Reliability of impedance cardiography in measuring central haemodynamics
DEFF Research Database (Denmark)
Mehlsen, J; Bonde, J; Stadeager, C;
1991-01-01
The purpose of the study described here was to investigate the reliability of impedance cardiography (IC) in measuring cardiac output (CO) and central blood volume. Absolute values and changes in these variables obtained by impedance cardiography and by isotope- or thermodilution techniques were...... healthy subjects and in 25 unmedicated patients with ischaemic heart disease. We obtained significant correlations between absolute values (y = 0.68x + 1.48) and changes (y = 1.00x + 0.0003) in CO measured by IC and isotope- or thermodilution. IC significantly overestimated absolute values of CO (P less...... suitable for repeated measurements in studies on the haemodynamic effects of physiological or pharmacological intervention. Impedance cardiography is sufficiently reliable for comparison of absolute values of CO between different groups of patients. We cannot recommend impedance cardiography for...
[Cardiac output monitoring by impedance cardiography in cardiac surgery].
Shimizu, H; Seki, S; Mizuguchi, A; Tsuchida, H; Watanabe, H; Namiki, A
1990-04-01
The cardiac output monitoring by impedance cardiography, NCCOM3, was evaluated in adult patients (n = 12) who were subjected to coronary artery bypass grafting. Values of cardiac output measured by impedance cardiography were compared to those by the thermodilution method. Changes of base impedance level used as an index of thoracic fluid volume were also investigated before and after cardiopulmonary bypass (CPB). Correlation coefficient (r) of the values obtained by thermodilution with impedance cardiography was 0.79 and the mean difference was 1.29 +/- 16.9 (SD)% during induction of anesthesia. During the operation, r was 0.83 and the mean difference was -14.6 +/- 18.7%. The measurement by impedance cardiography could be carried out through the operation except when electro-cautery was used. Base impedance level before CPB was significantly lower as compared with that after CPB. There was a negative correlation between the base impedance level and central venous pressure (CVP). No patients showed any signs suggesting lung edema and all the values of CVP, pulmonary artery pressure and blood gas analysis were within normal ranges. From the result of this study, it was concluded that cardiac output monitoring by impedance cardiography was useful in cardiac surgery, but further detailed examinations will be necessary on the relationship between the numerical values of base impedance and the clinical state of the patients. PMID:2362347
Comparison of Methods for Stroke Volume Computing from Impedance Cardiography
Czech Academy of Sciences Publication Activity Database
Jurák, Pavel; Halámek, Josef; Vondra, Vlastimil; Kára, T.; Nykodým, J.; Číp, Ondřej; Eisenberger, M.; Leinveber, P.; Fráňa, P.; Meluzín, J.; Souček, M.
Brno : Brno University of Technology, 2004, s. 69-71. ISBN 80-214-2633-0. ISSN 1211-412X. [Biosignal 2004 /17./. Brno (CZ), 23.06.2004-25.06.2004] R&D Projects: GA ČR GA102/02/1339 Keywords : Impedance Cardiography * Stroke Volume Subject RIV: FA - Cardiovascular Diseases incl. Cardiotharic Surgery
Coherent ensemble averaging techniques for impedance cardiography
Hurwitz, Barry E.; Shyu, Liang-Yu; Reddy, Sridhar P; Schneiderman, Neil; Nagel, Joachim H.
1990-01-01
EKG synchronized ensemble averaging of the impedance cardiogram tends to blur or suppress signal events due to signal jitter or event latency variability. Although ensemble averaging provides some improvement in the stability of the signal and signal to noise ratio under conditions of nonperiodic influences of respiration and motion, coherent averaging techniques were developed to determine whether further enhancement of the impedance cardiogram could be obtained. Physiological signals were o...
Czech Academy of Sciences Publication Activity Database
Jurák, Pavel; Halámek, Josef; Vondra, Vlastimil; Viščor, Ivo; Lipoldová, J.; Plachý, M.
2010-01-01
Roč. 37, - (2010), s. 1051-1054. ISSN 0276-6574 R&D Projects: GA AV ČR IAA200650801 Institutional research plan: CEZ:AV0Z20650511 Keywords : respiratory effect * thoracic impedance signal * impedance cardiography Subject RIV: JA - Electronics ; Optoelectronics, Electrical Engineering http://cinc.mit.edu/archives/2010/pdf/1051.pdf
The IMPACT shirt: textile integrated and portable impedance cardiography.
Ulbrich, Mark; Mühlsteff, Jens; Sipilä, Auli; Kamppi, Merja; Koskela, Anne; Myry, Manu; Wan, Tingting; Leonhardt, Steffen; Walter, Marian
2014-06-01
Measurement of hemodynamic parameters such as stroke volume (SV) via impedance cardiography (ICG) is an easy, non-invasive and inexpensive way to assess the health status of the heart. We present a possibility to use this technology for monitoring risk patients at home. The IMPACT Shirt (IMPedAnce Cardiography Textile) has been developed with integrated textile electrodes and textile wiring, as well as with portable miniaturized hardware. Several textile materials were characterized in vitro and in vivo to analyze their performance with regard to washability, and electrical characteristics such as skin-electrode impedance, capacitive coupling and subjective tactile feeling. The small lightweight hardware measures ECG and ICG continuously and transmits wireless data via Bluetooth to a mobile phone (Android) or PC for further analysis. A lithium polymer battery supplies the circuit and can be charged via a micro-USB. Results of a proof-of-concept trial show excellent agreement between SV assessed by a commercial device and the developed system. The IMPACT Shirt allows monitoring of SV and ECG on a daily basis at the patient's home. PMID:24846072
The IMPACT shirt: textile integrated and portable impedance cardiography
International Nuclear Information System (INIS)
Measurement of hemodynamic parameters such as stroke volume (SV) via impedance cardiography (ICG) is an easy, non-invasive and inexpensive way to assess the health status of the heart. We present a possibility to use this technology for monitoring risk patients at home. The IMPACT Shirt (IMPedAnce Cardiography Textile) has been developed with integrated textile electrodes and textile wiring, as well as with portable miniaturized hardware. Several textile materials were characterized in vitro and in vivo to analyze their performance with regard to washability, and electrical characteristics such as skin-electrode impedance, capacitive coupling and subjective tactile feeling. The small lightweight hardware measures ECG and ICG continuously and transmits wireless data via Bluetooth to a mobile phone (Android) or PC for further analysis. A lithium polymer battery supplies the circuit and can be charged via a micro-USB. Results of a proof-of-concept trial show excellent agreement between SV assessed by a commercial device and the developed system. The IMPACT Shirt allows monitoring of SV and ECG on a daily basis at the patient’s home. (paper)
Daralammouri, Yunis; Ayoub, Khubaib; Badrieh, Najwan; Lauer, Bernward
2016-01-01
Background Impedance cardiography (IC) is a noninvasive modality that utilizes changes in impedance across the thorax to assess hemodynamic parameters, including stroke volume (SV). This study compared aortic valve area (AVA) as assessed by a hybrid approach of transthoracic echocardiography (TTE) and impedance cardiography (IC) to AVA determined at cardiac catheterization using the Gorlin equation. Methods A total of 30 patients with moderate to severe aortic stenosis underwent AVA measureme...
Hemodynamic evaluation in pregnancy: limitations of impedance cardiography
International Nuclear Information System (INIS)
Impedance cardiography (ICG) has been proposed to estimate the stroke volume (SV) and cardiac output (CO) in various medical indications. The aim of this study was to explore the reliability of ICG during pregnancy with respect to SV and CO measurements. Blood pressure, heart rate and thoracic impedance were monitored during the course of pregnancy and related ICG patterns analyzed. We determined thoracic impedance (Z0), left ventricular ejection time (LVET) and the maximum value of the first derivative of the impedance waveform (dZ/dt)max beat-to-beat. The two main components of non-invasive SV estimation, LVET and (dZ/dt)max, decreased in week 35 relative to week 12 by 7.6% ± 4.8% and by 36.0% ± 14.0%, respectively, based on a decrease in R–R interval of 9.9% ± 9.7%, whereas changes in Z0 did not play a significant role. Furthermore, the decrease of (dZ/dt)max was greater in the third compared to the second trimester of pregnancy. Taking into account an 18.2% ± 6.8% body weight increase, the calculated SV decreased in week 35 relative to week 12 by 19.2% ± 14.3%. The normalized term ((dZ/dt)max)/Z0 explained 85–94% of the variance of SV change after week 20. These results were confirmed for fixed R–R interval at a constant value across the repeated measurements. Based on beat-to-beat analysis of the main components of non-invasive SV estimation by ICG, measured repeatedly throughout normal pregnancy, we conclude that SV calculations render invalid data. Our findings strongly suggest that CO cannot be reliably assessed with ICG in pregnant women. (paper)
International Nuclear Information System (INIS)
The cardiac resynchronization therapy is an effective treatment for systolic failure patients. Independent electrical stimulation of left and right ventricle corrects mechanical ventricular dyssynchrony. About 30–40% treated patients do not respond to therapy. In order to improve clinical outcome authors propose the two channels impedance cardiography for assessment of ventricular dyssynchrony. The proposed method is intended for validation of patients diagnosis and optimization of pacemaker settings for cardiac resynchronization therapy. The preliminary study has showed that bichannel impedance cardiography is a promising tool for assessment of ventricular dyssynchrony.
Blood Pressure and Impedance Cardiography duríng Tilt Table Test
Czech Academy of Sciences Publication Activity Database
Jurák, Pavel; Halámek, Josef; Vondra, Vlastimil; Plachý, M.; Fráňa, P.; Leinveber, Pavel
2009-01-01
Roč. 36, - (2009), s. 429-432. ISSN 0276-6574 R&D Projects: GA AV ČR IAA200650801 Institutional research plan: CEZ:AV0Z20650511 Keywords : blood pressure * heart rate * thoracic impedance cardiography Subject RIV: JA - Electronics ; Optoelectronics, Electrical Engineering http://cinc.mit.edu/archives/2009/pdf/0429.pdf
Bernstein, Donald P.
2010-01-01
Impedance cardiography (ICG) is a branch of bioimpedance pimarily concerned with the determination of left ventricular stroke volume (SV). As implemented, using the transthoracic approach, the technique involves applying a current field longitudinally across a segment of thorax by means of a constant magnitude, high frequency, low amplitude alternating current (AC). By Ohm's Law, the voltage difference measured within the current field is proportional to the electrical impedance Z (&Omega...
Czech Academy of Sciences Publication Activity Database
Viščor, Ivo; Jurák, Pavel; Vondra, Vlastimil; Halámek, Josef; Leinveber, Pavel
2009-01-01
Roč. 36, - (2009), s. 749-751. ISSN 0276-6574 R&D Projects: GA AV ČR IAA200650801; GA ČR GP102/07/P425 Institutional research plan: CEZ:AV0Z20650511 Keywords : Mueller maneuver * impedance cardiography * congestive heart failure Subject RIV: JA - Electronics ; Optoelectronics, Electrical Engineering http://cinc.mit.edu/archives/2009/pdf/0749.pdf
Juan Carlos Márquez Ruiz; Markus Rempfler; Fernando Seoane; Kaj Lindecrantz
2013-01-01
During the last decades the use of Electrical Bioimpedance (EBI) in the medical field has been subject of extensive research, especially since it is an affordable, harmless and non-invasive technology. In some specific applications such as body composition assessment where EBI has proven a good degree of effectiveness and reliability, the use of textile electrodes and measurement garments have shown a good performance and reproducible results. Impedance Cardiography (ICG) is another modality ...
Development of a wearable multi-frequency impedance cardiography device.
Weyer, Sören; Menden, Tobias; Leicht, Lennart; Leonhardt, Steffen; Wartzek, Tobias
2015-02-01
Cardiovascular diseases as well as pulmonary oedema can be early diagnosed using vital signs and thoracic bio-impedance. By recording the electrocardiogram (ECG) and the impedance cardiogram (ICG), vital parameters are captured continuously. The aim of this study is the continuous monitoring of ECG and multi-frequency ICG by a mobile system. A mobile measuring system, based on 'low-power' ECG, ICG and an included radio transmission is described. Due to the high component integration, a board size of only 6.5 cm×5 cm could be realized. The measured data can be transmitted via Bluetooth and visualized on a portable monitor. By using energy-efficient hardware, the system can operate for up to 18 hs with a 3 V battery, continuously sending data via Bluetooth. Longer operating times can be realized by decreased transfer rates. The relative error of the impedance measurement was less than 1%. The ECG and ICG measurements allow an approximate calculation of the heart stroke volume. The ECG and the measured impedance showed a high correlation to commercial devices (r=0.83, p<0.05). In addition to commercial devices, the developed system allows a multi-frequency measurement of the thoracic impedance between 5-150 kHz. PMID:25559781
Oleksiak, Justyna; Cybulski, Gerard
2014-11-01
The aim of this work was to create a computer program, written in LabVIEW, which enables the visualization and analysis of hemodynamic parameters. It allows the user to import data collected using ReoMonitor, an ambulatory monitoring impedance cardiography (AICG) device. The data include one channel of the ECG and one channel of the first derivative of the impedance signal (dz/dt) sampled at 200Hz and the base impedance signal (Z0) sampled every 8s. The program consist of two parts: a bioscope allowing the presentation of traces (ECG, AICG, Z0) and an analytical portion enabling the detection of characteristic points on the signals and automatic calculation of hemodynamic parameters. The detection of characteristic points in both signals is done automatically, with the option to make manual corrections, which may be necessary to avoid "false positive" recognitions. This application is used to determine the values of basic hemodynamic variables: pre-ejection period (PEP), left ventricular ejection time (LVET), stroke volume (SV), cardiac output (CO), and heart rate (HR). It leaves room for further development of additional features, for both the analysis panel and the data acquisition function.
A portable device to assess underwater changes of cardio dynamic variables by impedance cardiography
International Nuclear Information System (INIS)
Data concerning heart rate (HR), stroke volume (SV), and cardiac output (CO) during dynamic apnoea (DA) were collected from 10 healthy male, elite divers by means of an impedance cardiograph adapted to the underwater environment (C. O. Re., from 2C Technologies Inc, Italy). Three trials were performed by the divers in a 3-m-deep pool with a water temperature of 25°C: 3-minute head-out immersion during normal breathing (A), till exhaustion immersed at the surface (B) and at 3m depth (C). Both B and C conditions did not led to changes in HR, SV and CO compared to A. Data indicate that typical diving response consisting in a reduction of HR, SV and CO was not present during DA, probably due to sympathetic activation induced by exercise during DA, which partially obscured the effects of the diving response. Moreover, this study highlights the innovative role of our portable, impedance cardiography device, i.e. the C. O. Re., in easily assessing cardiodynamic changes in subjects engaged in exercise schedules including phases of underwater, dynamic apnoea.
Directory of Open Access Journals (Sweden)
Juan Carlos Márquez Ruiz
2013-10-01
Full Text Available During the last decades the use of Electrical Bioimpedance (EBI in the medical field has been subject of extensive research, especially since it is an affordable, harmless and non-invasive technology. In some specific applications such as body composition assessment where EBI has proven a good degree of effectiveness and reliability, the use of textile electrodes and measurement garments have shown a good performance and reproducible results. Impedance Cardiography (ICG is another modality of EBI that can benefit from the implementation and use of wearable sensors. ICG technique is based on continuous impedance measurements of a longitudinal segment across the thorax taken at a single frequency. The need for specific electrode placement on the thorax and neck can be easily ensured with the use of a garment with embedded textile electrodes, textrodes. The first step towards the implementation of ICG technology into a garment is to find out if ICG measurements with textile sensors give a good enough quality of the signal to allow the estimation of the fundamental ICG parameters. In this work, the measurement performance of a 2-belt set with incorporated textrodes for thorax and neck was compared against ICG measurements obtained with Ag/AgCl electrodes. The analysis was based on the quality of the fundamental ICG signals (∆Z, dZ/dt and ECG, systolic time intervals and other ICG parameters. The obtained results indicate the feasibility of using textrodes for ICG measurements with consistent measurements and relatively low data dispersion. Thus, enabling the development of measuring garments for ICG measurements.
Directory of Open Access Journals (Sweden)
Donald P Bernstein
2010-01-01
Full Text Available Impedance cardiography (ICG is a branch of bioimpedance pimarily concerned with the determination of left ventricular stroke volume (SV. As implemented, using the transthoracic approach, the technique involves applying a current field longitudinally across a segment of thorax by means of a constant magnitude, high frequency, low amplitude alternating current (AC. By Ohm's Law, the voltage difference measured within the current field is proportional to the electrical impedance Z (Ω. Without ventilatory or cardiac activity, Z is known as the transthoracic, static base impedance Z0. Upon ventricular ejection, a characteristic time dependent cardiac-synchronous pulsatile impedance change is obtained, ΔZ(t, which, when placed electrically in parallel with Z0, constitutes the time-variable total transthoracic impedance Z(t. ΔZ(t represents a dual-element composite waveform, which comprises both the radially-oriented volumetric expansion of and axially-directed forward blood flow within both great thoracic arteries. In its majority, however, ΔZ(t is known to primarily emanate from the ascending aorta. Conceptually, commonly implemented methods assume a volumetric origin for the peak systolic upslope of ΔZ(t, (i.e. dZ/dtmax, with the presumed units of Ω·s-1. A recently introduced method assumes the rapid ejection of forward flowing blood in earliest systole causes significant changes in the velocity-induced blood resistivity variation (Δρb(t, Ωcm·s-1, and it is the peak rate of change of the blood resistivity variation dρb(t/dtmax (Ωcm·s-2 that is the origin of dZ/dtmax. As a consequence of dZ/dtmax peaking in the time domain of peak aortic blood acceleration, dv/dtmax (cm·s-2, it is suggested that dZ/dtmax is an ohmic mean acceleration analog (Ω·s-2 and not a mean flow or velocity surrogate as generally assumed. As conceptualized, the normalized value, dZ/dtmax/Z0, is a dimensionless ohmic mean acceleration equivalent (s-2
Design and evaluation of a portable device for the measurement of bio-impedance cardiography
Shi, Qinghai; Heinig, Andreas; Kanoun, Olfa
2011-01-01
Electrical impedance of biological matter is known as electrical bio-impedance or simply as bio-impedance. Bio-impedance devices are of great value for monitoring the pathological and physiological status of biological tissues in clinical and home environments. The technological progress in instrumentation has significantly contributed to the progress that has been observed during the last past decades in impedance spectroscopy and electrical impedance cardiograph. Although bio-impedance is n...
Borodiciene, Jurgita; Gudaityte, Jurate; Macas, Andrius
2015-01-01
Background Although the prone position providing better exposure for anorectal surgery is required it can cause a reduction of cardiac output and cardiac index. The goal was to compare haemodynamic changes assessed by impedance cardiography during anorectal surgery under low-dose spinal anaesthesia in lithotomy and jack-knife position. Methods The prospective randomized controlled study included 104, ASA I-II adult patients admitted for elective minor anorectal surgery, assigned to be perform...
Variable impedance cardiography waveforms: how to evaluate the preejection period more accurately
Ermishkin, V. V.; Kolesnikov, V. A.; Lukoshkova, E. V.; Mokh, V. P.; Sonina, R. S.; Dupik, N. V.; Boitsov, S. A.
2012-12-01
Impedance method has been successfully applied for left ventricular function assessment during functional tests. The preejection period (PEP), the interval between Q peak in ECG and a specific mark on impedance cardiogram (ICG) which corresponds to aortic valve opening, is an important indicator of the contractility state and its neurogenic control. Accurate identification of ejection onset by ICG is often problematic, especially in the cardiologic patients, due to peculiar waveforms. An essential obstacle is variability of the shape of the ICG waveform during the exercise and subsequent recovery. A promissing solution can be introduction of an additional pulse sensor placed in the nearby region. We tested this idea in 28 healthy subjects and 6 cardiologic patients using a dual-channel impedance cardiograph for simultaneous recording from the aortic and neck regions, and an earlobe photoplethysmograph. Our findings suggest that incidence of abnormal complicated ICG waveforms increases with age. The combination of standard ICG with ear photoplethysmography and/or additional impedance channel significantly improves the efficacy and accuracy of PEP estimation.
Assessing cardiac preload by the Initial Systolic Time Interval obtained from impedance cardiography
Directory of Open Access Journals (Sweden)
Jan H Meijer
2010-01-01
Full Text Available The Initial Systolic Time Interval (ISTI, obtained from the electrocardiogram (ECG and impedance cardiogram (ICG, is considered to be a measure for the time delay between the electrical and mechanical activity of the heart and reflects an early active period of the cardiac cycle. The clinical relevance of this time interval is subject of study. This paper presents preliminary results of a pilot study investigating the use of ISTI in evaluating and predicting the circulatory response to fluid administration in patients after coronary artery bypass graft surgery, by comparing ISTI with cardiac output (CO responsiveness. Also the use of the pulse transit time (PTT, earlier recommended for this purpose, is investigated. The results show an inverse relationship between ISTI and CO at all moments of fluid administration and also an inverse relationship between the changes ΔISTI and ΔCO before and after full fluid administration. No relationships between PTT and CO or ΔPTT and ΔCO were found. It is concluded that ISTI is dependent upon preload, and that ISTI has the potential to be used as a clinical parameter assessing preload.
International Nuclear Information System (INIS)
The Initial Systolic Time Interval (ISTI), obtained from the electrocardiogram and impedance cardiogram, is considered to be a measure for the time delay between the electrical and mechanical activity of the heart. This time delay is influenced by the sympathetic nerve system. Therefore, an observational study was performed in a group of patients (SCI) with spinal cord injuries. The relationship between the ISTI and the total heart cycle (RR-interval) was established by varying the RR-interval using an exercise stimulus to increase the heart rate. The slope of this relationship was observed to be significantly higher in the SCI-group as compared with a control group, although there was no difference in ISTI in the range of common heart rates during the test between the groups. This slope and the ISTI was observed to be significantly different in an acute patient having a recent spinal cord injury at a high level. Because of the variety in injury levels and incompleteness of the injuries further, more specific research is necessary to draw decisive conclusions with respect to the contribution of autonomic nervous control on the ISTI in SCI, although the present observations are notable.
SASSMANNOVÁ, Anna
2007-01-01
Echocardiography is an investigation of heart via scan. This enables to intend the moving and the locality of heart structures via scan pulse waves which are repulsed with acoustic interfaces. Impedance measuring of the thorax hemodynamics is based on changes of electrical impedance. These changes happen mainly because of the heart function. By its rhytmical function the heart periodically changes the conditions of blood flow through all vessels. By this we can explain periodical changes of i...
The Device for the Two-Channel Impedance Cardiography
Czech Academy of Sciences Publication Activity Database
Vondra, Vlastimil; Verner, Petr; Viščor, Ivo
Brno : University of Technology, 2006, s. 116-118. ISBN 80-214-3152-0. ISSN 1211-412X. [BIOSIGNAL 2006 /18./. Brno (CZ), 28.06.2006-30.06.2006] R&D Projects: GA ČR GA102/06/0136 Institutional research plan: CEZ:AV0Z20650511 Keywords : bioimpedance * cardiac output Subject RIV: FS - Medical Facilities ; Equipment
Directory of Open Access Journals (Sweden)
Jon Kepa Balparda
2012-04-01
Full Text Available La cardiografía de impendancia (CGI representa un método no invasivo para la evaluación del estado hemodinámico latido a latido. Aunque se introdujo por primera vez hace más de 40 años, la CGI ha mostrado un resurgimiento en la última década, a partir de una serie de estudios clínicos que han demostrado su precisión en la estimación del volumen latido, tanto contra el "gold-standard" invasivo (termodilución, como contra los métodos de referencia no invasivos (ecocardiografía. Diversos estudios demuestran la utilidad de esta técnica en el manejo del paciente con falla cardíaca y en el enfoque diagnóstico y terapéutico de la hipertensión arterial, por lo cual constituyen actualmente dos de las aplicaciones clínicas más importantes de la CGI. En falla cardiaca, los cambios en el volumen de líquido del tórax y del gasto cardíaco evaluados por CGI, han demostrado ser predictores de descompensación aguda, incluso semanas antes del inicio de la sintomatología respiratoria; además, permiten identificar el origen cardiogénico o respiratorio de la disnea cuando el examen físico y los demás paraclínicos no son concluyentes. En los pacientes con hipertensión arterial no controlada o resistente, la CGI permite realizar una mejor caracterización del fenotipo hipertensivo y elegir la estrategia farmacológica más específica para intervenir la alteración hemodinámica predominante (resistencia vascular vs. gasto cardiaco elevado. En este artículo se realiza una revisión de los principios biofísicos de la CGI y su utilidad en la evaluación no invasiva del estado hemodinámico, así como una evaluación crítica de la literatura que da soporte a su aplicación clínica en el tratamiento de la falla cardíaca y la hipertensión arterial.Impedance cardiography (ICG represents a non-invasive method for hemodynamic assessment in a beat-to-beat basis. Since its introduction more than forty years ago, a renewed interest in the
Development of a Patch-Type Cardiography Monitoring System for Human Healthcare
Hamada, Hiroyuki; Jiang, Yonggang; Iga, Yuki; Okochi, Sayaka; Kanda, Kensuke; Fujita, Takayuki; Higuchi, Kohei; Maenaka, Kazusuke
This paper describes a wireless cardiography monitoring system for human healthcare. The system consists of a cardiography sensor module and a radio frequency (RF) module for data communication. And the P, Q, R, S, T signals was acquired by using the prototype cardiography sensor. Real-time cardiography monitoring was demonstrated with a RF base station which was located 10 m away from the sensing object.
Wavelet analysis of the impedance cardiogram waveforms
Podtaev, S.; Stepanov, R.; Dumler, A.; Chugainov, S.; Tziberkin, K.
2012-12-01
Impedance cardiography has been used for diagnosing atrial and ventricular dysfunctions, valve disorders, aortic stenosis, and vascular diseases. Almost all the applications of impedance cardiography require determination of some of the characteristic points of the ICG waveform. The ICG waveform has a set of characteristic points known as A, B, E ((dZ/dt)max) X, Y, O and Z. These points are related to distinct physiological events in the cardiac cycle. Objective of this work is an approbation of a new method of processing and interpretation of the impedance cardiogram waveforms using wavelet analysis. A method of computer thoracic tetrapolar polyrheocardiography is used for hemodynamic registrations. Use of original wavelet differentiation algorithm allows combining filtration and calculation of the derivatives of rheocardiogram. The proposed approach can be used in clinical practice for early diagnostics of cardiovascular system remodelling in the course of different pathologies.
Czech Academy of Sciences Publication Activity Database
Jurák, Pavel; Halámek, Josef; Vondra, Vlastimil; Viščor, Ivo; Lipoldová, J.; Plachý, M.
Vol. 37. New York: IEEE, 2010, s. 1051-1054. ISBN 978-1-4244-7318-2. ISSN 0276-6574. [Annual Conference of the Computing-in- Cardiology /37./. Belfast (IE), 26.09.2010-29.09.2010] R&D Projects: GA AV ČR IAA200650801 Institutional support: RVO:68081731 Keywords : electrical bioimpendance * cardiac-output Subject RIV: FS - Medical Facilities ; Equipment
DEFF Research Database (Denmark)
Osbak, Philip S; Henriksen, Jens Henrik Sahl; Kofoed, Klaus F;
2011-01-01
Abstract Background. Atrial fibrillation (AF) is associated with significant morbidity and mortality. To test the effect of interventions, knowledge of cardiac output (CO) is important. However, the irregular heart rate might cause some methods for determination of CO to have inherent weaknesses...
Energy Technology Data Exchange (ETDEWEB)
Kelbaek, H.; Svendsen, J.H.; Aldershvile, J.; Folke, K.; Nielsen, S.L.
The stroke volume (SV) was determined by first passage radionuclide cardiography and the left ventricular ejection fraction (LVEF) by multigated radionuclide cardiography in 20 patients with ischemic heart disease. The results were evaluated against those obtained by the invasive dye dilution or thermodilution and left ventricular cardioangiographic techniques. In a paired comparison the mean difference between the invasive and radionuclide SV was -1 ml (SED 3.1) with a correlation coefficient of 0.83 (p < 0.01). Radionuclide LVEF values also correlated well with cardioangiographic measurements, r = 0.93 (p < 0.001). LVEF determined by multigated radionuclide cardiography was, however, significantly lower than when measured by cardioangiography, the mean difference being 6% (p < 0.001). These findings suggest that radionuclide determinations of SV and LVEF are reliable. The discrepancy between the non-invasive and invasive LVEF values raises the question, whether LVEF is overestimated by cardioangiography or underestimated by radionuclide cardiography.
Two-channel high dynamic range bioimpedance monitor for cardiography
Czech Academy of Sciences Publication Activity Database
Viščor, Ivo; Vondra, Vlastimil; Halámek, Josef
Berlin : Springer, 2007, s. 225-228. ISBN 978-3-540-92840-9. [IFMBE: Electrical Bioimpedance /13./ and Electrical Impedance Tomography /8./. Graz (AT), 29.08.2007-02.09.2007] R&D Projects: GA ČR GP102/07/P425 Institutional research plan: CEZ:AV0Z20650511 Keywords : bioimpedance monitor * dynamic range * reactance * digital quadrature demodulation Subject RIV: FA - Cardiovascular Diseases incl. Cardiotharic Surgery
Couroussé, Damien
2007-01-01
Mechanical impedance is a transposition to mechanics of the term impedance that is used and defined in circuit theory. The theory of circuit (theory of Kirchhoff networks) is basically applicable to electric networks but can be considered more generally as a unifying simplified theory of physics available in several domains like mechanics, electromagnetism, aero-acoustics and fluids mechanics.
[Evaluation of orthostatic regulation by saddle support test using thoracic impedance].
Gugova, F K; Lapin, V V
2002-01-01
We investigated 21 healthy volunteers (10 males and 11 females, mean age 23 +/- 4 years). All the subjects have undergone two 20 min head-up tilt tests using tilt table "TRI W.G. inc." (USA): the first with footplate support and the second with bicycle saddle. Thoracic electrical impedance was measured using impedance cardiography according to Kubicek et al. The protocols included an initial period of 20 min of supine rest while baseline thoracic impedance, blood pressure and heart rate were recorded and then followed by a tilt to 65 degrees. Changes of impedance were measured at min 1, 2, 3, 5, 7, 10, 15, 20 after the procedure. Women had higher values of thoracic impedance both at rest and during the tilt test than men. The value of impedance of the chest negatively correlated with the body mass index. We suppose that an increase of impedance more than 15% may be related with pathological venous pooling. Thoracic impedance may be used to monitor changes of thoracic fluid volumes with posture and possibly to assess orthostatic regulation. The contribution of leg muscles in orthostatic regulation does not reflect values of thoracic impedance. PMID:12152425
International Nuclear Information System (INIS)
Computer assisted diagnosis using analysis of medical images is an area of active research in health informatics. This paper proposes a technique for indication of heart diseases by using information related to shapes of the left ventricle (LV). LV boundaries are tracked from echo-cardiography images taken from LV short axis view, corresponding to two disease conditions viz. dilated cardiomyopathy and hypertrophic cardiomyopathy, and discriminated from the normal condition. The LV shapes are modeled using shape histograms generated by plotting the frequency of normalized radii lengths drawn from the centroid to the periphery, against a specific number of bins. A 3-layer neural network activated by a log-sigmoid function is used to classify the shape histograms into one of the three classes. Experimentations on a dataset of 240 images show recognition accuracies of the order of 80%.
Influence of physiological sources on the impedance cardiogram analyzed using 4D FEM simulations
International Nuclear Information System (INIS)
Impedance cardiography is a simple and inexpensive method to acquire data on hemodynamic parameters. This study analyzes the influence of four dynamic physiological sources (aortic expansion, heart contraction, lung perfusion and erythrocyte orientation) on the impedance signal using a model of the human thorax with a high temporal resolution (125 Hz) based on human MRI data. Simulations of electromagnetic fields were conducted using the finite element method. The ICG signal caused by these sources shows very good agreement with the measured signals (r = 0.89). Standard algorithms can be used to extract characteristic points to calculate left ventricular ejection time and stroke volume (SV). In the presented model, the calculated SV equals the implemented left ventricular volume change of the heart. It is shown that impedance changes due to lung perfusion and heart contraction compensate themselves, and that erythrocyte orientation together with the aortic impedance basically form the ICG signal while taking its characteristic morphology from the aortic signal. The model is robust to conductivity changes of tissues and organ displacements. In addition, it reflects the multi-frequency behavior of the thoracic impedance. (paper)
The Influence of Heart and Lung Dynamics on the Impedance Cardiogram — A Simulative Analysis
Directory of Open Access Journals (Sweden)
M. Ulbrich
2011-01-01
Full Text Available Impedance cardiography (ICG is a simple and cheap method for acquiring hemodynamic parameters. Unfortunately, not all physiological influences on the ICG signal have yet been identified. In this work, the influence of heart and lung dynamics is analyzed using a simplified model of the human thorax with high temporal resolution. Simulations are conducted using the finite integration technique (FIT with temporal resolution of 103 Hz. It is shown that changes in heart volume as well as conductivity changes of the lung have a high impact on the ICG signal, if analyzed separately. Considering the sum signal of both physiological sources, it can be shown that they compensate each other and thus do not contribute to the signal. This finding supports Kubicek’s model.
Impedance model for nanostructures
Directory of Open Access Journals (Sweden)
R. S. Akhmedov
2007-06-01
Full Text Available The application of the impedance model for nanoelectronic quantum-mechanical structures modelling is described. Characteristics illustrating the efficiency of the model are presented.
Impedance and Collective Effects
Metral, E; Rumolo, R; Herr, W
2013-01-01
This document is part of Subvolume C 'Accelerators and Colliders' of Volume 21 'Elementary Particles' of Landolt-Börnstein - Group I 'Elementary Particles, Nuclei and Atoms'. It contains the Chapter '4 Impedance and Collective Effects' with the content: 4 Impedance and Collective Effects Introduction 4.1 Space Charge 4.2 Wake Fields and Impedances 4.3 Coherent Instabilities 4.4 Landau Damping 4.5 Two-Stream Effects (Electron Cloud and Ions) 4.6 Beam-Beam Effects 4.7 Numerical Modelling
Energy Technology Data Exchange (ETDEWEB)
Georges, R.; Vernejoul, P. de; Raynaud, C.; Blanchon, P.; Kellershohn, C.; Turiaf, J. [Commissariat a l' Energie Atomique, Saclay (France). Centre d' Etudes Nucleaires
1961-07-01
The authors used gamma cardiology during the abatement of 16 cases of asthma with a view to detecting heart attacks not otherwise visible with routine methods of examination: clinical, radiological and electro-cardio-graphical. In gamma cardiology, a radioactive indicator is used and its path followed in the cavities of the heart. The method makes it possible to study the circulation in the right heart, the pulmonary crossing, and the left heart, as well as evaluation of the heart-flow. As a result of their investigations the authors, after having discussed the significance of the data obtained with the method, suggest that it is possible by the use of gamma cardiography during the abatement of an asthma attack: 1- To confirm attacks of the right heart which have already been detected by ordinary methods. 2- To confirm the presence of modifications in the recorded curves which suggest, in the case of the left heart, possible attack; such on attack is also indicated, in a small number of cases, by electrocardiography curves. Some considerations are put forward by the authors concerning the physiopathology of attacks on the left heart. (authors) [French] Les auteurs ont fait appel a la gammacardiographie au decours de 16 cas d'etat de mal asthmatique, dans le but d'objectiver des atteintes cardiaques incapables d'etre revelees par les moyens d'exploration de routine: cliniques, radiologiques et electrocardiographiques. La gammacardiographie est un procede qui enregistre a l'aide d'un indicateur radioactif, le trajet de celui-ci dans les cavites cardiaques. Elle permet l'etude de la circulation dans le coeur droit, de la traversee pulmonaire, et le coeur gauche, de meme que le calcul du debit cardiaque. Au terme de leurs investigations, les auteurs apres avoir discute la signification des donnees fournies par la methode admettent que la gammacardiographie objective au secours de l'etat de mal asthmatique: 1- La confirmation de l
Impedance and component heating
Métral, E; Mounet, N; Pieloni, T; Salvant, B
2015-01-01
The impedance is a complex function of frequency, which represents, for the plane under consideration (longitudinal, horizontal or vertical), the force integrated over the length of an element, from a “source” to a “test” wave, normalized by their charges. In general, the impedance in a given plane is a nonlinear function of the test and source transverse coordinates, but it is most of the time sufficient to consider only the first few linear terms. Impedances can influence the motion of trailing particles, in the longitudinal and in one or both transverse directions, leading to energy loss, beam instabilities, or producing undesirable secondary effects such as excessive heating of sensitive components at or near the chamber wall, called beam-induced RF heating. The LHC performance limitations linked to impedances encountered during the 2010-2012 run are reviewed and the currently expected situation during the HL-LHC era is discussed.
Energy Technology Data Exchange (ETDEWEB)
Zotter, B. [European Organization for Nuclear Research, Geneva (Switzerland)
1996-08-01
This report describes a number of measurements and computations of the impedance of the Large Electron Positron collider LEP at CERN. The work has been performed over several years, together with D. Brandt, K. Cornelis, A. Hofmann, G. Sabbi and many others. The agreement between measurements of single bunch instabilities on the machine and computer simulations is in general excellent and gives confidence in the impedance model used. (author)
International Nuclear Information System (INIS)
The proton Storage Ring (PSR), now in operation at Los Alamos, is a fast-cycling high-current accumulator designed to produce intense 800 MeV proton pulses for driving a spallation neutron source. This paper presents graphs of calculated longitudinal and transverse coupling impedances vs frequency for various components in the PSR beamline. The impedances are estimated using simplified formulas for the effects of steps, cavities, monitor plates, and other discontinuities in the beam pipe for the ring. An examination of stability limits indicates that the longitudinal impedance per harmonic should be less than something in the hundreds of ohms range and the transverse impedance should be less than something of the order of a megohm/meter at low frequencies. For the PSR, the impedances due to the harmonic buncher and due to possible high-Q resonances in the bump magnet chambers might be significant. Simplified growth rate estimates using the real part of the transverse impedance indicate that the bumper magnet coils for the fast kicker plates might be contributing to an observed instability with onset at about 1013 protons per bunch
Kopp, Joachim; Slatyer, Tracy R; Wang, Xiao-Ping; Xue, Wei
2016-01-01
We consider a new class of thermal dark matter models, dubbed "Impeded Dark Matter", in which the mass splitting between the dark matter particles and their annihilation products is tiny. Compared to the previously proposed Forbidden Dark Matter scenario, the mass splittings we consider are much smaller, and are allowed to be either positive or negative. We demonstrate that either case can be easily realized without requiring tuning of model parameters. For negative mass splitting, we demonstrate that the annihilation cross-section for Impeded Dark Matter depends linearly on the dark matter velocity or may even be kinematically forbidden, making this scenario almost insensitive to constraints from the cosmic microwave background and from observations of dwarf galaxies. Accordingly, it may be possible for Impeded Dark Matter to yield observable signals in clusters or the Galactic center, with no corresponding signal in dwarfs. For positive mass splitting, we show that the annihilation cross-section is suppress...
Longitudinal impedance of RHIC
International Nuclear Information System (INIS)
The longitudinal impedance of the two RHIC rings has been measured using the effect of potential well distortion on longitudinal Schottky measurements. For the blue RHIC ring Im(Z/n) = 1.5±0.2?. For the yellow ring Im(Z/n) = 5.4±1?.
Gynecologic electrical impedance tomograph
Korjenevsky, A.; Cherepenin, V.; Trokhanova, O.; Tuykin, T.
2010-04-01
Electrical impedance tomography extends to the new and new areas of the medical diagnostics: lungs, breast, prostate, etc. The feedback from the doctors who use our breast EIT diagnostic system has induced us to develop the 3D electrical impedance imaging device for diagnostics of the cervix of the uterus - gynecologic impedance tomograph (GIT). The device uses the same measuring approach as the breast imaging system: 2D flat array of the electrodes arranged on the probe with handle is placed against the body. Each of the 32 electrodes of the array is connected in turn to the current source while the rest electrodes acquire the potentials on the surface. The current flows through the electrode of the array and returns through the remote electrode placed on the patient's limb. The voltages are measured relative to another remote electrode. The 3D backprojection along equipotential surfaces is used to reconstruct conductivity distribution up to approximately 1 cm in depth. Small number of electrodes enables us to implement real time imaging with a few frames per sec. rate. The device is under initial testing and evaluation of the imaging capabilities and suitability of usage.
Acoustic impedances of ear canals measured by impedance tube
DEFF Research Database (Denmark)
Ciric, Dejan; Hammershøi, Dorte
2007-01-01
During hearing sensitivity tests, the sound field is commonly generated by an earphone placed on a subject ear. One of the factors that can affect the sound transmission in the ear is the acoustic impedance of the ear canal. Its importance is related to the contribution of other elements involved...... in the transmission such as the earphone impedance. In order to determine the acoustic impedances of human ear canals, the standardized method for measurement of complex impedances used for the measurement of the audiometric earphone impedances is applied. It is based on the transfer function between...... two microphone locations in an impedance tube. The end of the tube representing the measurement plane is placed at the ear canal entrance. Thus, the impedance seen from the entrance inward is measured on 25 subjects. Most subjects participated in the previous measurement of the ratio between the...
Observations involving broadband impedance modelling
Energy Technology Data Exchange (ETDEWEB)
Berg, J.S. [Stanford Linear Accelerator Center, Menlo Park, CA (United States)
1996-08-01
Results for single- and multi-bunch instabilities can be significantly affected by the precise model that is used for the broadband impedance. This paper discusses three aspects of broadband impedance modelling. The first is an observation of the effect that a seemingly minor change in an impedance model has on the single-bunch mode coupling threshold. The second is a successful attempt to construct a model for the high-frequency tails of an r.f. cavity. The last is a discussion of requirements for the mathematical form of an impedance which follow from the general properties of impedances. (author)
Aortic Impedance in Little Mice
Reddy, Anilkumar K.; Taffet, George E.; Hartley, Craig J.
2008-01-01
The Little dwarf mouse lives 30% longer than its age-matched wild-type (WT) mouse. We determined aortic input impedance in 21 (8 Little, 13 WT) 4 month-old mice. Modulus of impedance was calculated from the Fourier transformed aortic pressure (P) and average luminal flow velocity (Vavg) as ∣Zi∣ = ∣P∣/∣Vavg∣. Characteristic impedance was estimated by averaging the 2nd-10th harmonic of the impedance moduli. We found the impedance modulus ∣Zi∣ to be similar in the 2 groups (WT vs. Little; mean±S...
RF impedance measurement calibration
International Nuclear Information System (INIS)
The intent of this note is not to explain all of the available calibration methods in detail. Instead, we will focus on the calibration methods of interest for RF impedance coupling measurements and attempt to explain: (1). The standards and measurements necessary for the various calibration techniques. (2). The advantages and disadvantages of each technique. (3). The mathematical manipulations that need to be applied to the measured standards and devices. (4). An outline of the steps needed for writing a calibration routine that operated from a remote computer. For further details of the various techniques presented in this note, the reader should consult the references
Microwave Impedance Measurement for Nanoelectronics
Directory of Open Access Journals (Sweden)
M. Randus
2011-04-01
Full Text Available The rapid progress in nanoelectronics showed an urgent need for microwave measurement of impedances extremely different from the 50Ω reference impedance of measurement instruments. In commonly used methods input impedance or admittance of a device under test (DUT is derived from measured value of its reflection coefficient causing serious accuracy problems for very high and very low impedances due to insufficient sensitivity of the reflection coefficient to impedance of the DUT. This paper brings theoretical description and experimental verification of a method developed especially for measurement of extreme impedances. The method can significantly improve measurement sensitivity and reduce errors caused by the VNA. It is based on subtraction (or addition of a reference reflection coefficient and the reflection coefficient of the DUT by a passive network, amplifying the resulting signal by an amplifier and measuring the amplified signal as a transmission coefficient by a common vector network analyzer (VNA. A suitable calibration technique is also presented.
International Nuclear Information System (INIS)
Studies have shown that blood-flow-induced change in electrical conductivity is of equal importance in assessment of the impedance cardiogram (ICG) as are volumetric changes attributed to the motion of heart, lungs and blood vessels. To better understand the sole effect of time-varying blood conductivity on the spatiotemporal distribution of trans-thoracic electric fields (i.e. ICG), this paper presents a segmented high-resolution (1 mm3) thoracic cardiovascular system, in which the time-varying pressures, flows and electrical conductivities of blood in different vessels are evaluated using a set of coupled nonlinear differential equations, red blood cell orientation and cardiac cycle functions. Electric field and voltage simulations are performed using two and four electrode configurations delivering a small alternating electric current to an anatomically realistic and electrically accurate model of modelled human torso. The simulations provide a three-dimensional electric field distribution and show that the time-varying blood conductivity alters the voltage potential difference between the electrodes by a maximum of 0.28% for a cardiac output of about 5 L min−1. As part of a larger study, it is hoped that this initial model will be useful in providing improved insights into blood-flow-related spatiotemporal electric field variations and assist in the optimal placement of electrodes in impedance cardiography experiments
Optically stimulated differential impedance spectroscopy
Maxey, Lonnie C; Parks, II, James E; Lewis, Sr., Samuel A; Partridge, Jr., William P
2014-02-18
Methods and apparatuses for evaluating a material are described. Embodiments typically involve use of an impedance measurement sensor to measure the impedance of a sample of the material under at least two different states of illumination. The states of illumination may include (a) substantially no optical stimulation, (b) substantial optical stimulation, (c) optical stimulation at a first wavelength of light, (d) optical stimulation at a second wavelength of light, (e) a first level of light intensity, and (f) a second level of light intensity. Typically a difference in impedance between the impedance of the sample at the two states of illumination is measured to determine a characteristic of the material.
DEFF Research Database (Denmark)
Goh, Ailian; Gao, Feng; Loh, Pon Chiang;
2007-01-01
control, and push up the overall system costs. Therefore, alternative topological solutions are of interest, and should preferably be implemented using only passive LC elements and diodes, connected as unique impedance networks. A number of possible network configurations are now investigated...... in this paper, and are respectively named as Z-source, H-source, EZ-source and their respective "inverted" variants. The presented impedance networks can either be used with a traditional voltage-source or current-source inverter, and can either be powered by a voltage or current source. All impedance networks...... the practicalities and performances of the described impedance networks....
Richardson, John G.
2009-11-17
An impedance estimation method includes measuring three or more impedances of an object having a periphery using three or more probes coupled to the periphery. The three or more impedance measurements are made at a first frequency. Three or more additional impedance measurements of the object are made using the three or more probes. The three or more additional impedance measurements are made at a second frequency different from the first frequency. An impedance of the object at a point within the periphery is estimated based on the impedance measurements and the additional impedance measurements.
Mechanical Impedance of Cerebral Material
Directory of Open Access Journals (Sweden)
Hédi Ben Ghozlen
2012-03-01
Full Text Available The tentative variation of the mechanical impedance, of a cylindrical sample of cerebral material, has been achieved by Vibrometer Laser according to the frequency. The studied matter is supposed homogeneous, isotropic and stationary. A multilayered mechanical model has been associated to the studied sample to simulate its vibration. The theoretical expression of mechanical impedance has been determined while taking the mechanical/electric analogy as a basis. A good adjustment of theoretical model parameters permitted us to have a good agreement theory/experience of the mechanical impedance variation according to the sample vibration frequency.
IMPEDANCE CHARACTERISTICS OF POLYFURAN FILMS
Institute of Scientific and Technical Information of China (English)
Liang Li; Xiao-bo Wan; Gi Xue
2002-01-01
Electrochemical impedance spectroscopy (EIS) was first used for the characterization of polyfuran (PFu) films that had been formed electrochemically on an Au electrode. The polyfuran was measured in high oxidation state, intermediate oxidation state and reduction state, respectively. As the oxidation level is increased, the ionic conductivity of PFu/BF4-increases. And impedance studies on PFu show that the anion BF4- appears to be mobile with a high diffusion coefficient of approximately 10-8 cm2 @ s-1.
[Monitoring cervical dilatation by impedance].
Salvat, J; Lassen, M; Sauze, C; Baud, S; Salvat, F
1992-01-01
Several different physics procedures have been tried to mechanize the recording of partograms. Can a measure of impedance of tissue Z using potential difference V, according to Ohm's law V = Z1, and 1 is a constant, be correlated with a measure of cervical dilatation using vaginal examination? This was our hypothesis. The tissue impedance meter was made to our design and applied according to a bipolar procedure. Our work was carried out on 28 patients. 10 patients were registered before labour started in order to test the apparatus and to record the impedance variations without labour taking place, and 18 patients were registered in labour to see whether there was any correlation. The level of impedance in the cervix without labour was 302.7 Ohms with a deviation of 8.2. Using student's t tests it was found that there was a significant correlation (p less than 0.001) in four measurements between the impedance measure and measures obtained by extrapolating the degrees of dilatation calculated from vaginal examination. This is a preliminary study in which we have defined the conditions that are necessary to confirm these first results and to further develop the method. PMID:1401774
Noncontact scanning electrical impedance imaging.
Liu, Hongze; Hawkins, Aaron; Schultz, Stephen; Oliphant, Travis E
2004-01-01
We are interested in applying electrical impedance imaging to a single cell because it has potential to reveal both cell anatomy and cell function. Unfortunately, classic impedance imaging techniques are not applicable to this small scale measurement due to their low resolution. In this paper, a different method of impedance imaging is developed based on a noncontact scanning system. In this system, the imaging sample is immersed in an aqueous solution allowing for the use of various probe designs. Among those designs, we discuss a novel shield-probe design that has the advantage of better signal-to-noise ratio with higher resolution compared to other probes. Images showing the magnitude of current for each scanned point were obtained using this configuration. A low-frequency linear physical model helps to relate the current to the conductivity at each point. Line-scan data of high impedance contrast structures can be shown to be a good fit to this model. The first two-dimensional impedance image of biological tissues generated by this technique is shown with resolution on the order of 100 mum. The image reveals details not present in the optical image. PMID:17271930
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Blaabjerg, Frede; Galigekere, Veda Prakash;
2016-01-01
A novel A-source impedance network is proposed in this letter. The A-source impedance network uses an autotransformer for realizing converters for any application that demand a very high dc voltage gain. The network utilizes a minimal turns ratio compared to other Magnetically Coupled Impedance...... Source (MCIS) networks to attain a high voltage gain. In addition, the proposed converter draws a continuous current from the source, and hence it is suitable for many types of renewable energy sources. The derived network expressions and theoretical analysis are finally validated experimentally...... with an example single-switch 400 W dc-dc converter. For the closed-loop control design and stability assessment, a small signal model and its analysis of the proposed network are also presented in brief....
Impedances of Laminated Vacuum Chambers
Energy Technology Data Exchange (ETDEWEB)
Burov, A.; Lebedev, V.; /Fermilab
2011-06-22
First publications on impedance of laminated vacuum chambers are related to early 70s: those are of S. C. Snowdon [1] and of A. G. Ruggiero [2]; fifteen years later, a revision paper of R. Gluckstern appeared [3]. All the publications were presented as Fermilab preprints, and there is no surprise in that: the Fermilab Booster has its laminated magnets open to the beam. Being in a reasonable mutual agreement, these publications were all devoted to the longitudinal impedance of round vacuum chambers. The transverse impedance and the flat geometry case were addressed in more recent paper of K. Y. Ng [4]. The latest calculations of A. Macridin et al. [5] revealed some disagreement with Ref. [4]; this fact stimulated us to get our own results on that matter. Longitudinal and transverse impendances are derived for round and flat laminated vacuum chambers. Results of this paper agree with Ref. [5].
Hybrid-Source Impedance Networks
DEFF Research Database (Denmark)
Li, Ding; Gao, Feng; Loh, Poh Chiang;
2010-01-01
Hybrid-source impedance networks have attracted attention among researchers because of their flexibility in performing buck-boost energy conversion. To date, three distinct types of impedance networks can be summarized for implementing voltage-type inverters with another three types summarized...... for current-type inverters. These impedance networks can in principle be combined into two generic network entities, before multiple of them can further be connected together by applying any of the two proposed generalized cascading concepts. The resulting two-level and three-level inverters implemented using...... the cascaded networks would have a higher output voltage gain and other unique advantages that currently have not been investigated yet. It is anticipated that these advantages would help the formed inverters find applications in photovoltaic and other renewable systems, where a high voltage gain is usually...
Short-circuit impedance measurement
DEFF Research Database (Denmark)
Pedersen, Knud Ole Helgesen; Nielsen, Arne Hejde; Poulsen, Niels Kjølstad
2003-01-01
Methods for estimating the short-circuit impedance in the power grid are investigated for various voltage levels and situations. The short-circuit impedance is measured, preferably from naturally occurring load changes in the grid, and it is shown that such a measurement system faces different...... kinds of problems at different locations in the grid. This means that the best measurement methodology changes depending on the location in the grid. Three typical examples with different measurement problems at 400 kV, 132 kV and 400 V voltage level are discussed....
