleague Carlo Caso passed away on July 7th, after several months of courageous fight against cancer. Carlo spent most of his scientific career at CERN, taking an active part in the experimental programme of the laboratory. His long and fruitful involvement in particle physics started in the sixties, in the Genoa group led by G. Tomasini. He then made several experiments using the CERN liquid hydrogen bubble chambers -first the 2000HBC and later BEBC- to study various facets of the production and decay of meson and baryon resonances. He later made his own group and joined the NA27 Collaboration to exploit the EHS Spectrometer with a rapid cycling bubble chamber as vertex detector. Amongst their many achievements, they were the first to measure, with excellent precision, the lifetime of the charmed D mesons. At the start of the LEP era, Carlo and his group moved to the DELPHI experiment, participating in the construction and running of the HPC electromagnetic c...
Pedro Medina Avendaño
1981-01-01
Carlos Vega Duarte tenía la sencillez de los seres elementales y puros. Su corazón era limpio como oro de aluvión. Su trato directo y coloquial ponía de relieve a un santandereano sin contaminaciones que amaba el fulgor de las armas y se encandilaba con el destello de las frases perfectas
Challenges of Monte Carlo Transport
Energy Technology Data Exchange (ETDEWEB)
Long, Alex Roberts [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2016-06-10
These are slides from a presentation for Parallel Summer School at Los Alamos National Laboratory. Solving discretized partial differential equations (PDEs) of interest can require a large number of computations. We can identify concurrency to allow parallel solution of discrete PDEs. Simulated particles histories can be used to solve the Boltzmann transport equation. Particle histories are independent in neutral particle transport, making them amenable to parallel computation. Physical parameters and method type determine the data dependencies of particle histories. Data requirements shape parallel algorithms for Monte Carlo. Then, Parallel Computational Physics and Parallel Monte Carlo are discussed and, finally, the results are given. The mesh passing method greatly simplifies the IMC implementation and allows simple load-balancing. Using MPI windows and passive, one-sided RMA further simplifies the implementation by removing target synchronization. The author is very interested in implementations of PGAS that may allow further optimization for one-sided, read-only memory access (e.g. Open SHMEM). The MPICH_RMA_OVER_DMAPP option and library is required to make one-sided messaging scale on Trinitite - Moonlight scales poorly. Interconnect specific libraries or functions are likely necessary to ensure performance. BRANSON has been used to directly compare the current standard method to a proposed method on idealized problems. The mesh passing algorithm performs well on problems that are designed to show the scalability of the particle passing method. BRANSON can now run load-imbalanced, dynamic problems. Potential avenues of improvement in the mesh passing algorithm will be implemented and explored. A suite of test problems that stress DD methods will elucidate a possible path forward for production codes.
Composite biasing in Monte Carlo radiative transfer
Baes, Maarten; Lunttila, Tuomas; Bianchi, Simone; Camps, Peter; Juvela, Mika; Kuiper, Rolf
2016-01-01
Biasing or importance sampling is a powerful technique in Monte Carlo radiative transfer, and can be applied in different forms to increase the accuracy and efficiency of simulations. One of the drawbacks of the use of biasing is the potential introduction of large weight factors. We discuss a general strategy, composite biasing, to suppress the appearance of large weight factors. We use this composite biasing approach for two different problems faced by current state-of-the-art Monte Carlo radiative transfer codes: the generation of photon packages from multiple components, and the penetration of radiation through high optical depth barriers. In both cases, the implementation of the relevant algorithms is trivial and does not interfere with any other optimisation techniques. Through simple test models, we demonstrate the general applicability, accuracy and efficiency of the composite biasing approach. In particular, for the penetration of high optical depths, the gain in efficiency is spectacular for the spe...
