Lawrence, L E
ABT-773 is a macrolide antibacterial agent under development by Abbott Laboratories and Taisho Pharmaceutical Co Ltd for the potential treatment of bacterial infection . As of February 2001, ABT-773 had entered phase III trials in the US . Japanese phase II trials were expected to commence in June 2000 and a phase II trial is being designed for respiratory infections, with Abbott expecting filing in March 2002 . The bioavailability of ABT-773 in humans is unaffected by food  and in a phase I, randomized, double-blind trial in healthy males only mild adverse effects, usually affecting the gastrointestinal system, were observed . Under an agreement, Abbott and Taisho are conducting joint research to discover new compounds; Abbott will have worldwide marketing, manufacturing and supply rights (except in Japan), and Taisho will receive royalties on Abbott's sales in consideration of granted rights. In Japan, the companies will co-market any resulting compounds . ABT-773 demonstrated good activity in vitro and in vivo against Streptococcus pneumoniae and Staphylococcus aureus , , and was highly potent even against macrolide-resistant ,  and invasive  S pneumoniae.
Lorenz, Hanns M
Adalimumab (D2E7), a human monoclonal antibody that binds to and neutralizes TNFa, is being developed by Abbott (formerly Knoll), under license from Cambridge Antibody Technology (CAT), for the potential treatment of inflammatory disorders such as rheumatoid arthritis (RA) and Crohn's disease. It is also being investigated for the potential treatment of coronary heart disease. Phase II studies for Crohn's disease and phase III for RA were ongoing throughout 2001. Limited data are only available for RA. In January 2002, it was reported that phase III trials of adalimumab for RA had been completed, but details have not been published in the primary literature so far. At this time CAT and Abbott expected to file for US approval in the second quarter of 2002 with a launch date anticipated for 2003. Phase III data are expected to be presented at the European League Against Rheumatism meeting in June 2002. In November 2000, Lehman Brothers predicted a US launch in June 2002 with peak US sales of $600 million in 2007 and a launch in non-US markets in 2003 with peak sales in these markets of $300 million in 2008. In December 2000, Merrill Lynch predicted regulatory clearance in the second half of 2003. The probability of adalimumab reaching the market is estimated to be 70%. In December 2000, Merrill Lynch predicted a 2003 launch, with estimated sales of pounds sterling 3.65 million in that year rising to pounds sterling 30.14 million in 2010. In March 2001, ABN AMRO predicted sales of $73 million in 2003 rising to $392 million in 2007.
.... FDA-2012-F-0138] Abbott Laboratories; Filing of Food Additive Petition AGENCY: Food and Drug... that Abbott Laboratories has filed a petition proposing that the food additive regulations be amended... given that a food additive petition (FAP 2A4788) has been filed by Abbott Laboratories, 3300 Stelzer Rd...
...] Abbott Laboratories et al.; Withdrawal of Approval of 35 New Drug Applications and 64 Abbreviated New... Tablets... Abbott Laboratories, PA77/Bldg. AP30-1E, 200 Abbott Park Rd., Abbott Park, IL 60064-6157. NDA... (diphenhydramine Healthcare. HCl)) Injection Preservative Free. NDA 010021 Placidyl Abbott Laboratories...
This dissertation discusses the analysis of marketing communication within Abbott Laboratories Ltd. division of Abbott Diabetes Care, which carry business within the market of pharmaceutical industry. Prior to the analysis, the focus is on the theoretical part of the traditional components of the communication mix. In the practical part, the theoretical knowledge is then applied to the chosen example of the company. On the basis of performed analysis, recommendations are proposed in order to ...
... DEPARTMENT OF LABOR Employment and Training Administration [TA-W-75,201] Abbott Laboratories..., applicable to workers of Abbott Laboratories, Diagnostics Division, including on-site leased workers from... (clerical) were employed on-site at the Irving, Texas location of Abbott Laboratories, Diagnostics Division...
... DEPARTMENT OF LABOR Employment and Training Administration [TA-W-82,379] Abbott Laboratories... February 22, 2013, applicable to workers of Abbott Laboratories, Diagnostic--Hematology division, including... Clara, California location of Abbott Laboratories, Diagnostic--Hematology Division. The Department has...
Downing, Nicholas S; Ross, Joseph S; Jackevicius, Cynthia A; Krumholz, Harlan M
The ongoing debate concerning the efficacy of fenofibrate has overshadowed an important aspect of the drug's history: Abbott Laboratories, the maker of branded fenofibrate, has produced several bioequivalent reformulations that dominate the market, although generic fenofibrate has been available for almost a decade. This continued use of branded formulations, which cost twice as much as generic versions of fenofibrate, imposes an annual cost of approximately $700 million on the US health care system. Abbott Laboratories maintained its dominance of the fenofibrate market in part through a complex switching strategy involving the sequential launch of branded reformulations that had not been shown to be superior to the first-generation product and patent litigation that delayed the approval of generic formulations. The small differences in dose of the newer branded formulations prevented their substitution with generics of older-generation products. As soon as direct generic competition seemed likely at the new dose level, where substitution would be allowed, Abbott would launch another reformulation, and the cycle would repeat. Based on the fenofibrate example, our objective is to describe how current policy can allow pharmaceutical companies to maintain market share using reformulations of branded medications, without demonstrating the superiority of next-generation products.
Carlson, Dawn M; Steinberg, Joyce L; Gordon, Gary
While significant advances in the treatment of cancer occured during the last half of the twentieth century, parallel decreases in overall cancer death rates were not observed. Cancer therapy remains an area of significant unmet medical need. Abbott's oncology research programs are focused on pioneering trageted, less toxic therapies, aimed at different aspects of tumor growth and development. Oncology drugs in development at Abbott target several mechanisms of cancer progression by interfering with multiple processes necessary for tumor growth: recruitment of a blood supply, cell proliferation, and the development of metastases. They include a selective endothelin A-receptor antagonist (atrasentan/Xinlay), 3 angiogenesis inhibitors (ABT 510, a thrombospondin mimetic: ABT-869, a multitargeted receptor tyrosine kinase inhibitor; and ABT 828, recombinant human plasminogen kringle 5), a cell proliferation inhibitor (ABT-751, an antimitotic agent), an apoptosis inducer (ABT 737, a Bcl-2 family inhibitor), and a poly(ADP-ribose)polymerase inhibitor.
ABT-510 is a small peptide thrombospondin-1 mimetic angiogenesis inhibitor under development by Abbott Laboratories for the potential treatment of solid tumors. ABT-510 is undergoing phase II clinical trials.
U.S. Department of Health & Human Services — This list includes products subject to recall since September 2010 related to infant formula distributed by Abbott. This list will be updated with publicly available...
Detection of HCV core antigen as direct marker of hepatitis C infection clearly improves diagnosis of this disease (especially reduction of window period) and brings broad clinical utilization. The company Abbott Laboratories offers fully automated laboratory test for measurement of HCV core antigen on ARCHITECT analyzers.
Gronowski, Ann M.; Copper, Susan; Baorto, David; Murray, Patrick R.
This study demonstrates that significant reproducibility problems can occur during routine use of the Abbott Laboratories LCx assay for Chlamydia trachomatis and Neisseria gonorrhoeae. These problems can go undetected by the quality control procedures outlined in the manufacturer's package insert. We outline here procedures for detecting and preventing contamination and reproducibility problems.
..., Notification of Proposed Production Activity, Abbott Laboratories, Inc., AbbVie, Inc. (Pharmaceutical Production), North Chicago, IL, Area Abbott Laboratories, Inc. (Abbott) and AbbVie, Inc. (AbbVie) submitted a... Abbott facilities to AbbVie, now designated as Subzone 22S (S-66-2012). Abbott and Abbvie are now...
U.S. Department of Health & Human Services — On September 22, 2010, Abbott issued a voluntary recall of certain Similac powdered infant formula after identifying a common warehouse beetle (both larvae and...
Education Law Center, Inc., Newark, NJ.
This document contains the following "Abbott Opinions": (1) "Early Childhood Education"; (2) "Adequate School Facilities"; (3) "Supplemental Programs and Whole School Reform in Elementary Schools"; (4) "Supplemental Programs in Middle and High Schools"; and (5) "Planning Programs and Budgets…
Wada, Carol K
Matrix metalloproteinases (MMPs) have been implicated in several pathologies. At Abbott Laboratories, the matrix metalloproteinases inhibitor drug discovery program has focused on the discovery of a potent, selective, orally bioavailable MMP inhibitor for the treatment of cancer. The program evolved from early succinate-based inhibitors to utilizing in-house technology such as SAR by NMR to develop a novel class of biaryl hydroxamate MMP inhibitors. The metabolic instability of the biaryl hydroxamates led to the discovery of a new class of N-formylhydroxylamine (retrohydroxamate) biaryl ethers, exemplified by ABT-770 (16). Toxicity issues with this pre-clinical candidate led to the discovery of another novel class of retrohydroxamate MMP inhibitors, the phenoxyphenyl sulfones such as ABT-518 (19j). ABT-518 is a potent, orally bioavailable, selective inhibitor of MMP-2 and 9 over MMP-1 that has been evaluated in Phase I clinical trials in cancer patients.
... withdrawing approval of a new drug application (NDA) for MERIDIA (sibutramine hydrochloride (HCl)) oral... requested that Abbott voluntarily withdraw MERIDIA (sibutramine HCl) oral capsules from the market, based on FDA's recent analysis of clinical trial data from the Sibutramine Cardiovascular Outcomes Trial (SCOUT...
Miler, Marijana; Nikolac Gabaj, Nora; Dukic, Lora; Simundic, Ana-Maria
The aim of the study was to estimate improvement of work efficiency in the laboratory after implementation of total laboratory automation (TLA) by Abbott Accelerator a3600 in the laboratory with measuring different key performance indicators (KPIs) before and after TLA implementation. The objective was also to recommend steps for defining KPIs in other laboratories. For evaluation of improvement 10 organizational and/or technical KPIs were defined for all phases of laboratory work and measured before (November 2013) and after (from 2015 to 2017) TLA implementation. Out of 10 defined KPIs, 9 were successfully measured and significantly improved. Waiting time for registration of samples in the LIS was significantly reduced from 16 (9-28) to 9 (6-16) minutes after TLA (P performed at core biochemistry analyzers which significantly reduced walking distance for sample management (for more than 800 m per worker) and number of tube touches (for almost 50%). Analyzers downtime and engagement time for analyzers maintenance was reduced for 50 h and 28 h per month, respectively. TLA eliminated manual dilution of samples with extreme results with sigma values increment from 3.4 to >6 after TLA. Although median turnaround time TAT for potassium and troponin was higher (for approximately 20 min), number of outliers with TAT >60 min expressed as sigma values were satisfying (>3). Implementation of the TLA improved the most of the processes in our laboratory with 9 out of 10 properly defined and measured KPIs. With proper planning and defining of KPIs, every laboratory could measure changes in daily workflow.
Taieb, Joëlle; Mendez Lozano, Daniel H; Benattar, Clarisse; Messaoudi, Chérif; Poüs, Christian
We assessed a new estradiol (E2) immunoassay on the Architect-i2000 (Abbott Laboratories) for monitoring ovulation stimulation for IVF-ET and re-establishing clinical cut-off points. The method has been modified to improve E2 measurements especially at normal and low concentrations. E2 was determined for 552 samples, from 83 women, presenting normal follicular status and undergoing 100 cycles of IVF treatment. We assessed the value of this assay for down-regulation of E2 concentration limit using gonadoliberin-releasing hormone agonist (GnRHa), and monitoring of the ovarian hyperstimulation, expected range of E2 per mature follicle prior to the administration of exogenous hCG and day 3 concentration limit. We compared results with our routine method (E2-6II Advia-Centaur; Siemens-Diagnostics) for which decision-making values were known. Considering E2 concentrations obtained with the new Architect-i2000 assay for patients treated with GnRHa for 2 weeks, the cutoff-point for ovarian down-regulation should be set down at 110 pmol/L to maintain 100% of sensitivity. Considering day 3 concentration limit determination, results were not significantly different from those obtained with our routine method. The mean E2 values per mature follicle fell into the range generally expected. E2 determination with the new E2 Architect-i2000 assay could be used to monitor ovulation, in patients undergoing IVF-ET, in combination with transvaginal ultrasound.
Cooper, B H; Prowant, S; Alexander, B; Brunson, D H
The Abbott yeast identification system (Abbott Laboratories, Diagnostics Division, Irving, Tex.) is a 24-h, instrumental method for identifying medically important yeasts, based on matrix analysis of 19 biochemical reactions and the germ tube test. The system was evaluated in two clinical laboratories by using 179 coded isolates, which included a high percentage of the less frequently encountered species. Based upon results with these coded isolates and from previously obtained laboratory dat...
Arikan, S; Rex, J H
November 1998, Aronex signed a licensing collaboration with Abbott Laboratories for the worldwide rights to nystatin LF .
This paper explores Maude Abbott's internationally significant career in medicine and her parallel commitment to caring for her sister, Alice Abbott. An examination of Abbott's life reveals the difficulties faced by an ambitious Canadian woman in medicine from the 1890s to the 1920s; difficulties compounded by caring for a sister with a mental illness. The Abbott archive suggests that it was far more difficult for a woman doctor to make the kind of sharp distinction between public and private life that might be expected of professional men.
Education Law Center, 2005
Except for school construction, there is no legislation to guide implementation of the programs and reforms ordered by the New Jersey Supreme Court in the landmark "Abbott v. Burke" case. Instead, in its 1998 "Abbott V decision," the Supreme Court directed the Commissioner of Education to provide standards and procedures to…
... DEPARTMENT OF COMMERCE Foreign-Trade Zones Board [B-91-2012] Foreign-Trade Zone 22--Chicago, Illinois, Authorization of Production Activity, Abbott Laboratories, Inc., AbbVie, Inc. (Pharmaceutical Production), North Chicago, Illinois, Area On December 14, 2012, Abbott Laboratories, Inc., and AbbVie, Inc...
Ballinger, Ronda; Gee, Mary Kay
This guide, which integrates adult basic education (ABE) curriculum, job skills for Abbott Laboratories, and work-related foundation skills, is designed for an instructional program in the skill areas of reading, writing, oral communications, mathematics, and problem solving. In addition to creating a uniform process and product to promote…
Huh, Hee Jae; Kim, Ji-Youn; Lee, Myoung-Keun; Lee, Nam Yong; Kim, Jong-Won; Ki, Chang-Seok
Long-term nucleoside analogue (NA) treatment leads to selection for drug-resistant mutations in patients undergoing hepatitis B virus (HBV) therapy. The Abbott RealTime HBV Sequencing assay (Abbott assay; Abbott Molecular Inc., Des Plaines, IL, USA) targets the reverse transcriptase region of the polymerase gene and as such has the ability to detect NA resistance-associated mutations in HBV. We evaluated the analytical performance of the Abbott assay and compared its diagnostic performance to that of a laboratory-developed nested-PCR and sequencing method. The analytical sensitivity of the Abbott assay was determined using a serially-diluted WHO International Standard. To validate the clinical performances of the Abbott assay and the laboratory-developed assay, 89 clinical plasma samples with various levels of HBV DNA were tested using both assays. The limit of detection of the Abbott assay, was 210IU/ml and it successfully detected mutations when the mutant types were present at levels ≥20%. Among 89 clinical specimens, 43 and 42 were amplification positive in the Abbott and laboratory-developed assays, respectively, with 87.6% overall agreement (78/89; 95% confidence interval [CI], 78.6-93.4). The Abbott assay failed to detect the minor mutant populations in two specimens, and therefore overall concordance was 85.3% (76/89), and the kappa value was 0.79 (95% CI, 0.67-0.90). The Abbott assay showed comparable diagnostic performance to laboratory-developed nested PCR followed by direct sequencing, and may be useful as a routine method for detecting HBV NA resistance-associated mutations in clinical laboratory settings. Copyright Â© 2016 Elsevier B.V. All rights reserved.
Impact of the New Abbott mPLUS feature on clinical laboratory efficiencies of abbott RealTime assays for detection of HIV-1, Hepatitis C Virus, Hepatitis B Virus, Chlamydia trachomatis, and Neisseria gonorrhoeae.
Lucic, Danijela; Jones, Sara; Wiesneth, Russ; Barry, Cathy; Webb, Erika; Belova, Larissa; Dolan, Peggy; Ho, Shiaolan; Abravaya, Klara; Cloherty, Gavin
Diagnostic laboratories are under increasing pressure to improve and expand their services. Greater flexibility in sample processing is a critical factor that can improve the time to results while reducing reagent waste, making laboratories more efficient and cost-effective. The introduction of the Abbott mPLUS feature, with the capacity for extended use of amplification reagents, significantly increases the flexibility of the m2000 platform and enables laboratories to customize their workflows based on sample arrival patterns. The flexibility in sample batch size offered by mPLUS enables significant reductions in processing times. For hepatitis B virus tests, a reduction in sample turnaround times of up to 30% (105 min) was observed for batches of 12 samples compared with those for batches of 24 samples; for Chlamydia trachomatis/Neisseria gonorrhoeae tests, the ability to run batches of 24 samples reduced the turnaround time by 83% (54 min) compared with that for batches of 48 samples. Excellent correlations between mPLUS and m2000 standard condition results were observed for all RealTime viral load assays evaluated in this study, with correlation r values of 0.998 for all assays tested. For the qualitative RealTime C. trachomatis/N. gonorrhoeae assay, the overall agreements between the two conditions tested were >98% for C. trachomatis and 100% for N. gonorrhoeae. Comparable precision results were observed for the two conditions tested for all RealTime assays. The enhanced mPLUS capability provides clinical laboratories with increased efficiencies to meet increasingly stringent turnaround time requirements without increased costs associated with discarding partially used amplification reagents.
Most of New Jersey's charter schools are located in the state's poorer, urban school districts, or "Abbott" districts, and exclusively serve students from those communities. A number of other schools are located outside of the Abbott districts but enroll students from these districts. Specifically, of the 50 charter schools operating in…
Sickinger, Eva; Braun, Hans-Bertram; Praast, Gerald; Stieler, Myriam; Gundlach, Cordelia; Birkenbach, Claudia; Prostko, John; Palafox, Mary Ann; Frias, Edwin; Hsu, Stephen; Matias, Matthew; Pucci, Dominick; Hausmann, Michael; Sagel, Ulrich; Smith, Darwin
Development of the ARCHITECT Toxo IgM assay has been done to assist the clinician in acute Toxoplasma gondii infection detection, especially in pregnant women. Its use, in conjunction with ARCHITECT Toxo IgG and Toxo Avidity assays, will provide an array of assays particularly useful in the monitoring of pregnant females to determine the risk of maternal transmission of the parasite. Specificity results from 2 testing sites, using populations of pregnant females, hospital patients, and blood donors, demonstrated that the assay has an overall resolved relative specificity of 99.89% (confidence interval, 99.68-99.98%). Relative specificity for pregnant female specimens was 99.95% (n = 2031). Excellent seroconversion sensitivity was observed for the ARCHITECT Toxo IgM assay, which was similar to the Abbott AxSYM Toxo IgM assay (Abbott Laboratories, Abbott Park, IL). In more than 90% of the panels tested, the 1st bleed detected in the serial bleeds was the same for both assays.
Grace Abbott's courageous struggles--to protect the rights of immigrants, to increase the role of women in government, and to improve the lives of all children--are filled with adventurous tales of the remarkable human ability to seek out suffering and to do something about it. "A Prairie Childhood" is an excerpt from the Grace Abbott biography…
Full Text Available C’est la première fois que Berenice Abbott est exposée à Paris. Les cent vingt images et trente documents présentés au Jeu de Paume sont regroupés en quatre grandes séries, qui correspondent aux quatre grandes phases de sa carrière photographique. La première partie retrace son œuvre de portraitiste, qui commence à Paris au début des années 1920, où elle photographie des anonymes, mais aussi beaucoup d’artistes et d’écrivains tels que Marcel Duchamp, Jean Cocteau, James Joyce, ou encore Djuna...
Cole, Laurence A; Khanlian, Sarah A
False-positive hCG results can lead to erroneous diagnoses and needless chemotherapy and surgery. In the last 2 years, eight publications described cases involving false-positive hCG tests; all eight involved the AxSym test. We investigated the source of this abundance of cases and a simple fix that may be used by clinical laboratories. False-positive hCG was primarily identified by absence of hCG in urine and varying or negative hCG results in alternative tests. Seventeen false-positive serum samples in the AxSym test were evaluated undiluted and at twofold dilution with diluent containing excess goat serum or immunoglobulin. We identified 58 patients with false-positive hCG, 47 of 58 due to the Abbott AxSym total hCGbeta test (81%). Sixteen of 17 of these "false-positive" results (mean 100 mIU/ml) became undetectable when tested again after twofold dilution. A simple twofold dilution with this diluent containing excess goat serum or immunoglobulin completely protected 16 of 17 samples from patients having false-positive results. It is recommended that laboratories using this test use twofold dilution as a minimum to prevent false-positive results.
New Jersey's urban--or "Abbott"--schools have improved at the preschool and elementary school level, but lag when it comes to middle and high school performance. These are the key findings of an Abbott Indicators Project report entitled, "The Abbott Districts in 2005-06: Progress and Challenges." The report was prepared by…
... Authority, Foreign-Trade Zone 153; Abbott Cardiovascular Systems, Inc., (Cardiovascular Devices), Riverside... of Abbott Cardiovascular Systems, Inc., within Sites 11-13 of FTZ 153, located in Riverside County... behalf of Abbott Cardiovascular Systems, Inc., as described in the application and Federal Register...
Jungkind, D L; DiRenzo, S A; Young, S J
The Abbott enzyme immunoassay (Abbott Laboratories, North Chicago, Ill.) for human T-cell lymphotropic virus type III (HTLV-III) antibody was evaluated to determine the effect of using heat-inactivated (56 degrees C for 30 min) serum as the sample. Each of 58 nonreactive serum samples gave a higher A492 value when tested after heat inactivation. Ten of the samples became reactive after heating. Heat-inactivated serum should not be used in the current Abbott HTLV-III antibody test, because thi...
Improving clinical laboratory efficiency: a time-motion evaluation of the Abbott m2000 RealTime and Roche COBAS AmpliPrep/COBAS TaqMan PCR systems for the simultaneous quantitation of HIV-1 RNA and HCV RNA.
Amendola, Alessandra; Coen, Sabrina; Belladonna, Stefano; Pulvirenti, F Renato; Clemens, John M; Capobianchi, M Rosaria
Diagnostic laboratories need automation that facilitates efficient processing and workflow management to meet today's challenges for expanding services and reducing cost, yet maintaining the highest levels of quality. Processing efficiency of two commercially available automated systems for quantifying HIV-1 and HCV RNA, Abbott m2000 system and Roche COBAS Ampliprep/COBAS TaqMan 96 (docked) systems (CAP/CTM), was evaluated in a mid/high throughput workflow laboratory using a representative daily workload of 24 HCV and 72 HIV samples. Three test scenarios were evaluated: A) one run with four batches on the CAP/CTM system, B) two runs on the Abbott m2000 and C) one run using the Abbott m2000 maxCycle feature (maxCycle) for co-processing these assays. Cycle times for processing, throughput and hands-on time were evaluated. Overall processing cycle time was 10.3, 9.1 and 7.6 h for Scenarios A), B) and C), respectively. Total hands-on time for each scenario was, in order, 100.0 (A), 90.3 (B) and 61.4 min (C). The interface of an automated analyzer to the laboratory workflow, notably system set up for samples and reagents and clean up functions, are as important as the automation capability of the analyzer for the overall impact to processing efficiency and operator hands-on time.
Lucas, Tamara; Villegas, Ana Maria
In 1999-2000, over one-third of all students in the 30 Abbott districts spoke a native language other than English, and more than one-tenth were considered limited English proficient (LEP). The proportions of LEP students varied considerably across the districts, but they comprised between 5% and 29% of total enrollments in 18 of the districts.…
M. Schutten (Martin); D. Peters (D.); N. Back (Nicole); A.W. van den Beld (Annewieke); B. Beuselinck (B.); V. Foulongne (V.); A.M. Geretti (Anna Maria); L. Pandiani (L.); M. Tiemann; H.G.M. Niesters (Bert)
textabstractThe analytical performances of the new Abbott RealTime hepatitis C virus (HCV) and human immunodeficiency virus type 1 viral load assays were compared at nine laboratories with different competitor assays. These included the Abbott LcX, Bayer Versant bDNA, Roche COBAS Amplicor, and Roche
Schutten, Martin; Peters, D; Back, N K T; Beld, M; Beuselinck, K; Foulongne, V; Geretti, A-M; Pandiani, L; Tiemann, C; Niesters, H G M
The analytical performances of the new Abbott RealTime hepatitis C virus (HCV) and human immunodeficiency virus type 1 viral load assays were compared at nine laboratories with different competitor assays. These included the Abbott LcX, Bayer Versant bDNA, Roche COBAS Amplicor, and Roche COBAS
Schutten, M.; Peters, D.; Back, N. K. T.; Beld, M.; Beuselinck, K.; Foulongne, V.; Geretti, A.-M.; Pandiani, L.; Tiemann, C.; Niesters, H. G. M.
The analytical performances of the new Abbott RealTime hepatitis C virus (HCV) and human immunodeficiency virus type 1 viral load assays were compared at nine laboratories with different competitor assays. These included the Abbott LcX, Bayer Versant bDNA, Roche COBAS Amplicor, and Roche COBAS
Ko, Kiwoong; Yu, Shinae; Lee, Eun Hee; Park, Hyosoon; Woo, Hee-Yeon; Kwon, Min-Jung
Various assays for detecting high-risk human papillomavirus (HR HPV) have been introduced recently, including the Abbott RealTime High-Risk HPV assay. We sought to compare the performance of Abbott PCR to Hybrid Capture 2 for the detection of HR HPV. A total of 941 cervical swab specimens were obtained. We submitted all specimens for HR HPV detection with HC2 and Abbott PCR, and then additionally analyzed discordant and concordant positive results using restriction fragment mass polymorphism (RFMP) genotyping analysis. HC2 detected one of 13 HR HPV types in 12.3% (116/941) of cases, while Abbott PCR detected one of 14 detectable HR HPV types in 12.9% (121/941) of cases. The overall agreement rate was 97.3% with a kappa coefficient of 0.879. Discordant results between these two assays were observed in 25 cases. HC2 showed a sensitivity of 90.0% and specificity of 95.9%, while Abbott PCR showed a sensitivity of 98.0% and specificity of 96.8% when using RFMP results as the gold standard. For HPV 16/18 detection, Abbott PCR showed 95.8%/88.9% sensitivity and 99.2%/99.8% specificity, respectively. The overall coinfection rate between HPV 16, 18 and non-16/18 was 9.9% (12/121) in Abbott PCR analysis. Considering its high agreement rate with HC2, higher sensitivity/specificity compared to HC2, and ability to differentiate HPV 16/18 from other HPV types, Abbott PCR could be a reliable laboratory testing method for the screening of HPV infections. © 2016 by the Association of Clinical Scientists, Inc.
Coorevits, L; Vanscheeuwijck, C; Traen, A; Bingé, L; Ryckaert, I; Padalko, E
We evaluated Copan FLOQSwabs next to Abbott swabs for the detection of Chlamydia trachomatis (CT) by Abbott RealTime PCR. We collected 1062 paired swabs from female sex workers. The study was divided in two arms, according to the order of swab collection. If the Abbott swab was collected first, 501 couples were concordant and two discordant (Abbott negative and Copan positive). If the Copan swab was collected first, 537 couples were concordant and 10 discordant (eight Abbott negative and Copan positive and two Abbott positive and Copan negative). All discordant samples contained low levels of C. trachomatis. Technical issues lead to retesting of 64 Copan and 21 Abbott swabs. Our results show that Copan FLOQSwabs can be used interchangeably with Abbott swabs. While appearing to have an advantage in detecting more positive samples, the use of Copan swabs led to a higher retesting rate due to technical errors.
Barnett, W. Steven; Jung, Kwanghee; Youn, Min-Jong; Frede, Ellen C.
New Jersey's Abbott Preschool program is of broad national and international interest because the Abbott program provides a model for building a high-quality system of universal pre-K through public-private partnerships that transform the existing system. The program offers high-quality pre-K to all children in 31 New Jersey communities with high…
Applewhite, Erain; Hirsch, Lesley
The New Jersey Supreme Court's 1998 ruling in Abbott v. Burke represents the first judicial directive in the nation that public education must include a high-quality, well-planned preschool program starting at age three. This decision applies to 30 urban school districts, known as the Abbott districts, that serve approximately 25 percent of the…
Wallemacq, Pierre; Maine, Gregory T; Berg, Keith; Rosiere, Thomas; Marquet, Pierre; Aimo, Giuseppe; Mengozzi, Giulio; Young, Julianna; Wonigeit, Kurt; Kretschmer, Robert; Wermuth, Bendicht; Schmid, Rainer W
The objective of this study was to evaluate the analytical performance of the Abbott ARCHITECT Cyclosporine (CsA) immunoassay in 7 clinical laboratories in comparison to liquid chromatography/tandem mass spectrometry (LC/MS/MS), Abbott TDx, Cobas Integra 800, and the Dade Dimension Xpand immunoassay. The ARCHITECT assay uses a whole blood specimen, a pretreatment step with organic reagents to precipitate proteins and extract the drug, followed by a 2-step automated immunoassay with magnetic microparticles coated with anti-CsA antibody and an acridinium-CsA tracer. Imprecision testing at the 7 evaluation sites gave a range of total % coefficient of variations of 7.5%-12.2% at 87.5 ng/mL, 6.6%-14.3% at 411 ng/mL, and 5.2%-10.7% at 916 ng/mL. The lower limit of quantification ranged from 12 to 20 ng/mL. Purified CsA metabolites AM1, AM1c, AM4N, AM9, and AM19 were tested in whole blood by the ARCHITECT assay and showed minimal cross-reactivity at all 7 sites. In particular, AM1 and AM9 cross-reactivity in the ARCHITECT assay, ranged from -2.5% to 0.2% and -0.8% to 2.2%, respectively, and was significantly lower than for the TDx assay, in which the values were 3.2% and 16.1%, respectively. Comparable testing of metabolites in the Dade Dimension Xpand assay at 2 evaluation sites showed cross-reactivity to AM4N (6.4% and 6.8%) and AM9 (2.6% and 3.6%) and testing on the Roche Integra 800 showed cross-reactivity to AM1c (2.4%), AM9 (10.7%), and AM19 (2.8%). Cyclosporine International Proficiency Testing Scheme samples, consisting of both pooled specimens from patients receiving CsA therapy as well as whole-blood specimens supplemented with CsA, were tested by the ARCHITECT assay at 6 sites and showed an average bias of -24 to -58 ng/mL versus LC/MSMS CsA and -2 to -37 ng/mL versus AxSYM CsA. Studies were performed with the ARCHITECT CsA assay on patient specimens with the following results: ARCHITECT CsA assay versus LC/MSMS, average bias of 31 ng/mL; ARCHITECT versus the
Chen, J H K; She, K K K; Kwong, T-C; Wong, O-Y; Siu, G K H; Leung, C-C; Chang, K-C; Tam, C-M; Ho, P-L; Cheng, V C C; Yuen, K-Y; Yam, W-C
The automated high-throughput Abbott RealTime MTB real-time PCR assay has been recently launched for Mycobacterium tuberculosis complex (MTBC) clinical diagnosis. This study would like to evaluate its performance. We first compared its diagnostic performance with the Roche Cobas TaqMan MTB assay on 214 clinical respiratory specimens. Prospective analysis of a total 520 specimens was then performed to further evaluate the Abbott assay. The Abbott assay showed a lower limit of detection at 22.5 AFB/ml, which was more sensitive than the Cobas assay (167.5 AFB/ml). The two assays demonstrated a significant difference in diagnostic performance (McNemar's test; P = 0.0034), in which the Abbott assay presented significantly higher area under curve (AUC) than the Cobas assay (1.000 vs 0.880; P = 0.0002). The Abbott assay demonstrated extremely low PCR inhibition on clinical respiratory specimens. The automated Abbott assay required only very short manual handling time (0.5 h), which could help to improve the laboratory management. In the prospective analysis, the overall estimates for sensitivity and specificity of the Abbott assay were both 100 % among smear-positive specimens, whereas the smear-negative specimens were 96.7 and 96.1 %, respectively. No cross-reactivity with non-tuberculosis mycobacterial species was observed. The superiority in sensitivity of the Abbott assay for detecting MTBC in smear-negative specimens could further minimize the risk in MTBC false-negative detection. The new Abbott RealTime MTB assay has good diagnostic performance which can be a useful diagnostic tool for rapid MTBC detection in clinical laboratories.
Full Text Available Objectives: Analytical and clinical verification of both old and new generations of the Abbott total 25-hydroxyvitamin D (25OHD assays, and an examination of reference Intervals. Methods: Determination of between-run precision, and Deming comparison between patient sample results for 25OHD on the Abbott Architect, DiaSorin Liaison and AB SCIEX API 4000 (LC-MS/MS. Establishment of uncertainty of measurement for 25OHD Architect methods using old and new generations of the reagents, and estimation of reference interval in healthy Irish population. Results: For between-run precision the manufacturer claims 2.8% coefficients of variation (CVs of 2.8% and 4.6% for their high and low controls, respectively. Our instrument showed CVs between 4% and 6.2% for all levels of the controls on both generations of the Abbott reagents. The between-run uncertainties were 0.28 and 0.36, with expanded uncertainties 0.87 and 0.98 for the old and the new generations of reagent, respectively. The difference between all methods used for patientsâ samples was within total allowable error, and the instruments produced clinically equivalent results. The results covered the medical decision points of 30, 40, 50 and 125Â nmol/L. The reference interval for total 25OHD in our healthy Irish subjects was lower than recommended levels (24â111 nmol/L. Conclusion: In a clinical laboratory Abbott 25OHD immunoassays are a useful, rapid and accurate method for measuring total 25OHD. The new generation of the assay was confirmed to be reliable, accurate, and a good indicator for 25OHD measurement. More study is needed to establish reference intervals that correctly represent the healthy population in Ireland.