Input impedance characteristics of microstrip structures
Directory of Open Access Journals (Sweden)
A. I. Nazarko
2015-06-01
Full Text Available Introduction. Electromagnetic crystals (EC and EC-inhomogeneities are one of the main directions of microstrip devices development. In the article the input impedance characteristics of EC- and traditional microstrip inhomogeneities and filter based on EC-inhomogeneities are investigated. Transmission coefficient characteristics. Transmission coefficient characteristics of low impedance EC- and traditional inhomogeneities are considered. Characteristics are calculated in the software package Microwave Studio. It is shown that the efficiency of EC-inhomogeneity is much higher. Input impedance characteristics of low impedance inhomogeneities. Dependences of input impedance active and reactive parts of EC- and traditional inhomogeneities are given. Dependences of the active part illustrate significant low impedance transformation of nominal impedance. The conditions of impedance matching of structure and input medium are set. Input impedance characteristics of high impedance inhomogeneities. Input impedance characteristics of high impedance EC- and traditional inhomogeneities are considered. It was shown that the band of transformation by high impedance inhomogeneities is much narrower than one by low impedance inhomogeneities. Characteristics of the reflection coefficient of inhomogeneities are presented. Input impedance characteristics of narrowband filter. The structure of narrowband filter based on the scheme of Fabry-Perot resonator is presented. The structure of the filter is fulfilled by high impedance EC-inhomogeneities as a reflectors. Experimental and theoretical amplitude-frequency characteristics of the filter are presented. Input impedance characteristics of the filter are shown. Conclusions. Input impedance characteristics of the structure allow to analyse its wave properties, especially resonant. EC-inhomogeneity compared with traditional microstrip provide substantially more significant transformation of the the input impedance.
The Aberdeen Impedance Imaging System.
Kulkarni, V; Hutchison, J M; Mallard, J R
1989-01-01
The Aberdeen Impedance Imaging System is designed to reconstruct 2 dimensional images of the average distribution of the amplitude and phase of the complex impedance within a 3 dimensional region. The system uses the four electrode technique in a 16 electrode split-array. The system hardware consists of task-orientated electronic modules for: driving a constant current, multiplexing the current drive, demultiplexing peripheral voltages, differential amplification, phase sensitive detection and low-pass filtration, digitisation with a 14 bit analog to digital converter (ADC), and -control logic for the ADC and multiplexors. A BBC microprocessor (Master series), initiates a controlled sequence for the collection of a number of data sets which are averaged and stored on disk. Image reconstruction is by a process of convolution-backprojection similar to the fan-beam reconstruction of computerised tomography and is also known as Equipotential Backprojection. In imaging impedance changes associated with fracture healing the changes may be large enough to allow retrieval of both the amplitude and phase of the complex impedance. Sequential imaging of these changes would necessitate monitoring electronic and electrode drift by imaging an equivalent region of the contralateral limb. Differential images could be retrieved when the image of the normal limb is the image template. Better characterisation of tissues would necessitate a cleaner retrieval of the quadrature signal. PMID:2742979
DEFF Research Database (Denmark)
Siwakoti, Yam Prasad; Loh, Poh Chiang; Blaabjerg, Frede;
2014-01-01
This letter introduces a new versatile Y-shaped impedance network for realizing converters that demand a very high-voltage gain, while using a small duty ratio. To achieve that, the proposed network uses a tightly coupled transformer with three windings, whose obtained gain is presently not match...
Small Signal Loudspeaker Impedance Emulator
DEFF Research Database (Denmark)
Iversen, Niels Elkjær; Knott, Arnold
2014-01-01
from driver to driver. Therefore, a loudspeaker emulator capable of adjusting its impedance to that of a given driver is desired for measurement purposes. This paper proposes a loudspeaker emulator circuit for small signals. Simulations and experimental results are compared and show that it is possible...
Reducing the SPS Machine Impedance
Collier, Paul; Guinand, R; Jiménez, J M; Rizzo, A; Spinks, Alan; Weiss, K
2002-01-01
The SPS as LHC Injector project has been working for some time to prepare the SPS for its role as final injector for the LHC. This included major work related to injection, acceleration, extraction and beam instrumentation for the LHC beams [1]. Measurements carried out with the high brightness LHC beam showed that a major improvement of the machine impedance would also be necessary [2]. In addition to removing all lepton related components (once LEP operation ended in 2000), the decision was made to shield the vacuum system pumping port cavities. These accidental cavities had been identified as having characteristic frequencies in the 1-1.5GHz range. Since the SPS vacuum system contains roughly 1000 of these cavities, they constitute a major fraction of the machine impedance. As removal of the ports and associated bellows is not possible, transition shields (PPS) had to be designed to insert within the pumping port cavities.
Impedance based automatic electrode positioning.
Miklody, Daniel; Hohne, Johannes
2015-08-01
The position of electrodes in electrical imaging and stimulation of the human brain is an important variable with vast influences on the precision in modeling approaches. Nevertheless, the exact position is obscured by many factors. 3-D Digitization devices can measure the distribution over the scalp surface but remain uncomfortable in application and often imprecise. We demonstrate a new approach that uses solely the impedance information between the electrodes to determine the geometric position. The algorithm involves multidimensional scaling to create a 3 dimensional space based on these impedances. The success is demonstrated in a simulation study. An average electrode position error of 1.67cm over all 6 subjects could be achieved. PMID:26736345
Tapping mode microwave impedance microscopy
Lai, K.
2009-01-01
We report tapping mode microwave impedance imaging based on atomic force microscope platforms. The shielded cantilever probe is critical to localize the tip-sample interaction near the tip apex. The modulated tip-sample impedance can be accurately simulated by the finite-element analysis and the result agrees quantitatively to the experimental data on a series of thin-film dielectric samples. The tapping mode microwave imaging is also superior to the contact mode in that the thermal drift in a long time scale is totally eliminated and an absolute measurement on the dielectric properties is possible. We demonstrated tapping images on working nanodevices, and the data are consistent with the transport results. © 2009 American Institute of Physics.
A compact broadband nonsynchronous noncommensurate impedance transformer
DEFF Research Database (Denmark)
Zhurbenko, Vitaliy; Kim, Kseniya; Narenda, Kumar
2012-01-01
Nonsynchronous noncommensurate impedance transformers consist of a combination of high‐ and low‐impedance transmission lines. High‐impedance lines have narrow tracks in strip and microstrip technology, which allows for high flexibility and miniaturization of the layout in comparison to the...... this article online at wileyonlinelibrary.com. DOI 10.1002/mop.26975...
Impedance analysis of acupuncture points and pathways
International Nuclear Information System (INIS)
Investigation of impedance characteristics of acupuncture points from acoustic to radio frequency range is addressed. Discernment and localization of acupuncture points in initial single subject study was unsuccessfully attempted by impedance map technique. Vector impedance analyses determined possible resonant zones in MHz region.
Anderson, Karl F. (Inventor)
1994-01-01
A constant current loop measuring system is provided for measuring a characteristic of an environment. The system comprises a first impedance positionable in the environment, a second impedance coupled in series with said first impedance and a parasitic impedance electrically coupled to the first and second impedances. A current generating device, electrically coupled in series with the first and second impedances, provides a constant current through the first and second impedances to produce first and second voltages across the first and second impedances, respectively, and a parasitic voltage across the parasitic impedance. A high impedance voltage measuring device measures a voltage difference between the first and second voltages independent of the parasitic voltage to produce a characteristic voltage representative of the characteristic of the environment.
Impedance spectroscopy of food mycotoxins
Bilyy, Oleksandr I.; Yaremyk, Roman Ya.; Kotsyumbas, Ihor Ya.; Kotsyumbas, Halyna I.
2012-01-01
A new analytical method of high-selective detection of mycotoxins in food and feed are considered. A method is based on optical registration the changes of conduct of the electric polarized bacterial agents in solution at the action of the external gradient electric fields. Measuring are conducted in integrated electrode-optical cuvette of the special construction, which provides the photometric analysis of forward motion of the objects registration in liquid solution under act of the enclosed electric field and simultaneous registration of kinetics of change of electrical impedance parameters solution and electrode system.
Pumping slots: impedances and power losses
Energy Technology Data Exchange (ETDEWEB)
Kurennoy, S. [Maryland Univ., College Park, MD (United States). Dept. of Physics
1996-08-01
Contributions of pumping slots to the beam coupling impedances and power losses in a B-factory ring are considered. While their leading contribution is to the inductive impedance, for high-intensity machines with short bunches like e{sup +}e{sup -} B-factories the real part of the impedance and related loss factors are also important. Using an analytical approach we calculate the coupling impedances and loss factors due to slots in a ring with an arbitrary cross section of the vacuum chamber. Effects of the slot tilt on the beam impedance are also considered, and restrictions on the tilt angle are derived from limitations on the impedance increase. The power leakage through the slots is discussed briefly. The results are applied to the KEK B-factory. (author)
Impedances for electron linacs and storage rings
International Nuclear Information System (INIS)
Some basic concepts and results are presented concerning the impedances of electron linacs and storage rings. The impedances of an accelerator or ring completely characterizes the interaction of the beam with its environment. Not only does the impedance (or its Fourier transform, the wake potential) determine the energy loss by a bunched beam to its environment, but it is also the chief ingredient required for any calculation of beam stability
High frequency impedances in European XFEL
Energy Technology Data Exchange (ETDEWEB)
Dohlus, Martin; Zagorodnov, Igor; Zagorodnova, Olga
2010-06-15
The method of the optical approximation is used to estimate the high frequency impedances of different vacuum chamber transitions of the European XFEL beam line. The approximations of the longitudinal impedances are obtained in terms of simple one-dimensional integrals. The transverse impedances are written in analytical closed form. The analytical results are compared with the results obtained by numerical solution of Maxwell's equations. (orig.)
Impedance-controlled ultrasound probe
Gilbertson, Matthew W.; Anthony, Brian W.
2011-03-01
An actuated hand-held impedance-controlled ultrasound probe has been developed. The controller maintains a prescribed contact state (force and velocity) between the probe and a patient's body. The device will enhance the diagnostic capability of free-hand elastography and swept-force compound imaging, and also make it easier for a technician to acquire repeatable (i.e. directly comparable) images over time. The mechanical system consists of an ultrasound probe, ball-screw-driven linear actuator, and a force/torque sensor. The feedback controller commands the motor to rotate the ball-screw to translate the ultrasound probe in order to maintain a desired contact force. It was found that users of the device, with the control system engaged, maintain a constant contact force with 15 times less variation than without the controller engaged. The system was used to determine the elastic properties of soft tissue.
Lorentz Force Electrical Impedance Tomography
Grasland-Mongrain, Pol; Chapelon, Jean-Yves; Lafon, Cyril
2014-01-01
This article describes a method called Lorentz Force Electrical Impedance Tomography. The electrical conductivity of biological tissues can be measured through their sonication in a magnetic field: the vibration of the tissues inside the field induces an electrical current by Lorentz force. This current, detected by electrodes placed around the sample, is proportional to the ultrasonic pressure, to the strength of the magnetic field and to the electrical conductivity gradient along the acoustic axis. By focusing at different places inside the sample, a map of the electrical conductivity gradient can be established. In this study experiments were conducted on a gelatin phantom and on a beef sample, successively placed in a 300 mT magnetic field and sonicated with an ultrasonic transducer focused at 21 cm emitting 500 kHz bursts. Although all interfaces are not visible, in this exploratory study a good correlation is observed between the electrical conductivity image and the ultrasonic image. This method offers...
Transverse Impedance of Ferrite Elements
Burov, A
2004-01-01
A specific feature of ferrites is that these materials behave either as metals or magneto-dielectrics, depending on the frequency range. Their magnetic permeability is a function of frequency as well. In this paper, the transverse impedance of a ferrite kicker is calculated. The method suggested in Ref. [1] is generalized here for ferrites. Namely, in [1] it is assumed that the electric field of the beam charge dipole is always perfectly shielded. In fact, this assumption requires the conductivity being high compared with the frequency. This is not necessarily true for the high frequencies of a proton single-bunch spectrum. That is why the dynamics of the electric shielding has to be taken into account for ferrite kickers. The generalized analytic result is applied for the ferrite MKE kickers at the CERN SPS, and a fairly good agreement with observations [2] is found.
Possibilities of electrical impedance tomography in gynecology
V, Trokhanova O.; A, Chijova Y.; B, Okhapkin M.; V, Korjenevsky A.; S, Tuykin T.
2013-04-01
The paper describes results of comprehensive EIT diagnostics of mammary glands and cervix. The data were obtained from examinations of 170 patients by EIT system MEM (multi-frequency electrical impedance mammograph) and EIT system GIT (gynecological impedance tomograph). Mutual dependence is discussed.
Esophageal Impedance Monitoring: Clinical Pearls and Pitfalls.
Ravi, Karthik; Katzka, David A
2016-09-01
The development of intraluminal esophageal impedance monitoring has improved our ability to detect and measure gastroesophageal reflux without dependence on acid content. This ability to detect previously unrecognized weak or nonacid reflux episodes has had important clinical implications in the diagnosis and management of gastroesophageal reflux disease (GERD). In addition, with the ability to assess bolus transit within the esophageal lumen, impedance monitoring has enhanced the recognition and characterization of esophageal motility disorders in patients with nonobstructive dysphagia. The assessment of the intraluminal movement of gas and liquid has also been proven to be of diagnostic value in conditions such as rumination syndrome and excessive belching. Further, alternative applications of impedance monitoring, such as the measurement of mucosal impedance, have provided novel insights into assessing esophageal mucosal integrity changes as a consequence of inflammatory change. Future applications for esophageal impedance monitoring also hold promise in esophageal conditions other than GERD. However, despite all of the clinical benefits afforded by esophageal impedance monitoring, important clinical and technical shortcomings limit its diagnostic value and must be considered when interpreting study results. Overinterpretation of studies or application of impedance monitoring in patients can have deleterious clinical implications. This review will highlight the clinical benefits and limitations of esophageal impedance monitoring and provide clinical pearls and pitfalls associated with this technology. PMID:27325223
Impedance of a slotted-pipe kicker
Energy Technology Data Exchange (ETDEWEB)
Feng Zhou [Academia Sinica, Beijing, BJ (China). Inst. of High Energy Physics
1996-08-01
This paper introduces the principle of a new slotted kicker simply, which is made by using vacuum pipe itself with proper slits as current conductors, and then, presents a rough estimation of its longitudinal and transverse impedance, respectively. Calculation shows that its impedance is reduced significantly compared to our present air-coil kicker. (author)
Impedance of Surface Footings on Layered Ground
DEFF Research Database (Denmark)
Andersen, Lars; Clausen, Johan Christian
2005-01-01
discussed. Based on the Green's function for a stratified half-space, the impedance of a surface footing with arbitrary shape is computed. A wind turbine foundation is analysed in the frequency range 0 to 3 Hz. Analyses show that soil stratification may lead to a significant changes in the impedance related...
Impedance of Surface Footings on Layered Ground
DEFF Research Database (Denmark)
Andersen, Lars; Clausen, Johan
2007-01-01
discussed. Based on the Green's function for a stratified half-space, the impedance of a surface footing with arbitrary shape is computed. A wind turbine foundation is analysed in the frequency range 0-3 Hz. Analyses show that soil stratification may lead to significant changes in the impedance related to...
New OTRA-Based Generalized Impedance Simulator
Ashish Gupta; Raj Senani; D. R. Bhaskar; SINGH, A.K.
2013-01-01
Operational transresistance amplifier (OTRA) has attracted considerable attention in the recent literature in several applications such as impedance simulation, universal biquad filter realization, realization of sinusoidal oscillators and multivibrators. However, to the best knowledge of the authors, any OTRA-based generalized impedance simulator circuits have not been reported so far. The purpose of this paper is to present such a circuit.
Analytic methods for calculating coupling impedances
International Nuclear Information System (INIS)
These lecture notes describe a variety of analytic techniques to calculate the longitudinal and transverse impedances of obstacles in a beam pipe. They also treat the effort to shield these impedances from the beam by appropriate use of thin conducting layers. (orig.)
Estimating the short-circuit impedance
DEFF Research Database (Denmark)
Nielsen, Arne Hejde; Pedersen, Knud Ole Helgesen; Poulsen, Niels Kjølstad
through a RLS-algorithm. The algorithms have been tested and implemented on a PC at a 132 kV substation supplying a rolling mill. Knowing the short-circuit impedance gives the rolling mill an opportunity to adjust the arc furnace operation to keep flicker below a certain level. Therefore, the PC performs...... a simultaneously measurement of impedance and flicker....
Far-infrared embedding impedance measurements
Neikirk, D. P.; Rutledge, D. B.
1984-01-01
A technique which allows the measurement of detector embedding impedance has been developed. By using a bismuth microbolometer as a variable resistance load the impedance of one element in a bow-tie antenna array operating at 94 GHz was inferred. The technique is frequency insensitive, and could be used throughout the far-infrared.
Identification of irradiated potatoes by impedance measurements
International Nuclear Information System (INIS)
Measuring the impedance was found to be a highly reliable and practical technique for identifying irradiated potatoes. Impedance was measured by puncturing a potato tuber with a steel electrode and passing a 3 -- 5 mA alternating current through it. Three parameters were determined: Z0/Z180 (impedance ratio at 5 kHz, 0 to 180 seconds after puncturing), Z sub(50k)/Z sub(0.5k) (impedance ratio at 50 kHz to 0.5 kHz) and Z sub(50k)/Z sub(5k) (impedance ratio at 50 kHz to 5 kHz). Among these, parameter Z sub(50k)/Z sub(5k) was the most favourable index. The technique allowed not only differentiation between unirradiated and irradiated potatoes but an estimation of the irradiation dose for up to six months after irradiation, independent of the potato storage condition. (author)
Fractional Order Element Based Impedance Matching
Radwan, Ahmed Gomaa
2014-06-24
Disclosed are various embodiments of methods and systems related to fractional order element based impedance matching. In one embodiment, a method includes aligning a traditional Smith chart (|.alpha.|=1) with a fractional order Smith chart (|.alpha.|.noteq.1). A load impedance is located on the traditional Smith chart and projected onto the fractional order Smith chart. A fractional order matching element is determined by transitioning along a matching circle of the fractional order Smith chart based at least in part upon characteristic line impedance. In another embodiment, a system includes a fractional order impedance matching application executed in a computing device. The fractional order impedance matching application includes logic that obtains a first set of Smith chart coordinates at a first order, determines a second set of Smith chart coordinates at a second order, and determines a fractional order matching element from the second set of Smith chart coordinates.
Rotor damage detection by using piezoelectric impedance
Qin, Y.; Tao, Y.; Mao, Y. F.
2016-04-01
Rotor is a core component of rotary machinery. Once the rotor has the damage, it may lead to a major accident. Thus the quantitative rotor damage detection method based on piezoelectric impedance is studied in this paper. With the governing equation of piezoelectric transducer (PZT) in a cylindrical coordinate, the displacement along the radius direction is derived. The charge of PZT is calculated by the electric displacement. Then, by the use of the obtained displacement and charge, an analytic piezoelectric impedance model of the rotor is built. Given the circular boundary condition of a rotor, annular elements are used as the analyzed objects and spectral element method is used to set up the damage detection model. The Electro-Mechanical (E/M) coupled impedance expression of an undamaged rotor is deduced with the application of a low-cost impedance test circuit. A Taylor expansion method is used to obtain the approximate E/M coupled impedance expression for the damaged rotor. After obtaining the difference between the undamaged and damaged rotor impedance, a rotor damage detection method is proposed. This method can directly calculate the change of bending stiffness of the structural elements, it follows that the rotor damage can be effectively detected. Finally, a preset damage configuration is used for the numerical simulation. The result shows that the quantitative damage detection algorithm based on spectral element method and piezoelectric impedance proposed in this paper can identify the location and the severity of the damaged rotor accurately.
Measurements of electrical impedance of biomedical objects.
Frączek, Marcin; Kręcicki, Tomasz; Moron, Zbigniew; Krzywaźnia, Adam; Ociepka, Janusz; Rucki, Zbigniew; Szczepanik, Zdzisław
2016-01-01
Some basic problems related to measurements of electrical impedance of biological objects (bioimpedance) have been presented in this paper. Particularly problems arising from impedance occurring at the sensor-tissue interface (interfacial impedances) in contact measuring methods have been discussed. The influence of finite values of impedances of the current source and voltage measuring device has also been taken into consideration. A model of the impedance sensor for the four-electrode measurement method containing the interfacial, source and measuring device impedances has been given and its frequency characteristics obtained by the computer simulation have been presented. The influence of these impedances on the shape of frequency characteristic of the sensor model has been discussed. Measurements of bioimpedance of healthy and anomalous soft tissues have been described. Some experimental results, particularly the frequency characteristics of bioimpedance, have been shown. The presented results of measurement show that bioimpedance can be a valuable source of information about the tissues, so measurement of bioimpedance can be a useful supplement to other medical diagnostic methods. PMID:27151250
Tracking of electrochemical impedance of batteries
Piret, H.; Granjon, P.; Guillet, N.; Cattin, V.
2016-04-01
This paper presents an evolutionary battery impedance estimation method, which can be easily embedded in vehicles or nomad devices. The proposed method not only allows an accurate frequency impedance estimation, but also a tracking of its temporal evolution contrary to classical electrochemical impedance spectroscopy methods. Taking into account constraints of cost and complexity, we propose to use the existing electronics of current control to perform a frequency evolutionary estimation of the electrochemical impedance. The developed method uses a simple wideband input signal, and relies on a recursive local average of Fourier transforms. The averaging is controlled by a single parameter, managing a trade-off between tracking and estimation performance. This normalized parameter allows to correctly adapt the behavior of the proposed estimator to the variations of the impedance. The advantage of the proposed method is twofold: the method is easy to embed into a simple electronic circuit, and the battery impedance estimator is evolutionary. The ability of the method to monitor the impedance over time is demonstrated on a simulator, and on a real Lithium ion battery, on which a repeatability study is carried out. The experiments reveal good tracking results, and estimation performance as accurate as the usual laboratory approaches.
Summary of the impedance working group
International Nuclear Information System (INIS)
The impedance working group concentrated on the LHC design during the workshop. They look at the impedance contributions of liner, beam position monitors, shielded bellows, experimental chambers, superconducting cavities, recombination chambers, space charge, kickers, and the resistive wall. The group concluded that the impedance budgeting and the conceptual designs of the vacuum chamber components looked basically sound. It also noted, not surprisingly, that a large amount of studies are to be carried out further, and it ventured to give a partial list of these studies
SPATIAL VARIABILITY OF PEDOZEMS MECHANICAL IMPEDANCE
Directory of Open Access Journals (Sweden)
Zhukov A.V.
2013-04-01
Full Text Available We studied the spatial variability of pedozem mechanical impedance in ResearchRemediation Center of the Dnipropetrovsk State Agrarian University in Ordzhonikidze. Thestatistical distribution of the soil mechanical impedance within the studied area is characterized by deviation from the normal law in 0–10 and 30–50 cm layers from the surface. 2D and 3D modeling shows the structural design of the soil as locations of high mechanical impedance which found in the soils with less hardness.
The electrochemical impedance of metal hydride electrodes
DEFF Research Database (Denmark)
Valøen, Lars Ole; Lasia, Andrzej; Jensen, Jens Oluf; Tunold, Reidar
2002-01-01
The electrochemical impedance responses for different laboratory type metal hydride electrodes were successfully modeled and fitted to experimental data for AB5 type hydrogen storage alloys as well as one MgNi type electrode. The models fitted the experimental data remarkably well. Several AC......, explaining the experimental impedances in a wide frequency range for electrodes of hydride forming materials mixed with copper powder, were obtained. Both charge transfer and spherical diffusion of hydrogen in the particles are important sub processes that govern the total rate of the electrochemical...... observed. The impedance analysis was found to be an efficient method for characterizing metal hydride electrodes in situ....
Estimating the short-circuit impedance
DEFF Research Database (Denmark)
Nielsen, Arne Hejde; Pedersen, Knud Ole Helgesen; Poulsen, Niels Kjølstad
1997-01-01
current are derived each period, and the short-circuit impedance is estimated from variations in these components created by load changes in the grid. Due to the noisy and dynamic grid with high harmonic distortion it is necessary to threat the calculated values statistical. This is done recursively......A method for establishing a complex value of the short-circuit impedance from naturally occurring variations in voltage and current is discussed. It is the symmetrical three phase impedance at the fundamental grid frequency there is looked for. The positive sequence components in voltage and...... through a RLS-algorithm. The algorithms have been tested and implemented on a PC at a 132 kV substation supplying a rolling mill. Knowing the short-circuit impedance gives the rolling mill an opportunity to adjust the arc furnace operation to keep flicker below a certain level. Therefore, the PC performs...
Interpretation of faradaic impedance for Corrosion monitoring
Energy Technology Data Exchange (ETDEWEB)
Itagaki, M.; Taya, A.; Imamura, M.; Saruwatari, R.; Watanabe, K. [Science University of Tokyo, Chiba (Japan)
2004-02-15
A polarization resistance is generally used to estimate the corrosion rate in the corrosion monitoring by an electrochemical impedance method. When the Faradaic impedance has a time constant due to the reaction intermediate, the electrochemical impedance describes more than one loop on the complex plane. For example, the electrochemical impedance of iron in acidic solution shows capacitive and inductive loops on the complex plane. In this case, the charge transfer resistance and the polarization resistance are determined at middle and low frequency ranges, respectively. Which should be selected for corrosion resistance in corrosion monitoring, the charge transfer resistance or the polarization resistance? In the present paper, the above-mentioned question is examined theoretically and experimentally
Impedance feedback control for scanning electrochemical microscopy.
Alpuche-Aviles, M A; Wipf, D O
2001-10-15
A new constant-distance imaging method based on the relationship between tip impedance and tip-substrate separation has been developed for the scanning electrochemical microscope. The tip impedance is monitored by application of a high-frequency ac voltage bias between the tip and auxiliary electrode. The high-frequency ac current is easily separated from the dc-level faradaic electrochemistry with a simple RC filter, which allows impedance measurements during feedback or generation/collection experiments. By employing a piezo-based feedback controller, we are able to maintain the impedance at a constant value and, thus, maintain a constant tip-substrate separation. Application of the method to feedback and generation/collection experiments with tip electrodes as small as 2 microm is presented. PMID:11681463
Impedance and collective effects in the LHC
Energy Technology Data Exchange (ETDEWEB)
Gareyte, J. [European Organization for Nuclear Research, Geneva (Switzerland)
1996-08-01
After a review of the main LHC parameters, and a brief description of the RF and vacuum systems, the coupling impedances of the main machine elements are given, as well as the resulting thresholds for instabilities. (author)
Interpretation of faradaic impedance for Corrosion monitoring
International Nuclear Information System (INIS)
A polarization resistance is generally used to estimate the corrosion rate in the corrosion monitoring by an electrochemical impedance method. When the Faradaic impedance has a time constant due to the reaction intermediate, the electrochemical impedance describes more than one loop on the complex plane. For example, the electrochemical impedance of iron in acidic solution shows capacitive and inductive loops on the complex plane. In this case, the charge transfer resistance and the polarization resistance are determined at middle and low frequency ranges, respectively. Which should be selected for corrosion resistance in corrosion monitoring, the charge transfer resistance or the polarization resistance? In the present paper, the above-mentioned question is examined theoretically and experimentally
Effective impedance modeling of metamaterial structures.
Dossou, Kokou B; Poulton, Christopher G; Botten, Lindsay C
2016-03-01
We present methods for retrieving the effective impedance of metamaterials from the Fresnel reflection coefficients at the interface between two semi-infinite media. The derivation involves the projection of rigorous modal expansions onto the dominant modes of the two semi-infinite media. It is shown that the effective impedance can also be written as a ratio of averaged field quantities. Thus, a number of effective impedance formulas, previously obtained by field averaging techniques, can also be derived from the scattering-based formalism by an appropriate choice of projection. Within the effective medium limit, it is observed that a simple semianalytic modeling technique based on the effective impedance can be used to reliably compute the reflection coefficients of metamaterials over a wide range of incidence angles. We use this technique to model planar metamaterial waveguides or surface modes. PMID:26974905
Study of high impedance magnetic alloy core
International Nuclear Information System (INIS)
J-PARC 3 GeV Rapid Cycling Synchrotron (RCS) and Main Ring (MR) employ RF cavities loaded with Magnetic Alloy (MA) cores to generate a high field gradient. The MA core shunt impedance Rp is a key parameter to increase the beam power. To achieve the high shunt impedance Rp, we increased the filling factor by a strong winding tension. As a result, the shunt impedance Rp was reduced in despite of the filling factor increase because the strong winding tension deteriorates the MA core magnetic properties significantly. On the other hand, the MA core with thin ribbons and the MA core that is magnetized by the rotation process show the high permeability in the accelerating frequency region, and those MA cores show the high shunt impedance Rp consequently. (author)
Wave impedance of an atomically thin crystal.
Merano, Michele
2015-11-30
I propose an expression for the electromagnetic wave impedance of a two-dimensional atomic crystal, and I deduce the Fresnel coefficients in terms of this quantity. It is widely known that a two-dimensional crystal can absorb light, if its conductivity is different from zero. It is less emphasized that they can also store a certain amount of electromagnetic energy. The concept of impedance is useful to quantify this point. PMID:26698783
CSR Impedance for Non-Ultrarelativistic Beams
Energy Technology Data Exchange (ETDEWEB)
Li, Rui [Jefferson Lab., Newport News, VA (United States); Tsai, Cheng Y. [Virginia Polytechnic Inst. and State Univ. (Virginia Tech), Blacksburg, VA (United States); Jefferson Lab., Newport News, VA (United States)
2015-09-01
For the analysis of the coherent synchrotron radiation (CSR)-induced microbunching gain in the low energy regime, such as when a high-brightness electron beam is transported through a low-energy merger in an energy-recovery linac (ERL) design, it is necessary to extend the CSR impedance expression in the ultrarelativistic limit to the non-ultrarelativistic regime. This paper presents our analysis of CSR impedance for general beam energies.
Surface impedance and the Casimir force
Bezerra, V.B.; Klimchitskaya, G. L.; Romero, C.
2001-01-01
The impedance boundary condition is used to calculate the Casimir force in configurations of two parallel plates and a shpere (spherical lens) above a plate at both zero and nonzero temperature. The impedance approach allows one to find the Casimir force between the realistic test bodies regardless of the electromagnetic fluctuations inside the media. Although this approach is an approximate one, it has wider areas of application than the Lifshitz theory of the Casimir force. The general form...
Directory of Open Access Journals (Sweden)
Dejan Krizaj
2012-10-01
Full Text Available Electrical impedance measurements of the suspensions have to take into account the double layer impedance that is due to a very thin charged layer formed at the electrode-electrolite interface. A dedicated measuring cell that enables variation of the distance between the electrodes was developed for investigation of electrical properties of suspensions using two electrode impedance measurements. By varying the distance between the electrodes it is possible to separate the double layer and the suspension impedance from the measured data. From measured and extracted impedances electrical lumped models have been developed. The error of non inclusion of the double layer impedance has been analyzed. The error depends on the frequency of the measurements as well as on the distance between the electrodes.
Surface impedance and the Casimir force
International Nuclear Information System (INIS)
The impedance boundary condition is used to calculate the Casimir force in configurations of two parallel plates and a sphere (spherical lens) above a plate at both zero and nonzero temperature. The impedance approach allows one to find the Casimir force between the realistic test bodies regardless of the electromagnetic fluctuations inside the media. Although this approach is an approximate one, it has wider areas of application than the Lifshitz theory of the Casimir force. The general formulas of the impedance approach to the theory of the Casimir force are given and the formal substitution is found for connecting it with the Lifshitz formula. The range of micrometer separations between the test bodies, which is interesting from the experimental point of view, is investigated in detail. It is shown that at zero temperature the results obtained on the basis of the surface impedance method are in agreement with those obtained in framework of the Lifshitz theory within a fraction of a percent. The temperature correction to the Casimir force from the impedance method coincides with that from the Lifshitz theory up to four significant figures. The case of millimeter separations that corresponds to the normal skin effect is also considered. At zero temperature the obtained results have good agreement with the Lifshitz theory. At nonzero temperature the impedance approach is not subject to the interpretation problems peculiar to the zero-frequency term of the Lifshitz formula in dissipative media
Analysis of formulas used in coupling impedance coaxial-wire measurements for distributed impedances
International Nuclear Information System (INIS)
In this paper the authors study the validity of coupling impedance bench measurements for distributed impedances, comparing the commonly used log formula to the result obtained applying a modified version of Bethe's theory of diffraction to a long slot in a coaxial beam pipe. The equations found provide a quantitative expression for the influence of the wire thickness used in the measurement of the real and imaginary part of the longitudinal impedance. The precision achievable in an actual measurement is therefore discussed. The method presented has also been applied in the presence of lumped impedances
Electromagnetic Wave Scattering By the Coated Impedance Cylinder
Directory of Open Access Journals (Sweden)
V.I. Vyunnik
2010-01-01
Full Text Available In this work the boundary conditions for the impedance circular cylinder coated by a low contrast dielectric thin layer are derived. Expression for the reduced impedance of the cylinder is obtained. Conditions and applicability limits of the proposed approach are defined. Influence of the coating impedance on the reduced impedance of the cylinder is investigated.
Non-contact scanning electrical impedance imaging.
Liu, Hongze; Hawkins, Aaron; Schultz, Stephen; Oliphant, Travis
2004-01-01
We are interested in applying electrical impedance imaging to a single cell because it has potential to reveal both cell anatomy and cell function. Unfortunately, classic impedance imaging techniques are not applicable to this small scale measurement due to their low resolution. In this paper, a different method of impedance imaging is developed based on a non-contact scanning system. In this system, the imaging sample is immersed in an aqueous solution allowing for the use of various probe designs. Among those designs, we discuss a novel shield-probe design that has the advantage of better signal-to-noise ratio with higher resolution compared to other probes. Images showing the magnitude of current for each scanned point were obtained using this configuration. A low-frequency linear physical model helps to relate the current to the conductivity at each point. Line-scan data of high impedance contrast structures can be shown to be a good fit to this model. The first two-dimensional impedance image of biological tissues generated by this technique is shown with resolution on the order of 100 mum. The image reveals details not present in the optical image. PMID:17271931
Ferrofluid Microwave Devices With Magnetically Controlled Impedances
Fannin, P. C.; Stefu, N.; Marin, C. N.; Malaescu, I.; Totoreanu, R.
2010-08-01
Ferrofluid filled transmission lines are microwave electronic devices. The complex dielectric permittivity and the complex magnetic permeability of a kerosene based ferrofluid with magnetite nanoparticles, in the frequency range (0.5-6) GHz were measured, for several values of polarising field, H. Afterwards, the input impedance of a short-circuited transmission line filled with this ferrofluid was computed using the equation Z = Zc tanh(γl). Here Zc and l are the characteristic impedance and the length of the coaxial line and γ is the propagation constant, depending on the dielectric and magnetic parameters of the material within the line. It is demonstrated how the impedance displays a frequency and polarizing field dependence, which has application in the design of magnetically controlled microwave devices.
Respiratory acoustic impedance in left ventricular failure.
Depeursinge, F B; Feihl, F; Depeursinge, C; Perret, C H
1989-12-01
The measurement of respiratory acoustic impedance (Zrs) by forced pseudorandom noise provides a simple means of assessing respiratory mechanics in nonintubated intensive care patients. To characterize the lung mechanical alterations induced by acute vascular congestion of the lung, Zrs was measured in 14 spontaneously breathing patients hospitalized for acute left ventricular failure. The Zrs data in the cardiac patients were compared with those of 48 semirecumbent normal subjects and those of 23 sitting asthmatic patients during allergen-induced bronchospasm. In the patients with acute left ventricular failure, the Zrs abnormalities noted were an excessive frequency dependence of resistance from 10 to 20 Hz and an abnormally low reactance at all frequencies, abnormalities qualitatively similar to those observed in the asthmatic patients but of lesser magnitude. Acute lung vascular congestion modifies the acoustic impedance of the respiratory system. Reflex-induced bronchospasm might be the main mechanism altering respiratory acoustic impedance in acute left ventricular failure. PMID:2582846
Coupling slots without shunt impedance drop
International Nuclear Information System (INIS)
It is well known that coupling slots between adjacent cells in a π-mode structure reduce shunt impedance per unit length with respect to single cell cavities. To design optimized coupling slots, one has to answer the following question: for a given coupling factor, what shape, dimension, position and number of slots lead to the lowest shunt impedance drop? A numerical study using the 3D code MAFIA has been carried out. The aim was to design the 352 MHz cavities for the high intensity proton accelerator of the TRISPAL project. The result is an unexpected set of four 'petal' slots. Such slots should lead to a quasi-negligible drop in shunt impedance: about -1% on average, for particle velocity from 0.4 c to 0.8 c. (author)
Impedance plethysmographic observations in thoracic outlet syndrome.
Directory of Open Access Journals (Sweden)
Nerurkar S
1990-07-01
Full Text Available Forty patients with symptoms of neuro-vascular compression in the upper extremities were subjected to impedance plethysmographic study using Parulkar′s method. Two patients recorded decreased blood flow (BFI in supine position and were diagnosed as having partial occlusion at subclavian level. Sixteen of the patients recorded decreased BFI on 90 degrees abduction and hyper-abduction. Twelve of these patients had radiological evidence of anomalous cervicle ribs. In remaining four patients extrinsic impression on the subclavian artery due to fibrous deposits was confirmed by arteriography. Remaining 22 patients recorded normal impedance plethysmograms. Impedance plethysmography thus provided a non-invasive modality for confirmation of vascular compression in thoracic outlet syndrome.
Electrical impedance measurement of irradiated potatoes
International Nuclear Information System (INIS)
Several chemical, biochemical and histological methods have been suggested for the identification of irradiated potatoes but these methods are either time consuming or lack the reliability and precision to be of much practical use. Measurement of electrical conductivity or impedance appears to be a simple and reliable technique. We have examined the suitability of electrical impedance method for potatoes grown in our country after exposing to a sprout inhibiting dose of 0.1 kGy. The results of this study are described. 10 refs., 3 figs., 2 tabs
Wave impedance retrieving via Bloch modes analysis
DEFF Research Database (Denmark)
Andryieuski, Andrei; Ha, S.; Sukhorukov, A.; Malureanu, Radu; Kivshar, Y.; Lavrinenko, Andrei
-ciples violation, like antiresonance behaviour with Im(ε) <0, Im(μ) <0. We employ the Bloch mode analysis of periodic metamaterials to extract the dominating (fundamental) Bloch mode. Then it is possible to determine the Bloch and wave impedances by the surface and volume aver-aging of the electromagnetic field of...... the Bloch mode, respectively. Case studies prove that our ap-proach can determine material and wave effective parameters of lossy and lossless metamaterials. In some examples when the passivity is violated we made further analysis and showed that this is due to the failure of concept of impedance...
Directory of Open Access Journals (Sweden)
Humayra Ferdous
2013-12-01
Full Text Available Focused Impedance Measurement (FIM is a technique where impedance can be measured with the optimum level of localization without much increase in complexity of measuring instrument. The electrodes are applied on the skin surface but the organs inside also contributes to the measurement, as the body is a volume conductor. In a healthy and disease free lung region, the air enters at breathe-in increases the impedance of the lung and impedance reduces during breathe-out. In contrast, for a diseased lung, where part of the lungs is filled with water or some fluid, air will not enter into this zone reducing impedance change between inspiration and expiration. With this idea, the current work had been executed to have general view of localised impedance change throughout thorax using 6-electrode FIM. This generated a matrix mapping from both the front and from the back of the thorax, which afterwards provided that how impedance change due to ventilation varies from frontal plane to back plane of human bodies.
Impedance-matched drilling telemetry system
Normann, Randy A.; Mansure, Arthur J.
2008-04-22
A downhole telemetry system that uses inductance or capacitance as a mode through which signal is communicated across joints between assembled lengths of pipe wherein efficiency of signal propagation through a drill string, for example, over multiple successive pipe segments is enhanced through matching impedances associated with the various telemetry system components.
Identification of irradiated potatoes by impedance measurements
International Nuclear Information System (INIS)
A variety of parameters drawn from impedance measurements are tested to distinguish between irradiated and non-irradiated potatoes. Some of these parameters are able to identify the irradiated potatoes. The identification is still possible after a storage time of 3 months. (author)
Identification of irradiated potatoes by impedance measurements
International Nuclear Information System (INIS)
Identification of irradiated potatoes (Alpha variety) by electrical impedance measurements has been carried out. Experiments were performed by passing ∼3m A alternating current through the potato tubers that were punctured with the galvanized metallic electrodes. The parameters Z0/Z180 (impedance ratio at 50 Hz zero to 180 seconds post puncturing) Z50 k/Z5 k, Z0.5 k/ Z50 k/Z0.05 k (impedance ratio at 50 khz, and 0.05 khz, respectively) were determined at various temperatures and the best temperature for the measurement was obtained. The selection of the identification parameter was based on its constancy over the post irrigation storage time (six months), as well as, its dependency on the magnitude of the absorbed dose.Based on the above criteria, the impedance ratio of Z50 k/Z5 k was determined to be the best identification parameter.The obtained empirical formulas allow to estimate the applied dose and also to differentiation between the irradiated and unirradiated potatoes at the temperature of the (20-30digc)
Detection of irradiated potatoes by impedance measurement
International Nuclear Information System (INIS)
Potato is one of the major food items to be treated with ionising radiation and potatoes are irradiated on a large scale in several countries. Every year around 15,000 t of potatoes are irradiated at doses of 60 to 150 Gy (average dose is about 100 Gy) in Japan. Although various methods to detect irradiated potatoes have been investigated, no established method has been reported. Measuring electrical conductivity or impedance of potatoes has been reported as a promising method for the detection of irradiated potatoes. In previous studies it has been found that the ratio of impedance magnitude at 50 kHz to that at 5 kHz, measured immediately after puncturing a potato tuber, is dependent upon the dose applied to the tuber, independent of storage temperature and stable during storage after irradiation. The aim of this study was to establish the optimum conditions for impedance measurement and to examine the applicability of the impedance measuring method to various cultivars (cv.) of potatoes. (author)
Variable high surge impedance loading line
Energy Technology Data Exchange (ETDEWEB)
Drummond, Marcio; Podporkin, Georgij; Ruffier, Aureo P.; Fernandes, Chester; Mesquita, Evanise M. de; Oliveira, Gloria Suzana G. de [Centro de Pesquisas de Energia Eletrica (CEPEL), Rio de Janeiro, RJ (Brazil)
1996-07-01
Taking advantage of the High Surge Impedance Loading Line technology and of it's physical characteristics, more specifically those related to the maximization of transmission capacity by increasing the number of sub-conductors, an innovative opportunity related to adaptive electrical parameters of the line was disclosed: the possibility of independently switching of phase sub-conductors to adapt the line impedance to transmission power. The aim of this procedure is to adapt the electrical parameters of the line in such way that Surge Impedance Loading is kept higher enough to make possible the transmitted power, imposing a system favorable operating condition. The assessment of this concept is presented in this paper including electrical studies for the definition of line configuration and steady state power system studies, considering the alternative of feeding power by a hypothetical 230 kV radial transmission line. The transmission distance was varied from 100 km to 300 km. technical and economical comparisons between this new transmission line technology and the conventional one are also presented. Transmission, Transmission Line, High Surge Impedance Loading Line, FACTs. (author)
Electrical Impedance Tomography Technology (EITT) Project
Oliva-Buisson, Yvette J.
2014-01-01
Development of a portable, lightweight device providing two-dimensional tomographic imaging of the human body using impedance mapping. This technology can be developed to evaluate health risks and provide appropriate medical care on the ISS, during space travel and on the ground.
Explicit Expressions of Impedances and Wake Functions
Energy Technology Data Exchange (ETDEWEB)
Ng, K.Y.; /Fermilab; Bane, K,; /SLAC
2012-06-11
Sections 3.2.4 and 3.2.5 of the Handbook of Accelerator Physics and Engineering on Landau damping are combined and updated. The new addition includes impedances and wakes for multi-layer beam pipe, optical model, diffraction model, and cross-sectional transition.
Explicit expressions of impedances and wake functions
Energy Technology Data Exchange (ETDEWEB)
Ng, K.Y.; /Fermilab; Bane, K,; /SLAC
2010-10-01
Sections 3.2.4 and 3.2.5 of the Handbook of Accelerator Physics and Engineering on Landau damping are combined and updated. The new addition includes impedances and wakes for multi-layer beam pipe, optical model, diffraction model, and cross-sectional transition.
Microwave transmission-line impedance data
Gunston, M A R
1996-01-01
A compendium of data for computing the characteristic impedance of transmission lines based on physical dimensions. Covers both conventional structures and unusual geometries, including coaxial, eccentric and elliptic coaxial, twin-wire, wire-above-ground, microstrip and derivatives, stripline, slabline and trough line. Also details numerous configurations of coupled lines.
Impedance and collective effects in the KEKB
Energy Technology Data Exchange (ETDEWEB)
Chin, Yongho [National Lab. for High Energy Physics, Tsukuba, Ibaraki (Japan); Oide, Katsunobu
1996-08-01
This paper focuses on beam instabilities due to single-beam collective effects, impedances from various beamline elements, ion trapping, photo-electrons, and other issues in the KEKB. We will also discuss the power deposition generated by a beam in the form of the Higher-Order-Mode (HOM) losses by interacting with its surroundings. (author)
Electrochemical Impedance Studies of SOFC Cathodes
DEFF Research Database (Denmark)
Hjelm, Johan; Søgaard, Martin; Wandel, Marie;
2007-01-01
Mixed ion- and electron-conducting composite electrodes consisting of doped ceria and perovskite have been studied by electrochemical impedance spectroscopy (EIS) at different temperatures and oxygen partial pressures. This paper aims to describe the different contributions to the polarisation im...
Electrical impedance spectroscopy and diagnosis of tendinitis
International Nuclear Information System (INIS)
There have been a number of studies that investigate the usefulness of bioelectric signals in diagnoses and treatment in the medical field. Tendinitis is a musculoskeletal disorder with a very high rate of occurrence. This study attempts to examine whether electrical impedance spectroscopy (EIS) can detect pathological changes in a tendon and find the exact location of the lesion. Experimental tendinitis was induced by injecting collagenase into one side of the patellar tendons in rabbits, while the other side was used as the control. After measuring the impedance in the tendinitis and intact tendon tissue, the dissipation factor was computed. The real component of impedance and the dissipation factor turned out to be lower in tendinitis than in intact tissues. Moreover, the tendinitis dissipation factor spectrum showed a clear difference from that of the intact tendon, indicating its usefulness as a tool for detecting the location of the lesion. Pathologic findings from the tissues that were obtained after measuring the impedance confirmed the presence of characteristics of tendinitis. In conclusion, EIS is a useful method for diagnosing tendinitis and detecting the lesion location in invasive treatment
International Nuclear Information System (INIS)
The purpose of this study is to compare measurement errors in two commercially available multi-frequency bioimpedance analyzers, a Xitron 4000B and an ImpediMed SFB7, including electrode impedance mismatch. The comparison was made using resistive electrical models and in ten human volunteers. We used three different electrical models simulating three different body segments: the right-side, leg and thorax. In the electrical models, we tested the effect of the capacitive coupling of the patient to ground and the skin–electrode impedance mismatch. Results showed that both sets of equipment are optimized for right-side measurements and for moderate skin–electrode impedance mismatch. In right-side measurements with mismatch electrode, 4000B is more accurate than SFB7. When an electrode impedance mismatch was simulated, errors increased in both bioimpedance analyzers and the effect of the mismatch in the voltage detection leads was greater than that in current injection leads. For segments with lower impedance as the leg and thorax, SFB7 is more accurate than 4000B and also shows less dependence on electrode mismatch. In both devices, impedance measurements were not significantly affected (p > 0.05) by the capacitive coupling to ground
Inter-Changeability of Impedance Devices for Lymphedema Assessment.
van Zanten, Malou; Piller, Neil; Ward, Leigh C
2016-06-01
Impedance technology is a popular technique for the early detection of lymphedema. The preferred approach is to use bioimpedance spectroscopy (BIS), with measurements being made with the subject lying supine, although attempts have been made to use single or multiple frequency impedance measurements obtained while the subject is standing. The aim of the present study was to determine the equivalence of these different approaches. Impedance measurements of the individual limbs of 37 healthy individuals were determined using both a stand-on, multi-frequency impedance device and a supine impedance spectroscopy instrument. Significant differences were found between the instruments in both absolute impedance values and, importantly, inter-limb impedance ratios. Since impedance ratios in healthy individuals provide the reference standard for detection of lymphedema, these data indicate that the methods are not interchangeable. Consideration of the errors associated with each method indicates that the BIS remains the preferred method for lymphedema detection. PMID:26574711
Segmental and whole body electrical impedance measurements in dialysis patients
Nescolarde Selva, Lexa
2006-01-01
The main objective of this thesis is to contribute to the prevention and control of the cardiovascular risk, hydration state and nutritional state in dialysis patients using non-invasive electrical impedance measurements. The thesis is structured in three parts with the following objectives: 1) to establish electrical impedance reference data for healthy Cuban population, 2)to improve the diagnostic based on impedance methods in Cuban hemodialysis (HD)patients and 3) to develop the impedance ...
Using A Particular Sampling Method for Impedance Measurement
Lentka Grzegorz
2014-01-01
The paper presents an impedance measurement method using a particular sampling method which is an alternative to DFT calculation. The method uses a sine excitation signal and sampling response signals proportional to current flowing through and voltage across the measured impedance. The object impedance is calculated without using Fourier transform. The method was first evaluated in MATLAB by means of simulation. The method was then practically verified in a constructed simple impedance measu...
The electrochemical impedance of metal hydride electrodes
DEFF Research Database (Denmark)
Valøen, Lars Ole; Lasia, Andrzej; Jensen, Jens Oluf;
2002-01-01
The electrochemical impedance responses for different laboratory type metal hydride electrodes were successfully modeled and fitted to experimental data for AB5 type hydrogen storage alloys as well as one MgNi type electrode. The models fitted the experimental data remarkably well. Several AC......, explaining the experimental impedances in a wide frequency range for electrodes of hydride forming materials mixed with copper powder, were obtained. Both charge transfer and spherical diffusion of hydrogen in the particles are important sub processes that govern the total rate of the electrochemical...... hydrogen absorption/desorption reaction. To approximate the experimental data, equations describing the current distribution in porous electrodes were needed. Indications of one or more parallel reduction/oxidation processes competing with the electrochemical hydrogen absorption/desorption reaction were...
Impedance effects in the CLIC damping rings
Koukovini-Platia, E; Mounet, N; Rumolo, G; Salvant, B
2011-01-01
Due to the unprecedented brilliance of the beams, the performance of the Compact Linear Collider (CLIC) damping rings (DR) is affected by collective effects. Single bunch instability thresholds based on a broad-band resonator model and the associated coherent tune shifts have been evaluated with the HEADTAIL code. Simulations performed for positive and negative values of chromaticity showed that higher order bunch modes can be potentially dangerous for the beam stability. This study also includes the effects of high frequency resistive wall impedance due to different coatings applied on the chambers of the wigglers for e-cloud mitigation and/or ultra-low vacuum pressure. The impact of the resistive wall wake fields on the transverse impedance budget is finally discussed.
Are patents impeding medical care and innovation?
Directory of Open Access Journals (Sweden)
E Richard Gold
2010-01-01
Full Text Available BACKGROUND TO THE DEBATE: Pharmaceutical and medical device manufacturers argue that the current patent system is crucial for stimulating research and development (R&D, leading to new products that improve medical care. The financial return on their investments that is afforded by patent protection, they claim, is an incentive toward innovation and reinvestment into further R&D. But this view has been challenged in recent years. Many commentators argue that patents are stifling biomedical research, for example by preventing researchers from accessing patented materials or methods they need for their studies. Patents have also been blamed for impeding medical care by raising prices of essential medicines, such as antiretroviral drugs, in poor countries. This debate examines whether and how patents are impeding health care and innovation.
Applications of Nonlinear Electrochemical Impedance Spectroscopy (NLEIS)
Adler, S. B.
2013-08-31
This paper reviews the use of nonlinear electrochemical impedance spectroscopy (NLEIS) in the analysis of SOFC electrode reactions. By combining EIS and NLEIS, as well as other independent information about an electrode material, it becomes possible to establish quantitative links between electrochemical kinetics and materials properties, even when systems are unstable with time. After a brief review of the method, this paper summarizes recent results analyzing the effects of Sr segregation in thin-film LSC electrodes. © The Electrochemical Society.