Monte Carlo simulations on SIMD computer architectures
Energy Technology Data Exchange (ETDEWEB)
Burmester, C.P.; Gronsky, R. [Lawrence Berkeley Lab., CA (United States); Wille, L.T. [Florida Atlantic Univ., Boca Raton, FL (United States). Dept. of Physics
1992-03-01
Algorithmic considerations regarding the implementation of various materials science applications of the Monte Carlo technique to single instruction multiple data (SMM) computer architectures are presented. In particular, implementation of the Ising model with nearest, next nearest, and long range screened Coulomb interactions on the SIMD architecture MasPar MP-1 (DEC mpp-12000) series of massively parallel computers is demonstrated. Methods of code development which optimize processor array use and minimize inter-processor communication are presented including lattice partitioning and the use of processor array spanning tree structures for data reduction. Both geometric and algorithmic parallel approaches are utilized. Benchmarks in terms of Monte Carlo updates per second for the MasPar architecture are presented and compared to values reported in the literature from comparable studies on other architectures.
Energy Technology Data Exchange (ETDEWEB)
Martin, E.; Gschwind, R.; Henriet, J.; Sauget, M.; Makovicka, L. [IRMA/Enisys/FEMTO-ST, Pole universitaire des Portes du Jura, place Tharradin, BP 71427, 2521 1 - Montbeliard cedex (France)
2010-07-01
In order to reduce the computing time needed by Monte Carlo codes in the field of irradiation physics, notably in dosimetry, the authors report the use of artificial neural networks in combination with preliminary Monte Carlo calculations. During the learning phase, Monte Carlo calculations are performed in homogeneous media to allow the building up of the neural network. Then, dosimetric calculations (in heterogeneous media, unknown by the network) can be performed by the so-learned network. Results with an equivalent precision can be obtained within less than one minute on a simple PC whereas several days are needed with a Monte Carlo calculation
NASA Space Radiation Transport Code Development Consortium.
Townsend, Lawrence W
2005-01-01
Recently, NASA established a consortium involving the University of Tennessee (lead institution), the University of Houston, Roanoke College and various government and national laboratories, to accelerate the development of a standard set of radiation transport computer codes for NASA human exploration applications. This effort involves further improvements of the Monte Carlo codes HETC and FLUKA and the deterministic code HZETRN, including developing nuclear reaction databases necessary to extend the Monte Carlo codes to carry out heavy ion transport, and extending HZETRN to three dimensions. The improved codes will be validated by comparing predictions with measured laboratory transport data, provided by an experimental measurements consortium, and measurements in the upper atmosphere on the balloon-borne Deep Space Test Bed (DSTB). In this paper, we present an overview of the consortium members and the current status and future plans of consortium efforts to meet the research goals and objectives of this extensive undertaking.
Allele coding in genomic evaluation
Directory of Open Access Journals (Sweden)
Christensen Ole F
2011-06-01
Full Text Available Abstract Background Genomic data are used in animal breeding to assist genetic evaluation. Several models to estimate genomic breeding values have been studied. In general, two approaches have been used. One approach estimates the marker effects first and then, genomic breeding values are obtained by summing marker effects. In the second approach, genomic breeding values are estimated directly using an equivalent model with a genomic relationship matrix. Allele coding is the method chosen to assign values to the regression coefficients in the statistical model. A common allele coding is zero for the homozygous genotype of the first allele, one for the heterozygote, and two for the homozygous genotype for the other allele. Another common allele coding changes these regression coefficients by subtracting a value from each marker such that the mean of regression coefficients is zero within each marker. We call this centered allele coding. This study considered effects of different allele coding methods on inference. Both marker-based and equivalent models were considered, and restricted