Lim, Myong Cheol; Lee, Do-Hoon; Hwang, Sang-Hyun; Hwang, Na Rae; Lee, Bomyee; Shin, Hye Young; Jun, Jae Kwan; Yoo, Chong Woo; Lee, Dong Ock; Seo, Sang-Soo; Park, Sang-Yoon; Joo, Jungnam
Human papillomavirus (HPV) testing based on cervical samples is important for use in cervical cancer screening. However, cervical sampling is invasive. Therefore, non-invasive methods for detecting HPV, such as urine samples, are needed. For HPV detection in urine samples, two real-time PCR (RQ-PCR) tests, Roche cobas 4800 test (Roche_HPV; Roche Molecular Diagnostics) and Abbott RealTime High Risk HPV test (Abbott_HPV; Abbott Laboratories) were compared to standard cervical samples. The performance of Roche_HPV and Abbott_HPV for HPV detection was evaluated at the National Cancer Center using 100 paired cervical and urine samples. The tests were also compared using urine samples stored at various temperatures and for a range of durations. The overall agreement between the Roche_HPV and Abbott_HPV tests using urine samples for any hrHPV type was substantial (86.0% with a kappa value of 0.7173), and that for HPV 16/18 was nearly perfect (99.0% with a kappa value of 0.9668). The relative sensitivities (based on cervical samples) for HPV 16/18 detection using Roche_HPV and Abbott_HPV with urine samples were 79.2% (95% CI; 57.9-92.9%) and 81.8% (95% CI; 59.7-94.8%), respectively. When the cut-off C T value for Abbott_HPV was extended to 40 for urine samples, the relative sensitivity of Abbott_HPV increased to 91.7% from 81.8% for HPV16/18 detection and to 87.0% from 68.5% for other hrHPV detection. The specificity was not affected by the change in the C T threshold. Roche_HPV and Abbott_HPV showed high concordance. However, HPV DNA detection using urine samples was inferior to HPV DNA detection using cervical samples. Interestingly, when the cut-off C T value was set to 40, Abbott_HPV using urine samples showed high sensitivity and specificity, comparable to those obtained using cervical samples. Fully automated DNA extraction and detection systems, such as Roche_HPV and Abbott_HPV, could reduce the variability in HPV detection and accelerate the standardization of HPV
Lim, Jinsook; Song, Kyung Eun; Song, Sang Hoon; Choi, Hyun-Jung; Koo, Sun Hoe; Kwon, Gye Choel
-The traceability of clinical results to internationally recognized and accepted reference materials and reference measurement procedures has become increasingly important. Therefore, the establishment of traceability has become a mandatory requirement for all in vitro diagnostics devices. -To evaluate the traceability of the Abbott Architect c8000 system (Abbott Laboratories, Abbott Park, Illinois), consisting of calibrators and reagents, across 4 different chemistry analyzers, and to evaluate its general performance on the Toshiba 2000FR NEO (Toshiba Medical Systems Corporation, Otawara-shi, Tochigi-ken, Japan). -For assessment of traceability, secondary reference materials were evaluated 5 times, and then bias was calculated. Precision, linearity, and carryover were determined according to the guidelines of the Clinical and Laboratory Standards Institute (Wayne, Pennsylvania). -The biases from 4 different analyzers ranged from -2.33% to 2.70% on the Toshiba 2000FR NEO, -2.33% to 5.12% on the Roche Hitachi 7600 (Roche Diagnostics International, Basel, Switzerland), -0.93% to 2.87% on the Roche Modular, and -2.16% to 2.86% on the Abbott Architect c16000. The total coefficients of variance of all analytes were less than 5%. The coefficients of determination (R(2)) were more than 0.9900. The carryover rate ranged from -0.54% to 0.17%. -Abbott clinical chemistry assays met the performance criteria based on desirable biological variation for precision, bias, and total error. They also showed excellent linearity and carryover. Therefore, these clinical chemistry assays were found to be accurate and reliable and are readily applicable on the various platforms used in this study.
Wright, James R; Alberti, Samuel J M M; Lyons, Christopher; Fraser, Richard S
- In the early 1900s, it was common practice to retain, prepare, and display instructive pathologic specimens to teach pathology to medical trainees and practitioners; these collections were called medical museums. Maude Abbott established her reputation by developing expertise in all aspects of medical museum work. She was a founder of the International Association of Medical Museums (later renamed the International Academy of Pathology) and became an internationally renowned expert on congenital heart disease. Her involvement in the Canadian Medical War Museum (CMWM) is less well known. - To explore Abbott's role in the development of the CMWM during and after World War I and to trace its history. - Available primary and secondary historical sources were reviewed. - Instructive pathologic specimens derived from Canadian soldiers dying during World War I were shipped to the Royal College of Surgeons in London, which served as a clearinghouse for museum specimens from Dominion forces. The Canadian specimens were repatriated to Canada, prepared by Abbott, and displayed at several medical meetings. Abbott, because she was a woman, could not enlist and so she reported to a series of enlisted physicians with no expertise in museology. Plans for a permanent CMWM building in Ottawa eventually failed and Abbott maintained the collection at McGill (Montreal, Quebec, Canada) until her death in 1940. We trace the CMWM after her death. - Sadly, after Abbott had meticulously prepared these precious teaching specimens so that their previous owners' ultimate sacrifice would continue to help their military brethren, the relics were bureaucratically lost.
Qiu, Ning; Li, Rui; Yu, Jian-Guo; Yang, Wen; Zhang, Wei; An, Yong; Li, Tong; Liu, Xue-En; Zhuang, Hui
To compare the performance of the Da-an real-time hepatitis B virus (HBV) DNA assay and Abbott RealTime HBV assay. HBV DNA standards as well as a total of 180 clinical serum samples from patients with chronic hepatitis B were measured using the Abbott and Da-an real-time polymerase chain reaction (PCR) assays. Correlation and Bland-Altman plot analysis was used to compare the performance of the Abbott and Da-an assays. The HBV DNA levels were logarithmically transformed for analysis. All statistical analyses were performed using SPSS for Windows version 18.0. The correlation between the two assays was analyzed by Pearson's correlation and linear regression. The Bland-Altman plots were used for the analysis of agreement between the two assays. A P value of Da-an assay were significantly correlated with the expected values of HBV DNA standards (r = 0.999, for Abbott; r = 0.987, for Da-an, P Da-an assay. Moreover, HBV DNA levels measured by the Abbott assay were significantly higher than those of the Da-an assay (6.23 ± 1.76 log IU/mL vs 5.46 ± 1.55 log IU/mL, P Da-an assay presented lower sensitivity and a narrower linear range as compared to the Abbott assay, suggesting the need to be improved.
Yu, Hoi-Ying Elsie; Agus, Michael; Kellogg, Mark D
Diagnosis and management of diabetic ketoacidosis (DKA) often rely on the measurement of urine ketones along with blood glucose, anion gap, and pH. These values, however, do not reliably reflect the severity of ketoacidosis. The Abbott Precision Xceed Pro® meter is an FDA-approved device that quantitatively measures β-hydroxybutyrate (BOH) in whole blood. This study was undertaken to determine whether the ketone meter meets the analytical criteria to aid DKA diagnosis and management in the hospital. 54 heparinized venous whole blood BOH concentrations from 27 diabetic patients were measured by the Abbott meter, and compared with the plasma BOH concentrations measured with Stanbio reagent (reference method). Measurements were done in the hospital central laboratory. Of the 54 pairs of specimens analyzed, 17 pairs displayed a difference of >15% between the two methods. Nearly all discrepant points occurred when BOH >5 mmol/L (reference method). Linearity evaluation revealed that the meter is not linear from 0.0 to 8.0 mmol/L, contrary to the claim by the manufacturer. Further, we identified acetoacetate, a metabolite commonly present in DKA patients, as a potential interfering substance for the meter BOH measurement. BOH measurements by the Abbott meter up to 3 mmol/L correlate well with the reference method, but become discrepant above that point. While this characteristic may be useful in the diagnosis of DKA, it may not allow clinicians to serially follow the response to therapy in hospitalized DKA patients with BOH values greater than 5 mmol/L (reference method). © 2011 John Wiley & Sons A/S.
Halfon, Philippe; Benmoura, Dominique; Agostini, Aubert; Khiri, Hacene; Penaranda, Guillaume; Martineau, Agnes; Blanc, Bernard
Abbott RealTime (RT) High-Risk (HR) HPV assay is a new qualitative real-time polymerase chain reaction (PCR) based assay for the detection of 14 HR HPV DNA. The assay can differentiate between the infection by HPV 16, HPV 18 and non-HPV 16/18 types through the distinct fluorescent labels on the type specific probes. To evaluate the clinical performance of the Abbott RT HR HPV test, in comparison with biopsy, Hybrid Capture II (HCII), and Linear Array (LA), for detection of high-grade disease (CIN2+). The study population consisted of 143 women who were included in three referral gynecology clinics in Marseilles (France) between March 2007 and June 2008. The clinical performance of the RT HR HPV assay, performed on the fully automated m2000 system, was compared with HCII and LA. HR HPV positivity rate was similar for all tests (Abbott RT HR HPV and HCII, 62%, and LA 63%). All tests had high sensitivities and negative predictive values for CIN2+ detection (>90%). The agreement between HCII and Abbott RT HR HPV, and between HCII and LA were 93% (k=0.85) and 96% (k=0.91) respectively. As expected, HPV16 or HPV18 positivity was greater in advanced grades of disease, especially in CIN2+ patients: 85% in CIN2+ vs. 33% in Abbott RT HR HPV assay is good and closely correlated with the two other assays. The automation and ability to identify type 16 and 18 make this a very attractive option for HPV testing in laboratories and potentially provides improved patient management. Copyright 2010. Published by Elsevier B.V.
He, Chao; Germer, Jeffrey J; Ptacek, Elizabeth R; Bommersbach, Carl E; Mitchell, P Shawn; Yao, Joseph D C
Hepatitis virus C (HCV) genotype (GT) determination and subtype (ST) differentiation (1a versus 1b) remain important for the selection of appropriate direct-acting antiviral (DAA) therapy. This study is a retrospective comparison of HCV GT and ST result distribution when using the Abbott RealTime HCV Genotype II assay (HCVGT II) alone and in combination with the Abbott RealTime HCV Genotype Plus RUO assay (HCVGT Plus) for routine testing of clinical serum specimens at a reference laboratory. HCVGT II results of specimens tested from June 2014 through January 2016 (period 1) were compared with combined results from HCVGT II and HCVGT Plus (HCVGT II/Plus) performed from January 2016 through January 2017 (period 2). A total of 44,127 and 25,361 specimens were tested during periods 1 and 2, respectively. Use of HCVGT II/Plus significantly reduced the frequency of GT 1 results without ST (0.4%) when compared to preliminary HCVGT II results during period 2 (5.3%; p < 0.01) and final HCVGT II results in period 1 (5.5%; p < 0.01). HCVGT II/Plus also resulted in GT 6 reactivity in 38 specimens with results of "HCV detected" (n = 17) or GT 1 (n = 21) following initial HCVGT II testing during period 2. When compared to the use of HCVGT II alone, HCVGT II/Plus significantly reduced the frequency of GT 1 without ST results observed in a large reference laboratory, while also enabling the identification of HCV GT 6. Copyright © 2018 Elsevier B.V. All rights reserved.
Flodin, Mats; Larsson, Anders
Glomerular filtration rate (GFR) is widely accepted as the best overall measure of kidney function. Cystatin C is a novel endogenous GFR marker that has been shown to be superior to creatinine for estimation of GFR in several studies. There is a need for cystatin C assays adapted to routine chemistry instrument to minimize turnaround times and allowing 24 h/day availability. We have evaluated a new cystatin C assay developed for Architect cSystem (Abbott Laboratories, Abbott Park, IL, USA). The cystatin C assay showed good agreement with the corresponding assay from Dade Behring (Deerfield, IL, USA). The assay has a very low total imprecision and a good linearity. The new cystatin C assay is an interesting alternative to current cystatin C assays. On an Architect cSystem the assay can be performed with the same turnaround times and availability as creatinine.
Ssebugenyi, I; Kizza, A; Mpoza, B; Aluma, G; Boaz, I; Newell, K; Laeyendecker, O; Shott, J P; Serwadda, D; Reynolds, S J
The need for viral load (VL) monitoring of HIV patients receiving antiretroviral therapy (ART) in resource-limited settings (RLS) has become apparent with studies showing the limitations of immunological monitoring. We compared the Abbott m2000 Real-Time (Abbott) HIV-1 assay with the Roche AMPLICOR Monitor v1.5 (Roche) HIV-1 assay over a range of VL concentrations. Three hundred and eleven plasma samples were tested, including 164 samples from patients on ART ≥ six months and 147 from ART-naïve patients. The Roche assay detected ≥400 copies/mL in 158 (50.8%) samples. Of these, Abbott produced 145 (91.8%) detectable results ≥400 copies/mL; 13 (8.2%) samples produced discrepant results. Concordance between the assays for detecting HIV-1 RNA ≥400 copies/mL was 95.8% (298/311). The sensitivity, specificity, positive predictive value and negative predictive value of Abbott to detect HIV-1 RNA ≥400 copies/mL were 91.8%, 100%, 100% and 92.2%, respectively. For the 151 samples with HIV-1 RNA ≥400 copies/mL for both assays, a good linear correlation was found (r = 0.81, P Abbott assay performed well in our setting, offering an alternative methodology for HIV-1 VL for laboratories with realtime polymerase chain reaction (PCR) capacity.
Dimeski, G; Jones, B W; Tilley, V; Greenslade, M N; Russell, A W
Both Roche and Abbott have released new glucose meter strips. They supply the entire Australian hospital market. The present study compared the performance of the new strips utilizing various specimen types (capillary, venous lithium heparin whole blood, venous lithium heparin plasma and serum) and evaluated how well they comply with the International Standards Organization (ISO) 15197 criteria. The study included imprecision, patient comparison and interference studies. Participants with and without diabetes were recruited to evaluate the performance of various specimen types against the Beckman DxC800 glucose method. The strips were tested for different interferences: galactose, maltose, lactose, Icodextrin, Intragam, paracetamol, sodium, ascorbic acid, variable strip storage temperature, haematocrit, haemolysis and lipaemia. The imprecision of the two strips was approximately 5% or less, except for the Abbott strip at very low values (1.4 mmol/L), approximately 7%. In total, 78% and 84%, respectively, of the results from the finger prick capillary specimens with the Roche (Accu-Chek Performa meter) and Abbott (Optium Xceed meter) strips, not 95% or greater as recommended by the ISO guideline, were within the recommended limits compared with reference plasma estimation on laboratory analysers. Galactose, ascorbic acid, haematocrit and sodium on the Roche and ascorbic acid and haematocrit on the Abbott strip continue to interfere to a variable degree with the glucose measurement. Analytically small differences exist between the glucose meter strips. The most significant analytical difference with the strips was at low glucose levels when compared with laboratory analyses and this may be of clinical importance. The impact of some of the interferences is variable between the two strips. Individuals, health-care professionals and health-care institutions should consider these data when selecting glucose meters for the management of people with diabetes mellitus.
Whitebook, Marcy; Ryan, Sharon; Kipnis, Fran; Sakai, Laura
In a series of New Jersey Supreme Court decisions known as Abbott v. Burke, the 28 (now 31) urban school districts serving the state's poorest students were ordered to create systems of high-quality preschool for all three- and four-year-old children, beginning in the 1999-2000 school year. The Abbott Preschool Program now serves approximately…
Association for Children of New Jersey, 2004
In an attempt to better prepare young children for the challenges of kindergarten and first grade, the Supreme Court of New Jersey, in its 1998 landmark decision of "Abbott v. Burke" (Abbott V), required the State's poorest school districts to implement high quality, intensive preschool for all 3-and 4-year old children. To take…
Sariguzel, Fatma Mutlu; Berk, Elife; Gokahmetoglu, Selma; Ercal, Baris Derya; Celik, Ilhami
The determination of HCV genotypes and subtypes is very important for the selection of antiviral therapy and epidemiological studies. The aim of this study was to evaluate the performance of Abbott Real Time HCV Genotype II assay in HCV genotyping of HCV infected patients in Kayseri, Turkey. One hundred patients with chronic hepatitis C admitted to our hospital were evaluated between June 2012 and December 2012, HCV RNA levels were determined by the COBAS® AmpliPrep/COBAS® TaqMan® 48 HCV test. HCV genotyping was investigated by the Abbott Real Time HCV Genotype II assay. With the exception of genotype 1, subtypes of HCV genotypes could not be determined by Abbott assay. Sequencing analysis was used as the reference method. Genotypes 1, 2, 3 and 4 were observed in 70, 4, 2 and 24 of the 100 patients, respectively, by two methods. The concordance between the two systems to determine HCV major genotypes was 100%. Of 70 patients with genotype 1, 66 showed infection with subtype 1b and 4 with subtype 1a by Abbott Real Time HCV Genotype II assay. Using sequence analysis, 61 showed infection with subtype 1b and 9 with subtype 1a. In determining of HCV genotype 1 subtypes, the difference between the two methods was not statistically significant (P>0.05). HCV genotype 4 and 3 samples were found to be subtype 4d and 3a, respectively, by sequence analysis. There were four patients with genotype 2. Sequence analysis revealed that two of these patients had type 2a and the other two had type 2b. The Abbott Real Time HCV Genotype II assay yielded results consistent with sequence analysis. However, further optimization of the Abbott Real Time HCV Genotype II assay for subtype identification of HCV is required.
Wallemacq, Pierre; Goffinet, Jean-Sebastien; O'Morchoe, Susan; Rosiere, Thomas; Maine, Gregory T; Labalette, Myriam; Aimo, Giuseppe; Dickson, Diana; Schmidt, Ed; Schwinzer, Reinhard; Schmid, Rainer W
The objective of this study was to evaluate the analytical performance of the Abbott ARCHITECT Tacrolimus immunoassay. Proficiency panels and specimens from a population of organ transplant recipients were analyzed in 6 clinical laboratories in Europe and the United States, and the results were compared with other methods. The ARCHITECT assay requires a whole blood specimen pretreatment step with methanol/zinc sulfate to precipitate protein and extract the drug, followed by a 30-minute immunoassay using anti-tacrolimus antibody-coated paramagnetic microparticles and an acridinium-tacrolimus tracer. The assay was free from hematocrit interference in the range 25%-55% and from interference by extremes of cholesterol, triglycerides, bilirubin, total protein, and uric acid. The total percent of coefficient of variations of the assay were 4.9%-7.6% at 3 ng/mL, 2.9%-4.6% at 8.6 ng/mL, and 3.1%-8.2% at 15.5 ng/mL. Limit of detection was Abbott IMx Tacrolimus II immunoassay (4 sites) yielded average biases between -0.94 and +0.26 ng/mL; ARCHITECT assay versus the Dade Dimension Tacrolimus immunoassay (2 sites) yielded average biases of -0.46 and +0.11 ng/mL; and ARCHITECT assay versus LC-MSMS methods at 2 sites yielded average biases of +0.51 and +1.63 ng/mL. Spearman correlation coefficients were >/=0.90 on all method comparisons. The ARCHITECT Tacrolimus assay is a semiautomated, robust, and highly sensitive immunoassay, representing an alternative approach for laboratories not equipped with LC-MSMS, and meets the 1 ng/mL recommendation of LOQ by the European Consensus Conference on Tacrolimus Optimization.
Gaze, David C; Prante, Christian; Dreier, Jens; Knabbe, Cornelius; Collet, Corinne; Launay, Jean-Marie; Franekova, Janka; Jabor, Antonin; Lennartz, Lieselotte; Shih, Jessie; del Rey, Jose Manuel; Zaninotto, Martina; Plebani, Mario; Collinson, Paul O
Galectin-3 is secreted from macrophages and binds and activates fibroblasts forming collagen. Tissue fibrosis is central to the progression of chronic heart failure (CHF). We performed a European multicentered evaluation of the analytical performance of the two-step routine and Short Turn-Around-Time (STAT) galectin-3 immunoassay on the ARCHITECT i1000SR, i2000SR, and i4000SR (Abbott Laboratories). We evaluated the assay precision and dilution linearity for both routine and STAT assays and compared serum and plasma, and fresh vs. frozen samples. The reference interval and biological variability were also assessed. Measurable samples were compared between ARCHITECT instruments and between the routine and STAT assays and also to a galectin-3 ELISA (BG Medicine). The total assay coefficient of variation (CV%) was 2.3%-6.2% and 1.7%-7.4% for the routine and STAT assays, respectively. Both assays demonstrated linearity up to 120 ng/mL. Galectin-3 concentrations were higher in plasma samples than in serum samples and correlated well between fresh and frozen samples (R=0.997), between the routine and STAT assays, between the ARCHITECT i1000 and i2000 instruments and with the galectin-3 ELISA. The reference interval on 627 apparently healthy individuals (53% male) yielded upper 95th and 97.5th percentiles of 25.2 and 28.4 ng/mL, respectively. Values were significantly lower in subjects younger than 50 years. The galectin-3 routine and STAT assays on the Abbott ARCHITECT instruments demonstrated good analytical performance. Further clinical studies are required to demonstrate the diagnostic and prognostic potential of this novel marker in patients with CHF.
This survey was conducted at John Abbott College to gauge teachers' responses to issues concerning their job satisfaction, interaction with colleagues, perceptions of student abilities, and perceptions concerning union negotiating priorities and areas of conflict within the institutional environment. Of the 75 teachers contacted, 47 returned…
Walker, Elaine M.
This study examined issues faced during implementation of school-based management (SBM) in New Jersey's special needs or Abbott districts, using a literature review, surveys of K-12 schools, and focus groups with central office administrators. The study examined forms of SBM, team operations, local autonomy versus state power, skills required to…
... finished product) include: resins, plastic tubing, stent components, plastic packaging, plastic clips... noted above. FTZ designation would further allow Abbott to realize logistical benefits through the use of weekly customs entry procedures. Customs duties also could possibly be deferred or reduced on...
The landmark New Jersey Supreme Court school funding case, "Abbott v. Burke", established the availability of preschool for all three- and four-year-olds living within the state's thirty-one poorest districts as a means of eradicating the effects of poverty. Longitudinal studies have shown the value of high quality preschool programs for…
... DEPARTMENT OF COMMERCE Foreign-Trade Zones Board [Order No. 1654] Approval for Expansion of Subzone 22F, Abbott Molecular, Inc. (Pharmaceutical and Molecular Diagnostic Products), Chicago, IL, Area Pursuant to its authority under the Foreign-Trade Zones Act of June 18, 1934, as amended (19 U.S.C. 81a-81u...
Wang, Linchuan; Chen, Wei; Yu, Yan
To date, there is a trend that the chemiluminescent microparticle immunoassays (CMIA) and electrochemiluminescence immunoassays (ECIA) technology gradually replacing the enzyme-linked immunosorbent assay (ELISA). But the performance such as the limit of quantitation (LOQ), precision, linear range of CMIA, or ECIA for serum markers of infectious diseases has rarely been reported. Using proficiency testing samples and standard materials, we confirmed the LOQ of the ELISA and the precision, linear range, LOQ, and instrument biases of the Abbott i2000 for eight serum markers. We used the Abbott i2000 and ELISAs to assess five HIV samples; the researchers were blinded to the true status of the samples. For the Abbott i2000, the coefficients of variation (CV) for the low, medium, and high concentration samples ranged from 1.06 to 12.74%, which were less than the allowable error; the linear ranges of HBsAg and HBsAb were 0.66-304.11 IU/ml and 8.16-1205.9 mIU/ml, respectively. For the Abbott i2000, the LOQs of HBsAg, HBsAb, HBeAg, HBeAb, HBcAb, anti-HCV, anti-TP, and anti-HIV were 0.026 IU/ml, 4 mIU/ml, 0.14 NCU/ml, 0.56 NCU/ml, 0.99 NCU/ml, 0.5 NCU/ml, 8.8 mIU/ml, and 1.92 NCU/ml, respectively, and these values were 0.16 IU/ml, 6.97 mIU/ml, 1.16 NCU/ml, 1.63 NCU/ml, 1.79 NCU/ml, 1.03 NCU/ml, 8.33 mIU/ml, and 1.3 NCU/ml, respectively, for the ELISA. When five HIV samples were blindly assessed, two cases were missed by the Abbott i2000 and the ELISA results were consistent with the expected results. The Abbott i2000 performed significantly better than the ELISA on HBV and HCV screening; however, for anti-TP and anti-HIV, the ELISA remained the preferred method. © 2016 Wiley Periodicals, Inc.
Blank, Fidela S J; Miller, Moses; Nichols, James; Smithline, Howard; Crabb, Gillian; Pekow, Penelope
The purpose of this study is to compare blood glucose levels measured by a point of care (POC) device to laboratory measurement using the same sample venous blood from patients with suspected diabetic ketoacidosis (DKA). A descriptive correlational design was used for this IRB-approved quality assurance project. The study site was the 50-bed BMC emergency department (ED) which has an annual census of over 100,000 patient visits. The convenience sample consisted of 54 blood samples from suspected DKA patients with orders for hourly blood draws for glucose measurement. Spearman correlations of the glucose POC values, reference lab values, and differences between the two, were evaluated. A chi-square test was used to evaluate the association between the acidosis status and FDA acceptability of POC values. Patient age range was 10-86 years; 63% were females; 46% had a final diagnosis of DKA. POC values underestimated glucose levels 93% of the time. There was a high correlation between the lab value and the magnitude of the difference, (lab minus POC value) indicating that the higher the true glucose value, the greater the difference between the lab and the POC value. A chi-square test showed no overall association between acidosis and FDA-acceptability. The POC values underestimated lab reported glucose levels in 50 of 54 cases even with the use of same venous sample sent to the lab, which make it highly unreliable for use in monitoring suspected DKA patients.
Full Text Available Joshua Kostera, Gregor Leckie, Klara Abravaya, Hong Wang Abbott Molecular, Abbott Laboratories, Des Plaines, IL, USA Introduction: The Abbott RealTime MTB RIF/INH Resistance Assay (RT MTB RIF/INH is an assay for the detection of rifampicin (RIF- and/or isoniazid (INH-resistant Mycobacterium tuberculosis (MTB. The assay can be used to test sputum, bronchial alveolar lavage, and N-Acetyl-L-Cysteine (NALC/NaOH pellets prepared from these samples. The assay can be used in direct testing mode, or in reflex mode following a MTB positive result produced by its companion assay, Abbott RT MTB. Methods: In this study, the direct testing mode was used to test paired sputum and NALC/NaOH pellets prepared from sputum collected from Bangladesh TB patients. One hundred and thirty two paired samples were tested. Results: The RT MTB RIF/INH inhibition rate was 0%. One hundred and twenty-two paired samples had results above the assay limit of detection and were analyzed by comparing with results from phenotypic drug sensitivity testing, GeneXpert MTB/RIF (Xpert, and MTBDR plus (Hain. RT MTB RIF/INH results were in good agreement with those of GeneXpert and Hain. Conclusion: The ability of this assay to detect RIF and INH resistance may contribute to the global control of multidrug resistant tuberculosis. Keywords: tuberculosis, rifampicin, isoniazid, resistance
Ko, Dae-Hyun; Hyun, Jungwon; Kim, Hyun Soo; Park, Min-Jeong; Kim, Jae-Seok; Park, Ji-Young; Shin, Dong Hoon; Cho, Hyoun Chan
Protein induced by vitamin K absence (PIVKA) is measured using various assays and is used to help diagnose hepatocellular carcinoma. The present study evaluated the analytical and clinical performances of the recently released Abbott Architect PIVKA assay. Precision, linearity, and correlation tests were performed in accordance with the Clinical Laboratory Standardization Institute guidelines. Sample type suitability was assessed using serum and plasma samples from the same patients, and the reference interval was established using sera from 204 healthy individuals. The assay had coefficients of variation of 3.2-3.5% and intra-laboratory variation of 3.6-5.5%. Linearity was confirmed across the entire measurable range. The Architect PIVKA assay was comparable to the Lumipulse PIVKA assay, and the plasma and serum samples provided similar results. The lower reference limit was 13.0 mAU/mL and the upper reference limit was 37.4 mAU/mL. The ability of the Architect PIVKA assay to detect hepatocellular carcinoma was comparable to that of the alpha-fetoprotein test and the Lumipulse PIVKA assay. The Architect PIVKA assay provides excellent analytical and clinical performance, is simple for clinical laboratories to adopt, and has improved sample type suitability that could broaden the assay's utility. © 2018 by the Association of Clinical Scientists, Inc.
The aim for this testing is to find a small panel of biomarkers (n=2-5) that can be tested on the Abbott ARCHITECT automated immunoassay platform for the early detection of hepatocellular carcinoma (HCC). This panel of biomarkers should perform significantly better than alpha-fetoprotein (AFP) alone based on multivariate statistical analysis. This testing of the EDRN reference set will help expedite the selection of a small panel of ARCHITECT biomarkers for the early detection of HCC. The panel of ARCHITECT biomarkers Abbott plans to test include: AFP, protein induced by vitamin K absence or antagonist-II (PIVKA-II), golgi protein 73 (GP73), hepatocellular growth factor (HGF), dipeptidyl peptidase 4 (DPP4) and DPP4/seprase (surface expressed protease) heterodimer hybrid. PIVKA-II is abnormal des-carboxylated prothrombin (DCP) present in vitamin K deficiency.
Fu, Yung-Chieh; Liao, I-Chuang; Chen, Hung-Mo; Yan, Jing-Jou
The Abbott RealTime MTB assay, launched in June 2014, has been shown to have a competitive performance in the detection of the Mycobacterium tuberculosis (MTB) complex in respiratory specimens. The present study was conducted to investigate the usefulness of the Abbott MTB Realtime assay in the detection of MTB in formalin-fixed paraffin-embedded (FFPE) tissues. A total of 96 FFPE specimens obtained from microbiologically proven MTB cases (N=60) and nontuberculous Mycobacterium cases (N=36) were analyzed. The performance of the Abbott MTB Realtime assay was compared with that of the Roche Cobas TaqMan MTB assay. The overall sensitivity and specificity of the Abbott assay were 63.3% and 97.2%, respectively, compared with 11.7% and 100% for the Cobas assay. The detection rate of the Abbott assay was much higher among 37 acid-fast-positive specimens than among 23 acid-fast-negative specimens (89.3% versus 21.7%, respectively). The detection rate of the assay was higher among 29 resection specimens than among 31 small biopsy specimens (86.2% versus 41.9%, respectively). Our results suggest that the Abbott RealTime MTB assay can be used to differentiate MTB from nontuberculous mycobacterial infections in acid-fast-positive FFPE tissues. © 2016 by the Association of Clinical Scientists, Inc.
Our earlier glycoproteomic studies have identified bisecting glycoslyation and core fucosylation changes on particular glycoproteins in endometrioid ovarian cancer tissues and plasma (Abbott et al, 2010, Proteomics). We have validated that these glycan changes occur on the same glycoproteins in serous ovarian cancer plasma using a lectin-pull down western blot assays. We would like to used pooled reference samples to develop a sensitive magnetic bead-based assay to detect these glycoproteins with bisecting and core fucosylation changes.
Full Text Available Introduction: In the HIV-1 infection, the increase in number of CD4 T lymphocytes and the viral load decline are the main indicators of the effectiveness of antiretroviral therapy. On average, 85% of patients receiving effective treatment has a persistent suppression of plasma viral load below the detection limit (<50 copies/mL of clinically used viral load assays, regardless of treatment regimen in use. It is known, however, that, even when viremia is reduced below the sensitivity limit of current diagnostic assays, the virus persists in “reservoirs” and traces of free virions can be detected in plasma.There is a considerable interest to investigate the clinical significance of residual viremia. Advances in molecular diagnostics allows nowadays to couple a wide dynamic range to a high sensitivity.The Abbott Real-time HIV-1 test is linear from 40 to 107 copies/mL and provides, below 40 copies/mL, additional information such as “<40cp/mL, target detected” or “target not detected”. The HIV-1 detection is verified by the max-Ratio algorithm software.We assessed the test sensitivity when the qualitative response is considered as well. Methods: A ‘probit’ analysis was performed using dilutions of the HIV-1 RNA Working Reagent 1 for NAT assays (NIBSC code: 99/634, defined in IU/mL and different from that used by the manufacturer (VQA,Virology Quality Assurance Laboratory of the AIDS Clinical Trial Group for standardization and definition of performances.The sample input volume (0.6 mL was the same used in clinical routine. A total of 196 replicates at concentrations decreasing from 120 to 5 copies/mL, in three different sessions, have been tested.The ‘probit’ analysis (binomial dose-response model, 95% “hit-rate” has been carried out on the SAS 9.1.3 software package. Results: The sensitivity of the “<40cp/mL, target detected” response was equal to 28,76 copies/mL, with 95% confidence limits between 22,19 and 52,27 copies
Gaydos, C. A.; Cartwright, C. P.; Colaninno, P.; Welsch, J.; Holden, J.; Ho, S. Y.; Webb, E. M.; Anderson, C.; Bertuzis, R.; Zhang, L.; Miller, T.; Leckie, G.; Abravaya, K.; Robinson, J.