Learning collaborative impedance-based robot behaviors
Rozo, Leonel; Calinon, Sylvain; Caldwell, Darwin,; Jiménez Schlegl, Pablo; Torras, Carme
2013-01-01
Research in learning from demonstration has focused on transferring movements from humans to robots. However, a need is arising for robots that do not just replicate the task on their own, but that also interact with humans in a safe and natural way to accomplish tasks cooperatively. Robots with variable impedance capabilities opens the door to new challenging applications, where the learning algorithms must be extended by encapsulating force and vision information. In this paper we propose a...
Impedance Matched Absorptive Thermal Blocking Filters
Wollack, E J; Rostem, K; U-Yen, K
2014-01-01
We have designed, fabricated and characterized absorptive thermal blocking filters for cryogenic microwave applications. The transmission line filter's input characteristic impedance is designed to match $50\\,\\Omega$ and its response has been validated from 0-to-50\\,GHz. The observed return loss in the 0-to-20\\,GHz design band is greater than $20\\,$dB and shows graceful degradation with frequency. Design considerations and equations are provided that enable this approach to be scaled and modified for use in other applications.
New methods of measuring normal acoustic impedance
Wayman, James L.
1984-01-01
In recent years new methods based on signal processing technical have been developed to measure the normal acoustic impedance of materials. These methods proved to be considerably faster easier to implement than the SRW method rhey replace. Mathematical, hardware and software aspects of these techniques are discussed and results obtained over a frequency range of 200-4000 Hz for several architectural materials are presented. NPS Foundation Research Program http://archive....
Bioelectrical Impedance Assessment of Wound Healing
Lukaski, Henry C.; Moore, Micheal
2012-01-01
Objective assessment of wound healing is fundamental to evaluate therapeutic and nutritional interventions and to identify complications. Despite availability of many techniques to monitor wounds, there is a need for a safe, practical, accurate, and effective method. A new method is localized bioelectrical impedance analysis (BIA) that noninvasively provides information describing cellular changes that occur during healing and signal complications to wound healing. This article describes the ...
Detection of irradiated potatoes by impedance measurements
International Nuclear Information System (INIS)
The impedance ratio at 5kHz to 50kHz (Z6K/Z50K) determined at 22degC at an apical region of potato tuber which was pre-incubated at 22degC for 3 days or longer resulted in the best detection of radian treatment. Irradiated potatoes of 10 cultivars could be detected with this method, and potatoes 'Danshaku' commercially irradiated at Shihoro could be distinguished from unirradiated 'Danshaku'. (author)
Electrical impedance studies of uranium oxide
International Nuclear Information System (INIS)
The thesis presents data on the electrical properties of uranium oxide at temperatures from 1700K to 4.2K, and pressures between 25 K bar and 70 K bar. The impedance data were analysed using the technique of complex plane representation to establish the conductivity and dielectric constant of uranium dioxide. The thermophysical data were compared with previously reported experimental and theoretical work on uranium dioxide and other fluorite structured oxides. (U.K.)
Estimating the Transverse Impedance in the Fermilab Recycler
Energy Technology Data Exchange (ETDEWEB)
Ainsworth, Robert [Fermilab; Adamson, Philip [Fermilab; Burov, Alexey [Fermilab; Kourbanis, Ioanis [Fermilab; Yang, Ming-Jen [Fermilab
2016-06-01
Impedance could represent a limitation of running high intensity bunches in the Fermilab recycler. With high intensity upgrades foreseen, it is important to quantify the impedance. To do this, studies have been performed measuring the tune shift as a function of bunch intensity allowing the transverse impedance to be derived.
Impedance of a beam tube with antechamber
International Nuclear Information System (INIS)
A beam vacuum chamber was proposed to allow synchrotron light to radiate from a circulating electron beam into an antechamber containing photon targets, pumps, etc. To determine the impedance, electromagnetic measurements were carried out on a section of chamber using for low frequencies a current-carrying wire and for up to 16 GHz, a resonance perturbation method. Because the response of such a chamber would depend on upstream and downstream restrictions of aperture yet to be determined, the resonance studies were analyzed in some generality. The favorable conclusion of these studies is that the antechamber makes practically no contribution to either the longitudinal or the transverse impedances. The small beam tube with slot-isolated antechamber responds as a small cylinder up to the frequency (∼ 15 GHz) where energy can propagate transversely in the narrow slot. The important function of the slot is to prevent the beam from coupling to the many lower-frequency modes of the antechamber. If cut off frequency were higher than the slot-coupling at 15 GHz, only an increased density of modes is expected, with little change in average beam impedances
Bioelectrical impedance analysis of bovine milk fat
Veiga, E. A.; Bertemes-Filho, P.
2012-12-01
Three samples of 250ml at home temperature of 20°C were obtained from whole, low fat and fat free bovine UHT milk. They were analysed by measuring both impedance spectra and dc conductivity in order to establish the relationship between samples related to fat content. An impedance measuring system was developed, which is based on digital oscilloscope, a current source and a FPGA. Data was measured by the oscilloscope in the frequency 1 kHz to 100 kHz. It was showed that there is approximately 7.9% difference in the conductivity between whole and low fat milk whereas 15.9% between low fat and free fat one. The change of fatness in the milk can be significantly sensed by both impedance spectra measurements and dc conductivity. This result might be useful for detecting fat content of milk in a very simple way and also may help the development of sensors for measuring milk quality, as for example the detection of mastitis.
Bioelectrical impedance analysis of bovine milk fat
International Nuclear Information System (INIS)
Three samples of 250ml at home temperature of 20°C were obtained from whole, low fat and fat free bovine UHT milk. They were analysed by measuring both impedance spectra and dc conductivity in order to establish the relationship between samples related to fat content. An impedance measuring system was developed, which is based on digital oscilloscope, a current source and a FPGA. Data was measured by the oscilloscope in the frequency 1 kHz to 100 kHz. It was showed that there is approximately 7.9% difference in the conductivity between whole and low fat milk whereas 15.9% between low fat and free fat one. The change of fatness in the milk can be significantly sensed by both impedance spectra measurements and dc conductivity. This result might be useful for detecting fat content of milk in a very simple way and also may help the development of sensors for measuring milk quality, as for example the detection of mastitis.
PEP-X IMPEDANCE AND INSTABILITY CALCULATIONS
Energy Technology Data Exchange (ETDEWEB)
Bane, K.L.F.; Lee, L.-Q.; Ng, C.; Stupakov, G.; au Wang, L.; Xiao, L.; /SLAC
2010-08-25
PEP-X, a next generation, ring-based light source is designed to run with beams of high current and low emittance. Important parameters are: energy 4.5 GeV, circumference 2.2 km, beam current 1.5 A, and horizontal and vertical emittances, 185 pm by 8 pm. In such a machine it is important that impedance driven instabilities not degrade the beam quality. In this report they study the strength of the impedance and its effects in PEP-X. For the present, lacking a detailed knowledge of the vacuum chamber shape, they create a straw man design comprising important vacuum chamber objects to be found in the ring, for which they then compute the wake functions. From the wake functions they generate an impedance budget and a pseudo-Green function wake representing the entire ring, which they, in turn, use for performing microwave instability calculations. In this report they, in addition, consider in PEP-X the transverse mode-coupling, multi-bunch transverse, and beam-ion instabilities.
Application of impedance spectroscopy to SOFC research
Energy Technology Data Exchange (ETDEWEB)
Hsieh, G.; Mason, T.O. [Northwestern Univ., Evanston, IL (United States); Pederson, L.R. [Pacific Northwest National Lab., Richland, WA (United States)
1996-12-31
With the resurgence of interest in solid oxide fuel cells and other solid state electrochemical devices, techniques originally developed for characterizing aqueous systems are being adapted and applied to solid state systems. One of these techniques, three-electrode impedance spectroscopy, is particularly powerful as it allows characterization of subcomponent and interfacial properties. Obtaining accurate impedance spectra, however, is difficult as reference electrode impedance is usually non-negligible and solid electrolytes typically have much lower conductance than aqueous solutions. Faidi et al and Chechirlian et al have both identified problems associated with low conductivity media. Other sources of error are still being uncovered. Ford et al identified resistive contacts with large time constants as a possibility, while Me et al showed that the small contact capacitance of the reference electrode was at fault. Still others show that instrument limitations play a role. Using the voltage divider concept, a simplified model that demonstrates the interplay of these various factors, predicts the form of possible distortions, and offers means to minimize errors is presented.
Qureshi, Tabassum-Ur-Razaq; Chatwin, Chris; Huber, Nicolas; Zarafshani, Ali; Tunstall, Benjamin; Wang, Wei
2010-01-01
The current source is a key component in bio-impedance measurement systems. The accuracy of the current source can be measured in terms of its output impedance together with other parameters, with certain applications demanding extremely high output impedance. This paper presents an investigation and comparison of different current source designs based on the Enhanced Howland circuit combined with a General Impedance Converter (GIC) circuit using both ideal and non-ideal operational amplifier...
Impedance analysis of fibroblastic cell layers measured by electric cell-substrate impedance sensing
Lo, Chun-Min; Ferrier, Jack
1998-06-01
Impedance measurements of cell layers cultured on gold electrode surfaces obtained by electric cell-substrate impedance sensing provide morphological information such as junctional resistance and cell-substrate separation. Previously, a model that assumes that cells have a disklike shape and that electric currents flow radially underneath the ventral cell surface and then through the paracellular space has been used to theoretically calculate the impedance of the cell-covered electrode. In this paper we propose an extended model of impedance analysis for cell layers where cellular shape is rectangular. This is especially appropriate for normal fibroblasts in culture. To verify the model, we analyze impedance data obtained from four different kinds of fibroblasts that display a long rectangular shape. In addition, we measure the average cell-substrate separation of human gingival fibroblasts at different temperatures. At temperatures of 37, 22, and 4 °C, the average separation between ventral cell surface and substratum are 46, 55, and 89 nm, respectively.
Using A Particular Sampling Method for Impedance Measurement
Directory of Open Access Journals (Sweden)
Lentka Grzegorz
2014-08-01
Full Text Available The paper presents an impedance measurement method using a particular sampling method which is an alternative to DFT calculation. The method uses a sine excitation signal and sampling response signals proportional to current flowing through and voltage across the measured impedance. The object impedance is calculated without using Fourier transform. The method was first evaluated in MATLAB by means of simulation. The method was then practically verified in a constructed simple impedance measurement instrument based on a PSoC (Programmable System on Chip. The obtained calculation simplification recommends the method for implementation in simple portable impedance analyzers destined for operation in the field or embedding in sensors.
Wakefield and impedance studies of a liner using MAFIA
Energy Technology Data Exchange (ETDEWEB)
Chou, W.; Barts, T.
1993-03-01
The liner is a perforated beam tube that is coaxial with an outer bore tube. The 3D code MAFIA (version 3.1) is used to study the wakefields, impedances, and resonances of this structure. The short-range wakes and low-frequency (below the cutoff) impedances are in agreement with the theoretical model. The long-range wakes and high-frequency resonances are associated with the distribution of the holes (or slots). The dependence of the impedance on the size, shape, and pattern of the holes (or slots) is studied. The impact of the liner impedance on the Superconducting Super Collider impedance budget is discussed.
Wakefield and impedance studies of a liner using MAFIA
Energy Technology Data Exchange (ETDEWEB)
Chou, W.; Barts, T. (SSC Laboratory, Dallas, Texas 75237 (United States))
1993-12-25
The liner is a perforated beam tube which is coaxial with an outer bore tube. The 3D code MAFIA version 3.1 is used to study the wakefields, impedances, and resonances of this structure. The short range wakes and low frequency (below the cutoff) impedances are in agreement with the theoretical model. The long range wakes and high frequency resonances are associated with the distribution of the holes (or slots). The dependence of the impedance on the size, shape, and pattern of the holes (or slots) is studied. The impact of the liner impedance on the SSC impedance budget is discussed.
Wakefield and impedance studies of a liner using MAFIA
Chou, W.; Barts, T.
1993-12-01
The liner is a perforated beam tube which is coaxial with an outer bore tube. The 3D code MAFIA version 3.1 is used to study the wakefields, impedances, and resonances of this structure. The short range wakes and low frequency (below the cutoff) impedances are in agreement with the theoretical model. The long range wakes and high frequency resonances are associated with the distribution of the holes (or slots). The dependence of the impedance on the size, shape, and pattern of the holes (or slots) is studied. The impact of the liner impedance on the SSC impedance budget is discussed.
Impedance spectroscopy and electrical modeling of electrowetting on dielectric devices
International Nuclear Information System (INIS)
Using impedance spectroscopy, we have determined models for the elements which determine the ac electrical behavior in electrowetting on dielectric (EWOD) systems. Three commonly used EWOD electrode configurations were analyzed. In each case, the impedance can be modeled by a combination of elements, including the solution resistance, the capacitance of the dielectric layer, and the constant phase impedance of the electrode double layers. The sensitivity of the system’s impedance to variations in the electrowetted area is also analyzed for these common configurations. We also demonstrate that the impedance per unit area of typical EWOD systems is invariant to bias voltage. (paper)
Impedance Modeling of Solid Oxide Fuel Cell Cathodes
DEFF Research Database (Denmark)
Mortensen, Jakob Egeberg; Søgaard, Martin; Jacobsen, Torben
2010-01-01
partial pressure range 0.028-1.00 atm. The recorded impedance spectra were successfully analyzed using the developed impedance model in the investigated temperature and oxygen partial pressure range. It is also demonstrated that the model can be used to predict how impedance spectra evolve with different......A 1-dimensional impedance model for a solid oxide fuel cell cathode is formulated and applied to a cathode consisting of 50/50 wt% strontium doped lanthanum cobaltite and gadolinia doped ceria. A total of 42 impedance spectra were recorded in the temperature range: 555-852°C and in the oxygen...
Superconducting fault current-limiter with variable shunt impedance
Llambes, Juan Carlos H; Xiong, Xuming
2013-11-19
A superconducting fault current-limiter is provided, including a superconducting element configured to resistively or inductively limit a fault current, and one or more variable-impedance shunts electrically coupled in parallel with the superconducting element. The variable-impedance shunt(s) is configured to present a first impedance during a superconducting state of the superconducting element and a second impedance during a normal resistive state of the superconducting element. The superconducting element transitions from the superconducting state to the normal resistive state responsive to the fault current, and responsive thereto, the variable-impedance shunt(s) transitions from the first to the second impedance. The second impedance of the variable-impedance shunt(s) is a lower impedance than the first impedance, which facilitates current flow through the variable-impedance shunt(s) during a recovery transition of the superconducting element from the normal resistive state to the superconducting state, and thus, facilitates recovery of the superconducting element under load.
A review of impedance measurements of whole cells.
Xu, Youchun; Xie, Xinwu; Duan, Yong; Wang, Lei; Cheng, Zhen; Cheng, Jing
2016-03-15
Impedance measurement of live biological cells is widely accepted as a label free, non-invasive and quantitative analytical method to assess cell status. This method is easy-to-use and flexible for device design and fabrication. In this review, three typical techniques for impedance measurement, i.e., electric cell-substrate impedance sensing, Impedance flow cytometry and electric impedance spectroscopy, are reviewed from the aspects of theory, to electrode design and fabrication, and applications. Benefiting from the integration of microelectronic and microfluidic techniques, impedance sensing methods have expanded their applications to nearly all aspects of biology, including living cell counting and analysis, cell biology research, cancer research, drug screening, and food and environmental safety monitoring. The integration with other techniques, the fabrication of devices for certain biological assays, and the development of point-of-need diagnosis devices is predicted to be future trend for impedance sensing techniques. PMID:26513290
Effective surface impedance of a high temperature superconductor
International Nuclear Information System (INIS)
Highlights: •Effective surface impedance depends on London penetration depth and operating temperature, T. •The effective surface resistance and the effective surface reactance are important. •The surface resistance determines the electromagnetic power dissipation. -- Abstract: In this work, we will investigate theoretically the effective surface impedance of a high temperature superconducting thin film in the microwave range. This investigation will be according to varying the thickness of the superconducting thin film and the operating temperature, based on means of impedance transformations. The microwave surface impedance is related to the fundamental properties of high-Tc superconductors. The study of surface impedance has two important aspects. The first one is the real part of the surface impedance; the surface resistance determines the electromagnetic power dissipation through the superconductor thin film. The second is the imaginary part of the surface impedance; the surface reactance is proportional to the magnetic-field penetration depth
Effective surface impedance of a high temperature superconductor
Energy Technology Data Exchange (ETDEWEB)
Aly, Arafa H., E-mail: arafa16@yahoo.com; Sabra, Walied
2013-12-15
Highlights: •Effective surface impedance depends on London penetration depth and operating temperature, T. •The effective surface resistance and the effective surface reactance are important. •The surface resistance determines the electromagnetic power dissipation. -- Abstract: In this work, we will investigate theoretically the effective surface impedance of a high temperature superconducting thin film in the microwave range. This investigation will be according to varying the thickness of the superconducting thin film and the operating temperature, based on means of impedance transformations. The microwave surface impedance is related to the fundamental properties of high-T{sub c} superconductors. The study of surface impedance has two important aspects. The first one is the real part of the surface impedance; the surface resistance determines the electromagnetic power dissipation through the superconductor thin film. The second is the imaginary part of the surface impedance; the surface reactance is proportional to the magnetic-field penetration depth.
Acoustic impedances of audiometric earphones coupled to different loads
DEFF Research Database (Denmark)
Ciric, Dejan; Hammershøi, Dorte
The acoustic impedance of an audiometric earphone is one of the factors that can affect sound transmission through the ear during tests of hearing sensitivity. Similar situation exists during calibration of the earphone where its impedance can affect sound transmission through a coupler. The...... importance of this impedance is related to the contribution of other elements involved in transmission such as ear canal impedance or impedance of the coupler seen from outside. In order to determine the acoustic impedances of five audiometric earphones, the standardized method for measurement of complex......, audiometric earphones are coupled to different loads. Thus, they are placed on different terminations of the tube including metal plate, artificial rubber pinna and upper part of the standardized coupler. The results show that the impedances of earphones are different, but they also differ from radiation...
International Nuclear Information System (INIS)
We report on the impedance mapping of in vitro cellular morphology by electrical impedance spectroscopy, using microelectrodes. A micro multielectrode system was designed, fabricated, assembled, tested and demonstrated for the monitoring of anchorage-dependent cell behavior and morphology. This system allowed continuous, label-free, quantitative monitoring and visualization of cell adhesion, spreading, proliferation and detachment due to cell cycle processes as well as cell–drug interaction, with spatio-temporal resolution. OvCa429 ovarian cancer cells were monitored in vitro over a period of 70 hours by inoculating the cell suspension directly on the multielectrode device. The phase angle of impedance was observed to develop a distinctive shape as a result of cell attachment and proliferation. The shape of the phase angle curve reverted back to the pre-attachment shape upon detachment of cells from the substrate, caused by the addition of trypsin to the cell culture medium. The impedance data of the cell culture were then successfully modeled as a multi-parametric equivalent circuit. The model incorporated both interfacial and cell-layer impedance parameters. Upon addition of trypsin, the cell-layer parameters showed a marked decline and were eventually eliminated from the multi-parametric model, confirming the correlation of the model to the electrode–cell–electrolyte system. These experiments demonstrate the applicability of the impedance mapping technique in visualizing and quantifying physiological changes in the cell layer due to cellular processes as well as the effect of external chemical stimulus on cells (cell–drug interaction)
Method for conducting nonlinear electrochemical impedance spectroscopy
Adler, Stuart B.; Wilson, Jamie R.; Huff, Shawn L.; Schwartz, Daniel T.
2015-06-02
A method for conducting nonlinear electrochemical impedance spectroscopy. The method includes quantifying the nonlinear response of an electrochemical system by measuring higher-order current or voltage harmonics generated by moderate-amplitude sinusoidal current or voltage perturbations. The method involves acquisition of the response signal followed by time apodization and fast Fourier transformation of the data into the frequency domain, where the magnitude and phase of each harmonic signal can be readily quantified. The method can be implemented on a computer as a software program.
Magnetic field dependence of rf surface impedance
International Nuclear Information System (INIS)
In this paper the surface impedance, Zs, is calculated for type-II superconductors subjected to a static magnetic field and small-amplitude microwave radiation. A complex penetration depth is calculated by using a model of vortex dynamics including a linear pinning restoring force and a viscous drag force. The static magnetic field dependence of the surface resistance Rs and surface reactance Xs is found by examining the dependence of the complex penetration depth on the length scales for low-field penetration, pinning penetration, and flux-flow penetration. In turn, from Rs, the static magnetic field dependence of the rate of energy dissipation is found
Transferring human impedance regulation skills to robots
Ajoudani, Arash
2016-01-01
This book introduces novel thinking and techniques to the control of robotic manipulation. In particular, the concept of teleimpedance control as an alternative method to bilateral force-reflecting teleoperation control for robotic manipulation is introduced. In teleimpedance control, a compound reference command is sent to the slave robot including both the desired motion trajectory and impedance profile, which are then realized by the remote controller. This concept forms a basis for the development of the controllers for a robotic arm, a dual-arm setup, a synergy-driven robotic hand, and a compliant exoskeleton for improved interaction performance.
Development of the impedance void meter
Energy Technology Data Exchange (ETDEWEB)
Chung, Moon Ki; Song, Chul Hwa; Won, Soon Yeon; Kim, Bok Deuk [Korea Atomic Energy Research Institute, Taejon (Korea, Republic of)
1994-06-01
An impedance void meter is developed to measure the area-averaged void fraction. Its basic principle is based on the difference in the electrical conductivity between phases. Several methods of measuring void fraction are briefly reviewed and the reason why this type of void meter is chosen to develop is discussed. Basic principle of the measurement is thoroughly described and several design parameters to affect the overall function are discussed in detail. As example of applications is given for vertical air-water flow. It is shown that the current design has good dynamic response as well as very fine spatial resolution. (Author) 47 refs., 37 figs.
Adaptive techniques in electrical impedance tomography reconstruction
International Nuclear Information System (INIS)
We present an adaptive algorithm for solving the inverse problem in electrical impedance tomography. To strike a balance between the accuracy of the reconstructed images and the computational efficiency of the forward and inverse solvers, we propose to combine an adaptive mesh refinement technique with the adaptive Kaczmarz method. The iterative algorithm adaptively generates the optimal current patterns and a locally-refined mesh given the conductivity estimate and solves for the unknown conductivity distribution with the block Kaczmarz update step. Simulation and experimental results with numerical analysis demonstrate the accuracy and the efficiency of the proposed algorithm. (paper)
Low impedance kicker for high intensity accelerator
International Nuclear Information System (INIS)
A kicker magnet is one of the most important components in a synchrotron. To kick the beam from injection line to synchrotron orbit or opposite direction, a ferrite material is used to achieve a fast rise time. In case of a high intensity accelerator, the material is heated by beam current. Sometimes, it is reported that characteristic of material is changed when the material reaches Curie temperature. In this paper, we propose a low impedance kicker which consists of thin magnetic alloy ribbon. The structure of the new kicker is suitable to avoid to receive the beam effects. (author)
Impedance matching for broadband piezoelectric energy harvesting
International Nuclear Information System (INIS)
This paper presents a system design for broadband piezoelectric energy harvesting by means of impedance matching. An inductive load impedance is emulated by controlling the output current of the piezoelectric harvester with a bipolar boost converter. The reference current is derived from the low pass filtered voltage measured at the harvester terminals. In order to maximize the harvested power especially for nonresonant frequencies the filter parameters are adjusted by a simple optimization algorithm. However the amount of harvested power is limited by the efficiency of the bipolar boost converter. Therefore an additional switch in the bipolar boost converter is proposed to reduce the capacitive switching losses. The proposed system is simulated using numerical parameters of available discrete components. Using the additional switch, the harvested power is increased by 20%. The proposed system constantly harvests 80% of the theoretically available power over frequency. The usable frequency range of ±4Hz around the resonance frequency of the piezoelectric harvester is mainly limited due to the boost converter topology. This comparison does not include the power dissipation of the control circuit
Electrochemical Impedance Spectroscopy of Conductive Polymer Coatings
Calle, Luz Marina; MacDowell, Louis G.
1996-01-01
Electrochemical impedance spectroscopy (EIS) was used to investigate the corrosion protection performance of twenty nine proprietary conductive polymer coatings for cold rolled steel under immersion in 3.55 percent NaCl. Corrosion potential as well as Bode plots of the data were obtained for each coating after one hour immersion, All coatings, with the exception of one, have a corrosion potential that is higher in the positive direction than the corrosion potential of bare steel under the same conditions. Group A consisted of twenty one coatings with Bode plots indicative of the capacitive behavior characteristic of barrier coatings. An equivalent circuit consisting of a capacitor in series with a resistor simulated the experimental EIS data for these coatings very well. Group B consisted of eight coatings that exhibited EIS spectra showing an inflection point which indicates that two time constants are present. This may be caused by an electrochemical process taking place which could be indicitive of coating failing. These coatings have a lower impedance that those in Group A.
AC Impedance Behaviour of Black Diamond Films
Institute of Scientific and Technical Information of China (English)
Haitao YE; Olivier GAUDIN; Richard B.JACKMAN
2005-01-01
The first measurement of impedance on free-standing diamond films from 0.1 Hz to 10 MHz up to 300℃ were reported. A wide range of chemical vapour deposition (CVD) materials were investigated, but here we concentrate are well fitted to a RC parallel circuit model and the equivalent resistance and capacitance for the diamond films have been estimated using the Zview curve fitting. The results show only one single semicircle response at each temperature measured. It was found that the resistance decreases from 62 MΩ at room temperature to 4 kΩ at300℃, with an activation energy around 0.51 eV. The equivalent capacitance is maintained at the level of 100 pF up to 300℃ suggesting that the diamond grain boundaries are dominating the conduction. At 400℃, the impedance at low frequencies shows a linear tail, which can be explained that the AC polarization of diamond/Au interface occurs.
Study on Impedance Characteristics of Aircraft Cables
Directory of Open Access Journals (Sweden)
Weilin Li
2016-01-01
Full Text Available Voltage decrease and power loss in distribution lines of aircraft electric power system are harmful to the normal operation of electrical equipment and may even threaten the safety of aircraft. This study investigates how the gap distance (the distance between aircraft cables and aircraft skin and voltage frequency (variable frequency power supply will be adopted for next generation aircraft will affect the impedance of aircraft cables. To be more precise, the forming mechanism of cable resistance and inductance is illustrated in detail and their changing trends with frequency and gap distance are analyzed with the help of electromagnetic theoretical analysis. An aircraft cable simulation model is built with Maxwell 2D and the simulation results are consistent with the conclusions drawn from the theoretical analysis. The changing trends of the four core parameters of interest are analyzed: resistance, inductance, reactance, and impedance. The research results can be used as reference for the applications in Variable Speed Variable Frequency (VSVF aircraft electric power system.
Fisher information analysis in electrical impedance tomography
International Nuclear Information System (INIS)
This paper provides a quantitative analysis of the optimal accuracy and resolution in electrical impedance tomography (EIT) based on the Cramér–Rao lower bound. The imaging problem is characterized by the forward operator and its Jacobian. The Fisher information operator is defined for a deterministic parameter in a real Hilbert space and a stochastic measurement in a finite-dimensional complex Hilbert space with a Gaussian measure. The connection between the Fisher information and the singular value decomposition (SVD) based on the maximum likelihood (ML) criterion (the ML-based SVD) is established. It is shown that the eigenspaces of the Fisher information provide a suitable basis to quantify the trade-off between the accuracy and the resolution of the (nonlinear) inverse problem. It is also shown that the truncated ML-based pseudo-inverse is a suitable regularization strategy for a linearized problem, which exploits sufficient statistics for estimation within these subspaces. The statistical-based Cramér–Rao lower bound provides a complement to the deterministic upper bounds and the L-curve techniques that are employed with linearized inversion. To this end, electrical impedance tomography provides an interesting example where the eigenvalues of the SVD usually do not exhibit a very sharp cut-off, and a trade-off between the accuracy and the resolution may be of practical importance. A numerical study of a hypothetical EIT problem is described, including a statistical analysis of the model errors due to the linearization. (paper)
Electrical impedance tomography: so close to touching the holy grail
2014-01-01
Electrical impedance tomography is a new technology giving us lung imaging that may allow lung function to be monitored at the bedside. Several applications have been studied to guide mechanical ventilation at the bedside with electrical impedance tomography. Positive end-expiratory pressure trials guided by electrical impedance tomography are relevant in terms of recruited volume or homogeneity of the lung. Tidal impedance variation is a new parameter of electrical impedance tomography that may help physicians with ventilator settings in acute respiratory distress syndrome patients. This parameter is able to identify the onset of overdistention in the nondependent part and recruitment in the dependent part. Electrical impedance tomography presents a big step forward in mechanical ventilation. PMID:25041593
Validation of a Numerical Method for Determining Liner Impedance
Watson, Willie R.; Jones, Michael G.; Tanner, Sharon E.; Parrott, Tony L.
1996-01-01
This paper reports the initial results of a test series to evaluate a method for determining the normal incidence impedance of a locally reacting acoustically absorbing liner, located on the lower wall of a duct in a grazing incidence, multi-modal, non-progressive acoustic wave environment without flow. This initial evaluation is accomplished by testing the methods' ability to converge to the known normal incidence impedance of a solid steel plate, and to the normal incidence impedance of an absorbing test specimen whose impedance was measured in a conventional normal incidence tube. The method is shown to converge to the normal incident impedance values and thus to be an adequate tool for determining the impedance of specimens in a grazing incidence, multi-modal, nonprogressive acoustic wave environment for a broad range of source frequencies.
Transverse beam coupling impedance of the CERN Proton Synchrotron
Persichelli, S.; Migliorati, M.; Biancacci, N.; Gilardoni, S.; Metral, E.; Salvant, B.
2016-04-01
Beam coupling impedance is a fundamental parameter to characterize the electromagnetic interaction of a particle beam with the surrounding environment. Synchrotron machine performances are critically affected by instabilities and collective effects triggered by beam coupling impedance. In particular, transverse beam coupling impedance is expected to impact beam dynamics of the CERN Proton Synchrotron (PS), since a significant increase in beam intensity is foreseen within the framework of the LHC Injectors Upgrade (LIU) project. In this paper we describe the study of the transverse beam coupling impedance of the PS, taking into account the main sources of geometrical impedance and the contribution of indirect space charge at different energies. The total machine impedance budget, determined from beam-based dedicated machine measurement sessions, is also discussed and compared with the theoretical model.
Harmonic analysis for identification of nonlinearities in impedance spectroscopy
Energy Technology Data Exchange (ETDEWEB)
Kiel, M.; Bohlen, O.; Sauer, D.U. [Electrochemical Energy Conversion and Storage Systems Group, Institute for Power Electronics and Electrical Drives (ISEA), RWTH Aachen University (Germany)
2008-10-30
Though impedance is only defined for linear systems, impedance spectroscopy is also successfully applied to nonlinear systems such as fuel cells and batteries. The influence of nonlinearities on measurement results in impedance spectroscopy is therefore discussed on a theoretical and simulative basis. The basis is a simplified Randles model of an electrochemical cell, on which a simulated impedance spectroscopy in galvanostatic mode is performed. For the investigation the focus is on the Butler-Vollmer equation in order to describe the nonlinearity. Furthermore, a linear model for comparison is used, in which the Butler-Volmer nonlinearity is replaced by a linear resistor to show the differences in impedance measurement. In order to find a correlation, also the occurring harmonics are observed. The results are discussed and several methods are suggested for maintaining a quasi-linear impedance measurement by controlling the amplitude of the excitation signal. (author)
Transverse impedance measurement in RHIC and the AGS
Energy Technology Data Exchange (ETDEWEB)
Biancacci, Nicolo [European Organization for Nuclear Research (CERN), Geneva (Switzerland); Blaskiewicz, M. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.; Dutheil, Y. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.; Liu, C. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.; Mernick, M. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.; Minty, M. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.; White, S. M. [Brookhaven National Lab. (BNL), Upton, NY (United States). Collider-Accelerator Dept.
2014-05-12
The RHIC luminosity upgrade program aims for an increase of the polarized proton luminosity by a factor 2. To achieve this goal a significant increase in the beam intensity is foreseen. The beam coupling impedance could therefore represent a source of detrimental effects for beam quality and stability at high bunch intensities. For this reason it is essential to quantify the accelerator impedance budget and the major impedance sources, and possibly cure them. In this MD note we summarize the results of the 2013 transverse impedance measurements in the AGS and RHIC. The studies have been performed measuring the tune shift as a function of bunch intensity and deriving the total accelerator machine transverse impedance. For RHIC, we could obtain first promising results of impedance localization measurements as well.
Active impedance metasurface with full 360° reflection phase tuning
Zhu, Bo O.; Junming Zhao; Yijun Feng
2013-01-01
Impedance metasurface is composed of electrical small scatters in two dimensional plane, of which the surface impedance can be designed to produce desired reflection phase. Tunable reflection phase can be achieved by incorporating active element into the scatters, but the tuning range of the reflection phase is limited. In this paper, an active impedance metasurface with full 360° reflection phase control is presented to remove the phase tuning deficiency in conventional approach. The unit ce...
Positive impedance humidity sensors via single-component materials
Jingwen Qian; Zhijian Peng; Zhenguang Shen; Zengying Zhao; Guoliang Zhang; Xiuli Fu
2016-01-01
Resistivity-type humidity sensors have been investigated with great interest due to the increasing demands in industry, agriculture and daily life. To date, most of the available humidity sensors have been fabricated based on negative humidity impedance, in which the electrical resistance decreases as the humidity increases, and only several carbon composites have been reported to present positive humidity impedance. However, here we fabricate positive impedance humidity sensors only via sing...
Impedance studies for the PS Finemet loaded longitudinal damper
Persichelli, S; Paoluzzi, M; Salvant, B
2014-01-01
The impedance of the Finemet® loaded longitudinal damper cavity, installed in the CERN Proton Synchrotron straight section 02 during the Long Shutdown 2013-2014, has been evaluated [1]. Time domain simulations with CST Particle Studio have been performed in order to get the longitudinal and transverse impedance of the device and make a comparison with the longitudinal impedance that was measured for a single cell prototype.
Effects of temperature on the electrical impedance of piezoelectric elements
Krishnamurthy, Karthik Chandran
1996-01-01
A structural health monitoring technique, developed at the Center for Intelligent Material Systems and Structures, employs piezoelectric (PZT) materials for tracking the structural impedance to qualitatively identify damage. The mechanical impedance of a structure is a function of the structure's mass, stiffness, damping, and structural boundary conditions. Changes in any of the above-mentioned properties lead to a change in the mechanical impedance of the structure and a chang...
On the Passivity Based Impedance Control of Flexible Joint Robots
Ott, Christian; Albu-Schäffer, Alin; Kugi, A; Hirzinger, Gerd
2008-01-01
In this work a novel type of impedance controllers for flexible joint robots is proposed. As a target impedance a desired stiffness and damping are considered without inertia shaping. For this problem two controllers of different complexity are proposed. Both have a cascaded structure with an inner torque feedback loop and an outer impedance controller. For the torque feedback, a physical interpretation as a scaling of the motor inertia is given, which allows to incorpora...
Qualitative Model of the Input Impedance of Rectangular Microstrip Antenna
Directory of Open Access Journals (Sweden)
Saeed Reza Ostadzadeh
Full Text Available In this paper, a fuzzy-based approach is proposed so as to predict the input impedance of the rectangular microstrip antenna. In the proposed approach, at first, behavior of single microstrip antenna is represented as simple and unchanged membership functions, and the feed probe effect on the input impedance is then extracted as simple curves so that the input impedance of microstrip antenna in despite of other existing models is efficiently predicted.
Line Impedance Estimation Using Active and Reactive Power Variations
DEFF Research Database (Denmark)
Timbus, Adrian Vasile; Rodriguez, Pedro; Teodorescu, Remus;
2007-01-01
This paper proposes an estimation method of power system impedance based on power variations caused by a distributed power generation system (DPGS) at the point of common coupling (PCC). The proposed algorithm is computationally simple and uses the voltage variations at the point of common coupling...... (PCC) caused by the variations of the power delivered to utility network to derive the value of grid impedance. Accurate estimation of both resistive and inductive part of the impedance is obtained, as the results presented show....
Impedance characterization of PV modules in outdoor conditions
DEFF Research Database (Denmark)
Oprea, Matei-lon; Thorsteinsson, Sune; Spataru, Sergiu;
2016-01-01
Impedance spectroscopy (IS) has been used for laboratory characterizations of photovoltaic (PV) technologies under well controlled conditions. This work applies IS for outdoor characterization of PV panels, in order to observe the effect of irradiance (G) and temperature (T) on the PV module......’s impedance spectrum, and further construct an impedance model that can link environmental changes to the model’s parameters. To achieve this, an optimized setup has been developed for long-term impedance spectra monitoring synchronised with accurate irradiance and temperature data. Preliminary results show...... clear correlation between the determined parameters and ambient conditions....
Transverse impedance measurements in RHIC and the AGS
Biancacci, N; Blaskiewicz, M; Liu, C; Mernick, K; Minty, M; White, S
2014-01-01
The RHIC luminosity upgrade program aims for an increase of the polarized proton luminosity by a factor 2. To achieve this goal a significant increase in the beam intensity is foreseen. The beam coupling impedance represents a source of detrimental effects for beam quality and stability at high bunch intensities. In this paper, we evaluate the global transverse impedance in both the AGS and RHIC with measurements of tune shift as a function of bunch intensity. The results are compared to past measurements and the present impedance model. First attempts at transverse impedance localization are as well presented for the RHIC Blue ring.
In vivo impedance spectroscopy of deep brain stimulation electrodes
Lempka, Scott F.; Miocinovic, Svjetlana; Johnson, Matthew D.; Vitek, Jerrold L.; McIntyre, Cameron C.
2009-08-01
Deep brain stimulation (DBS) represents a powerful clinical technology, but a systematic characterization of the electrical interactions between the electrode and the brain is lacking. The goal of this study was to examine the in vivo changes in the DBS electrode impedance that occur after implantation and during clinically relevant stimulation. Clinical DBS devices typically apply high-frequency voltage-controlled stimulation, and as a result, the injected current is directly regulated by the impedance of the electrode-tissue interface. We monitored the impedance of scaled-down clinical DBS electrodes implanted in the thalamus and subthalamic nucleus of a rhesus macaque using electrode impedance spectroscopy (EIS) measurements ranging from 0.5 Hz to 10 kHz. To further characterize our measurements, equivalent circuit models of the electrode-tissue interface were used to quantify the role of various interface components in producing the observed electrode impedance. Following implantation, the DBS electrode impedance increased and a semicircular arc was observed in the high-frequency range of the EIS measurements, commonly referred to as the tissue component of the impedance. Clinically relevant stimulation produced a rapid decrease in electrode impedance with extensive changes in the tissue component. These post-operative and stimulation-induced changes in impedance could play an important role in the observed functional effects of voltage-controlled DBS and should be considered during clinical stimulation parameter selection and chronic animal research studies.
Comprehensive characterization of thermophysical properties in solids using thermal impedance
Martínez-Flores, J. J.; Licea-Jiménez, L.; Pérez García, S. A.; Rodríguez-Viejo, J.; Alvarez-Quintana, J.
2012-11-01
Thermal impedance Zth(iω) is a way of defining the thermophysical characteristics and behavior of thermal systems. Existing photoacoustic and photothermal approaches based on thermal impedance formalism merely allows a partial thermal characterization of the materials (generally, either thermal diffusivity or thermal effusivity). In this work, a new approach based on the thermal impedance concept in terms of its characteristic thermal time constant is developed from thermal quadrupoles formalism. The approach outlined in this contribution presents a set of analytical equations in which through a single measurement of thermal impedance is sufficient to obtain a comprehensive characterization of the thermophysical properties of solid materials in a simple way.
Proceedings of the impedance and bunch instability workshop
Energy Technology Data Exchange (ETDEWEB)
1990-04-01
This report discusses the following topics: impedance and bunch lengthening; single bunch stability in the ESRF; a longitudinal mode-coupling instability model for bunch lengthening; high-frequency behavior of longitudinal coupling impedance; beam-induced energy spreads at beam-pipe transitions; on the calculation of wake functions using MAFIA-T3 code; preliminary measurements of the bunch length and the impedance of LEP; measurements and simulations of collective effects in the CERN SPS; bunch lengthening in the SLC damping rings; and status of impedance measurements for the spring-8 storage ring.
Impedance of the PEP-II DIP screen
Energy Technology Data Exchange (ETDEWEB)
Ng, C.-K. [Stanford Linear Accelerator Center, Menlo Park, CA (United States); Weiland, T.
1996-08-01
The vacuum chamber of a storage ring normally consists of periodically spaced pumping slots. The longitudinal impedance of slots are analyzed in this paper. It is found that although the broad-band impedance is tolerable, the narrow-band impedance, as a consequence of the periodicity of the slots, may exceed the stability limit given by natural damping with no feedback system on. Based on this analysis, the PEP-II distributed-ion-pump (DIP) screen uses long grooves with hidden holes cut halfway to reduce both the broad-band and narrow-band impedances. (author)
Esophageal impedance baseline according to different time intervals
Directory of Open Access Journals (Sweden)
Ummarino Dario
2012-06-01
Full Text Available Abstract Background The impedance baseline has been shown to reflect esophageal integrity, and to be decreased in patients with esophagitis. However, different methods for the determination of the impedance baseline have not been compared. Methods The median impedance baseline was calculated in 10 consecutive multichannel intraluminal impedance recordings in children with non-erosive reflux disease. All children underwent an endoscopy with a biopsy as part of the clinical work-up to exclude esophagitis. The impedance baseline was obtained both by including and excluding all impedance episodes (IE; reflux, swallows and gas episodes during the full recording, and during the first 1-minute period without an IE every hour (method 1, every 2 hours (method 2 or every 4 hours (method 3. The impedance baseline obtained during the full recording was set at 100%, and the variation (difference in impedance baseline for the different methods and variability (difference in impedance baseline during one analysis period were assessed. Results None of the participants had esophagitis. The mean difference over the six channels between the impedance baseline over the total recording with and without IE was approximately 2.5%, and comparable for each channel (range 0.47% to 5.55%. A mean of 1,028 IEs were excluded in each tracing, and it took between 4 and 24 hours to delete all events in one tracing. The difference in the impedance baseline obtained with and without IEs was mainly caused by the gas episodes in the upper channels and swallows in the lower channels. The median impedance baseline according to the three one-minute analysis methods was comparable to the median impedance baseline according to the 24 hour analysis. Conclusions The automatic determination of the median impedance baseline over the total tracing including IEs is an adequate method. In isolated tracings with numerous IEs, the calculation of the median impedance baseline over one minute
Validation of an Impedance Education Method in Flow
Watson, Willie R.; Jones, Michael G.; Parrott, Tony L.
2004-01-01
This paper reports results of a research effort to validate a method for educing the normal incidence impedance of a locally reacting liner, located in a grazing incidence, nonprogressive acoustic wave environment with flow. The results presented in this paper test the ability of the method to reproduce the measured normal incidence impedance of a solid steel plate and two soft test liners in a uniform flow. The test liners are known to be locally react- ing and exhibit no measurable amplitude-dependent impedance nonlinearities or flow effects. Baseline impedance spectra for these liners were therefore established from measurements in a conventional normal incidence impedance tube. A key feature of the method is the expansion of the unknown impedance function as a piecewise continuous polynomial with undetermined coefficients. Stewart's adaptation of the Davidon-Fletcher-Powell optimization algorithm is used to educe the normal incidence impedance at each Mach number by optimizing an objective function. The method is shown to reproduce the measured normal incidence impedance spectrum for each of the test liners, thus validating its usefulness for determining the normal incidence impedance of test liners for a broad range of source frequencies and flow Mach numbers. Nomenclature
Lower leg electrical impedance after distal bypass surgery
DEFF Research Database (Denmark)
Belanger, G K; Bolbjerg, M L; Heegaard, N H; Wiik, A; Schroeder, T V; Secher, N H
1998-01-01
Electrical impedance was determined in 13 patients following distal bypass surgery to evaluate lower leg oedema as reflected by its circumference. Tissue injury was assessed by the plasma concentration of muscle enzymes. After surgery, the volume of the control lower leg increased from 1250 (816...... concentration of troponin I (n = 8). In conclusion, tissue injury was reflected by increases in muscle enzymes in plasma. We found an inverse correlation between lower leg electrical impedance and volume, but the deviation in electrical impedance was approximately twice that of the leg volume. Electrical...... impedance appears to be a useful method for the evaluation of lower leg oedema after distal bypass surgery....
Distribution transformer specifiers should reconsider altering natural impedances
Energy Technology Data Exchange (ETDEWEB)
Wiegand, D
1991-02-01
Loss evaluation formulae cause modern transformer manufacturers to produce designs which are optimized to achieve the most competitive balance between evaluated losses and cost. If impedance is not presented as a constraint, its value may be quite high or low, and could lead to concern about fault protection where impedance is low, or regulation where impedance is high. The effects on impedance of evaluation formulae from three major Canadian utilities are tabulated. Reasons for the setting of constraints on impedance include the avoidance of an unacceptable level of voltage regulation by setting an upper limit, avoiding failure of the transformer due to load-side faults by applying a lower limit, coordinating system protection devices, or matching impedances of similar transformers purchased from different manufacturers. An urban type of formula will likely result in a transformer with low impedance down to 1.3%, causing concern about the ability of the transformer to withstand short circuit conditions, however the probability of this occurrence is very low, and circuit impedance increases rapidly as one moves from the tranformer to the load location. Thus, a transformer with an impedance value that appears low is not likely to make a system any less reliable, and it is recommended that the optimum loss design be accepted in such situations. 2 figs., 5 tabs.
Impedance Localization Measurements using AC Dipoles in the LHC
Biancacci, Nicolo; Papotti, Giulia; Persson, Tobias; Salvant, Benoit; Tomás, Rogelio
2016-01-01
The knowledge of the LHC impedance is of primary importance to predict the machine performance and allow for the HL-LHC upgrade. The developed impedance model can be benchmarked with beam measurements in order to assess its validity and limit. This is routinely done, for example, moving the LHC collimator jaws and measuring the induced tune shift. In order to localize possible unknown impedance sources, the variation of phase advance with intensity between beam position monitors can be measured. In this work we will present the impedance localization measurements performed at injection in the LHC using AC dipoles as exciter as well as the underlying theory.
Proceedings of the impedance and bunch instability workshop
International Nuclear Information System (INIS)
This report discusses the following topics: impedance and bunch lengthening; single bunch stability in the ESRF; a longitudinal mode-coupling instability model for bunch lengthening; high-frequency behavior of longitudinal coupling impedance; beam-induced energy spreads at beam-pipe transitions; on the calculation of wake functions using MAFIA-T3 code; preliminary measurements of the bunch length and the impedance of LEP; measurements and simulations of collective effects in the CERN SPS; bunch lengthening in the SLC damping rings; and status of impedance measurements for the spring-8 storage ring
Electrode contact impedance sensitivity to variations in geometry
International Nuclear Information System (INIS)
Electrode contact impedance is a crucial factor in physiological measurements and can be an accuracy-limiting factor when performing electroencephalography and electrical impedance tomography. In this work, standard flat electrodes and micromachined multipoint spiked electrodes are characterized with a finite-element method electromagnetic solver and the dependence of the contact impedance on geometrical factors is explored. It is found that flat electrodes are sensitive to changes in the outer skin layer properties related to hydration and thickness, while spike electrodes are not. The impedance as a function of the effective contact area, number of spikes and penetration depth has also been studied and characterized. (paper)
The vacuum impedance and unit systems
Kitano, M
2006-01-01
In electromagnetism, the vacuum impedance $Z_0$ is a universal constant, which is as important as the velocity of light $c$ in vacuum. Unfortunately, however, its significance does not seem to be appreciated so well and sometimes the presence itself is ignored. It is partly because in the Gaussian system of units, which has widely been used for long time, $Z_0$ is a dimensionless constant and of unit magnitude. In this paper, we clarify that $Z_0$ is a fundamental parameter in electromagnetism and plays major roles in the following scenes: reorganizing the structure of the electromagnetic formula in reference to the relativity; renormalizing the quantities toward natural unit systems starting from the SI unit system; and defining the magnitudes of electromagnetic units.
Coupled Transmission Lines as Impedance Transformer
DEFF Research Database (Denmark)
Jensen, Thomas; Zhurbenko, Vitaliy; Krozer, Viktor;
2007-01-01
A theoretical investigation of the use of a coupled line section as an impedance transformer is presented. We show how to properly select the terminations of the coupled line structures for effective matching of real and complex loads in both narrow and wide frequency ranges. The corresponding...... circuit configurations and the design procedures are proposed. Synthesis relations are derived and provided for efficient matching circuit construction. Design examples are given to demonstrate the flexibility and limitations of the design methods and to show their validity for practical applications....... Wideband matching performance with relative bandwidth beyond 100% and return loss > 20 dB is demonstrated both theoretically and experimentally. Good agreement is achieved between the measured and predicted performance of the coupled line transformer section....
Quartz tuning fork based microwave impedance microscopy
Cui, Yong-Tao; Ma, Eric Yue; Shen, Zhi-Xun
2016-06-01
Microwave impedance microscopy (MIM), a near-field microwave scanning probe technique, has become a powerful tool to characterize local electrical responses in solid state samples. We present the design of a new type of MIM sensor based on quartz tuning fork and electrochemically etched thin metal wires. Due to a higher aspect ratio tip and integration with tuning fork, such design achieves comparable MIM performance and enables easy self-sensing topography feedback in situations where the conventional optical feedback mechanism is not available, thus is complementary to microfabricated shielded stripline-type probes. The new design also enables stable differential mode MIM detection and multiple-frequency MIM measurements with a single sensor.
The sensitivity in Electrical Impedance Tomography
Directory of Open Access Journals (Sweden)
A. I. Rybin
2013-12-01
Full Text Available Introduction. The concept of sensitivity in Electrical Impedance Tomography is introduced (first – fourth type. The experimental researches measuring the voltages on the phantom outline are conducted on the created layout (for uniform cylindrical vessel with brine and placed inhomogeneities in a vessel. The main part. The inverse problems are solved for simulated on PC phantom (the third type sensitivity and from measured results (the fourth type sensitivity by conductivity zones method using regularization by A. Tykhonov. The sensitivity to conductivity increasing of elements inside the phantom is significantly less than the sensitivity to resistance increasing. The results of measured voltages processing and the results of projection reconstruction (obtained from mathematical model and from measured results are described. Conclusions. The satisfactory agreements of reconstruction results between themselves and with mathematical and measured phantoms are shown.
Relating membrane potential to impedance spectroscopy
Directory of Open Access Journals (Sweden)
Eugen Gheorghiu
2011-12-01
Full Text Available Non-invasive, label-free assessment of membrane potential of living cells is still a challenging task. The theory linking membrane potential to the low frequency α dispersion exhibited by suspensions of spherical shelled particles (presenting a net charge distribution on the inner side of the shell has been pioneered in our previous studies with emphasis on the permittivity spectra. Whereas α dispersion is related to a rather large variation exhibited by the permittivity spectrum, we report that the related decrement presented by the impedance magnitude spectrum is either extremely small, or occurs (for large cells at very small frequencies (~mHz explaining the lack of experimental bioimpedance data on the matter. We stress that appropriate choice of the parameters (as revealed by the microscopic model may enable access to membrane potential as well as to other relevant parameters when investigating living cells and charged lipid vesicles. We analyse the effect on the low frequency of the permittivity and impedance spectra of: I. Parameters pertaining to cell membrane i.e. (i membrane potential (through the amount of the net charge on the inner side of the membrane, (ii size of the cells/vesicles, (iii conductivity of the membrane; II. Parameters of the extra cellular medium (viscosity and conductivity. The applicability of the study has far reaching implications for basic (life sciences (providing non-invasive access to the dynamics of relevant cell parameters as well as for biosensing applications, e.g. assessment of cytotoxicity of a wide range of stimuli. doi:10.5617/jeb.214 J Electr Bioimp, vol. 2, pp. 93-97, 2011
Bioelectric impedance phase angle in breast carcinoma
Directory of Open Access Journals (Sweden)
Ruchi Tyagi
2014-01-01
Full Text Available Context: Worldwide breast cancer is the most frequently diagnosed life threatening cancer and the leading cause of death in women. Bioelectric impedance analysis (BIA affords an emerging opportunity to assess prognosis because of its ability to non invasively assess cell and plasma membrane structure and function by means of phase angle. Aims: To compare the phase angle between patients of breast cancer and their matched control with the help of BIA. Settings and Design: After taking clearance from ethical committee, a total of 34 female cases of histologically proven infiltrating ductal breast carcinoma were included from the surgery IPD, department of surgery. Equal numbers of the matched controls were recruited from the friends and relatives of cases. Materials and Methods: Bio Electrical Impedance Analyzer (BIA BODY STAT QUAD SCAN 4000 was used to measure resistance (R and reactance (Xc by recording a voltage drop in applied current. Phase angle is the ratio of reactance to resistance and is a measure of cell vitality. Statistical analysis used: Unpaired "t" test was applied. Results: In control group, the phase angle showed a mean of 5.479 whereas in test group, it showed a mean value of 4.726. The P value showed a significant difference (P < 0.0001. The smaller the phase angle values were higher was the tumor, nodes, metastases (TNM staging. The phase angles differed significantly from the healthy age matched control values. Conclusions: This study demonstrated that phase angle is a strong predictor of severity of breast cancer and differed significantly between the two groups.