maximum likelihood and Bayesian methods were used in inference. Results Theoretical derivations showed that parameter estimates and estimated marker effects in marker-based models are the same irrespective of the allele coding, provided that the model has a fixed general mean. For the equivalent models, the same results hold, even though different allele coding methods lead to different genomic relationship matrices. Calculated genomic breeding values are independent of allele coding when the estimate of the general mean is included into the values. Reliabilities of estimated genomic breeding values calculated using elements of the inverse of the coefficient matrix depend on the allele coding because different allele coding methods imply different models. Finally, allele coding affects the mixing of Markov chain Monte Carlo algorithms, with the centered coding being
Conceptual design and Monte Carlo simulations of the AGATA array
Energy Technology Data Exchange (ETDEWEB)
Farnea, E., E-mail: Enrico.Farnea@pd.infn.i [Istituto Nazionale di Fisica Nucleare, Sezione di Padova, Padova (Italy); Recchia, F.; Bazzacco, D. [Istituto Nazionale di Fisica Nucleare, Sezione di Padova, Padova (Italy); Kroell, Th. [Institut fuer Kernphysik, Technische Universitaet Darmstadt, Darmstadt (Germany); Podolyak, Zs. [Department of Physics, University of Surrey, Guildford (United Kingdom); Quintana, B. [Departamento de Fisica Fundamental, Universidad de Salamanca, Salamanca (Spain); Gadea, A. [Instituto de Fisica Corpuscular, CSIC-Universidad de Valencia, Valencia (Spain)
2010-09-21
The aim of the Advanced GAmma Tracking Array (AGATA) project is the construction of an array based on the novel concepts of pulse shape analysis and {gamma}-ray tracking with highly segmented Ge semiconductor detectors. The conceptual design of AGATA and its performance evaluation under different experimental conditions has required the development of a suitable Monte Carlo code. In this article, the description of the code as well as simulation results relevant for AGATA, are presented.
Energy Technology Data Exchange (ETDEWEB)
Aryal, Prakash; Molloy, Janelle A. [Department of Radiation Medicine, University of Kentucky, Lexington, Kentucky 40536 (United States); Rivard, Mark J., E-mail: mark.j.rivard@gmail.com [Department of Radiation Oncology, Tufts University School of Medicine, Boston, Massachusetts 02111 (United States)
2014-02-15
Purpose: To investigate potential causes for differences in TG-43 brachytherapy dosimetry parameters in the existent literature for the model IAI-125A{sup 125}I seed and to propose new standard dosimetry parameters. Methods: The MCNP5 code was used for Monte Carlo (MC) simulations. Sensitivity of dose distributions, and subsequently TG-43 dosimetry parameters, was explored to reproduce historical methods upon which American Association of Physicists in Medicine (AAPM) consensus data are based. Twelve simulation conditions varying{sup 125}I coating thickness, coating mass density, photon interaction cross-section library, and photon emission spectrum were examined. Results: Varying{sup 125}I coating thickness, coating mass density, photon cross-section library, and photon emission spectrum for the model IAI-125A seed changed the dose-rate constant by up to 0.9%, about 1%, about 3%, and 3%, respectively, in comparison to the proposed standard value of 0.922 cGy h{sup −1} U{sup −1}. The dose-rate constant values by Solberg et al. [“Dosimetric parameters of three new solid core {sup 125}I brachytherapy sources,” J. Appl. Clin. Med. Phys. 3, 119–134 (2002)], Meigooni et al. [“Experimental and theoretical determination of dosimetric characteristics of IsoAid ADVANTAGE™ {sup 125}I brachytherapy source,” Med. Phys. 29, 2152–2158 (2002)], and Taylor and Rogers [“An EGSnrc Monte Carlo-calculated database of TG-43 parameters,” Med. Phys. 35, 4228–4241 (2008)] for the model IAI-125A seed and Kennedy et al. [“Experimental and Monte Carlo determination of the TG-43 dosimetric parameters for the model 9011 THINSeed™ brachytherapy source,” Med. Phys. 37, 1681–1688 (2010)] for the model 6711 seed were +4.3% (0.962 cGy h{sup −1} U{sup −1}), +6.2% (0.98 cGy h{sup −1} U{sup −1}), +0.3% (0.925 cGy h{sup −1} U{sup −1}), and −0.2% (0.921 cGy h{sup −1} U{sup −1}), respectively, in comparison to the proposed standard
Chemical application of diffusion quantum Monte Carlo
Reynolds, P. J.; Lester, W. A., Jr.