A multicenter clinical study was conducted to evaluate the performance characteristics of the Abbott RealTime CT/NG assay, a multiplex real-time PCR assay, for simultaneous detection of Chlamydia trachomatis and Neisseria gonorrhoeae. The specimens were collected from a total of 3,832 male and female subjects at 16 geographically diverse sites. Specimens included male and female urine samples, male urethral swabs, female endocervical swabs, and self-collected and clinician-collected vaginal swabs. Specimens were tested with the automated Abbott RealTime CT/NG assay, Aptima Combo 2 assay (Gen-Probe), ProbeTec ET CT/GC assay (Becton Dickinson), and culture for N. gonorrhoeae. The Aptima Combo 2 assay, the ProbeTec assay, and the N. gonorrhoeae culture were used as the reference assays. For each subject, a patient infected status (PIS) was determined based on the combined results from the reference assays. The overall prevalence in female subjects was 8.9% for C. trachomatis and 3.8% for N. gonorrhoeae. The overall male prevalence was 18.2% for C. trachomatis and 16.7% for N. gonorrhoeae. The overall sensitivity and specificity of the Abbott RealTime CT/NG assay were 92.4% and 99.2% for C. trachomatis and 96.9% and 99.7% for N. gonorrhoeae, respectively. In comparison, the sensitivity and specificity, respectively, for the Aptima Combo 2 assay were 94.5% and 99.0% for C. trachomatis and 96.1% and 99.5% for N. gonorrhoeae, and those for the ProbeTec ET assay were 90.3% and 99.5% for C. trachomatis and 92.0% and 97.3% for N. gonorrhoeae in this study. The Abbott RealTime CT/NG assay offers C. trachomatis and N. gonorrhoeae dual detection with high sensitivity and specificity. The automated assay provides a useful alternative nucleic acid amplification assay for clinical laboratories and clinicians. PMID:20668135
Gaydos, C A; Cartwright, C P; Colaninno, P; Welsch, J; Holden, J; Ho, S Y; Webb, E M; Anderson, C; Bertuzis, R; Zhang, L; Miller, T; Leckie, G; Abravaya, K; Robinson, J
A multicenter clinical study was conducted to evaluate the performance characteristics of the Abbott RealTime CT/NG assay, a multiplex real-time PCR assay, for simultaneous detection of Chlamydia trachomatis and Neisseria gonorrhoeae. The specimens were collected from a total of 3,832 male and female subjects at 16 geographically diverse sites. Specimens included male and female urine samples, male urethral swabs, female endocervical swabs, and self-collected and clinician-collected vaginal swabs. Specimens were tested with the automated Abbott RealTime CT/NG assay, Aptima Combo 2 assay (Gen-Probe), ProbeTec ET CT/GC assay (Becton Dickinson), and culture for N. gonorrhoeae. The Aptima Combo 2 assay, the ProbeTec assay, and the N. gonorrhoeae culture were used as the reference assays. For each subject, a patient infected status (PIS) was determined based on the combined results from the reference assays. The overall prevalence in female subjects was 8.9% for C. trachomatis and 3.8% for N. gonorrhoeae. The overall male prevalence was 18.2% for C. trachomatis and 16.7% for N. gonorrhoeae. The overall sensitivity and specificity of the Abbott RealTime CT/NG assay were 92.4% and 99.2% for C. trachomatis and 96.9% and 99.7% for N. gonorrhoeae, respectively. In comparison, the sensitivity and specificity, respectively, for the Aptima Combo 2 assay were 94.5% and 99.0% for C. trachomatis and 96.1% and 99.5% for N. gonorrhoeae, and those for the ProbeTec ET assay were 90.3% and 99.5% for C. trachomatis and 92.0% and 97.3% for N. gonorrhoeae in this study. The Abbott RealTime CT/NG assay offers C. trachomatis and N. gonorrhoeae dual detection with high sensitivity and specificity. The automated assay provides a useful alternative nucleic acid amplification assay for clinical laboratories and clinicians.
Schalin-Jäntti, Camilla; Tanner, Pirjo; Välimäki, Matti J; Hämäläinen, Esa
Current serum TSH reference intervals have been criticized as they were established from unselected background populations. A special concern is that the upper limit, which defines subclinical hypothyroidism, is too high. The objective was to redefine the TSH reference interval in the adult Finnish population. The current reference interval for the widely used Abbott Architect method in Finland is 0.4-4.0 mU/L. Serum TSH and free T4 concentrations were derived from 606 healthy, non-pregnant, 18-91-year-old Finns from the Nordic Reference Interval Project (NORIP) and the possible effects of age, sex and thyroid peroxidase antibody (TPOAb) status were evaluated. After excluding TPOAb-positive subjects and outliers, a reference population of 511 subjects was obtained. In the reference population, no statistically significant gender- or age-specific differences in mean TSH (1.55 ± 3.30 mU/L) or TSH reference intervals were observed. The new reference interval was 0.5-3.6 mU/L (2.5th-97.5th percentiles). The current upper TSH reference limit is 10% too high. A TSH > 3.6 mU/L, confirmed with a repeat TSH sampling, may indicate subclinical hypothyroidism. Differences in ethnicity, regional iodine-intake and analytical methods underline the need for redefining the TSH reference interval in central laboratories in different countries.
Tam, Kingsley King-Gee; Leung, Kenneth Siu-Sing; To, Sabrina Wai-Chi; Siu, Gilman Kit-Hang; Lau, Terrence Chi-Kong; Shek, Victor Chi-Man; Tse, Cindy Wing-Sze; Wong, Samson Sai-Yin; Ho, Pak-Leung; Yam, Wing-Cheong
Abbott RealTime MTB (Abbott-RT) in conjunction with Abbott RealTime MTB RIF/INH Resistance (Abbott-RIF/INH) is a new, high-throughput automated nucleic acid amplification platform (Abbott-MDR) for detection of Mycobacterium tuberculosis complex (MTBC) and the genotypic markers for rifampicin (RIF) and isoniazid (INH) resistance directly from respiratory specimens. This prospective study evaluated the diagnostic performance of this new platform for MTBC and multidrug-resistant tuberculosis (MDR-TB) using 610 sputum specimens in a tuberculosis high-burden setting. Using conventional culture results and clinical background as reference standards, Abbott-RT exhibited an overall sensitivity and specificity of 95.2% and 99.8%, respectively. Genotypic RIF/INH resistance of 178 "MTB detected" specimens was subsequently analyzed by Abbott-RIF/INH. Compared to phenotypic drug susceptibility test results, Abbott-RIF/INH detected resistance genotypic markers in 84.6% MDR-TB, 80% mono-RIF-resistant and 66.7% mono-INH-resistant specimens. Two of the RIF-resistant specimens carried a novel single, nonsense mutation at rpoB Q513 and in silico simulation demonstrated that the truncated RpoB protein failed to bind with other subunits for transcription. Overall, Abbott-MDR platform provided high throughput and reliable diagnosis of MDR-TB within a TB high-burden region. Copyright © 2017 Elsevier Inc. All rights reserved.
Aloisio, Elena; Carnevale, Assunta; Pasqualetti, Sara; Birindelli, Sarah; Dolci, Alberto; Panteghini, Mauro
Automatic photometric determination of the hemolysis index (HI) on serum and plasma samples is central to detect potential interferences of in vitro hemolysis on laboratory tests. When HI is above an established cut-off for interference, results may suffer from a significant bias and undermine clinical reliability of the test. Despite its undeniable importance for patient safety, the analytical performance of HI estimation is not usually checked in laboratories. Here we evaluated for the first time the random source of measurement uncertainty of HI determination on the two Abbott Architect c16000 platforms in use in our laboratory. From January 2016 to September 2017, we collected data from daily photometric determination of HI on a fresh-frozen serum pool with a predetermined HI value of ~100 (corresponding to ~1g/L of free hemoglobin). Monthly and cumulative CVs were calculated. During 21months, 442 and 451 measurements were performed on the two platforms, respectively. Monthly CVs ranged from 0.7% to 2.7% on c16000-1 and from 0.8% to 2.5% on c16000-2, with a between-platform cumulative CV of 1.82% (corresponding to an expanded uncertainty of 3.64%). Mean HI values on the two platforms were just slightly biased (101.3 vs. 103.1, 1.76%), but, due to the high precision of measurements, this difference assumed statistical significance (p<0.0001). Even though no quality specifications are available to date, our study shows that the HI measurement on Architect c16000 platform has nice reproducibility that could be considered in establishing the state of the art of the measurement. Copyright © 2018 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
Establishment of detailed reference values for luteinizing hormone, follicle stimulating hormone, estradiol, and progesterone during different phases of the menstrual cycle on the Abbott ARCHITECT® analyzer
Stricker, Reto; Eberhart, Raphael; Chevailler, Marie-Christine; Quinn, Frank A.; Bischof, Paul; Stricker, René
During a normal menstrual cycle, serum levels of luteinizing hormone (LH), follicle-stimulating hormone (FSH), estradiol, and progesterone can vary widely between cycles for the same woman, as well as between different woman. Reliable reference values based on the local population are important for correct interpretation of laboratory results. The purpose of our study was to determine detailed reference values for these hormones throughout the menstrual cycle using the Abbott ARCHITECT system...
Opinnäytetyö suoritettiin THL:n Tautiriskiyksikön analyyttisen biokemian laboratoriossa (TLAB). Työssä validoitiin uusi entsymaattinen Abbott Architect c8000 HbA1c -menetelmä, jota käytetään diabetekseen liittyvissä tutkimuksissa. Validoinnilla haluttiin varmistaa uuden mittaustekniikaltaan erilaisen menetelmän toimivuus. Menetelmävertailussa komparatiivisena menetelmänä oli laboratoriossa rutiinikäytössä ollut Abbottin immunoturbidimetrinen HbA1c-menetelmä. Uusi entsymaattinen menetelmä peru...
Ozarda, Yesim; Ichihara, Kiyoshi; Aslan, Diler; Aybek, Hulya; Ari, Zeki; Taneli, Fatma; Coker, Canan; Akan, Pinar; Sisman, Ali Riza; Bahceci, Onur; Sezgin, Nurzen; Demir, Meltem; Yucel, Gultekin; Akbas, Halide; Ozdem, Sebahat; Polat, Gurbuz; Erbagci, Ayse Binnur; Orkmez, Mustafa; Mete, Nuriye; Evliyaoglu, Osman; Kiyici, Aysel; Vatansev, Husamettin; Ozturk, Bahadir; Yucel, Dogan; Kayaalp, Damla; Dogan, Kubra; Pinar, Asli; Gurbilek, Mehmet; Cetinkaya, Cigdem Damla; Akin, Okhan; Serdar, Muhittin; Kurt, Ismail; Erdinc, Selda; Kadicesme, Ozgur; Ilhan, Necip; Atali, Dilek Sadak; Bakan, Ebubekir; Polat, Harun; Noyan, Tevfik; Can, Murat; Bedir, Abdulkerim; Okuyucu, Ali; Deger, Orhan; Agac, Suret; Ademoglu, Evin; Kaya, Ayşem; Nogay, Turkan; Eren, Nezaket; Dirican, Melahat; Tuncer, GulOzlem; Aykus, Mehmet; Gunes, Yeliz; Ozmen, Sevda Unalli; Kawano, Reo; Tezcan, Sehavet; Demirpence, Ozlem; Degirmen, Elif
A nationwide multicenter study was organized to establish reference intervals (RIs) in the Turkish population for 25 commonly tested biochemical analytes and to explore sources of variation in reference values, including regionality. Blood samples were collected nationwide in 28 laboratories from the seven regions (≥400 samples/region, 3066 in all). The sera were collectively analyzed in Uludag University in Bursa using Abbott reagents and analyzer. Reference materials were used for standardization of test results. After secondary exclusion using the latent abnormal values exclusion method, RIs were derived by a parametric method employing the modified Box-Cox formula and compared with the RIs by the non-parametric method. Three-level nested ANOVA was used to evaluate variations among sexes, ages and regions. Associations between test results and age, body mass index (BMI) and region were determined by multiple regression analysis (MRA). By ANOVA, differences of reference values among seven regions were significant in none of the 25 analytes. Significant sex-related and age-related differences were observed for 10 and seven analytes, respectively. MRA revealed BMI-related changes in results for uric acid, glucose, triglycerides, high-density lipoprotein (HDL)-cholesterol, alanine aminotransferase, and γ-glutamyltransferase. Their RIs were thus derived by applying stricter criteria excluding individuals with BMI >28 kg/m2. Ranges of RIs by non-parametric method were wider than those by parametric method especially for those analytes affected by BMI. With the lack of regional differences and the well-standardized status of test results, the RIs derived from this nationwide study can be used for the entire Turkish population.
Schneider, George J; Kuper, Kevin G; Abravaya, Klara; Mullen, Carolyn R; Schmidt, Marion; Bunse-Grassmann, Astrid; Sprenger-Haussels, Markus
Automated sample preparation systems must meet the demands of routine diagnostics laboratories with regard to performance characteristics and compatibility with downstream assays. In this study, the performance of QIAGEN EZ1 DSP Virus Kit on the BioRobot EZ1 DSP was evaluated in combination with the Abbott RealTime HIV-1, HCV, and HBV assays, followed by thermalcycling and detection on the Abbott m2000rt platform. The following performance characteristics were evaluated: linear range and precision, sensitivity, cross-contamination, effects of interfering substances and correlation. Linearity was observed within the tested ranges (for HIV-1: 2.0-6.0 log copies/ml, HCV: 1.3-6.9 log IU/ml, HBV: 1.6-7.6 log copies/ml). Excellent precision was obtained (inter-assay standard deviation for HIV-1: 0.06-0.17 log copies/ml (>2.17 log copies/ml), HCV: 0.05-0.11 log IU/ml (>2.09 log IU/ml), HBV: 0.03-0.07 log copies/ml (>2.55 log copies/ml)), with good sensitivity (95% hit rates for HIV-1: 50 copies/ml, HCV: 12.5 IU/ml, HBV: 10 IU/ml). No cross-contamination was observed, as well as no negative impact of elevated levels of various interfering substances. In addition, HCV and HBV viral load measurements after BioRobot EZ1 DSP extraction correlated well with those obtained after Abbott m2000sp extraction. This evaluation demonstrates that the QIAGEN EZ1 DSP Virus Kit provides an attractive solution for fully automated, low throughput sample preparation for use with the Abbott RealTime HIV-1, HCV, and HBV assays.
Salmona, Maud; Fourati, Slim; Feghoul, Linda; Scieux, Catherine; Thiriez, Aline; Simon, François; Resche-Rigon, Matthieu; LeGoff, Jérôme
Accurate quantification of Epstein-Barr virus (EBV) load in blood is essential for the management of post-transplant lymphoproliferative disorders. The automation of DNA extraction and amplification may improve accuracy and reproducibility. We evaluated the EBV PCR Kit V1 with fully automated DNA extraction and amplification on the m2000 system (Abbott assay). Conversion factor between copies and international units (IU), lower limit of quantification, imprecision and linearity were determined in a whole blood (WB) matrix. Results from 339 clinical WB specimens were compared with a home-brew real-time PCR assay used in our laboratory (in-house assay). The conversion factor between copies and IU was 3.22 copies/IU. The lower limit of quantification (LLQ) was 1000 copies/mL. Intra- and inter-assay coefficients of variation were 3.1% and 7.9% respectively for samples with EBV load higher than the LLQ. The comparison between Abbott assay and in-house assay showed a good concordance (kappa = 0.77). Loads were higher with the Abbott assay (mean difference = 0.62 log10 copies/mL). The EBV PCR Kit V1 assay on the m2000 system provides a reliable and easy-to-use method for quantification of EBV DNA in WB. Copyright © 2016 Elsevier Inc. All rights reserved.
Bond, GR; Steele, PE; Uges, DRA
Case report: A 13-yr-old girl overdosed on 48 x 150 mg venlafaxine (Effexor XR(R)). She was taking venlafaxine regularly for depression. Her only other medications included topical Benzamycin and pyridoxine 50 mg daily for acne. The Abbott AxSYM(R) assay was positive only for phencyclidine, but
Ismail, Ashrafali M.; Sivakumar, Jayashree; Anantharam, Raghavendran; Dayalan, Sujitha; Samuel, Prasanna; Fletcher, Gnanadurai J.; Gnanamony, Manu; Abraham, Priya
Virological monitoring of hepatitis B virus (HBV) DNA is critical to the management of HBV infection. With several HBV DNA quantification assays available, it is important to use the most efficient testing system for virological monitoring. In this study, we evaluated the performance characteristics and comparability of three HBV DNA quantification systems: Abbott HBV real-time PCR (Abbott PCR), artus HBV real-time PCR with QIAamp DNA blood kit purification (artus-DB), and artus HBV real-time PCR with the QIAamp DSP virus kit purification (artus-DSP). The lower limits of detection of these systems were established against the WHO international standards for HBV DNA and were found to be 1.43, 82, and 9 IU/ml, respectively. The intra-assay and interassay coefficients of variation of plasma samples (1 to 6 log10 IU/ml) ranged between 0.05 to 8.34% and 0.16 to 3.48% for the Abbott PCR, 1.53 to 26.85% and 0.50 to 12.89% for artus-DB, and 0.29 to 7.42% and 0.94 to 3.01% for artus-DSP, respectively. Ninety HBV clinical samples were used for comparison of assays, and paired quantitative results showed strong correlation by linear regression analysis (artus-DB with Abbott PCR, r = 0.95; Abbott PCR with artus-DSP, r = 0.97; and artus-DSP with artus-DB, r = 0.94). Bland-Altman analysis showed a good level of agreement for Abbott PCR and artus-DSP, with a mean difference of 0.10 log10 IU/ml and limits of agreement of −0.91 to 1.11 log10 IU/ml. No genotype-specific bias was seen in all three systems for HBV genotypes A, C, and D, which are predominant in this region. This finding illustrates that the Abbott real-time HBV and artus-DSP systems show more comparable performance than the artus-DB system, meeting the current guidelines for assays to be used in the management of hepatitis B. PMID:21795507
Ismail, Ashrafali M; Sivakumar, Jayashree; Anantharam, Raghavendran; Dayalan, Sujitha; Samuel, Prasanna; Fletcher, Gnanadurai J; Gnanamony, Manu; Abraham, Priya
Virological monitoring of hepatitis B virus (HBV) DNA is critical to the management of HBV infection. With several HBV DNA quantification assays available, it is important to use the most efficient testing system for virological monitoring. In this study, we evaluated the performance characteristics and comparability of three HBV DNA quantification systems: Abbott HBV real-time PCR (Abbott PCR), artus HBV real-time PCR with QIAamp DNA blood kit purification (artus-DB), and artus HBV real-time PCR with the QIAamp DSP virus kit purification (artus-DSP). The lower limits of detection of these systems were established against the WHO international standards for HBV DNA and were found to be 1.43, 82, and 9 IU/ml, respectively. The intra-assay and interassay coefficients of variation of plasma samples (1 to 6 log(10) IU/ml) ranged between 0.05 to 8.34% and 0.16 to 3.48% for the Abbott PCR, 1.53 to 26.85% and 0.50 to 12.89% for artus-DB, and 0.29 to 7.42% and 0.94 to 3.01% for artus-DSP, respectively. Ninety HBV clinical samples were used for comparison of assays, and paired quantitative results showed strong correlation by linear regression analysis (artus-DB with Abbott PCR, r = 0.95; Abbott PCR with artus-DSP, r = 0.97; and artus-DSP with artus-DB, r = 0.94). Bland-Altman analysis showed a good level of agreement for Abbott PCR and artus-DSP, with a mean difference of 0.10 log(10) IU/ml and limits of agreement of -0.91 to 1.11 log(10) IU/ml. No genotype-specific bias was seen in all three systems for HBV genotypes A, C, and D, which are predominant in this region. This finding illustrates that the Abbott real-time HBV and artus-DSP systems show more comparable performance than the artus-DB system, meeting the current guidelines for assays to be used in the management of hepatitis B.
Babady, N. Esther; Germer, Jeffrey J.; Yao, Joseph D. C.
No significantly discordant results were observed between the Abbott RealTime HIV-1 assay and the COBAS AmpliPrep/COBAS TaqMan HIV-1 Test (CTM) among 1,190 unique clinical plasma specimens obtained from laboratories located in 40 states representing all nine U.S. geographic regions and previously yielding “target not detected” results by CTM.
Babady, N Esther; Germer, Jeffrey J; Yao, Joseph D C
No significantly discordant results were observed between the Abbott RealTime HIV-1 assay and the COBAS AmpliPrep/COBAS TaqMan HIV-1 Test (CTM) among 1,190 unique clinical plasma specimens obtained from laboratories located in 40 states representing all nine U.S. geographic regions and previously yielding "target not detected" results by CTM.
Mallory, Melanie A; Lucic, Danijela X; Sears, Mitchell T; Cloherty, Gavin A; Hillyard, David R
HCV genotyping is a critical tool for guiding initiation of therapy and selecting the most appropriate treatment regimen. To evaluate the concordance between the Abbott GT II assay and genotyping by sequencing subregions of the HCV 5'UTR, core and NS5B. The Abbott assay was used to genotype 127 routine patient specimens and 35 patient specimens with unusual subtypes and mixed infection. Abbott results were compared to genotyping by 5'UTR, core and NS5B sequencing. Sequences were genotyped using the NCBI non-redundant database and the online genotyping tool COMET. Among routine specimens, core/NS5B sequencing identified 93 genotype 1s, 13 genotype 2s, 15 genotype 3s, three genotype 4s, two genotype 6s and one recombinant specimen. Genotype calls by 5'UTR, core, NS5B sequencing and the Abbott assay were 97.6% concordant. Core/NS5B sequencing identified two discrepant samples as genotype 6 (subtypes 6l and 6u) while Abbott and 5'UTR sequencing identified these samples as genotype 1 with no subtype. The Abbott assay subtyped 91.4% of genotype 1 specimens. Among the 35 rare specimens, the Abbott assay inaccurately genotyped 3k, 6e, 6o, 6q and one genotype 4 variant; gave indeterminate results for 3g, 3h, 4r, 6m, 6n, and 6q specimens; and agreed with core/NS5B sequencing for mixed specimens. The Abbott assay is an automated HCV genotyping method with improved accuracy over 5'UTR sequencing. Samples identified by the Abbott assay as genotype 1 with no subtype may be rare subtypes of other genotypes and thus require confirmation by another method. Copyright © 2014 Elsevier B.V. All rights reserved.
Jun-Jin Peng; Wen-Chang Xiang; Shao-Hong Cai
We consider the conserved charge of static black holes with squashed horizons in the Einstein-Maxwell-dilaton theory via both the Abbott-Deser-Tekin (ADT) method and its off-shell generalization.We first make use of the original ADT method to compute the mass of the dilaton squashed black holes in terms of three different reference spacetimes,which are the asymptotic geometry,the fiat background and the spacetime of the KaluzaKlein monopole with boundary matched to the original metric,respectively.Each mass satisfies the first law of black hole thermodynamics,although the mass computed on the basis of the boundary matching the KaluzaKlein monopole is different from that of the other two reference spacetimes.Then the mass of the black holes is evaluated through the off-shell generalized ADT method.
Georgescu, C.; Cristea, G. C.; Dima, C.; Deleanu, L.
The paper presents the change of functional parameters defined on the Abbott-Firestone curve in order to evaluate the surface quality of the balls from the four ball tester, after tests done with several vegetable oils. The tests were done using two grades of rapeseed oil (degummed and refined) and two grades of soybean oil (coarse and degummed) and a common transmission oil (T90). Test parameters were 200 N and 0.576 m/s (1500 rpm) for 60 minutes. For the refined rapeseed oil, the changes in shape of the Abbott-Firestone curves are more dramatic, these being characterized by high values of Spk (the average value for the wear scars on the three balls), thus being 40% of the sum Svk + Sk + Spk, percentage also obtained for the soybean oil, but the value Spk being lower. For the degummed soybean oil, the profile height of the wear scars are taller than those obtained after testing the coarse soybean oil, meaning that the degumming process has a negative influence on the worn surface quality and the lubricating capacity of this oil. Comparing the surface quality of the wear scars on fixed tested balls is a reliable method to point out the lubricant properties of the vegetable oils, especially if they are compared to a “classical” lubricant as a non-additivated transmission mineral oil T90. The best surface after testing was obtained for the soybean oil, followed by T90 oil and the degummed grades of the soybean oil and rapeseed oil (these three giving very close values for the functional parameters), but the refined rapeseed oil generated the poorest quality of the wear scars on the balls, under the same testing conditions.
Muldrew, Kenneth L; Lovett, Jennie L
BK polyomavirus (BKPyV) quantification is useful for monitoring renal transplant patient response to therapy. The Abbott m2000 RealTime System employed by some clinical laboratories to perform US Food and Drug Administration-approved assays can also be used to develop in-house assays such as the one presented here. This study aimed to validate an in-house quantitative real-time PCR assay targeting the BKPyV major capsid VP1 gene for assessment of viral load using the Abbott m2000 RealTime System. BKPyV load was measured in 95 urine and plasma samples previously tested for BKPyV by one of three laboratories (46 BKPyV-positive samples consisting of 35 plasma and 11 urine samples; 49 samples negative for BKPyV consisting of 47 plasma and two urine samples). Two additional plasma specimens from the College of American Pathologists proficiency testing survey were also analysed. Precision studies were performed by diluting a high-viral-titre patient sample into BKPyV-negative pooled plasma to create high-positive (6.16 log10 copies ml(-1)) and low-positive (3.16 log10 copies ml(-1)) samples. For precision studies of inter-assay variability, a high-positive (7.0 log10 copies ml(-1)) and a low-positive (3.0 log10 copies ml(-1)) sample were measured in 20 separate runs. The assay's limit of quantification and limit of detection were 2.70 and 2.25 log10 copies ml(-1), respectively. The assay was linear from 2.70 to 9.26 log10 copies ml(-1). Of the 48 known positives, 43 were detected as positive, with three reported by the reference laboratory as values lower than the limit of detection. Two known positives at 3.27 and 3.80 log10 copies ml(-1) tested negative by the m2000 BKPyV assay. Of the 49 known negative samples, 48 were negative by the m2000 BKPyV load assay, with one sample confirmed positive by a reference laboratory. Qualitative analysis prior to discrepancy testing demonstrated a sensitivity of 89.58 % and a specificity of 97.96 %. Precision studies
José Marcos de Castro Nunes
Full Text Available Specimens of Dotyophycus pacificum I. A. Abbott were found during a survey of Rhodophyta on the coast of Bahia state. The samples were taken from 23-36 meters depth and the specimens found were studied in detail and compared to other morphologically similar species. This is the first time that the genus Dotyophycus is cited for the Atlantic Ocean.Durante estudo sobre as rodofíceas do litoral do estado da Bahia foram encontrados exemplares de Dotyophycus pacificum Abbott em coletas realizadas a 23-36 metros de profundidade. Os espécimes foram estudados detalhadamente e comparados a espécies morfologicamente semelhantes. Esta é a primeira ocorrência de D. pacificum para o oceano Atlântico.
Tesija Kuna, Andrea; Dukic, Kristina; Nikolac Gabaj, Nora; Miler, Marijana; Vukasovic, Ines; Langer, Sanja; Simundic, Ana-Maria; Vrkic, Nada
To compare the analytical performances of the enzymatic method (EM) and capillary electrophoresis (CE) for hemoglobin A1c (HbA1c) measurement. Imprecision, carryover, stability, linearity, method comparison, and interferences were evaluated for HbA1c via EM (Abbott Laboratories, Inc) and CE (Sebia). Both methods have shown overall within-laboratory imprecision of less than 3% for International Federation of Clinical Chemistry and Laboratory Medicine (IFCC) units (<2% National Glycohemoglobin Standardization Program [NGSP] units). Carryover effects were within acceptable criteria. The linearity of both methods has proven to be excellent (R2 = 0.999). Significant proportional and constant difference were found for EM, compared with CE, but were not clinically relevant (<5 mmol/mol; NGSP <0.5%). At the clinically relevant HbA1c concentration, stability observed with both methods was acceptable (bias, <3%). Triglyceride levels of 8.11 mmol per L or greater showed to interfere with EM and fetal hemoglobin (HbF) of 10.6% or greater with CE. The enzymatic method proved to be comparable to the CE method in analytical performances; however, certain interferences can influence the measurements of each method.
Alfredo Jácome Roca
Full Text Available
La Academia Nacional de Medicina con la colaboración de Laboratorios Abbott otorgó en noviembre de 2011 la quinta versión del Premio a las Ciencias Médicas, uno en las áreas de ciencias médicas y experimentales y otro en el área de ciencias clínicas. Presentamos aquí algunas notas sobre los trabajos ganadores y los que obtuvieron menciones honoríficas
El Premio en el área de las Ciencias Básicas lo obtuvieron Carlos Vélez Pardo y Marlene Jiménez del Río, profesores del Grupo de Neurociencias de la Facultad de Medicina de la Universidad de Antioquia en Medellín. El trabajo ganador se tituló: “MECANISMOS MOLECULARES DE PÉRDIDA NEURONAL Y DE CITOPROTECCIÓN EN MODE-LOS IN VITRO E IN VIVO DE LA ENFERMEDAD DE ALZHEIMER Y PARKINSON” [1,2].
Saiki, M.K.; Martin, B.A.
This study was conducted to gain a better understanding of fishery resources in Abbotts Lagoon, Point Reyes National Seashore. During February/March, May, August, and November 1999, fish were sampled with floating variable-mesh gill nets and small minnow traps from as many as 14 sites in the lagoon. Water temperature, dissolved oxygen, pH, total ammonia(NH3 + NH4+), salinity, turbidity, water depth, and bottom substrate composition were also measured at each site. A total of 2,656 fish represented by eight species was captured during the study. Gill nets captured Sacramento perch, Archoplites interruptus; largemouth bass, Micropterus salmoides; Pacific herring, Clupea pallasi; prickly sculpin, Cottus asper, silver surfperch, Hyperprosopon ellipticum; longfin smelt, Spirinchus thaleichthys; and striped bass, Morone saxatilis; whereas minnow traps captured Sacramento perch; prickly sculpin; and threespine stickleback, Gasterosteus aculeatus. Cluster analysis (Ward's minimum variance method of fish catch statistics identified two major species assemblages-the first dominated by Sacramento perch and, to a lesser extent, by largemouth bass, and the second dominated by Pacific herring and threespine stickleback. Simple discriminant analysis of environmental variables indicated that salinity contributed the most towards separating the two assemblages.
Sheffield, Catherine A; Kane, Michael P; Bakst, Gary; Busch, Robert S; Abelseth, Jill M; Hamilton, Robert A
This study compared the accuracy and precision of four value-added glucose meters. Finger stick glucose measurements in diabetes patients were performed using the Abbott Diabetes Care (Alameda, CA) Optium, Diagnostic Devices, Inc. (Miami, FL) DDI Prodigy, Home Diagnostics, Inc. (Fort Lauderdale, FL) HDI True Track Smart System, and Arkray, USA (Minneapolis, MN) HypoGuard Assure Pro. Finger glucose measurements were compared with laboratory reference results. Accuracy was assessed by a Clarke error grid analysis (EGA), a Parkes EGA, and within 5%, 10%, 15%, and 20% of the laboratory value criteria (chi2 analysis). Meter precision was determined by calculating absolute mean differences in glucose values between duplicate samples (Kruskal-Wallis test). Finger sticks were obtained from 125 diabetes patients, of which 90.4% were Caucasian, 51.2% were female, 83.2% had type 2 diabetes, and average age of 59 years (SD 14 years). Mean venipuncture blood glucose was 151 mg/dL (SD +/-65 mg/dL; range, 58-474 mg/dL). Clinical accuracy by Clarke EGA was demonstrated in 94% of Optium, 82% of Prodigy, 61% of True Track, and 77% of the Assure Pro samples (P Abbott Optium was significantly more accurate than the other meter systems, whereas the HDI True Track was significantly less accurate and less precise compared to the other meter systems.
Helmersson-Karlqvist, Johanna; Flodin, Mats; Havelka, Aleksandra Mandic; Xu, Xiao Yan; Larsson, Anders
Serum/plasma albumin is an important and widely used laboratory marker and it is important that we measure albumin correctly without bias. We had indications that the immunoturbidimetric method on Cobas c 501 and the bromocresol purple (BCP) method on Architect 16000 differed, so we decided to study these methods more closely. A total of 1,951 patient requests with albumin measured with both the Architect BCP and Cobas immunoturbidimetric methods were extracted from the laboratory system. A comparison with fresh plasma samples was also performed that included immunoturbidimetric and BCP methods on Cobas c 501 and analysis of the international protein calibrator ERM-DA470k/IFCC. The median difference between the Abbott BCP and Roche immunoturbidimetric methods was 3.3 g/l and the Roche method overestimated ERM-DA470k/IFCC by 2.2 g/l. The Roche immunoturbidimetric method gave higher values than the Roche BCP method: y = 1.111x - 0.739, R² = 0.971. The Roche immunoturbidimetric albumin method gives clearly higher values than the Abbott and Roche BCP methods when analyzing fresh patient samples. The differences between the two methods were similar at normal and low albumin levels. © 2016 Wiley Periodicals, Inc.
Ollier, Laurence; Laffont, Catherine; Kechkekian, Aurore; Doglio, Alain; Giordanengo, Valérie
Routine use of the automated chemiluminescent microparticle immunoassay Abbott ARCHITECT anti-HBc for diagnosis of hepatitis B is limited in case of borderline reactive sera with low signal close to the cut-off index. In order to determine the significance of anti-HBc detection when borderline reactivity occurs using the ARCHITECT anti-HBc assay, a comparative study was designed. 3540 serum samples collected over a 2-month period in the hospital of Nice were examined for markers of HBV infection (HBsAg, anti-HBs and anti-HBc). One hundred seven samples with sufficient volume and with borderline reactivity by the ARCHITECT assay were tested by two other anti-HBc assays, a microparticle enzyme immunoassay (MEIA, AxSYM Core, Abbott Laboratories, IL, USA) and an enzyme linked fluorescent assay (ELFA, VIDAS Anti-HBc Total II, bioMérieux, Lyon, France). Only 46 samples were confirmed by the AxSYM and the VIDAS assays. Additional serological information linked to patient history showed that the remaining samples (61) were false positives (11), had low titer of anti-HBc antibodies (13), or were inconclusive (37). This comparative study highlighted the existence of a grey zone around the cut-off index. Confirmative results through a different immunoassay are needed to confirm the diagnosis of HBV on borderline reactive sera using the ARCHITECT anti-HBc assay.