Grant, Caroline A; Pham, Trang; Hough, Judith; Riedel, Thomas; Stocker, Christian; Schibler, Andreas
2011-01-01
Introduction Electrical impedance tomography (EIT) has been shown to be able to distinguish both ventilation and perfusion. With adequate filtering the regional distributions of both ventilation and perfusion and their relationships could be analysed. Several methods of separation have been suggested previously, including breath holding, electrocardiograph (ECG) gating and frequency filtering. Many of these methods require interventions inappropriate in a clinical setting. This study ther...
Bogónez Franco, Francisco; Nescolarde Selva, Lexa Digna; Bragós Bardia, Ramon; Rosell Ferrer, Francisco Javier; Yandiola, Iñigo
2009-01-01
The purpose of this study is to compare measurement errors in two commercially available multi-frequency bioimpedance analyzers, a Xitron 4000B and an ImpediMed SFB7, including electrode impedance mismatch. The comparison was made using resistive electrical models and in ten human volunteers. We used three different electrical models simulating three different body segments: the right-side, leg and thorax. In the electrical models, we tested the effect of the capacitive coupling of the ...
Broadband electrical impedance spectroscopy for dynamic electrical bio-impedance characterization
Sánchez Terrones, Benjamín
2012-01-01
The electrical impedance of biological samples is known in the literature as Electrical Bioimpedance (EBI). The Electrical Bioimpedance enables to characterize physiological conditions and events that are interesting for physiological research and medical diagnosis. Although the Electrical Bioimpedance weakness is that it depends on many physiological parameters, on the other hand, it is suitable for many medical applications where minimally invasive and real-time measurements with simple and...
Nguyen, D. T.; Kosobrodov, R.; Barry, M. A.; Chik, W.; Pouliopoulos, J.; Oh, T. I.; Thiagalingam, A.; McEwan, A.
2013-04-01
Recent studies in animal models suggest that the use of small volume boluses of NaCl as an impedance contrast agent can significantly improve pulmonary perfusion imaging by Electrical Impedance Tomography (EIT). However, these studies used highly concentrated NaCl solution (20%) which may have adverse effects on the patients. In a pilot experiment, we address this problem by comparing a number of different Impedance Contrast Boluses (ICBs). Conductivity changes in the lungs of a sheep after the injection of four different ICBs were compared, including three NaCl-based ICBs and one glucose-based ICB. The following procedure was followed for each ICB. Firstly, ventilation was turned off to provide an apneic window of approximately 40s to image the conductivity changes due to the ICB. Each ICB was then injected through a pig-tail catheter directly into the right atrium. EIT images were acquired throughout the apnea to capture the conductivity change. For each ICB, the experiment was repeated three times. The three NaCl-based ICB exhibited similar behaviour in which following the injection of each of these ICBs, the conductivity of each lung predictably increased. The effect of the ICB of 5% glucose solution was inconclusive. A small decrease in conductivity in the left lung was observed in two out of three cases and none was discernible in the right lung.
Modifying the acoustic impedance of polyurea-based composites
Nantasetphong, Wiroj; Amirkhizi, Alireza V.; Jia, Zhanzhan; Nemat-Nasser, Sia
2013-04-01
Acoustic impedance is a material property that depends on mass density and acoustic wave speed. An impedance mismatch between two media leads to the partial reflection of an acoustic wave sent from one medium to another. Active sonar is one example of a useful application of this phenomenon, where reflected and scattered acoustic waves enable the detection of objects. If the impedance of an object is matched to that of the surrounding medium, however, the object may be hidden from observation (at least directly) by sonar. In this study, polyurea composites are developed to facilitate such impedance matching. Polyurea is used due to its excellent blast-mitigating properties, easy casting, corrosion protection, abrasion resistance, and various uses in current military technology. Since pure polyurea has impedance higher than that of water (the current medium of interest), low mass density phenolic microballoon particles are added to create composite materials with reduced effective impedances. The volume fraction of particles is varied to study the effect of filler quantity on the acoustic impedance of the resulting composite. The composites are experimentally characterized via ultrasonic measurements. Computational models based on the method of dilute-randomly-distributed inclusions are developed and compared with the experimental results. These experiments and models will facilitate the design of new elastomeric composites with desirable acoustic impedances.
Flip-Chip Carrier Would Match Microwave FET Impedances
Huang, H. C.
1982-01-01
Proposed field-effect transistor consists of three cells which make up one complete FET pellet. Pellet is flip-chip mounted on carrier with source grounded gate and drain posts connected directly to impedance-matching transmission-line segments. Impedance transformers are part of mounting and contact strips.
Surface impedance in the anomalous skin effect regime
Chrzanowski, Janusz; Kirkiewicz, Józef
2008-12-01
An analytical solution of the surface impedance is obtained using the kinetic equation with the collision integral that takes into account the Fermi liquid effects. It is assumed that the reflection of electrons is purely diffusive. Particular attention is paid to the influence of external magnetic field and polarization of the incident wave on the real and imagine part of the surface impedance.
The Impedance Response of Semiconductors: An Electrochemical Engineering Perspective.
Orazem, Mark E.
1990-01-01
Shows that the principles learned in the study of mass transport, thermodynamics, and kinetics associated with electrochemical systems can be applied to the transport and reaction processes taking place within a semiconductor. Describes impedance techniques and provides several graphs illustrating impedance data for diverse circuit systems. (YP)
Development of impedance matching technologies for ICRF antenna arrays
International Nuclear Information System (INIS)
All high power ICRF heating systems include devices for matching the input impedance of the antenna array to the generator output impedance. For most types of antennas used, the input impedance is strongly time-dependent on timescales as rapid as 10-4 s, while the rf generators used are capable of producing full power only into a stationary load impedance. Hence, the dynamic response of the matching method is of great practical importance. In this paper, world-wide developments in this field over the past decade are reviewed. These techniques may be divided into several classes. The edge plasma parameters that determine the antenna array's input impedance may be controlled to maintain a fixed load impedance. The frequency of the rf source can be feedback controlled to compensate for changes in the edge plasma conditions, or fast variable tuning elements in the transmission line between the generator output and the antenna input connections can provide the necessary time-varying impedance transformation. In lossy passive schemes, reflected power due to the time-varying impedance of the antenna array is diverted to a dummy load. Each of these techniques can be applied to a pre-existing antenna system. If a new antenna is to be designed, recent advances allow the antenna array to have the intrinsic property of presenting a constant load to the feeding transmission lines despite the varying load seen by each antenna in the array
Effect of Feeding and Suction on Gastric Impedance Spectroscopy Measurements.
Beltran, Nohra E; Sánchez-Miranda, Gustavo; Sacristan, Emilio
2015-01-01
A specific device and system has been developed and tested for clinical monitoring of gastric mucosal reactance in the critically ill as an early warning of splanchnic hypoperfusion associated with shock and sepsis. This device has been proven effective in clinical trials and is expected to become commercially available next year. The system uses a combination nasogastric tube and impedance spectroscopy probe as a single catheter. Because this device has a double function, the question is: Does enteral feeding or suction affect the gastric reactance measurements? This study was designed to evaluate the effect of feeding and suction on the measurement of gastric impedance spectroscopy in healthy volunteers. Impedance spectra were obtained from the gastric wall epithelia of 18 subjects. The spectra were measured for each of the following conditions: postinsertion of gastric probe, during active suction, postactive suction, and during enteral feeding (236 ml of nutritional supplement). Impedance spectra were reproducible in all volunteers under all conditions tested. There was a slight increase in impedance parameters after suction, and a decrease in impedance after feeding; however, these observed differences were insignificant compared to patient-to-patient variability, and truly negligible compared with previously observed changes associated with splanchnic ischemia in critically ill patients. Our results demonstrate that suction or feeding when using the impedance spectro-metry probe/nasogastric tube does not significantly interfere with gastric impedance spectrometer measurements. PMID:26226020
Synthesis of adaptive impedance control for bipedal robot mechanisms
Directory of Open Access Journals (Sweden)
Petrović Milena
2008-01-01
Full Text Available The paper describes the impedance algorithm in locomotion of humanoid robot with proposed parameter modulation depending on the gate phase. The analysis shows influence of walking speed and foot elevation on regulator's parameters. Chosen criterion cares for footpath tracking and needed energy for that way of walking. The experiments give recommendation for impedance regulator tuning.
Reconstruction of a potential from the impedance boundary map
Isaev, Mikhail
2012-01-01
We give formulas and equations for finding generalized scattering data for the Schr\\"odinger equation in open bounded domain at fixed energy from the impedance boundary map (or Robin-to-Robin map). Combining these results with results of the inverse scattering theory we obtain efficient methods for reconstructing potential from the impedance boundary map.
Development on electromagnetic impedance function modeling and its estimation
Sutarno, D.
2015-09-01
Today the Electromagnetic methods such as magnetotellurics (MT) and controlled sources audio MT (CSAMT) is used in a broad variety of applications. Its usefulness in poor seismic areas and its negligible environmental impact are integral parts of effective exploration at minimum cost. As exploration was forced into more difficult areas, the importance of MT and CSAMT, in conjunction with other techniques, has tended to grow continuously. However, there are obviously important and difficult problems remaining to be solved concerning our ability to collect process and interpret MT as well as CSAMT in complex 3D structural environments. This talk aim at reviewing and discussing the recent development on MT as well as CSAMT impedance functions modeling, and also some improvements on estimation procedures for the corresponding impedance functions. In MT impedance modeling, research efforts focus on developing numerical method for computing the impedance functions of three dimensionally (3-D) earth resistivity models. On that reason, 3-D finite elements numerical modeling for the impedances is developed based on edge element method. Whereas, in the CSAMT case, the efforts were focused to accomplish the non-plane wave problem in the corresponding impedance functions. Concerning estimation of MT and CSAMT impedance functions, researches were focused on improving quality of the estimates. On that objective, non-linear regression approach based on the robust M-estimators and the Hilbert transform operating on the causal transfer functions, were used to dealing with outliers (abnormal data) which are frequently superimposed on a normal ambient MT as well as CSAMT noise fields. As validated, the proposed MT impedance modeling method gives acceptable results for standard three dimensional resistivity models. Whilst, the full solution based modeling that accommodate the non-plane wave effect for CSAMT impedances is applied for all measurement zones, including near-, transition
Development on electromagnetic impedance function modeling and its estimation
International Nuclear Information System (INIS)
Today the Electromagnetic methods such as magnetotellurics (MT) and controlled sources audio MT (CSAMT) is used in a broad variety of applications. Its usefulness in poor seismic areas and its negligible environmental impact are integral parts of effective exploration at minimum cost. As exploration was forced into more difficult areas, the importance of MT and CSAMT, in conjunction with other techniques, has tended to grow continuously. However, there are obviously important and difficult problems remaining to be solved concerning our ability to collect process and interpret MT as well as CSAMT in complex 3D structural environments. This talk aim at reviewing and discussing the recent development on MT as well as CSAMT impedance functions modeling, and also some improvements on estimation procedures for the corresponding impedance functions. In MT impedance modeling, research efforts focus on developing numerical method for computing the impedance functions of three dimensionally (3-D) earth resistivity models. On that reason, 3-D finite elements numerical modeling for the impedances is developed based on edge element method. Whereas, in the CSAMT case, the efforts were focused to accomplish the non-plane wave problem in the corresponding impedance functions. Concerning estimation of MT and CSAMT impedance functions, researches were focused on improving quality of the estimates. On that objective, non-linear regression approach based on the robust M-estimators and the Hilbert transform operating on the causal transfer functions, were used to dealing with outliers (abnormal data) which are frequently superimposed on a normal ambient MT as well as CSAMT noise fields. As validated, the proposed MT impedance modeling method gives acceptable results for standard three dimensional resistivity models. Whilst, the full solution based modeling that accommodate the non-plane wave effect for CSAMT impedances is applied for all measurement zones, including near-, transition
Development on electromagnetic impedance function modeling and its estimation
Energy Technology Data Exchange (ETDEWEB)
Sutarno, D., E-mail: Sutarno@fi.itb.ac.id [Earth Physics and Complex System Division Faculty of Mathematics and Natural Sciences Institut Teknologi Bandung (Indonesia)
2015-09-30
Today the Electromagnetic methods such as magnetotellurics (MT) and controlled sources audio MT (CSAMT) is used in a broad variety of applications. Its usefulness in poor seismic areas and its negligible environmental impact are integral parts of effective exploration at minimum cost. As exploration was forced into more difficult areas, the importance of MT and CSAMT, in conjunction with other techniques, has tended to grow continuously. However, there are obviously important and difficult problems remaining to be solved concerning our ability to collect process and interpret MT as well as CSAMT in complex 3D structural environments. This talk aim at reviewing and discussing the recent development on MT as well as CSAMT impedance functions modeling, and also some improvements on estimation procedures for the corresponding impedance functions. In MT impedance modeling, research efforts focus on developing numerical method for computing the impedance functions of three dimensionally (3-D) earth resistivity models. On that reason, 3-D finite elements numerical modeling for the impedances is developed based on edge element method. Whereas, in the CSAMT case, the efforts were focused to accomplish the non-plane wave problem in the corresponding impedance functions. Concerning estimation of MT and CSAMT impedance functions, researches were focused on improving quality of the estimates. On that objective, non-linear regression approach based on the robust M-estimators and the Hilbert transform operating on the causal transfer functions, were used to dealing with outliers (abnormal data) which are frequently superimposed on a normal ambient MT as well as CSAMT noise fields. As validated, the proposed MT impedance modeling method gives acceptable results for standard three dimensional resistivity models. Whilst, the full solution based modeling that accommodate the non-plane wave effect for CSAMT impedances is applied for all measurement zones, including near-, transition
Finite difference time domain implementation of surface impedance boundary conditions
Beggs, John H.; Luebbers, Raymond J.; Yee, Kane S.; Kunz, Karl S.
1991-01-01
Surface impedance boundary conditions are employed to reduce the solution volume during the analysis of scattering from lossy dielectric objects. In the finite difference solution, they also can be utilized to avoid using small cells, made necessary by shorter wavelengths in conducting media throughout the solution volume. The standard approach is to approximate the surface impedance over a very small bandwidth by its value at the center frequency, and then use that result in the boundary condition. Here, two implementations of the surface impedance boundary condition are presented. One implementation is a constant surface impedance boundary condition and the other is a dispersive surface impedance boundary condition that is applicable over a very large frequency bandwidth and over a large range of conductivities. Frequency domain results are presented in one dimension for two conductivity values and are compared with exact results. Scattering width results from an infinite square cylinder are presented as a two dimensional demonstration. Extensions to three dimensions should be straightforward.
Impedance Coordinative Control for Cascaded Converter in Bidirectional Application
DEFF Research Database (Denmark)
Tian, Yanjun; Deng, Fujin; Chen, Zhe;
2015-01-01
Two stage cascaded converters are widely used in bidirectional applications, but the negative impedance may cause system instability. Actually the impedance interaction is much different between forward power flow and reversed power flow, which will introduce more uncertainty to the system...... difference between forward and reversed power flow. This paper addresses the analysis with the topology of cascaded dual-active-bridge converter (DAB) with inverter, and the proposed control method can also be implemented in unidirectional applications and other general cascaded converter system. The...... stability. This paper proposes a control method for the constant power controlled converter in cascaded system, and consequently it can change the negative impedance of constant power converter into resistive impedance, which will improve the cascaded system stability, as well as merge the impedance...
Institute of Scientific and Technical Information of China (English)
Jean-Luc Fellahi; Vincent Caille; Cyril Charron; Pierre-Hervé Deschamps-Berger; Antoine Vieillard-Baron; 黄建廷
2011-01-01
背景 胸阻抗心动图(ICG),一直被认为是一种无创性、连续性、可独立操作且经济有效的方法,用于监测心输出量.本研究比较了在静息状态下改变健康志愿者血流动力学参数时,胸阻抗心动图(Niccomo装置)与经胸超声多普勒心动图对心脏指数(CI)的测量情况.方法 本研究共纳入了25例健康志愿者(7例男性,18例女性;平均年龄36±6岁;体表面积1.75±O.17 m2),分别在3种实验条件进行测定:基础水平、呼气末正压通气(+10 cm H2O)、下半身加压(用医疗抗休克裤对腹部施加30 cm H2O的压力).结果 在所有测量中,胸阻抗心动图的信号质量>89%.胸阻抗心动图与多普勒心动图测定的心脏指数(CITTE和CICG)间存在着较弱但有统计学意义的相关性(r=0.36;P=0.002).2种技术测得的心脏指数的一致性数值为0.94 L·min-1·m-2(95%可信区间0.77～1.11);一致性可变范围为-0.47～2.35 L·min-1·m-2,误差百分率为53%.施加呼气末正压+10 cm H2O(r=0.21;P=0.31)或医疗抗休克裤(r=0.22;P=0.30)后,CITTE和CIICG的变化率之间的相关性没有统计学意义.结论 采用胸阻抗心动图与多普勒心动图测量健康志愿者静息状态下的CI,两者测量所得CI的绝对值相关性较差,缺乏一致性.在血流动力学变化时,使用Niccomo装置测量心脏指数的变化也缺乏可靠性.
Journal and Wave Bearing Impedance Calculation Software
Hanford, Amanda; Campbell, Robert
2012-01-01
The wave bearing software suite is a MALTA application that computes bearing properties for user-specified wave bearing conditions, as well as plain journal bearings. Wave bearings are fluid film journal bearings with multi-lobed wave patterns around the circumference of the bearing surface. In this software suite, the dynamic coefficients are outputted in a way for easy implementation in a finite element model used in rotor dynamics analysis. The software has a graphical user interface (GUI) for inputting bearing geometry parameters, and uses MATLAB s structure interface for ease of interpreting data. This innovation was developed to provide the stiffness and damping components of wave bearing impedances. The computational method for computing bearing coefficients was originally designed for plain journal bearings and tilting pad bearings. Modifications to include a wave bearing profile consisted of changing the film thickness profile given by an equation, and writing an algorithm to locate the integration limits for each fluid region. Careful consideration was needed to implement the correct integration limits while computing the dynamic coefficients, depending on the form of the input/output variables specified in the algorithm.
Electrochemical impedance spectroscopy of oxidized porous silicon
International Nuclear Information System (INIS)
We present a study of the electrochemical oxidation process of porous silicon. We analyze the effect of the layer thickness (1.25–22 μm) and of the applied current density (1.1–11.1 mA/cm2, values calculated with reference to the external samples surface) on the oxidation process by comparing the galvanostatic electrochemical impedance spectroscopy (EIS) measurements and the optical specular reflectivity of the samples. The results of EIS were interpreted using an equivalent circuit to separate the contribution of different sample parts. A different behavior of the electrochemical oxidation process has been found for thin and thick samples: whereas for thin samples the oxidation process is univocally related to current density and thickness, for thicker samples this is no more true. Measurements by Energy Dispersive Spectroscopy using a Scanning Electron Microscopy confirmed that the inhomogeneity of the electrochemical oxidation process is increased by higher thicknesses and higher currents. A possible explanation is proposed to justify the different behavior of thin and thick samples during the electrochemical process. - Highlights: • A multidisciplinary approach on porous Si electrochemical oxidation is proposed. • Electrochemical, optical, and structural characterizations are used. • Layer thickness and oxidation current effects are shown. • An explanation of the observed behavior is proposed
Electrochemical impedance spectroscopy of oxidized porous silicon
Energy Technology Data Exchange (ETDEWEB)
Mula, Guido, E-mail: guido.mula@unica.it [Dipartimento di Fisica, Università degli Studi di Cagliari, Cittadella Universitaria di Monserrato, S.P. 8 km 0.700, 09042 Cagliari (Italy); Tiddia, Maria V. [Dipartimento di Fisica, Università degli Studi di Cagliari, Cittadella Universitaria di Monserrato, S.P. 8 km 0.700, 09042 Cagliari (Italy); Ruffilli, Roberta [Nanochemistry, Istituto Italiano di Tecnologia, Via Morego 30, 16163 Genova (Italy); Falqui, Andrea [Nanochemistry, Istituto Italiano di Tecnologia, Via Morego 30, 16163 Genova (Italy); Dipartimento di Scienze Chimiche e Geologiche, Università degli Studi di Cagliari, Cittadella Universitaria di Monserrato, S.P. 8 km 0.700, 09042 Cagliari (Italy); Palmas, Simonetta; Mascia, Michele [Dipartimento di Ingegneria Meccanica Chimica e dei Materiali, Università degli Studi di Cagliari, Piazza d' Armi, 09126 Cagliari (Italy)
2014-04-01
We present a study of the electrochemical oxidation process of porous silicon. We analyze the effect of the layer thickness (1.25–22 μm) and of the applied current density (1.1–11.1 mA/cm{sup 2}, values calculated with reference to the external samples surface) on the oxidation process by comparing the galvanostatic electrochemical impedance spectroscopy (EIS) measurements and the optical specular reflectivity of the samples. The results of EIS were interpreted using an equivalent circuit to separate the contribution of different sample parts. A different behavior of the electrochemical oxidation process has been found for thin and thick samples: whereas for thin samples the oxidation process is univocally related to current density and thickness, for thicker samples this is no more true. Measurements by Energy Dispersive Spectroscopy using a Scanning Electron Microscopy confirmed that the inhomogeneity of the electrochemical oxidation process is increased by higher thicknesses and higher currents. A possible explanation is proposed to justify the different behavior of thin and thick samples during the electrochemical process. - Highlights: • A multidisciplinary approach on porous Si electrochemical oxidation is proposed. • Electrochemical, optical, and structural characterizations are used. • Layer thickness and oxidation current effects are shown. • An explanation of the observed behavior is proposed.
DEFF Research Database (Denmark)
Nielsen, Jimmi; Hjelm, Johan
2014-01-01
It was shown through a comprehensive impedance spectroscopy study that the impedance of the classic composite LSM:YSZ (lanthanum strontium manganite and yttria stabilized zirconia) solid oxide fuel cell (SOFC) cathode can be described well with porous electrode theory. Furthermore, it was illustr...... electrode theory is the most suitable framework for any type of porous composite SOFC electrode evaluation.......It was shown through a comprehensive impedance spectroscopy study that the impedance of the classic composite LSM:YSZ (lanthanum strontium manganite and yttria stabilized zirconia) solid oxide fuel cell (SOFC) cathode can be described well with porous electrode theory. Furthermore, it was...... illustrated through a literature review on SOFC electrodes that porous electrode theory not only describes the classic LSM:YSZ SOFC cathode well, but SOFC electrodes in general. The extensive impedance spectroscopy study of LSM:YSZ cathodes consisted of measurements on cathodes with three different sintering...
Impedance spectroscopy for the detection and identification of unknown toxins
Riggs, B. C.; Plopper, G. E.; Paluh, J. L.; Phamduy, T. B.; Corr, D. T.; Chrisey, D. B.
2012-06-01
Advancements in biological and chemical warfare has allowed for the creation of novel toxins necessitating a universal, real-time sensor. We have used a function-based biosensor employing impedance spectroscopy using a low current density AC signal over a range of frequencies (62.5 Hz-64 kHz) to measure the electrical impedance of a confluent epithelial cell monolayer at 120 sec intervals. Madin Darby canine kidney (MDCK) epithelial cells were grown to confluence on thin film interdigitated gold electrodes. A stable impedance measurement of 2200 Ω was found after 24 hrs of growth. After exposure to cytotoxins anthrax lethal toxin and etoposide, the impedance decreased in a linear fashion resulting in a 50% drop in impedance over 50hrs showing significant difference from the control sample (~20% decrease). Immunofluorescent imaging showed that apoptosis was induced through the addition of toxins. Similarities of the impedance signal shows that the mechanism of cellular death was the same between ALT and etoposide. A revised equivalent circuit model was employed in order to quantify morphological changes in the cell monolayer such as tight junction integrity and cell surface area coverage. This model showed a faster response to cytotoxin (2 hrs) compared to raw measurements (20 hrs). We demonstrate that herein that impedance spectroscopy of epithelial monolayers serves as a real-time non-destructive sensor for unknown pathogens.
Measuring the Acoustic Impedance of Pipes and Musical Instruments
Jaeger, Herbert
2007-05-01
Using a small electret microphone and a piezo-buzzer we have constructed a simple impedance transducer to measure the input impedance of air columns, such as cylindrical pipes, as well as musical instruments. The input impedance of an air column is given as the ratio of the pressure to the volume flow of air at the input of the air column. The microphone serves as the pressure transducer, while the piezo-buzzer is controlled to provide a constant velocity amplitude. Therefore the microphone signal is proportional to the acoustical impedance and, if required, can be calibrated using a simple air column for which the impedance can be calculated. This impedance transducer is currently in use as demonstration equipment for a physical acoustics class. It is simple to use and robust, so that it is well-suited for an undergraduate introductory laboratory environment. This talk will discuss the function of the impedance transducer and show examples of the type of measurements that can be performed. To cite this abstract, use the following reference: http://meetings.aps.org/link/BAPS.2007.OSS07.C1.1
Characterizing aging effects of lithium ion batteries by impedance spectroscopy
International Nuclear Information System (INIS)
Impedance spectroscopy is one of the most promising methods for characterizing aging effects of portable secondary batteries online because it provides information about different aging mechanisms. However, application of impedance spectroscopy 'in the field' has some higher requirements than for laboratory experiments. It requires a fast impedance measurement process, an accurate model applicable with several batteries and a robust method for model parameter estimation. In this paper, we present a method measuring impedance at different frequencies simultaneously. We propose to use a composite electrode model, capable to describe porous composite electrode materials. A hybrid method for parameter estimation based on a combination of evolution strategy and Levenberg-Marquardt method allowed a robust and fast parameter calculation. Based on this approach, an experimental investigation of aging effects of a lithium ion battery was carried out. After 230 discharge/charge cycles, the battery showed a 14% decreased capacity. Modeling results show that series resistance, charge transfer resistance and Warburg coefficient changed thereby their values by approximately 60%. A single frequency impedance measurement, usually carried out at 1 kHz, delivers only information about series resistance. Impedance spectroscopy allows additionally the estimation of charge transfer resistance and Warburg coefficient. This fact and the high sensitivity of model parameters to capacity change prove that impedance spectroscopy together with an accurate modeling deliver information that significantly improve characterization of aging effects
A comparison study of electrodes for neonate electrical impedance tomography
International Nuclear Information System (INIS)
Electrical impedance tomography (EIT) is an imaging technique that has the potential to be used for studying neonate lung function. The properties of the electrodes are very important in multi-frequency EIT (MFEIT) systems, particularly for neonates, as the skin cannot be abraded to reduce contact impedance. In this work, the impedance of various clinical electrodes as a function of frequency is investigated to identify the optimum electrode type for this application. Six different types of self-adhesive electrodes commonly used in general and neonatal cardiology have been investigated. These electrodes are Ag/AgCl electrodes from the Ambu® Cardiology Blue sensors range (BR, NF and BRS), Kendall (KittyCat(TM) and ARBO®) and Philips 13953D electrodes. In addition, a textile electrode without gel from Textronics was tested on two subjects to allow comparison with the hydrogel-based electrodes. Two- and four-electrode measurements were made to determine the electrode-interface and tissue impedances, respectively. The measurements were made on the back of the forearm of six healthy adult volunteers without skin preparation with 2.5 cm electrode spacing. Impedance measurements were carried out using a Solartron SI 1260 impedance/gain-phase analyser with a frequency range from 10 Hz to 1 MHz. For the electrode-interface impedance, the average magnitude decreased with frequency, with an average value of 5 kΩ at 10 kHz and 337 Ω at 1 MHz; for the tissue impedance, the respective values were 987 Ω and 29 Ω. Overall, the Ambu BRS, Kendall ARBO® and Textronics textile electrodes gave the lowest electrode contact impedance at 1 MHz. Based on the results of the two-electrode measurements, simple RC models for the Ambu BRS and Kendall-ARBO and Textronics textile electrodes have been derived for MFEIT applications
Measurements of the SPS transverse impedance in 2000
Arduini, Gianluigi; Cornelis, Karel; Klem, J T; Zimmermann, Frank; Zorzano-Mier, M P
2001-01-01
We report on measurements of coherent tune shifts, head-tail growth rates, and current-dependent betatron phase advances at the CERN SPS in the year 2000. Comparing results obtained at two different energies shows that there is no notable contribution from space charge. Within the measurement resolution the impedance is the same as in 1999, consistent with the expected small effect from changes to ony a small number of pumping ports. In 2000, data were taken over an expanded range of chromaticities, which increases the sensitivity to the impedance frequency distribution. Measuremeents of the current-dependent phase advance around the ring help localizing the most important impedance sources.
Literature Review on Recent Magnetocardiography and Impedance-Magnetocardiography Technologies
Energy Technology Data Exchange (ETDEWEB)
Kandori, A.; Miyashita, T.; Ogata, K. [Advaned Research Laboratory, Hitachi, Ltd, Mito (Japan); and others
2006-10-15
We have developed magnetocardiography(MCG) and impedance magnetocardiography(I-MCG) for detecting heart disease by using dc-SQUID technology. The MCG system, using low-Tc SQUID, is being applied commercially for diagnosing heart disease. Using the low-Tc MCG system, many clinical studies on detection of abnormality have been performed. Furthermore, we have developed a portable MCG system using high-Tc SQUID. For detecting changes in kinetic impedance in the heart, an I-MCG system has been demonstrated. The I-MCG system could detect the mechanical movement of the heart. In this report, we review current clinical applications of magnetocardiography and impedance magnetocardiography.
MEASURED TRANSVERSE COUPLING IMPEDANCE OF RHIC INJECTION AND ABORT KICKERS
International Nuclear Information System (INIS)
Concerns regarding possible transverse instabilities in RHIC and the SNS pointed to the need for measurements of the transverse coupling impedance of ring components. The impedance of the RHIC injection and abort kicker was measured using the conventional method based on the S21 forward transmission coefficient. A commercial 450 Ω twin-wire Lecher line were used and the data was interpreted via the log-formula. All measurements, were performed in test stands fully representing operational conditions including pulsed power supplies and connecting cables. The measured values for the transverse coupling impedance in kick direction and perpendicular to it are comparable in magnitude, but differ from Handbook predictions
PHEMT Distributed Power Amplifier Adopting Broadband Impedance Transformer
DEFF Research Database (Denmark)
Narendra, K.; Limiti, E.; Paoloni, C.; Zhurbenko, Vitaliy
2013-01-01
A non-uniform drain line distributed power amplifier (DPA) employing a broadband impedance transformer is presented. The DPA is based on GaAs PHEMT technology. The impedance transformer employs asymmetric coupled lines and transforms a low output impedance of the amplifier to a standard 50 Ω...... transmission line. The output power of approximately 600 mW, with an associated gain of 9 dB and PAE greater than 30 percent, is demonstrated in the frequency range from 10 to 1800 MHz....
Impedance-type measurements using XPS
International Nuclear Information System (INIS)
An impedance type of measurement using X-ray photoelectron spectroscopy is applied for probing charging/discharging dynamics of a sample containing Rb deposited on a SiO2/Si substrate containing an octadecyltrichlorosilane (OTS) bilayer coating. The OTS bilayer coatings have possible use as anti-relaxation wall coatings for alkali atom vapor cells in miniaturized instruments such as chip-scale atomic clocks, and/or magnetometers. The measurement consists of the application of bipolar square wave pulses of ±10.0 V amplitude to the sample with varying frequencies in the range of 10-2 to 102 Hz while recording X-ray photoemission data. For a conducting sample this type of measurement twins all the photoemission peaks at -10.0 and +10.0 eV positions at all frequencies with exactly 20.0 eV difference between them. However, for samples amenable to charging, the difference between the twinned peaks is less than 20.0 eV, and gradually decreases at correspondingly lower frequencies. For the sample under consideration here at 0.01 Hz, the twinned O1s and Si2p peaks, representing the SiO2 substrate, are separated by 18.2 eV, displaying a 1.8 eV charging shift. These positions differ from those of the C1s (18.0 eV) representing the OTS bilayer and the Rb3d peaks (18.1 eV). These results reveal that the Rb is electrically (perhaps also chemically) isolated from the OTS bilayer, which may be correlated with the improved performance of the OTS bilayers as anti-relaxation coatings in these alkali atom magnetometer cells.
Volume dependence of respiratory impedance in infants.
Peták, F; Hayden, M J; Hantos, Z; Sly, P D
1997-10-01
We previously studied low-frequency respiratory impedance (Zrs) data at an elevated lung volume to separate airway and tissue mechanical properties in normal infants (Am. I. Respir. Crit. Care Med. 1996; 154:161-166). The aim of the present study was to determine the volume dependence of the airway and tissue mechanics by extending Zrs measurements to lower lung volumes. Zrs spectra between 0.5 and 21 Hz were measured in supine sleeping infants (n = 8; 7 to 26 mo of age) at mean transrespiratory pressures (Ptr[mean]) of 20, 10, and 0 cm H2O, during periods of apnea induced by inflating the infants' lungs to a pressure of 20 cm H2O through a face mask. At each inflation pressure, a model containing airway resistance (Raw) and inertance (law) and tissue damping (G) and elastance (H) was fitted to Zrs data. At FRC, the values of Raw, law, G, and H were 20.6+/-4.9 (SD) cm H2O x s/L, 0.037+/-0.014 cm H2O x s2/L, 39.6+/-10.3 cm H2O/L, and 147+/-35 cm H2O/L, respectively. Increase of Ptr(mean) caused a monotonous decrease in Raw (42+/-7% of the value at FRC), while law remained constant. The tissue parameters were minimal at a Ptr(mean) of 10 cm H2O (68+/-10% and 78+/-6% in G and H, respectively) and significantly higher at both 0 and 20 cm H2O. Although Zrs measurements can be made in most infants at lung volumes as low as FRC, an inflation pressure of 20 cm H2O provides a higher success rate and is therefore a more suitable condition for general use. PMID:9351618
Qin, F. X.; Peng, H. X.; Popov, V. V.; Phan, M. H.
2011-02-01
Composites consisting of glass-coated amorphous microwire Co 68.59Fe 4.84Si 12.41B 14.16 and 913 E-glass prepregs were designed and fabricated. The influences of tensile stress, annealing and number of composite layers on the giant magneto-impedance (GMI) and giant stress-impedance (GSI) effects in these composites were investigated systematically. It was found that the application of tensile stress along the microwire axis or an increase in the number of composite layers reduced the GMI effect and increased the circular anisotropy field, while the annealing treatment had a reverse effect. The value of matrix-wire interfacial stress calculated via the GMI profiles coincided with the value of the applied effective tensile stress to yield similar GMI profiles. Enhancement of the GSI effect was achieved in the composites relative to their single microwire inclusion. These findings are important for the development of functional microwire-based composites for magnetic- and stress-sensing applications. They also open up a new route for probing the interfacial stress in fibre-reinforced polymer (FRP) composites.
EXPERIMENTAL RESEARCH ON EVALUATING STRUCTURE DAMAGE WITH PIEZOELECTRIC DYNAMIC IMPEDANCE
Institute of Scientific and Technical Information of China (English)
无
2002-01-01
A dynamic impedance-based structural health monitoring technique is introduced. According to the direct and the converse piezoelectric property of piezoelectric materials, the piezoceramic ( PZT ) can be used as an actuator and a sensor synchronously. If damages like cracks, holes, debonding or loose connections are presented in the structure, the physical variations of the structure will cause the mechanical impedance modified. On the basis of introducing the principle and the theory, the experiment and the analysis on some damages of the structure are studied by means of the dynamic impedance technique. On the view of experiment, kinds of structural damages are evaluated by the information of dynamic impedance in order to validate the feasibility of the method.
Bioelectrical impedance analysis for bovine milk: Preliminary results
Bertemes-Filho, P.; Valicheski, R.; Pereira, R. M.; Paterno, A. S.
2010-04-01
This work reports the investigation and analysis of bovine milk quality by using biological impedance measurements using electrical impedance spectroscopy (EIS). The samples were distinguished by a first chemical analysis using Fourier transform midinfrared spectroscopy (FTIR) and flow citometry. A set of milk samples (100ml each) obtained from 17 different cows in lactation with and without mastitis were analyzed with the proposed technique using EIS. The samples were adulterated by adding distilled water and hydrogen peroxide in a controlled manner. FTIR spectroscopy and flow cytometry were performed, and impedance measurements were made in a frequency range from 500Hz up to 1MHz with an implemented EIS system. The system's phase shift was compensated by measuring saline solutions. It was possible to show that the results obtained with the Bioelectrical Impedance Analysis (BIA) technique may detect changes in the milk caused by mastitis and the presence of water and hydrogen peroxide in the bovine milk.
Studying of acid-gas pipelines corrosion with impedance spectroscopy
Energy Technology Data Exchange (ETDEWEB)
Neshati, J.; Fardi, M.R.; Ghassem, H. [Corrosion Department, NIOC-RIPI, Pazhooheshgah Bulevard, Khairabad Junction, Old Qom Road Tehran (Iran)
2004-07-01
In this research, the acid-gas pipelines of a gas refinery were simulated in laboratory. Acid gas is normally the feed of sulfur recovery plant (SRP) in a gas refinery. For studying corrosion kinetic and related mechanisms the impedance spectroscopy was used. Impedance diagrams were simulated by Boukamp1988 software. It was found that the simulated systems can be equated to a circuit with two time constants. For studying corrosion rate changes a type of inhibitor was utilized. The inhibitor used in this work was an imidazoline, an appropriate based inhibitor formulated with the commercial grade imidazoline and dimmer - trimer acid. It was shown that impedance spectroscopy technique can be used for corrosion monitoring of acid gas pipelines in gas refineries. The impedance spectroscopy will be tried in due course as a suitable technique in field for corrosion control of acid-gas pipelines. (authors)
Iterative Reconstruction Methods for Hybrid Inverse Problems in Impedance Tomography
DEFF Research Database (Denmark)
Hoffmann, Kristoffer; Knudsen, Kim
2014-01-01
For a general formulation of hybrid inverse problems in impedance tomography the Picard and Newton iterative schemes are adapted and four iterative reconstruction algorithms are developed. The general problem formulation includes several existing hybrid imaging modalities such as current density...
Impedance of rigid bodies in one-dimensional elastic collisions
Santos, Janilo; Nelson, Osman Rosso
2012-01-01
In this work we study the problem of one-dimensional elastic collisions of billiard balls, considered as rigid bodies, in a framework very different from the classical one presented in text books. Implementing the notion of impedance matching as a way to understand eficiency of energy transmission in elastic collisions, we find a solution which frames the problem in terms of this conception. We show that the mass of the ball can be seen as a measure of its impedance and verify that the problem of maximum energy transfer in elastic collisions can be thought of as a problem of impedance matching between different media. This approach extends the concept of impedance, usually associated with oscillatory systems, to system of rigid bodies.
Analysis of impedance characteristics of IMPATT-generators
Directory of Open Access Journals (Sweden)
F. B. Bereziuk
2007-10-01
Full Text Available Model of power solid-state oscillators in a millimeter wave band is presented. This model based on non-stationary impedance characteristics of pulse oscillating impact avalanche transit time diodes.
Low profile conformal antenna arrays on high impedance substrate
Singh, Hema; Jha, Rakesh Mohan
2016-01-01
This book presents electromagnetic (EM) design and analysis of dipole antenna array over high impedance substrate (HIS). HIS is a preferred substrate for low-profile antenna design, owing to its unique boundary conditions. Such substrates permit radiating elements to be printed on them without any disturbance in the radiation characteristics. Moreover HIS provides improved impedance matching, enhanced bandwidth, and increased broadside directivity owing to total reflection from the reactive surface and high input impedance. This book considers different configurations of HIS for array design on planar and non-planar high-impedance surfaces. Results are presented for cylindrical dipole, printed dipole, and folded dipole over single- and double-layered square-patch-based HIS and dogbone-based HIS. The performance of antenna arrays is analyzed in terms of performance parameters such as return loss and radiation pattern. The design presented shows acceptable return loss and mainlobe gain of radiation pattern. Thi...
Motion discrimination of throwing a baseball using forearm electrical impedance
International Nuclear Information System (INIS)
The extroversion or hyperextension of elbow joint cause disorders of elbow joint in throwing a baseball. A method, which is easy handling and to measure motion objectively, can be useful for evaluation of throwing motion. We investigated a possibility of motion discrimination of throwing a baseball using electrical impedance method. The parameters of frequency characteristics (Cole-Cole arc) of forearm electrical impedance were measured during four types of throwing a baseball. Multiple discriminant analysis was used and the independent variables were change ratios of 11 parameters of forearm electrical impedance. As results of 120 data with four types of throwing motion in three subjects, hitting ratio was very high and 95.8%. We can expect to discriminate throwing a baseball using multiple discriminant analysis of impedance parameters.
Motion discrimination of throwing a baseball using forearm electrical impedance
Nakamura, Takao; Kusuhara, Toshimasa; Yamamoto, Yoshitake
2013-04-01
The extroversion or hyperextension of elbow joint cause disorders of elbow joint in throwing a baseball. A method, which is easy handling and to measure motion objectively, can be useful for evaluation of throwing motion. We investigated a possibility of motion discrimination of throwing a baseball using electrical impedance method. The parameters of frequency characteristics (Cole-Cole arc) of forearm electrical impedance were measured during four types of throwing a baseball. Multiple discriminant analysis was used and the independent variables were change ratios of 11 parameters of forearm electrical impedance. As results of 120 data with four types of throwing motion in three subjects, hitting ratio was very high and 95.8%. We can expect to discriminate throwing a baseball using multiple discriminant analysis of impedance parameters.
Optimization design of power efficiency of exponential impedance transformer
International Nuclear Information System (INIS)
The paper investigates the optimization design of power efficiency of exponential impedance transformer with analytic method and numerical method. In numerical calculation, a sine wave Jantage with hypothesis of rising edge equivalence is regarded as the forward-going Jantage at input of transformer, and its dominant angular frequency is determined by typical rise-time of actual Jantage waveforms. At the same time, dissipative loss in water dielectric is neglected. The numerical results of three typical modes of impedance transformation, viz. linear mode, saturation mode and steep mode,are compared. Pivotal factors which affect the power efficiency of exponential impedance transformer are discussed, and a certain extent quantitative range of intermediate variables and accordance coefficients are obtained. Finally, the paper discusses some important issues in actual design, such as insulation safety factor in structure design, effects of coupling capacitance on impedance calculation, and dissipative loss in water dielectric. (authors)
Digital synthetic impedance for application in vibration damping
Nečásek, J.; Václavík, J.; Marton, P.
2016-02-01
In this work we present construction details of a precision, standalone, and compact digital synthetic impedance for application in the field of vibration damping. The presented device is based on an embedded ARM microcontroller with external AD and DA converters and a special analog front-end. The performance of the device is tested by comparing the actually synthesized impedance with several prescribed impedances and shows very good match. Fine-tuning ability of the device, which is crucial for the considered application, is also demonstrated and reaches as small step as 0.1% for the most complicated impedance structure and drops below the level of direct measurability with less complex structures. The real application in vibration damping is demonstrated on a simple and well understood case of a one-dimensional vibrating spring-mass system with piezoelectric actuator embedded as the interface between source of vibrations and vibrating mass.
A valveless micro impedance pump driven by electromagnetic actuation
Rinderknecht, Derek; Hickerson, Anna Iwaniec; Gharib, Morteza
2005-01-01
Over the past two decades, a variety of micropumps have been explored for various applications in microfluidics such as control of pico- and nanoliter flows for drug delivery as well as chemical mixing and analysis. We present the fabrication and preliminary experimental studies of flow performance on the micro impedance pump, a previously unexplored method of pumping fluid on the microscale. The micro impedance pump was constructed of a simple thin-walled tube coupled at either end to glass ...
MD 349: Impedance Localization with AC-dipole
Biancacci, Nicolo; Metral, Elias; Salvant, Benoit; Papotti, Giulia; Persson, Tobias Hakan Bjorn; Tomas Garcia, Rogelio; CERN. Geneva. ATS Department
2016-01-01
The purpose of this MD is to measure the distribution of the transverse impedance of the LHC by observing the phase advance variation with intensity between the machine BPMs. Four injected bunches with different intensities are excited with an AC dipole and the turn by turn data is acquired from the BPM system. Through post-processing analysis the phase variation along the machine is depicted and, from this information, first conclusions of the impedance distribution can be drawn.
Effects of Human Arm Impedance on Dynamics Learning and Generalization
Darainy, Mohammad; Mattar, Andrew A. G.; Ostry, David J.
2009-01-01
Previous studies have demonstrated anisotropic patterns of hand impedance under static conditions and during movement. Here we show that the pattern of kinematic error observed in studies of dynamics learning is associated with this anisotropic impedance pattern. We also show that the magnitude of kinematic error associated with this anisotropy dictates the amount of motor learning and, consequently, the extent to which dynamics learning generalizes. Subjects were trained to reach to visual t...
The wave impedance of an atomically thin crystal
Merano, Michele
2015-01-01
I propose an expression for the electromagnetic wave impedance of a two-dimensional atomic crystal, and I deduce the Fresnel coefficients in terms of this quantity. It is widely known that a two-dimensional crystal can absorb light, if its conductivity is different from zero. It is less emphasized that they can also store a certain amount of electromagnetic energy. The concept of impedance is useful to quantify this point.
Pore Characteristics of Chitosan Scaffolds Studied by Electrochemical Impedance Spectroscopy
Tully-Dartez, Stephanie; Cardenas, Henry E.; Sit, Ping-Fai Sidney
2009-01-01
In this study, a novel approach, electrochemical impedance spectroscopy (EIS), was used to examine the pore characteristics of chitosan scaffolds under aqueous conditions. The EIS was run with a constant current of 0.1 mA with the frequency sweep of 106 to 10−4 Hz. The resulting complex impedance measurement was then used to calculate porosity, which was determined to be 71%. Scanning electron microscopy (SEM) and mercury intrusion porosimetry (MIP), two commonly used methods for scaffold cha...
A Batteryless Sensor ASIC for Implantable Bio-impedance Applications
Rodriguez, Saul; Ollmar, Stig; Waqar, Muhammad; Rusu, Ana
2015-01-01
The measurement of the biological tissue's electrical impedance is an active research field that has attracted a lot of attention during the last decades. Bio-impedances are closely related to a large variety of physiological conditions; therefore, they are useful for diagnosis and monitoring in many medical applications. Measuring living tissues, however, is a challenging task that poses countless technical and practical problems, in particular if the tissues need to be measured under the sk...
Experiences with a two terminal-pair digital impedance bridge
Callegaro, Luca; D'Elia, Vincenzo; Kampik, Marian; Kim, Dan Bee; Ortolano, Massimo; Pourdanesh, Faranak
2014-01-01
This paper describes the realization of a two terminal-pair digital impedance bridge and the test measurements performed with it. The bridge, with a very simple architecture, is based on a commercial two-channel digital signal synthesizer and a synchronous detector. The bridge can perform comparisons between impedances having arbitrary phase and magnitude ratio: its balance is achieved automatically in less than a minute. $R$-$C$ comparisons with calibrated standards, at kHz frequency and 100...
An Analysis of the Acoustic Input Impedance of the Ear
Withnell, Robert H.; Gowdy, Lauren E.
2013-01-01
Ear canal acoustics was examined using a one-dimensional lossy transmission line with a distributed load impedance to model the ear. The acoustic input impedance of the ear was derived from sound pressure measurements in the ear canal of healthy human ears. A nonlinear least squares fit of the model to data generated estimates for ear canal radius, ear canal length, and quantified the resistance that would produce transmission losses. Derivation of ear canal radius has application to quantify...
Impedance of tissue-mimicking phantom material under compression
Barry Belmont; Dodde, Robert E.; Shih, Albert J.
2013-01-01
The bioimpedance of tissues under compression is a field in need of study. While biological tissues can become compressed in a myriad of ways, very few experiments have been conducted to describe the relationship between the passive electrical properties of a material (impedance/admittance) during mechanical deformation. Of the investigations that have been conducted, the exodus of fluid from samples under compression has been thought to be the cause of changes in impedance, though until now ...
Ultrasonic flow measurement and wall acoustic impedance effects.
Willatzen, M
2004-03-01
An examination of the influence of wall acoustic impedance effects on sound propagation in flowing liquids confined by cylindrical walls is presented. Special focus is given to the importance of the wall acoustic impedance value for ultrasonic flow meter performance. The mathematical model presented allows any radially-dependent axial flow profile to be examined in the linear flow acoustics regime where fluid flow speed is much smaller than the fluid sound speed everywhere in the fluid medium. PMID:14996531
A microscopy technique based on bio-impedance sensors
Yúfera, A.; Huertas, Gloria; Olmo, Alberto
2012-01-01
It is proposed a microscopy for cell culture applications based on impedance sensors. The imagined signals are measured with the Electrical Cell-Substrate Spectroscopy (ECIS) technique, by identifying the cell area. The proposed microscopy allows real-time monitoring inside the incubator, reducing the contamination risk by human manipulation. It requires specific circuits for impedance measurements, a two-dimensional sensor array (pixels), and employing electrical models to decode efficiently...
The coupling impedance of the RHIC injection kicker system
International Nuclear Information System (INIS)
In this paper, results from impedance measurements on the RHIC injection kickers are reported. The kicker is configured as a ''C'' cross section magnet with interleaved ferrite and high-permittivity dielectric sections to achieve a travelling wave structure. The impedance was measured using the wire method in which a resistive match provides a smooth transition from the network analyzer to the reference line in the set-up. Accurate results are obtained by interpreting the forward scattering coefficient via the log-formula. The four kickers with their ceramic beam tubes contribute a Z/n=0.22 Ω/ring in the interesting frequency range from 0.1 to 1 GHz, and less above. At frequencies above ∼ 100 MHz, the impedance is ferrite dominated and not affected by the kicker terminations. Below 100 MHz, the Blumlein pulser with the ∼75 m feeding cables is visible in the impedance but makes no significant contribution to the results. The measurements show that the kicker coupling impedance is tolerable without the need for impedance reducing measures
The coupling impedance of the RHIC injection kicker system
International Nuclear Information System (INIS)
In this paper, results from impedance measurements on the RHIC injection kickers are reported. The kicker is configured as a open-quotes Cclose quotes cross section magnet with interleaved ferrite and high-permittivity dielectric sections to achieve a travelling wave structure. The impedance was measured using the wire method in which a resistive match provides a smooth transition from the network analyzer to the reference line in the set-up. Accurate results are obtained by interpreting the forward scattering coefficient via the log-formula. The four kickers with their ceramic beam tubes contribute a Z/n = 0.22 Ω/ring in the interesting frequency range from 0.1 to 1 GHz, and less above. At frequencies above ∼100 MHZ, the impedance is ferrite dominated and not affected by the kicker terminations. Below 100 MHz, the Blumlein pulser with the ∼75 m feeding cables is visible in the impedance but makes no significant contribution to the results. The measurements show that the kicker coupling impedance is tolerable without the need for impedance reducing measures
Optical approximation in the theory of geometric impedance
International Nuclear Information System (INIS)
In this paper we introduce an optical approximation into the theory of impedance calculation, one valid in the limit of high frequencies. This approximation neglects diffraction effects in the radiation process, and is conceptually equivalent to the approximation of geometric optics in electromagnetic theory. Using this approximation, we derive equations for the longitudinal impedance for arbitrary offsets, with respect to a reference orbit, of source and test particles. With the help of the Panofsky-Wenzel theorem we also obtain expressions for the transverse impedance (also for arbitrary offsets). We further simplify these expressions for the case of the small offsets that are typical for practical applications. Our final expressions for the impedance, in the general case, involve two dimensional integrals over various cross-sections of the transition. We further demonstrate, for several known axisymmetric examples, how our method is applied to the calculation of impedances. Finally, we discuss the accuracy of the optical approximation and its relation to the diffraction regime in the theory of impedance. (orig.)
Positive impedance humidity sensors via single-component materials.