1983-10-01
The diffusion quantum Monte Carlo (QMC) method gives a stochastic solution to the Schroedinger equation. As an example the singlet-triplet splitting of the energy of the methylene molecule CH2 is given. The QMC algorithm was implemented on the CYBER 205, first as a direct transcription of the algorithm running on our VAX 11/780, and second by explicitly writing vector code for all loops longer than a crossover length C. The speed of the codes relative to one another as a function of C, and relative to the VAX is discussed. Since CH2 has only eight electrons, most of the loops in this application are fairly short. The longest inner loops run over the set of atomic basis functions. The CPU time dependence obtained versus the number of basis functions is discussed and compared with that obtained from traditional quantum chemistry codes and that obtained from traditional computer architectures. Finally, preliminary work on restructuring the algorithm to compute the separate Monte Carlo realizations in parallel is discussed.
Comparison over the nuclear analysis of the HCLL blanket for the European DEMO
Energy Technology Data Exchange (ETDEWEB)
Jaboulay, Jean-Charles, E-mail: jean-charles.jaboulay@cea.fr [CEA-Saclay, DEN, DM2S, F-91191 Gif-sur-Yvette (France); Aiello, Giacomo; Aubert, Julien [CEA-Saclay, DEN, DM2S, F-91191 Gif-sur-Yvette (France); Villari, Rosaria [ENEA, UTFUS-TECN, Via E. Fermi 4, 00044 Frascati, Rome (Italy); Fischer, Ulrich [Karlsruhe Institute of Technology, 76344 Eggenstein-Leopoldshafen, Karlsruhe (Germany)
2016-11-01
Highlights: • A complete nuclear analysis of the DEMO HCLL has been carried out at CEA with the TRIPOLI-4{sup ®} Monte Carlo code. • The DEMO tokamak model was generated by the CAD import tool McCad. • The HCLL blankets were implemented using a previous MCNP model developed at ENEA. • A good agreement is observed between the results obtained at CEA with TRIPOLI-4 and JEFF-3.1.1 and whose obtained at ENEA with MCNP and FENDL-2.1. - Abstract: This paper presents the comparison over the nuclear analysis of the European DEMO with HCLL blanket carried out with the TRIPOLI-4{sup ®} Monte Carlo code and the JEFF-3.1.1 nuclear data library and with the MCNP5 Monte Carlo code and the FENDL-2.1 nuclear data library. The MCNP5 analysis was conducted firstly by ENEA with a detailed 3D model describing all the HCLL blanket internal structures. This MCNP5 model was converted into TRIPOLI-4{sup ®} representation for performing the nuclear analysis at CEA with the objective to demonstrate consistency between both analyses. A very good agreement was obtained for all of the relevant nuclear responses (neutron wall loading, tritium breeding ratio, nuclear heating, neutron flux distribution, etc.), validating CEA’s nuclear analysis approach, based on TRIPOLI-4{sup ®} Monte Carlo code and JEFF-3.1.1 nuclear data library, for the European DEMO.
Bayoumi, T A; Reda, S M; Saleh, H M
2012-01-01
Radioactive waste generated from the nuclear applications should be properly isolated by a suitable containment system such as, multi-barrier container. The present study aims to evaluate the isolation capacity of a new multi-barrier container made from cement and clay and including borate waste materials. These wastes were spiked by (137)Cs and (60)Co radionuclides to simulate that waste generated from the primary cooling circuit of pressurized water reactors. Leaching of both radionuclides in ground water was followed and calculated during ten years. Monte Carlo (MCNP5) simulations computed the photon flux distribution of the multi-barrier container, including radioactive borate waste of specific activity 11.22KBq/g and 4.18KBq/g for (137)Cs and (60)Co, respectively, at different periods of 0, 15.1, 30.2 and 302 years. The average total flux for 100cm radius of spherical cell was 0.192photon/cm(2) at initial time and 2.73×10(-4)photon/cm(2) after 302 years. Maximum waste activity keeping the surface radiation dose within the permissible level was calculated and found to be 56KBq/g with attenuation factors of 0.73cm(-1) and 0.6cm(-1) for cement and clay, respectively. The average total flux was 1.37×10(-3)photon/cm(2) after 302 years. Monte Carlo simulations revealed that the proposed multi-barrier container is safe enough during transportation, evacuation or rearrangement in the disposal site for more than 300 years. Copyright © 2011 Elsevier Ltd. All rights reserved.