Stellrecht, K A; Espino, A A; Maceira, V P; Nattanmai, S M; Butt, S A; Wroblewski, D; Hannett, G E; Musser, K A
Clostridium difficile-associated diarrhea is a well-recognized complication of antibiotic use. Historically, diagnosing C. difficile has been difficult, as antigen assays are insensitive and culture-based methods require several days to yield results. Nucleic acid amplification tests (NAATs) are quickly becoming the standard of care. We compared the performance of two automated investigational/research use only (IUO/RUO) NAATs for the detection of C. difficile toxin genes, the IMDx C. difficile for Abbott m2000 Assay (IMDx) and the BD Max Cdiff Assay (Max). A prospective analysis of 111 stool specimens received in the laboratory for C. difficile testing by the laboratory's test of record (TOR), the BD GeneOhm Cdiff Assay, and a retrospective analysis of 88 specimens previously determined to be positive for C. difficile were included in the study. One prospective specimen was excluded due to loss to follow-up discrepancy analysis. Of the remaining 198 specimens, 90 were positive by all three methods, 9 were positive by TOR and Max, and 3 were positive by TOR only. One negative specimen was initially inhibitory by Max. The remaining 95 specimens were negative by all methods. Toxigenic C. difficile culture was performed on the 12 discrepant samples. True C. difficile-positive status was defined as either positive by all three amplification assays or positive by toxigenic culture. Based on this definition, the sensitivity and specificity were 96.9% and 95% for Max and 92.8% and 100% for IMDx. In summary, both highly automated systems demonstrated excellent performance, and each has individual benefits, which will ensure that they will both have a niche in clinical laboratories.
Alfredo Jácome Roca
Full Text Available
El Premio en el área de las Ciencias Básicas lo obtuvieron profesores de la Universidad Nacional de Colombia en Bogotá, pertenecientes al Laboratorio de Biología Molecular de Virus de la Facultad de Medicina y al Departamento de Química de la Facultad de Ciencias. Se trata de los investigadores Carlos Arturo Guerrero Fonseca, Martha Calderón, Orlando Acosta y Fanny Guzmán.
El trabajo se titula "Interferencia de la infección por rotavirus mediante la inhibición de la actividad de
la proteína disulfuro isomerasa (PDI de la membrana celular de las líneas MA 104 y Caen-2".
Sridhar, Siddharth; Yip, Cyril C Y; Chan, Jasper F W; To, Kelvin K W; Cheng, Vincent C C; Yuen, Kwok-Yung
The Abbott RealTime HCV Genotype II assay (Abbott-RT-HCV assay) is a real-time PCR based genotyping method for hepatitis C virus (HCV). This study measured the impact of inter-genotypic recombination and probe cross-reactivity on the performance of the Abbott-RT-HCV assay. 517 samples were genotyped using the Abbott-RT-HCV assay over a one-year period, 34 (6.6%) were identified as HCV genotype 1 without further subtype designation raising the possibility of inaccurate genotyping. These samples were subjected to confirmatory sequencing. 27 of these 34 (79%) samples were genotype 1b while five (15%) were genotype 6. One HCV isolate was an inter-genotypic 1a/4o recombinant. This is a novel natural HCV recombinant that has never been reported. Inter-genotypic recombination and probe cross-reactivity can affect the accuracy of the Abbott-RT-HCV assay, both of which have significant implications on antiviral regimen choice. Confirmatory sequencing of ambiguous results is crucial for accurate genotyping. Copyright © 2018 Elsevier Inc. All rights reserved.
Tang, Ning; Pahalawatta, Vihanga; Frank, Andrea; Bagley, Zowie; Viana, Raquel; Lampinen, John; Leckie, Gregor; Huang, Shihai; Abravaya, Klara; Wallis, Carole L
HIV RNA suppression is a key indicator for monitoring success of antiretroviral therapy. From a logistical perspective, viral load (VL) testing using Dried Blood Spots (DBS) is a promising alternative to plasma based VL testing in resource-limited settings. To evaluate the analytical and clinical performance of the Abbott RealTime HIV-1 assay using a fully automated one-spot DBS sample protocol. Limit of detection (LOD), linearity, lower limit of quantitation (LLQ), upper limit of quantitation (ULQ), and precision were determined using serial dilutions of HIV-1 Virology Quality Assurance stock (VQA Rush University), or HIV-1-containing armored RNA, made in venous blood. To evaluate correlation, bias, and agreement, 497 HIV-1 positive adult clinical samples were collected from Ivory Coast, Uganda and South Africa. For each HIV-1 participant, DBS-fingerprick, DBS-venous and plasma sample results were compared. Correlation and bias values were obtained. The sensitivity and specificity were analyzed at a threshold of 1000 HIV-1 copies/mL generated using the standard plasma protocol. The Abbott HIV-1 DBS protocol had an LOD of 839 copies/mL, a linear range from 500 to 1×10 7 copies/mL, an LLQ of 839 copies/mL, a ULQ of 1×10 7 copies/mL, and an inter-assay SD of ≤0.30 log copies/mL for all tested levels within this range. With clinical samples, the correlation coefficient (r value) was 0.896 between DBS-fingerprick and plasma and 0.901 between DBS-venous and plasma, and the bias was -0.07 log copies/mL between DBS-fingerprick and plasma and -0.02 log copies/mL between DBS-venous and plasma. The sensitivity of DBS-fingerprick and DBS-venous was 93%, while the specificity of both DBS methods was 95%. The results demonstrated that the Abbott RealTime HIV-1 assay with DBS sample protocol is highly sensitive, specific and precise across a wide dynamic range and correlates well with plasma values. The Abbott RealTime HIV-1 assay with DBS sample protocol provides an
Hinić, Vladimira; Feuz, Kinga; Turan, Selda; Berini, Andrea; Frei, Reno; Pfeifer, Karin; Goldenberger, Daniel
Rapid and reliable diagnosis is crucial for correct management of tuberculosis. The Abbott RealTime MTB Assay represents a novel qualitative real-time PCR assay for direct detection of M. tuberculosis-complex (MTB) DNA from respiratory samples. The test targets two highly conserved sequences, the multi-copy insertion element IS6110 and the protein antigen B (PAB) gene of MTB, allowing even the detection of IS6610-deficient strains. We evaluated this commercial diagnostic test by analyzing 200 respiratory and, for the first time, 87 non-respiratory clinical specimens from our tertiary care institution and compared its results to our IS6110-based in-house real-time PCR for MTB as well as MTB culture. Overall sensitivity for Abbott RealTime MTB was 100% (19/19) in smear positive and 87.5% (7/8) in smear negative specimens, while the specificity of the assay was 100% (260/260). For both non-respiratory smear positive and smear negative specimens Abbott RealTime MTB tests showed 100% (8/8) sensitivity and 100% (8/8) specificity. Cycle threshold (Ct) value analysis of 16 MTB positive samples showed a slightly higher Ct value of the Abbott RealTime MTB test compared to our in-house MTB assay (mean delta Ct = 2.55). In conclusion, the performance of the new Abbott RealTime MTB Assay was highly similar to culture and in-house MTB PCR. We document successful analysis of 87 non-respiratory samples with the highly automated Abbott RealTime MTB test with no inhibition observed. Copyright © 2017 Elsevier Ltd. All rights reserved.
Çuhadar, Serap; Köseoğlu, Mehmet; Çinpolat, Yasemin; Buğdaycı, Güler; Usta, Murat; Semerci, Tuna
Extremely high glucose concentrations have been shown to interfere with creatinine assays especially with Jaffe method in peritoneal dialysate. Because diabetes is the fastest growing chronic disease in the world, laboratories study with varying glucose concentrations. We investigated whether different levels of glucose spiked in serum interfere with 21 routine chemistry and thyroid assays at glucose concentrations between 17-51 mmol/L. Baseline (group I) serum pool with glucose concentration of 5.55 (5.44-5.61) mmol/L was prepared from patient sera. Spiking with 20% dextrose solution, sample groups were obtained with glucose concentrations: 17.09, 34.52, and 50.95 mmol/L (group II, III, IV, respectively). Total of 21 biochemistry analytes and thyroid tests were studied on Abbott c8000 and i2000sr with commercial reagents. Bias from baseline value was checked statistically and clinically. Creatinine increased significantly by 8.74%, 31.66%, 55.31% at groups II, III, IV, respectively with P values of < 0.001. At the median glucose concentration of 50.95 mmol/L, calcium, albumin, chloride and FT4 biased significantly clinically (-0.85%, 1.63%, 0.65%, 7.4% with P values 0.138, 0.214, 0.004, < 0.001, respectively). Remaining assays were free of interference. Among the numerous biochemical parameters studied, only a few parameters are affected by dramatically increased glucose concentration. The creatinine measurements obtained in human sera with the Jaffe alkaline method at high glucose concentrations should be interpreted with caution. Other tests that were affected with extremely high glucose concentrations were calcium, albumin, chloride and FT4, hence results should be taken into consideration in patients with poor diabetic control.
Poljak, Mario; Ostrbenk, Anja; Seme, Katja; Ucakar, Veronika; Hillemanns, Peter; Bokal, Eda Vrtacnik; Jancar, Nina; Klavs, Irena
The clinical performance of the Abbott RealTime High Risk HPV (human papillomavirus) test (RealTime) and that of the Hybrid Capture 2 HPV DNA test (hc2) were prospectively compared in the population-based cervical cancer screening setting. In women >30 years old (n = 3,129), the clinical sensitivity of RealTime for detection of cervical intraepithelial neoplasia of grade 2 (CIN2) or worse (38 cases) and its clinical specificity for lesions of less than CIN2 (3,091 controls) were 100% and 93.3%, respectively, and those of hc2 were 97.4% and 91.8%, respectively. A noninferiority score test showed that the clinical specificity (P laboratories. RealTime can be considered to be a reliable and robust HPV assay clinically comparable to hc2 for the detection of CIN2+ lesions in a population-based cervical cancer screening setting.
Guichet, Emilande; Aghokeng, Avelin; Eymard-Duvernay, Sabrina; Vidal, Nicole; Ayouba, Ahidjo; Mpoudi Ngole, Eitel; Delaporte, Eric; Ciaffi, Laura; Peeters, Martine
With the increasing demand of HIV viral load (VL) tests in resource-limited countries (RLCs) there is a need for assays at affordable cost and able to quantify all known HIV-1 variants. VLs obtained with a recently developed open and polyvalent universal HIV-1/SIVcpz/SIVgor RT-qPCR were compared to Abbott RealTime HIV-1 assay in Cameroon. On 474 plasma samples, characterized by a wide range of VLs and a broad HIV-1 group M genetic diversity, 97.5% concordance was observed when using the lower detection limit of each assay. When using the threshold of 3.00 log 10 copies/mL, according to WHO guidelines to define virological failure (VF) in RLCs, the concordance was 94.7%, 360/474 versus 339/474 patients were identified with VF with the new assay and Abbott RealTime HIV-1, respectively. Higher VLs were measured with the new assay, +0.47 log 10 copies/mL (95% CI; 0.42-0.52) as shown with Bland-Altman analysis. Eleven samples from patients on VF with drug resistance were not detected by Abbott RealTime HIV-1 versus two only with the new assay. Overall, our study showed that the new assay can be easily implemented in a laboratory in RLCs with VL experience and showed good performance on a wide diversity of HIV-1 group M variants. Copyright © 2016 Elsevier B.V. All rights reserved.
Schutten, Martin; Fries, E; Burghoorn-Maas, C; Niesters, H G M
BACKGROUND: Despite FDA approval and CE marking of commercial tests, manufacturer independent testing of technical aspects is important. OBJECTIVES: To evaluate the analytical performance of the new Abbott RealTime HCV and HIV-1 viral load tests. STUDY DESIGN: Sensitivity, specificity and
Antonucci, FrancescoPaolo; Cento, Valeria; Sorbo, Maria Chiara; Manuelli, Matteo Ciancio; Lenci, Ilaria; Sforza, Daniele; Di Carlo, Domenico; Milana, Martina; Manzia, Tommaso Maria; Angelico, Mario; Tisone, Giuseppe; Perno, Carlo Federico; Ceccherini-Silberstein, Francesca
We evaluated the performance of a rapid method to quantify HCV-RNA in the hepatic and extrahepatic compartments, by using for the first time the Abbott RealTime HCV-assay. Non-tumoral (NT), tumoral (TT) liver samples, lymph nodes and ascitic fluid from patients undergoing orthotopic-liver-transplantation (N=18) or liver resection (N=4) were used for the HCV-RNA quantification; 5/22 patients were tested after or during direct acting antivirals (DAA) treatment. Total RNA and DNA quantification from tissue-biopsies allowed normalization of HCV-RNA concentrations in IU/μg of total RNA and IU/10 6 liver-cells, respectively. HCV-RNA was successfully quantified with high reliability in liver biopsies, lymph nodes and ascitic fluid samples. Among the 17 untreated patients, a positive and significant HCV-RNA correlation between serum and NT liver-samples was observed (Pearson: rho=0.544, p=0.024). Three DAA-treated patients were HCV-RNA "undetectable" in serum, but still "detectable" in all tested liver-tissues. Differently, only one DAA-treated patient, tested after sustained-virological-response, showed HCV-RNA "undetectability" in liver-tissue. HCV-RNA was successfully quantified with high reliability in liver bioptic samples and extrahepatic compartments, even when HCV-RNA was "undetectable" in serum. Abbott RealTime HCV-assay is a good diagnostic tool for HCV quantification in intra- and extra-hepatic compartments, whenever a bioptic sample is available. Copyright © 2017 Elsevier B.V. All rights reserved.
Comparison of the clinical performances of the AdvanSure HPV Screening Real-Time PCR, the Abbott Real-Time High-Risk HPV Test, and the Hybrid Capture High-Risk HPV DNA Test for Cervical Cancer Screening.
Chung, Hae-Sun; Hahm, Chorong; Lee, Miae
The clinical performance of three human papillomavirus (HPV) DNA commercial assays for cervical cancer screening was evaluated; the AdvanSure HPV Screening Real-Time PCR (AdvanSure PCR; LG Life Sciences) that was developed recently for the detection of both high-risk and low-risk genotypes, the Abbott RealTime High-Risk HPV Test (Abbott PCR; Abbott Molecular) and the Hybrid Capture High-Risk HPV DNA test (HC2; Qiagen). The three different HPV DNA tests were compared using cytology samples obtained from 619 women who underwent routine cervical cancer screening. The gold-standard assay was histopathological confirmation of cervical intraepithelial neoplasia of grade 2 or worse. The clinical sensitivities of the AdvanSure PCR, the Abbott PCR and the HC2 for the detection of cervical intraepithelial neoplasia of grade 2 or worse were 95.5%, 95.5% and 100%, respectively, while the clinical specificities were 61.6%, 86.4% and 83.3%, respectively. There were no significant differences in the clinical sensitivities of the Abbott PCR and the AdvanSure PCR compared to the HC2. The clinical specificities of the Abbott PCR and the AdvanSure PCR for the detection of HPV types 16/18 were 97.8% and 98.5%, respectively. For cervical cancer screening, all three tests showed relatively good clinical sensitivities, but the AdvanSure PCR had lower clinical specificity than the Abbott PCR and the HC2. The AdvanSure PCR and the Abbott PCR assays have the advantage of being automated and the ability to distinguish between HPV types 16/18 and other HPV types. The two real-time PCR assays could be useful tools in HPV testing for cervical cancer screening. Copyright © 2014 Elsevier B.V. All rights reserved.
Sarrazin, Christoph; Dierynck, Inge; Cloherty, Gavin; Ghys, Anne; Janssen, Katrien; Luo, Donghan; Witek, James; Buti, Maria; Picchio, Gaston; De Meyer, Sandra
Protease inhibitor (PI)-based response-guided triple therapies for hepatitis C virus (HCV) infection are still widely used. Noncirrhotic treatment-naive and prior relapser patients receiving telaprevir-based treatment are eligible for shorter, 24-week total therapy if HCV RNA is undetectable at both weeks 4 and 12. In this study, the concordance in HCV RNA assessments between the Roche High Pure System/Cobas TaqMan and Abbott RealTime HCV RNA assays and the impacts of different HCV RNA cutoffs on treatment outcome were evaluated. A total of 2,629 samples from 663 HCV genotype 1 patients receiving telaprevir/pegylated interferon/ribavirin in OPTIMIZE were analyzed using the High Pure System and reanalyzed using Abbott RealTime (limits of detection, 15.1 IU/ml versus 8.3 IU/ml; limits of quantification, 25 IU/ml versus 12 IU/ml, respectively). Overall, good concordance was observed between the assays. Using undetectable HCV RNA at week 4, 34% of the patients would be eligible for shorter treatment duration with Abbott RealTime versus 72% with the High Pure System. However, using Abbott RealTime, a similar proportion (74%) would be eligible. Of the patients receiving 24-week total therapy, 87% achieved a sustained virologic response with undetectable HCV RNA by the High Pure System or Abbott RealTime; however, 92% of the patients with undetectable HCV RNA by Abbott RealTime achieved a sustained virologic response. Using undetectable HCV RNA as the cutoff, the more sensitive Abbott RealTime assay would identify fewer patients eligible for shorter treatment than the High Pure System. Our data confirm the Abbott RealTime assay, to determine eligibility for shortened PI-based HCV treatment. (The study was registered with ClinicalTrials.gov under registration no. NCT01241760.). Copyright © 2015, American Society for Microbiology. All Rights Reserved.
Brokenshire, M K; Say, P J; van Vonno, A H; Wong, C
OBJECTIVE: To evaluate the commercial microparticle enzyme immunoassay (MEIA), Abbott IMx Select Chlamydia, for the detection of Chlamydia trachomatis in women and to compare its performance with endocervical cell culture. Also, to determine whether sampling the urethral site is an important part of chlamydial diagnosis in women. SETTING: The Auckland, Manukau, and Waitakere Sexual Health Clinics, Auckland, New Zealand and the Department of Clinical Microbiology, Auckland Hospital, Auckland, ...
Lavoie, S; Caswell, D; Gill, M J; Kadkhoda, K; Charlton, C L; Levett, P N; Hatchette, T; Garceau, R; Maregmen, J; Mazzulli, T; Needle, R; Kadivar, K; Kim, J
False-reactivity in HIV-negative specimens has been detected in HIV fourth-generation antigen/antibody or 'combo' assays which are able to detect both anti-HIV-1/HIV-2 antibodies and HIV-1 antigen. We sought to characterize these specimens and determine the effect of heterophilic interference. Specimens previously testing as false-reactive on the Abbott ARCHITECT HIV Ag/Ab combo assay and re-tested on a different (Siemens ADVIA Centaur HIV Ag/Ab) assay. A subset of these specimens were also pre-treated with heterophilic blocking agents and re-tested on the Abbott assay. Here we report that 95% (252/264) of clinical specimens that were repeatedly reactive on the Abbott ARCHITECT HIV Ag/Ab combo assay (S/Co range, 0.94-678) were negative when re-tested on a different fourth generation HIV combo assay (Siemens ADVIA Centaur HIV Ag/Ab). All 264 samples were subsequently confirmed to be HIV negative. On a small subset (57) of specimens with available volume, pre-treatment with two different reagents (HBT; Heterophilic Blocking Tube, NABT; Non-Specific Blocking Tube) designed to block heterophilic antibody interference either eliminated (HBT) or reduced (NABT) the false reactivity when re-tested on the ARCHITECT HIV Ag/Ab combo assay. Our results suggest that the Abbott ARCHITECT HIV Ag/Ab combo assay can be prone to heterophilic antibody interference. Crown Copyright © 2018. Published by Elsevier B.V. All rights reserved.
Fumi, M; Martins, D; Pancione, Y; Sale, S; Rocco, V
B-chronic lymphocytic leukemia CLL, a neoplastic clonal disorder with monomorphous small B lymphocytes with scanty cytoplasm and clumped chromatin, can be morphologically differentiated in typical and atypical forms with different prognosis: Smudge cells (Gumprecht's shadows) are one of the well-known features of the typical CLL and are much less inconsistent in other different types CLPD. Abbott Cell-Dyn Sapphire uses the fluorescence after staining with the DNA fluorochrome propidium iodide for the measurement of nucleated red blood cells (NRBCs) and nonviable cells (FL3+ cell fraction): We have studied the possible correlation between presence and number of morphologically identifiable smudge cells on smears and the percentage of nonviable cells produced by Cell-Dyn Sapphire. 305 blood samples from 224 patients with B-cell lymphoproliferative disorders and 40 healthy blood donors were analyzed by CBC performed by Cell-Dyn Sapphire, peripheral blood smear, and immunophenotype characterization. FL3+ fraction in CLPD directly correlated with the percentage of smudge cells and is significantly increased in patients with typical B-CLL. This phenomenon is much less evident in patients with atypical/mixed B-CLL and B-NHL. In small laboratories without FCM and cytogenetic, smudge cells%, can be utilized as a preliminary diagnostic and prognostic tool in differential diagnosis of CLPD. © 2014 John Wiley & Sons Ltd.
Yin, Lianli; Tang, Yinghua; Chen, Xiang; Sun, Yifan
Follicle-stimulating hormone (FSH) and luteinizing hormone (LH) regulate the growth and reproductive activity of gonadal tissue and determine the concentration of LH is essential for the prediction of ovulation. Collectively, FSH and LH are important measurements to ascertain the causes of infertility as well as diagnosing disorders such as polycystic ovary syndrome and pituitary and gonadal dysfunction. This study compares the correlation between LH and FSH measurements during examination with two different systems, Architect i2000sr (Abbott Laboratories; Lake Bluff, IL, USA) and Cobas e601 (Roche; Geneva, Switzerland), and assesses the differences between these systems. Serum analysis was performed for 95 patients using both the Cobas e601 and Architect i2000sr systems. The method used to compare the systems was Passing-Bablok regression analysis with a Bland-Altman agreement plot. Inter-rater agreement was analyzed using a concordance correlation coefficient. Architect i2000sr and Cobas e601 have strong correlations in their LH and FSH results. However, the Bland-Altman plot shows that LH and FSH measurements in Cobas e601 are about 1.31 times and 1.26 times higher than those in Architect i2000sr, respectively. Passing-Bablok regression analysis also shows significant proportional deviation between them. The difference between the test results for LH and FSH in Cobas e601 and Architect i2000sr indicate that the results from one system cannot be directly used to evaluate the other system.
Enraizamento de estacas caulinares de kiwi (Actinidia chinensis Planch cv Abbott tratadas com auxinas e boro Rooting of kiwi stem cuttings (Actinidia chinensis Planch. cv Abbott treated with auxins and boron
Full Text Available O trabalho teve como finalidade, estudar o efeito de várias auxinas sintéticas em formulações comerciais e do boro, sobre o enraizamento de estacas caulinares de kiwi (Actinidia chinensis Planch, cv Abbott.. As estacas utilizadas continham dois nós e duas folhas cortadas ao meio, com aproximadamente 10 cm de comprimento, onde o corte basal em bisel foi realizado logo abaixo de um nó e o apical acima do outro nó. O efeito das auxinas, sobre o enraizamento de estacas caulinares de kiwi foi verificado mediante os seguintes tratamentos, aplicados sobre as bases das estacas: T1 H(20; T2 (NAA 300 ppm; T3 (IBA 300 ppm; T4 (NAA 300 ppm + B; T5 (IBA 300 ppm + B; T6 (NAA 0,5%-pó e T7 (IBA 0,5%-pó. Após o tratamento das estacas, estas foram plantadas em bandejas de enraizamento, contendo vermiculita pura e colocadas em câmara de nebulização, onde permaneceram por 120 dias, até a sua coleta. Para a avaliação do efeito de auxinas e do ácido bórico, sobre o enraizamento de estacas caulinares de kiwi, foram realizadas as seguintes observações: 1. porcentagem de estacas enraizadas; 2. análise de açúcares redutores e açúcares totais (em g/100 g de matéria seca; 3. análise de triptofano (em µg/100 mg de matéria seca. Os resultados obtidos no processo de enraizamento de estacas caulinares de kiwi (Actinidia chinensis Planch. variedade Abbott, levou a concluir que o inverno e outono foram as melhores épocas de coleta dos ramos de auxinas para a confecção das estacas. O processo de enraizamento foi ainda incrementado com a aplicação exógena na base das estacas, sendo que o alto teor de açúcares redutores e totais beneficiou a maior porcentagem de enraizamento.This study was carried out to evaluate the effects of some synthetical auxins and boron trade formulations in the rooting of stem cuttings of some kiwi (Actinidia chinensis Planch varieties. The experiment was carried out in a misty nebulization chamber in the Botany
Performance comparison of the 4th generation Bio-Rad Laboratories GS HIV Combo Ag/Ab EIA on the EVOLIS™ automated system versus Abbott ARCHITECT HIV Ag/Ab Combo, Ortho Anti-HIV 1+2 EIA on Vitros ECi and Siemens HIV-1/O/2 enhanced on Advia Centaur.
Mitchell, Elizabeth O; Stewart, Greg; Bajzik, Olivier; Ferret, Mathieu; Bentsen, Christopher; Shriver, M Kathleen
A multisite study was conducted to evaluate the performance of the Bio-Rad 4th generation GS HIV Combo Ag/Ab EIA versus Abbott 4th generation ARCHITECT HIV Ag/Ab Combo. The performance of two 3rd generation EIAs, Ortho Diagnostics Anti-HIV 1+2 EIA and Siemens HIV 1/O/2 was also evaluated. Study objective was comparison of analytical HIV-1 p24 antigen detection, sensitivity in HIV-1 seroconversion panels, specificity in blood donors and two HIV false reactive panels. Analytical sensitivity was evaluated with International HIV-1 p24 antigen standards, the AFFSAPS (pg/mL) and WHO 90/636 (IU/mL) standards; sensitivity in acute infection was compared on 55 seroconversion samples, and specificity was evaluated on 1000 negative blood donors and two false reactive panels. GS HIV Combo Ag/Ab demonstrated better analytical HIV antigen sensitivity compared to ARCHITECT HIV Ag/Ab Combo: 0.41 IU/mL versus 1.2 IU/mL (WHO) and 12.7 pg/mL versus 20.1 pg/mL (AFSSAPS); GS HIV Combo Ag/Ab EIA also demonstrated slightly better specificity compared to ARCHITECT HIV Ag/Ab Combo (100% versus 99.7%). The 4th generation HIV Combo tests detected seroconversion 7-11 days earlier than the 3rd generation HIV antibody only EIAs. Both 4th generation immunoassays demonstrated excellent performance in sensitivity, with the reduction of the serological window period (7-11 days earlier detection than the 3rd generation HIV tests). However, GS HIV Combo Ag/Ab demonstrated improved HIV antigen analytical sensitivity and slightly better specificity when compared to ARCHITECT HIV Ag/Ab Combo assay, with higher positive predictive values (PPV) for low prevalence populations. Copyright © 2013 Elsevier B.V. All rights reserved.
Full Text Available Traditionally, gambling and problem gambling research relies on cross-sectional and retrospective designs. This has compromised identification of temporal relationships and causal inference. To overcome these problems a new questionnaire, the Jonsson-Abbott Scale (JAS, was developed and used in a large, prospective, general population study, The Swedish Longitudinal Gambling Study (Swelogs. The JAS has 11 items and seeks to identify early indicators, examine relationships between indicators and assess their capacity to predict future problem progression. The aims of the study were to examine psychometric properties of the JAS (internal consistency and dimensionality and predictive validity with respect to increased gambling risk and problem gambling onset. The results are based on repeated interviews with 3818 participants. The response rate from the initial baseline wave was 74%. The original sample consisted of a random, stratified selection from the Swedish population register aged between 16 and 84. The results indicate an acceptable fit of a three-factor solution in a confirmatory factor analysis with ‘Over consumption,’ ‘Gambling fallacies,’ and ‘Reinforcers’ as factors. Reinforcers, Over consumption and Gambling fallacies were significant predictors of gambling risk potential and Gambling fallacies and Over consumption were significant predictors of problem gambling onset (incident cases at 12 month follow up. When controlled for risk potential measured at baseline, the predictor Over consumption was not significant for gambling risk potential at follow up. For incident cases, Gambling fallacies and Over consumption remained significant when controlled for risk potential. Implications of the results for the development of problem gambling, early detection, prevention, and future research are discussed.
Chien, Tzu-I; Chen, Hui-Hou; Kao, Jau-Tsuen
B-type natriuretic peptide (BNP) and N-terminal pro-brain natriuretic peptide (NT-proBNP) are small cardiac hormones released from the heart. They can be used as an important aid to diagnose congestive heart failure (CHF). We compared the performances of the Abbott AxSYM and Roche Elecsys 2010 for the measurement of BNP and NT-proBNP. The first method uses a microparticle enzyme-linked immunoassay, whereas the other uses chemiluminescent immunometric assay. The CVs using pooled sera ranged from 3.7% to 12.7% for the AxSYM and 0.9% to 2.2% for the Elecsys 2010. The Passing and Bablok regression was Elecsys 2010 NT-proBNP=7.23xAxSYM BNP+2.53. The BNP in EDTA plasma was more stable than in serum. The immunoreactivity difference of NT-proBNP in serum or EDTA plasma was within 10% when stored at 4 degrees Celsius or 25 degrees Celsius for 72 h. Receiver operating characteristic (ROC) curves were different for both assays, and the areas under the curves were 0.704 and 0.841 for the AxSYM and Elecsys 2010 method, respectively. Both assays were not entirely specific for heart failure. The precision and stability for NT-proBNP was better than for BNP in serum. It is important to use method-appropriate reference ranges (or cutoff) for the BNP and NT-proBNP, respectively, in the assessment of CHF.
Jonsson, Jakob; Abbott, Max W; Sjöberg, Anders; Carlbring, Per
Traditionally, gambling and problem gambling research relies on cross-sectional and retrospective designs. This has compromised identification of temporal relationships and causal inference. To overcome these problems a new questionnaire, the Jonsson-Abbott Scale (JAS), was developed and used in a large, prospective, general population study, The Swedish Longitudinal Gambling Study (Swelogs). The JAS has 11 items and seeks to identify early indicators, examine relationships between indicators and assess their capacity to predict future problem progression. The aims of the study were to examine psychometric properties of the JAS (internal consistency and dimensionality) and predictive validity with respect to increased gambling risk and problem gambling onset. The results are based on repeated interviews with 3818 participants. The response rate from the initial baseline wave was 74%. The original sample consisted of a random, stratified selection from the Swedish population register aged between 16 and 84. The results indicate an acceptable fit of a three-factor solution in a confirmatory factor analysis with 'Over consumption,' 'Gambling fallacies,' and 'Reinforcers' as factors. Reinforcers, Over consumption and Gambling fallacies were significant predictors of gambling risk potential and Gambling fallacies and Over consumption were significant predictors of problem gambling onset (incident cases) at 12 month follow up. When controlled for risk potential measured at baseline, the predictor Over consumption was not significant for gambling risk potential at follow up. For incident cases, Gambling fallacies and Over consumption remained significant when controlled for risk potential. Implications of the results for the development of problem gambling, early detection, prevention, and future research are discussed.
Analytical and clinical performance characteristics of the Abbott RealTime MTB RIF/INH Resistance, an assay for the detection of rifampicin and isoniazid resistant Mycobacterium tuberculosis in pulmonary specimens.
Kostera, Joshua; Leckie, Gregor; Tang, Ning; Lampinen, John; Szostak, Magdalena; Abravaya, Klara; Wang, Hong
Clinical management of drug-resistant tuberculosis patients continues to present significant challenges to global health. To tackle these challenges, the Abbott RealTime MTB RIF/INH Resistance assay was developed to accelerate the diagnosis of rifampicin and/or isoniazid resistant tuberculosis to within a day. This article summarizes the performance of the Abbott RealTime MTB RIF/INH Resistance assay; including reliability, analytical sensitivity, and clinical sensitivity/specificity as compared to Cepheid GeneXpert MTB/RIF version 1.0 and Hain MTBDRplus version 2.0. The limit of detection (LOD) of the Abbott RealTime MTB RIF/INH Resistance assay was determined to be 32 colony forming units/milliliter (cfu/mL) using the Mycobacterium tuberculosis (MTB) strain H37Rv cell line. For rifampicin resistance detection, the Abbott RealTime MTB RIF/INH Resistance assay demonstrated statistically equivalent clinical sensitivity and specificity as compared to Cepheid GeneXpert MTB/RIF. For isoniazid resistance detection, the assay demonstrated statistically equivalent clinical sensitivity and specificity as compared to Hain MTBDRplus. The performance data presented herein demonstrate that the Abbott RealTime MTB RIF/INH Resistance assay is a sensitive, robust, and reliable test for realtime simultaneous detection of first line anti-tuberculosis antibiotics rifampicin and isoniazid in patient specimens. Copyright © 2016 The Author. Published by Elsevier Ltd.. All rights reserved.
Manlutac, Anna Liza M; Giesick, Jill S; McVay, Patricia A
HIV screening assays have gone through several generations of development in an effort to narrow the "window period" of detection. Utilizing a fourth generation HIV screening assay has the potential to detect earlier HIV infection, thus reducing HIV-1 transmission. To identify acute infections to decrease HIV transmission in San Diego County. Serum specimens were collected from clients seen by multiple submitters in San Diego County. All acceptable specimens were screened using the 4th Gen Combo Assay. Initially reactive specimens were repeated in duplicate and if repeatedly reactive, were confirmed by HIV-1 Immunofluorescent Antibody Assay (IFA). IFA negative/inconclusive specimens were sent for HIV-1 NAT and HIV-2 antibody testing to referral laboratories. BioRad Multispot HIV-1/HIV-2 Rapid Test was also performed on a subset of specimens. Of 14,559 specimens received in 20 months, 14,517 specimens were tested. Of the 14,517 specimens that were tested, a total of 279 (1.9%) specimens were CIA repeatedly reactive and 240 of the 279 confirmed by HIV-1 IFA. Thirty-nine gave IFA negative/inconclusive result and 30 were further tested for HIV-1 NAT and 36 for HIV-2 antibody. Thirteen specimens were considered false positives by CIA and 17 specimens were classified as acute infections. Eleven of 39 IFA negative/inconclusive specimens were further tested by Multispot. Five of the 11 were positive by Multispot. The fourth generation Abbott ARCHITECT HIV Ag/Ab Combo Assay identified 17 patients who may have been missed by the prior HIV-1 screening assay used at San Diego County Public Health Laboratory. Copyright © 2013 Elsevier B.V. All rights reserved.