Qian, Jingwen; Peng, Zhijian; Shen, Zhenguang; Zhao, Zengying; Zhang, Guoliang; Fu, Xiuli
2016-01-01
Resistivity-type humidity sensors have been investigated with great interest due to the increasing demands in industry, agriculture and daily life. To date, most of the available humidity sensors have been fabricated based on negative humidity impedance, in which the electrical resistance decreases as the humidity increases, and only several carbon composites have been reported to present positive humidity impedance. However, here we fabricate positive impedance humidity sensors only via single-component WO3-x crystals. The resistance of WO3-x crystal sensors in response to relative humidity could be tuned from a negative to positive one by increasing the compositional x. And it was revealed that the positive humidity impedance was driven by the defects of oxygen vacancy. This result will extend the application field of humidity sensors, because the positive humidity impedance sensors would be more energy-efficient, easier to be miniaturized and electrically safer than their negative counterparts for their lower operation voltages. And we believe that constructing vacancies in semiconducting materials is a universal way to fabricate positive impedance humidity sensors. PMID:27150936
Development of impedance void meter for evaluation of flow symmetry
International Nuclear Information System (INIS)
Highlights: • New impedance void meter design to evaluate flow symmetry. • Area-averaged measurements agree well with previous impedance meter designs. • Quadrant measurements agree well with local conductivity probe measurements. - Abstract: Often it is useful or even necessary to know whether the void fraction is evenly distributed in two-phase flows. In industrial systems, unevenly distributed flows can be a factor in vibrations and other undesirable occurrences. In bubble column chemical reactors, uneven distribution of the void can result in reduced reaction efficiency. In experiments, it is often desirable to have symmetrically distributed void for local measurements since this allows for more rapid data collection. Current systems for determining the symmetry of the void distribution in distributed flows either suffer from very long measurement time requirements or cause significant disturbances to the flow. To address the need for approximate evaluation of flow symmetry without causing excessive disturbance to the flow, a new impedance void meter design has been developed and tested under a variety of conditions. The ability of the new design to evaluate the void distribution symmetry has been tested through comparison with conductivity probe measurements of local void fraction, and the accuracy of the impedance meter has been verified through comparison with the conductivity probe measurements and with typical impedance void meter measurements
Determination of soil ionic concentration using impedance spectroscopy
Pandey, Gunjan; Kumar, Ratnesh; Weber, Robert J.
2013-05-01
This paper presents a novel approach to estimate the soil ionic concentration by way of multi-frequency impedance measurements and using the quasi-static dielectric mixing models to infer the various ionic concentrations. In our approach, the permittivity of the soil dielectric mixture is measured using impedance spectroscopy and the results are used as input parameters to dielectric mixing models, combined with the debye-type dielectric relaxation models. We observe that the dielectric mixing models work well for low RF (radio-frequency) range and help in determining the individual ionic concentration in a multi-component soil mixture. Using the fact that the permittivity of a dielectric mixture is proportional to its impedance, we validated our approach by making multi-frequency impedance measurements of a soil mixture at different concentrations of various components. The method provides a good estimate of individual components such as air, water and ions like nitrates. While the paper is written with the perspective of soil constituent concentration determination, the underlying principle of determining individual component concentration using multi-frequency impedance measurement is applicable more generally in areas such as characterizing biological systems like pathogens, quality control of pharmaceuticals etc.
Wideband impedance spectrum analyzer for process automation applications
Doerner, Steffen; Schneider, Thomas; Hauptmann, Peter R.
2007-10-01
For decades impedance spectroscopy is used in technical laboratories and research departments to investigate effects or material characteristics that affect the impedance spectrum of the sensor. Establishing this analytical approach for process automation and stand-alone applications will deliver additional and valuable information beside traditional measurement techniques such as the measurement of temperature, flow rate, and conductivity, among others. As yet, most of the current impedance analysis methods are suited for laboratory applications only since they involve stand-alone network analyzers that are slow, expensive, large, or immobile. Furthermore, those systems offer a large range of functionality that is not being used in process control and other fields of application. We developed a sensor interface based on high speed direct digital signal processing offering wideband impedance spectrum analysis with high resolution for frequency adjustment, excellent noise rejection, very high measurement rate, and convenient data exchange to common interfaces. The electronics has been implemented on two small circuit boards and it is well suited for process control applications such as monitoring phase transitions, characterization of fluidal systems, and control of biological processes. The impedance spectrum analyzer can be customized easily for different measurement applications by adapting the appropriate sensor module. It has been tested for industrial applications, e.g., dielectric spectroscopy and high temperature gas analysis.
Ring impedance and stored current for the photon factory
International Nuclear Information System (INIS)
The impedance of the Photon Factory ring is computed using estimates for individual vacuum chamber component impedances, and computer results for the impedance of the RF cavities. The total single-bunch loss impedance is expected to be about 2.5 MΩ at a bunch length of 2.0 cm. This is lower than the SPEAR impedance (per unit length of ring circumference) by about a factor of 5. Thus, the threshold current for single bunch instabilities which limit the beam current will probably be on the order of 150 - 200 mA. There should be no problem in reaching a stored current of 500 mA with 312 bunches. RF and beam parameters, such as stored current, klystron power, synchrotron radiation power, higher mode power, cavity power and reflected power are computed as a function of energy for two operating regions: at a constant beam current of 500 mA for lower energies where a klystron power of less than 650 kW is required, and at a constant klystron power of 650 kW at higher energies. Results are given for operation with and without a wiggler, and for both the single-bunch and 312-bunch modes. (author)
Application of Vertical Electrodes in Microfluidic Channels for Impedance Analysis
Directory of Open Access Journals (Sweden)
Qiang Li
2016-05-01
Full Text Available This paper presents a microfluidic device with electroplated vertical electrodes in the side walls for impedance measurement. Based on the proposed device, the impedance of NaCl solutions with different concentrations and polystyrene microspheres with different sizes was measured and analyzed. The electroplating and SU-8-PDMS (SU-8-poly(dimethylsiloxane bonding technologies were firstly integrated for the fabrication of the proposed microfluidic device, resulting in a tightly three-dimensional structure for practical application. The magnitude of impedance of the tested solutions in the frequency range of 1 Hz to 100 kHz was analyzed by the Zennium electrochemical workstation. The results show that the newly designed microfluidic device has potential for impedance analysis with the advantages of ease of fabrication and the integration of 3D electrodes in the side walls. The newly designed impedance sensor can distinguish different concentrations of polystyrene microspheres and may have potential for cell counting in biological areas. By integrating with other techniques such as dielectrophoresis (DEP and biological recognition technology, the proposed device may have potential for the assay to identify foodborne pathogen bacteria.
Application of plant impedance for diagnosing plant disease
Xu, Huirong; Jiang, Xuesong; Zhu, Shengpan; Ying, Yibin
2006-10-01
Biological cells have components acting as electrical elements that maintain the health of the cell by regulation of the electrical charge content. Plant impedance is decided by the state of plant physiology and pathology. Plant physiology and pathology can be studies by measuring plant impedance. The effect of Cucumber Mosaic Virus red bean isolate (CMV-RB) on electrical resistance of tomato leaves was studied by the method of impedance measurement. It was found that the value of resistance of tomato leaves infected with CMV-RB was smaller than that in sound plant leaves. This decrease of impedances in leaf tissue was occurred with increased severity of disease. The decrease of resistance of tomato leaves infected with CMV-RB could be detected by electrical resistance detecting within 4 days after inoculation even though significant visible differences between the control and the infected plants were not noted, so that the technique for measurement of tomato leaf tissue impedance is a rapid, clever, simple method on diagnosis of plant disease.
Force control for mechanoinduction of impedance variation in cellular organisms
International Nuclear Information System (INIS)
Constantly exposed to various forms of mechanical forces inherent in their physical environment (such as gravity, stress induced by fluid flow or cell–cell interactions, etc), cellular organisms sense such forces and convert them into biochemical signals through the processes of mechanosensing and mechanotransduction that eventually lead to biological changes. The effect of external forces on the internal structures and activities in a cellular organism may manifest in changes its physical properties, such as impedance. Studying variation in the impedance of a cellular organism induced by the application of an external mechanical force represents a meaningful endeavor (from a biosystems perspective) in exploring the complex mechanosensing and mechanotransduction mechanisms that govern the behavior of a cellular organism under the influence of external mechanical stimuli. In this paper we describe the development of an explicit force-feedback control system for exerting an indentation force on a cellular organism while simultaneously measuring its impedance. To demonstrate the effectiveness of this force-control system, we have conducted experiments using zebrafish embryos as a test model of a cellular organism. We report experimental results demonstrating that the application of a properly controlled external force leads to a significant change in the impedance of a zebrafish embryo. These results offer support for a plausible explanation that activities of pore canals in the chorion are responsible for the observed change in impedance.
Impedance simulation for LEReC booster cavity transformed from ERL gun cavity
Energy Technology Data Exchange (ETDEWEB)
Liu, Chuyu [Brookhaven National Lab. (BNL), Upton, NY (United States)
2015-11-24
Wake impedance induced energy spread is a concern for the low energy cooling electron beam. The impedance simulation of the booster cavity for the LEReC projection is presented in this report. The simulation is done for both non-relativistic and ultra-relativistic cases. The space charge impedance in the first case is discussed. For impedance budget consideration of the electron machine, only simulation of the geometrical impedance in the latter case is necessary since space charge is considered separately.
Active impedance metasurface with full 360° reflection phase tuning
Zhu, Bo O.; Zhao, Junming; Feng, Yijun
2013-01-01
Impedance metasurface is composed of electrical small scatters in two dimensional plane, of which the surface impedance can be designed to produce desired reflection phase. Tunable reflection phase can be achieved by incorporating active element into the scatters, but the tuning range of the reflection phase is limited. In this paper, an active impedance metasurface with full 360° reflection phase control is presented to remove the phase tuning deficiency in conventional approach. The unit cell of the metasurface is a multiple resonance structure with two resonance poles and one resonance zero, capable of providing 360° reflection phase variation and active tuning within a finite frequency band. Linear reflection phase tuning can also be obtained. Theoretical analysis and simulation are presented and validated by experiment at microwave frequency. The proposed approach can be applied to many cases where fine and full phase tuning is needed, such as beam steering in reflectarray antennas. PMID:24162366
Electrical impedance tomography of the 1995 OGI gasoline release
International Nuclear Information System (INIS)
Electrical impedance tomography (EIT) was used to image the plume resulting from a release of 378 liters (100 gallons) of gasoline into a sandy acquifer. Images were made in 5 planes before and 5 times during the release, to generate a detailed picture of the spatial as well as the temporal development of the plume as it spread at the water table. Information of the electrical impedance (both in phase and out of phase voltages) was used or several different frequencies to produce images. We observed little dispersion in the images either before or after the gasoline entered the acquifer. Likewise, despite some laboratory measurements of impedances, there was no evidence of a change in the reactance in the soil because of the gasoline
Complex Impedance Spectroscopic Properties of BaVO Ceramics
Directory of Open Access Journals (Sweden)
Praveen Khatri
2008-01-01
Full Text Available The polycrystalline sample of Ba3V2O8 was prepared by a high-temperature solid-state reaction technique. The effect of temperature on impedance parameters was studied using an impedance analyzer in a wide frequency range (102-106 Hz. The real and imaginary parts of complex impedance trace semicircles in the complex plane. The temperature-dependent plots reveal the presence of both bulk and grain boundary effects above 125∘C. The bulk resistance of the material decreases with rise in temperature. This exhibits a typical negative temperature coefficient of resistance (NTCR behavior of the material. The modulus analysis suggests a possible hopping mechanism for electrical transport processes of the material. The nature of variation of dc conductivity suggests the Arrhenius type of electrical conductivity.
Road Impedance Model Study under the Control of Intersection Signal
Directory of Open Access Journals (Sweden)
Yunlin Luo
2015-01-01
Full Text Available Road traffic impedance model is a difficult and critical point in urban traffic assignment and route guidance. The paper takes a signalized intersection as the research object. On the basis of traditional traffic wave theory including the implementation of traffic wave model and the analysis of vehicles’ gathering and dissipating, the road traffic impedance model is researched by determining the basic travel time and waiting delay time. Numerical example results have proved that the proposed model in this paper has received better calculation performance compared to existing model, especially in flat hours. The values of mean absolute percentage error (MAPE and mean absolute deviation (MAD are separately reduced by 3.78% and 2.62 s. It shows that the proposed model has feasibility and availability in road traffic impedance under intersection signal.
Different radiation impedance models for finite porous materials
DEFF Research Database (Denmark)
Nolan, Melanie; Jeong, Cheol-Ho; Brunskog, Jonas;
2015-01-01
The Sabine absorption coefficients of finite absorbers are measured in a reverberation chamber according to the international standard ISO 354. They vary with the specimen size essentially due to diffraction at the specimen edges, which can be seen as the radiation impedance differing from the...... infinite case. Thus, in order to predict the Sabine absorption coefficients of finite porous samples, one can incorporate models of the radiation impedance. In this study, different radiation impedance models are compared with two experimental examples. Thomasson’s model is compared to Rhazi’s method when...... coupled to the transfer matrix method (TMM). These methods are found to yield comparable results when predicting the Sabine absorption coefficients of finite porous materials. Discrepancies with measurement results can essentially be explained by the unbalance between grazing and non-grazing sound field...
Fiber Materials AC Impedance Characteristics and Principium Analysis
Wang, Jianjun; Li, Xiaofeng
With an invariable amplitude and variable frequency inspiriting, impedance of fiber materials rapidly decrease at first and then increase speedy followed with increasing of signal frequency. For the impedance curve of frequency is section of bathtub, this phenomenon is defined as alternating current electric conductive bathtub effect of fiber material. With analysis tools,of circuit theory and medium polarization theory, the phenomenon can be deeply detected that in AC electric field there are four different kind of currents in fiber material: absorbing current, conductance current, charging current and superficial current. With more analyzing it's discovered this phenomenon can be explained by medium polarize theory. Make using of fiber AC electric conductivity bathtub effect, fast testing equipment on fiber moisture regain can be invent, and disadvantages of conventional impedance technique, such as greatness test error and electrode polarization easily. This paper affords directions to design novel speediness fiber moisture test equipments in theory.
Sheath impedance effects in very high frequency plasma experiments
International Nuclear Information System (INIS)
The frequency dependence (13.56 MHz to 70 MHz) of the ion energy distribution at the ground electrode was measured by mass spectrometry in a symmetrical capacitive argon discharge. Reduced sheath impedance at Very High Frequency allows high levels of plasma power and substrate ion flux whilst maintaining low levels of ion energy and electrode voltage. The lower limit of ion bombardment energy is fixed by the sheath floating potential at high frequency, in contrast to low frequencies where only the rf voltage amplitude is determinant. The capacitive sheaths are thinner at high frequencies which accentuates the high frequency reduction in sheath impedance. It is argued that the frequency dependence of sheath impedance is responsible for the principal characteristics of Very High Frequency plasmas. The measurements are summarised by simple physical descriptions and compared with a Particle-In-Cell simulation. (author) figs., tabs., refs
The effect of vocal tract impedance on the vocal folds
DEFF Research Database (Denmark)
Agerkvist, Finn T.; Selamtzis, Andreas
2011-01-01
The importance of the interaction between the acoustic impedance of the vocal tract with the flow across the vocal cords is well established. In this paper we are investigating the changes in vocal tract impedance when using the different modes of phonation according to Sadolin [1], going from the...... soft levels of the Neutral mode to the high levels of the fully ‘metallic’ Edge mode. The acoustic impedance of vocal tract as seen from the mouth opening is measured via a microphone placed close to the mouth when exciting the system with a volume velocity source [2]. At the same time a Laryngograph...... frontend is used to measure the electroglottograph signal which reflects the opening and closing pattern of the vocal folds. The measurements were carried out for all four modes (Neutral, Curbing, Overdrive and Edge) for the vowel [a] in three different pitches: C3(131 Hz), G3 (196 Hz) and C4 (262Hz) . The...
Broadband excitation for short-time impedance spectroscopy
International Nuclear Information System (INIS)
Frequency domain impedance measurements are still the common approach in assessing passive electrical properties of cells and tissues. However, due to the time requirements for sweeping over a frequency range for performing spectroscopy, they are not suited for recovering fast impedance changes of biological objects. The use of broad bandwidth excitation and monitoring the response as a function of time will greatly reduce the measurement time. The widespread usage of a square wave excitation is simple but not always the best choice. Here we consider different waveforms for excitation and discuss not only the advantages but also their limitations. Measurements in a miniaturized chamber where frequency and time domain measurements are compared show the suitability of different waveforms as excitation signals for the measurements of bio-impedance. The chirp excitation has been found to be most promising in terms of frequency range, signal-to-noise ratio and crest factor
Impedance measurement system for a microwave-induced plasma
International Nuclear Information System (INIS)
An impedance measurement system for an atmospheric microwave-induced plasma is proposed using equivalent transmission line theory and a de-embedding process of an input impedance. An argon plasma was generated by using a microwave-induced discharge. The generator consists of a 2.45 GHz solid state power oscillator (SSPO), a circulator, a directional coupler, a three-stub tuner, a gas flow meter, and a plasma jet. The microwave source power was varied from 10 to 25 W, and the argon gas flux was varied from 1 to 5 L/min at atmospheric pressure. The experimental results for the microwave-induced plasma show the impedance behaviors of the resistance and the inductive reactance. The near unity voltage standing wave ratio (VSWR) from the experiment results ensures that the proposed measurement system is valid.
Impedance measurement system for a microwave-induced plasma
Energy Technology Data Exchange (ETDEWEB)
Kim, Jae Duk; Kim, Sang Hoon; Kim, Hyoung Jong; Shin, Suk Woo; Choi, Jin Joo; Na, Young Ho [Kwangwoon University, Seoul (Korea, Republic of)
2012-03-15
An impedance measurement system for an atmospheric microwave-induced plasma is proposed using equivalent transmission line theory and a de-embedding process of an input impedance. An argon plasma was generated by using a microwave-induced discharge. The generator consists of a 2.45 GHz solid state power oscillator (SSPO), a circulator, a directional coupler, a three-stub tuner, a gas flow meter, and a plasma jet. The microwave source power was varied from 10 to 25 W, and the argon gas flux was varied from 1 to 5 L/min at atmospheric pressure. The experimental results for the microwave-induced plasma show the impedance behaviors of the resistance and the inductive reactance. The near unity voltage standing wave ratio (VSWR) from the experiment results ensures that the proposed measurement system is valid.
Ground Loop Impedance of Long EHV Cable Lines
DEFF Research Database (Denmark)
Ohno, Teruo; Bak, Claus Leth; Sørensen, Thomas K.
2012-01-01
The distance protection scheme without communication is often applied to the backup protection of EHV cable lines. For a reliable operation of a ground distance relay, the ground loop impedance of EHV cable lines needs to have a linear relationship to the distance from the relay location to the...... literature. Through the calculation of the ground loop impedance for cable lines, it has been found that, for long EHV cable lines, the reliable operation of the ground distance relay is possible with a typical relay setting. Effects of parameters, such as substation grounding, cable layouts and...... fault location. The discontinuity of the ground loop impedance at cross-bonding points may have an ill effect on the reliable operation of the ground distance relay. However, the cause and parameters of the discontinuity and its effects on the ground distance relay protection have not been discussed in...
Electrical Modeling and Impedance Analysis of Biological Cells
Directory of Open Access Journals (Sweden)
Gowri Sree V.
2014-03-01
Full Text Available It was proved that the external electric field intensity has significant effects on the biological systems. The applied electric field intensity changes the electrical behavior of the cell systems. The impact of electric field intensity on the cell systems should be studied properly to optimize the electric field treatments of biological systems. Based on the cell dimensions and its dielectric properties, an electrical equivalent circuit for an endosperm cell in rice was developed and its total impedance and capacitance were verified with measurement results. The variations of impedance and conductance with respect to applied impulse voltage at different frequencies were plotted. This impedance analysis method can be used to determine the optimum voltage level for electric field treatment and also to determine the cell rupture due to electric field applications.
Characterization of high impedance connecting links for Bolometric detectors
Giachero, A.; Gotti, C.; Maino, M.; Pessina, G.
2013-08-01
High impedance connecting links and cables are tested at low frequency in terms of their parasitic impedance to ground and to neighboring connecting links. These parameters must be well characterized with detectors operated at low temperature, especially when the very front-end is at room temperature, which results in a long link. This is the case of the LUCIFER experiment, an array of crystals where every event of interest produces two signals, one composed of phonons, the other of photons. The parasitic impedance is usually considered to be the parallel combination of a resistance and a capacitance. We characterized both and found that from the static measurements the capacitance of the cable resulted much larger. On the basis of this result we optimized the measurement set-up and developed a model to account for this behavior.
Characterization of high impedance connecting links for Bolometric detectors
International Nuclear Information System (INIS)
High impedance connecting links and cables are tested at low frequency in terms of their parasitic impedance to ground and to neighboring connecting links. These parameters must be well characterized with detectors operated at low temperature, especially when the very front-end is at room temperature, which results in a long link. This is the case of the LUCIFER experiment, an array of crystals where every event of interest produces two signals, one composed of phonons, the other of photons. The parasitic impedance is usually considered to be the parallel combination of a resistance and a capacitance. We characterized both and found that from the static measurements the capacitance of the cable resulted much larger. On the basis of this result we optimized the measurement set-up and developed a model to account for this behavior
Impedance-Source Networks for Electric Power Conversion Part II
DEFF Research Database (Denmark)
Siwakoti, Yam P.; Peng, Fang Zheng; Blaabjerg, Frede;
2015-01-01
-source-network-based power converters has been covered in a previous paper and main topologies were discussed from an application point of view. Now Part II provides a comprehensive review of the most popular control and modulation strategies for impedance-source network-based power converters/inverters. These methods are......Impedance-source networks cover the entire spectrum of electric power conversion applications (dc-dc, dc-ac, ac-dc, ac-ac) controlled and modulated by different modulation strategies to generate the desired dc or ac voltage and current at the output. A comprehensive review of various impedance...... topology at a certain power level, switching frequency and demanded dynamic response....
A Wireless Multi-Sensor Dielectric Impedance Spectroscopy Platform
Directory of Open Access Journals (Sweden)
Seyed Alireza Ghaffari
2015-09-01
Full Text Available This paper describes the development of a low-cost, miniaturized, multiplexed, and connected platform for dielectric impedance spectroscopy (DIS, designed for in situ measurements and adapted to wireless network architectures. The platform has been tested and used as a DIS sensor node on ZigBee mesh and was able to interface up to three DIS sensors at the same time and relay the information through the network for data analysis and storage. The system is built from low-cost commercial microelectronics components, performs dielectric spectroscopy ranging from 5 kHz to 100 kHz, and benefits from an on-the-fly calibration system that makes sensor calibration easy. The paper describes the microelectronics design, the Nyquist impedance response, the measurement sensitivity and accuracy, and the testing of the platform for in situ dielectric impedance spectroscopy applications pertaining to fertilizer sensing, water quality sensing, and touch sensing.
Using FOCUS to determine the radiation impedance for square transducers
Jennings, Matthew R.; McGough, Robert J.
2012-10-01
The power radiated by an ultrasound transducer is calculated with the radiation resistance, which is the real part of the radiation impedance. For circular transducers, an analytical solution for the radiation impedance is known, but an analytical expression for the radiation impedance is not available for rectangular or square transducers. To determine the radiation resistance in FOCUS, the pressure on the surface of a square transducer is computed with the fast nearfield method, and then the force on the transducer face is computed by integrating the pressure. Results using this approach are numerically evaluated for a range of ka values from 0.1 to 16. The pressure on the transducer face is also computed with the Rayleigh-Sommerfeld integral, and the results are compared. The numerical value of the radiation resistance computed with FOCUS and with the Rayleigh-Sommerfeld integral converge to the same value, although FOCUS calculates the same result in about one-quarter of the time.
Nuclear EMP: stripline test method for measuring transfer impedance
International Nuclear Information System (INIS)
A method for measuring the transfer impedance of flat metal joints for frequencies to 100 MHz has been developed which makes use of striplines. The stripline method, which has similarities to the quadraxial method used for cylindrical components, is described and sets of test results are given. The transfer impedance of a simple joint is modeled as a spurious hyperbolic curve, and a close curve fit to transfer impedance test data from various samples is demonstrated for both the stripline and the quadraxial methods. Validity checks of the test data are discussed using the curve model and other criteria. The method was developed for testing riveted joints which form the avionics bays on B-1s. The joints must provide shielding from EMP currents
Impedance-stabilized positive corona discharge and its decontamination properties
Energy Technology Data Exchange (ETDEWEB)
Horak, P; Khun, J, E-mail: pavel.horak@vscht.c [Department of Physics and Measurements, Faculty of Chemical Engineering, Institute of Chemical Technology, Technicka 5, 166 28 Praha 6 (Czech Republic)
2010-04-01
The point-to-plane DC corona discharge in air at atmospheric pressure was stabilized by a serially connected ballast impedance. The ballast impedance was implemented by a resistor-capacitor group connected in parallel. In the case of connecting the serial impedance into the electric circuit of a negative corona, the transition into a spark takes place at parameters similar to those of a non-stabilized discharge. In contrast, in the case of a positive corona, the discharge does not undergo a transition into a spark, but rather into a mode of periodic streamers. We measured the bactericidal effect of the stabilized discharge. The experiments showed that after a 2-minute exposure the quantity of surviving bacteria decreased from 95% for a non-stabilized discharge down to 5% for a stabilized one.
Impedance-stabilized positive corona discharge and its decontamination properties
International Nuclear Information System (INIS)
The point-to-plane DC corona discharge in air at atmospheric pressure was stabilized by a serially connected ballast impedance. The ballast impedance was implemented by a resistor-capacitor group connected in parallel. In the case of connecting the serial impedance into the electric circuit of a negative corona, the transition into a spark takes place at parameters similar to those of a non-stabilized discharge. In contrast, in the case of a positive corona, the discharge does not undergo a transition into a spark, but rather into a mode of periodic streamers. We measured the bactericidal effect of the stabilized discharge. The experiments showed that after a 2-minute exposure the quantity of surviving bacteria decreased from 95% for a non-stabilized discharge down to 5% for a stabilized one.
An impact analysis of the fault impedance on voltage sags
Energy Technology Data Exchange (ETDEWEB)
Ramos, Alessandro Candido Lopes [CELG - Companhia Energetica de Goias, Goiania, GO (Brazil). Generation and Transmission. System' s Operation Center], E-mail: alessandro.clr@celg.com.br; Batista, Adalberto Jose [Federal University of Goias (UFG), Goiania, GO (Brazil)], E-mail: batista@eee.ufg.br; Leborgne, Roberto Chouhy [Federal University of Rio Grande do Sul (UFRGS), Porto Alegre, RS (Brazil)], E-mail: rcl@ece.ufrgs.br; Emiliano, Pedro Henrique Mota, E-mail: ph@phph.com.br
2009-07-01
This paper presents an impact analysis of the fault impedance, in terms of its module and angle, on voltage sags caused by faults. Symmetrical and asymmetrical faults are simulated, at transmission and distribution lines, by using a frequency-domain fault simulation software called ANAFAS. Voltage sags are monitored at buses where sensitive end-users are connected. In order to overcome some intrinsic limitations of this software concerning its automatic execution for several cases, a computational tool was developed in Java programming language. This solution allows the automatic simulation of cases including the effect of the fault position, the fault type, and the proper fault impedance. The main conclusion is that the module and angle of the fault impedance can have a significant influence on voltage sag depending on the fault characteristics. (author)
Impedance Interaction Modeling and Analysis for Bidirectional Cascaded Converters
DEFF Research Database (Denmark)
Tian, Yanjun; Deng, Fujin; Chen, Zhe;
2015-01-01
more uncertainty to the system stability. An investigation is performed here for showing that the forward and reverse interactions are prominently different in terms of dynamics and stability even though the cascaded converter control remains unchanged. An important guideline has been drawn for the......For the cascaded converter system, the output impedance of source converter interacts with the input impedance of load converter, and the interaction may cause the system instability. In bidirectional applications, when the power flow is reversed, the impedance interaction also varies, which brings...... control of the cascaded converter. That is when voltage mode converter working as the load converter; the constant power mode converter as the source converter, the system is more stable. The concluded findings have been verified by simulation and experimental results....
Advanced impedance modeling of solid oxide electrochemical cells
DEFF Research Database (Denmark)
Graves, Christopher R.; Hjelm, Johan
2014-01-01
Impedance spectroscopy is a powerful technique for detailed study of the electrochemical and transport processes that take place in fuel cells and electrolysis cells, including solid oxide cells (SOCs). Meaningful analysis of impedance measurements is nontrivial, however, because a large number of...... modeling parameters are fit to the many processes which often overlap in the same frequency ranges. Also, commonly used equivalent circuit (EC) models only provide zero-dimensional (0-D) approximations of the processes of the two electrodes, electrolyte and gas transport. Employing improved analytical...... electrode and 2-D gas transport models which have fewer unknown parameters for the same number of processes, (ii) use of a new model fitting algorithm, “multi-fitting”, in which multiple impedance spectra are fit simultaneously with parameters linked based on the variation of measurement conditions, (iii...
Microgrid Reactive and Harmonic Power Sharing Using Enhanced Virtual Impedance
DEFF Research Database (Denmark)
He, Jinwei; Wei Li, Yun; Guerrero, Josep M.;
2013-01-01
only a line current feed-forward term is added to the DG voltage reference, the proposed virtual impedances at fundamental and harmonic frequencies are realized using DG line current and point of common coupling (PCC) voltage feed-forward terms, respectively. With this modification, the mismatched DG...... feeder impedances can be properly compensated, resulting in accurate reactive and harmonic power sharing at the same time. In addition, this paper shows that the microgrid PCC harmonic voltages can be mitigated by reducing the magnitude of DG unit equivalent harmonic impedance. Finally, an improved...... proportional plus resonant voltage controller is developed to ensure accurate power sharing and PCC harmonic voltage compensation without using any fundamental/harmonic component detections....
Electrical impedance imaging of water distribution in the root zone
Newill, P.; Karadaglić, D.; Podd, F.; Grieve, B. D.; York, T. A.
2014-05-01
The paper describes a technique that is proposed for imaging water transport in and around the root zone of plants using distributed measurements of electrical impedance. The technique has the potential to analyse sub-surface phenotypes, for instance drought tolerance traits in crop breeding programmes. The technical aim is to implement an automated, low cost, instrument for high-throughput screening. Ultimately the technique is targeted at in-field, on-line, measurements. For demonstration purposes the present work considers measurements on laboratory scale rhizotrons housing growing maize plants. Each rhizotron is fitted with 60 electrodes in a rectangular array. To reduce electrochemical effects the capacitively coupled contactless conductivity (C4D) electrodes have an insulating layer on the surface and the resistance of the bulk material is deduced from spectroscopic considerations. Electrical impedance is measured between pairs of electrodes to build up a two-dimensional map. A modified electrical model of such electrodes is proposed which includes the resistive and reactive components of both the insulating layer and the bulk material. Measurements taken on a parallel-plate test cell containing water confirm that the C4D technique is able to measure electrical impedance. The test cell has been used to explore the effects of water content, compaction and temperature on measurements in soil. Results confirm that electrical impedance measurements are very sensitive to moisture content. Impedance fraction changes up to 20% are observed due to compaction up to a pressure of 0.21 kg cm-2 and a temperature fraction sensitivity of about 2%/°C. The effects of compaction and temperature are most significant under dry conditions. Measurements on growing maize reveal the changes in impedance across the rhizotron over a period of several weeks. Results are compared to a control vessel housing only soil.
Correcting electrode impedance effects in broadband SIP measurements
Huisman, Johan Alexander; Zimmermann, Egon; Esser, Odilia; Haegel, Franz-Hubert; Vereecken, Harry
2016-04-01
Broadband spectral induced polarization (SIP) measurements of the complex electrical resistivity can be affected by the contact impedance of the potential electrodes above 100 Hz. In this study, we present a correction procedure to remove electrode impedance effects from SIP measurements. The first step in this correction procedure is to estimate the electrode impedance using a measurement with reversed current and potential electrodes. In a second step, this estimated electrode impedance is used to correct SIP measurements based on a simplified electrical model of the SIP measurement system. We evaluated this new correction procedure using SIP measurements on water because of the well-defined dielectric properties. It was found that the difference between the corrected and expected phase of the complex electrical resistivity of water was below 0.1 mrad at 1 kHz for a wide range of electrode impedances. In addition, SIP measurements on a saturated unconsolidated sediment sample with two types of potential electrodes showed that the measured phase of the electrical resistivity was very similar (difference SIP measurements on variably saturated unconsolidated sand were made. Here, the plausibility of the phase of the electrical resistivity was improved for frequencies up to 1 kHz, but errors remained for higher frequencies due to the approximate nature of the electrode impedance estimates and some remaining unknown parasitic capacitances that led to current leakage. It was concluded that the proposed correction procedure for SIP measurements improved the accuracy of the phase measurements by an order of magnitude in the kHz frequency range. Further improvement of this accuracy requires a method to accurately estimate parasitic capacitances in situ.
Fuzzy modeling of electrical impedance tomography images of the lungs
Directory of Open Access Journals (Sweden)
Harki Tanaka
2008-01-01
Full Text Available OBJECTIVES: Aiming to improve the anatomical resolution of electrical impedance tomography images, we developed a fuzzy model based on electrical impedance tomography's high temporal resolution and on the functional pulmonary signals of perfusion and ventilation. INTRODUCTION: Electrical impedance tomography images carry information about both ventilation and perfusion. However, these images are difficult to interpret because of insufficient anatomical resolution, such that it becomes almost impossible to distinguish the heart from the lungs. METHODS: Electrical impedance tomography data from an experimental animal model were collected during normal ventilation and apnea while an injection of hypertonic saline was administered. The fuzzy model was elaborated in three parts: a modeling of the heart, the pulmonary ventilation map and the pulmonary perfusion map. Image segmentation was performed using a threshold method, and a ventilation/perfusion map was generated. RESULTS: Electrical impedance tomography images treated by the fuzzy model were compared with the hypertonic saline injection method and computed tomography scan images, presenting good results. The average accuracy index was 0.80 when comparing the fuzzy modeled lung maps and the computed tomography scan lung mask. The average ROC curve area comparing a saline injection image and a fuzzy modeled pulmonary perfusion image was 0.77. DISCUSSION: The innovative aspects of our work are the use of temporal information for the delineation of the heart structure and the use of two pulmonary functions for lung structure delineation. However, robustness of the method should be tested for the imaging of abnormal lung conditions. CONCLUSIONS: These results showed the adequacy of the fuzzy approach in treating the anatomical resolution uncertainties in electrical impedance tomography images.
Fuzzy modeling of electrical impedance tomography images of the lungs
Energy Technology Data Exchange (ETDEWEB)
Tanaka, Harki; Ortega, Neli Regina Siqueira; Galizia, Mauricio Stanzione [Universidade de Sao Paulo (USP), SP (Brazil). Faculdade de Medicina. Medical Informatics; Borges, Joao Batista; Amato, Marcelo Britto Passos [Universidade de Sao Paulo (USP), SP (Brazil). Faculdade de Medicina. Dept. of Experimental Pneumology]. E-mail: harki_t@yahoo.com
2008-07-01
Objectives: Aiming to improve the anatomical resolution of electrical impedance tomography images, we developed a fuzzy model based on electrical impedance tomography's high temporal resolution and on the functional pulmonary signals of perfusion and ventilation. Introduction: Electrical impedance tomography images carry information about both ventilation and perfusion. However, these images are difficult to interpret because of insufficient anatomical resolution, such that it becomes almost impossible to distinguish the heart from the lungs. Methods: Electrical impedance tomography data from an experimental animal model were collected during normal ventilation and apnoea while an injection of hypertonic saline was administered. The fuzzy model was elaborated in three parts: a modeling of the heart, the pulmonary ventilation map and the pulmonary perfusion map. Image segmentation was performed using a threshold method, and a ventilation/perfusion map was generated. Results: Electrical impedance tomography images treated by the fuzzy model were compared with the hypertonic saline injection method and computed tomography scan images, presenting good results. The average accuracy index was 0.80 when comparing the fuzzy modeled lung maps and the computed tomography scan lung mask. The average ROC curve area comparing a saline injection image and a fuzzy modeled pulmonary perfusion image was 0.77. Discussion: The innovative aspects of our work are the use of temporal information for the delineation of the heart structure and the use of two pulmonary functions for lung structure delineation. However, robustness of the method should be tested for the imaging of abnormal lung conditions. Conclusions: These results showed the adequacy of the fuzzy approach in treating the anatomical resolution uncertainties in electrical impedance tomography images. (author)
Impedance reduction mechanisms in a magnetically immersed electron diode
International Nuclear Information System (INIS)
The successful operation of a magnetically-immersed electron diode using inductive-voltage-adder (IVA) technology for radiography requires a stable impedance for > 30 ns. In an IVA, the voltages from many acceleration gaps are added in series along a magnetically-insulated transmission line. The final voltage is applied to a thin needle that is immersed in a 10--50 Tesla solenoidal magnetic field. An electron beam is produced in a small spot at the anode target (10 MV). The electron current flowing off the needle is determined by the space-charge limited flow in a long pipe. Ion-emitting plasmas are produced from direct beam heating of the target and from radiation emitted from the several eV target surface that heats the outer walls of the anode (1--4.5-cm radius). Plasma ions are attracted to the electron beam and provide a degree of charge neutralization. The presence of these neutralizing ions reduces the diode impedance (nominally 300 Ohms), enhancing the electron current. The author is studying the impedance reduction mechanisms with the hybrid simulation code IPROP. He finds ions emitted from the anode walls lead to a reduced but stable impedance consistent with bipolar flow. If the ions are permitted to strip in ion-ion collisions, however, a runaway loss of impedance occurs that provides diode current well above the bipolar limit. Assuming nitrogen ion emission, he has qualitatively reproduced impedance behavior observed in experiments on the Hermes III IVA accelerator at Sandia National Laboratories. Several mitigation schemes are being investigated
Design of optimized impedance transformer for ICRF antenna in LHD
International Nuclear Information System (INIS)
Highlights: ► We developed optimization method of impedance transformer for ICRF antenna. ► Power loss will be one-third with the optimized impedance transformer. ► Possibility of damage on the transmission line will be drastically reduced. ► High performance will be kept in the wide antenna impedance region. -- Abstract: A pair of ion cyclotron range of frequencies (ICRF) antennas in the large helical device (LHD), HAS antennas showed high efficiency in minority ion heating. However the low loading resistance must be increased to prevent breakdown in transmission line. Moreover, the voltage and the current around the feed-through must be reduced to protect it. For these purpose, we developed a design procedure of the impedance transformer for HAS antennas. To optimize the transformer, the inner conductors were divided into several segments and the radii of them were given discretely and independently. The maximum effective loading resistance was obtained by using all combinations of radii within the limitations of the voltage and current at the feed-through and the electric field on the transformer. To get a precise solution, this procedure was repeated several times by narrowing the range of radii of inner conductors. Then the optimized impedance transformer was designed by smoothing the radii of inner conductors. For the typical discharge, the voltage and current at the feed-through were reduced to the half of the original values with the same power. The effective loading resistance was increased to more than four times. High performance is expected in wide impedance region
Electrical impedance imaging of water distribution in the root zone
International Nuclear Information System (INIS)
The paper describes a technique that is proposed for imaging water transport in and around the root zone of plants using distributed measurements of electrical impedance. The technique has the potential to analyse sub-surface phenotypes, for instance drought tolerance traits in crop breeding programmes. The technical aim is to implement an automated, low cost, instrument for high-throughput screening. Ultimately the technique is targeted at in-field, on-line, measurements. For demonstration purposes the present work considers measurements on laboratory scale rhizotrons housing growing maize plants. Each rhizotron is fitted with 60 electrodes in a rectangular array. To reduce electrochemical effects the capacitively coupled contactless conductivity (C4D) electrodes have an insulating layer on the surface and the resistance of the bulk material is deduced from spectroscopic considerations. Electrical impedance is measured between pairs of electrodes to build up a two-dimensional map. A modified electrical model of such electrodes is proposed which includes the resistive and reactive components of both the insulating layer and the bulk material. Measurements taken on a parallel-plate test cell containing water confirm that the C4D technique is able to measure electrical impedance. The test cell has been used to explore the effects of water content, compaction and temperature on measurements in soil. Results confirm that electrical impedance measurements are very sensitive to moisture content. Impedance fraction changes up to 20% are observed due to compaction up to a pressure of 0.21 kg cm−2 and a temperature fraction sensitivity of about 2%/°C. The effects of compaction and temperature are most significant under dry conditions. Measurements on growing maize reveal the changes in impedance across the rhizotron over a period of several weeks. Results are compared to a control vessel housing only soil. (paper)
THE COUPLING IMPEDANCE OF THE RHIC INJECTION KICKER SYSTEM
International Nuclear Information System (INIS)
IN THIS PAPER, RESULTS FROM IMPEDANCE MEASUREMENTS ON THE RHIC INJECTION KICKERS ARE REPORTED. THE KICKER IS CONFIGURED AS A ''C'' CROSS SECTION MAGNET WITH INTERLEAVED FERRITE AND HIGH-PERMITTIVITY DIELECTRIC SECTIONS TO ACHIEVE A TRAVELLING WAVE STRUCTURE. THE IMPEDANCE WAS MEASURED USING THE WIRE METHOD, AND ACCURATE RESULTS ARE OBTAINED BY INTERPRETING THE FORWARD SCATTERING COEFFICIENT VIA THE LONG-FORMULA. THE FOUR KICKERS WITH THEIR CERAMIC BEAM TUBES CONTRIBUE AT Z/N-0.22 OMEGA/RING IN THE INTERESTING FREQUENCY RANGE FROM 0.1 TO 1 BHZ, AND LESS ABOVE
Impedance technique for measuring dielectrophoretic collection of microbiological particles
Allsopp, D W E; Brown, A P; Betts, W B
1999-01-01
Measurement of the impedance change resulting from the collection of microbiological particles at coplanar electrodes is shown to be an effective and potentially quantitative method of detecting dielectrophoresis. Strong correlations between the frequency-dependent dielectrophoretic collection characteristics measured by impedance change and those observed using an established counting method based on image analysis have been obtained for Escherichia coli. In addition it is shown that the new electrical method can be used to sense dielectrophoretic collection of 19 nm diameter latex beads, particles too small to be resolved by conventional optical detection systems. (author)
[An instrument for estimating human body composition using impedance measurement].
Yin, J; Peng, C
1997-03-01
According to the impedance feature of biological tissue, the instrument was designed at 1, 5, 10, 50, 100kHz to measure human impedance, and then to calculate human FAT, FFM, FAT%, TBW, ECW, ICW and so on. A 8031 singlechip microprocessor contacuting used as a control center in the instrument. The part of electric circuit contacuting human body in the instrument was unreally earthing. The instrument was safty, effective, repeatable, and easily manpulative. Prelimintary clinical experiment showed the results measured with the instrument could effectively reflect practical, status of human composition. PMID:9647623
A Case of Aerophagia Diagnosed by Multichannel Intraluminal Impedance Monitoring.
Sohn, Ki Chang; Jeong, Young Hoon; Jo, Dong Ho; Heo, Won Gak; Yeom, Dong Han; Choi, Suck Chei; Ryu, Han Seung
2015-11-01
Aerophagia is a disorder caused by abnormal accumulation of air in the gastrointestinal tract as a result of repetitive and frequent inflow of air through the mouth. For the diagnosis of this condition, it is difficult to objectively measure the air swallowing. However, multichannel intraluminal impedance monitoring facilitates the differential diagnosis between normal air swallowing and pathologic aerophagia, and can aid in the determination of the frequency and amount of air swallowed. In this report, in addition to a literature review, we describe a case of 36-year-old man with abdominal distension who was diagnosed with aerophagia using esophageal impedance monitoring and was treated with clonazepam. PMID:26586352
The beam coupling impedance model of CERN Proton Synchrotron
Persichelli, Serena; Migliorati, Mauro; Salvant, Benoit
The research activity described in this thesis work is dedicated to developing a longitudinal and a transverse beam coupling impedance model for the CERN Proton Synchrotron (PS), in the framework of the Large Hadron Collider (LHC) Injector Upgrade (LIU) project. The study allows a better understanding of the instability threshold of the machine, helping predicting the effects of the current increase planned for the upgrade program. Furthermore, the knowledge of the machine beam coupling impedance model allows improving the stability of beams injected into the LHC chain, in prevision for the particle collision energy increase in program for LHC physics experiments.
Experiences with a two terminal-pair digital impedance bridge
Callegaro, Luca; Kampik, Marian; Kim, Dan Bee; Ortolano, Massimo; Pourdanesh, Faranak
2014-01-01
This paper describes the realization of a two terminal-pair digital impedance bridge and the test measurements performed with it. The bridge, with a very simple architecture, is based on a commercial two-channel digital signal synthesizer and a synchronous detector. The bridge can perform comparisons between impedances having arbitrary phase and magnitude ratio: its balance is achieved automatically in less than a minute. $R$-$C$ comparisons with calibrated standards, at kHz frequency and 100 kohm magnitude level, give ratio errors of the order of $10^{-6}$, with potential for further improvements.
The frequency characteristics of medium voltage distribution system impedances
Directory of Open Access Journals (Sweden)
Liviu Emil Petrean
2009-10-01
Full Text Available In this paper we present the frequency characteristics of impedances involved in the electrical equivalent circuit of a large medium voltage distribution system. These impedances influence harmonics distortions propagation occurring due to the nonsinusoidal loads. We analyse the case of a 10 kV large urban distribution system which supplies industrial, commercial and residential customers. The influence of various parameters of the distribution network on the frequency characteristics are presented, in order to assess the interaction of harmonic distortion and distribution system network.
Diffusion and conversion impedance in solid oxide fuel cells
DEFF Research Database (Denmark)
Jacobsen, Torben; Hendriksen, Peter Vang; Koch, Søren
2008-01-01
In fuel cell electrodes where the thin, active electrode layer is supported by a porous layer the diffusion processes as well as the change in the gas phase composition due to the conversion of reactants will influence the cell impedance. Two models are developed for the description of these...... processes. The first is a CSTR model based on a well convected gas supply volume equipped with gas inlet and outlet. Since the current distribution on the electrode surface is uniform, the impedance is derived for perturbation from a steady state and contributions from kinetics are directly additive. In the...
Line impedance estimation using model based identification technique
DEFF Research Database (Denmark)
Ciobotaru, Mihai; Agelidis, Vassilios; Teodorescu, Remus
into the operation of the grid-connected power converters. This paper describes a quasi passive method for estimating the line impedance of the distribution electricity network. The method uses the model based identification technique to obtain the resistive and inductive parts of the line impedance....... The quasi-passive behaviour of the proposed method comes from the combination of the non intrusive behaviour of the passive methods with a better accuracy of the active methods. The simulation results reveal the good accuracy of the proposed method....
Short circuit currents calculation by using the impedance correction method
Energy Technology Data Exchange (ETDEWEB)
Schaefer, W.
1985-01-01
By introduction of correction factors for the impedances of generators and power station unit transformers into the known guidings according to VDE 0102 (basis principle of short circuit current calculation) it is possible to limit the deviation to less then +-5% of the value of the complete superposition method apart from some exceptions. The correction factors are valid for calculation of the maximum short circuit currents considering the practically admissible operating conditions. The separated correction of the impedances of generators and power station unit transformers can be used without restriction for all short circuit points of interest.
Construction of Tunnel Diode Oscillator for AC Impedance Measurement
Shin, J. H.; Kim, E.
2014-03-01
We construct a tunnel diode oscillator (TDO) to study electromagnetic response of a superconducting thin film. Highly sensitive tunnel diode oscillators allow us to detect extremely small changes in electromagnetic properties such as dielectric constant, ac magnetic susceptibility and magnetoresistance. A tunnel diode oscillator is a self-resonant oscillator of which resonance frequency is primarily determined by capacitance and inductance of a resonator. Amplitude of the signal depends on the quality factor of the resonator. The change in the impedance of the sample electromagnetic coupled to one of inductors in the resonator alters impedance of the inductor, and leads to the shift in the resonance frequency and the change of the amplitude.
Theory of fractional order elements based impedance matching networks
Radwan, Ahmed
2011-03-01
Fractional order circuit elements (inductors and capacitors) based impedance matching networks are introduced for the first time. In comparison to the conventional integer based L-type matching networks, fractional matching networks are much simpler and versatile. Any complex load can be matched utilizing a single series fractional element, which generally requires two elements for matching in the conventional approach. It is shown that all the Smith chart circles (resistance and reactance) are actually pairs of completely identical circles. They appear to be single for the conventional integer order case, where the identical circles completely overlap each other. The concept is supported by design equations and impedance matching examples. © 2010 IEEE.
Impedance Budget For Crab Cavity In MEIC Electron Ring
International Nuclear Information System (INIS)
The Medium Energy Electron-Ion Collider (MEIC) at Jefferson Lab has been envisioned as a first stage high energy particle accelerator beyond the 12 GeV upgrade of CEBAF. The estimate of impedance budget is important from the view point of beam stability and matching with other accelerator components driving currents. The detailed study of impedance budget for electron ring has been performed by considering the current design parameters of the e-ring. A comprehensive picture of the calculations involved in this study has been illustrated in the paper.
Two port network analysis for three impedance based oscillators
Said, Lobna A.
2011-12-01
Two-port network representations are applied to analyze complex networks which can be dissolved into sub-networks connected in series, parallel or cascade. In this paper, the concept of two-port network has been studied for oscillators. Three impedance oscillator based on two port concept has been analyzed using different impedance structures. The effect of each structure on the oscillation condition and the frequency of oscillation have been introduced. Two different implementations using MOS and BJT have been introduced. © 2011 IEEE.
Design methodology to enhance high impedance surfaces performances
Directory of Open Access Journals (Sweden)
Michael Grelier
2014-04-01
Full Text Available A methodology is introduced for designing wideband, compact and ultra-thin high impedance surfaces (HIS. A parametric study is carried out to examine the effect of the periodicity on the electromagnetic properties of an HIS. This approach allows designers to reach the best trade-off for HIS performances.
Organic electrochemical transistors for cell-based impedance sensing
Energy Technology Data Exchange (ETDEWEB)
Rivnay, Jonathan, E-mail: rivnay@emse.fr, E-mail: owens@emse.fr; Ramuz, Marc; Hama, Adel; Huerta, Miriam; Owens, Roisin M., E-mail: rivnay@emse.fr, E-mail: owens@emse.fr [Department of Bioelectronics, Ecole des Mines de St. Etienne, 13541 Gardanne (France); Leleux, Pierre [Department of Bioelectronics, Ecole des Mines de St. Etienne, 13541 Gardanne (France); Microvitae Technologies, Pole d' Activite Y. Morandat, 13120 Gardanne (France)
2015-01-26
Electrical impedance sensing of biological systems, especially cultured epithelial cell layers, is now a common technique to monitor cell motion, morphology, and cell layer/tissue integrity for high throughput toxicology screening. Existing methods to measure electrical impedance most often rely on a two electrode configuration, where low frequency signals are challenging to obtain for small devices and for tissues with high resistance, due to low current. Organic electrochemical transistors (OECTs) are conducting polymer-based devices, which have been shown to efficiently transduce and amplify low-level ionic fluxes in biological systems into electronic output signals. In this work, we combine OECT-based drain current measurements with simultaneous measurement of more traditional impedance sensing using the gate current to produce complex impedance traces, which show low error at both low and high frequencies. We apply this technique in vitro to a model epithelial tissue layer and show that the data can be fit to an equivalent circuit model yielding trans-epithelial resistance and cell layer capacitance values in agreement with literature. Importantly, the combined measurement allows for low biases across the cell layer, while still maintaining good broadband signal.
Effect of rib-cage structure on acoustic chest impedance
DEFF Research Database (Denmark)
Zimmermann, Niels Henrik; Møller, Henrik; Hansen, John; Hammershøi, Dorte; Rubak, Per
When a stethoscope is placed on the surface of the chest, the coupler picks up sound from heart and lungs transmitted through the tissues of the ribcage and from the surface of the skin. If the acoustic impedance of the chest surface is known, it is possible to optimize the coupler for picking up...
Contactless Impedance Sensors and Their Application to Flow Measurements
Directory of Open Access Journals (Sweden)
Karel Štulík
2013-02-01
Full Text Available The paper provides a critical discussion of the present state of the theory of high-frequency impedance sensors (now mostly called contactless impedance or conductivity sensors, the principal approaches employed in designing impedance flow-through cells and their operational parameters. In addition to characterization of traditional types of impedance sensors, the article is concerned with the use of less common sensors, such as cells with wire electrodes or planar cells. There is a detailed discussion of the effect of the individual operational parameters (width and shape of the electrodes, detection gap, frequency and amplitude of the input signal on the response of the detector. The most important problems to be resolved in coupling these devices with flow-through measurements in the liquid phase are also discussed. Examples are given of cell designs for continuous flow and flow-injection analyses and of detection systems for miniaturized liquid chromatography and capillary electrophoresis. New directions for the use of these sensors in molecular biology and chemical reactors and some directions for future development are outlined.