Monte Carlo techniques for analyzing deep penetration problems
Energy Technology Data Exchange (ETDEWEB)
Cramer, S.N.; Gonnord, J.; Hendricks, J.S.
1985-01-01
A review of current methods and difficulties in Monte Carlo deep-penetration calculations is presented. Statistical uncertainty is discussed, and recent adjoint optimization of splitting, Russian roulette, and exponential transformation biasing is reviewed. Other aspects of the random walk and estimation processes are covered, including the relatively new DXANG angular biasing technique. Specific items summarized are albedo scattering, Monte Carlo coupling techniques with discrete ordinates and other methods, adjoint solutions, and multi-group Monte Carlo. The topic of code-generated biasing parameters is presented, including the creation of adjoint importance functions from forward calculations. Finally, current and future work in the area of computer learning and artificial intelligence is discussed in connection with Monte Carlo applications. 29 refs.
Latorre, Jose I
2015-01-01
There exists a remarkable four-qutrit state that carries absolute maximal entanglement in all its partitions. Employing this state, we construct a tensor network that delivers a holographic many body state, the H-code, where the physical properties of the boundary determine those of the bulk. This H-code is made of an even superposition of states whose relative Hamming distances are exponentially large with the size of the boundary. This property makes H-codes natural states for a quantum memory. H-codes exist on tori of definite sizes and get classified in three different sectors characterized by the sum of their qutrits on cycles wrapped through the boundaries of the system. We construct a parent Hamiltonian for the H-code which is highly non local and finally we compute the topological entanglement entropy of the H-code.
Iterative acceleration methods for Monte Carlo and deterministic criticality calculations
Energy Technology Data Exchange (ETDEWEB)
Urbatsch, T.J.
1995-11-01
If you have ever given up on a nuclear criticality calculation and terminated it because it took so long to converge, you might find this thesis of interest. The author develops three methods for improving the fission source convergence in nuclear criticality calculations for physical systems with high dominance ratios for which convergence is slow. The Fission Matrix Acceleration Method and the Fission Diffusion Synthetic Acceleration (FDSA) Method are acceleration methods that speed fission source convergence for both Monte Carlo and deterministic methods. The third method is a hybrid Monte Carlo method that also converges for difficult problems where the unaccelerated Monte Carlo method fails. The author tested the feasibility of all three methods in a test bed consisting of idealized problems. He has successfully accelerated fission source convergence in both deterministic and Monte Carlo criticality calculations. By filtering statistical noise, he has incorporated deterministic attributes into the Monte Carlo calculations in order to speed their source convergence. He has used both the fission matrix and a diffusion approximation to perform unbiased accelerations. The Fission Matrix Acceleration method has been implemented in the production code MCNP and successfully applied to a real problem. When the unaccelerated calculations are unable to converge to the correct solution, they cannot be accelerated in an unbiased fashion. A Hybrid Monte Carlo method weds Monte Carlo and a modified diffusion calculation to overcome these deficiencies. The Hybrid method additionally possesses reduced statistical errors.
Kubilius, Jonas
2014-01-01
Sharing code is becoming increasingly important in the wake of Open Science. In this review I describe and compare two popular code-sharing utilities, GitHub and Open Science Framework (OSF). GitHub is a mature, industry-standard tool but lacks focus towards researchers. In comparison, OSF offers a one-stop solution for researchers but a lot of functionality is still under development. I conclude by listing alternative lesser-known tools for code and materials sharing.