Full Text Available Hepatitis C virus (HCV RNA measurement before, during and after antiviral therapy has become an essential tool in the management of interferon-based treatment of HCV-related infections. Conventional Polymerase Chain Reaction (PCR has been largely used to obtain quantitative data, but laborious, time-consuming post-PCR handling steps are required to gain valuable results. Real time (RT PCR now provides advantages over end-point (EP PCR due to its improved rapidity, sensitivity, reproducibility and the reduced risk of carry-over contamination, and has now proven itself to be valuable for the more precise monitoring of viral load kinetics and assessing antiviral response.The Abbott Real-Time HCV-RNA is a recently introduced assay for the automated processing of clinical samples and HCV-RNA quantitation: its basic technology relies on use of fluorescent linear probes (dynamic range using 0.5 ml as input target= 12-108 IU/mL and a hybridization/detection step at low temperature (35°C, which allows target mismatches to be tolerated. To determine the clinical application of the Abbott Real-Time assay and defining its correlation with the Bayer Versant bDNA v.3 assay, 68 consecutive samples from unselected HCV-infected patients were retrospectively analysed with RT and the results obtained using the two tests compared.A good correlation was found between RT-PCR and bDNA: 97% of samples tested had a result within a 0.5 log HCV IU/mL difference (bias=0.15 log, whereas 6 samples negative with bDNA gave positive results with Abbott RT (range, 1.89-3.07 log IU/mL and “in-house” qualitative RT-PCR assays.
Amendola, Alessandra; Bloisi, Maria; Marsella, Patrizia; Sabatini, Rosella; Bibbò, Angela; Angeletti, Claudio; Capobianchi, Maria Rosaria
Numerous studies investigating clinical significance of HIV-1 minimal residual viremia (MRV) suggest potential utility of assays more sensitive than those routinely used to monitor viral suppression. However currently available methods, based on different technologies, show great variation in detection limit and input plasma volume, and generally suffer from lack of standardization. In order to establish new tools suitable for routine quantification of minimal residual viremia in patients under virological suppression, some modifications were introduced into standard procedure of the Abbott RealTime HIV-1 assay leading to a "modified" and an "ultrasensitive" protocols. The following modifications were introduced: calibration curve extended towards low HIV-1 RNA concentration; 4 fold increased sample volume by concentrating starting material; reduced volume of internal control; adoption of "open-mode" software for quantification. Analytical performances were evaluated using the HIV-1 RNA Working Reagent 1 for NAT assays (NIBSC). Both tests were applied to clinical samples from virologically suppressed patients. The "modified" and the "ultrasensitive" configurations of the assay reached a limit of detection of 18.8 (95% CI: 11.1-51.0 cp/mL) and 4.8 cp/mL (95% CI: 2.6-9.1 cp/mL), respectively, with high precision and accuracy. In clinical samples from virologically suppressed patients, "modified" and "ultrasensitive" protocols allowed to detect and quantify HIV RNA in 12.7% and 46.6%, respectively, of samples resulted "not-detectable", and in 70.0% and 69.5%, respectively, of samples "detected laboratories for measuring MRV. Copyright © 2011 Elsevier B.V. All rights reserved.
Coyne, S; Lacour, B; Hennequin-Le Meur, C
In order to build a continuous quality improvement approach for control of glucose meters in clinical divisions at Necker-Enfants Malades hospital, the analytical performances (precision and accuracy) of 2 glucose meters have been evaluated in our laboratory according to SFBC recommendations. Fifty-six heparinized whole blood specimens from patients and thirty-nine from healthy volunteers were analyzed on each of the two meters and compared to plasma glucose measurement on the Roche Hitachi 917 system. The correlation coefficient was 0.938 for Optium Xceed and 0.911 for One Touch Ultra. However, 14.7% and 18.9% of the results (n = 95) for respectively Optium Xceed and One Touch Ultra were discordant, i.e. higher than a 20% difference compared to reference blood glucose concentrations. Inaccuracy was more important for low glucose concentrations (glucose concentrations. Capillary blood glucose concentrations must therefore be interpreted with caution concerning the diagnosis of hypoglycemia and treatment of unstable patients. Moreover, quality control of glucose meters (blood glucose determinations concurrently at bedside and in the laboratory) is difficult to perform. It also raises questions about the responsibility of "point-of-care testing", an area still subject to discussion.
Krintus, Magdalena; Kozinski, Marek; Boudry, Pascal; Capell, Nuria Estañ; Köller, Ursula; Lackner, Karl; Lefèvre, Guillaume; Lennartz, Lieselotte; Lotz, Johannes; Herranz, Antonio Mora; Nybo, Mads; Plebani, Mario; Sandberg, Maria B; Schratzberger, Wolfgang; Shih, Jessie; Skadberg, Øyvind; Chargui, Ahmed Taoufik; Zaninotto, Martina; Sypniewska, Grazyna
International recommendations highlight the superior value of cardiac troponins (cTns) for early diagnosis of myocardial infarction along with analytical requirements of improved precision and detectability. In this multicenter study, we investigated the analytical performance of a new high sensitive cardiac troponin I (hs-cTnI) assay and its 99th percentile upper reference limit (URL). Laboratories from nine European countries evaluated the ARCHITECT STAT high sensitive troponin I (hs-TnI) immunoassay on the ARCHITECT i2000SR/i1000SR immunoanalyzers. Imprecision, limit of blank (LoB), limit of detection (LoD), limit of quantitation (LoQ) linearity of dilution, interferences, sample type, method comparisons, and 99th percentile URLs were evaluated in this study. Total imprecision of 3.3%-8.9%, 2.0%-3.5% and 1.5%-5.2% was determined for the low, medium and high controls, respectively. The lowest cTnI concentration corresponding to a total CV of 10% was 5.6 ng/L. Common interferences, sample dilution and carryover did not affect the hs-cTnI results. Slight, but statistically significant, differences with sample type were found. Concordance between the investigated hs-cTnI assay and contemporary cTnI assay at 99th percentile cut-off was found to be 95%. TnI was detectable in 75% and 57% of the apparently healthy population using the lower (1.1 ng/L) and upper (1.9 ng/L) limit of the LoD range provided by the ARCHITECT STAT hs-TnI package insert, respectively. The 99th percentile values were gender dependent. The new ARCHITECT STAT hs-TnI assay with improved analytical features meets the criteria of high sensitive Tn test and will be a valuable diagnostic tool.
Møller, Jens Kjølseth; Pedersen, Lisbeth Nørum; Persson, Kenneth
In an analytical methods comparison study on clinical samples, the Abbott RealTime CT new formulation assay (m2000 real-time PCR) consisting of a duplex PCR targeting different parts of the cryptic plasmid in Chlamydia trachomatis was compared with version 2 of the Roche COBAS(R) TaqMan(R) CT ass...
'Redemption between politics and ontology: Agamben on the coming politics' [Review] Abbott, M (2014) The figure of this world: Agamben and the question of political ontology; Whyte, J (2014) Catastrophe and redemption: the political thought of Giorgio Agamben
Review of: \\ud \\ud Abbott, M. (2014) The figure of this world: Agamben and the question of political ontology. Edinburgh: Edinburgh University Press, 9780748684090\\ud \\ud Whyte, J. (2014) Catastrophe and redemption: the political thought of Giorgio Agamben. Albany, NY: SUNY Press, 9781438448527
Gustavo Malagón Londoño
Full Text Available
Una vez más cumplimos uno de los objetivos primordiales de la Academia, cual es: estimular la investigación entre los profesionales de la salud residentes en Colombia y premiar la excelencia de quienes se hayan destacado con trabajos originales e inéditos en los campos de las ciencias clínicas y de las áreas básicas. A la convocatoria abierta para el concurso fue sorprendente y grato para esta Academia recibir una respuesta positiva de más de 100 profesionales estudiosos y consagrados a la investigación quienes por la importancia de los temas y el manejo metodológico, serio y responsable científicamente, llevaron al jurado calificador a una difícil selección de los mejores, para lo cual la preselección inicial de los 50, propició la más cuidadosa y ardua labor, habida cuenta de la sobresaliente calidad de los trabajos presentados.
Esto condujo a una primera conclusión: la del progreso en las características de excelencia de los investigadores colombianos y luego a la de la seriedad con que respondieron a la convocatoria. Por lo mismo la labor del Coordinador del concurso y de todo el equipo de reconocidos profesionales que integraron el grupo calificador resultó ardua y en extremo difícil; solo el altruismo, la consagración y la experiencia del Académico Hernando Groot Liévano y la seriedad, el conocimiento y dedicación del grupo de expertos que lo secundó para la evaluación cualitativa, permitió que los trabajos presentados fueran analizados exhaustivamente, para culminar el proceso de selección y cumplir el cronograma dispendioso para llegar a esta fecha solemne de entrega de los premios.
Debo reconocer con gratitud a nombre de la Academia no solo la ejemplar dedicación del coordinador y los examinadores, sino la colaboración de los Laboratorios Abbott; estos demostraron laudable compromiso con las disciplinas del conocimiento, lo cual explica el apoyo solidario
Kish, Laszlo B.; Gingl, Zoltan; Mingesz, Robert; Vadai, Gergely; Smulko, Janusz; Granqvist, Claes-Göran
A recent paper by Gunn-Allison-Abbott (GAA) [L. J. Gunn et al., Scientific Reports 4 (2014) 6461] argued that the Kirchhoff-law-Johnson-noise (KLJN) secure key exchange system could experience a severe information leak. Here we refute their results and demonstrate that GAA's arguments ensue from a serious design flaw in their system. Specifically, an attenuator broke the single Kirchhoff-loop into two coupled loops, which is an incorrect operation since the single loop is essential for the security in the KLJN system, and hence GAA's asserted information leak is trivial. Another consequence is that a fully defended KLJN system would not be able to function due to its built-in current-comparison defense against active (invasive) attacks. In this paper we crack GAA's scheme via an elementary current-comparison attack which yields negligible error probability for Eve even without averaging over the correlation time of the noise.
Nikolac Gabaj, Nora; Miler, Marijana; Vrtarić, Alen; Hemar, Marina; Filipi, Petra; Kocijančić, Marija; Šupak Smolčić, Vesna; Ćelap, Ivana; Šimundić, Ana-Maria
The aim of our study was to perform verification of serum indices on three clinical chemistry platforms. This study was done on three analyzers: Abbott Architect c8000, Beckman Coulter AU5800 (BC) and Roche Cobas 6000 c501. The following analytical specifications were verified: precision (two patient samples), accuracy (sample with the highest concentration of interferent was serially diluted and measured values compared to theoretical values), comparability (120 patients samples) and cross reactivity (samples with increasing concentrations of interferent were divided in two aliquots and remaining interferents were added in each aliquot. Measurements were done before and after adding interferents). Best results for precision were obtained for the H index (0.72%-2.08%). Accuracy for the H index was acceptable for Cobas and BC, while on Architect, deviations in the high concentration range were observed (y=0.02 [0.01-0.07]+1.07 [1.06-1.08]x). All three analyzers showed acceptable results in evaluating accuracy of L index and unacceptable results for I index. The H index was comparable between BC and both, Architect (Cohen's κ [95% CI]=0.795 [0.692-0.898]) and Roche (Cohen's κ [95% CI]=0.825 [0.729-0.922]), while Roche and Architect were not comparable. The I index was not comparable between all analyzer combinations, while the L index was only comparable between Abbott and BC. Cross reactivity analysis mostly showed that serum indices measurement is affected when a combination of interferences is present. There is heterogeneity between analyzers in the hemolysis, icteria, lipemia (HIL) quality performance. Verification of serum indices in routine work is necessary to establish analytical specifications.
O' Brien, Kevin C. [University of Illinois, Champaign, IL (United States)
The work summarized in this report is the first step towards a project that will re-train and create jobs for personnel in the coal industry and continue regional economic development to benefit regions impacted by previous downturns. The larger project is aimed at capturing ~300 tons/day (272 metric tonnes/day) CO2 at a 90% capture rate from existing coal- fired boilers at the Abbott Power Plant on the campus of University of Illinois (UI). It will employ the Linde-BASF novel amine-based advanced CO2 capture technology, which has already shown the potential to be cost-effective, energy efficient and compact at the 0.5-1.5 MWe pilot scales. The overall objective of the project is to design and install a scaled-up system of nominal 15 MWe size, integrate it with the Abbott Power Plant flue gas, steam and other utility systems, and demonstrate the viability of continuous operation under realistic conditions with high efficiency and capacity. The project will also begin to build a workforce that understands how to operate and maintain the capture plants by including students from regional community colleges and universities in the operation and evaluation of the capture system. This project will also lay the groundwork for follow-on projects that pilot utilization of the captured CO2 from coal-fired power plants. The net impact will be to demonstrate a replicable means to (1) use a standardized procedure to evaluate power plants for their ability to be retrofitted with a pilot capture unit; (2) design and construct reliable capture systems based on the Linde-BASF technology; (3) operate and maintain these systems; (4) implement training programs with local community colleges and universities to establish a workforce to operate and maintain the systems; and (5) prepare to evaluate at the large pilot scale level various methods to utilize the resulting captured CO2. Towards the larger project goal, the UI-led team, together
Culture for detection of Neisseria gonorrhoeae (NG) is being replaced by molecular assays, but difficulties are observed with false positive and negatives results, especially for extragenital samples. This study evaluates the Abbott CT\\/NG Real-Time assay and a real-time porA pseudogene assay. Samples (n = 600) from a mixed prevalence Irish population include 164 male urines with corresponding urethral swabs, 58 endocervical swabs, 173 male pharyngeal swabs, 205 male rectal swabs, 36 NG clinical isolates and 26 commensal Neisseria species isolates. There was a 100% concordance between the Abbott CT\\/NG Real-Time and the porA assay. The positivity rate was 1.2%, 1.7%, 8.1% and 5.8% for FVU\\/urethral swabs, endocervical, pharyngeal and rectal swabs, respectively. These results were compared to culture and discrepancies were found with nine pharyngeal and three rectal swabs. Seven of the 12 discrepant positive samples were sequenced and were confirmed "true positives". The sensitivity and specificity of the molecular assays was 100%. The sensitivity of the culture-based testing was 100% for urogenital samples but 36% and 75% for pharyngeal and rectal swabs, respectively. The combined Abbott CT\\/NG and porA assays provide a valuable alternative to culture and also generate a significant increase in the diagnosis of pharyngeal and rectal NG infection.
The Abbott RealTime High Risk HPV test is a clinically validated human papillomavirus assay for triage in the referral population and use in primary cervical cancer screening in women 30 years and older: a review of validation studies.
Poljak, Mario; Oštrbenk, Anja
Human papillomavirus (HPV) testing has become an essential part of current clinical practice in the management of cervical cancer and precancerous lesions. We reviewed the most important validation studies of a next-generation real-time polymerase chain reaction-based assay, the RealTime High Risk HPV test (RealTime)(Abbott Molecular, Des Plaines, IL, USA), for triage in referral population settings and for use in primary cervical cancer screening in women 30 years and older published in peer-reviewed journals from 2009 to 2013. RealTime is designed to detect 14 high-risk HPV genotypes with concurrent distinction of HPV-16 and HPV-18 from 12 other HPV genotypes. The test was launched on the European market in January 2009 and is currently used in many laboratories worldwide for routine detection of HPV. We concisely reviewed validation studies of a next-generation real-time polymerase chain reaction (PCR)-based assay: the Abbott RealTime High Risk HPV test. Eight validation studies of RealTime in referral settings showed its consistently high absolute clinical sensitivity for both CIN2+ (range 88.3-100%) and CIN3+ (range 93.0-100%), as well as comparative clinical sensitivity relative to the currently most widely used HPV test: the Qiagen/Digene Hybrid Capture 2 HPV DNA Test (HC2). Due to the significantly different composition of the referral populations, RealTime absolute clinical specificity for CIN2+ and CIN3+ varied greatly across studies, but was comparable relative to HC2. Four validation studies of RealTime performance in cervical cancer screening settings showed its consistently high absolute clinical sensitivity for both CIN2+ and CIN3+, as well as comparative clinical sensitivity and specificity relative to HC2 and GP5+/6+ PCR. RealTime has been extensively evaluated in the last 4 years. RealTime can be considered clinically validated for triage in referral population settings and for use in primary cervical cancer screening in women 30 years and older.
François, Catherine; Segard, Christine; Bouvier, Maryline; Stefanski, Martine; Pannier, Christine; Zawadzki, Patricia; Roussel, Catherine; Hecquet, Denise; Duverlie, Gilles; Brochot, Etienne; Castelain, Sandrine
This study compared the performance of 3 automated immunoassays, Architect ® (Abbott), Immulite ® (Siemens) and Liaison ® (Diasorin), for Epstein-Barr virus (EBV) serology. Ninety-one serum samples collected in Amiens University Hospital were analyzed for the presence of Viral Capsid Antigen (VCA) IgG and IgM and Epstein-Barr Nuclear Antigen (EBNA) IgG. The agreement between the 3 assays was calculated for each marker individually and for determination of the EBV profile, based on interpretation of the combination of these 3 EBV markers. Although similar results were obtained with Architect ® and Liaison ® , several discordant results were observed with Immulite ® , particularly for EBNA IgG. A large number of EBNA IgG-positive results were observed, which interfered with interpretation of the EBV profile. In contrast, Immulite ® performed similarly to the 2 other assays for detection of VCA IgM. Copyright © 2017 Elsevier Inc. All rights reserved.
Full Text Available Recently, Gunn, Allison and Abbott (GAA [http://arxiv.org/pdf/1402.2709v2.pdf] proposed a new scheme to utilize electromagnetic waves for eavesdropping on the Kirchhoff-law-Johnson-noise (KLJN secure key distribution. We proved in a former paper [Fluct. Noise Lett. 13 (2014 1450016] that GAA’s mathematical model is unphysical. Here we analyze GAA’s cracking scheme and show that, in the case of a loss-free cable, it provides less eavesdropping information than in the earlier (Bergou-Scheuer-Yariv mean-square-based attack [Kish LB, Scheuer J, Phys. Lett. A 374:2140-2142 (2010], while it offers no information in the case of a lossy cable. We also investigate GAA’s claim to be experimentally capable of distinguishing—using statistics over a few correlation times only—the distributions of two Gaussian noises with a relative variance difference of less than 10-8. Normally such distinctions would require hundreds of millions of correlations times to be observable. We identify several potential experimental artifacts as results of poor KLJN design, which can lead to GAA’s assertions: deterministic currents due to spurious harmonic components caused by ground loops, DC offset, aliasing, non-Gaussian features including non-linearities and other non-idealities in generators, and the timederivative nature of GAA’s scheme which tends to enhance all of these artifacts.
Tamimi, Waleed; Albanyan, Esam; Altwaijri, Yasmin; Tamim, Hani; Alhussein, Fahad
Reference intervals for pubertal characteristics are influenced by genetic, geographic, dietary and socioeconomic factors. Therefore, the aim of this study was to establish age-specific reference intervals of glucose and lipid levels among local school children. This was cross-sectional study, conducted among Saudi school children. Fasting blood samples were collected from 2149 children, 1138 (53%) boys and 1011 (47%) girls, aged 6 to 18 years old. Samples were analyzed on the Architect c8000 Chemistry System (Abbott Diagnostics, USA) for glucose, cholesterol, triglycerides, HDL and LDL. Reference intervals were established by nonparametric methods between the 2.5th and 97.5th percentiles. Significant differences were observed between boys and girls for cholesterol and triglycerides levels in all age groups (P glucose levels except at age 12 to 13 years. Saudi children have comparable serum cholesterol levels than their Western counterparts. This may reflect changing dietary habits and increasing affluence in Saudi Arabia. Increased lipid screening is anticipated, and these reference intervals will aid in the early assessment of cardiovascular and diabetes risk in Saudi pediatric populations.
Quantitation of estrogen receptor in seventy-five specimens of breast cancer: comparison between an immunoassay (Abbott ER-EIA monoclonal) and a [3H]estradiol binding assay based on isoelectric focusing in polyacrylamide gel
Pousette, A.; Gustafsson, S.A.; Thoernblad, A.M.N.; Nordgren, A.; Saellstroem, J.Li.; Lindgren, A.; Sundelin, P.; Gustafsson, J.A.
Quantitation of estrogen receptor has been performed in cytosol prepared from 75 specimens of breast cancer tissue from patients who had not received hormonal therapy. The study was performed in order to compare an immunoassay (Abbott Laboratories, North Chicago, IL) with our currently used method for estrogen receptor analysis based on isoelectric focusing of [ 3 H]estradiol-receptor complex in polyacrylamide gels. Using linear regression analysis, a regression coefficient (slope) of 1.30 and a correlation coefficient of 0.75 were calculated. The differences in results between the two methods are probably partly explained by the fact that the ligand-based method only measures unoccupied receptor, whereas the immunoassay detects the total amount of receptor, resulting in generally slightly higher concentrations with the latter method. However, in five of 75 specimens the ligand-based method gave a considerably higher concentration of estrogen receptor. This was most probably explained by partial proteolysis resulting in the formation of receptor fragment(s), which was undetectable with the immunoassay but detectable with the ligand-based method. These observations underline the importance of careful handling of specimens during the whole immunoassay procedure
Zeh, Clement; Ndiege, Kenneth; Inzaule, Seth; Achieng, Rebecca; Williamson, John; Chih-Wei Chang, Joy; Ellenberger, Dennis; Nkengasong, John
Routine HIV viral load testing is not widely accessible in most resource-limited settings, including Kenya. To increase access to viral load testing, alternative sample types like dried blood spots (DBS), which overcome the logistic barriers associated with plasma separation and cold chain shipment need to be considered and evaluated. The current study evaluated matched dried blood spots (DBS) and dried plasma spots (DPS) against plasma using the Abbott M 2000 (Abbott) and Roche Cobas Ampliprep/Cobas TaqMan (CAP/CTM) quantitative viral load assays in western Kenya. Matched plasma DBS and DPS were obtained from 200 HIV-1 infected antiretroviral treatment (ART)-experienced patients attending patient support centers in Western Kenya. Standard quantitative assay performance parameters with accompanying 95% confidence intervals (CI) were assessed at the assays lower detection limit (400cps/ml for CAP/CTM and 550cps/ml for Abbott) using SAS version 9.2. Receiver operating curves (ROC) were further used to assess viral-load thresholds with best assay performance (reference assay CAP/CTM plasma). Using the Abbott test, the sensitivity and specificity, respectively, for DPS were (97.3%, [95%CI: 93.2-99.2] and 98.1% [95%CI: 89.7-100]) and those for DBS (93.9% [95%CI: 88.8-97.2] and 88.0% [95%CI: 82.2-92.4]). The correlation and agreement using paired plasma and DPS/DBS were strong, with r2 = 90.5 and rc = 68.1. The Bland-Altman relative percent change was 95.3 for DPS, (95%CI: 90.4-97.7) and 73.6 (95%CI: 51.6-86.5) for DBS. Using the CAP/CTM assay, the sensitivity for DBS was significantly higher compared to DPS (100.0% [95% CI: 97.6-100.0] vs. 94.7% [95%CI: 89.8-97.7]), while the specificity for DBS was lower: 4%, [95% CI: 0.4-13.7] compared to DPS: 94.0%, [95% CI: 83.5-98.7]. When compared under different clinical relevant thresholds, the accuracy for the Abbott assay was 95% at the 1000cps/ml cut-off with a sensitivity and specificity of 96.6% [95% CI 91.8-98.7] and 90
Bruegel, Mathias; Nagel, Dorothea; Funk, Manuela; Fuhrmann, Petra; Zander, Johannes; Teupser, Daniel
Various types of automated hematology analyzers are used in clinical laboratories. Here, we performed a side-by-side comparison of five current top of the range routine hematology analyzers in the setting of a university hospital central laboratory. Complete blood counts (CBC), differentials, reticulocyte and nucleated red blood cell (NRBC) counts of 349 patient samples, randomly taken out of routine diagnostics, were analyzed with Cell-Dyn Sapphire (Abbott), DxH 800 (Beckman Coulter), Advia 2120i (Siemens), XE-5000 and XN-2000 (Sysmex). Inter-instrument comparison of CBCs including reticulocyte and NRBC counts and investigation of flagging quality in relation to microscopy were performed with the complete set of samples. Inter-instrument comparison of five-part differential was performed using samples without atypical cells in blood smear (n=292). Automated five-part differentials and NRBCs were additionally compared with microscopy. The five analyzers showed a good concordance for basic blood count parameters. Correlations between instruments were less well for reticulocyte counts, NRBCs, and differentials. The poorest concordance for NRBCs with microscopy was observed for Advia 2120i (Kendall's τb=0.37). The highest flagging sensitivity for blasts was observed for XN-2000 (97% compared to 65%-76% for other analyzers), whereas overall specificity was comparable between different instruments. To the best of our knowledge, this is the most comprehensive side-by-side comparison of five current top of the range routine hematology analyzers. Variable analyzer quality and parameter specific limitations must be considered in defining laboratory algorithms in clinical practice.
Park, Seungman; Kang, Youjin; Kim, Dong Geun; Kim, Eui-Chong; Park, Sung Sup; Seong, Moon-Woo
The detection of high-risk (HR) HPV in cervical cancer screening is important for early diagnosis of cervical cancer or pre-cancerous lesions. We evaluated the analytical and clinical performances of 3 HR HPV assays in Gynecology patients. A total of 991 specimens were included in this study: 787 specimens for use with a Hybrid Capture 2 (HC2) and 204 specimens for a HPV DNA microarray (DNA Chip). All specimens were tested using an Abbott RealTime High Risk HPV assay (Real-time HR), PGMY PCR, and sequence analysis. Clinical sensitivities for severe abnormal cytology (severe than high-grade squamous intraepithelial lesion) were 81.8% for Real-time HR, 77.3% for HC2, and 66.7% for DNA Chip, and clinical sensitivities for severe abnormal histology (cervical intraepithelial neoplasia grade 2+) were 91.7% for HC2, 87.5% for Real-time HR, and 73.3% for DNA Chip. As compared to results of the sequence analysis, HC2, Real-time HR, and DNA Chip showed concordance rates of 94.3% (115/122), 90.0% (117/130), and 61.5% (16/26), respectively. The HC2 assay and Real-time HR assay showed comparable results to each other in both clinical and analytical performances, while the DNA Chip assay showed poor clinical and analytical performances. The Real-time HR assay can be a good alternative option for HR HPV testing with advantages of allowing full automation and simultaneous genotyping of HR types 16 and 18. Copyright © 2013 Elsevier Inc. All rights reserved.
Qin, Yu; Zhang, Hongyun; Marlowe, Natalia; Fei, Mandong; Yu, Judy; Lei, Xiaoqin; Yu, Lulu; Zhang, Jia; Cao, Di; Ma, Li; Chen, Wen
HIV+/AIDS women have an increased risk of developing into CIN and cervical cancer compared to the general population. Limited medical resource and the lack of AIDS relevant knowledge impair the coverage and efficiency of cervical cancer screening. To compare the clinical performance of self-collected dry storage medium (FTA Elute card) and physician-collected PreservCyt medium in detection of high risk human papillomavirus (HR HPV) among HIV-1 positive population. Three hundred HIV-1 positive women (aged 25-65) were recruited from Yunnan infectious hospital. Two cervicovaginal samples were collected from each participant: one was collected by the women themselves and applied on a FTA Elute card; the other one was collected by a physician and stored in PreservCyt solution. All the samples were tested for 14 HR HPV using Abbott RealTime High Risk HPV assay. Biopsies were taken for histological diagnosis if any abnormal impression was noticed under colposcopy. 291 (97.0%) of participants were eligible for this study. 101 (34.70%) participants were found HR HPV positive in both FTA card and PreservCyt samples, and 19 (6.53%) women were diagnosed as CIN2+. The HR HPV positive rate on samples collected by FTA Elute card and PreservCyt solution was 42.61% and 39.86%, respectively. The overall agreement was 87% (kappa=0.731) between FTA card and PreservCyt. The clinical sensitivity and specificity of FTA card and PreservCyt were 100%, 61.39% and 100%, 64.33%, respectively. In this study, FTA Elute card demonstrated a good performance on self-collected sample for HR HPV detection in HIV-1 positive population. For the women from low-resource area with HIV-1 infection, FTA Elute card could be an attractive sample collection method for cervical cancer screening. Copyright Â© 2016 Elsevier B.V. All rights reserved.
Worlock, A; Blair, D; Hunsicker, M; Le-Nguyen, T; Motta, C; Nguyen, C; Papachristou, E; Pham, J; Williams, A; Vi, M; Vinluan, B; Hatzakis, A
The Aptima HCV Quant Dx assay (Aptima assay) is a fully automated quantitative assay on the Panther® system. This assay is intended for confirmation of diagnosis and monitoring of HCV RNA in plasma and serum specimens. The purpose of the testing described in this paper was to evaluate the performance of the Aptima assay. The analytical sensitivity, analytical specificity, precision, and linearity of the Aptima assay were assessed. The performance of the Aptima assay was compared to two commercially available HCV assays; the Abbott RealTime HCV assay (Abbott assay, Abbott Labs Illinois, USA) and the Roche COBAS Ampliprep/COBAS Taqman HCV Quantitative Test v2.0 (Roche Assay, Roche Molecular Systems, Pleasanton CA, USA). The 95% Lower Limit of Detection (LoD) of the assay was determined from dilutions of the 2nd HCV WHO International Standard (NIBSC 96/798 genotype 1) and HCV positive clinical specimens in HCV negative human plasma and serum. Probit analysis was performed to generate the 95% predicted detection limits. The Lower Limit of Quantitation (LLoQ) was established for each genotype by diluting clinical specimens and the 2nd HCV WHO International Standard (NIBSC 96/798 genotype 1) in HCV negative human plasma and serum. Specificity was determined using 200 fresh and 536 frozen HCV RNA negative clinical specimens including 370 plasma specimens and 366 serum specimens. Linearity for genotypes 1 to 6 was established by diluting armored RNA or HCV positive clinical specimens in HCV negative serum or plasma from 8.08 log IU/mL to below 1 log IU/mL. Precision was tested using a 10 member panel made by diluting HCV positive clinical specimens or spiking armored RNA into HCV negative plasma and serum. A method comparison was conducted against the Abbott assay using 1058 clinical specimens and against the Roche assay using 608 clinical specimens from HCV infected patients. In addition, agreement between the Roche assay and the Aptima assay using specimens with low
Rapid virological response assessment by Abbott RealTime hepatitis C virus assay for predicting sustained virological responses in patients with hepatitis C virus genotype 1 treated with pegylated-interferon and ribavirin
Full Text Available The lower limits of virus detection of hepatitis C virus (HCV RNA detection assays are continuously improving. We aimed to assess the utility of more precise definition of 4th week viral load [rapid virological response (RVR] in predicting sustained virological response (SVR in HCV genotype 1 patients treated with pegylated-interferon (PEG-IFN and ribavirin. Clinical data of treatment-naïve HCV genotype 1 patients were retrospectively collected from 2009 to 2014. Patients were grouped according to 4th week viral load as follows: undetectable (n = 90 and detectable but not quantifiable (< 12 IU/mL, n = 27. All patients received PEG-IFNα-2a or -2b and ribavirin for 24 weeks. Serum HCV RNA levels were measured by Abbott RealTime (ART; Abbott Molecular, Abbott Park, IL, USA HCV assay. SVR was 95.5% and 63% in the undetectable group and < 12 IU/mL group of 4th week viral load, respectively. The between-group difference in SVR was significant (p < 0.001. We determined 4th week viral load was independently associated with SVR (odds ratio = 19.28; p = 0.002 and a good predictor of SVR [area under the curve (AUC = 0.775; p = 0.001]. ART HCV assays had a stronger SVR predictive value in HCV genotype 1 patients, indicating that only the undetectable group of 4th week viral load patients measured by ART HCV assay should be considered for shorter treatment time (24 weeks with PEG-IFN and ribavirin.
Full Text Available Abstract Background Point of care (POC glucose meters are routinely used as a screening tool for hypoglycemia in a neonatal setting. Glucose meters however, lack the same accuracy as laboratory instruments for glucose measurement. In this study we investigated potential reasons for this inaccuracy and established a cut off value for confirmatory testing. Methods In this prospective study, all patients in the neonatal intensive care unit who had a plasma glucose test ordered were eligible to participate. Demographic information, sample collection information (nine variables and a recent hematocrit value were recorded for each sample. Glucose measurements were taken at the bedside on the glucose meter (RN PCx as well as in the laboratory on both the glucose meter (LAB PCx and the laboratory analyzer (PG. Data were analyzed by simple and mixed-effects regression analysis and by analysis of a receiver operator characteristics (ROC curve. Results There were 475 samples analyzed from 132 patients. RN PCx values were higher than PG values (mean = 4.9%, while LAB PCx results were lower (mean = -5.2% than PG values. Only 31% of the difference between RN PCx – PG and 46% of the difference for LAB PCx – PG could be accounted for by the variables tested. The largest proportion of variance between PCx and PG measurements was explained by hematocrit (about 30% with a greater effect seen at glucose concentrations ≤4.0 mmol/L (≤72 mg/dL(48% and 40% for RN PCx and LAB PCx, respectively. The ROC analysis showed that for detection of all cases of hypoglycemia (PG Conclusion The large difference between glucose results obtained by PCx glucose meter compared to the laboratory analyzer can be explained in part by hematocrit and low glucose concentration. These results emphasize that the glucose meter is useful only as a screening device for neonatal hypoglycemia and that a screening cut off value must be established.