Impedance characterization of PV modules in outdoor conditions
DEFF Research Database (Denmark)
Oprea, Matei-lon; Thorsteinsson, Sune; Spataru, Sergiu;
2016-01-01
Impedance spectroscopy (IS) has been used for laboratory characterizations of photovoltaic (PV) technologies under well controlled conditions. This work applies IS for outdoor characterization of PV panels, in order to observe the effect of irradiance (G) and temperature (T) on the PV module...
Spectral magnetotelluric impedances for an anisotropic layered conductor
Czech Academy of Sciences Publication Activity Database
Pek, Josef
2002-01-01
Roč. 50, č. 4 (2002), s. 619-643. ISSN 0001-5725 R&D Projects: GA ČR GA205/99/0917; GA MŠk ME 185 Institutional research plan: CEZ:AV0Z3012916 Keywords : magnetotelluric impedance * anisotropic conductor * electromagnetic induction Subject RIV: DE - Earth Magnetism, Geodesy, Geography
Impedance of a rough conducting boundary in a layered medium
International Nuclear Information System (INIS)
The authors construct an operator for the equivalent impedance of a rough surface which bounds a layered medium, and discuss the physical approach by which it is generated. They give an example of the application of this operator in computing the attenuation of eigenmodes of an integrated optical waveguide with a rough substrate
A method for suppressing cardiogenic oscillations in impedance pneumography
International Nuclear Information System (INIS)
The transthoracic electrical impedance signal originates from the cardiac and respiratory functions. In impedance pneumography (IP) the lung function is assessed and the cardiac impedance signal, cardiogenic oscillations (CGOs), is considered an additive noise in the measured signal. In order to accurately determine pulmonary flow parameters from the signal, the CGO needs to be attenuated without distorting the respiratory part of the signal. We assessed the suitability of a filtering technique, originally described by Schuessler et al (1998 Ann. Biomed. Eng. 26 260–7) for an esophageal pressure signal, for CGO attenuation in the IP signal. The technique is based on ensemble averaging the CGO events using the electrocardiogram (ECG) R-wave as the trigger signal. Lung volume is known to affect the CGO waveforms. Therefore we modified the filtering method to produce a lung volume-dependent parametric model of the CGO waveform. A simultaneous recording of ECG, IP and pneumotachograph (PNT) was conducted on 41 healthy, sitting adults. The performance of the proposed method was compared to a low-pass filter and a Savitzky–Golay filter in terms of CGO attenuation and respiratory signal distortion. The method was found to be excellent, exhibiting CGO attenuation of 35.0±12.5 dB (mean±SD) and minimal distortion of the respiratory part of the impedance signal
Development of a current monitor using a negative impedance circuit
International Nuclear Information System (INIS)
We developed a beam current transformer which appropriates for monitoring beam of an accelerator having operating period of a few seconds. The beam monitor is a new type CT which can measure DC component of beam using a negative impedance circuit. In this report, we describe stability of a time constant, temperature and frequency characteristic of the CT. (author)
EU Regulations Impede Market Introduction of GM Forest Trees.
Custers, René; Bartsch, Detlef; Fladung, Matthias; Nilsson, Ove; Pilate, Gilles; Sweet, Jeremy; Boerjan, Wout
2016-04-01
Biotechnology can greatly improve the efficiency of forest tree breeding for the production of biomass, energy, and materials. However, EU regulations impede the market introduction of genetically modified (GM) trees so their socioeconomic and environmental benefits are not realized. European policy makers should concentrate on a science-based regulatory process. PMID:26897457
Impedance and instability threshold estimates in the main injector I
Energy Technology Data Exchange (ETDEWEB)
Martens, M.A.; Ng, K.Y.
1994-03-01
One of the important considerations in the design of the Main Injector is the beam coupling impedances in the vacuum chamber and the stability of the beam. Along with the higher intensities comes the possibility of instabilities which lead to growth in beam emittances and/or the loss of beam. This paper makes estimations of the various impedances and instability thresholds based on impedance estimations and measurements. Notably missing from this paper is any analysis of transition crossing and its potential limitations on beam intensity and beam emittance. Future work should consider this issue. The body of the work contains detailed analysis of the various impedance estimations and instability threshold calculations. The calculations are based on the Main Injector beam intensity of 6 x 10{sup 10} protons per bunch, 95% normalized transverse emittances of 20{pi} mm-mrad, and 95% normalized longitudinal emittance of 0.1 eV-s at 8.9 GeV injection energy and 0.25 eV-s at 150 GeV flattop energy. The conclusions section summarizes the results in the paper and is meant to be readable by itself without referring to the rest of the paper. Also in the conclusion section are recommendations for future investigations.
Electrochemical Impedance Study of Reduction Kinetics of the Pesticide Vinclozoline
Czech Academy of Sciences Publication Activity Database
Pospíšil, Lubomír; Sokolová, Romana; Colombini, M. P.; Giannarelli, S.; Fuoco, R.
2000-01-01
Roč. 67, - (2000), s. 305-312. ISSN 0026-265X R&D Projects: GA MŠk OC D15.10; GA ČR GA203/97/1048 Institutional research plan: CEZ:AV0Z4040901 Keywords : electrochemical impedance * pesticides * vinclozoline Subject RIV: CG - Electrochemistry Impact factor: 0.884, year: 2000
Three-dimensional conjugate gradients inversion of magnetotelluric impedance tensor
International Nuclear Information System (INIS)
Complete text of publication follows. We have developed a three-dimensional conjugate gradients inversion algorithm of magnetotelluric (MT) impedance tensor. This inversion algorithm can be used to invert 3D MT area data and 2D MT profile data. The importance of including the on-diagonal impedance tensor terms, Zxx and Zyy, in 3D inversion is considered in the synthetic example. The results from the 3D inversion of synthetic 3D area data indicate that the results using all impedance tensor elements, Zxx, Zxy, Zyx and Zyy, in the inversion are better than those without using on-diagonal elements. Therefore, 3D inversion should include all tensor elements if possible. The results from the 3D inversion of synthetic 2D profile data show that reasonable images for structures beneath and near a single MT profile can be obtained if all tensor elements are used in the inversion of the profile data. Currently, most of MT data are still collected on 2D profiles. This inversion algorithm of impedance tensor can promote the practical application of 3D MT inversion. We also use this inversion algorithm to invert the real data in the kayabe area in Japan. From the trial inversion with the synthetic and real data, the validity and practicability of this inversion algorithm is verified.
Impedance-Source Networks for Electric Power Conversion Part I
DEFF Research Database (Denmark)
Siwakoti, Yam P.; Peng, Fang Zheng; Blaabjerg, Frede;
2015-01-01
Impedance networks cover the entire of electric power conversion from dc (converter, rectifier), ac (inverter), to phase and frequency conversion (ac-ac) in a wide range of applications. Various converter topologies have been reported in the literature to overcome the limitations and problems of ...
Organic electrochemical transistors for cell-based impedance sensing
Rivnay, Jonathan; Ramuz, Marc; Leleux, Pierre; Hama, Adel; Huerta, Miriam; Owens, Roisin M.
2015-01-01
Electrical impedance sensing of biological systems, especially cultured epithelial cell layers, is now a common technique to monitor cell motion, morphology, and cell layer/tissue integrity for high throughput toxicology screening. Existing methods to measure electrical impedance most often rely on a two electrode configuration, where low frequency signals are challenging to obtain for small devices and for tissues with high resistance, due to low current. Organic electrochemical transistors (OECTs) are conducting polymer-based devices, which have been shown to efficiently transduce and amplify low-level ionic fluxes in biological systems into electronic output signals. In this work, we combine OECT-based drain current measurements with simultaneous measurement of more traditional impedance sensing using the gate current to produce complex impedance traces, which show low error at both low and high frequencies. We apply this technique in vitro to a model epithelial tissue layer and show that the data can be fit to an equivalent circuit model yielding trans-epithelial resistance and cell layer capacitance values in agreement with literature. Importantly, the combined measurement allows for low biases across the cell layer, while still maintaining good broadband signal.
Mutual Radiation Impedance of Uncollapsed CMUT Cells with Different Radii
Ozgurluk, Alper; Atalar, Abdullah; Koymen, Hayrettin
2015-01-01
A polynomial approximation is proposed for the mutual acoustic impedance between uncollapsed capacitive micromachined ultrasonic transducer (CMUT) cells with different radii in an infinite rigid baffle. The resulting approximation is employed in simulating CMUTs with a circuit model. A very good agreement is obtained with the corresponding finite element simulation (FEM) result.
Calculation of longitudinal CSR impedance in curved chamber
International Nuclear Information System (INIS)
Coherent synchrotron radiation (CSR) fields are generated when a bunched beam moves along a curved trajectory. A new code, named CSRZ, was developed using finite difference method to calculate the longitudinal CSR impedance for a beam moving along a curved chamber. The method adopted in our code was originated by T. Agoh and K. Yokoya [1]. It solves the parabolic equation in the frequency domain in a curvilinear coordinate system. In our studies, the chamber has uniform rectangular crosssection along the beam trajectory, which is the same as that in [1]. But the curvature of the beam trajectory is freed, and then we can investigate the CSR impedance of a single or a series of bending magnets. The calculation results indicate that the shielding effect due to outer chamber wall can be well explained by a simple optical approximation model at high frequencies. With an approximation of a wiggling chamber inside a wiggler, the coherent wiggler radiation (CWR) impedance has also been studied. Due to chamber shielding, the CWR impedance exhibits narrow peaks at frequencies satisfying the resonant conditions. (author)
Improved techniques of impedance calculation and localization in particle accelerators
Biancacci, Nicolò; Migliorati, Mauro; Métral, Elias; Salvant, Benoit
In this thesis we mainly focus on particle accelerators applied to high energy physics research where a fundamental parameter, the luminosity, is maximized in order to increase the rate of particle collisions useful to particle physicists. One way to increase this parameter is to increase the intensity of the circulating beams which is limited by the onset of collective effects that may drive the beam unstable and eventually provoke beam losses or reduce the beam quality required by the particle physics experiments. One major cause of collective effects is the beam coupling impedance, a quantity that quantifies the effect of the fields scattered by a beam passing through any accelerator device. The development of an impedance budget is required in those machines that are planning substantial upgrades as shown in this thesis for the CERN PS case. The main source of impedance in the CERN LHC are the collimators. Within an impedance reduction perspective, in order to reach the goals of the planned upgrades, it ...
Identification of Critical Transmission Limits in Injection Impedance Plane
DEFF Research Database (Denmark)
Jóhannsson, Hjörtur; Østergaard, Jacob; Nielsen, Arne Hejde
2012-01-01
the partial derivatives of the variables P,Q and V in respect to each other become zero. In addition to the mappingof the critical lines, some characteristic lines are mapped as well. These include the mapping of the lines of constant P,Q,Vand d from the PQV-surface into the impedance plane. All of...
The Current Distribution and Characteristic Impedance of Multiwire Hexaedric Antenna
Z. Novacek
1996-01-01
Both the current distribution and characteristic impedance variations along a hexaedric antenna are analysed in this paper. The presented arrangement by auxiliary wires may ensure equal currents in all antenna wires. The results are required for calculations of the field in the operating volume and may be used for antenna design.
Pulse wave velocity measured by Multichannel whole body impedance plethysmograph
Czech Academy of Sciences Publication Activity Database
Viščor, Ivo; Jurák, Pavel; Plešinger, Filip; Vondra, Vlastimil; Halámek, Josef; Soukup, L.; Matějková, M.
Brno : Society for Medical Innovation and Technology, 2015. s. 70. [SMIT 2015. International Conference of the Society for Medical Innovation and Technology/27./. 10.09.2015-12.09.2015, Brno] Institutional support: RVO:68081731 Keywords : electrical impedance * plethysmograph * pulse wave velocity * arterial stiffness Subject RIV: BH - Optics, Masers, Lasers
Impedance Spectra of Activating/Passivating Solid Oxide Electrodes
DEFF Research Database (Denmark)
Mogensen, Mogens Bjerg; Sun, Xiufu; Koch, Søren;
2014-01-01
Fuel Cells (SOFC) and Electrolyser Cells (SOEC) exist. Not all of them are irreversible, especially not over short periods. A reversible degradation is termed “passivation” and the reverse is then “activation”. These processes may exhibit themselves in the Electrochemical Impedance Spectra (EIS) as...
Effect of rib-cage structure on acoustic chest impedance
DEFF Research Database (Denmark)
Zimmermann, Niels Henrik; Møller, Henrik; Hansen, John;
2011-01-01
When a stethoscope is placed on the surface of the chest, the coupler picks up sound from heart and lungs transmitted through the tissues of the ribcage and from the surface of the skin. If the acoustic impedance of the chest surface is known, it is possible to optimize the coupler for picking up...
Determination of Complex Microcalorimeter Parameters with Impedance Measurements
Saab, T.; Bandler, S. R.; Chervenak, J.; Figueroa-Feliciano, E.; Finkbeiner, F.; Iyomoto, N.; Kelley, R.; Kilbourne, C. A.; Lindeman, M. A.; Porter, F. S.; Sadleir, J.
2005-01-01
The proper understanding and modeling of a microcalorimeter s response requires the accurate knowledge of a handful of parameters, such as C, G, alpha, . . . . While a few of these, such 8s the normal state resistance and the total thermal conductance to the heat bath (G) are directly determined from the DC IV characteristics, some others, notoriously the heat capacity (C) and alpha, appear in degenerate combinations in most measurable quantities. The case of a complex microcalorimeter, i.e. one in which the absorber s heat capacity is connected by a finite thermal impedance to the sensor, and subsequently by another thermal impedance to the heat bath, results in an added ambiguity in the determination of the individual C's and G's. In general, the dependence of the microcalorimeter s complex impedance on these parameters varies with frequency. This variation allows us to determine the individual parameters by fitting the prediction of the microcalorimeter model to the impedance data. We describe in this paper our efforts at characterizing the Goddard X-ray microcalorimeters. Using the parameters determined with this method we them compare the pulse shape and noise spectra predicted by the microcalorimeter model to data taken with the same devices.
Impedance matching of rf-excited CO2-lasers
International Nuclear Information System (INIS)
The rf-excitation technique is based on the application of a sinusoidal voltage in the regime above 10 MHz to obtain a glow discharge. Specific requirements are necessary to ensure an efficient and reflection free power transmission from the generator to the load. In this paper, the authors discusses impedance matching of rf-excited CO2-lasers. In the case of power matching the maximum power transmission is obtained, whereas impedance matching avoids reflection of the incident rf-power which would cause a standing wave at the line. Two or three element lumped circuits are sufficient to match the impedance of an rf-plasma to the characteristic impedance of the transmission line. In the authors' case, the eight laser discharge sections were driven by their own generators, each connected by a 50 Ω transmission line and a Collins filter. This Collins filter ensures an ideal matching for the total range of input power so that the total incident rf-power is utilized in the plasma
Electrochemical Impedance Spectroscopy to Characterize Inflammatory Atherosclerotic Plaques
Yu, Fei; Dai, Xiaohu; Beebe, Tyler; Hsiai, Tzung
2011-01-01
Despite advances in diagnosis and therapy, atherosclerotic cardiovascular disease remains the leading cause of morbidity and mortality in the Western world. Predicting metabolically active atherosclerotic lesions has remained an unmet clinical need. We hereby developed an electrochemical strategy to characterize the inflammatory states of high-risk atherosclerotic plaques. Using the concentric bipolar microelectrodes, we sought to demonstrate distinct Electrochemical Impedance Spectroscopic (...
Does Dividend Tax Impede Competition for Corporate Charters?
DEFF Research Database (Denmark)
Lai, Tat-kei; Ng, Travis
High dividend income tax in the U.S. can impede state competition in the market for corporate charters. We offer a model to formalize the mechanism through which dividend tax lowers the incentives for a state legislator to refrain from enacting takeover regulations. We test a key driver within th...
An impedance model of the relativistic heavy ion collider, RHIC
International Nuclear Information System (INIS)
This paper is an abbreviated version of a comprehensive and detailed analysis of RHIC instabilities soon to be published as a RHIC project report. It emphasises longitudinal impedance modeling and design choices in RHIC, wile a companion paper emphasises instability calculations
Does strategic planning enhance or impede innovation and firm performance?
Song, Michael; Im, Subin; van der Bij, Hans; Song, Lisa Z.
2011-01-01
Does strategic planning enhance or impede innovation and firm performance? The current literature provides contradictory views. This study extends the resource-advantage theory to examine the conditions in which strategic planning increases or decreases the number of new product development projects
BIOELECTRICAL IMPEDANCE VECTOR ANALYSIS IDENTIFIES SARCOPENIA IN NURSING HOME RESIDENTS
Loss of muscle mass and water shifts between body compartments are contributing factors to frailty in the elderly. The body composition changes are especially pronounced in institutionalized elderly. We investigated the ability of single-frequency bioelectrical impedance analysis (BIA) to identify b...
Impedance and instability threshold estimates in the main injector I
International Nuclear Information System (INIS)
One of the important considerations in the design of the Main Injector is the beam coupling impedances in the vacuum chamber and the stability of the beam. Along with the higher intensities comes the possibility of instabilities which lead to growth in beam emittances and/or the loss of beam. This paper makes estimations of the various impedances and instability thresholds based on impedance estimations and measurements. Notably missing from this paper is any analysis of transition crossing and its potential limitations on beam intensity and beam emittance. Future work should consider this issue. The body of the work contains detailed analysis of the various impedance estimations and instability threshold calculations. The calculations are based on the Main Injector beam intensity of 6 x 1010 protons per bunch, 95% normalized transverse emittances of 20π mm-mrad, and 95% normalized longitudinal emittance of 0.1 eV-s at 8.9 GeV injection energy and 0.25 eV-s at 150 GeV flattop energy. The conclusions section summarizes the results in the paper and is meant to be readable by itself without referring to the rest of the paper. Also in the conclusion section are recommendations for future investigations
Electrode-less measurement of cell layers impedance
Czech Academy of Sciences Publication Activity Database
Krůšek, Jan; Ďaďo, S.
2014-01-01
Roč. 63, č. 6 (2014), s. 705-711. ISSN 0862-8408 R&D Projects: GA ČR(CZ) GBP304/12/G069 Institutional support: RVO:67985823 Keywords : cell impedance measurement * transepithelial resistance Subject RIV: ED - Physiology Impact factor: 1.293, year: 2014
Impedance characteristics of the Bz diode on the LION accelerator
International Nuclear Information System (INIS)
The LION accelerator at Cornell University is being used to study the characteristics of the applied B/sub z/, or 'barrel' diode. This 0.8 TW, 4 ohm, ion accelerator has the ability to take several shots per day, and hence alloys systematic scans to be performed. An important result of a recent series of experiments is that the diode impedance remains relatively constant, decaying only slowly, during the 50 nsec pulse. When the diode is operated with a 4.5 mm gap and a 21 kG insulating magnetic field, the typical diode parameters, are a voltage of 1 MV and a total current of 250 kA, leading to a diode impedance of 4 ohms and power of 0.25 TW. The diode impedance decays with a 100 nsec time constant. The ion beams have peak currents of roughly 125 kA and typical impedances of Bohms, which decays with a time constant of 25 nsec. The Child-Langmuir gap was approximately 2 mm and closed with a velocity of roughly 2X10/sup 6/ cm/sec. Current experimental work is aimed at characterizing the impedance of the B/sub z/ diode as a function of the applied magnetic field, the A-K gap, the anode curvature, and the anode groove parameters. In addition, the effect of changing the voltage rise with a plasma opening switch and of adding an electron limiter is examined. The ion beam quality is examined at the focus of the barrel diode with a swept Thomson parabola and various Rutherford scattering diagnostics
Reflections on the history of electrochemical impedance spectroscopy
International Nuclear Information System (INIS)
The history of electrochemical impedance spectroscopy (EIS) is briefly reviewed, starting with the foundations laid by Heaviside in the late 19th century in the form of Linear Systems Theory (LST). Warburg apparently was the first to extend the concept of impedance to electrochemical systems at the turn of the 19th century, when he derived the impedance function for a diffusional process that still bears his name. Impedance spectroscopy was next employed extensively using reactive bridges to measure the capacitance of ideally polarizable electrodes (mostly mercury), leading to the development of models for the electrified interface. However, it was the invention of the potentiostat in the 1940s and the development of frequency response analyzers in the 1970s that led to the use of EIS in exploring electrochemical and corrosion mechanisms, primarily because of their ability to probe electrochemical systems at very low frequencies. These inventions have led to an explosion in the use of EIS for exploring a wide range of systems and processes, ranging from conduction in the solid and liquid states, ionic and electronic conduction in polymers, heterogeneous reaction mechanisms, and the important phenomenon of passivity. It is evident that the use of EIS in identifying reaction mechanisms makes use of pattern recognition, currently through inspection. It is argued that, in the future development of EIS, reaction mechanism analysis (RMA) would be most efficiently done by using artificial neural networks operating in the pattern recognition mode. This strategy would require the creation of libraries of reaction mechanisms for which the theoretical impedance functions are known
Impedance Analysis of Silicon Nanowire Lithium Ion Battery Anodes
Ruffo, Riccardo
2009-07-02
The impedance behavior of silicon nanowire electrodes has been investigated to understand the electrochemical process kinetics that influences the performance when used as a high-capacity anode in a lithium ion battery. The ac response was measured by using impedance spectroscopy in equilibrium conditions at different lithium compositions and during several cycles of charge and discharge in a half cell vs. metallic lithium. The impedance analysis shows the contribution of both surface resistance and solid state diffusion through the bulk of the nanowires. The surface process is dominated by a solid electrolyte layer (SEI) consisting of an inner, inorganic insoluble part and several organic compounds at the outer interface, as seen by XPS analysis. The surface resistivity, which seems to be correlated with the Coulombic efficiency of the electrode, grows at very high lithium contents due to an increase in the inorganic SEI thickness. We estimate the diffusion coefficient of about 2 × 10 -10 cm 2/s for lithium diffusion in silicon. A large increase in the electrode impedance was observed at very low lithium compositions, probably due to a different mechanism for lithium diffusion inside the wires. Restricting the discharge voltage to 0.7 V prevents this large impedance and improves the electrode lifetime. Cells cycled between 0.07 and 0.70 V vs. metallic lithium at a current density of 0.84 A/g (C/5) showed good Coulombic efficiency (about 99%) and maintained a capacity of about 2000 mAh/g after 80 cycles. © 2009 American Chemical Society.
Dynamic impedance of piles in visco-elastic material considering axial loads
Institute of Scientific and Technical Information of China (English)
JIANG Jian-guo; ZHOU Xu-hong; ZHANG Jia-sheng
2005-01-01
The dynamic impedance function of pile in visco-elastie material considering axial loads under lateral dynamic force was analyzed, and the beam dynamic differential equation was used to induce the dynamic impedance function. After analyzing the edge conditions, the dynamic impedance functions were deduced. Contrasted with the result that does not consider axial loads, the axial loads have obvious influence on the dynamic impedance function.And the results show that the dimensionless prarmeter of the dynamic impedance will change from 6 % to 9 % when considering axial loads, and dimensionless prarmeter of the dynamic impedance of the coupling horizontal-sway will increase by 31 %.
Modern Trends in Imaging XI: Impedance Measurements in the Biomedical Sciences
Directory of Open Access Journals (Sweden)
Frederick D. Coffman
2012-01-01
Full Text Available Biological organisms and their component organs, tissues and cells have unique electrical impedance properties. Impedance properties often change with changes in structure, composition, and metabolism, and can be indicative of the onset and progression of disease states. Over the past 100 years, instruments and analytical methods have been developed to measure the impedance properties of biological specimens and to utilize these measurements in both clinical and basic science settings. This chapter will review the applications of impedance measurements in the biomedical sciences, from whole body analysis to impedance measurements of single cells and cell monolayers, and how cellular impedance measuring instruments can now be used in high throughput screening applications.
Energy Technology Data Exchange (ETDEWEB)
Petracca, S. [Salerno Univ. (Italy)
1996-08-01
Debye potentials, the Lorentz reciprocity theorem, and (extended) Leontovich boundary conditions can be used to obtain simple and accurate analytic estimates of the longitudinal and transverse coupling impedances of (piecewise longitudinally uniform) multi-layered pipes with non simple transverse geometry and/or (spatially inhomogeneous) boundary conditions. (author)
International Nuclear Information System (INIS)
The device performances of organic thin film transistors are often limited by the metal–organic interface because of the disordered molecular layers at the interface and the energy barriers against the carrier injection. It is important to study the local impedance at the interface without being affected by the interface morphology. We combined frequency modulation atomic force microscopy with scanning impedance microscopy (SIM) to sensitively measure the ac responses of the interface to an ac voltage applied across the interface and the dc potential drop at the interface. By using the frequency-modulation SIM (FM-SIM) technique, we characterized the interface impedance of a Pt electrode and a single pentacene grain as a parallel circuit of a contact resistance and a capacitance. We found that the reduction of the contact resistance was caused by the reduction of the energy level mismatch at the interface by the FM-SIM measurements, demonstrating the usefulness of the FM-SIM technique for investigation of the local interface impedance without being affected by its morphology
Kimura, Tomoharu; Kobayashi, Kei; Yamada, Hirofumi
2015-08-01
The device performances of organic thin film transistors are often limited by the metal-organic interface because of the disordered molecular layers at the interface and the energy barriers against the carrier injection. It is important to study the local impedance at the interface without being affected by the interface morphology. We combined frequency modulation atomic force microscopy with scanning impedance microscopy (SIM) to sensitively measure the ac responses of the interface to an ac voltage applied across the interface and the dc potential drop at the interface. By using the frequency-modulation SIM (FM-SIM) technique, we characterized the interface impedance of a Pt electrode and a single pentacene grain as a parallel circuit of a contact resistance and a capacitance. We found that the reduction of the contact resistance was caused by the reduction of the energy level mismatch at the interface by the FM-SIM measurements, demonstrating the usefulness of the FM-SIM technique for investigation of the local interface impedance without being affected by its morphology.
Pliquett, Uwe
2013-04-01
Over recent years advanced measurement methods have facilitated outstanding achievements not only in medical instrumentation but also in biotechnology. Impedance measurement is a simple and innocuous way to characterize materials. For more than 40 years biological materials, most of them based on cells, have been characterized by means of electrical impedance for quality control of agricultural products, monitoring of biotechnological or food processes or in health care. Although the list of possible applications is long, very few applications successfully entered the market before the turn of the century. This was, on the one hand, due to the low specificity of electrical impedance with respect to other material properties because it is influenced by multiple factors. On the other hand, equipment and methods for many potential applications were not available. With the appearance of microcontrollers that could be easily integrated in applications at the beginning of the 1980s, impedance measurement advanced as a valuable tool in process optimization and lab automation. However, established methods and data processing were mostly used in a new environment. This has changed significantly during the last 10 years with a dramatic growth of the market for medical instrumentation and also for biotechnological applications. Today, advanced process monitoring and control require fast and highly parallel electrical characterization which in turn yields incredible data volumes that must be handled in real time. Many newer developments require miniaturized but precise sensing methods which is one of the main parts of Lab-on-Chip technology. Moreover, biosensors increasingly use impedometric transducers, which are not compatible with the large expensive measurement devices that are common in the laboratory environment. Following the achievements in the field of bioimpedance measurement, we will now witness a dramatic development of new electrode structures and electronics
Micro-Horn Arrays for Ultrasonic Impedance Matching
Rao, Shanti; Palmer, Dean
2009-01-01
Thin-layered structures containing arrays of micromachined horns, denoted solid micro-horn arrays (SMIHAs), have been conceived as improved means of matching acoustic impedances between ultrasonic transducers and the media with which the transducers are required to exchange acoustic energy. Typically, ultrasonic transducers (e.g., those used in medical imaging) are piezoelectric or similar devices, which produce small displacements at large stresses. However, larger displacements at smaller stresses are required in the target media (e.g., human tissues) with which acoustic energy is to be exchanged. Heretofore, efficiencies in transmission of acoustic energy between ultrasonic transducers and target media have been severely limited because substantial mismatches of acoustic impedances have remained, even when coupling material layers have been interposed between the transducers and the target media. In contrast, SMIHAs can, in principle, be designed to effect more nearly complete acoustic impedance matching, leading to power transmission efficiencies of 90 percent or even greater. The SMIHA concept is based on extension, into the higher-frequency/ lower-wavelength ultrasonic range, of the use of horns to match acoustic impedances in the audible and lower-frequency ultrasonic ranges. In matching acoustic impedance in transmission from a higher-impedance acoustic source (e.g., a piezoelectric transducer) and a lowerimpedance target medium (e.g., air or human tissue), a horn acts as a mechanical amplifier. The shape and size of the horn can be optimized for matching acoustic impedance in a specified frequency range. A typical SMIHA would consist of a base plate, a face plate, and an array of horns that would constitute pillars that connect the two plates (see figure). In use, the base plate would be connected to an ultrasonic transducer and the face plate would be placed in contact with the target medium. As at lower frequencies, the sizes and shapes of the pillars
International Nuclear Information System (INIS)
Pulmonary vascular input impedance better characterizes right ventricular (RV) afterload and disease outcomes in pulmonary hypertension compared to the standard clinical diagnostic, pulmonary vascular resistance (PVR). Early efforts to measure impedance were not routine, involving open-chest measurement. Recently, the use of pulsed-wave (PW) Doppler-measured velocity to non-invasively estimate instantaneous flow has made impedance measurement more practical. One critical concern remains with clinical use: the measurement uncertainty, especially since previous studies only incorporated random error. This study utilized data from a large pediatric patient population to comprehensively examine the systematic and random error contributions to the total impedance uncertainty and determined the least error prone methodology to compute impedance from among four different methods. We found that the systematic error contributes greatly to the total uncertainty and that one of the four methods had significantly smaller propagated uncertainty; however, even when this best method is used, the uncertainty can be large for input impedance at high harmonics and for the characteristic impedance modulus. Finally, we found that uncertainty in impedance between normotensive and hypertensive patient groups displays no significant difference. It is concluded that clinical impedance measurement would be most improved by advancements in instrumentation, and the best computation method is proposed for future clinical use of the input impedance
Measurement of the longitudinal coupling impedance in the Brookhaven AGS
International Nuclear Information System (INIS)
The imaginary part of the longitudinal coupling impedance has been measured as a function of energy from 5 to approx. 28 GeV. This impedance is proportional to Δf = (f/sub q/--2f/sub d/) where f/sub d/ is the coherent dipole frequency and f/sub q/ the coherent quadrupole frequency. These frequencies are obtained by stimulating coupled bunch oscillations. If the dominant impedance is due to inductive wall plus space charge effects, then one has (Z/n) = j[Ω0L--g0Z0/2BETAγ.2]1 where L is the inductance per turn and Ω0 = 2πf0 the particle rotation frequency. The expression (Z/n) = 4jΔfπ2hV0cos phi/sub s/B3/3I0f/sub d/ can be used to find the impedance if the synchrotron phase space distribution is proportional to (1--r2)/sub 1/2/. I0 is the current per bunch, B = f0 x tau0 the bunch length and V0 is the external voltage. For a distribution given by (1--r2) the right hand side should be multiplied by 27/4π2. If the latter is assumed, an inductive impedance of 20.4 Ω is obtained with a null at approx. = 6 Gev (γ/sub tr/ = 8.5) for a transverse emittance of 22 πμ rad-m. At 5 GeV the reactance is negative but larger than the simple relation assumed for (Z/n) would predict. If the bunches are parabolic, then the inductive impedance would be 29.7 Ω with a null again at 6.6 GeV but only for an emittance of 2.5 μrad-m. Again the 5 GeV reactance is much too large. The significance of these results is discussed
Wave impedances of drill strings and other periodic media
Drumheller, Douglas S.
2002-12-01
It is commonly known that wave reflections are caused by abrupt spatial variations in the physical parameter called wave impedance. When a material contains a spatially periodic distribution of wave impedances some very interesting and complex wave propagation phenomena will occur. Two examples of such periodic structures immediately come to mind: the first is a sandwiched structure of two types of plates, say for example, identical layers of thin steel plates interspersed with identical thick aluminum plates; and the second is a large number of identical long thin pipes that are connected from end to end with identical short heavy threaded couplings. The pipe assembly is our primary concern here because it represents the drill string, used worldwide to drill for natural energy resources. We want to understand how waves propagate through drill strings because we want to use them as a means of communication. But while the second structure is our primary concern, it is the study of the first structure, composed of layers, that is the truly historical problem and the source of much of our understanding of this rich set of wave physics. Traditionally, wave propagation in periodic media has been studied as an eigenvalue problem. The eigenvalues themselves yield information about phase velocities, group velocities, passbands, and stopbands. Most often the analysis has stopped there and the eigenvectors have been ignored. Here we turn our attention to the eigenvectors, using them to evaluate the impedance of the periodic structure with particular emphasis on the periodic drill string. As you might expect the impedance of the drill string is a complex number, which is evaluated from a very complicated expression. However, we have discovered that the impedance at two physical locations along the length of each piece of drill pipe in the drill string always reduces to a real number. This is immensely important because it allows us to match the impedance of the drill string
Architecture, modeling, and analysis of a plasma impedance probe
Jayaram, Magathi
Variations in ionospheric plasma density can cause large amplitude and phase changes in the radio waves passing through this region. Ionospheric weather can have detrimental effects on several communication systems, including radars, navigation systems such as the Global Positioning Sytem (GPS), and high-frequency communications. As a result, creating models of the ionospheric density is of paramount interest to scientists working in the field of satellite communication. Numerous empirical and theoretical models have been developed to study the upper atmosphere climatology and weather. Multiple measurements of plasma density over a region are of marked importance while creating these models. The lack of spatially distributed observations in the upper atmosphere is currently a major limitation in space weather research. A constellation of CubeSat platforms would be ideal to take such distributed measurements. The use of miniaturized instruments that can be accommodated on small satellites, such as CubeSats, would be key to achieving these science goals for space weather. The accepted instrumentation techniques for measuring the electron density are the Langmuir probes and the Plasma Impedance Probe (PIP). While Langmuir probes are able to provide higher resolution measurements of relative electron density, the Plasma Impedance Probes provide absolute electron density measurements irrespective of spacecraft charging. The central goal of this dissertation is to develop an integrated architecture for the PIP that will enable space weather research from CubeSat platforms. The proposed PIP chip integrates all of the major analog and mixed-signal components needed to perform swept-frequency impedance measurements. The design's primary innovation is the integration of matched Analog-to-Digital Converters (ADC) on a single chip for sampling the probes current and voltage signals. A Fast Fourier Transform (FFT) is performed by an off-chip Field-Programmable Gate Array (FPGA
A method to separate process contributions in impedance spectra by variation of test conditions
DEFF Research Database (Denmark)
Jensen, Søren Højgaard; Hauch, Anne; Hendriksen, Peter Vang; Mogensen, Mogens Bjerg; Bonanos, Nikolaos; Jacobsen, Torben
2007-01-01
Many processes contribute to the overall impedance of an electrochemical cell, and these may be difficult to separate in the impedance spectrum. Here, we present an investigation of a solid oxide fuel cell based on differences in impedance spectra due to a change of operating parameters and present...... the result as the derivative of the impedance with respect to ln(f). The method is used to separate the anode and cathode contributions and to identify various types of processes....
DEFF Research Database (Denmark)
Kim, Kseniya; Zhurbenko, Vitaliy; Johansen, Tom Keinicke;
2012-01-01
Nonsynchronous noncommensurate impedance transformers consist of a combination of high- and low-impedance transmission lines. The present work is dedicated to the synthesis of such transformers. They are analyzed employing wave transmission matrices. The performance of the transformer is compared...... to a traditional tapered line impedance transformer. The increase in bandwidth of nonsynchronous noncommensurate impedance transformers typically leads to shortening the transformer length, which makes the transformer attractive for applications, where a wide operating band and high transformation...
High accuracy particle analysis using sheathless microfluidic impedance cytometry.
Spencer, Daniel; Caselli, Federica; Bisegna, Paolo; Morgan, Hywel
2016-07-01
This paper describes a new design of microfluidic impedance cytometer enabling accurate characterization of particles without the need for focusing. The approach uses multiple pairs of electrodes to measure the transit time of particles through the device in two simultaneous different current measurements, a transverse (top to bottom) current and an oblique current. This gives a new metric that can be used to estimate the vertical position of the particle trajectory through the microchannel. This parameter effectively compensates for the non-uniform electric field in the channel that is an unavoidable consequence of the use of planar parallel facing electrodes. The new technique is explained and validated using numerical modelling. Impedance data for 5, 6 and 7 μm particles are collected and compared with simulations. The method gives excellent coefficient of variation in (electrical) radius of particles of 1% for a sheathless configuration. PMID:27241585
Gemstone Grinding Process Improvement by using Impedance Force Control
Directory of Open Access Journals (Sweden)
Hamprommarat Chumpol
2015-01-01
Full Text Available Chula Automatic Faceting Machine has been developed by The Advance Manufacturing Research Lab, Chulalongkorn University to support Thailand Gems-Industry. The machine has high precision motion control by using position and force control. A contact stiffness model is used to estimate grinding force. Although polished gems from the Faceting Machine have uniform size and acceptable shape, the force of the grinding and polishing process cannot be maintain constant and has some fluctuation due to indirect force control. Therefor this research work propose a new controller for this process based on an impedance direct force control to improve the gemstone grinding performance during polishing process. The grinding force can be measured through motor current. The results show that the polished gems by using impedance direct force control can maintain uniform size as well as good shape and high quality surface.
Acoustic Wavefront Manipulation: Impedance Inhomogeneity and Extraordinary Reflection
Zhao, Jiajun; Chen, Zhining; Li, Baowen
2013-01-01
Optical wavefront can be manipulated by interfering elementary beams with phase inhomogeneity. Therefore a surface allowing huge, abrupt and position-variant phase change would enable all possibilities of wavefront engineering. However, one may not have the luxury of efficient abrupt-phase-changing materials in acoustics. This motivates us to establish a counterpart mechanism for acoustics, in order to empower the wide spectrum of novel acoustic applications. Remarkably, the proposed impedance-governed generalized Snell's law (IGSL) of reflection is distinguished from that in optics. Via the manipulation of inhomogeneous acoustic impedance, extraordinary reflection can be tailored for unprecedented wavefront manipulation while ordinary reflection can be surprisingly switched on or off. Our results may power the acoustic-wave manipulation and engineering. We demonstrate novel acoustic applications by planar surfaces designed with IGSL.
Impedance function of a group of vertical piles
International Nuclear Information System (INIS)
Impedance and transfer functions of a group of vertical piles located in any desired configuration in plan in a horizontally stratified soil layer are derived. Hysteretic and radiation damping are accounted for. The method separates the piles and the soil, introducing unknown interaction forces. The total flexibility matrix of the soil is constructed, superposing the (complex) flexibility coefficients caused by the interaction forces of a single pile only. The dependence of the impedance and transfer functions on the oscllating frequency for foundations with different numbers of piles is investigated. Pile-soil-pile interaction is shown to be very important for all modes of vibration. The procedure is used in the seismic analysis of a reactor building. (Author)
Traceable calibration of impedance heads and artificial mastoids
Scott, D. A.; Dickinson, L. P.; Bell, T. J.
2015-12-01
Artificial mastoids are devices which simulate the mechanical characteristics of the human head, and in particular of the bony structure behind the ear. They are an essential tool in the calibration of bone-conduction hearing aids and audiometers. With the emergence of different types of artificial mastoids in the market, and the realisation that the visco-elastic part of these instruments changes over time, the development of a method of traceable calibration of these devices without relying on commercial software has become important for national metrology institutes. This paper describes commercially available calibration methods, and the development of a traceable calibration method including the traceable calibration of the impedance head used to measure the mechanical impedance of the artificial mastoid.
Traceable calibration of impedance heads and artificial mastoids
International Nuclear Information System (INIS)
Artificial mastoids are devices which simulate the mechanical characteristics of the human head, and in particular of the bony structure behind the ear. They are an essential tool in the calibration of bone-conduction hearing aids and audiometers. With the emergence of different types of artificial mastoids in the market, and the realisation that the visco-elastic part of these instruments changes over time, the development of a method of traceable calibration of these devices without relying on commercial software has become important for national metrology institutes. This paper describes commercially available calibration methods, and the development of a traceable calibration method including the traceable calibration of the impedance head used to measure the mechanical impedance of the artificial mastoid. (paper)
High resolution impedance manometric findings in dysphagia of Huntington's disease
Institute of Scientific and Technical Information of China (English)
Tae Hee Lee; Joon Seong Lee; Wan Jung Kim
2012-01-01
Conventional manometry presents significant challenges,espedally in assessment of pharyngeal swallowing,because of the asymmetry and deglutitive movements of oropharyngeal structures.It only provides information about intraluminal pressure and thus it is difficult to study functional details of esophageal motility disorders.New technology of solid high resolution impedance manometry (HRIM),with 32 pressure sensors and 6 impedance sensors,is likely to provide better assessment of pharyngeal swallowing as well as more information about esophageal motility disorders.However,the clinical usefulness of application of HRIM in patients with oropharyngeal dysphagia or esophageal dysphagia is not known.We experienced a case of Huntington's disease presenting with both oropharyngeal and esophageal dysphagia,in which HRIM revealed the mechanism of oropharyngeal dysphagia and provided comprehensive information about esophageal dysphagia.
Structural damage identification using piezoelectric impedance and Bayesian inference
Shuai, Q.; Zhou, K.; Tang, J.
2015-04-01
Structural damage identification is a challenging subject in the structural health monitoring research. The piezoelectric impedance-based damage identification, which usually utilizes the matrix inverse-based optimization, may in theory identify the damage location and damage severity. However, the sensitivity matrix is oftentimes ill-conditioned in practice, since the number of unknowns may far exceed the useful measurements/inputs. In this research, a new method based on intelligent inference framework for damage identification is presented. Bayesian inference is used to directly predict damage location and severity using impedance measurement through forward prediction and comparison. Gaussian process is employed to enrich the forward analysis result, thereby reducing computational cost. Case study is carried out to illustrate the identification performance.
Dielectrophoresis and dielectrophoretic impedance detection of adenovirus and rotavirus
Nakano, Michihiko; Ding, Zhenhao; Suehiro, Junya
2016-01-01
The aim of this study is the electrical detection of pathogenic viruses, namely, adenovirus and rotavirus, using dielectrophoretic impedance measurement (DEPIM). DEPIM consists of two simultaneous processes: dielectrophoretic trapping of the target and measurement of the impedance change and increase in conductance with the number of trapped targets. This is the first study of applying DEPIM, which was originally developed to detect bacteria suspended in aqueous solutions, to virus detection. The dielectric properties of the viruses were also investigated in terms of their dielectrophoretic behavior. Although their estimated dielectric properties were different from those of bacteria, the trapped viruses increased the conductance of the microelectrode in a manner similar to that in bacteria detection. We demonstrated the electrical detection of viruses within 60 s at concentrations as low as 70 ng/ml for adenovirus and 50 ng/ml for rotavirus.
A CMOS Electrochemical Impedance Spectroscopy (EIS) Biosensor Array.
Manickam, Arun; Chevalier, Aaron; McDermott, Mark; Ellington, Andrew D; Hassibi, Arjang
2010-12-01
In this paper, we present a fully integrated biosensor 10 × 10 array in a standard complementary metal-oxide semiconducor process, which takes advantage of electrochemical impedance spectroscopy (EIS). We also show that this system is able to detect various biological analytes, such as DNA and proteins, in real time and without the need for molecular labels. In each pixel of this array, we implement a biocompatible Au electrode transducer and embedded sensor circuitry which takes advantage of the coherent detector to measure the impedance of the associated electrode-electrolyte interface. This chip is capable of concurrently measuring admittance values as small as 10(-8) Ω(-1) within the array with the detection dynamic range of more than 90 dB in the frequency range of 10 Hz-50 MHz. PMID:23850755
Time-domain based impedance measurement: strengths and drawbacks
International Nuclear Information System (INIS)
While most applications of bio-impedance measurements for characterization of cells or tissue do not have any time requirements, newer developments like real time monitoring of moving cells or membrane recovery after high voltage application depend on fast measurement. This is not compatible with sweeping the frequency through the desired range while assessing magnitude and phase of the material under test (MUT). A high speed is only achievable by using broad-bandwidth excitation signals and monitoring the response in time-domain. Time-domain based methods can be distinguished by the excitation signals as well as by assessing the transmission or reflection behavior of the MUT. Although there is good agreement regarding the advantages of fast measurements, time-domain measurements are often rejected because of low precision and noise sensitivity. This paper points not only the advantages of impedance measurements in time domain but shows also drawbacks together with possible solutions.
Impedance of tissue-mimicking phantom material under compression
Directory of Open Access Journals (Sweden)
Barry Belmont
2013-02-01
Full Text Available The bioimpedance of tissues under compression is a field in need of study. While biological tissues can become compressed in a myriad of ways, very few experiments have been conducted to describe the relationship between the passive electrical properties of a material (impedance/admittance during mechanical deformation. Of the investigations that have been conducted, the exodus of fluid from samples under compression has been thought to be the cause of changes in impedance, though until now was not measured directly. Using a soft tissue-mimicking phantom material (tofu whose passive electrical properties are a function of the conducting fluid held within its porous structure, we have shown that the mechanical behavior of a sample under compression can be measured through bioimpedance techniques.
Experimental Impedance of Single Liner Elements with Bias Flow
Follet, J. I.; Betts, J. F.; Kelly, Jeffrey J.; Thomas, Russell H.
2000-01-01
An experimental investigation was conducted to generate a high quality database, from which the effects of a mean bias flow on the acoustic impedance of lumped-element single-degree-of-freedom liners was determined. Acoustic impedance measurements were made using the standard two-microphone method in the NASA Langley Normal Incidence Tube. Each liner consisted of a perforated sheet with a constant-area cavity. Liner resistance was shown to increase and to become less frequency and sound pressure level dependent as the bias flow was increased. The resistance was also consistently lower for a negative bias flow (suction) than for a positive bias flow (blowing) of equal magnitude. The slope of the liner reactance decreased with increased flow.
Beam steering and impedance matching of plasmonic horn nanoantennas
Afridi, Adeel
2016-01-01
In this paper, we study a plasmonic horn nanoantenna on a metal-backed substrate. The horn nanoantenna structure consists of a two-wire transmission line (TWTL) flared at the end. We analyze the effect of the substrate thickness on the nanoantenna's radiation pattern, and demonstrate beam steering in a broad range of elevation angles. Furthermore, we analyze the effect of the ground plane on the impedance matching between the antenna and the TWTL, and observe that the ground plane increases the back reflection into the waveguide. To reduce the reflection, we develop a transmission line model to design an impedance matching section which leads to 99.75% power transmission to the nanoantenna.
On accurate differential measurements with electrochemical impedance spectroscopy
Kernbach, S; Kernbach, O
2016-01-01
This paper describes the impedance spectroscopy adapted for analysis of small electrochemical changes in fluids. To increase accuracy of measurements the differential approach with temperature stabilization of fluid samples and electronics is used. The impedance analysis is performed by the single point DFT, signal correlation, calculation of RMS amplitudes and interference phase shift. For test purposes the samples of liquids and colloids are treated by fully shielded electromagnetic generators and passive cone-shaped structures. Fluidic samples collected from different geological locations are also analysed. In all tested cases we obtained different results for impacted and non-impacted samples, moreover, a degradation of electrochemical stability after treatment is observed. This method is used in laboratory analysis of weak emissions and ensures a high repeatability of results.
Development of a radio frequency excited local impedance probe
International Nuclear Information System (INIS)
Local void fraction measurements were made with a Karlsruhe type impedance probe. The probe was operated at radiofrequency to minimize sensitivity to liquid phase resistivity. Two types of signal thresholding were used: level and derivative. A dual beam X-ray system was used as a calibration standard for the radio frequency excited impedance probe. Calibration was performed in vertical air/water flows. Derivative thresholding was found to be preferable to level thresholding, however, in both schemes hydrodynamic and surface tension effects were observed below a liquid superficial velocity of 0.5 m/s. Table salt (NaCl) was added to the water to verify the probe's response to changing water resistivity. Derivative thresholding appeared to work quite well but level thresholding was found to be inadequate due to the change in capacitance. (orig.)
An electrochemical impedance model for integrated bacterial biofilms
International Nuclear Information System (INIS)
Bacterial cells attachment onto solid surfaces and the following growth into mature microbial biofilms may result in highly antibiotic resistant biofilms. Such biofilms may be incidentally formed on tissues or implanted devices, or intentionally formed by directed deposition of microbial sensors on whole-cell bio-chip surface. A new method for electrical characterization of the later on-chip microbial biofilm buildup is presented in this paper. Measurement of impedance vs. frequency in the range of 100 mHz to 400 kHz of Escherichia coli cells attachment to indium-tin-oxide-coated electrodes was carried out while using optical microscopy estimating the electrode area coverage. We show that impedance spectroscopy measurements can be interpreted by a simple electrical equivalent model characterizing both attachment and growth of the biofilm. The correlation of extracted equivalent electrical lumped components with the visual biofilm parameters and their dependence on the attachment and growth phases is confirmed.
Permeability deduced from impedance measurements at microwave frequencies
International Nuclear Information System (INIS)
We present both circumferential, μphi, and parallel, μparallel, permeabilities of a Co68.15Fe4.35Si12.5B15 amorphous wire. We determined μphi from impedance measurements up to 10 GHz using a network analyzer with the wire as the central conductor of a coaxial line. We determined μparallel from impedance variations up to 1 GHz with the wire mounted as the core of a conductive loop driven by a microwave signal of a network analyzer. We report here on the design of the sample holders, on the measurements and on the methods used for their analysis, including the details of the effect of sample mounting on the results
Bench measurements of coupling impedance of AGS Booster components
International Nuclear Information System (INIS)
Quantifying instability thresholds for modern synchrotrons and storage rings requires some knowledge of the accelerator's coupling impedance. To this end, the wire technique has been implemented to measure the longitudinal coupling impedance of AGS Booster devices. The techniques are being refined to allow measurement of RHIC devices at higher frequencies. All the measurements are performed using an HP 8753 Network Analyzer controlled via GPIB by a Macintosh computer. The computer provides an environment for automated data acquisition, data analysis, and report generation. Resistive matches between the 50ω analyzer cables and the 300ω pipe-and-wire structure allow the use of a simple response calibration in the measurement of S21 to 400MHz. Results from ferrite loaded rf cavities, position monitors and kickers are presented. 4 refs., 4 figs
Determination of complex microcalorimeter parameters with impedance measurements
International Nuclear Information System (INIS)
The proper understanding and modeling of a microcalorimeter's response requires accurate knowledge of a handful of parameters, such as C, G, α. While a few of these parameters are directly determined from the IV characteristics, some others, notoriously the heat capacity (C) and α, appear in degenerate combinations in most measurable quantities. The consideration of a complex microcalorimeter leads to an added ambiguity in the determination of the parameters. In general, the dependence of the microcalorimeter's complex impedance on these various parameters varies with frequency. This dependence allows us to determine individual parameters by fitting the prediction of the microcalorimeter model to impedance data. In this paper we describe efforts at characterizing the Goddard X-ray microcalorimeters. With the parameters determined by this method, we compare the pulse shape and noise spectra predictions to data taken with the same devices
Meandered-line antenna with integrated high-impedance surface.
Energy Technology Data Exchange (ETDEWEB)
Forman, Michael A.
2010-09-01
A reduced-volume antenna composed of a meandered-line dipole antenna over a finite-width, high-impedance surface is presented. The structure is novel in that the high-impedance surface is implemented with four Sievenpiper via-mushroom unit cells, whose area is optimized to match the meandered-line dipole antenna. The result is an antenna similar in performance to patch antenna but one fourth the area that can be deployed directly on the surface of a conductor. Simulations demonstrate a 3.5 cm ({lambda}/4) square antenna with a bandwidth of 4% and a gain of 4.8 dBi at 2.5 GHz.
Impedance matching wireless power transmission system for biomedical devices.