Directory of Open Access Journals (Sweden)
Cecilia Maya
2004-12-01
Full Text Available El método Monte Carlo se aplica a varios casos de valoración de opciones financieras. El método genera una buena aproximación al comparar su precisión con la de otros métodos numéricos. La estimación que produce la versión Cruda de Monte Carlo puede ser aún más exacta si se recurre a metodologías de reducción de la varianza entre las cuales se sugieren la variable antitética y de la variable de control. Sin embargo, dichas metodologías requieren un esfuerzo computacional mayor por lo cual las mismas deben ser evaluadas en términos no sólo de su precisión sino también de su eficiencia.
Monte Carlo and nonlinearities
Dauchet, Jérémi; Blanco, Stéphane; Caliot, Cyril; Charon, Julien; Coustet, Christophe; Hafi, Mouna El; Eymet, Vincent; Farges, Olivier; Forest, Vincent; Fournier, Richard; Galtier, Mathieu; Gautrais, Jacques; Khuong, Anaïs; Pelissier, Lionel; Piaud, Benjamin; Roger, Maxime; Terrée, Guillaume; Weitz, Sebastian
2016-01-01
The Monte Carlo method is widely used to numerically predict systems behaviour. However, its powerful incremental design assumes a strong premise which has severely limited application so far: the estimation process must combine linearly over dimensions. Here we show that this premise can be alleviated by projecting nonlinearities on a polynomial basis and increasing the configuration-space dimension. Considering phytoplankton growth in light-limited environments, radiative transfer in planetary atmospheres, electromagnetic scattering by particles and concentrated-solar-power-plant productions, we prove the real world usability of this advance on four test-cases that were so far regarded as impracticable by Monte Carlo approaches. We also illustrate an outstanding feature of our method when applied to sharp problems with interacting particles: handling rare events is now straightforward. Overall, our extension preserves the features that made the method popular: addressing nonlinearities does not compromise o...
Energy Technology Data Exchange (ETDEWEB)
Wollaber, Allan Benton [Los Alamos National Lab. (LANL), Los Alamos, NM (United States)
2016-06-16
This is a powerpoint presentation which serves as lecture material for the Parallel Computing summer school. It goes over the fundamentals of the Monte Carlo calculation method. The material is presented according to the following outline: Introduction (background, a simple example: estimating π), Why does this even work? (The Law of Large Numbers, The Central Limit Theorem), How to sample (inverse transform sampling, rejection), and An example from particle transport.
Directory of Open Access Journals (Sweden)
Pedro Medina Avendaño
1981-01-01
Full Text Available Carlos Vega Duarte tenía la sencillez de los seres elementales y puros. Su corazón era limpio como oro de aluvión. Su trato directo y coloquial ponía de relieve a un santandereano sin contaminaciones que amaba el fulgor de las armas y se encandilaba con el destello de las frases perfectas
The effect of load imbalances on the performance of Monte Carlo algorithms in LWR analysis
Energy Technology Data Exchange (ETDEWEB)
Siegel, A.R., E-mail: siegela@mcs.anl.gov [Argonne National Laboratory, Nuclear Engineering Division (United States); Argonne National Laboratory, Mathematics and Computer Science Division (United States); Smith, K., E-mail: kord@mit.edu [Massachusetts Institute of Technology, Department of Nuclear Science and Engineering (United States); Romano, P.K., E-mail: romano7@mit.edu [Massachusetts Institute of Technology, Department of Nuclear Science and Engineering (United States); Forget, B., E-mail: bforget@mit.edu [Massachusetts Institute of Technology, Department of Nuclear Science and Engineering (United States); Felker, K., E-mail: felker@mcs.anl.gov [Argonne National Laboratory, Mathematics and Computer Science Division (United States)
2013-02-15
A model is developed to predict the impact of particle load imbalances on the performance of domain-decomposed Monte Carlo neutron transport algorithms. Expressions for upper bound performance “penalties” are derived in terms of simple machine characteristics, material characterizations and initial particle distributions. The hope is that these relations can be used to evaluate tradeoffs among different memory decomposition strategies in next generation Monte Carlo codes, and perhaps as a metric for triggering particle redistribution in production codes.