Balion, Cynthia; Grey, Vijaylaxmi; Ismaila, Afisi; Blatz, Susan; Seidlitz, Wendy
Abstract Background Point of care (POC) glucose meters are routinely used as a screening tool for hypoglycemia in a neonatal setting. Glucose meters however, lack the same accuracy as laboratory instruments for glucose measurement. In this study we investigated potential reasons for this inaccuracy and established a cut off value for confirmatory testing. Methods In this prospective study, all patients in the neonatal intensive care unit who had a plasma glucose test ordered were eligible to ...
?uhadar, Serap; K?seo?lu, Mehmet; ?inpolat, Yasemin; Bu?dayc?, G?ler; Usta, Murat; Semerci, Tuna
Abstract Introduction: Extremely high glucose concentrations have been shown to interfere with creatinine assays especially with Jaffe method in peritoneal dialysate. Because diabetes is the fastest growing chronic disease in the world, laboratories study with varying glucose concentrations. We investigated whether different levels of glucose spiked in serum interfere with 21 routine chemistry and thyroid assays at glucose concentrations between 17-51 mmol/L. Materials and methods: Base...
Balion, Cynthia; Grey, Vijaylaxmi; Ismaila, Afisi; Blatz, Susan; Seidlitz, Wendy
Point of care (POC) glucose meters are routinely used as a screening tool for hypoglycemia in a neonatal setting. Glucose meters however, lack the same accuracy as laboratory instruments for glucose measurement. In this study we investigated potential reasons for this inaccuracy and established a cut off value for confirmatory testing. In this prospective study, all patients in the neonatal intensive care unit who had a plasma glucose test ordered were eligible to participate. Demographic information, sample collection information (nine variables) and a recent hematocrit value were recorded for each sample. Glucose measurements were taken at the bedside on the glucose meter (RN PCx) as well as in the laboratory on both the glucose meter (LAB PCx) and the laboratory analyzer (PG). Data were analyzed by simple and mixed-effects regression analysis and by analysis of a receiver operator characteristics (ROC) curve. There were 475 samples analyzed from 132 patients. RN PCx values were higher than PG values (mean = 4.9%), while LAB PCx results were lower (mean = -5.2%) than PG values. Only 31% of the difference between RN PCx--PG and 46% of the difference for LAB PCx--PG could be accounted for by the variables tested. The largest proportion of variance between PCx and PG measurements was explained by hematocrit (about 30%) with a greater effect seen at glucose concentrations LAB PCx, respectively). The ROC analysis showed that for detection of all cases of hypoglycemia (PG < 2.6 mmol/L)(PG < 47 mg/dL) the PCx screening cut off value would need to be set at 3.8 mmol/L (68 mg/dL) requiring 20% of all samples to have confirmatory analysis by the laboratory method. The large difference between glucose results obtained by PCx glucose meter compared to the laboratory analyzer can be explained in part by hematocrit and low glucose concentration. These results emphasize that the glucose meter is useful only as a screening device for neonatal hypoglycemia and that a screening
Dupuy, Anne Marie; Hurstel, Rémy; Bargnoux, Anne Sophie; Badiou, Stéphanie; Cristol, Jean Paul
Rheumatoid factor (RF) consists of autoantibodies and because of its heterogeneity its determination is not easy. Currently, nephelometry and Elisa method are considered as reference methods. Due to consolidation, many laboratories have fully automated turbidimetric apparatus, and specific nephelemetric systems are not always available. In addition, nephelemetry is more accurate, but time consuming, expensive, and requires a specific device, resulting in a lower efficiency. Turbidimetry could be an attractive alternative. The turbidimetric RF test from Diagam meets the requirements of accuracy and precision for optimal clinical use, with an acceptable measuring range, and could be an alternative in the determination of RF, without the associated cost of a dedicated instrument, making consolidation and saving blood possible.
Reinsberg, J.; Jost, E.; Van der Ven, H.
Test performance of two automated progesterone assays available on the immunoassay analysers Abbott AxSYM and Technicon Immuno 1, respectively, was evaluated in comparison with the radioimmunoassay Progesterone MAIA. For assessment of test performance imprecision, functional sensitivity and linearity of dilution was examined. Correlation with the manual radioimmunoassay was assessed using 122 serum samples over the range 0-110 nmol/L. Imprecision studies revealed for the AxSYM Progesterone within-run CV's of 1.8-6.4% and day-to-day CV's of 3.5-9.7% (concentration range 2.3-75 nmol/L); Immuno 1 Progesterone: within-run CV's 1.0-7.3%, day-to-day CV's 2.3-7.7% (concentration range 1.2-60 nmol/L). The functional sensitivity was <1.7 nmol/L for the AxSYM Progesterone and <1.1 nmol/L for the Immuno 1 Progesterone. With the AxSYM Progesterone the mean recovery after dilution from five samples was 102% (89-107%), from one sample only 69-80% was recovered; with the Immuno 1 Progesterone the mean recovery was 95% (80-105%). Despite of a quite good overall correlation (coefficients 0.972 and 0.981) the relationship of both assays to the Progesterone MAIA significantly deviate from linearity with a considerably higher slope within the lower concentration range. The relationship between the automated assays was linear over the entire concentration range (Immuno = 1.207 * AxSYM + 1; r = 0.986). The time to first result was 20 min for the AxSYM Progesterone, 45 min for the Immuno 1 Progesterone and 90 min for the Progesterone MAIA. The evaluated progesterone assays both exhibit an excellent precision and a high degree of sensitivity. They offer a rapid and flexible method for progesterone determination which may be especially useful for the monitoring of ovarian stimulation during in-vitro fertilization. (author)
Donovan, Laura Fasbach
When New Jersey set out to level the educational playing field between the state's poorest and wealthiest school districts a decade ago, one of the mission's most important components was ensuring teacher quality met the same high standards regardless of zip code. Research shows, a teacher's own education and training in early childhood education…
Human papillomavirus detection using the Abbott RealTime high-risk HPV tests compared with conventional nested PCR coupled to high-throughput sequencing of amplification products in cervical smear specimens from a Gabonese female population.
Moussavou-Boundzanga, Pamela; Koumakpayi, Ismaël Hervé; Labouba, Ingrid; Leroy, Eric M; Belembaogo, Ernest; Berthet, Nicolas
Cervical cancer is the fourth most common malignancy in women worldwide. However, screening with human papillomavirus (HPV) molecular tests holds promise for reducing cervical cancer incidence and mortality in low- and middle-income countries. The performance of the Abbott RealTime High-Risk HPV test (AbRT) was evaluated in 83 cervical smear specimens and compared with a conventional nested PCR coupled to high-throughput sequencing (HTS) to identify the amplicons. The AbRT assay detected at least one HPV genotype in 44.57% of women regardless of the grade of cervical abnormalities. Except for one case, good concordance was observed for the genotypes detected with the AbRT assay in the high-risk HPV category determined with HTS of the amplicon generated by conventional nested PCR. The AbRT test is an easy and reliable molecular tool and was as sensitive as conventional nested PCR in cervical smear specimens for detection HPVs associated with high-grade lesions. Moreover, sequencing amplicons using an HTS approach effectively identified the genotype of the hrHPV identified with the AbRT test.
CLSI-based transference of the CALIPER database of pediatric reference intervals from Abbott to Beckman, Ortho, Roche and Siemens Clinical Chemistry Assays: direct validation using reference samples from the CALIPER cohort.
Estey, Mathew P; Cohen, Ashley H; Colantonio, David A; Chan, Man Khun; Marvasti, Tina Binesh; Randell, Edward; Delvin, Edgard; Cousineau, Jocelyne; Grey, Vijaylaxmi; Greenway, Donald; Meng, Qing H; Jung, Benjamin; Bhuiyan, Jalaluddin; Seccombe, David; Adeli, Khosrow
The CALIPER program recently established a comprehensive database of age- and sex-stratified pediatric reference intervals for 40 biochemical markers. However, this database was only directly applicable for Abbott ARCHITECT assays. We therefore sought to expand the scope of this database to biochemical assays from other major manufacturers, allowing for a much wider application of the CALIPER database. Based on CLSI C28-A3 and EP9-A2 guidelines, CALIPER reference intervals were transferred (using specific statistical criteria) to assays performed on four other commonly used clinical chemistry platforms including Beckman Coulter DxC800, Ortho Vitros 5600, Roche Cobas 6000, and Siemens Vista 1500. The resulting reference intervals were subjected to a thorough validation using 100 reference specimens (healthy community children and adolescents) from the CALIPER bio-bank, and all testing centers participated in an external quality assessment (EQA) evaluation. In general, the transferred pediatric reference intervals were similar to those established in our previous study. However, assay-specific differences in reference limits were observed for many analytes, and in some instances were considerable. The results of the EQA evaluation generally mimicked the similarities and differences in reference limits among the five manufacturers' assays. In addition, the majority of transferred reference intervals were validated through the analysis of CALIPER reference samples. This study greatly extends the utility of the CALIPER reference interval database which is now directly applicable for assays performed on five major analytical platforms in clinical use, and should permit the worldwide application of CALIPER pediatric reference intervals. Copyright © 2013 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
Khoo, T-L; Xiros, N; Guan, F; Orellana, D; Holst, J; Joshua, D E; Rasko, J E J
The CELL-DYN Emerald is a compact bench-top hematology analyzer that can be used for a three-part white cell differential analysis. To determine its utility for analysis of human and mouse samples, we evaluated this machine against the larger CELL-DYN Sapphire and Sysmex XT2000iV hematology analyzers. 120 human (normal and abnormal) and 30 mouse (normal and abnormal) samples were analyzed on both the CELL-DYN Emerald and CELL-DYN Sapphire or Sysmex XT2000iV analyzers. For mouse samples, the CELL-DYN Emerald analyzer required manual recalibration based on the histogram populations. Analysis of the CELL-DYN Emerald showed excellent precision, within accepted ranges (white cell count CV% = 2.09%; hemoglobin CV% = 1.68%; platelets CV% = 4.13%). Linearity was excellent (R² ≥ 0.99), carryover was minimal (laboratories or as a backup system in larger laboratories. © 2012 John Wiley & Sons Ltd.
Ghys, Timothy; Goedhuys, Wim; Spincemaille, Katrien; Gorus, Frans; Gerlo, Erik
Glucose testing at the bedside has become an integral part of the management strategy in diabetes and of the careful maintenance of normoglycemia in all patients in intensive care units. We evaluated two point-of-care glucometers for the determination of plasma-equivalent blood glucose. The Precision PCx and the Accu-Chek Inform glucometers were evaluated. Imprecision and bias relative to the Vitros 950 system were determined using protocols of the Clinical Laboratory Standards Institute (CLSI). The effects of low, normal, and high hematocrit levels were investigated. Interference by maltose was also studied. Within-run precision for both instruments ranged from 2-5%. Total imprecision was less than 5% except for the Accu-Chek Inform at the low level (2.9 mmol/L). Both instruments correlated well with the comparison instrument and showed excellent recovery and linearity. Both systems reported at least 95% of their values within zone A of the Clarke Error Grid, and both fulfilled the CLSI quality criteria. The more stringent goals of the American Diabetes Association, however, were not reached. Both systems showed negative bias at high hematocrit levels. Maltose interfered with the glucose measurements on the Accu-Chek Inform but not on the Precision PCx. Both systems showed satisfactory imprecision and were reliable in reporting plasma-equivalent glucose concentrations. The most stringent performance goals were however not met.
Full Text Available In the last years the trend of centralizing small laboratories in large reference centers led to a careful evaluation of the diagnostic profiles. In the serology laboratory of Microbiology Unit, St. Orsola-Malpighi Hospital, Bologna, Italy the choice has been to combine random access analyzers (ARCHITECT Abbott and ELISA analyzers (BEPIII Dade Behring.
... Implementation Plan for Abbott Laboratories AGENCY: Environmental Protection Agency (EPA). ACTION: Final rule... site-specific rulemaking for certain of Abbott Laboratories' (Abbott) tunnel dryers and fluid bed... rulemaking for certain of Abbott Laboratories' (Abbott) tunnel dryers and fluid bed dryers. (i) Incorporation...
A European multicientre study on the comparison of HBV viral loads between VERIS HBV assay and Roche COBAS® TAQMAN® HBV test, Abbott RealTime HBV assay, Siemens VERSANT HBV assay, and Qiagen artus HBV RG kit.
Braun, Patrick; Delgado, Rafael; Drago, Monica; Fanti, Diana; Fleury, Hervé; Izopet, Jacques; Lombardi, Alessandra; Marcos, MaAngeles; Sauné, Karine; O'Shea, Siobhan; Pérez-Rivilla, Alfredo; Ramble, John; Trimoulet, Pascale; Vila, Jordi; Whittaker, Duncan; Artus, Alain; Rhodes, Daniel
Hepatitis B viral load testing is essential to treatment and monitoring decisions in patients with chronic Hepatitis B. Beckman Coulter has developed the VERIS HBV Assay (Veris) for use on the fully automated DxN VERIS Molecular Diagnostics System. 1 OBJECTIVES: To evaluate the clinical performance of the Veris HBV Assay at multiple EU laboratories STUDY DESIGN: Method comparison was performed with a total of 344 plasma specimens from HBV infected patients tested with Veris and COBAS ® TaqMan ® HBV Test (Cobas), 207 specimens tested with Veris and RealTime HBV Assay (RealTime), 86 specimens tested with Veris and VERSANT ® HBV Assay (Versant), and 74 specimens tested with Veris and artus ® HBV RG PCR kit (artus). Bland-Altman analysis showed average bias of -0.46 log 10 IU/mL between Veris and Cobas, -0.46 log 10 IU/mL between Veris and RealTime, -0.36 log 10 IU/mL between Veris and Versant, and -0.12 log 10 IU/mL between Veris and artus. Bias was consistent across the assay range. Patient monitoring results using Veris demonstrated similar viral load trends over time to Cobas, RealTime, and artus. The VERIS HBV Assay demonstrated comparable clinical performance, with varying degrees of negative bias, compared to other currently marketed assays for HBV DNA monitoring. This negative bias should be taken into consideration if switching monitoring methods to Veris. Copyright © 2017 Elsevier B.V. All rights reserved.
Braun, Patrick; Delgado, Rafael; Drago, Monica; Fanti, Diana; Fleury, Hervé; Hofmann, Jörg; Izopet, Jacques; Kühn, Sebastian; Lombardi, Alessandra; Mancon, Alessandro; Marcos, Mª Angeles; Mileto, Davide; Sauné, Karine; O'Shea, Siobhan; Pérez-Rivilla, Alfredo; Ramble, John; Trimoulet, Pascale; Vila, Jordi; Whittaker, Duncan; Artus, Alain; Rhodes, Daniel
Viral load monitoring is essential for patients under treatment for HIV. Beckman Coulter has developed the VERIS HIV-1 Assay for use on the novel, automated DxN VERIS Molecular Diagnostics System. ¥ OBJECTIVES: Evaluation of the clinical performance of the new quantitative VERIS HIV-1 Assay at multiple EU laboratories. Method comparison with the VERIS HIV-1 Assay was performed with 415 specimens at 5 sites tested with COBAS ® AmpliPrep/COBAS ® TaqMan ® HIV-1 Test, v2.0, 169 specimens at 3 sites tested with RealTime HIV-1 Assay, and 202 specimens from 2 sites tested with VERSANT HIV-1 Assay. Patient monitoring sample results from 4 sites were also compared. Bland-Altman analysis showed the average bias between VERIS HIV-1 Assay and COBAS HIV-1 Test, RealTime HIV-1 Assay, and VERSANT HIV-1 Assay to be 0.28, 0.39, and 0.61 log 10 cp/mL, respectively. Bias at low end levels below 1000cp/mL showed predicted bias to be <0.3 log 10 cp/mL for VERIS HIV-1 Assay versus COBAS HIV-1 Test and RealTime HIV-1 Assay, and <0.5 log 10 cp/mL versus VERSANT HIV-1 Assay. Analysis on 174 specimens tested with the 0.175mL volume VERIS HIV-1 Assay and COBAS HIV-1 Test showed average bias of 0.39 log 10 cp/mL. Patient monitoring results using VERIS HIV-1 Assay demonstrated similar viral load trends over time to all comparators. The VERIS HIV-1 Assay for use on the DxN VERIS System demonstrated comparable clinical performance to COBAS ® HIV-1 Test, RealTime HIV-1 Assay, and VERSANT HIV-1 Assay. Copyright © 2017 Elsevier B.V. All rights reserved.
... Services and Kelly Services. 75,274 Abbott Laboratories, Abbott Park, IL....... February 14, 2010... Moines, IA........ January 28, 2010. Inc., D/B/A UPS; Des Moines Billing Site. 75,156 Abbott Point of...
Saint Francis Health System (SFHS) consists of three hospitals and one clinic: Saint Francis Hospital (SFH); Broken Arrow Medical Center; Laureate Psychiatric Hospital; and Warren Clinic. SFHS has 670 physicians on staff and serves medical (oncology, orthopedic, neurology, and renal), surgical, cardiac, women and infant, pediatric, transplant, and trauma patients in Tulsa County, Oklahoma, which has a population of 660,000. SFH incorporates 706 staffed beds, including 126 pediatric beds and 119 critical care beds. Each year, the health system averages 38,000 admissions, 70,000 emergency department visits, 25,000 surgeries, and 3,500 births. Saint Francis Laboratory is located within the main hospital facility (SFH) and functions as a core lab for the health system. The lab also coordinates lab services with Saint Francis Heart Hospital, a physician-system joint venture. The Optimal Equipment Configuration (OEC) Project was designed by the Clinical Laboratory Services division of Premier, a group purchasing organization, with the goal of determining whether laboratories could improve efficiency and decrease unit cost by using a single-source vendor. Participants included seven business partners (Abbott, Bayer, Beckman/Coulter, Dade/Behring, J&J/ Ortho, Olympus, and Roche) and 21 laboratory sites (a small, mid-sized, and large site for each vendor). SFH laboratory staff embraced Premier's concept and viewed the OEC project as an opportunity to "energize" laboratory operations. SFH partnered with Abbott, their primary equipment vendor, for the project. Using resources and tools made available through the project, the laboratory was re-engineered to simplify workflow, increase productivity, and decrease costs by adding automation and changing to centralized specimen processing. Abbott and SFH shared a common vision for the project and enhanced their partnership through increased communication and problem solving. Abbott's area representatives provided for third
Khalil, O S; Zurek, T F; Tryba, J; Hanna, C F; Hollar, R; Pepe, C; Genger, K; Brentz, C; Murphy, B; Abunimeh, N
We describe a multichannel heterogeneous immunoassay analyzer in which a sample is split between disposable reaction trays in a group of linear tracks. The system's pipettor uses noninvasive sensing of the sample volume and disposable pipet tips. Each assay track has (a) a conveyor belt for moving reaction trays to predetermined functional stations, (b) temperature-controlled tunnels, (c) noncontact transfer of the reaction mixture between incubation and detection wells, and (d) single-photon counting to detect a chemiluminescence (CL) signal from the captured immunochemical product. A novel disposable reaction tray, with separate reaction and detection wells and self-contained fluid removal, is used in conjunction with the transfer device on the track to produce a carryover-free system. The linear immunoassay track has nine predetermined positions for performing individual assay steps. Assay step sequence and timing is selected by changing the location of the assay modules between these predetermined positions. The assay methodology, a combination of microparticle capture and direct detection of a CL signal on a porous matrix, offers excellent sensitivity, specificity, and ease of automation. Immunoassay configurations have been tested for hepatitis B surface antigen and for antibodies to hepatitis B core antigen, hepatitis C virus, human immunodeficiency virus I and II, and human T-cell leukemia virus I and II.
ing 40 min, I noted the olive-brown female was visiting a nest cavity near the top of a large dead tree, next to and entwined within the canopy of the olive (Fig. 1). The occupied cavity was one of three holes near the terminus, and on the underside of a rotten branch, and I estimated its height above ground at approximately ...
Full Text Available Objective To investigate the effects of different laboratory test reagents for hepatitis C virus (HCV antibody through a comparative analysis. Methods A total of 207 samples which tested positive by four anti-HCV screening reagents commonly used in the laboratories in China (Kehua, Xinchuang, Wantai, and Abbott were included. HCV RNA nucleic acid amplification (NAT was performed, and if NAT results were negative, recombinant immunoblot assay (RIBA was performed for further confirmation. The test results of these four screening reagents were compared, and their S/CO values and true positive rates were analyzed. Results Of all the 205 samples testing positive by any one reagent, 191 (93.2% tested positive by the four reagents, and 14 (6.8% were tested inconsistently by the four reagents. The positive predictive values of Xinchuang, Kehua, Wantai, and Abbott reagents were 88.2% (180/204, 93.8% (180/192, 91.4% (180/197, and 90.0% (180/200, respectively. The S/CO thresholds with a positive predictive value of ≥95% for Xinchuang, Kehua, Wantai, and Abbott reagents were 9.0, 4.0, 5.0, and 7.0, respectively. Conclusion Xinchuang, Kehua, Wantai, and Abbott reagents have significantly different S/CO thresholds with a positive predictive value of ≥95%, which are significantly different from those in other domestic laboratories. Each laboratory should establish an applicable S/CO threshold with a positive predictive value of ≥95%, in order to reduce the sample size for confirmatory test.
Bugdayci, Guler; Oguzman, Hamdi; Arattan, Havva Yasemin; Sasmaz, Guler
The use of Reference Change Values (RCV) has been advocated as very useful for monitoring individuals. Most of these are performed for monitoring individuals in acute situations and for following up the improvement or deterioration of chronic diseases. In our study, we aimed at evaluating the RCV calculation for 24 clinical chemistry analytes widely used in clinical laboratories and the utilization of this data. Twenty-four serum samples were analyzed with Abbott kits (Abbott Laboratories, Abbott Park, IL, USA), manufactured for use with the Architect c8000 (Abbott Laboratories, Abbott Park, IL, USA) auto-analyzer. We calculated RCV using the following formula: RCV = Z x 2 1/2x (CVA2 + CVw2)1/2. Four reference change values (RCV) were calculated for each analyte using four statistical probabilities (0.95, and 0.99, unidirectional and bidirectional). Moreover, by providing an interval after identifying upper and lower limits with the Reference Change Factor (RCF), serially measured tests were calculated by using two formulas: exp (Z x 2 1/2 x (CV(A)2 + CVw2)½/100) for RCF(UP) and (1/RCF(UP)) for RCF(DOWN). RCVs of these analytes were calculated as 14.63% for glucose, 29.88% for urea, 17.75% for ALP, 53.39% for CK, 46.98% for CK-MB, 21.00% amylase, 8.00% for total protein, 8.70% for albumin, 51.08% for total bilirubin, 86.34% for direct bilirubin, 6.40% for calcium, 15.03% for creatinine, 21.47% for urate, 14.19% for total cholesterol, 46.62% for triglyceride, 20.51% for HDL-cholesterol, 29.59% for AST, 46.31% for ALT, 31.54% for GGT, 20.92% for LDH, 19.75% for inorganic phosphate, 3.05% for sodium, 11.75% for potassium, 4.44% for chloride (RCV, p laboratories. RCV could be available as a tool for making clinical decision, especially when monitoring individuals.
Fei, Yang; Wang, Wei; He, Falin; Zhong, Kun; Wang, Zhiguo
The aim of this study was to use Six Sigma(SM) (Motorola Trademark Holdings, Libertyville, IL) techniques to analyze the quality of point-of-care (POC) glucose testing measurements quantitatively and to provide suggestions for improvement. In total, 151 laboratories in China were included in this investigation in 2014. Bias and coefficient of variation were collected from an external quality assessment and an internal quality control program, respectively, for POC glucose testing organized by the National Center for Clinical Laboratories. The σ values and the Quality Goal Index were used to evaluate the performance of POC glucose meters. There were 27, 30, 57, and 37 participants in the groups using Optium Xceed™ (Abbott Diabetes Care, Alameda, CA), Accu-Chek(®) Performa (Roche, Basel, Switzerland), One Touch Ultra(®) (Abbott), and "other" meters, respectively. The median of the absolute value of percentage difference varied among different lots and different groups. Among all the groups, the Abbott One Touch Ultra group had the smallest median of absolute value of percentage difference except for lot 201411, whereas the "other" group had the largest median in all five lots. More than 85% of participate laboratories satisfied the total allowable error (TEa) requirement in International Organization for Standardization standard 15197:2013, and 85.43% (129/151) of laboratories obtained intralaboratory coefficient of variations less than 1/3TEa. However, Six Sigma techniques suggested that 41.72% (63/151) to 65.56% (99/151) of the laboratories needed to improve their POC glucose testing performance, in either precision, trueness, or both. Laboratories should pay more attention on the practice of POC glucose testing and take actions to improve their performance. Only in this way can POC glucose testing really function well in clinical practice.
... NDA 007707, held by Abbott Laboratories, and initially approved on June 28, 1951. PHENURONE is an oral anticonvulsant indicated for the treatment of epilepsy. In a letter dated May 14, 2003, Abbott Laboratories...
... Rd., Tallaght, Dublin 24, Ireland. Abbott Laboratories, North Chicago, IL 60064, has informed FDA of... 21 CFR parts 522 and 529 to make conforming changes to Abbott Laboratories' product listings. This... paragraph (c)(1), revise the entry for ``Abbott Laboratories'' and remove the entry for ``RMS Laboratories...
... Implementation Plan for Abbott Laboratories AGENCY: Environmental Protection Agency (EPA). ACTION: Proposed rule... its SIP. These amendments consist of a site- specific rulemaking for certain of Abbott Laboratories' (Abbott) tunnel dryers and fluid bed dryers. This site-specific rule revision is approvable because it...
Pfisterer, M.; Bertel, O.; Bonetti, P.O.
or refute this hypothesis, we set up an 11-center 4-country prospective trial of 2260 consecutive patients treated with >= 3.0-mm stents only, randomized to receive Cypher (Johnson & Johnson, Miami Lakes, FL), Vision (Abbott Vascular, Abbott Laboratories, IL), or Xience stents (Abbott Vascular). Only...