Lum, Kin Yun; Lindén, Maria; Tan, Tian Swee
2015-01-01
For medical application, the efficiency and transmission distance of the wireless power transfer (WPT) are always the main concern. Research has been showing that the impedance matching is one of the critical factors for dealing with the problem. However, there is not much work performed taking both the source and load sides into consideration. Both sides matching is crucial in achieving an optimum overall performance, and the present work proposes a circuit model analysis for design and implementation. The proposed technique was validated against experiment and software simulation. Result was showing an improvement in transmission distance up to 6 times, and efficiency at this transmission distance had been improved up to 7 times as compared to the impedance mismatch system. The system had demonstrated a near-constant transfer efficiency for an operating range of 2cm-12cm. PMID:25980873
Nuclear radiation-warning detector that measures impedance
Savignac, Noel Felix; Gomez, Leo S; Yelton, William Graham; Robinson, Alex; Limmer, Steven
2013-06-04
This invention is a nuclear radiation-warning detector that measures impedance of silver-silver halide on an interdigitated electrode to detect light or radiation comprised of alpha particles, beta particles, gamma rays, X rays, and/or neutrons. The detector is comprised of an interdigitated electrode covered by a layer of silver halide. After exposure to alpha particles, beta particles, X rays, gamma rays, neutron radiation, or light, the silver halide is reduced to silver in the presence of a reducing solution. The change from the high electrical resistance (impedance) of silver halide to the low resistance of silver provides the radiation warning that detected radiation levels exceed a predetermined radiation dose threshold.
Robust recursive impedance estimation for automotive lithium-ion batteries
Fridholm, Björn; Wik, Torsten; Nilsson, Magnus
2016-02-01
Recursive algorithms, such as recursive least squares (RLS) or Kalman filters, are commonly used in battery management systems to estimate the electrical impedance of the battery cell. However, these algorithms can in some cases run into problems with bias and even divergence of the estimates. This article illuminates problems that can arise in the online estimation using recursive methods, and lists modifications to handle these issues. An algorithm is also proposed that estimates the impedance by separating the problem in two parts; one estimating the ohmic resistance with an RLS approach, and another one where the dynamic effects are estimated using an adaptive Kalman filter (AKF) that is novel in the battery field. The algorithm produces robust estimates of ohmic resistance and time constant of the battery cell in closed loop with SoC estimation, as demonstrated by both in simulations and with experimental data from a lithium-ion battery cell.
Evaluation of the broadband longitudinal impedance of the CERN PS
Damerau, H; Hancock, S; Migliorati, M; Palumbo, L; Persichelli, S; CERN. Geneva. ATS Department
2012-01-01
The CERN Proton Synchrotron (PS) produces beams well beyond its original parameters and especially the high-brightness beams delivered to the Large Hadron Collider are close to present stability limitations. Within the framework of the LHC Injectors Upgrade project, a signicant increase in beam intensity is planned. With this background, collective effects triggered by the self-induced electromagnetic elds, which are generated by the interaction of the beam with the surrounding environment, may play an important role in beam stability and machine performance. In this paper we study the short range wakeelds and the corresponding longitudinal broadband coupling impedance of the PS by measuring the incoherent quadrupole synchrotron frequency as a function of beam intensity and comparing the results with the evaluation of the contribution of several machine installations to the total impedance budget.
Object impedance control for cooperative manipulation - Theory and experimental results
Schneider, Stanley A.; Cannon, Robert H., Jr.
1992-01-01
This paper presents the dynamic control module of the Dynamic and Strategic Control of Cooperating Manipulators (DASCCOM) project at Stanford University's Aerospace Robotics Laboratory. First, the cooperative manipulation problem is analyzed from a systems perspective, and the desirable features of a control system for cooperative manipulation are discussed. Next, a control policy is developed that enforces a controlled impedance not of the individual arm endpoints, but of the manipulated object itself. A parallel implementation for a multiprocessor system is presented. The controller fully compensates for the system dynamics and directly controls the object internal forces. Most importantly, it presents a simple, powerful, intuitive interface to higher level strategic control modules. Experimental results from a dual two-link-arm robotic system are used to compare the object impedance controller with other strategies, both for free-motion slews and environmental contact.
Design and Analysis of Impedance Pumps Utilizing Electromagnetic Actuation
Yu-Hisang Wang; Yao-Wen Tsai; Chien-Hsiung Tsai; Chia-Yen Lee; Lung-Ming Fu
2010-01-01
This study designs and analyzes an impedance pump utilizing an electromagnetic actuator. The pump is designed to have three major components, namely a lower glass substrate patterned with a copper micro-coil, a microchannel, and an upper glass cover plate attached a magnetic PDMS diaphragm. When a current is passed through the micro-coil, an electromagnetic force is established between the coil and the magnetic diaphragm. The resulting deflection of the PDMS diaphragm creates an acoustic impe...
Impedance microflow cytometry for viability studies of microorganisms
Di Berardino, Marco; Hebeisen, Monika; Hessler, Thomas; Ziswiler, Adrian; Largiadèr, Stephanie; Schade, Grit
2011-02-01
Impedance-based Coulter counters and its derivatives are widely used cell analysis tools in many laboratories and use normally DC or low frequency AC to perform these electrical analyses. The emergence of micro-fabrication technologies in the last decade, however, provides a new means of measuring electrical properties of cells. Microfluidic approaches combined with impedance spectroscopy measurements in the radio frequency (RF) range increase sensitivity and information content and thus push single cell analyses beyond simple cell counting and sizing applications towards multiparametric cell characterization. Promising results have been shown already in the fields of cell differentiation and blood analysis. Here we emphasize the potential of this technology by presenting new data obtained from viability studies on microorganisms. Impedance measurements of several yeast and bacteria strains performed at frequencies around 10 MHz enable an easy discrimination between dead and viable cells. Moreover, cytotoxic effects of antibiotics and other reagents, as well as cell starvation can also be monitored easily. Control analyses performed with conventional flow cytometers using various fluorescent dyes (propidium iodide, oxonol) indicate a good correlation and further highlight the capability of this device. The label-free approach makes on the one hand the use of usually expensive fluorochromes obsolete, on the other hand practically eliminates laborious sample preparation procedures. Until now, online cell monitoring was limited to the determination of viable biomass, which provides rather poor information of a cell culture. Impedance microflow cytometry, besides other aspects, proposes a simple solution to these limitations and might become an important tool for bioprocess monitoring applications in the biotech industry.
Physically based Impedance Modelling of Lithium-Ion Cells
Illig, Jörg
2014-01-01
In this book, a new procedure to analyze lithium-ion cells is introduced. The cells are disassembled to analyze their components in experimental cell housings. Then, Electrochemical Impedance Spectroscopy, time domain measurements and the Distribution function of Relaxation Times are applied to obtain a deep understanding of the relevant loss processes. This procedure yields a notable surplus of information about the electrode contributions to the overall internal resistance of the cell.
Stepped Impedance Resonators for High Field Magnetic Resonance Imaging
Akgun, Can E.; DelaBarre, Lance; Yoo, Hyoungsuk; Sohn, Sung-Min; Snyder, Carl J.; Adriany, Gregor; Ugurbil, Kamil; Gopinath, Anand; Vaughan, J. Thomas
2014-01-01
Multi-element volume radio-frequency (RF) coils are an integral aspect of the growing field of high field magnetic resonance imaging (MRI). In these systems, a popular volume coil of choice has become the transverse electromagnetic (TEM) multi-element transceiver coil consisting of microstrip resonators. In this paper, to further advance this design approach, a new microstrip resonator strategy in which the transmission line is segmented into alternating impedance sections referred to as step...
AC Complex Impedance Analysis of Doped Strontium Titanate Multifunctional Ceramics
Institute of Scientific and Technical Information of China (English)
无
2002-01-01
Doped SrTiO3 capacitor-varistor multifunctional ceramics were fabricated by a single sintering process. AC compleximpedance analysis was performed to investigate electrical features ofgrains and grain boundaries for both as-reducedceramic and reoxidized ceramics. The results showed that the as-reduced ceramic exhibited inductive response athigh frequencies above 2 MHz, which is attributed to the contribution of electron behavior in semiconducting grains.The high frequency inductive response disappeared in impedance plots of reoxidized ceramics.
Impedance matching through a single passive fractional element
Radwan, Ahmed Gomaa
2012-07-01
For the first time, a generalized admittance Smith chart theory is introduced to represent fractional order circuit elements. The principles of fractional order matching circuits are described. We show that for fractional order α < 1, a single parallel fractional element can match a wider range of load impedances as compared to its series counterpart. Several matching examples demonstrate the versatility of fractional order series and parallel element matching as compared to the conventional approach. © 2012 IEEE.
Low Power Implantable ASIC forBio-Impedance Measurements
Hussain Waqar, Muhammad
2012-01-01
Electrical bio-impedance can give a lot of insight into the basic physiological parameters of human body and concentration of glucose is one of those vital parameters.In order to control diabetes mellitus, it is critically essential to maintain blood glucose concentrations within the normal physiological range to avoid diabetes related complications.Consequently, accurate in-vivo and in-vitro measurement of glucose concentration in physiological uids has long been a central goal of bio sensor...
Variations of respiratory impedance with lung volume in bronchial hyperreactivity.
van den Elshout, F J; van de Woestijne, K P; Folgering, H T
1990-08-01
The total respiratory impedance was measured at various frequencies (4 to 52 Hz) with a pseudo-random-noise forced oscillation technique (FOT). The apparatus (Oscillaire) was connected with a spirometer forming a closed respiratory circuit in which gas concentrations were kept constant. Measurements were made in 15 healthy subjects (group 1) and in 30 asthmatic patients with bronchial hyperreactivity, subdivided into group 2 treated only with inhaled beta 2-mimetics (n = 15) and in group 3 using both beta 2-mimetics and steroids in inhalation (n = 15). No significant differences were found between the impedance data obtained with the Oscillaire alone and those obtained with the Oscillaire connected with the spirometer circuit. The impedance was measured at FRC level, and at FRC +1 L and -1 L. The relative changes of the resistance at 8 Hz were -23.2 percent (13.8) at FRC +1 L and +40.9 percent (29.3) at FRC -1 L relative to the values at resting FRC. This inverse relation between airway resistance and lung volume was similar in all groups. The average reactance decreased at FRC -1 L in all groups. However, at FRC +1 L the average reactance increased 50.6 percent in group 2 and 94.2 percent in group 3, but decreased in group 1. Concomitant changes were observed in the resonant frequency and in the frequency dependence of resistance. Because of these qualitatively different responses of the impedance data to changes in lung volume (both for the whole group and for each individual) between healthy subjects and asthmatic patients, this test might be useful for the diagnosis of bronchial hyperreactivity. PMID:2376168
Measuring Water Flow in Soil Using Electrical Impedance Tomography
Kaipio, Paula
2014-01-01
Abstract The topic of this thesis was the monitoring of water flow in soil using Electrical Impedance Tomography (EIT). The aim was to investigate the suitability of EIT for imaging water flow in soil. Laboratory and field measurements were carried out for investigating the feasibility of EIT imaging of water flow in soil. Water flow in soil was three-dimensionally (3D) imaged from the measurements. Imaging of flowing water and other fluids in soil can be utilized in investigating gro...
The Impedance of the Ceramic Chamber in J-PARC
Shobuda, Yoshihiro; Ohmi, Kazuhito; Toyama, Takeshi
2005-01-01
The ceramic chamber is adopted at the RCS (rapid cycling synchrotron) in J-PARC. The copper stripes are on the outer surface of the chamber in order to shield the electro-magnetic field produced by the beam. The inner surface of the chamber is coated by TiN to suppress the secondary electron emission. In this paper, we calculate the strength of electro-magnetic field produced by the beam and evaluate the impedance of this ceramic chamber.
Voltammetric and impedance studies of inosine-5'-monophosphate and hypoxanthine
Oliveira-Brett, Ana Maria; Silva, Luís A.; Farace, Giosi; Vadgama, Pankaj; Christopher M. A. Brett
2003-01-01
The oxidation mechanism and adsorption of inosine 5'-monophosphate and hypoxanthine were investigated in solutions of different pH using voltammetric and impedance methods at glassy carbon electrodes. For both compounds, the pH dependence from differential pulse voltammetry showed that the same number of electrons and protons are involved in the rate-determining step of the electrochemical reaction. In the case of hypoxanthine, it was also possible to study the effect of different concentrati...
Effect of External Vibration on PZT Impedance Signature
Yaowen Yang; Aiwei Miao
2008-01-01
Piezoelectric ceramic Lead Zirconate Titanate (PZT) transducers, working on the principle of electromechanical impedance (EMI), are increasingly applied for structural health monitoring (SHM) in aerospace, civil and mechanical engineering. The PZT transducers are usually surface bonded to or embedded in a structure and subjected to actuation so as to interrogate the structure at the desired frequency range. The interrogation results in the electromechanical admittance (inverse of EMI) signatu...
Some Nonlinear Reconstruction Algorithms for Electrical Impedance Tomography
Energy Technology Data Exchange (ETDEWEB)
Berryman, J G
2001-03-09
An impedance camera [Henderson and Webster, 1978; Dines and Lytle, 1981]--or what is now more commonly called electrical impedance tomography--attempts to image the electrical impedance (or just the conductivity) distribution inside a body using electrical measurements on its boundary. The method has been used successfully in both biomedical [Brown, 1983; Barber and Brown, 1986; J. C. Newell, D. G. Gisser, and D. Isaacson, 1988; Webster, 1990] and geophysical applications [Wexler, Fry, and Neurnan, 1985; Daily, Lin, and Buscheck, 1987], but the analysis of optimal reconstruction algorithms is still progressing [Murai and Kagawa, 1985; Wexler, Fry, and Neurnan, 1985; Kohn and Vogelius, 1987; Yorkey and Webster, 1987; Yorkey, Webster, and Tompkins, 1987; Berryman and Kohn, 1990; Kohn and McKenney, 1990; Santosa and Vogelius, 1990; Yorkey, 1990]. The most common application is monitoring the influx or efflux of a highly conducting fluid (such as brine in a porous rock or blood in the human body) through the volume being imaged. For biomedical applications, this met hod does not have the resolution of radiological methods, but it is comparatively safe and inexpensive and therefore provides a valuable alternative when continuous monitoring of a patient or process is desired. The following discussion is intended first t o summarize the physics of electrical impedance tomography, then to provide a few details of the data analysis and forward modeling requirements, and finally to outline some of the reconstruction algorithms that have proven to be most useful in practice. Pointers to the literature are provided throughout this brief narrative and the reader is encouraged to explore the references for more complete discussions of the various issues raised here.
Complex Impedance as a Diagnostic Tool for Characterizing Thermal Detectors
Vaillancourt, J E
2004-01-01
The complex a.c. impedance of a bolometer or microcalorimeter detector is easily measured and can be used to determine thermal time constants, thermal resistances, heat capacities, and sensitivities. Accurately extracting this information requires an understanding of the electrical and thermal properties of both the detector and the measurement system. We show that this is a practical method for measuring parameters in detectors with moderately complex thermal systems.
Numerical evaluation of self and mutual earth return impedances
Denisa Stet; Dan D. Micu; Levente Czumbil; Emil Simion
2014-01-01
The paper presents an evaluation of analytical self and mutual impedances formulas of lines with earth return, taking into account the ground correction terms. The determined formulas contain semi-infinite integral terms which are calculated by a novel stable and efficient numerical integration scheme in order to overcome the involved oscillation problems. It might seek approximations of the semi-infinite integrals by replacing an exponential or algebraic function, the objective being to perm...
Modes and exceptional points in waveguides with impedance boundary conditions
Midya, Bikashkali
2016-01-01
A planar waveguide with impedance boundary, composed of non-perfect metallic plates, and with passive or active dielectric filling is considered. We show the possibility of selective mode guiding and amplification when homogeneous pump is added to the dielectric, and analyze differences in TE and TM mode propagation. Such a non-conservative system is also shown to feature exceptional points, for specific and experimentally tunable parameters, which are described for a particular case of transparent dielectric.
SOFC modelling and parameter identification by means of impedance spectroscopy
Leonide, André
2010-01-01
A high resolution electrochemical impedance spectroscopy study on anode supported single cells (ASC) is presented. The cells were characterised over a broad range of operating conditions, including different temperatures, current densities and various cathode and anode gas compositions. The analysis of the distribution of relaxation times combined with the numeric accuracy of a CNLS fit enabled the identification of five different processes contributing to the total polarisation loss of an...
Giant magneto-impedance in nanocrystalline glass-covered microwires
International Nuclear Information System (INIS)
Giant magneto-impedance (GMI) effect has been studied in Fe73.5Nb3Cu1Si13.5B9 glass-covered microwires after heat treatment in vacuum at different temperatures up to 600 deg C for 1 h. We present studies on the frequency dependence of the GMI in the range of 1 kHz-10 MHz and the dependence of magnetostriction constant on annealing time
On second order shape optimization methods for electrical impedance tomography
Afraites, Lekbir; Dambrine, Marc; Kateb, Djalil
2007-01-01
This paper is devoted to the analysis of a second order method for recovering the \\emph{a priori} unknown shape of an inclusion $\\omega$ inside a body $\\Omega$ from boundary measurement. This inverse problem - known as electrical impedance tomography - has many important practical applications and hence has focussed much attention during the last years. However, to our best knowledge, no work has yet considered a second order approach for this problem. This paper aims to fill that void: we in...
Deurenberg, P.R.M.; Deurenberg-Yap, M.; Schouten, F.J.M.
2002-01-01
Methods: Anthropometric parameters were measured in addition to impedance (100 kHz) of the total body, arms and legs. Impedance indexes were calculated as height2/impedance. Arm length (span) and leg length (sitting height), wrist and knee width were measured from which body build indices were calcu
Low-frequency oscillations of the impedance of electrolyte moving in an electro-osmotic regime
Kompan, M. E.; Malyshkin, V. G.; Goffman, V. G.
2014-07-01
The impedance of a liquid electrolyte has been studied under the conditions of electro-osmotic flow. It is found that the impedance exhibits oscillations in the region of subhertz frequencies, which are related to resonant mechanical oscillations arising in the flow. Assumptions concerning the type of these resonant oscillations are formulated based on the observed spectrum of impedance oscillations.
Electrical Impedance of Acupuncture Meridians: The Relevance of Subcutaneous Collagenous Bands
Ahn, Andrew C.; Min Park; Shaw, Jessica R; McManus, Claire A.; Kaptchuk, Ted J.; Langevin, Helene M.
2010-01-01
Background: The scientific basis for acupuncture meridians is unknown. Past studies have suggested that acupuncture meridians are physiologically characterized by low electrical impedance and anatomically associated with connective tissue planes. We are interested in seeing whether acupuncture meridians are associated with lower electrical impedance and whether ultrasound-derived measures – specifically echogenic collagenous bands - can account for these impedance differences. Methods/Results...
Time-domain fitting of battery electrochemical impedance models
Alavi, S. M. M.; Birkl, C. R.; Howey, D. A.
2015-08-01
Electrochemical impedance spectroscopy (EIS) is an effective technique for diagnosing the behaviour of electrochemical devices such as batteries and fuel cells, usually by fitting data to an equivalent circuit model (ECM). The common approach in the laboratory is to measure the impedance spectrum of a cell in the frequency domain using a single sine sweep signal, then fit the ECM parameters in the frequency domain. This paper focuses instead on estimation of the ECM parameters directly from time-domain data. This may be advantageous for parameter estimation in practical applications such as automotive systems including battery-powered vehicles, where the data may be heavily corrupted by noise. The proposed methodology is based on the simplified refined instrumental variable for continuous-time fractional systems method ('srivcf'), provided by the Crone toolbox [1,2], combined with gradient-based optimisation to estimate the order of the fractional term in the ECM. The approach was tested first on synthetic data and then on real data measured from a 26650 lithium-ion iron phosphate cell with low-cost equipment. The resulting Nyquist plots from the time-domain fitted models match the impedance spectrum closely (much more accurately than when a Randles model is assumed), and the fitted parameters as separately determined through a laboratory potentiostat with frequency domain fitting match to within 13%.
Proposal for measurement of earth impedance using variable frequency injection
International Nuclear Information System (INIS)
Measurement of the impedance of earthing systems of high voltage installations at power fundamental frequency has been studied extensively, and standards are available which provide guidance on methods of measurement and types of instrumentation. In the UK, the staged-fault test is seldom performed. Therefore, it is necessary to measure earth impedance without any circuit interruption by injecting current into the earthing system using an auxiliary test circuit. This type of 'fall-of-potential' test generally uses low voltage energization and produces test currents of the order of only a few amperes. At this level of current, potential measurements are made difficult due to the presence of relatively high standing voltages in the earthing system. In this paper, a new earth impedance measurement system which is able to overcome the effects of system noise and provide accurate measurements is described. Initial laboratory tests which resulted in optimized instrument settings and subsequent field tests demonstrated successful application at a range of operational electricity installations
Optimal impedance on transmission of Lorentz force EMATs
Isla, Julio; Seher, Matthias; Challis, Richard; Cegla, Frederic
2016-02-01
Electromagnetic-acoustic transducers (EMATs) are attractive for non-destructive inspections because direct contact with the specimen under test is not required. This advantage comes at a high cost in sensitivity and therefore it is important to optimise every aspect of an EMAT. The signal strength produced by EMATs is in part determined by the coil impedance regardless of the transduction mechanism (e.g. Lorentz force, magnetostriction, etc.). There is very little literature on how to select the coil impedance that maximises the wave intensity; this paper addresses that gap. A transformer circuit is used to model the interaction between the EMAT coil and the eddy currents that are generated beneath the coil in the conducting specimen. Expressions for the coil impedances that satisfy the maximum efficiency and maximum power transfer conditions on transmission are presented. To support this analysis, a tunable coil that consists of stacked identical thin layers independently accessed is used so that the coil inductance can be modified while leaving the radiation pattern of the EMAT unaffected.
Enhanced modeling of magnetic impedance sensing system for damage detection
International Nuclear Information System (INIS)
There has been recent interest in utilizing the magneto-mechanical coupling characteristics of a magnetic transducer to perform impedance-based damage detection of electrically conductive structures. This approach is non-contact in nature and has potential advantages in many applications. One important parameter in this approach is the lift-off distance, i.e. the distance from the transducer to the structure monitored, the change of which changes the magneto-mechanical coupling. In the past, the magneto-mechanical coupling is extracted completely or partially from experiment in an ad hoc manner. A predictive capability of magneto-mechanical coupling under given lift-off distance would play a significant role in damage detection practice and in sensor design/optimization. In this research, we formulate detailed first-principles-based modeling of a magnetic impedance transducer. In particular, the complete electrical effect of the structure is explicitly taken into consideration. Comprehensive analyses and experiments are carried out, which validate the underlying hypothesis as well as the accuracy of the new model proposed for impedance response prediction. (paper)
Investigations of lithium–sulfur batteries using electrochemical impedance spectroscopy
International Nuclear Information System (INIS)
Highlights: ► Li–S battery was investigated during cycling by means of EIS. ► An equivalent circuit is proposed for interpretation of impedance contributions. ► Diminution of charge transfer resistance at middle frequency due to cell degradation. ► High charge transfer resistance at high frequency due high polysulfide concentration. ► Formation of isolating layer after cycling (AFM). -- Abstract: The electrochemical behavior of lithium–sulfur (Li–S) batteries was investigated by means of electrochemical impedance spectroscopy (EIS). Measurements were performed in equidistant charge intervals at different depths of discharge and charge during the first cycle. Additionally, the degradation of the cells was analyzed for up to 50 cycles. An equivalent electrical circuit is proposed to simulate the electrochemical processes and to quantify the impedance contributions of Li–S batteries. EIS as a function of the cycle number shows an increased capacity for fading correlated with a decrease in the charge transfer resistance of the cathode. Atomic force microscopy (AFM) was also used to provide information about changes in the electrical conductivity of the cathode surface as they are related to the building of isolating films
Embedded silver PDMS electrodes for single cell electrical impedance spectroscopy
Wei, Yuan; Xu, Zhensong; Cachia, Mark A.; Nguyen, John; Zheng, Yi; Wang, Chen; Sun, Yu
2016-09-01
This paper presents a microfluidic device with wide channels and embedded AgPDMS electrodes for measuring the electrical properties of single cells. The work demonstrates the feasibility of using a large channel design and embedded electrodes for impedance spectroscopy to circumvent issues such as channel clogging and limited device re-usability. AgPDMS electrodes were formed on channel sidewalls for impedance detection and cell electrical properties measurement. Equivalent circuit models were used to interpret multi-frequency impedance data to quantify each cell’s cytoplasm conductivity and specific membrane capacitance. T24 cells were tested to validate the microfluidic system and modeling results. Comparisons were then made by measuring two leukemia cell lines (AML-2 and HL-60) which were found to have different cytoplasm conductivity values (0.29 ± 0.15 S m‑1 versus 0.47 ± 0.20 S m‑1) and specific membrane capacitance values (41 ± 25 mF m‑2 versus 55 ± 26 mF m‑2) when the cells were flown through the wide channel and measured by the AgPDMS electrodes.
Effect of Accelerator Impedance on Electron Cloud Instability
Allen, Brian; Muggli, Patric; Fischer, Wolfram; Blaskiewicz, Michael; Katsouleas, Thomas
2009-11-01
Interaction between a beam and electron clouds (e-cloud) present in circular accelerators is known to limit accelerator performances through instabilities, beam loss, beam-blowup, and the resulting reduced luminosity. The RHIC beam is most susceptible to instabilities as it crosses energy transition (γt=22.9) and it is posited that ring impedance could play a role in the development of instabilities during this transition. We use the quasi-static particle in cell code QuickPIC to describe the interaction between the RHIC Au beam and the electron cloud. In QuickPIC the electron cloud density is uniform around the ring and the beam has a constant beta function given by the accelerator circumference and the beam tune. We incorporate in the current QuickPIC version the ring impedance for a circular accelerator and we take a first look at the effect this impedance has on the beam and e-cloud interaction for typical RHIC parameters.
Beam impedance measurements on the ALS sector tank
International Nuclear Information System (INIS)
The 10 m long ALS curved sector tank is formed from two shells out of which the beam chamber is machined. Vacuum pumping and photon stops are located in an antechamber connected to the beam tube through a 1cm slot. In order to determine whether the beam is significantly coupled to the antechamber, measurements of longitudinal beam impedance were performed up to 26 GHz, well above the cutoff frequency of the beam pipe. Two different schemes were used: In the first, the wire method was adapted for use above cutoff; in the second, the impedance was detected from the response to TM-waves propagated in the aperture without a wire. Temperature at various locations in the setup was recorded for later phase corrections. Antennas were placed in the antechamber to detect radiated power or possible resonances. A reference measurement was made with the slot sealed by a flexible gasket of knitted wire. The seal was then removed and the response with antechamber recorded. The response was checked by inserting obstacles of known impedance. Both measurement methods provided low numbers with Z/n<0.001 Ohm over the whole frequency range. No resonances attributable to the antechamber were observed. 3 refs., 6 figs
Revealing alteration of membrane structures during ischema using impedance spectroscopy
Directory of Open Access Journals (Sweden)
Mihaela Gheorghiu
2002-11-01
Full Text Available Alterations of membrane structure and function are essential characteristics of cells undergoing ischemia. Noninvasive monitoring of tissue alterations during ischemia and the estimation of the reversibility domain (corresponding to organ capability to fully recover its functions after shifting back to normal blood perfusion are important for biomedical applications allowing better time management during surgical interventions, especially in organ transplantation. Due to it’s capability to reveal inhomogeneities, as well as it’s noninvasive character, impedance spectroscopy was used for continuous monitoring of the progression of excised tissue samples during ischemia. We have developed a fast, noninvasive, automated method for quantitative analysis of impedance spectra of tissue samples, capable of revealing, through characteristic parameters (dispersion amplitudes, time constants and distribution parameters membrane based microscopic processes like the closure ofgap-junctions (a characteristic of the early alterations of ischemic tissues in the reversibility phase. Microscopic and equivalent circuit modeling was used to probe the effect of closure of cell connections and of changes in electrical properties of cell constituents on impedance spectra. We have developed a normalizing procedure emphasizing the pattern of ischemic alterations and enabling the comparison of different data sets.
Application of impedance measurement techniques to accelerating cavity mode characterization
International Nuclear Information System (INIS)
Impedance measurements, using a central wire to simulate the electron beam, were performed on a 52 MHz accelerating cavity at the National Synchrotron Light Source (NSLS). This cavity was recently installed in the X-ray storage ring at the NSLS as a part of an upgrade of the ring. To damp higher-order modes (HOM) in this cavity, damping antennas have been installed. We implemented the impedance measurement technique to characterize the cavity modes up to 1 GHz and confirm the effectiveness of the damping antennas. Scattering parameters were measured using a network analyzer (HP 8510B) with a personal computer as a controller. Analysis based on S and T parameters for the system was used to solve for the cavity impedance, Z(ω), as a function of the measured transmission response, S21(ω). Search techniques were used to find the shunt resistance Rsh, and Q from the calculated Z(ω) for different modes. Our results for R / Q showed good agreement with URMEL simulations. The values of Q were compared with other independent Q measurement techniques. Our analytical technique offers an alternative approach for cases where full thru-reflection-line (TRL) calibration is not feasible and a more time-effective technique for obtaining R / Q, compared with the bead-pull method. (orig.)
NEW BIOTESTING METHOD WITH THE APPLICATION OF MODERN IMPEDANCE TECHNOLOGIES
Directory of Open Access Journals (Sweden)
Sibirtsev V.S.
2015-03-01
Full Text Available The paper deals with new concepts of biotesting method updating. Modern conductometric technologies and the analysis of microbial «growth curves» are used. The registration occurs in a real time mode for the set of parallel samples. Results are shown for comparison of the proposed impedance biotesting technique with standard cultural determination method for total amount of microorganismes in the tested samples. Results are presented for practical application of the proposed impedance biotesting technique to the analysis as inhibitory action of clorhexidine disinfectant on the vital activity of Escherichia coli, as milk ripening process at the presence of various microorganisms species and protein preparations. The impedance biotesting method, proposed in the present work, provides high level of its own data convergence with the data, being received as a result of application of standard cultural biotesting techniques. Thus, the proposed method has such advantages, as: an opportunity of the detailed information reception about dynamics change of magnitude of population and intensity of test microorganisms metabolism, significant reduction of the culture media amount used, as well as researcher's temporary and labor efforts while the analyses realization, and the growth of analysis objectivity.
Impedance spectroscopic characterization of gadolinium substituted cobalt ferrite ceramics
International Nuclear Information System (INIS)
Gadolinium (Gd) substituted cobalt ferrites (CoFe2−xGdxO4, referred to CFGO) with variable Gd content (x = 0.0–0.4) have been synthesized by solid state ceramic method. The crystal structure and impedance properties of CFGO compounds have been evaluated. X-ray diffraction measurements indicate that CFGO crystallize in the inverse spinel phase. The CFGO compounds exhibit lattice expansion due to substitution of larger Gd ions into the crystal lattice. Impedance spectroscopy analysis was performed under a wide range of frequency (f = 20 Hz–1 MHz) and temperature (T = 303–573 K). Electrical properties of Gd incorporated Co ferrite ceramics are enhanced compared to pure CoFe2O4 due to the lattice distortion. Impedance spectroscopic analysis illustrates the variation of bulk grain and grain-boundary contributions towards the electrical resistance and capacitance of CFGO materials with temperature. A two-layer heterogeneous model consisting of moderately conducting grain interior (ferrite-phase) regions separated by insulating grain boundaries (resistive-phase) accurately account for the observed temperature and frequency dependent electrical characteristic of CFGO ceramics.
Study of surfactant-skin interactions by skin impedance measurements.
Lu, Guojin; Moore, David J
2012-02-01
The stratum corneum (SC) plays a very critical physiological role as skin barrier in regulating water loss through the skin and protects the body from a wide range of physical and chemical exogenous insults. Surfactant-containing formulations can induce skin damage and irritation owing to surfactant absorption and penetration. It is generally accepted that reduction in skin barrier properties occurs only after surfactants have penetrated/permeated into the skin barrier. To mitigate the harshness of surfactant-based cleansing products, penetration/permeation of surfactants should be reduced. Skin impedance measurements have been taken in vitro on porcine skin using vertical Franz diffusion cells to investigate the impact of surfactants, temperature and pH on skin barrier integrity. These skin impedance results demonstrate excellent correlation with other published methods for assessing skin damage and irritation from different surfactant chemistry, concentration, pH, time of exposure and temperature. This study demonstrates that skin impedance can be utilized as a routine approach to screen surfactant-containing formulations for their propensity to compromise the skin barrier and hence likely lead to skin irritation. PMID:21923733
Portable audio electronics for impedance-based measurements in microfluidics
International Nuclear Information System (INIS)
We demonstrate the use of audio electronics-based signals to perform on-chip electrochemical measurements. Cell phones and portable music players are examples of consumer electronics that are easily operated and are ubiquitous worldwide. Audio output (play) and input (record) signals are voltage based and contain frequency and amplitude information. A cell phone, laptop soundcard and two compact audio players are compared with respect to frequency response; the laptop soundcard provides the most uniform frequency response, while the cell phone performance is found to be insufficient. The audio signals in the common portable music players and laptop soundcard operate in the range of 20 Hz to 20 kHz and are found to be applicable, as voltage input and output signals, to impedance-based electrochemical measurements in microfluidic systems. Validated impedance-based measurements of concentration (0.1–50 mM), flow rate (2–120 µL min−1) and particle detection (32 µm diameter) are demonstrated. The prevailing, lossless, wave audio file format is found to be suitable for data transmission to and from external sources, such as a centralized lab, and the cost of all hardware (in addition to audio devices) is ∼10 USD. The utility demonstrated here, in combination with the ubiquitous nature of portable audio electronics, presents new opportunities for impedance-based measurements in portable microfluidic systems. (technical note)
Design and Analysis of Impedance Pumps Utilizing Electromagnetic Actuation
Directory of Open Access Journals (Sweden)
Yu-Hisang Wang
2010-04-01
Full Text Available This study designs and analyzes an impedance pump utilizing an electromagnetic actuator. The pump is designed to have three major components, namely a lower glass substrate patterned with a copper micro-coil, a microchannel, and an upper glass cover plate attached a magnetic PDMS diaphragm. When a current is passed through the micro-coil, an electromagnetic force is established between the coil and the magnetic diaphragm. The resulting deflection of the PDMS diaphragm creates an acoustic impedance mismatch within the microchannel, which results in a net flow. In performing the analysis, simulated models of the magnetic field, the diaphragm displacement and the flow rate are developed using Ansoft/Maxwell3D, ANSYS FEA and FLUENT 6.3 CFD software, respectively. Overall, the simulated results reveal that a net flow rate of 52.8 μL/min can be obtained using a diaphragm displacement of 31.5 μm induced by a micro-coil input current of 0.5 A. The impedance pump proposed in this study provides a valuable contribution to the ongoing development of Lab-on-Chips (LoCs systems.
A comprehensive impedance journey to continuous microbial fuel cells.
Sevda, Surajbhan; Chayambuka, Kudakwashe; Sreekrishnan, T R; Pant, Deepak; Dominguez-Benetton, Xochitl
2015-12-01
The aim of the present work was to characterize the impedance response of an air-cathode MFC operating in a continuous mode and to determine intrinsic properties that define its performance which are crucial to be controlled for scalability purposes. The limiting step on electricity generation is the anodic electrochemically-active biofilm, independently of the external resistance, Rext, utilized. However, for Rext below 3 kΩ the internal impedance of the bioanode remained invariable, in good correspondence to the power density profile. The hydraulic retention time (HRT) had an effect on the impedance of both the bioanode and the air-cathode and especially on the overall MFC. The lowest HRT at which the MFC was operable was 3h. Yet, the variation on the HRT did not have a significant impact on power generation. A two constant phase element-model was associated with the EIS response of both bioanode and air-cathode, respectively. Consistency was found between the CPE behaviour and the normal power-law distribution of local resistivity with a uniform dielectric constant, which represented consistent values with the electrical double layer, the Nernst diffusion layer and presumably the biofilm thickness. These results have future implications on MFC monitoring and control, as well as in providing critical parameters for scale-up. PMID:25921205
Dielectric and impedance spectroscopic studies of neodymium gallate
Sakhya, Anup Pradhan; Dutta, Alo; Sinha, T. P.
2016-05-01
The AC electrical properties of a polycrystalline neodymium gallate, NdGaO3 (NGO), synthesized by the sol-gel method have been investigated by employing impedance spectroscopy in the frequency range from 42 Hz to 5 MHz and in the temperature range from 323 K to 593 K. The X-ray diffraction analysis shows that the compound crystallizes in the orthorhombic phase with Pbnm space group at room temperature. Two relaxation processes with different relaxation times are observed from the impedance as well as modulus spectroscopic measurements, which have been attributed to the grain and the grain boundary effects at different temperatures in NGO. The complex impedance data are analyzed by an electrical equivalent circuit consisting of a resistance and a constant phase element in parallel. It has been observed that the value of the capacitance and the resistance associated with the grain boundary is higher than those associated with the grain. The temperature dependent electrical conductivity shows the negative temperature coefficient of resistance. The frequency dependent conductivity spectra are found to follow the power law.
Directory of Open Access Journals (Sweden)
Elin Ericsson
Full Text Available Ventilator-induced or ventilator-associated lung injury (VILI/VALI is common and there is an increasing demand for a tool that can optimize ventilator settings. Electrical impedance tomography (EIT can detect changes in impedance caused by pulmonary ventilation and perfusion, but the effect of changes in the position of the body and in the placing of the electrode belt on the impedance signal have not to our knowledge been thoroughly evaluated. We therefore studied ventilation-related and perfusion-related changes in impedance during spontaneous breathing in 10 healthy subjects in five different body positions and with the electrode belt placed at three different thoracic positions using a 32-electrode EIT system. We found differences between regions of interest that could be attributed to changes in the position of the body, and differences in impedance amplitudes when the position of the electrode belt was changed. Ventilation-related changes in impedance could therefore be related to changes in the position of both the body and the electrode belt. Perfusion-related changes in impedance were probably related to the interference of major vessels. While these findings give us some insight into the sources of variation in impedance signals as a result of changes in the positions of both the body and the electrode belt, further studies on the origin of the perfusion-related impedance signal are needed to improve EIT further as a tool for the monitoring of pulmonary ventilation and perfusion.
Ericsson, Elin; Tesselaar, Erik; Sjöberg, Folke
2016-01-01
Ventilator-induced or ventilator-associated lung injury (VILI/VALI) is common and there is an increasing demand for a tool that can optimize ventilator settings. Electrical impedance tomography (EIT) can detect changes in impedance caused by pulmonary ventilation and perfusion, but the effect of changes in the position of the body and in the placing of the electrode belt on the impedance signal have not to our knowledge been thoroughly evaluated. We therefore studied ventilation-related and perfusion-related changes in impedance during spontaneous breathing in 10 healthy subjects in five different body positions and with the electrode belt placed at three different thoracic positions using a 32-electrode EIT system. We found differences between regions of interest that could be attributed to changes in the position of the body, and differences in impedance amplitudes when the position of the electrode belt was changed. Ventilation-related changes in impedance could therefore be related to changes in the position of both the body and the electrode belt. Perfusion-related changes in impedance were probably related to the interference of major vessels. While these findings give us some insight into the sources of variation in impedance signals as a result of changes in the positions of both the body and the electrode belt, further studies on the origin of the perfusion-related impedance signal are needed to improve EIT further as a tool for the monitoring of pulmonary ventilation and perfusion. PMID:27253433
Optimal multisine excitation design for broadband electrical impedance spectroscopy
International Nuclear Information System (INIS)
Electrical impedance spectroscopy (EIS) can be used to characterize biological materials in applications ranging from cell culture to body composition, including tissue and organ state. The emergence of cell therapy and tissue engineering opens up a new and promising field of application. While in most cases classical measurement techniques based on a frequency sweep can be used, EIS based on broadband excitations enables dynamic biological systems to be characterized when the measuring time and injected energy are a constraint. Myocardial regeneration, cell characterization in micro-fluidic systems and dynamic electrical impedance tomography are all examples of such applications. The weakness of such types of fast EIS measuring techniques resides in their intrinsic loss of accuracy. However, since most of the practical applications have no restriction over the excitation used, the input power spectrum can be appropriately designed to maximize the accuracy obtained from the measurements. This paper deals with the problem of designing the optimal multisine excitation for electrical bioimpedance measurements. The optimal multisine is obtained by the minimization of the Cramer–Rao lower bound, or what is the same, by maximizing the accuracy obtained from the measurements. Furthermore, because no analytical solution exists for global optimization involving time and frequency domains jointly, this paper presents the multisine optimization approach partially in both domains and then combines the results. As regards the frequency domain approach, a novel contribution is made for the multisine amplitude power spectrum. In the time domain, multisine is optimized by reducing its crest factor. Moreover, the impact on the information and accuracy of the impedance spectrum obtained from using different multisine amplitude power spectra is discussed, as well as the number of frequencies and frequency distributions. The theory is supported by a set of validation measurements
Optimal multisine excitation design for broadband electrical impedance spectroscopy
Sanchez, B.; Vandersteen, G.; Bragos, R.; Schoukens, J.
2011-11-01
Electrical impedance spectroscopy (EIS) can be used to characterize biological materials in applications ranging from cell culture to body composition, including tissue and organ state. The emergence of cell therapy and tissue engineering opens up a new and promising field of application. While in most cases classical measurement techniques based on a frequency sweep can be used, EIS based on broadband excitations enables dynamic biological systems to be characterized when the measuring time and injected energy are a constraint. Myocardial regeneration, cell characterization in micro-fluidic systems and dynamic electrical impedance tomography are all examples of such applications. The weakness of such types of fast EIS measuring techniques resides in their intrinsic loss of accuracy. However, since most of the practical applications have no restriction over the excitation used, the input power spectrum can be appropriately designed to maximize the accuracy obtained from the measurements. This paper deals with the problem of designing the optimal multisine excitation for electrical bioimpedance measurements. The optimal multisine is obtained by the minimization of the Cramer-Rao lower bound, or what is the same, by maximizing the accuracy obtained from the measurements. Furthermore, because no analytical solution exists for global optimization involving time and frequency domains jointly, this paper presents the multisine optimization approach partially in both domains and then combines the results. As regards the frequency domain approach, a novel contribution is made for the multisine amplitude power spectrum. In the time domain, multisine is optimized by reducing its crest factor. Moreover, the impact on the information and accuracy of the impedance spectrum obtained from using different multisine amplitude power spectra is discussed, as well as the number of frequencies and frequency distributions. The theory is supported by a set of validation measurements when
High Dynamic Range Complex Impedance Measurement System for Petrophysical Usage
Chen, R.; He, X.; Yao, H.; Tan, S.; Shi, H.; Shen, R.; Yan, C.; Zeng, P.; He, L.; Qiao, N.; Xi, F.; Zhang, H.; Xie, J.
2015-12-01
Spectral induced polarization method (SIP) or complex resistivity method is increasing its application in metalliferous ore exploration, hydrocarbon exploration, underground water exploration, monitoring of environment pollution, and the evaluation of environment remediation. And the measurement of complex resistivity or complex impedance of rock/ore sample and polluted water plays a fundamental role in improving the application effect of SIP and the application scope of SIP. However, current instruments can't guaranty the accuracy of measurement when the resistance of sample is less than 10Ω or great than 100kΩ. A lot of samples, such as liquid, polluted sea water, igneous rock, limestone, and sandstone, can't be measured with reliable complex resistivity result. Therefore, this problem projects a shadow in the basic research and application research of SIP. We design a high precision measurement system from the study of measurement principle, sample holder, and measurement instrument. We design input buffers in a single board. We adopt operation amplifier AD549 in this system because of its ultra-high input impedance and ultra-low current noise. This buffer is good in acquiring potential signal across high impedance sample. By analyzing the sources of measurement error and errors generated by the measurement system, we propose a correction method to remove the error in order to achieve high quality complex impedance measurement for rock and ore samples. This measurement system can improve the measurement range of the complex impedance to 0.1 Ω ~ 10 GΩ with amplitude error less than 0.1% and phase error less than 0.1mrad when frequency ranges as 0.01 Hz ~ 1 kHz. We tested our system on resistors with resistance as 0.1Ω ~ 10 GΩ in frequency range as 1 Hz ~ 1000 Hz, and the measurement error is less than 0.1 mrad. We also compared the result with LCR bridge and SCIP, we can find that the bridge's measuring range only reaches 100 MΩ, SCIP's measuring range
Wang, Jing
2015-01-01
Electrical bioimpedance measurement is widely used today for monitoring body condition. The applications include and go beyond, body composition assessment, nutritional status evaluation, and cancer detection. The modalities for interpreting the impedance information have also developed quickly over the recent decades from single frequency bioimpedance analysis to spectrum and to images. Bioimpedance is measured by computing the relationship between voltage and current. In a current based bio...
Güren, Onan; Çayören, Mehmet; Tükenmez Ergene, Lale; Akduman, Ibrahim
2014-10-01
A new microwave imaging method that uses microwave contrast agents is presented for the detection and localization of breast tumours. The method is based on the reconstruction of breast surface impedance through a measured scattered field. The surface impedance modelling allows for representing the electrical properties of the breasts in terms of impedance boundary conditions, which enable us to map the inner structure of the breasts into surface impedance functions. Later a simple quantitative method is proposed to screen breasts against malignant tumours where the detection procedure is based on weighted cross correlations among impedance functions. Numerical results demonstrate that the method is capable of detecting small malignancies and provides reasonable localization.
Characterization of electro-acoustics impedance and its application to active noise control
Institute of Scientific and Technical Information of China (English)
HOU Hong; YANG Jianhua
2004-01-01
Characteristics of radiation impedance and its inducing variation of electrical impedance for a controllable source have been investigated. An impedance-based error criterion has been proposed and its application to Active Noise Control is demonstrated through a coil driven loudspeaker. A general formula of radiation impedance is derived for two control strategies, according to the criterion of total acoustic power output. The radiation impedances of some commonly used sound sources are calculated. We discuss in detail the relation between variation of the input electrical impedance and radiation impedance for the two control strategies. The measured data of the input electrical impedance from a loudspeaker agree fairly well with theoretical analysis. An AC- bridge circuit is designed in order to measure the weak variation of electrical impedance resulted from radiation impedance. The bridge relative output is unique for a certain control strategy, from which an impedance-based error criterion is then proposed and the implementation of its application to an active control system is analyzed.Numerical results of such criterion are presented. An analogue control system is set up and experiments are carried out in a semi-anechoic chamber to verify the new control approach.
Diagnosis of PEMFC stack failures via electrochemical impedance spectroscopy
Merida-Donis, Walter Roberto
Two failure modes related to water management in Proton Exchange Membrane fuel cells (dehydration and flooding) were investigated using electrochemical impedance spectroscopy as a diagnosis tool. It was hypothesised that each failure mode corresponds to changes in the overall stack impedance that are observable in different frequency ranges. This hypothesis was corroborated experimentally. The experimental implementation required new testing hardware and techniques. A four-cell stack capable of delivering individually conditioned reactants to each cell was designed, built, tested, and characterised under a variety of operating conditions. This stack is the first reported prototype of its type. The stack was used to perform galvanostatic, impedance measurements in situ. The measurements were made at three different temperatures (62, 70 and 80°C), covering the current density range 0.1 to 1.0 A cm-2 , and the frequency range 0.1 to 4 x 105 Hz. The recorded data represent the first reported set of measurements covering these ranges. The failure modes were simulated on individual cells within the stack. The effects on individual cell and stack impedance were studied by measuring the changes in stack and cell impedances under flooding or dehydration conditions. Dehydration effects were measurable over a wide frequency range (0.5 to 105 Hz). In contrast, flooding effects were measurable in a narrower frequency range (0.5 to 102 Hz). Using these results, separate or concurrent impedance measurements in these frequency ranges (or narrow bands thereof) can be used to discern and identify the two failure modes quasi-instantaneously. Such detection was not possible with pre-existing, do techniques. The measured spectra were modelled by a simple equivalent circuit whose time constants corresponded to ideal (RC) and distributed (Warburg) components. The model was robust enough to fit all the measured spectra (for single cells and the stack), under normal and simulated
Directory of Open Access Journals (Sweden)
Yangkyu Park
2016-01-01
Full Text Available Purpose. To distinguish between normal (SV-HUC-1 and cancerous (TCCSUP human urothelial cell lines using microelectrical impedance spectroscopy (μEIS. Materials and Methods. Two types of μEIS devices were designed and used in combination to measure the impedance of SV-HUC-1 and TCCSUP cells flowing through the channels of the devices. The first device (μEIS-OF was designed to determine the optimal frequency at which the impedance of two cell lines is most distinguishable. The μEIS-OF trapped the flowing cells and measured their impedance at a frequency ranging from 5 kHz to 1 MHz. The second device (μEIS-RT was designed for real-time impedance measurement of the cells at the optimal frequency. The impedance was measured instantaneously as the cells passed the sensing electrodes of μEIS-RT. Results. The optimal frequency, which maximized the average difference of the amplitude and phase angle between the two cell lines (p<0.001, was determined to be 119 kHz. The real-time impedance of the cell lines was measured at 119 kHz; the two cell lines differed significantly in terms of amplitude and phase angle (p<0.001. Conclusion. The μEIS-RT can discriminate SV-HUC-1 and TCCSUP cells by measuring the impedance at the optimal frequency determined by the μEIS-OF.
Park, Yangkyu; Kim, Hyeon Woo; Yun, Joho; Seo, Seungwan; Park, Chang-Ju; Lee, Jeong Zoo; Lee, Jong-Hyun
2016-01-01
Purpose. To distinguish between normal (SV-HUC-1) and cancerous (TCCSUP) human urothelial cell lines using microelectrical impedance spectroscopy (μEIS). Materials and Methods. Two types of μEIS devices were designed and used in combination to measure the impedance of SV-HUC-1 and TCCSUP cells flowing through the channels of the devices. The first device (μEIS-OF) was designed to determine the optimal frequency at which the impedance of two cell lines is most distinguishable. The μEIS-OF trapped the flowing cells and measured their impedance at a frequency ranging from 5 kHz to 1 MHz. The second device (μEIS-RT) was designed for real-time impedance measurement of the cells at the optimal frequency. The impedance was measured instantaneously as the cells passed the sensing electrodes of μEIS-RT. Results. The optimal frequency, which maximized the average difference of the amplitude and phase angle between the two cell lines (p < 0.001), was determined to be 119 kHz. The real-time impedance of the cell lines was measured at 119 kHz; the two cell lines differed significantly in terms of amplitude and phase angle (p < 0.001). Conclusion. The μEIS-RT can discriminate SV-HUC-1 and TCCSUP cells by measuring the impedance at the optimal frequency determined by the μEIS-OF. PMID:26998490
Effect of borehole design on electrical impedance tomography measurements
Mozaffari, Amirpasha; Huisman, Johan Alexander; Treichel, Andrea; Zimmermann, Egon; Kelter, Matthias; Vereecken, Harry
2015-04-01
Electrical Impedance Tomography (EIT) is a sophisticated non-invasive tool to investigate the subsurface in engineering and environmental studies. To increase the depth of investigation, EIT measurements can be made in boreholes. However, the presence of the borehole may affect EIT measurements. Here, we aim to investigate the effect of different borehole components on EIT measurements using 2,5-D and 3D finite element modeling and unstructured meshes. To investigate the effect of different borehole components on EIT measurements, a variety of scenarios were designed. In particular, the effect of the water-filled borehole, the PVC casing, and the gravel filter were investigated relative to complex resistivity simulations for a homogenous medium with chain and electrode modules. It was found that the results of the complex resistivity simulations were best understood using the sensitivity distribution of the electrode configuration under consideration. In all simulations, the sensitivity in the vicinity of the borehole was predominantly negative. Therefore, the introduction of the water-filled borehole caused an increase in the real part of the impedance, and a decrease (more negative) in the imaginary part of the simulated impedance. The PVC casing mostly enhanced the effect of the water-filled borehole described above, although this effect was less clear for some electrode configuration. The effect of the gravel filter mostly reduced the effect of the water-filled borehole with PVC casing. For EIT measurements in a single borehole, the highest simulated phase error was 12% for a Wenner configuration with electrode spacing of 0.33 m. This error decreased with increasing electrode spacing. In the case of cross-well configurations, the error in the phase shit was as high as 6%. Here, it was found that the highest errors occur when both current electrodes are located in the same borehole. These results indicated that cross-well measurements are less affected by the
A phase field model of electrochemical impedance spectroscopy
Gathright, William
One of the fundamental tenets of Material Science is the link between microstructure and material properties. As such, there is a need for a microstructure-sensitive model of electrochemistry. Phase field models have been designed to simulate systems with complex and evolving microstructures such as eutectic solidification and dendrite growth. The goal of this work is to extend phase field models into electrochemistry: with a chemical reaction model and a method of simulating electro-chemical impedance spectroscopy (EIS). The model given in the present work also forms the foundation for a microstructure-sensitive model of electrochemistry. EIS is a widely-used and powerful diagnostic technique in which the frequency-dependant impedance is measured. Though popular, data from EIS can be notoriously difficult to interpret. The present work also presents simulated EIS data, as well as explanations into the origins of common Nyquist plot features. At high-frequency, an analytic expression for the resistances is derived by analogy to Ohm's law. At low-frequency, the value for the resistance can be predicted by a simulated DC experiment. High- frequency capacitance originates in a difference in the current between the electrode and electrolyte. Low-frequency capacitance is defined by an “effective" surface charge, calculated by integrating the current over time rather than the charge density over distance. Depressed semicircle constant-phase element (CPE) behavior is also observed in the simulated data. Simulations with fast reaction kinetics exhibit power-law CPE impedance behavior, while simulations with a slow or no reaction are best explained by a combination of finite-length diffusion and electromigration. The model developed in this work is a tool to simulate, study, and interpret EIS data. Ultimately, it will serve as the foundation for a microstructure-sensitive model of electrochemistry.