A Monte Carlo algorithm for degenerate plasmas
Energy Technology Data Exchange (ETDEWEB)
Turrell, A.E., E-mail: a.turrell09@imperial.ac.uk; Sherlock, M.; Rose, S.J.
2013-09-15
A procedure for performing Monte Carlo calculations of plasmas with an arbitrary level of degeneracy is outlined. It has possible applications in inertial confinement fusion and astrophysics. Degenerate particles are initialised according to the Fermi–Dirac distribution function, and scattering is via a Pauli blocked binary collision approximation. The algorithm is tested against degenerate electron–ion equilibration, and the degenerate resistivity transport coefficient from unmagnetised first order transport theory. The code is applied to the cold fuel shell and alpha particle equilibration problem of inertial confinement fusion.
Energy Technology Data Exchange (ETDEWEB)
Marcus, Ryan C. [Los Alamos National Laboratory
2012-07-24
Overview of this presentation is (1) Exascale computing - different technologies, getting there; (2) high-performance proof-of-concept MCMini - features and results; and (3) OpenCL toolkit - Oatmeal (OpenCL Automatic Memory Allocation Library) - purpose and features. Despite driver issues, OpenCL seems like a good, hardware agnostic tool. MCMini demonstrates the possibility for GPGPU-based Monte Carlo methods - it shows great scaling for HPC application and algorithmic equivalence. Oatmeal provides a flexible framework to aid in the development of scientific OpenCL codes.
Directory of Open Access Journals (Sweden)
Charlie Samuya Veric
2001-12-01
Full Text Available The importance of Carlos Bulosan in Filipino and Filipino-American radical history and literature is indisputable. His eminence spans the pacific, and he is known, diversely, as a radical poet, fictionist, novelist, and labor organizer. Author of the canonical America Iis the Hearts, Bulosan is celebrated for chronicling the conditions in America in his time, such as racism and unemployment. In the history of criticism on Bulosan's life and work, however, there is an undeclared general consensus that views Bulosan and his work as coherent permanent texts of radicalism and anti-imperialism. Central to the existence of such a tradition of critical reception are the generations of critics who, in more ways than one, control the discourse on and of Carlos Bulosan. This essay inquires into the sphere of the critical reception that orders, for our time and for the time ahead, the reading and interpretation of Bulosan. What eye and seeing, the essay asks, determine the perception of Bulosan as the angel of radicalism? What is obscured in constructing Bulosan as an immutable figure of the political? What light does the reader conceive when the personal is brought into the open and situated against the political? the essay explores the answers to these questions in Bulosan's loving letters to various friends, strangers, and white American women. The presence of these interrogations, the essay believes, will secure ultimately the continuing importance of Carlos Bulosan to radical literature and history.
2009-01-01
On 7 April CERN will be holding a symposium to mark the 75th birthday of Carlo Rubbia, who shared the 1984 Nobel Prize for Physics with Simon van der Meer for contributions to the discovery of the W and Z bosons, carriers of the weak interaction. Following a presentation by Rolf Heuer, lectures will be given by eminent speakers on areas of science to which Carlo Rubbia has made decisive contributions. Michel Spiro, Director of the French National Institute of Nuclear and Particle Physics (IN2P3) of the CNRS, Lyn Evans, sLHC Project Leader, and Alan Astbury of the TRIUMF Laboratory will talk about the physics of the weak interaction and the discovery of the W and Z bosons. Former CERN Director-General Herwig Schopper will lecture on CERN’s accelerators from LEP to the LHC. Giovanni Bignami, former President of the Italian Space Agency and Professor at the IUSS School for Advanced Studies in Pavia will speak about his work with Carlo Rubbia. Finally, Hans Joachim Sch...