Abbott, B. P.; Abbott, R.; Abbott, T. D.; Abernathy, M. R.; Acernese, F.; Ackley, K.; Adams, C.; Adams, T.; Addesso, P.; Adhikari, R. X.; Adya, V. B.; Affeldt, C.; Agathos, M.; Agatsuma, K.; Aggarwal, N.; Aguiar, O. D.; Aiello, L.; Ain, A.; Ajith, P.; Allen, B.; Allocca, A.; Altin, P. A.; Ananyeva, A.; Anderson, S. B.; Anderson, W. G.; Appert, S.; Arai, K.; Araya, M. C.; Areeda, J. S.; Arnaud, N.; Arun, K. G.; Ascenzi, S.; Ashton, G.; Ast, M.; Aston, S. M.; Astone, P.; Aufmuth, P.; Aulbert, C.; Avila-Alvarez, A.; Babak, S.; Bacon, P.; Bader, M. K. M.; Baker, P. T.; Baldaccini, F.; Ballardin, G.; Ballmer, S. W.; Barayoga, J. C.; Barclay, S. E.; Barish, B. C.; Barker, D.; Barone, F.; Barr, B.; Barsotti, L.; Barsuglia, M.; Barta, D.; Bartlett, J.; Bartos, I.; Bassiri, R.; Basti, A.; Batch, J. C.; Baune, C.; Bavigadda, V.; Bazzan, M.; Beer, C.; Bejger, M.; Belahcene, I.; Belgin, M.; Bell, A. S.; Berger, B. K.; Bergmann, G.; Berry, C. P. L.; Bersanetti, D.; Bertolini, A.; Betzwieser, J.; Bhagwat, S.; Bhandare, R.; Bilenko, I. A.; Billingsley, G.; Billman, C. R.; Birch, J.; Birney, R.; Birnho Ltz, O.; Biscans, S.; Bisht, A.; Bitossi, M.; Biwer, C.; Bizouard, M. A.; Blackburn, J. K.; Blackman, J.; Blair, C. D.; Blair, D. G.; Blair, R. M.; Bloemen, S.; Bock, O.; Boer, M.; Bogaert, G.; Bohe, A.; Bondu, F.; Bonnand, R.; Boom, B. A.; Bork, R.; Boschi, V.; Bose, S.; Bouffanais, Y.; Bozzi, A.; Bradaschia, C.; Brady, P. R.; Braginsky, V. B.; Branchesi, M.; Brau, J. E.; Briant, T.; Brillet, A.; Brinkmann, M.; Brisson, V.; Brockill, P.; Broida, J. E.; Brooks, A. F.; Brown, D. A.; Brown, D. D.; Brown, N. M.; Brunett, S.; Buchanan, C. C.; Buikema, A.; Bulik, T.; Bulten, H. J.; Buonanno, A.; Buskulic, D.; Buy, C.; Byer, R. L.; Cabero, M.; Cadonati, L.; Cagnoli, G.; Cahillane, C.; Calderon Bustillo, J.; Callister, T. A.; Calloni, E.; Camp, J. B.; Canepa, M.; Cannon, K. C.; Cao, H.; Cao, J.; Capano, C. D.; Capocasa, E.; Carbognani, F.; Caride, S.; Casanueva Diaz, J.; Casentini, C.; Caudill, S.; Cavaglia, M.; Cavalier, F.; Cavalieri, R.; Cella, G.; Cepeda, C. B.; Baiardi, L. C.; Cerretani, G.; Cesarini, E.; Chamberlin, S. J.; Chan, M.; Chao, S.; Charlton, P.; Chassande-Mottin, E.; Cheeseboro, B. D.; Chen, H. Y.; Chen, Y.; Cheng, H.-P.; Chincarini, A.; Chiummo, A.; Chmiel, T.; Cho, H. S.; Cho, M.; Chow, J. H.; Christensen, N.; Chu, Q.; Chua, A. J. K.; Chua, S.; Chung, S.; Ciani, G.; Clara, F.; Clark, J. A.; Cleva, F.; Cocchieri, C.; Coccia, E.; Cohadon, P.-F.; Colla, A.; Collette, C. G.; Cominsky, L.; Constancio, M.; Conti, L.; Cooper, S. J.; Corbitt, T. R.; Cornish, N.; Corsi, A.; Cortese, S.; Costa, C. A.; Coughlin, M. W.; Coughlin, S. B.; Coulon, J.-P.; Countryman, S. T.; Couvares, P.; Covas, P. B.; Cowan, E. E.; Coward, D. M.; Cowart, M. J.; Coyne, D. C.; Coyne, R.; Creighton, J. D. E.; Creighton, T. D.; Cripe, J.; Crowder, S. G.; Cullen, T. J.; Cumming, A.; Cunningham, L.; Cuoco, E.; Del Canton, T.; Danilishin, S. L.; D'Antonio, S.; Danzmann, K.; Dasgupta, A.; da Silva Costa, C. F.; Dattilo, V.; Dave, I.; Davier, M.; Davies, G. S.; Davis, D.; Daw, E. J.; Day, B.; Day, R.; de, S.; Debra, D.; Debreczeni, G.; Degallaix, J.; de Laurentis, M.; Deleglise, S.; Del Pozzo, W.; Denker, T.; Dent, T.; Dergachev, V.; De Rosa, R.; Derosa, R. T.; Desalvo, R.; Devenson, J.; Devine R. C, .; Dhurandhar, S.; Diaz, M. C.; di Fiore, L.; di Giovanni M.; di Girolamo, T.; di Lieto, A.; di Pace, S.; di Palma, I.; di Virgilio A.; Doctor, Z.; Dolique, V.; Donovan, F.; Dooley, K. L.; Doravari, S.; Dorrington, I.; Douglas, R.; Dovale Alvarez, M.; Downes, T. P.; Drago, M.; Drever, R. W. P.; Driggers, J. C.; Du, Z.; Ducrot, M.; Dwyer, S. E.; Edo, T. B.; Edwards, M. C.; Effler, A.; Eggenstein, H.-B.; Ehrens, P.; Eichholz, J.; Eikenberry, S. S.; Eisenstein, R. A.; Essick, R. C.; Etienne, Z.; Etzel, T.; Evans, M.; Evans, T. M.; Everett, R.; Factourovich, M.; Fafone, V.; Fair, H.; Fairhurst, S.; Fan, X.; Farinon, S.; Farr, B.; Farr, W. M.; Fauchon-Jones, E. J.; Favata, M.; Fays, M.; Fehrmann, H.; Fejer, M. M.; Fernandez Galiana, A.; Ferrante, I.; Ferreira, E. C.; Ferrini, F.; Fidecaro, F.; Fiori, I.; Fiorucci, D.; Fisher, R. P.; Flaminio, R.; Fletcher, M.; Fong, H.; Forsyth, S. S.; Fournier, J.-D.; Frasca, S.; Frasconi, F.; Frei, Z.; Freise, A.; Frey, R.; Frey, V.; Fries, E. M.; Fritschel, P.; Frolov, V. V.; Fulda, P.; Fyffe, M.; Gabbard, H.; Gadre, B. U.; Gaebel, S. M.; Gair, J. R.; Gammaitoni, L.; Gaonkar, S. G.; Garufi, F.; Gaur, G.; Gayathri, V.; Gehrels, N.; Gemme, G.; Genin, E.; Gennai, A.; George, J.; Gergely, L.; Germain, V.; Ghonge, S.; Ghosh, A.; Ghosh, A.; Ghosh, S.; Giaime, J. A.; Giardina, K. D.; Giazotto, A.; Gill, K.; Glaefke, A.; Goetz, E.; Goetz, R.; Gondan, L.; Gonzalez, G.; Gonzalez Castro, J. M.; Gopakumar, A.; Gorodetsky, M. L.; Gossan, S. E.; Gosselin, M.; Gouaty, R.; Grado, A.; Graef, C.; Granata, M.; Grant, A.; Gras, S.; Gray, C.; Greco, G.; Green, A. C.; Groot, P.; Grote, H.; Grunewald, S.; Guidi, G. M.; Guo, X.; Gupta, A.; Gupta, M. K.; Gushwa, K. E.; Gustafson, E. K.; Gustafson, R.; Hacker, J. J.; Hall, B. R.; Hall, E. D.; Hammond, G.; Haney, M.; Hanke, M. M.; Hanks, J.; Hanna, C.; Hanson, J.; Hardwick, T.; Harms, J.; Harry, G. M.; Harry, I. W.; Hart, M. J.; Hartman, M. T.; Haster, C.-J.; Haughian, K.; Healy, J.; Heidmann, A.; Heintze, M. C.; Heitmann, H.; Hello, P.; Hemming, G.; Hendry, M.; Heng, I. S.; Hennig, J.; Henry, J.; Heptonstall, A. W.; Heurs, M.; Hild, S.; Hoak, D.; Hofman, D.; Holt, K.; Holz, D. E.; Hopkins, P.; Hough, J.; Houston, E. A.; Howell, E. J.; Hu, Y. M.; Huerta, E. A.; Huet, D.; Hughey, B.; Husa, S.; Huttner, S. H.; Huynh-Dinh, T.; Indik, N.; Ingram, D. R.; Inta, R.; Isa, H. N.; Isac, J.-M.; Isi, M.; Isogai, T.; Iyer, B. R.; Izumi, K.; Jacqmin, T.; Jani, K.; Jaranowski, P.; Jawahar, S.; Jimenez-Forteza, F.; Johnson, W. W.; Jones, D. I.; Jones, R.; Jonker, R. J. G.; Ju, L.; Junker, J.; Kalaghatgi, C. V.; Kalogera, V.; Kandhasamy, S.; Kang, G.; Kanner, J. B.; Karki, S.; Karvinen, K. S.; Kasprzack, M.; Katsavounidis, E.; Katzman, W.; Kaufer, S.; Kaur, T.; Kawabe, K.; Kefelian, F.; Keitel, D.; Kelley, D. B.; Kennedy, R.; Key, J. S.; Khalili, F. Y.; Khan, I.; Khan, S.; Khan, Z.; Khazanov, E. A.; Kijbunchoo, N.; Kim, C.; Kim, J. C.; Kim, W.; Kim, W.; Kim, Y.-M.; Kimbrell, S. J.; King, E. J.; King, P. J.; Kirchhoff, R.; Kissel, J. S.; Klein, B.; Kleybolte, L.; Klimenko, S.; Koch, P.; Koehlenbeck, S. M.; Koley, S.; Kondrashov, V.; Kontos, A.; Korobko, M.; Korth, W. Z.; Kowalska, I.; Kozak, D. B.; Kramer, C.; Kringel, V.; Krishnan, B.; Krolak, A.; Kuehn, G.; Kumar, P.; Kumar, R.; Kuo, L.; Kutynia, A.; Lackey, B. D.; Landry, M.; Lang, R. N.; Lange, J.; Lantz, B.; Lanza, R. K.; Lartaux-Vollard, A.; Lasky, P. D.; Laxen, M.; Lazzarini, A.; Lazzaro, C.; Leaci, P.; Leavey, S.; Lebigot, E. O.; Lee, C. H.; Lee, H. K.; Lee, H. M.; Lee, K.; Lehmann, J.; Lenon, A.; Leonardi, M.; Leong, J. R.; Leroy, N.; Letendre, N.; Levin, Y.; Li, T. G. F.; Libson, A.; Littenberg, T. B.; Liu, J.; Lockerbie, N. A.; Lombardi, A. L.; London, L. T.; Lord, J. E.; Lorenzini, M.; Loriette, V.; Lormand, M.; Losurdo, G.; Lough, J. D.; Lousto, C. O.; Lovelace, G.; Luck, H.; Lundgren, A. P.; Lynch, R.; Ma, Y.; Macfoy, S.; Machenschalk, B.; Macinnis, M.; MacLeod, D. M.; Magana-Sandoval, F.; Majorana, E.; Maksimovic, I.; Malvezzi, V.; Man, N.; Mandic, V.; Mangano, V.; Mansell, G. L.; Manske, M.; Mantovani, M.; Marchesoni, F.; Marion, F.; Marka, S.; Marka, Z.; Markosyan, A. S.; Maros, E.; Martelli, F.; Martellini, L.; Martin, I. W.; Martynov, D. V.; Mason, K.; Masserot, A.; Massinger, T. J.; Masso-Reid, M.; Mastrogiovanni, S.; Matichard, F.; Matone, L.; Mavalvala, N.; Mazumder, N.; McCarthy, R.; McClelland, D. E.; McCormick, S.; McGra, Th C.; McGuire, S. C.; McIntyre, G.; McIver, J.; McManus, D. J.; McRae, T.; McWilliams, S. T.; Meacher, D.; Meadors, G. D.; Meidam, J.; Melatos, A.; Mendell, G.; Mendoza-Gandara, D.; Mercer, R. A.; Merilh, E. L.; Merzougui, M.; Meshkov, S.; Messenger, C.; Messick, C.; Metzdorff, R.; Meyers, P. M.; Mezzani, F.; Miao, H.; Michel, C.; Middleton, H.; Mikhailov, E. E.; Milano, L.; Miller, A. L.; Miller, A.; Miller, B. B.; Miller, J.; Millhouse, M.; Minenkov, Y.; Ming, J.; Mirshekari, S.; Mishra, C.; Mitra, S.; Mitrofanov, V. P.; Mitselmakher, G.; Mittleman, R.; Moggi, A.; Mohan, M.; Mohapatra, S. R. P.; Montani, M.; Moore, B. C.; Moore, C. J.; Moraru, D.; Moreno, G.; Morriss, S. R.; Mours, B.; Mow-Lowry, C. M.; Mueller, G.; Muir, A. W.; Mukherjee, A.; Mukherjee, D.; Mukherjee, S.; Mukund, N.; Mullavey, A.; Munch, J.; Muniz, E. A. M.; Murray, P. G.; Mytidis, A.; Napier, K.; Nardecchia, I.; Naticchioni, L.; Nelemans, G.; Nelson, T. J. N.; Neri, M.; Nery, M.; Neunzert, A.; Newport, J. M.; Newton, G.; Nguyen, T. T.; Nielsen, A. B.; Nissanke, S.; Nitz, A.; Noack, A.; Nocera, F.; Nolting, D.; Normandin, M. E. N.; Nuttall, L. K.; Oberling, J.; Ochsner, E.; Oelker, E.; Ogin, G. H.; Oh, J. J.; Oh, S. H.; Ohme, F.; Oliver, M.; Oppermann, P.; Oram, R. J.; O'Reilly, B.; O'Shaughnessy, R.; Ottaway, D. J.; Overmier, H.; Owen, B. J.; Pace, A. E.; Page, J.; Pai, A.; Pai, S. A.; Palamos, J. R.; Palashov, O.; Palomba, C.; Pal-Singh, A.; Pan, H.; Pankow, C.; Pannarale, F.; Pant, B. C.; Paoletti, F.; Paoli, A.; Papa, M. A.; Paris, H. R.; Parker, W.; Pascucci, D.; Pasqualetti, A.; Passaquieti, R.; Passuello, D.; Patricelli, B.; Pearlstone, B. L.; Pedraza, M.; Pedurand, R.; Pekowsky, L.; Pele, A.; Penn, S.; Perez, C. J.; Perreca, A.; Perri, L. M.; Pfeiffer, H. P.; Phelps, M.; Piccinni, O. J.; Pichot, M.; Piergiovanni, F.; Pierro, V.; Pillant, G.; Pinard, L.; Pinto, I. M.; Pitkin, M.; Poe, M.; Poggiani, R.; Popolizio, P.; Post, A.; Powell, J.; Prasad, J.; Pratt, J. W. W.; Predoi, V.; Prestegard, T.; Prijatelj, M.; Principe, M.; Privitera, S.; Prix, R.; Prodi, G. A.; Prokhorov, L. G.; Puncken, O.; Punturo, M.; Puppo, P.; Purrer, M.; Qi, H.; Qin, J.; Qiu, S.; Quetschke, V.; Quintero E. A.; QuitzoW-James, R.; Raab, F. J.; Rabeling, D. S.; Radkins, H.; Raffai, P.; Raja, S.; Rajan, C.; Rakhmanov, M.; Rapagnani, P.; Raymond, V.; Razzano, M.; Re, V.; Read, J.; Regimbau, T.; Rei, L.; Reid, S.; Reitze, D. H.; Rew, H.; Reyes, S. D.; Rhoades, E.; Ricci, F.; Riles, K.; Rizzo, M.; Robertson, N. A.; Robie, R.; Robinet, F.; Rocchi, A.; Rolland, L.; Rollins, J. G.; Roma, V. J.; Romano, R.; Romie, J. H.; Rosinska, D.; Rowan, S.; Rudiger, A.; Ruggi, P.; Ryan, K.; Sachdev, S.; Sadecki, T.; Sadeghian, L.; Sakellariadou, M.; Salconi, L.; Saleem, M.; Salemi, F.; Samajdar, A.; Sammut, L.; Sampson, L. M.; Sanchez, E. J.; Sandberg, V.; Sanders, J. R.; Sassolas, B.; Sathyaprakash, B. S.; Saulson, P. R.; Sauter, O.; Savage, R. L.; Sawadsky, A.; Schale, P.; Scheuer, J.; Schmidt, E.; Schmidt, J.; Schmidt, P.; Schnabel, R.; Schofield, R. M. S.; Schonbeck, A.; Schreiber, E.; Schuette, D.; Schutz, B. F.; Schwalbe, S. G.; Scott, J.; Scott, S. M.; Sellers, D.; Sengupta, A. S.; Sentenac, D.; Sequino, V.; Sergeev, A.; Setyawati, Y.; Shaddock, D. A.; Shaffer, T. J.; Shahriar, M. S.; Shapiro, B.; Shawhan P.; Sheperd, A.; Shoemaker, D. H.; Shoemaker, D. M.; Siellez, K.; Siemens, X.; Sieniawska, M.; Sigg, D.; Silva, A. D.; Singer, A.; Singer, L. P.; Singh, A.; Singh, R.; Singhal, A.; Sintes, A. M.; Slagmolen, B. J. J.; Smith, B.; Smith, J. R.; Smith, R. J. E.; Son, E. J.; Sorazu, B.; Sorrentino, F.; Souradeep, T.; Spencer, A. P.; Srivastava, A. K.; Staley, A.; Steinke, M.; Steinlechner, J.; Steinlechner, S.; Steinmeyer, D.; Stephens, B. C.; Stevenson, S. P.; Stone, R.; Strain, K. A.; Straniero, N.; Stratta, G.; Strigin, S. E.; Sturani, R.; Stuver, A. L.; Summerscales, T. Z.; Sun, L.; Sunil, S.; Sutton, P. J.; Swinkels, B. L.; Szczepanczyk, M. J.; Tacca, M.; Talukder, D.; Tanner, D. B.; Tapai, M.; Taracchini, A.; Taylor, R.; Theeg, T.; Thomas, E. G.; Thomas, M.; Thomas, P.; Thorne, K. A.; Thrane, E.; Tippens, T.; Tiwari, S.; Tiwari, V.; Tokmakov, K. V.; Toland, K.; Tomlinson, C.; Tonelli, M.; Tornasi, Z.; Torrie, C. I.; Toyra, D.; Travasso, F.; Traylor, G.; Trifiro, D.; Trinastic, J.; Tringali, M. C.; Trozzo, L.; Tse, M.; Tso, R.; Turconi, M.; Tuyenbayev, D.; Ugolini, D.; Unnikrishnan, C. S.; Urban, A. L.; Usman, S. A.; Vahlbruch, H.; Vajente, G.; Valdes, G.; van Bakel, N.; van Beuzekom, M.; van den Brand, J. F. J.; van den Broeck, C.; Vander-Hyde, D. C.; van der Schaaf, L.; van Heijningen, J. V.; van Veggel, A. A.; Vardaro, M.; Varma, V.; Vass, S.; Vasuth, M.; Vecchio, A.; Vedovato, G.; Veitch, J.; Veitch, P. J.; Venkateswara, K.; Venugopalan, G.; Verkindt, D.; Vetrano, F.; Vicere, A.; Viets, A. D.; Vinciguerra, S.; Vine, D. J.; Vinet, J.-Y.; Vitale, S.; Vo, T.; Vocca, H.; Vorvick, C.; Voss, D. V.; Vousden, W. D.; Vyatchanin, S. P.; Wade, A. R.; Wade, L. E.; Wade, M.; Walker, M.; Wallace, L.; Walsh, S.; Wang, G.; Wang, H.; Wang, M.; Wang, Y.; Ward, R. L.; Warner, J.; Was, M.; Watchi, J.; Weaver, B.; Wei, L.-W.; Weinert, M.; Weinstein, A. J.; Weiss, R.; Wen, L.; Wessels, P.; Westphal, T.; Wette, K.; Whelan, J. T.; Whiting, B. F.; Whittle, C.; Williams, D.; Williams, R. D.; Williamson, A. R.; Willis, J. L.; Willke, B.; Wimmer, M. H.; Winkler, W.; Wipf, C. C.; Wittel, H.; Woan, G.; Woehler, J.; Worden, J.; Wright, J. L.; Wu, D. S.; Wu, G.; Yam, W.; Yamamoto, H.; Yancey, C. C.; Yap, M. J.; Yu, H.; Yu, H.; Yvert, M.; Zadrozny, A.; Zangrando, L.; Zanolin, M.; Zendri, J.-P.; Zevin, M.; Zhang, L.; Zhang, M.; Zhang, T.; Zhang, Y.; Zhao, C.; Zhou, M.; Zhou, Z.; Zhu, S. J.; Zhu, X. J.; Zucker, M. E.; Zweizig, J.; Buchner, S.; Cognard, I.; Corongiu, A.; Freire, P. C. C.; Guillemot, L.; Hobbs, G. B.; Kerr, M.; Lyne, A. G.; Possenti, A.; Ridolfi, A.; Shannon, R. M.; Stappers, B. W.; Weltevrede, P.; (The Ligo Scientific Collaboration)
We have obtained timings for 200 known pulsars. Timing was performed using the 42ft telescope and Lovell telescope at Jodrell Bank (UK), the 26m telescope at Hartebeesthoek (South Africa), the Parkes radio telescope (Australia), the Nancay Decimetric Radio Telescope (France), the Arecibo Observatory (Puerto Rico) and the Fermi Large Area Telescope (LAT). Of these, 122 have been targeted in previous campaigns (Aasi+ 2014, J/ApJ/785/119), while 78 are new to this search. (1 data file).
Krintus, Magdalena; Kozinski, Marek; Boudry, Pascal
BACKGROUND: International recommendations highlight the superior value of cardiac troponins (cTns) for early diagnosis of myocardial infarction along with analytical requirements of improved precision and detectability. In this multicenter study, we investigated the analytical performance of a ne...
We are evaluating whether detection of a tumor-specific N-linked glycosylation known as B 1,6 branched N-glycan present on the glycoprotein periostin in breast cancer will be useful as a new biomarker for the detection of breast cancer in patient plasma and serum. We have completed an initial study using samples with known inavasive ductal breast carcinoma diagnosis and the results look very promising. Therefore, we would like to proceed with our analysis of this potential biomarker for breast cancer diagnosis by analyzing the blinded samples in breast reference set 1.
Peng, Jun-Jin; Xiang, Wen-Chang; Cai, Shao-Hong
Not Available Supported by the National Natural Science Foundation of China under Grant Nos 11275157 and 11505036, the Doctoral Research Fund of Guizhou Normal University in 2014, the Technology Department of Guizhou Province Fund under Grant No 2114, and the Science and Technology Innovation Talent Team of Guizhou Province under Grant No (2015)4015.
Ponessa, Joan; Boylan, Ellen
This report on preschool facilities analyzes regulations proposed by the New Jersey Department of Education (NJDOE) to implement the Educational Facilities Construction and Financing Act. (EFCFA). EFCFA, which authorizes and governs New Jersey's public school construction program, was enacted in July 2000 to implement the State Supreme Court's…
This paper deals with four countries that, like Rwanda, suffered economic and social collapse in the early 1990s. It develops a sociologically informed understanding of what influences the well-being of people living in Kazakhstan and Kyrgyzstan (Central Asian Republics) and Armenia and Georgia ( the Caucasus), four of ...
Park, Yongjung; Hong, Duck Jin; Shin, Saeam; Cho, Yonggeun; Kim, Hyon-Suk
We evaluated quantitative hepatitis B surface antigen (qHBsAg) assays and a hepatitis B virus (HBV) core-related antigen (HBcrAg) assay. A total of 529 serum samples from patients with hepatitis B were tested. HBsAg levels were determined by using the Elecsys (Roche Diagnostics, Indianapolis, IN) and Architect (Abbott Laboratories, Abbott Park, IL) qHBsAg assays. HBcrAg was measured by using Lumipulse HBcrAg assay (Fujirebio, Tokyo, Japan). Serum aminotransferases and HBV DNA were respectively quantified by using the Hitachi 7600 analyzer (Hitachi High-Technologies, Tokyo, Japan) and the Cobas AmpliPrep/Cobas TaqMan test (Roche). Precision of the qHBsAg and HBcrAg assays was assessed, and linearity of the qHBsAg assays was verified. All assays showed good precision performance with coefficients of variation between 4.5% and 5.3% except for some levels. Both qHBsAg assays showed linearity from 0.1 to 12,000.0 IU/mL and correlated well (r = 0.9934). HBsAg levels correlated with HBV DNA (r = 0.3373) and with HBcrAg (r = 0.5164), and HBcrAg also correlated with HBV DNA (r = 0.5198; P < .0001). This observation could provide impetus for further research to elucidate the clinical usefulness of the qHBsAg and HBcrAg assays.
... the Sibutramine Cardiovascular Outcomes Trial (SCOUT) (M01- 392), for new drug application (NDA) 20-632, MERIDIA (sibutramine hydrochloride monohydrate) Capsules, sponsored by Abbott Laboratories, for...
Meyer, Jay J; Kuo, Annie F; Olson, Randall J
To determine capsular breakage risk from contact by phacoemulsification needles by machine and tip type. Experimental laboratory investigation. Infiniti (Alcon, Inc.) with Intrepid cartridges and Signature (Abbott Medical Optics, Inc.) phacoemulsification machines were tested using 19- and 20-gauge sharp and rounded tips. Actual and unoccluded flow vacuum were determined at 550 mm Hg, bottle height of 75 cm, and machine-indicated flow rate of 60 mL/minute. Breakage from brief tip contact with a capsular surrogate and human cadaveric lenses was calculated. Nineteen-gauge tips had more flow and less unoccluded flow vacuum than 20-gauge tips for both machines, with highest unoccluded flow vacuum in the Infiniti. The 19-gauge sharp tip was more likely than the 20-gauge sharp tip to cause surrogate breakage for Signature with micropulse and Ellips (Abbott Medical Optics, Inc.) ultrasound at 100% power. For Infiniti using OZil (Alcon, Inc.) ultrasound, 20-gauge sharp tips were more likely than 19-gauge sharp tips to break the membrane. For cadaveric lenses, using rounded 20-gauge tips at 100% power, breakage rates were micropulse (2.3%), Ellips (2.3%), OZil (5.3%). Breakage rates for sharp 20-gauge Ellips tips were higher than for rounded tips. Factors influencing capsular breakage may include active vacuum at the tip, flow rate, needle gauge, and sharpness. Nineteen-gauge sharp tips were more likely than 20-gauge tips to cause breakage in lower vacuum methods. For higher-vacuum methods, breakage is more likely with 20-gauge than with 19-gauge tips. Rounded-edge tips are less likely than sharp-edged tips to cause breakage. Copyright 2010 Elsevier Inc. All rights reserved.
.... Testosterone gel, marketed by Abbott under the brand name Androgel, is a prescription gel used to treat adult... proposed transaction. With its resources, capabilities, strong reputation, and experience manufacturing and marketing generic products, Watson is well-positioned to replicate the competition that would be lost with...
Bennett, Michael; Dent, Catherine L.; Ma, Qing; Dastrala, Sudha; Grenier, Frank; Workman, Ryan; Syed, Hina; Ali, Salman; Barasch, Jonathan; Devarajan, Prasad
Background and objectives: The authors have previously shown that urine neutrophil gelatinase-associated lipocalin (NGAL), measured by a research ELISA, is an early predictive biomarker of acute kidney injury (AKI) after cardiopulmonary bypass (CPB). In this study, whether an NGAL immunoassay developed for a standardized clinical platform (ARCHITECT analyzer®, Abbott Diagnostics Division, Abbott Laboratories, Abbott Park, IL) can predict AKI after CPB was tested.
Terri WEISSMAN, The Realisms of Berenice Abbott : Documentary Photography and Political Action/Sharon CORWIN, Jessica MAY et Terri WEISSMAN, American Modern : Documentary Photography by Abbott, Evans, and Bourke-White
La tradition documentaire américaine des années 1930 garderait-elle quelque vertu – malgré les assauts parfois dévastateurs menés il y a une génération de cela par la critique d’inspiration poststructuraliste ? Est-on encore fondé à postuler l’éventualité d’une contribution sociale et politique digne d’intérêt à propos d’une photographie de type journalistique ou documentaire sans se voir immédiatement opposer quelques conclusions définitives sur la nature prédatrice de l’outil photographique...
This report on Long Range Facilities Plans (LRFPs) analyzes regulations proposed by the New Jersey Department of Education (NJDOE) to implement the Educational Facilities Construction and Financing Act. (EFCFA). EFCFA, which authorizes and governs New Jersey's public school construction program, was enacted in July 2000 to implement the State…
.... Corporation, West Plains Division, Regal Beloit Corporation. 75,201 Abbott Laboratories, Irving, TX February 9... Jeanerette Distribution Center; Fruit of the Loom; Leased Workers Spherion. 74,902 Abbott Diabetes Care, Inc... Subject firm Location Impact date 75,246 Deluxe Laboratories......... Hollywood, CA 75,246A Deluxe...
supplement, guinea pig and fish chow (Ralston Purina Co.), desiccated hog liver powder,-and brewer’s yeast (ICN Pharmaceuticals Inc). The dog food, hog...110 - .115 17 15.680 -.095 .064 ___39 36.034 -.090 .082 REFERENCES CITED Abbott , W. S. 1925. A method of computing the effectiveness of an insecticide...Trop. Med. 24:131-174. Reid, J. A. 1968. Anopheline mosquitoes of Malaya and Borneo. Stud. Inst. Med. Res. Malaysia , Kuala Lumpur. Reid, J. A. 1970
Saraswathy, T S; Rozainanee, M Z; Asshikin, R Nurul; Zainah, S
Rubella infection in pregnant women during the first trimester of pregnancy can lead to fetal anomalies, commonly known as congenital rubella syndrome (CRS). The objective of our study was to analyze the serological test results among infants suspected of having CRS aged Malaysia were examined for rubella specific IgM and IgG antibodies using a Axsym, automated analyzer (Abbott Laboratories). Forty-eight samples were positive for rubella specific IgM antibodies and 494 samples were positive for rubella specific IgG antibodies. These were then age stratified and their clinical history reviewed for any CRS symptoms. Fifteen of 38 rubella IgM positive infants (39.5%) aged < 3 months, had a clinical appearance compatible with CRS. However, only 1 IgM positive infant aged 3 to 6 months and one infant aged 7 to 11 months had clinical appearance compatible with CRS. The most common abnormal findings in these cases were congenital heart defects and cataracts. Forty-eight point eight percent of IgM positive cases and 53.1% of IgG positive cases, had inadequate information in the chart to determine the presence of CRS. Clinical findings and timely laboratory diagnosis to determine the presence of CRS are important in infants born with congenital defects. Physicians should also be aware of the appropriate interpretation of these findings.
Weykamp, C W; Penders, T J; Miedema, K; Muskiet, F A; van der Slik, W
We investigated the effect of calibration with lyophilized calibrators on whole-blood glycohemoglobin (glyHb) results. One hundred three laboratories, using 20 different methods, determined glyHb in two lyophilized calibrators and two whole-blood samples. For whole-blood samples with low (5%) and high (9%) glyHb percentages, respectively, calibration decreased overall interlaboratory variation (CV) from 16% to 9% and from 11% to 6% and decreased intermethod variation from 14% to 6% and from 12% to 5%. Forty-seven laboratories, using 14 different methods, determined mean glyHb percentages in self-selected groups of 10 nondiabetic volunteers each. With calibration their overall mean (2SD) was 5.0% (0.5%), very close to the 5.0% (0.3%) derived from the reference method used in the Diabetes Control and Complications Trial. In both experiments the Abbott IMx and Vision showed deviating results. We conclude that, irrespective of the analytical method used, calibration enables standardization of glyHb results, reference values, and interpretation criteria.
Nino-Fong, R; Collins, TM; Chan, CB
Introduction: The present investigation was designed to investigate the accuracy and precision of lactate measurement obtained with contemporary biosensors (Chiron Diagnostics, Nova Biomedical) and standard enzymatic pho-tometric procedures (Sigma Diagnostics, Abbott Laboratories, Analyticon).
Mar 12, 2015 ... 1Department of Medical Laboratory Technology, Kwame Nkrumah University of Science and .... increases, as in diabetes mellitus and in metabolic acidosis during .... Matthew CA, Frank P, Hurst, Kevin C, Abbott MD, Robert M.
Peripherals neuropathies are common complications of diabetes. ... We retrospectively studied the data of 110 diabetics admitted in the electromyography laboratory of the ..... Rajabally A. Yusuf, Sarasamma Priyarenjini, Abbott J. Richard.
Keywords: Camel Milk, Serum glucose, Lipid profile, Diabetes. INTRODUCTION. Diabetes is ... all products of Randox Laboratories, Switzerland. Fresh camel milk samples .... Abbott, R.D., Wilson, P.W., Kannel, W.B. and. Castelli, W.P. (1988).
Sep 1, 1990 ... MRC Blood Platelet Research Unit, Department of Haema- tology, University of .... was made isosmotic with a buffered saline-glucose-citrate solu- tion10 by adding 9 ... assayed with commercial kits (pF4 - Abbott Laboratories;.
2007 hearing of the House Oversight and Government Reform Subcommittee on Information Policy, Gregory Wilshusen, director of Information Technology at...Abbott Laboratories in Chicago, recruiters are reaching out to College students by offering flexible work schedules, telecommuting , full tuition
Berlet, Matthew H.; Steffen, Diana; Shaughness, George; Hanner, James
Severe complications can and do occur when central venous catheters are inadvertently placed into subclavian arteries. Two cases are discussed that describe how these inadvertent arterial punctures can be closed using the Perclose device (Abbott Laboratories, Redwood City, CA, USA)
Popular culture requires readily identifiable villains. Subcultural groups often serve this role, creating controversies. Controversies based on religion are especially bitter. As a rule, religion in the movies is inoffensively sentimental, but "The Da Vinci Code" is both popular and provocative, treading on the dangerous ground of Jesus's…
Tamimi, Waleed; Albanyan, Esam; Altwaijri, Yasmin; Tamim, Hani; Alhussein, Fahad
Reference intervals for pubertal characteristics are influenced by genetic, geographic, dietary and socioeconomic factors. Therefore, the aim of this study was to establish age-specific reference intervals of glucose and lipid levels among local school children. This was cross-sectional study, conducted among Saudi school children. Fasting blood samples were collected from 2149 children, 1138 (53%) boys and 1011 (47%) girls, aged 6 to 18 years old. Samples were analyzed on the Architect c8000...
Reed, Derek W
...), fundamentally changing how warfare is being conducted. Network centric warfare (NCW) systems are being rushed to the field and are offered as a solution for the fog of war and as a way to reduce manpower costs...
De Keukeleire, Steven; Desmet, Stefanie; Lagrou, Katrien; Oosterlynck, Julie; Verhulst, Manon; Van Besien, Jessica; Saegeman, Veroniek; Reynders, Marijke
The performance of Elecsys Syphilis was compared to Architect Syphilis TP and Reformulated Architect Syphilis TP. The overall sensitivity and specificity were 98.4% and 99.5%, 97.7% and 97.1%, and 99.2% and 99.7% respectively. The assays are comparable and considered adequate for syphilis screening. Copyright © 2016 Elsevier Inc. All rights reserved.
Rytuba, James J.; Hothem, Roger L.; Brussee, Brianne E.; Goldstein, Daniel N.
Executive Summary Stable-isotope data indicate that there are three sources of water that effect the composition and Hg concentration of waters in Harley Gulch: (1) meteoric water that dominates water chemistry during the wet season; (2) thermal water effluent from the Turkey Run mine that effects the chemistry at sample site HG1; and (3) cold connate groundwater that dominates water chemistry during the dry season as it upwells and reaches the surface. The results from sampling executed for this study suggest four distinct areas in Harley Gulch: (1) the contaminated West Fork of Harley Gulch, consisting of the stream immediately downstream from the mine area and the wetlands upstream from Harley Gulch canyon (sample sites HG1-HG2, (2) the East Fork of Harley Gulch, where no mining has occurred (sample site HG3), (3) sample sites HG4-HG7, where a seasonal influx of saline groundwater alters stream chemistry, and (4) sample sites HG7-HG10, downstream in Harley Gulch towards the confluence with Cache Creek.
Bezeaud, A; Rosenswajg, M; Guillin, M C
Five commercially available assays for hepatitis delta (HD) virus markers were compared for sensitivity, specificity, and reproducibility: three assays for antibody (anti-HD), provided by Diagnostics Pasteur, Organon Teknika, and Abbott Laboratories, and two assays for antigen (HD Ag), from Pasteur and Organon Teknika. The assay from Organon Teknika is the less sensitive assay for anti-HD detection. Although the sensitivities of the Pasteur and Abbott assays for anti-HD detection are similar,...
Pfützner, Andreas; Musholt, Petra B; Schipper, Christina; Demircik, Filiz; Hengesbach, Carina; Flacke, Frank; Sieber, Jochen; Forst, Thomas
Hematocrit (HCT) is known to be a confounding factor that interferes with many blood glucose (BG) measurement technologies, resulting in wrong readings. Dynamic electrochemistry has been identified as one possible way to correct for these potential deviations. The purpose of this laboratory investigation was to assess the HCT stability of four BG meters known to employ dynamic electrochemistry (BGStar and iBGStar, Sanofi; Wavesense Jazz, AgaMatrix; Wellion Linus, MedTrust) in comparison with three other devices (GlucoDock, Medisana; OneTouch Verio Pro, LifeScan; FreeStyle Freedom InsuLinx, Abbott-Medisense). Venous heparinized blood was immediately aliquoted after draw and manipulated to contain three different BG concentrations (60-90, 130-160, and 280-320 mg/dl) and five different HCT levels (25%, 35%, 45%, 55%, and 60%). After careful oxygenation to normal blood oxygen pressure, each of the resulting 15 different samples was measured six times with three devices and three strip lots of each meter. The YSI Stat 2300 served as laboratory reference method. Stability to HCT influence was assumed when less than 10% difference occurred between the highest and lowest mean glucose deviations in relation to HCT concentrations [hematocrit interference factor (HIF)]. Five of the investigated self-test meters showed a stable performance with the different HCT levels tested in this investigation: BGStar (HIF 4.6%), iBGStar (6.6%), Wavesense Jazz (4.1%), Wellion Linus (8.5%), and OneTouch Verio Pro (6.2%). The two other meters were influenced by HCT (FreeStyle InsuLinx 17.8%; GlucoDock 46.5%). In this study, meters employing dynamic electrochemistry, as used in the BGStar and iBGStar devices, were shown to correct for potential HCT influence on the meter results. Dynamic electrochemistry appears to be an effective way to handle this interfering condition. © 2013 Diabetes Technology Society.
Pfützner, Andreas; Musholt, Petra B.; Schipper, Christina; Demircik, Filiz; Hengesbach, Carina; Flacke, Frank; Sieber, Jochen; Forst, Thomas
Background Hematocrit (HCT) is known to be a confounding factor that interferes with many blood glucose (BG) measurement technologies, resulting in wrong readings. Dynamic electrochemistry has been identified as one possible way to correct for these potential deviations. The purpose of this laboratory investigation was to assess the HCT stability of four BG meters known to employ dynamic electrochemistry (BGStar and iBGStar, Sanofi; Wavesense Jazz, AgaMatrix; Wellion Linus, MedTrust) in comparison with three other devices (GlucoDock, Medisana; OneTouch Verio Pro, LifeScan; FreeStyle Freedom InsuLinx, Abbott-Medisense). Methods Venous heparinized blood was immediately aliquoted after draw and manipulated to contain three different BG concentrations (60–90, 130–160, and 280–320 mg/dl) and five different HCT levels (25%, 35%, 45%, 55%, and 60%). After careful oxygenation to normal blood oxygen pressure, each of the resulting 15 different samples was measured six times with three devices and three strip lots of each meter. The YSI Stat 2300 served as laboratory reference method. Stability to HCT influence was assumed when less than 10% difference occurred between the highest and lowest mean glucose deviations in relation to HCT concentrations [hematocrit interference factor (HIF)]. Results Five of the investigated self-test meters showed a stable performance with the different HCT levels tested in this investigation: BGStar (HIF 4.6%), iBGStar (6.6%), Wavesense Jazz (4.1%), Wellion Linus (8.5%), and OneTouch Verio Pro (6.2%). The two other meters were influenced by HCT (FreeStyle InsuLinx 17.8%; GlucoDock 46.5%). Conclusions In this study, meters employing dynamic electrochemistry, as used in the BGStar and iBGStar devices, were shown to correct for potential HCT influence on the meter results. Dynamic electrochemistry appears to be an effective way to handle this interfering condition. PMID:24351179
laboratory diagnostics in hematology, biochemistry, immunology, oncology, microbiology and metabolic diseases. A prominent consideration was given to value added to laboratory medicine in the frame of managed care and quality management. The Conference was accompanied by a laboratory equipment and reagents exhibition with the participation of nine companies, also sponsors of the conference. Three workshops pointed out the presence of the most well known of them: Abbott, Roche and Nobis.