Optimization of Acoustic Pressure Measurements for Impedance Eduction
Jones, M. G.; Watson, W. R.; Nark, D. M.
2007-01-01
As noise constraints become increasingly stringent, there is continued emphasis on the development of improved acoustic liner concepts to reduce the amount of fan noise radiated to communities surrounding airports. As a result, multiple analytical prediction tools and experimental rigs have been developed by industry and academia to support liner evaluation. NASA Langley has also placed considerable effort in this area over the last three decades. More recently, a finite element code (Q3D) based on a quasi-3D implementation of the convected Helmholtz equation has been combined with measured data acquired in the Langley Grazing Incidence Tube (GIT) to reduce liner impedance in the presence of grazing flow. A new Curved Duct Test Rig (CDTR) has also been developed to allow evaluation of liners in the presence of grazing flow and controlled, higher-order modes, with straight and curved waveguides. Upgraded versions of each of these two test rigs are expected to begin operation by early 2008. The Grazing Flow Impedance Tube (GFIT) will replace the GIT, and additional capabilities will be incorporated into the CDTR. The current investigation uses the Q3D finite element code to evaluate some of the key capabilities of these two test rigs. First, the Q3D code is used to evaluate the microphone distribution designed for the GFIT. Liners ranging in length from 51 to 610 mm are investigated to determine whether acceptable impedance eduction can be achieved with microphones placed on the wall opposite the liner. This analysis indicates the best results are achieved for liner lengths of at least 203 mm. Next, the effects of moving this GFIT microphone array to the wall adjacent to the liner are evaluated, and acceptable results are achieved if the microphones are placed off the centerline. Finally, the code is used to investigate potential microphone placements in the CDTR rigid wall adjacent to the wall containing an acoustic liner, to determine if sufficient fidelity can be
Electrical impedance tomography and the fast multipole method
Bikowski, Jutta; Mueller, Jennifer L.
2004-10-01
A 3-D linearization-based reconstruction algorithm for Electrical Impedance Tomography suitable for breast cancer detection using data collected on a rectangular array was introduced by Mueller et al. [IEEE Biomed. Eng., 46(11), 1999]. By considering the scenario as an electrostatic problem, it is possible to model the electrodes with various charges, facilitating the use of the Fast Multipole Method (FMM) for calculating particle interactions and also supporting the use of different electrode models. In this paper the use of FMM is explained and results in form of reconstructed images from experimental data show that this method is an improvement.
Impedance Spectroscopy Measurements of Different Microstructure Plasma Sprayed TBC's
International Nuclear Information System (INIS)
Free standing samples of plasma sprayed zirconia partially stabilized with yettria were prepared with different spraying parameters. The different spraying parameters produces different amounts of pores and micro-tracks. Additionally, the effect of heat treatment at 1200 degree C for 15 h has been studied. Impedance spectroscopy (IS) measurements were performed on as- sprayed and heat-treated specimens at 600 degree C was used to study the effect of porosity and micro-cracks on the electric properties of samples. Specimen's microstructure was examined by optical microscopy
A new method to measure the acoustic surface impedance outdoors
International Nuclear Information System (INIS)
In the European countries noise pollution is considered to be one of the most important environmental problems. With respect to traffic noise, different researchers are working on the reduction of noise at the source, on the modelling of the acoustic absorption of the road structure and on the effects of the pavement on the propagation. The aim of this paper is to propose a new method to measure the acoustic impedance of surfaces located outdoors, which allows us to further noise propagation models, in order to evaluate exactly the noise exposure. (authors)
Chronic impedance spectroscopy of an endovascular stent-electrode array
Opie, Nicholas L.; John, Sam E.; Rind, Gil S.; Ronayne, Stephen M.; Grayden, David B.; Burkitt, Anthony N.; May, Clive N.; O’Brien, Terence J.; Oxley, Thomas J.
2016-08-01
Objective. Recently, we reported a minimally invasive stent-electrode array capable of recording neural signals from within a blood vessel. We now investigate the use of electrochemical impedance spectroscopy (EIS) measurements to infer changes occurring to the electrode–tissue interface from devices implanted in a cohort of sheep for up to 190 days. Approach. In a cohort of 15 sheep, endovascular stent-electrode arrays were implanted in the superior sagittal sinus overlying the motor cortex for up to 190 days. EIS was performed routinely to quantify viable electrodes for up to 91 days. An equivalent circuit model (ECM) was developed from the in vivo measurements to characterize the electrode–tissue interface changes occurring to the electrodes chronically implanted within a blood vessel. Post-mortem histological assessment of stent and electrode incorporation into the wall of the cortical vessels was compared to the electrical impedance measurements. Main results. EIS could be used to infer electrode viability and was consistent with x-ray analysis performed in vivo, and post-mortem evaluation. Viable electrodes exhibited consistent 1 kHz impedances across the 91 day measurement period, with the peak resistance frequency for the acquired data also stable over time. There was a significant change in 100 Hz phase angles, increasing from ‑67.8° ± 8.8° at day 0 to ‑43.8° ± 0.8° at day 91, which was observed to stabilize after eight days. ECM’s modeled to the data suggested this change was due to an increase in the capacitance of the electrode–tissue interface. This was supported by histological assessment with >85% of the implanted stent struts covered with neointima and incorporated into the blood vessel within two weeks. Conclusion. This work demonstrated that EIS could be used to determine the viability of electrode implanted chronically within a blood vessel. Impedance measurements alone were not observed to be a useful predictor of alterations
Impedance adaptation methods of the piezoelectric energy harvesting
Kim, Hyeoungwoo
In this study, the important issues of energy recovery were addressed and a comprehensive investigation was performed on harvesting electrical power from an ambient mechanical vibration source. Also discussed are the impedance matching methods used to increase the efficiency of energy transfer from the environment to the application. Initially, the mechanical impedance matching method was investigated to increase mechanical energy transferred to the transducer from the environment. This was done by reducing the mechanical impedance such as damping factor and energy reflection ratio. The vibration source and the transducer were modeled by a two-degree-of-freedom dynamic system with mass, spring constant, and damper. The transmissibility employed to show how much mechanical energy that was transferred in this system was affected by the damping ratio and the stiffness of elastic materials. The mechanical impedance of the system was described by electrical system using analogy between the two systems in order to simply the total mechanical impedance. Secondly, the transduction rate of mechanical energy to electrical energy was improved by using a PZT material which has a high figure of merit and a high electromechanical coupling factor for electrical power generation, and a piezoelectric transducer which has a high transduction rate was designed and fabricated. The high g material (g33 = 40 [10-3Vm/N]) was developed to improve the figure of merit of the PZT ceramics. The cymbal composite transducer has been found as a promising structure for piezoelectric energy harvesting under high force at cyclic conditions (10--200 Hz), because it has almost 40 times higher effective strain coefficient than PZT ceramics. The endcap of cymbal also enhances the endurance of the ceramic to sustain ac load along with stress amplification. In addition, a macro fiber composite (MFC) was employed as a strain component because of its flexibility and the high electromechanical coupling
Impedance Controller Tuned by Particle Swarm Optimization for Robotic Arms
Directory of Open Access Journals (Sweden)
Haifa Mehdi
2011-11-01
Full Text Available This paper presents an efficient and fast method for fine tuning the controller parameters of robot manipulators in constrained motion. The stability of the robotic system is proved using a Lyapunov‐based impedance approach whereas the optimal design of the controller parameters are tuned, in offline, by a Particle Swarm Optimization (PSO algorithm. For designing the PSOmethod,differentindexperformancesare considered in both joint and Cartesian spaces. A 3DOF manipulator constrained to a circular trajectory is finally used to validate the performances of the proposed approach. The simulation results show the stability and the performances of the proposed approach.
Determination of time delay between ventricles contraction using impedance measurements
International Nuclear Information System (INIS)
The paper presents a novel approach to assessment of ventricular dyssynchrony basing on multichannel electrical impedance measurements. Using a proper placement of electrodes, the sensitivity approach allows estimating time difference between chambers contraction from over determined nonlinear system of equations. The theoretical considerations which include Finite Element Method simulations were verified using measurements on healthy 28 year's old woman. The nonlinear least squares method was applied to obtain a time difference between heart chambers contraction. The obtained value was in a good agreement with theoretical values found in literature.
Impedance calculations for power cables to primary coolant pump motors
International Nuclear Information System (INIS)
The LOFT primary system motor generator sets are located in Room B-239 and are connected to the primary coolant pumps by means of a power cable. The calculated average impedance of this cable is 0.005323 ohms per unit resistance and 0.006025 ohms per unit reactance based on 369.6 kVA and 480 volts. The report was written to show the development of power cable parameters that are to be used in the SICLOPS (Simulation of LOFT Reactor Coolant Loop Pumping System) digital computer program as written in LTR 1142-16 and also used in the pump coastdowns for the FSAR Analysis
Cervical cancer detection by electrical impedance in a Colombian setting
International Nuclear Information System (INIS)
Electrical properties of normal and neoplastic cervical tissues in a heterogeneous group of 56 Colombian women were studied by electrical impedance spectroscopy and a model based on the Generalized Effective-Medium Theory of Induced Polarization (GEMTIP). Differences between the electrical bioimpedance spectra were correlated with cellular and tissue parameters. The analysis performed by the proposed model suggest that the number of different types of cellular layers that form the biological tissue, the intracellular and extracellular conductivity could be used to explain the differences between electrical bioimpedance spectra in normal and neoplastic tissues.
Detection of influenza virus A by using impedance biosensors
Czech Academy of Sciences Publication Activity Database
Meric, B.; Fojt, Lukáš; Özsöz, M.; Vetterl, Vladimír
Rovinj, 2008. s. 281. ISBN 978-9řš-6894-33-8. [1st Regional Symposium on Electrochemistry of South-East Europe. 04.05.2008-08.05.2008, Rovinj] R&D Projects: GA MŠk(CZ) 1M0528; GA ČR(CZ) GA202/08/1688; GA AV ČR(CZ) KAN200040651 Institutional research plan: CEZ:AV0Z50040507; CEZ:AV0Z50040702 Keywords : biosensor * electrochemical impedance spectroscopy * hybridization Subject RIV: BO - Biophysics
A transient model to simulate HTPEM fuel cell impedance spectra
DEFF Research Database (Denmark)
Vang, Jakob Rabjerg; Andreasen, Søren Juhl; Kær, Søren Knudsen
2012-01-01
laboratory. A simultaneous fit to both curves is not achieved. In order to investigate the effects of the fitting parameters on the simulation results, a parameter variation study is carried out. It is concluded that some of the fitting parameters assume values which are not realistic. In order to remedy......This paper presents a spatially resolved transient fuel cell model applied to the simulation of high temperature PEM fuel cell impedance spectra. The model is developed using a 2D finite volume method approach. The model is resolved along the channel and across the membrane. The model considers...
Ionic conductivity measurements of zirconia under pressure using impedance spectroscopy
Takebe, H; Ohtaka, O; Fukui, H; Yoshiasa, A; Yamanaka, T; Ota, K; Kikegawa, T
2002-01-01
We have set up an electrical conductivity measurement system under high-pressure and high-temperature conditions with a multi-anvil high-pressure apparatus using an AC complex impedance method. With this system, we have successfully measured the electrical conductivity of stabilized ZrO sub 2 (Y sub 2 O sub 3 -ZrO sub 2 solid solution) under pressures up to 5 GPa in the temperature range from 300 to 1200 K. The electrical conductivities obtained under pressure are compatible with those of previous results measured at ambient pressure.
RF impedance measurements of DC atmospheric micro-discharges
Overzet, Lawrence J; Mandra, Monali; Goeckner, Matthew; Dufour, Thierry; Dussart, Remi; Lefaucheux, Philippe
2016-01-01
The available diagnostics for atmospheric micro-plasmas remain limited and relatively complex to implement; so we present a radio-frequency technique for diagnosing a key parameter here. The technique allows one to estimate the dependencies of the electron density by measuring the RF-impedance of the micro-plasma and analyzing it with an appropriate equivalent circuit. This technique is inexpensive, can be used in real time and gives reasonable results for argon and helium DC micro-plasmas in holes over a wide pressure range. The electron density increases linearly with current in the expected range consistent with normal glow discharge behavior.
Numerical evaluation of self and mutual earth return impedances
Directory of Open Access Journals (Sweden)
Denisa Stet
2014-12-01
Full Text Available The paper presents an evaluation of analytical self and mutual impedances formulas of lines with earth return, taking into account the ground correction terms. The determined formulas contain semi-infinite integral terms which are calculated by a novel stable and efficient numerical integration scheme in order to overcome the involved oscillation problems. It might seek approximations of the semi-infinite integrals by replacing an exponential or algebraic function, the objective being to permit analytic integration. Since there is no good systematic method for making these replacements, their success depends directly on the intuition and ingenuity, taking into account that in practice the integrand has limited accuracy.
CdS based novel photo-impedance light sensor
International Nuclear Information System (INIS)
A novel photo-impedance (hybrid photocapacitive-photoresistive) sensor for visible light employs gated cadmium sulfide photoresistance connecting two fixed capacitances. The changes of the photoresistance and of the metal–insulator (dielectric)-semiconductor (MIS) under illumination affect the overall device capacitance. The MIS and photoresistor frequency dispersion allows for increasing the dynamic range by measuring the capacitance response at lower frequencies. By making this device a part of the tank circuit of a radio frequency oscillator, the information on illumination intensity can be relayed as the signal frequency and detected remotely. (paper)
Design study of a movable mask with low beam impedance
International Nuclear Information System (INIS)
A novel design of a movable-mask (collimator) for high current accelerators is proposed. The mask head is supported by a ceramics rod to reduce the interference with beam. One side of the rod is coated by a thin conductive material to avoid charge up of the head. The head is made of graphite or ceramics coated with copper to avoid the direct damage by an intense beam. The impedances of trapped modes, the loss factors and so on were evaluated by simulation codes. The frequencies and the Q factors of the trapped modes were also measured using a cold model, and compared with calculated ones. (author)
Effective impedance for predicting the existence of surface states
Xiao, Meng; Huang, Xueqin; Fang, Anan; Chan, C. T.
2016-03-01
We build an effective impedance for two-dimensional (2D) photonic crystals (PCs) comprising a rectangular lattice of dielectric cylinders with the incident electric field polarized along the axis of the cylinders. In particular, we discuss the feasibility of constructing an effective impedance for the case where the Bloch wave vector is far away from the center of Brillouin zone, where the optical response of the PC is necessarily anisotropic, and hence the effective description becomes inevitably angle dependent. We employ the scattering theory and treat the 2D system as a stack of 1D arrays. We consider only the zero-order interlayer diffraction, and all the higher order diffraction terms of interlayer scattering are ignored. This approximation works well when the higher order diffraction terms are all evanescent waves and the interlayer distance is far enough for them to decay out. Scattering theory enables the calculation of transmission and reflection coefficients of a finite-sized slab, and we extract the effective parameters such as the effective impedance (Ze) and the effective refractive index (ne) using a parameter retrieval method. We note that ne is uniquely defined only in a very limited region of the reciprocal space. (nek0a ≪1 , where k0 is the wave vector inside the vacuum and a is thickness of the slab for retrieval), but Ze is uniquely defined and has a well-defined meaning inside a much larger domain in the reciprocal space. For a lossless system, the effective impedance Ze is purely real for the pass band and purely imaginary in the band gaps. Using the sign of the imaginary part of Ze, we can classify the band gaps into two groups, and this classification explains why there is usually no surface state on the boundary of typical fully gapped PCs composed of a lattice of dielectric cylinders. This effective medium approach also allows us to predict the dispersion of surface states even when the surface wave vectors are well beyond the zone
Magdić, K.; Horvat-Radošević, V.
2013-01-01
This article describes the basic principles of the Electrochemical Impedance Spectroscopy, EIS, technique and its application in the characterization of electrode materials and electrochemical devices for energy conversion and storage. The concept of impedance and the necessary steps of impedance spectra analysis are explained. Basic relations for impedance of particular electro- chemical processes occurring in the system(s) are derived. Electri...
A MEMS-based valveless impedance pump utilizing electromagnetic actuation
International Nuclear Information System (INIS)
This study presents a planar valveless impedance-based micro pump for biomedical applications. The micro pump comprises four major components, namely a lower glass substrate containing a copper micro coil, a microchannel, an upper glass cover plate and a PDMS diaphragm with a magnet mounted on its upper surface. When a current is passed through the micro coil, an electromagnetic force is established between the coil and the magnet. The resulting deflection of the PDMS diaphragm creates an acoustic impedance mismatch within the microchannel, which results in a net flow. The performance of the micro pump is characterized both experimentally and numerically using Ansoft/Maxwell3D FEA software. The results show that the mechanical integrity of the micro pump is assured provided that the diaphragm deflection does not exceed 110 µm. This deflection is obtained by supplying the micro coil with an input current of 0.6 A, and results in a flow rate of 7.2 ml min−1 when the PDMS membrane is driven by an actuating frequency of 200 Hz
A MEMS-based valveless impedance pump utilizing electromagnetic actuation
Lee, Chia-Yen; Chang, Hsien-Tsung; Wen, Chih-Yung
2008-03-01
This study presents a planar valveless impedance-based micro pump for biomedical applications. The micro pump comprises four major components, namely a lower glass substrate containing a copper micro coil, a microchannel, an upper glass cover plate and a PDMS diaphragm with a magnet mounted on its upper surface. When a current is passed through the micro coil, an electromagnetic force is established between the coil and the magnet. The resulting deflection of the PDMS diaphragm creates an acoustic impedance mismatch within the microchannel, which results in a net flow. The performance of the micro pump is characterized both experimentally and numerically using Ansoft/Maxwell3D FEA software. The results show that the mechanical integrity of the micro pump is assured provided that the diaphragm deflection does not exceed 110 µm. This deflection is obtained by supplying the micro coil with an input current of 0.6 A, and results in a flow rate of 7.2 ml min-1 when the PDMS membrane is driven by an actuating frequency of 200 Hz.
A review on electrical impedance tomography for pulmonary perfusion imaging
International Nuclear Information System (INIS)
Although electrical impedance tomography (EIT) for ventilation monitoring is on the verge of clinical trials, pulmonary perfusion imaging with EIT remains a challenge, especially in spontaneously breathing subjects. In anticipation of more research on this subject, we believe a thorough review is called for. In this paper, findings related to the physiological origins and electrical characteristics of this signal are summarized, highlighting properties that are particularly relevant to EIT. The perfusion impedance change signal is significantly smaller in amplitude compared with the changes due to ventilation. Therefore, the hardware used for this purpose must be more sensitive and more resilient to noise. In previous works, some signal- or image-processing methods have been required to separate these two signals. Three different techniques are reviewed in this paper, including the ECG-gating method, frequency-domain-filtering-based methods and a principal-component-analysis-based method. In addition, we review a number of experimental studies on both human and animal subjects that employed EIT for perfusion imaging, with promising results in the diagnosis of pulmonary embolism (PE) and pulmonary arterial hypertension as well as other potential applications. In our opinion, PE is most likely to become the main focus for perfusion EIT in the future, especially for heavily instrumented patients in the intensive care unit (ICU). (paper)
An instrument for simultaneous EQCM impedance and SECM measurements.
Gollas, B; Bartlett, P N; Denuault, G
2000-01-15
A novel combination of an electrochemical quartz crystal microbalance (EQCM) and a scanning electrochemical microscope (SECM) has been built. Unlike conventional EQCMs, the instrument described here allows rapid in situ measurement of the modulus of the quartz crystal's transfer function. Data analysis in the complex plane for the Butterworth-Van Dyke (BVD) equivalent circuit yields the real and the imaginary components R (damping resistance) and XL (reactive inductance) of the crystal's electroacoustic impedance around its resonant frequency of 10 MHz. The influence of different tip shapes of an approaching microelectrode on the electroacoustic impedance of the quartz crystal was studied and found to be minimal for certain geometries. The capability of the EQCM/SECM instrument was tested in cyclic voltammetric plating/stripping experiments using a copper(I) chloride solution of high concentration in 1 M HCl. Four parameters, XL, R, the substrate, and the tip current, can be recorded simultaneously. Depletion layer effects were observed and could be corrected for to yield accurate current efficiencies for potentiodynamic and potentiostatic copper plating. The amperometric response of the SECM tip positioned closely to the substrate reflects the concentration changes of electroactive ions in the diffusion layer of the substrate electrode. PMID:10658330
Grinding process monitoring based on electromechanical impedance measurements
Marchi, Marcelo; Guimarães Baptista, Fabricio; de Aguiar, Paulo Roberto; Bianchi, Eduardo Carlos
2015-04-01
Grinding is considered one of the last processes in precision parts manufacturing, which makes it indispensable to have a reliable monitoring system to evaluate workpiece surface integrity. This paper proposes the use of the electromechanical impedance (EMI) method to monitor the surface grinding operation in real time, particularly the surface integrity of the ground workpiece. The EMI method stands out for its simplicity and for using low-cost components such as PZT (lead zirconate titanate) piezoelectric transducers. In order to assess the feasibility of applying the EMI method to the grinding process, experimental tests were performed on a surface grinder using a CBN grinding wheel and a SAE 1020 steel workpiece, with PZT transducers mounted on the workpiece and its holder. During the grinding process, the electrical impedance of the transducers was measured and damage indices conventionally used in the EMI method were calculated and compared with workpiece wear, indicating the surface condition of the workpiece. The experimental results indicate that the EMI method can be an efficient and cost-effective alternative for monitoring precision workpieces during the surface grinding process.
Studies of deionization and impedance spectroscopy for blood analyzer
Kwong, Charlotte C.; Li, Nan; Ho, Chih-Ming
2005-11-01
Blood analysis provides vital information for health conditions. For instance, typical infection response is correlated to an elevated White Blood Cell (WBC) count, while low Red Blood Cell (RBC) count, hemoglobin and hematocrit are caused by anemia or internal bleeding. We are developing two essential modules, deionization (DI) chip and microfluidic cytometer with impedance spectroscopy flow, for enabling the realization of a single platform miniaturized blood analyzer. In the proposed analyzer, blood cells are preliminarily sorted by Dielectrophoretic (DEP) means into sub-groups, differentiated and counted by impedance spectroscopy in a flow cytometer. DEP techniques have been demonstrated to stretch DNA, align Carbon Nanotubes (CNT) and trap cells successfully. However, DEP manipulation does not function in biological media with high conductivity. The DI module is designed to account for this challenge. H Filter will serve as an ion extraction platform in a microchamber. Sample and buffer do not mix well in micro scale allowing the ions being extracted by diffusion without increasing the volume. This can keep the downstream processing time short. Micro scale hydrodynamic focusing is employed to place single cell passing along the central plane of the flow cytometer module. By applying an AC electrical field, suspended cells are polarized, membrane capacitance C m, cytoplasm conductivity σ c, and cytoplasm permittivity ɛ c will vary as functions of frequency. Tracing back the monitored current, the numbers of individual cell species can be evaluated.
Circuit modeling of the electrical impedance: I. Neuromuscular disease
International Nuclear Information System (INIS)
Multifrequency electrical impedance myography (MFEIM) in the 3–300 kHz range was applied to 68 subjects representing 19 different neuromuscular diseases, and the impedances analyzed using the 5-element circuit model. Depending on severity, the ‘cellular’ parameters r2, r3, 1/c1 and 1/c2 were found to be as much as 10- to 20-fold larger than for normal subjects (taking age and girth into account), but in almost every case the extracellular fluid parameter r1 was at most only marginally affected. Strong correlations are found between r2 and 1/c1, but in the case of ALS that breaks down when c1 (representing the muscle fiber membrane capacitance) falls below half the normal value. Also, c2 (tentatively associated with intracellular organelle membranes) was found to be the most sensitive to disease progress in ALS, about three times more so than the 50 kHz phase, already suggested for use in clinical drug testing. We conclude that following parameters obtained using the combined MFEIM/5-element circuit analysis scheme offer a reliable, non-invasive and objective way of characterizing muscle in neuromuscular disease or during clinical drug testing. (paper)
Simplified signal processing for impedance spectroscopy with spectrally sparse sequences
International Nuclear Information System (INIS)
Classical method for measurement of the electrical bio-impedance involves excitation with sinusoidal waveform. Sinusoidal excitation at fixed frequency points enables wide variety of signal processing options, most general of them being Fourier transform. Multiplication with two quadrature waveforms at desired frequency could be easily accomplished both in analogue and in digital domains, even simplest quadrature square waves can be considered, which reduces signal processing task in analogue domain to synchronous switching followed by low pass filter, and in digital domain requires only additions. So called spectrally sparse excitation sequences (SSS), which have been recently introduced into bio-impedance measurement domain, are very reasonable choice when simultaneous multifrequency excitation is required. They have many good properties, such as ease of generation and good crest factor compared to similar multisinusoids. Typically, the usage of discrete or fast Fourier transform in signal processing step is considered so far. Usage of simplified methods nevertheless would reduce computational burden, and enable simpler, less costly and less energy hungry signal processing platforms. Accuracy of the measurement with SSS excitation when using different waveforms for quadrature demodulation will be compared in order to evaluate the feasibility of the simplified signal processing. Sigma delta modulated sinusoid (binary signal) is considered to be a good alternative for a synchronous demodulation.
Biomass measurement of living Lumbriculus variegatus with impedance spectroscopy
Directory of Open Access Journals (Sweden)
Martina Sammer
2014-12-01
Full Text Available Impedance spectroscopy is a useful tool for non-invasive and real time measurements of cell suspensions and a variety of biological tissues. The objective of this study was the investigation of the dielectric properties of living aquatic worms (Lumbriculus variegatus using impedance spectroscopy in a frequency range between 100 Hz and 10 MHz. We demonstrate a linear relation between the worm biomass and the phase response of the signal thereby providing a quick and precise method to determine the biomass of aquatic worms in situ. Possible applications for non-destructive online biomass monitoring of aquatic worms and other aqueous organisms are discussed. Furthermore, we show that groups of worms fed different diets can be distinguished by the method presented. These results reveal a close relationship between the nutritional composition of the worms and the measured phase response. We also demonstrate that the phase response at 90 kHz does not depend on the worm size. In contrast, the response function for the signal at 440 Hz reveals a linear correlation of average individual worm size and phase. Therefore, we conclude that the measured phase response at 90 kHz qualifies as a measure of the total amount of worm biomass present in the measuring cell, whereas the phase measurement at 440 Hz can be used to estimate the average individual worm size.
SVM for prostate cancer using electrical impedance measurements
International Nuclear Information System (INIS)
Biopsies are currently the 'gold standard' method for identifying cancer of the prostate. While biopsies yield very accurate information regarding the area they sample, they are performed at discrete points and provide no information on the adjacent tissue. To enhance procedural accuracy, biopsies of a large number of sites are routinely carried out. Although more accurate, this method is both more complex and nevertheless remains discrete. In this paper, we evaluate the advantages of using bio-impedance information as the input for a support vector machines (SVMs) classifier to overcome these limitations. In this method, the biopsy probes are used as electrodes to obtain electrical impedance data during each biopsy sample. Using a computer model of the prostate, a SVM was trained and tested. Different tumor shapes and conductivity values, and the classifier's ability to generalize to these different properties, were examined. We demonstrate that by using this classifier the number of biopsies can be reduced and valuable information concerning the adjacent tissue which was not biopsied can be generated
Electrical impedance myography at frequencies up to 2 MHz
International Nuclear Information System (INIS)
Extension of the frequency range of electrical impedance myography (EIM) to 2 MHz discloses a major rise in the reactance of muscle above 3–500 kHz, together with a slow decrease in the resistance consonant with the Kramers–Kronig relations. This 'upturn' phenomenon is found when the distant current electrode configuration of EIM is employed, but not when the current electrodes are placed close to the voltage measuring area. In that case the impedance qualitatively mimics the commonly used 3-element resistor–capacitor model. The possibility that the upturn is an artifact of the measuring system rather than a true property of the tissue is examined in detail. In particular, experiments are reported which argue against the transmission line mechanism proposed to explain similar increases in reactance in some high frequency whole-body BIA studies. Also, scaling of X versus R plots for muscle segments of different lengths strongly suggests that the upturn is as much a property of the underlying tissue as is the low frequency maximum in reactance
Modeling of low-frequency pulmonary impedance in dogs.
Hantos, Z; Daróczy, B; Csendes, T; Suki, B; Nagy, S
1990-03-01
The mechanical impedance of the lungs (ZL) was measured in open-chest dogs with small-amplitude pseudorandom volume oscillations between 0.125 and 5 Hz, at mean transpulmonary pressures (Ptp) of 0.2, 0.4, and 0.8 kPa. At the lowest frequencies, the pulmonary resistance showed a marked negative frequency dependence and mirrored the changes in the reactance with altered Ptp. The ZL data were evaluated on the basis of two models, each containing the same airway compartment with a resistance and an inertance. The tissue impedance (Zti) in model 1 was represented with two compliances and a resistance (L. E. Mount. J. Physiol. Lond. 127: 157-167, 1955), whereas in model 2 a two-parameter formulation implying rate-independent dissipated work and frequency-dependent elastance (J. Hildebrandt. J. Appl. Physiol. 28: 365-372, 1970) was employed. The estimation of model parameters showed that model 2 was superior to model 1 in both fitting performance and parameter insensitivity to weighting in the fitting criterion. The model 2 coefficients of damping and elastance, characterizing the real and imaginary parts of Zti, respectively, depended on the lung distension and were closely correlated. Although ZL exhibited a slight dependence on the peak-to-peak volume excursion, at a given oscillatory volume no inconsistency with linear tissue viscoelasticity was detected. PMID:2341352
Tunable acoustic radiation pattern assisted by effective impedance boundary
Qian, Feng; Quan, Li; Wang, Li-Wei; Liu, Xiao-Zhou; Gong, Xiu-Fen
2016-02-01
The acoustic wave propagation from a two-dimensional subwavelength slit surrounded by metal plates decorated with Helmholtz resonators (HRs) is investigated both numerically and experimentally in this work. Owing to the presence of HRs, the effective impedance of metal surface boundary can be manipulated. By optimizing the distribution of HRs, the asymmetric effective impedance boundary will be obtained, which contributes to generating tunable acoustic radiation pattern such as directional acoustic beaming. These dipole-like radiation patterns have high radiation efficiency, no fingerprint of sidelobes, and a wide tunable range of the radiation pattern directivity angle which can be steered by the spatial displacements of HRs. Project supported by the National Basic Research Program of China (Grant Nos. 2012CB921504 and 2011CB707902), the National Natural Science Foundation of China (Grant No.11474160), the Fundamental Research Funds for Central Universities, China (Grant No. 020414380001), the State Key Laboratory of Acoustics, Chinese Academy of Sciences (Grant No. SKLOA201401), the Priority Academic Program Development of Jiangsu Higher Education Institutions, and the Scientific Research Foundation for the Returned Overseas Chinese Scholars, State Education Ministry.
Kinetic Damping in the Spectrum of the Spherical Impedance Probe
Oberrath, Jens; Brinkmann, Ralf Peter
2015-09-01
Active plasma resonance spectroscopy is a widely used diagnostic method and several probes in different designs have been invented. One of them is the Spherical Impedance Probe. Its resonance behavior and the influence of kinetic effects on it can be described by a general kinetic model presented by the authors. It was theoretically shown that kinetic effects are responsible for a broadening of the resonance peak in the spectrum. However, the broadening of the resonance peak in a kinetically determined spectrum in the geometry of an existing probe is not evaluated, yet. We present such a spectrum of the Spherical Impedance Probe. Therefore, the general solution of the model is expanded in an orthonormal system of basis-functions. This expansion is truncated to determine an approximated spectrum. Its resonance peak shows clearly a broadening compared to a peak in a spectrum, which is determined by a fluiddynamical model. The authors acknowledge the support by the Research Service of Leuphana University Lueneburg, the Deutsche Forschungsgemeinschaft via the Ruhr University Research School and the Federal Ministry of Education and Research in frame of the PluTO+ projekt.
Determinants of pulmonary perfusion measured by electrical impedance tomography.
Smit, Henk J; Vonk Noordegraaf, Anton; Marcus, J Tim; Boonstra, Anco; de Vries, Peter M; Postmus, Pieter E
2004-06-01
Electrical impedance tomography (EIT) is a non-invasive imaging technique for detecting blood volume changes that can visualize pulmonary perfusion. The two studies reported here tested the hypothesis that the size of the pulmonary microvascular bed, rather than stroke volume (SV), determines the EIT signal. In the first study, the impedance changes relating to the maximal pulmonary pulsatile blood volume during systole (Delta Z(sys)) were measured in ten healthy subjects, ten patients diagnosed with chronic obstructive pulmonary disease, who were considered to have a reduced pulmonary vascular bed, and ten heart failure patients with an assumed low cardiac output but with a normal lung parenchyma. Mean Delta Z(sys) (SD) in these groups was 261 (34)x10(-5), 196 (39)x10(-5) ( Pbicycle while SV was measured by means of magnetic resonance imaging. The Delta Z(sys) at rest was 352 (53)x10(-5 ) and 345 (112)x10(-5 )AU during exercise (P=NS), whereas SV increased from 83 (21) to 105 (34) ml (P<0.05). The EIT signal likely reflects the size of the pulmonary microvascular bed, since neither a low cardiac output nor a change in SV of the heart appear to influence EIT. PMID:14985995
Numerical calculation of radiation pattern and impedance of column plasma
International Nuclear Information System (INIS)
When the frequency of the incident electromagnetic wave is lower than that of the plasma, the high-density and low-temperature plasma column is equivalent to a good conductor. However, because of the dispersive relation and the uneven distribution of density, the wave vector propagating along the plasmas is not a real constant, but a complex quantity varying with the change in the space. So, it is hard to calculate the radiation pattern, radiation impedance and gain of the column plasma with the normal methods used with the metallic antenna. In this paper, the space distribution of the wave vectors of the signal along the column plasma is calculated from the dispersive relation of the non-magnetized columnar plasma. And then, according to the antenna equation, the radiation pattern, radiation impedance and gain of the columnar plasma with even or uneven axial density distribution can be derived. The experimental results show that the numerical values calculated by this new method are much more accurate. This method will be of great importance in the application of the plasmas in the antenna transportation. (authors)
TCSC impedance regulator applied to the second benchmark model
Energy Technology Data Exchange (ETDEWEB)
Hamel, J.P.; Dessaint, L.A. [Ecole de Technologie Superieure, Montreal, PQ (Canada). Dept. of Electrical Engineering; Champagne, R. [Ecole de Technologie Superieure, Montreal, PQ (Canada). Dept. of Software and IT Engineering; Pare, D. [Institut de Recherche d' Hydro-Quebec, Varennes, PQ (Canada)
2008-07-01
Due to the combination of electrical demand growth and the high cost of building new power transmission lines, series compensation is increasingly used in power systems all around the world. Series compensation has been proposed as a new way to transfer more power on existing lines. By adding series compensation to an existing line (a relatively small change), the power transfer can be increased significantly. One of the means used for line compensation is the addition of capacitive elements in series with the line. This paper presented a thyristor-controlled series capacitor (TCSC) model that used impedance as reference, had individual controls for each phase, included a linearization module and considered only the fundamental frequency for impedance computations, without using any filter. The model's dynamic behavior was validated by applying it to the second benchmark model for subsynchronous resonance (SSR). Simulation results from the proposed model, obtained using EMTP-RV and SimPowerSystems were demonstrated. It was concluded that SSR was mitigated by the proposed approach. 19 refs., 19 figs.
Characterization of Physiological Glucose Concentration Using Electrical Impedance Spectroscopy
Directory of Open Access Journals (Sweden)
Quazi Delwar Hossain
2013-01-01
Full Text Available Non-invasive glucose monitoring is crucial for effective diabetes mellitus treatment while a sound correlation of a non-invasive parameter to glucose level variation is quite challenging. This paper presents characterization of glucose concentrations using Electrical Impedance Spectroscopy (EIS in three different solutions: 1 0.9% NaCl, 2 Saline (NaCl 1.3gm, KCl 0.75gm, Na3C6H5O7 1.45gm, D-glucose 6.75gm in 500mL and 3 Human Blood for every 25mg/dl change of glucose in total 150ml solution. A rectangular current pulse of 1.5s duration with 1mA peak is applied to the solutions and corresponding voltage is acquired across the solutions with Agilent InfiniiVision 7000B Series oscilloscope and Matlab R2011a Instrument Control Toolbox. The circuit proposed for current injection and voltage acquisition requires only two electrodes would reduce electrode polarization and skin irritation greatly which is a major concern in many previous works use generally four electrodes. Experimental results show sound correlation between EIS and blood glucose concentration. It is clearly found from the EIS that the DC impedance of solutions increases linearly with the increment in glucose concentrations.
Simplified signal processing for impedance spectroscopy with spectrally sparse sequences
Annus, P.; Land, R.; Reidla, M.; Ojarand, J.; Mughal, Y.; Min, M.
2013-04-01
Classical method for measurement of the electrical bio-impedance involves excitation with sinusoidal waveform. Sinusoidal excitation at fixed frequency points enables wide variety of signal processing options, most general of them being Fourier transform. Multiplication with two quadrature waveforms at desired frequency could be easily accomplished both in analogue and in digital domains, even simplest quadrature square waves can be considered, which reduces signal processing task in analogue domain to synchronous switching followed by low pass filter, and in digital domain requires only additions. So called spectrally sparse excitation sequences (SSS), which have been recently introduced into bio-impedance measurement domain, are very reasonable choice when simultaneous multifrequency excitation is required. They have many good properties, such as ease of generation and good crest factor compared to similar multisinusoids. Typically, the usage of discrete or fast Fourier transform in signal processing step is considered so far. Usage of simplified methods nevertheless would reduce computational burden, and enable simpler, less costly and less energy hungry signal processing platforms. Accuracy of the measurement with SSS excitation when using different waveforms for quadrature demodulation will be compared in order to evaluate the feasibility of the simplified signal processing. Sigma delta modulated sinusoid (binary signal) is considered to be a good alternative for a synchronous demodulation.
Nonlocal impedances and the Casimir entropy at low temperatures
International Nuclear Information System (INIS)
The problem with the temperature dependence of the Casimir force is investigated. Specifically, the entropy behavior in the low temperature limit, which caused debates in the literature, is analyzed. It is stressed that the behavior of the relaxation frequency in the T→0 limit does not play a physical role since the anomalous skin effect dominates in this range. In contrast with the previous works, where the approximate Leontovich impedance was used for analysis of nonlocal effects, we give description of the problem in terms of exact nonlocal impedances. It is found that the Casimir entropy is going to zero at T→0 only in the case when s polarization does not contribute to the classical part of the Casimir force. However, the entropy approaching zero from the negative side that, in our opinion, cannot be considered as thermodynamically satisfactory. The resolution of the negative entropy problem proposed in the literature is analyzed and it is shown that it cannot be considered as complete. The crisis with the thermal Casimir effect is stressed
Femoral Test Bed for Impedance Controlled Surgical Instrumentation
Directory of Open Access Journals (Sweden)
Christian Brendle
2012-01-01
Full Text Available The risk for patients during the standard procedure of revision of cemented artificial hip joints is unsatisfactorily highdue to its high level of invasiveness and limited access to the operative field. To reduce this risk we are developing anImpedance Controlled Surgical Instrumentation (ICOS system, which aims to establish real-time control during a BoneCement (BC milling process. For this, the relationship between the thickness of the BC and its frequency-dependentelectrical impedance is used to estimate the residual BC thickness. The aim is to avoid unintended cutting of boneby detecting the passage of the BC/bone boundary layer by the milling head. In a second step, an estimation of theresidual BC thickness will be used to improve process control. As a first step towards demonstrating the feasibility ofour approach, presented here are experimental studies to characterize the BC permittivity and to describe the process indetail. The results show that the permittivity properties of BC are dominated by its polymethyl methacrylate (PMMAfraction. Thus, PMMA can be used as a substitute for future experiments. Furthermore, a Femoral Test Bed (FTB wasdesigned. Using this setup we show it is feasible to accurately distinguish between slightly different thicknesses of BC.
Electrical conduction in polycrystalline CVD diamond: Temperature dependent impedance measurements
Energy Technology Data Exchange (ETDEWEB)
Ye, H.; Williams, O.A.; Jackman, R.B. [Electronic and Electrical Engineering, University College London, Torrington Place, London WC1E 7JE (United Kingdom); Rudkin, R.; Atkinson, A. [Department of Materials, Imperial College London, South Kensington, London SW7 2AZ (United Kingdom)
2002-10-16
In this paper, we report the first measurement of impedance on freestanding diamond films from 0.1 Hz to 10 MHz up to 300 C. A wide range of CVD materials have been investigated, but here we concentrate on 'black' diamond grown by MWPECVD. The Cole-Cole (Z' via Z{sup ''}) plots are well fitted to a RC parallel circuit model and the equivalent resistance and capacitance for the diamond films have been estimated using the Zview curve fitting. The results show only one single semicircle response at each temperature measured. It was found that the resistance decreases from 62 M{omega} at room temperature to 4 k{omega} at 300 C, with an activation energy around 0.51 eV. The equivalent capacitance is maintained at the level of 10{sup 2} pF up to 300 C, suggesting that the diamond grain boundaries are dominating the conduction. At 400 C, the impedance at low frequencies shows a linear tail, which can be explained that the ac polarization of diamond/Au interface occurs. (Abstract Copyright [2002], Wiley Periodicals, Inc.)
An Electrochemical Impedance Spectroscopy System for Monitoring Pineapple Waste Saccharification
Conesa, Claudia; Ibáñez Civera, Javier; Seguí, Lucía; Fito, Pedro; Laguarda-Miró, Nicolás
2016-01-01
Electrochemical impedance spectroscopy (EIS) has been used for monitoring the enzymatic pineapple waste hydrolysis process. The system employed consists of a device called Advanced Voltammetry, Impedance Spectroscopy & Potentiometry Analyzer (AVISPA) equipped with a specific software application and a stainless steel double needle electrode. EIS measurements were conducted at different saccharification time intervals: 0, 0.75, 1.5, 6, 12 and 24 h. Partial least squares (PLS) were used to model the relationship between the EIS measurements and the sugar determination by HPAEC-PAD. On the other hand, artificial neural networks: (multilayer feed forward architecture with quick propagation training algorithm and logistic-type transfer functions) gave the best results as predictive models for glucose, fructose, sucrose and total sugars. Coefficients of determination (R2) and root mean square errors of prediction (RMSEP) were determined as R2 > 0.944 and RMSEP 0.973 and RMSEP < 0.486 for artificial neural networks (ANNs), respectively. Therefore, a combination of both an EIS-based technique and ANN models is suggested as a promising alternative to the traditional laboratory techniques for monitoring the pineapple waste saccharification step. PMID:26861317
An Electrochemical Impedance Spectroscopy System for Monitoring Pineapple Waste Saccharification
Directory of Open Access Journals (Sweden)
Claudia Conesa
2016-02-01
Full Text Available Electrochemical impedance spectroscopy (EIS has been used for monitoring the enzymatic pineapple waste hydrolysis process. The system employed consists of a device called Advanced Voltammetry, Impedance Spectroscopy & Potentiometry Analyzer (AVISPA equipped with a specific software application and a stainless steel double needle electrode. EIS measurements were conducted at different saccharification time intervals: 0, 0.75, 1.5, 6, 12 and 24 h. Partial least squares (PLS were used to model the relationship between the EIS measurements and the sugar determination by HPAEC-PAD. On the other hand, artificial neural networks: (multilayer feed forward architecture with quick propagation training algorithm and logistic-type transfer functions gave the best results as predictive models for glucose, fructose, sucrose and total sugars. Coefficients of determination (R2 and root mean square errors of prediction (RMSEP were determined as R2 > 0.944 and RMSEP < 1.782 for PLS and R2 > 0.973 and RMSEP < 0.486 for artificial neural networks (ANNs, respectively. Therefore, a combination of both an EIS-based technique and ANN models is suggested as a promising alternative to the traditional laboratory techniques for monitoring the pineapple waste saccharification step.
Elliptic cylinder geometry for distinguishability analysis in impedance tomography.
Saka, Birsen; Yilmaz, Atila
2004-01-01
Electrical impedance tomography (EIT) is a technique that computes the cross-sectional impedance distribution within the body by using current and voltage measurements made on the body surface. It has been reported that the image reconstruction is distorted considerably when the boundary shape is considered to be more elliptical than circular as a more realistic shape for the measurement boundary. This paper describes an alternative framework for determining the distinguishability region with a finite measurement precision for different conductivity distributions in a body modeled by elliptic cylinder geometry. The distinguishable regions are compared in terms of modeling error for predefined inhomogeneities with elliptical and circular approaches for a noncircular measurement boundary at the body surface. Since most objects investigated by EIT are noncircular in shape, the analytical solution for the forward problem for the elliptical cross section approach is shown to be useful in order to reach a better assessment of the distinguishability region defined in a noncircular boundary. This paper is concentrated on centered elliptic inhomogeneity in the elliptical boundary and an analytic solution for this type of forward problem. The distinguishability performance of elliptical cross section with cosine injected current patterns is examined for different parameters of elliptical geometry. PMID:14723501
BPM Design and Impedance Considerations for a Rotatable Collimator for the LHC Collimation Upgrade
Energy Technology Data Exchange (ETDEWEB)
Smith, Jeffrey Claiborne; /SLAC; Keller, Lewis; /SLAC; Lundgren, Steven; /SLAC; Markiewicz, Thomas; /SLAC; Young, Andrew; /SLAC
2010-08-26
The Phase II upgrade to the LHC collimation system calls for complementing the 30 high robust Phase I graphite secondary collimators with 30 high Z Phase II collimators. This paper reports on BPM and impedance considerations and measurements of the integrated BPMs in the prototype rotatable collimator to be installed in the Super Proton Synchrotron (SPS) at CERN. The BPMs are necessary to align the jaws with the beam. Without careful design the beam impedance can result in unacceptable heating of the chamber wall or beam instabilities. The impedance measurements involve utilizing both a single displaced wire and two wires excited in opposite phase to disentangle the driving and detuning transverse impedances. Trapped mode resonances and longitudinal impedance are to also be measured and compared with simulations. These measurements, when completed, will demonstrate the device is fully operational and has the impedance characteristics and BPM performance acceptable for installation in the SPS.
Impedance matching and emission properties of optical antennas in a nanophotonic circuit
Huang, Jer-Shing; Biagioni, Paolo; Hecht, Bert
2008-01-01
An experimentally realizable prototype nanophotonic circuit consisting of a receiving and an emitting nano antenna connected by a two-wire optical transmission line is studied using finite-difference time- and frequency-domain simulations. To optimize the coupling between nanophotonic circuit elements we apply impedance matching concepts in analogy to radio frequency technology. We show that the degree of impedance matching, and in particular the impedance of the transmitting nano antenna, can be inferred from the experimentally accessible standing wave pattern on the transmission line. We demonstrate the possibility of matching the nano antenna impedance to the transmission line characteristic impedance by variations of the antenna length and width realizable by modern microfabrication techniques. The radiation efficiency of the transmitting antenna also depends on its geometry but is independent of the degree of impedance matching. Our systems approach to nanophotonics provides the basis for realizing gener...
Damage Assessment of Aerospace Structural Components by Impedance Based Health Monitoring
Gyekenyesi, Andrew L.; Martin, Richard E.; Sawicki, Jerzy T.; Baaklini, George Y.
2005-01-01
This paper addresses recent efforts at the NASA Glenn Research Center at Lewis Field relating to the set-up and assessment of electro-mechanical (E/M) impedance based structural health monitoring. The overall aim is the application of the impedance based technique to aeronautic and space based structural components. As initial steps, a laboratory was created, software written, and experiments conducted on aluminum plates in undamaged and damaged states. A simulated crack, in the form of a narrow notch at various locations, was analyzed using piezoelectric-ceramic (PZT: lead, zirconate, titarate) patches as impedance measuring transducers. Descriptions of the impedance quantifying hardware and software are provided as well as experimental results. In summary, an impedance based health monitoring system was assembled and tested. The preliminary data showed that the impedance based technique was successful in recognizing the damage state of notched aluminum plates.
A new three-dimensional electromechanical impedance model for an embedded dual-PZT transducer
Wang, Dansheng; Li, Zhi; Zhu, Hongping
2016-07-01
In the past twenty years, the electromechanical (EM) impedance technique has been investigated extensively in the mechanical, aviation and civil engineering fields. Many different EM impedance models have been proposed to characterize the interaction between the surface-bonded PZT transducer and the host structure. This paper formulates a new three-dimensional EM impedance model characterizing the interaction between an embedded circle dual-PZT transducer and the host structure based on the effective impedance concept. The proposed model is validated by experimental results from a group of smart cement cubes, in which three circle dual-PZT transducers are embedded respectively. In addition, a new EM impedance measuring method for the dual-PZT transducer is also introduced. In the measuring method, only a common signal generator and an oscilloscope are needed, by which the exciting and receiving voltage signals are obtained respectively. Combined with fast Fourier transform the EM impedance signatures of the dual-PZT transducers are obtained.
The influence of acupuncture on the impedance measured by four electrodes on meridians.
Zhang, W; Xu, R; Zhu, Z
1999-01-01
The impedance on the pericardium merdian near Quze (P3) and control points in 12 cases was measured by a four electrodes impedance instrument. The amplitude of impedance was recorded before, during and after needling the Neiguan. The result showed that the mean impedance on the meridian and control points before the needling were 52.8 +/- 11.0 (omega) and 61.7 +/- 10.3 (omega) respectively which had significant difference (P0.05). The impedance changed back to 51.1 +/- 11.3 (omega) and 59.9 +/- 11.0 (omega) on the meridian and control points respectively during the 5-10 minutes after withdrawing the needling. In some cases, impedance changed intermittently during the needling. The experiment implies that interstitial fluid increases during the needling by axon reflection and blood capillary expanding which may be one of the mechanisms of acupuncture regulation. PMID:10768415
Magdić, Katja; Horvat-Radošević, Višnja
2013-01-01
This article describes the basic principles of the Electrochemical Impedance Spectroscopy, EIS, technique and its application in the characterization of electrode materials and electrochemical devices for energy conversion and storage. The concept of impedance and the necessary steps of impedance spectra analysis are explained. Basic relations for impedance of particular electro- chemical processes occurring in the system(s) are derived. Electrical equivalent circuits as impedance analogues f...
Frequency and Temporal Identification of a Li-ion Polymer Battery Model Using Fractional Impedance
Montaru M.; Pelissier S.
2008-01-01
The modelling of high power Lithium batteries may work as a powerful tool in the sizing of batteries pack in hybrid vehicles. Moreover, this tool can be used to evaluate accurately the battery characteristics at each level of its lifetime. Until now, the model structure differs according to its end-use due to simulation limitation. Indeed, impedance measurements obtained by electrochemical impedance spectroscopy needs non-integer order impedance in order to correctly model diffusive phen...
A comparative study of feed impedances of a circular and three different spiral antenna arrays
Ramachandra, V.; Dey, K. K.; Khastgir, P.
1982-08-01
Mathematical derivation of feed impedances of the dipole radiators of a circular and three different spiral arrays has been made for both constant and progressive phase distributions. A set of graphs, which show the variations of these impedances and their components from radiator to radiator, has been obtained. Spiral distributions of impedances can be observed on the Argand diagram in the cases of the three spiral arrays.
Elyas Rafiee; Mansour Farzam; Mohammad Ali Golozar; Ali Ashrafi
2013-01-01
Variation of electrochemical impedance with dislocation density was investigated using electrochemical impedance spectroscopy (EIS). For this purpose, EIS measurements were carried out on 10, 20, 30, 40, and 50% cold-rolled commercially pure copper in 0.1 M NaCl (pH = 2) solution. Nyquist plots illustrated that the electrochemical reactions are controlled by both charge transfer and diffusion process. Increasing dislocation density, the magnitude of electrochemical impedance of samples was de...