2009-01-01
On 7 April CERN will be holding a symposium to mark the 75th birthday of Carlo Rubbia, who shared the 1984 Nobel Prize for Physics with Simon van der Meer for contributions to the discovery of the W and Z bosons, carriers of the weak interaction. Following a presentation by Rolf Heuer, lectures will be given by eminent speakers on areas of science to which Carlo Rubbia has made decisive contributions. Michel Spiro, Director of the French National Institute of Nuclear and Particle Physics (IN2P3) of the CNRS, Lyn Evans, sLHC Project Leader, and Alan Astbury of the TRIUMF Laboratory will talk about the physics of the weak interaction and the discovery of the W and Z bosons. Former CERN Director-General Herwig Schopper will lecture on CERN’s accelerators from LEP to the LHC. Giovanni Bignami, former President of the Italian Space Agency, will speak about his work with Carlo Rubbia. Finally, Hans Joachim Schellnhuber of the Potsdam Institute for Climate Research and Sven Kul...
DEFF Research Database (Denmark)
Cox, Geoff
Speaking Code begins by invoking the “Hello World” convention used by programmers when learning a new language, helping to establish the interplay of text and code that runs through the book. Interweaving the voice of critical writing from the humanities with the tradition of computing and software...
2014-12-01
QPSK Gaussian channels . .......................................................................... 39 vi 1. INTRODUCTION Forward error correction (FEC...Capacity of BSC. 7 Figure 5. Capacity of AWGN channel . 8 4. INTRODUCTION TO POLAR CODES Polar codes were introduced by E. Arikan in [1]. This paper...Under authority of C. A. Wilgenbusch, Head ISR Division EXECUTIVE SUMMARY This report describes the results of the project “More reliable wireless
Monte Carlo Techniques for Nuclear Systems - Theory Lectures
Energy Technology Data Exchange (ETDEWEB)
Brown, Forrest B. [Los Alamos National Lab. (LANL), Los Alamos, NM (United States). Monte Carlo Methods, Codes, and Applications Group; Univ. of New Mexico, Albuquerque, NM (United States). Nuclear Engineering Dept.
2016-11-29
These are lecture notes for a Monte Carlo class given at the University of New Mexico. The following topics are covered: course information; nuclear eng. review & MC; random numbers and sampling; computational geometry; collision physics; tallies and statistics; eigenvalue calculations I; eigenvalue calculations II; eigenvalue calculations III; variance reduction; parallel Monte Carlo; parameter studies; fission matrix and higher eigenmodes; doppler broadening; Monte Carlo depletion; HTGR modeling; coupled MC and T/H calculations; fission energy deposition. Solving particle transport problems with the Monte Carlo method is simple - just simulate the particle behavior. The devil is in the details, however. These lectures provide a balanced approach to the theory and practice of Monte Carlo simulation codes. The first lectures provide an overview of Monte Carlo simulation methods, covering the transport equation, random sampling, computational geometry, collision physics, and statistics. The next lectures focus on the state-of-the-art in Monte Carlo criticality simulations, covering the theory of eigenvalue calculations, convergence analysis, dominance ratio calculations, bias in Keff and tallies, bias in uncertainties, a case study of a realistic calculation, and Wielandt acceleration techniques. The remaining lectures cover advanced topics, including HTGR modeling and stochastic geometry, temperature dependence, fission energy deposition, depletion calculations, parallel calculations, and parameter studies. This portion of the class focuses on using MCNP to perform criticality calculations for reactor physics and criticality safety applications. It is an intermediate level class, intended for those with at least some familiarity with MCNP. Class examples provide hands-on experience at running the code, plotting both geometry and results, and understanding the code output. The class includes lectures & hands-on computer use for a variety of Monte Carlo calculations
Energy Technology Data Exchange (ETDEWEB)
Viana, Rodrigo Sartorelo Salemi
2014-07-01
The NSECT (Neutron Stimulated Emission Computed Tomography) figures as a new spectrographic technique able to evaluate in vivo the concentration of elements using the inelastic scattering reaction (n,n'). Since its introduction, several improvements have been proposed with the aim of investigating applications for clinical diagnosis and reduction of absorbed dose associated with CT acquisition. In this context, two new diagnostic applications are presented using spectroscopic and tomographic approaches from NSECT. A new methodology has also been proposed to optimize