Remco van Horssen
Full Text Available Objectives: Lactate is a major parameter in medical decision making. During labor, it is an indicator for fetal acidosis and immediate intervention. In the Emergency Department (ED, rapid analysis of lactate/blood gas is crucial for optimal patient care. Our objectives were to cross-compare-for the first time-two point-of-care testing (POCT lactate devices with routine laboratory results using novel tight precision targets and evaluate different lactate cut-off concentrations to predict metabolic acidosis. Design and methods: Blood samples from the delivery room (n=66 and from the ED (n=85 were analyzed on two POCT devices, the StatStrip-Lactate (Nova Biomedical and the iSTAT-1 (CG4+ cassettes, Abbott, and compared to the routine laboratory analyzer (ABL-735, Radiometer. Lactate concentrations were cross-compared between these analyzers. Results: The StatStrip correlated well with the ABL-735 (R=0.9737 and with the iSTAT-1 (R=0.9774 for lactate in umbilical cord blood. Lactate concentrations in ED samples measured on the iSTAT-1 and ABL-735 showed a correlation coefficient of R=0.9953. Analytical imprecision was excellent for lactate and pH, while for pO2 and pCO2 the coefficient of variation was relatively high using the iSTAT-1. Conclusion: Both POCT devices showed adequate analytical performance to measure lactate. The StatStrip can indicate metabolic acidosis in 1 Î¼l blood and will be implemented at the delivery room. Keywords: Lactate, Point-of-care testing, Blood gas, Fetal acidosis
... ecological research values at the Abbott Creek Research Natural Area. The withdrawal created by PLO No. 6856... the protection of the unique natural and ecological research values at the Abbott Creek Research...
Coley-Grant, Deon; Herbert, Mike; Cornes, Michael P; Barlow, Ian M; Ford, Clare; Gama, Rousseau
We studied the impact on reference intervals, classification of patients with hypoalbuminaemia and albumin infusion prescriptions on changing from a bromocresol green (BCG) to a bromocresol purple (BCP) serum albumin assay. Passing-Bablok regression analysis and Bland-Altman plot were used to compare Abbott BCP and Roche BCG methods. Linear regression analysis was used to compare in-house and an external laboratory Abbott BCP serum albumin results. Reference intervals for Abbott BCP serum albumin were derived in two different laboratories using pathology data from adult patients in primary care. Prescriptions for 20% albumin infusions were compared one year before and one year after changing the albumin method. Abbott BCP assay had a negative bias of approximately 6 g/L compared with Roche BCG method.There was good agreement (y = 1.04 x - 1.03; R(2 )= 0.9933) between in-house and external laboratory Abbott BCP results. Reference intervals for the serum albumin Abbott BCP assay were 31-45 g/L, different to those recommended by Pathology Harmony and the manufacturers (35-50 g/L). Following the change in method there was a large increase in the number of patients classified as hypoalbuminaemic using Pathology Harmony references intervals (32%) but not when retrospectively compared to locally derived reference intervals (16%) compared with the previous year (12%). The method change was associated with a 44.6% increase in albumin prescriptions. This equated to an annual increase in expenditure of £35,234. We suggest that serum albumin reference intervals be method specific to prevent misclassification of albumin status in patients. Change in albumin methodology may have significant impact on hospital resources. © The Author(s) 2015.
Smitz, J.; Schiettecatte, J.; Stierteghem, A.C. van; Jonckheer, M.H.
Four commerically available monoclonal immunoradiometric methods for the assay of TSH are tested. These four kits are: RIA-GNOST TSH code 0CPL of Behring, SUCROSEP TSH IRMA of Boots, TSH-IRMA-CT-100 of Medgenix, TSH-RIA bead II Ultrasensitive of Abbott. The accuracy, sensitivity and reproducibility of the four kits are compared. The methods are also clinically evaluated. The authors concluded that the kits of Abbott, Behring and Boots are suitable for use in the routine laboratory. 4 refs.; 4 figs.; 7 tabs
To determine the prevalence of vaginitis, syphilis and HIV infection in women in ... have a significant impact on infected people's health as well as on health care .... for both HIV-1 and H1V-2 antibodies using HIV 1/2 EIA. (Abbott Laboratories ...
Touw, D J; de Graaf, A I; de Goede, P
The limits of quantitation of the assay of tobramycin in serum by the fluorescence polarization immunoassay system marketed by Abbott Laboratories (TDxFLx system) are 0.1 and 10.0 mg/L. For some pharmacokinetic studies, however, a more sensitive analysis is needed. The sensitivity of the TDxFLx
...., 235 East 42nd 50 mg, 75 mg, and St., New York, NY 100 mg. 10017. NDA 020239 Kytril (granisetron... Kytril (granisetron Do. HCl) Tablets, EQ 1 mg (base), EQ 2 mg (base). NDA 020336 DynaCirc CR GlaxoSmith...) Abbott Laboratories. Capsules, 100 mg. NDA 021238 Kytril (granisetron Hoffman-La Roche, HCl) Oral...
Curry, Susan J; Byers, Tim; Hewitt, Maria Elizabeth
... competences and with regard for appropriate balance. Support for this project was provided by the National Cancer Institute; the Centers for Disease Control and Prevention; the American Cancer Society; Abbott Laboratories; the American Society of Clinical Oncology; Amgen, Inc.; Aventis; and the United Health Care Foundation. The views pre...
Principal Investigator: J. Michael Pierce, PhDInstitution: University of Georgia, Athens, GA Our project, Discovery and Development of Cancer Glycomarkers, is a joint collaboration between our laboratories at the CCRC, which include Karen Abbott, Lance Wells, Kevin Dobbin, and Mike Tiemeyer, those at TGen, in Phoenix, AZ, Daniel Von Hoff, Haiyong Han, and Mike Demeure, and
Rau, R.; Simianer, S.; Riel, P.L.C.M. van; Putte, L.B.A. van de; Kruger, K.; Schattenkirchner, M.; Allaart, C.F.; Breedveld, F.C.; Kempeni, J.; Beck, K.; Kupper, H.
OBJECTIVE: This randomized, placebo-controlled, double-blind, Phase 1 study assessed the magnitude, onset, and duration of response with intravenous (i.v.) and subcutaneous (s.c.) adalimumab (Humira, Abbott Laboratories) combined with methotrexate (MTX) in patients with active rheumatoid arthritis
Ulsen; J.van*; Michel; M.F.*; Strik; R.van*; Joost; T.H.van*; Stolz; E.*; Eijk; R.V.W.van
De Gonozyme test (Abbott Laboratories), een nieuwe enzyme immunoassay (EIA) voor het aantonen van Neisseria gonorrhoeae werd geevalueerd in een grote groep promiscue vrouwen. Als de EIA werd uitgevoerd met materiaal afkomstig van de cervix, bedroeg de prevalentie van gonorroe 8,2%. Vergeleken
Here are given the results of the firm BASF for the year 2000. The turnover has reached a historical record; these exceptional results are justified by the last sales and acquiring of the firm BASF and in particular by the cession of its pharmaceutical activities to the Abbott laboratories. (O.M.)
Pedersen, Stephen Wørlich; Clausen, Jesper Bøje; Gregerslund, Mie Manon
, (Duodopa®, Abbott Laboratories) is delivered continuously through a percutaneous endoscopic gastrostomy with the inner tube placed in the duodenum by means of a device (CADD legacy Duodopa pump (CE 0473)). The therapy implies continuous dopaminergic delivery directly to the duodenum and is therefore...
...-- Tulsa, OK 08/02/13 08/01/13 Wholesale Customer Application Support Team (Workers). 82953 Abbott Laboratories, Santa Clara, CA........ 08/02/13 08/01/13 including on-site leased workers from ATR Int'l (State...
... as the test product while another silymarin tablet brand, Silliver® (Abbott Laboratories Pak Ltd) was the reference product. The tablets were administered to healthy male volunteers orally at a dose of 200 mg following an overnight fast according to a randomized cross-over design. Scheduled blood samples were collected ...
Soroprevalência da hepatite B e avaliação da resposta imunológica à vacinação contra a hepatite B por via intramuscular e intradérmica em profissionais de um laboratório de saúde pública Hepatitis B seroprevalence and evaluation of immune response to hepatitis B vaccination using intramuscular and intradermal routes in public health laboratory employees
Regina Célia Moreira
using commercial kits (Abbott® Laboratories. Employees were submitted to the conventional three-dose vaccination by intramuscular route. To those employees who did not respond to intramuscular vaccination, 5 µg doses of Engerix® B were then administered by intradermal route up to nine doses. RESULTS: Four hundred and four healthcare workers were enrolled in this study. Initially, two (0.5% and 42 (10.4% were HBsAg and anti-HBs reagent, respectively. Among the 360 negative volunteers, 316 (87.8% received three vaccine doses and in 259 of them, serum samples were collected to evaluate vaccine efficacy. Among them, 242 (93.4% showed antibodies titer higher than 10 UI/l. Intradermal vaccination was carried out in five volunteers and all of them responded to this vaccine administration route. CONCLUSION: The prevalence of hepatitis B was not higher than in general population. Intradermal vaccine administration could be a good alternative in people that did not respond to previous intramuscular route.
Golchert, N. W.; ESH/QA Oversight
Argonne National Laboratory is a place where scientists and engineers come together to open up new possibilities for the future. Argonne has brought us many important projects in the past. It was at Argonne that researchers confirmed that Beethoven suffered from lead poisoning, and it was through the amazing Access Grid, pioneered at Argonne, that researchers in the United States were able to aid doctors on the other side of the world who were fighting the SARS outbreak. Researchers at Argonne are currently researching and developing new strategies in areas as varied as advanced nuclear reactors and other energy sources, medicine, and environmental science that will likely prove to be just as significant as Argonne's past achievements. Nuclear reactor development has been a priority at Argonne since its beginning. Argonne is very involved with the development of alternate strategies for safely treating and disposing of nuclear wastes. The first designs and prototypes of most of the nuclear reactors producing energy around the world today were originally conceived and tested by Argonne. While it may seem intimidating to live near a nuclear research site, the community surrounding Argonne is in no danger. The laboratory's Environmental Management Program provides Argonne's neighbors with quantitative risk data and has determined that the Argonne site is very safe. As a U.S. Department of Energy laboratory, Argonne has always been interested in finding new and more efficient energy sources. Current energy projects include fuel efficient cars, new batteries and fuel cells, and the conservation of U.S. oil and gas resources. The U.S. Department of Energy recently named Argonne the lead laboratory to test and evaluate new technologies for plug-in hybrid vehicles. Pharmaceutical companies use Argonne in their research, including a study discovering the structure of the HIV virus. Conducted at Argonne's Advanced Photon Source, this landmark research led Abbott Labs to
Oyaert, Matthijs; Van Maerken, Tom; Bridts, Silke; Van Loon, Silvi; Laverge, Heleen; Stove, Veronique
Point-of-care blood gas test results may benefit therapeutic decision making by their immediate impact on patient care. We evaluated the (pre-)analytical performance of a novel cartridge-type blood gas analyzer, the GEM Premier 5000 (Werfen), for the determination of pH, partial carbon dioxide pressure (pCO 2 ), partial oxygen pressure (pO 2 ), sodium (Na + ), potassium (K + ), chloride (Cl - ), ionized calcium ( i Ca 2+ ), glucose, lactate, and total hemoglobin (tHb). Total imprecision was estimated according to the CLSI EP5-A2 protocol. The estimated total error was calculated based on the mean of the range claimed by the manufacturer. Based on the CLSI EP9-A2 evaluation protocol, a method comparison with the Siemens RapidPoint 500 and Abbott i-STAT CG8+ was performed. Obtained data were compared against preset quality specifications. Interference of potential pre-analytical confounders on co-oximetry and electrolyte concentrations were studied. The analytical performance was acceptable for all parameters tested. Method comparison demonstrated good agreement to the RapidPoint 500 and i-STAT CG8+, except for some parameters (RapidPoint 500: pCO 2 , K + , lactate and tHb; i-STAT CG8+: pO 2 , Na + , i Ca 2+ and tHb) for which significant differences between analyzers were recorded. No interference of lipemia or methylene blue on CO-oximetry results was found. On the contrary, significant interference for benzalkonium and hemolysis on electrolyte measurements were found, for which the user is notified by an interferent specific flag. Identification of sample errors from pre-analytical sources, such as interferences and automatic corrective actions, along with the analytical performance, ease of use and low maintenance time of the instrument, makes the evaluated instrument a suitable blood gas analyzer for both POCT and laboratory use. Copyright © 2018 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
Abou El Hassan, Mohamed; Stoianov, Alexandra; Araújo, Petra A T; Sadeghieh, Tara; Chan, Man Khun; Chen, Yunqi; Randell, Edward; Nieuwesteeg, Michelle; Adeli, Khosrow
The CALIPER program has established a comprehensive database of pediatric reference intervals using largely the Abbott ARCHITECT biochemical assays. To expand clinical application of CALIPER reference standards, the present study is aimed at transferring CALIPER reference intervals from the Abbott ARCHITECT to Beckman Coulter AU assays. Transference of CALIPER reference intervals was performed based on the CLSI guidelines C28-A3 and EP9-A2. The new reference intervals were directly verified using up to 100 reference samples from the healthy CALIPER cohort. We found a strong correlation between Abbott ARCHITECT and Beckman Coulter AU biochemical assays, allowing the transference of the vast majority (94%; 30 out of 32 assays) of CALIPER reference intervals previously established using Abbott assays. Transferred reference intervals were, in general, similar to previously published CALIPER reference intervals, with some exceptions. Most of the transferred reference intervals were sex-specific and were verified using healthy reference samples from the CALIPER biobank based on CLSI criteria. It is important to note that the comparisons performed between the Abbott and Beckman Coulter assays make no assumptions as to assay accuracy or which system is more correct/accurate. The majority of CALIPER reference intervals were transferrable to Beckman Coulter AU assays, allowing the establishment of a new database of pediatric reference intervals. This further expands the utility of the CALIPER database to clinical laboratories using the AU assays; however, each laboratory should validate these intervals for their analytical platform and local population as recommended by the CLSI. Copyright © 2015 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
Higgins, Victoria; Truong, Dorothy; Woroch, Amy; Chan, Man Khun; Tahmasebi, Houman; Adeli, Khosrow
Evidence-based reference intervals (RIs) are essential to accurately interpret pediatric laboratory test results. To fill gaps in pediatric RIs, the Canadian Laboratory Initiative on Pediatric Reference Intervals (CALIPER) project developed an age- and sex-specific pediatric RI database based on healthy pediatric subjects. Originally established for Abbott ARCHITECT assays, CALIPER RIs were transferred to assays on Beckman, Roche, Siemens, and Ortho analytical platforms. This study provides transferred reference intervals for 29 biochemical assays for the Ortho VITROS 5600 Chemistry System (Ortho). Based on Clinical Laboratory Standards Institute (CLSI) guidelines, a method comparison analysis was performed by measuring approximately 200 patient serum samples using Abbott and Ortho assays. The equation of the line of best fit was calculated and the appropriateness of the linear model was assessed. This equation was used to transfer RIs from Abbott to Ortho assays. Transferred RIs were verified using 84 healthy pediatric serum samples from the CALIPER cohort. RIs for most chemistry analytes successfully transferred from Abbott to Ortho assays. Calcium and CO 2 did not meet statistical criteria for transference (r 2 reference intervals, 29 successfully verified with approximately 90% of results from reference samples falling within transferred confidence limits. Transferred RIs for total bilirubin, magnesium, and LDH did not meet verification criteria and are not reported. This study broadens the utility of the CALIPER pediatric RI database to laboratories using Ortho VITROS 5600 biochemical assays. Clinical laboratories should verify CALIPER reference intervals for their specific analytical platform and local population as recommended by CLSI. Copyright © 2018 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
For a company to maintain its competitive edge in today's global market every opportunity to gain an advantage must be exploited. Many companies are strategically focusing on improved utilization of existing equipment as well as regulatory compliance. Abbott Laboratories is no exception. Pharmaceutical companies such as Abbott Laboratories realize that reliability and availability of their production equipment is critical to be successful and competitive. Abbott Laboratories, like many of our competitors, is working to improve safety, minimize downtime and maximize the productivity and efficiency of key production equipment such as the pressure vessels utilized in our processes. The correct strategy in obtaining these objectives is to perform meaningful inspection with prioritization based on hazard analysis and risk. The inspection data gathered in Abbott Laboratories pressure vessel program allows informed decisions leading to improved process control. The results of the program are reduced risks to the corporation and employees when operating pressure retaining equipment. Accurate and meaningful inspection methods become the cornerstone of a program allowing proper preventative maintenance actions to occur. Successful preventative/predictive maintenance programs must utilize meaningful nondestructive evaluation techniques and inspection methods. Nondestructive examination methods require accurate useful tools that allow rapid inspection for the entire pressure vessel. Results from the examination must allow the owner to prove compliance of all applicable regulatory laws and codes. At Abbott Laboratories the use of advanced NDE techniques, primarily B-scan ultrasonics, has provided us with the proper tools allowing us to obtain our objectives. Abbott Laboratories uses B-scan ultrasonics utilizing a pulse echo pitch catch technique to provide essential data on our pressure vessels. Equipment downtime is reduced because the nondestructive examination usually takes
Federal Laboratory Consortium — The Bioassay Laboratory is an accredited laboratory capable of conducting standardized and innovative environmental testing in the area of aquatic ecotoxicology. The...
Federal Laboratory Consortium — Naval Academy Hydromechanics LaboratoryThe Naval Academy Hydromechanics Laboratory (NAHL) began operations in Rickover Hall in September 1976. The primary purpose of...
Kim, J.; Swantee, C.; Lee, B.; Gunning, H.; Chow, A.; Sidaway, F.; Sherlock, C.; Garceau, R.; Dimech, W.; Malloch, L.
In June 2005, a pilot program was implemented in Canadian laboratories to monitor the performance of the Abbott human immunodeficiency virus types 1 and 2 (HIV-1/2) gO enzyme immunoassay (EIA). Two different external quality control (QC) reagents and a “real-time” software analysis program were evaluated. In November 2005, higher-than-expected calibrator rate values in these kits were first reported at the Ontario Ministry of Health (Etobicoke), followed by the Alberta Provincial Public Healt...
Sanghvi, Ajit; Diven, Warren; Seltman, Howard; Starzl, Thomas
A new procedure for measuring cyclosporine in plasma has been introduced by Abbott Laboratories, involving their TDx instrumentation and fluorescence polarization immunoassay. Radioimmunoassay (RIA) and high-performance liquid chromatography are currently the conventional methods for measuring cyclosporine in plasma and whole blood. In an effort to find a method that will decrease the radioactive hazard, the reagent and supply cost, and the labor requirements associated with RIA procedures, w...
National Laboratories dealing with national security issues. He has served as a member of th, ,rtment of State Policy Planning Staff (1978-1981), and...together with Australia and New Zealand, of Malaysia when it con- fronted Indonesia); or bilateral (such as agreements or commitments for the...293., 5, See Cyril Falls, The Great War, 1914-1918 (New York: Putnam 1959), pp.. 227-230 for a brief account of the campaign. 6. See P, Abbott and D
Patel, Supen R
Abstract Introduction Sarcoidosis is an inflammatory disorder characterized by the presence of non-caseating granulomas in affected organs. The presence of CD4-positive T lymphocytes and macrophages in affected organs suggests an ongoing immune response. Systemic corticosteroids remain the mainstay of treatment, but therapy is often limited by adverse effects. This is the first report of the use of adalimumab (HUMIRA®, Abbott Laboratories, North Chicago, IL, USA), an anti-tumor necrosis facto...
glucose , hematocrit (Hct), hemoglobin (Hb), sodium (Na+), potassium (K+), and ionized calcium (iCa2+) using i-STAT cartridges ( Abbott Laboratories...animals were observed for 200 minutes. Blood chemistry and physiological parameters were recorded. Tissue samples from lung and small intestine were...seemingly acceptable medical therapy and surgical intervention.4 The first physiologic response to severe blood loss is activation of the neuroendocrine
temperature before measurements. A 1-mL aliquot was then pipetted into a kaolin tube to ini- tiate coagulation and 340-mL samples were pipetted into TEG...human ranges: PT, PTT, and fibrinogen ranges from Dade Behring. PLT count from Abbott Laboratories. Citrated Kaolin TEG values from Haemoscope. SONDEEN...crystalloid only (data courtesy of LTC R.S. Kotwal, MD, MPH). Military units operating in isolated, dispersed areas have been known to carry 2 to 4 units of
Piergies, A A; Sainati, S; Roth-Schechter, B
To determine in healthy volunteers (men and women; 18 to 40 years old) the potential cross-reactivity of Ambien (zolpidem) and/or its metabolites with drugs that are screened by the Syva EMIT II and the Abbott ADx urine drug screens assays. Open-label, fixed-treatment sequence of 1 night each of treatment with zolpidem (10 mg) and temazepam (15 mg). Clinical Pharmacology Unit within a teaching hospital. Over a 24-hour period, presence or absence of positive results on the Syva EMIT II or the Abbott ADx urine drug assay system, each performed at two different laboratory assay sites. Following ingestion of zolpidem, no subject had any positive response in either laboratory to the Syva EMIT II or the Abbott ADx urine drug screen assays at 0, 4, 8, 12, and 24 hours postdose. During the same time period, all subjects had measurable zolpidem plasma concentrations at 1.5 and 8 hours postdose, with mean concentrations of 115.2 ng/mL and 30.1 ng/mL, respectively (in agreement with its half-life of 2.5 hours). The positive response rate at 10 hours after ingestion of Restoril (temazepam) among the four laboratory/assay combinations ranged from 36.8% to 73.7%, a range that is within the reported response rates for these tests. These data indicate that zolpidem will not cross-react in standard urine drug screens with benzodiazepines, opiates, barbiturates, cocaine, cannabinoids, or amphetamines.
Salsamendi, Jason; Pereira, Keith; Baker, Reginald; Bhatia, Shivank S; Narayanan, Govindarajan
Transplant renal artery stenosis (TRAS) is a vascular complication frequently seen because of increase in the number of renal transplantations. Early diagnosis and management is essential to optimize a proper graft function. Currently, the endovascular treatment of TRAS using angioplasty and/or stenting is considered the treatment of choice with the advantage that it does not preclude subsequent surgical correction. Treatment of TRAS with the use of stents, particularly in tortuous transplant renal anatomy presents a unique challenge to an interventional radiologist. In this study, we present three cases from our practice highlighting the use of a balloon-expandable Multi-Link RX Ultra coronary stent system (Abbott Laboratories, Abbott Park, Illinois, USA) for treating high grade focal stenosis along very tortuous renal arterial segments. Cobalt-Chromium alloy stent scaffold provides excellent radial force, whereas the flexible stent design conforms to the vessel course allowing for optimal stent alignment.
Imam, Atique; Carter, Ranjana M. S.; Phillips-Hughes, Jane; Boardman, Philip; Uberoi, Raman
The StarClose device (Abbott Vascular Devices; Abbott Laboratories, Redwood City, CA) utilizes an externally placed Nitinol clip to achieve arterial closure following femoral artery puncture. The objectives of this study were to assess the efficacy and complications of the StarClose device in patients undergoing interventional radiological procedures. Preprocedural clotting status, pulse and blood pressure, severity of vessel calcification, sheath size, and time to deployment were recorded. Postdeployment complications immediately postprocedure, at 1 h, at 2 h, and at 1 week were recorded. A duplex scan was performed in the first 10 patients to assess any immediate vascular complications. Deployments were successful in 96% achieving immediate hemostasis. Mean deployment time was 48 s. There were no major complications. The StarClose device was found to have a high technical and clinical efficacy
De Backer, Ole; Luk, Ngai H V; Søndergaard, Lars
Following the success of transcatheter aortic and pulmonary valve implantation, there is a large interest in transcatheter mitral valve interventions to treat severe mitral regurgitation (MR). With the exception for the MitraClipTM (Abbott, Abbott Park, IL, USA) edge-to-edge leaflet plication...
Rands et Abbott,. 1964) et à la mosaïque (Abbott, 1961), deux maladies dues à un virus, ainsi qu'à la gommose. (Hughes, 1964) provoquée par une bactérie. En réponse à cette contrainte, des hybrides interspécifiques entre S.
Larsen, Kristian; Højbjerg, Karin
I professionslitteraturen beskrives profession overordnet som mere eller mindre autonome, faglige grupper, der i et historisk perspektiv udvikler sig i et samspil med samfundsmæssige institutioner samt i relationer til andre faggrupper Abbott 1988, Abbott 2005a og b, Carlhed 2011, Friedson, Weber...
.... Comments by e-mail are encouraged. FOR FURTHER INFORMATION CONTACT: Helene Abbott at (202) 287-1593 or via e-mail: firstname.lastname@example.org . SUPPLEMENTARY INFORMATION: Background Section-by-Section Analysis... the Idaho National Lab and has been in use since the late 1990s. The PIDS system is used by both DOE...
Abbott, B. P.; Abbott, R.; Abbott, T. D.; Abernathy, M. R.; Acernese, F.; Ackley, K.; Adams, C.; Phythian-Adams, A.T.; Addesso, P.; Adhikari, R. X.; Adya, V. B.; Affeldt, C.; Agathos, M.; Agatsuma, K.; Aggarwal, N.T.; Aguiar, O. D.; Aiello, L.; Ain, A.; Ajith, P.; Allen, B.; Allocca, A.; Altin, P. A.; Anderson, S. B.; Anderson, W. G.; Arai, K.; Araya, M. C.; Arceneaux, C. C.; Areeda, J. S.; Arnaud, N.; Arun, K. G.; Ascenzi, S.; Ashton, G.; Ast, M.; Aston, S. M.; Astone, P.; Aufmuth, P.; Aulbert, C.; Babak, S.; Bacon, P.; Bader, M. K. M.; Baker, P. T.; Baldaccini, F.; Ballardin, G.; Ballmer, S. W.; Barayoga, J. C.; Barclay, S. E.; Barish, B. C.; Barker, R.D.; Barone, F.; Barr, B.; Barsotti, L.; Barsuglia, M.; Barta, D.; Bartlett, J.; Bartos, I.; Bassiri, R.; Basti, A.; Batch, J. C.; Baune, C.; Bavigadda, V.; Bazzan, M.; Bejger, M.; Bell, A. S.; Berger, B. K.; Bergmann, G.; Berry, C. P. L.; Bersanetti, D.; Bertolini, A.; Betzwieser, J.; Bhagwat, S.; Bhandare, R.; Bilenko, I. A.; Billingsley, G.; Birch, M.J.; Birney, R.; Birnholtz, O.; Biscans, S.; Bisht, A.; Bitossi, M.; Biwer, C.; Bizouard, M. A.; Blackburn, J. K.; Blair, C. D.; Blair, D. G.; Blair, R. M.; Bloemen, A.L.S.; Bock, O.; Boer, M.; Bogaert, J.G.; Bogan, C.; Bohe, A.; Bond, T.C; Bondu, F.; Bonnand, R.; Boom, B. A.; Bork, R.; Boschi, V.; Bose, S.; Bouffanais, Y.; Bozzi, A.; Bradaschia, C.; Brady, P. R.; Braginsky, V. B.; Branchesi, M.; Brau, J. E.; Briant, T.; Brillet, A.; Brinkmann, M.; Brisson, V.; Brockill, P.; Broida, J. E.; Brooks, A. F.; Brown, A.D.; Brown, D.; Brown, N. M.; Brunett, S.; Buchanan, C. C.; Buikema, A.; Bulik, T.; Bulten, H. J.; Buonanno, A.; Buskulic, D.; Buy, C.; Byer, R. L.; Cabero, M.; Cadonati, L.; Cagnoli, G.; Cahillane, C.; Bustillo, J. Calderon; Callister, T. A.; Calloni, E.; Camp, J. B.; Cannon, K. C.; Cao, J.; Capano, C. D.; Capocasa, E.; Carbognani, F.; Caride, S.; Diaz, J. Casanueva; Casentini, J.; Caudill, S.; Cavaglia, M.; Cavalier, F.; Cavalieri, R.; Cella, G.; Cepeda, C. B.; Baiardi, L. Cerboni; Cerretani, G.; Cesarini, E.; Chamberlin, S. J.; Chan, M.; Chao, D. S.; Charlton, P.; Chassande-Mottin, E.; Cheeseboro, B. D.; Chen, H. Y.; Chen, Y; Cheng, C.; Chincarini, A.; Chiummo, A.; Cho, H. S.; Cho, M.; Chow, J. H.; Christensen, N.; Chu, Qian; Chua, S. E.; Chung, E.S.; Ciani, G.; Clara, F.; Clark, J. A.; Cleva, F.; Coccia, E.; Cohadon, P. -F.; Colla, A.; Collette, C. G.; Cominsky, L.; Constancio, M., Jr.; Conte, A.; Conti, L.; Cook, D.; Corbitt, T. R.; Cornish, N.; Corsi, A.; Cortese, S.; Costa, A.C.; Coughlin, M. W.; Coughlin, S. B.; Coulon, J. -P.; Countryman, S. T.; Couvares, P.; Cowan, E. E.; Coward, D. M.; Cowart, M. J.; Coyne, D. C.; Coyne, R.; Craig, K.; Creighton, J. D. E.; Cripe, J.; Crowder, S. G.; Cumming, A.; Cunningham, A.L.; Cuoco, E.; Dal Canton, T.; Danilishin, S. L.; D'Antonio, S.; Danzmann, K.; Darman, N. S.; Dasgupta, A.; Da Silva Costa, C. F.; Dattilo, V.; Dave, I.; Davier, M.; Davies, G. S.; Daw, E. J.; Day, R.; De, S.; Debra, D.; Debreczeni, G.; Degallaix, J.; De laurentis, M.; Deleglise, S.; Del Pozzo, W.; Denker, T.; Dent, T.; Dergachev, V.A.; Rosa, R.; DeRosa, R. T.; DeSalvo, R.; Devine, R. C.; Dhurandhar, S.; Diaz, M. C.; Di Fiore, L.; Giovanni, M.G.; Di Girolamo, T.; Di Lieto, A.; Di Pace, S.; Di Palma, I.; Di Virgilio, A.; Dolique, V.; Donovan, F.; Dooley, K. L.; Doravari, S.; Douglas, R.; Downes, T. P.; Drago, M.; Drever, R. W. P.; Driggers, J. C.; Ducrot, M.; Dwyer, S. E.; Edo, T. B.; Edwards, M. C.; Effler, A.; Eggenstein, H. -B.; Ehrens, P.; Eichholz, J.; Eikenberry, S. S.; Engels, W.; Essick, R. C.; Etienne, Z.; Etzel, T.; Evans, T. M.; Evans, T. M.; Everett, R.; Factourovich, M.; Fafone, V.; Fair, H.; Fairhurst, S.; Fan, X.M.; Fang, Q.; Farinon, S.; Farr, B.; Farr, W. M.; Fauchon-Jones, E. J.; Favata, M.; Fays, M.; Fehrmann, H.; Fejer, M. M.; Fenyvesi, E.; Ferrante, I.; Ferreira, E. C.; Ferrini, F.; Fidecaro, F.; Fiori, I.; Fiorucci, D.; Fisher, R. P.; Flaminio, R.; Fletcher, M; Fournier, J. -D.; Frasca, S.; Frasconi, F.; Frei, Z.; Freise, A.; Frey, R.; Frey, V.; Fritschel, P.; Frolov, V. V.; Fulda, P.; Fyffe, M.; Gabbard, H. A. G.; Gaebel, S. M.; Gair, J. R.; Gammaitoni, L.; Gaonkar, S. G.; Garufi, F.; Gaur, G.; Gehrels, N.; Gemme, G.; Geng, P.; Genin, E.; Gennai, A.; George, J.; Gergely, L.; Germain, V.; Ghosh, Abhirup; Ghosh, Archisman; Ghosh, S.; Giaime, J. A.; Giardina, K. D.; Giazotto, A.; Gill, K.P.; Glaefke, A.; Goetz, E.; Goetz, R.; Gondan, L.; Gonzalez, R.G.; Castro, J. M. Gonzalez; Gopakumar, A.; Gordon, N. A.; Gorodetsky, M. L.; Gossan, S. E.; Lee-Gosselin, M.; Gouaty, R.; Grado, A.; Graef, C.; Graff, P. B.; Granata, M.; Grant, A.; Gras, S.; Gray, C.M.; Greco, G.; Green, A. C.; Groot, P.; Grote, H.; Grunewald, S.; Guidi, G. M.; Guo, X.; Gupta, A.; Gupta, M. K.; Gushwa, K. E.; Gustafson, E. K.; Gustafson, R.; Hacker, J. J.; Buffoni-Hall, R.; Hall, E. D.; Hammond, G.L.; Haney, M.; Hanke, M. M.; Hanks, J.; Hanna, C.; Hannam, M. D.; Hanson, P.J.; Hardwick, T.; Harms, J.; Harry, G. M.; Harry, I. W.; Hart, M. J.; Hartman, M. T.; Haster, C. -J.; Haughian, K.; Healy, J.; Heidmann, A.; Heintze, M. C.; Heitmann, H.; Hello, P.; Hemming, G.; Hendry, M.; Heng, I. S.; Hennig, J.; Henry, J.A.; Heptonstall, A. W.; Heurs, M.; Hild, S.; Hoak, D.; Hofman, D.; Holt, K.; Holz, D. E.; Hopkins, P.; Hough, J.; Houston, E. A.; Howell, E. J.; Hu, Y. M.; Huang, S.; Huerta, E. A.; Huet, D.; Hughey, B.; Husa, S.; Huttner, S. H.; Huynh-Dinh, T.; Indik, N.; Ingram, D. R.; Inta, R.; Isa, H. N.; Isac, J. -M.; Isi, M.; Isogai, T.; Iyer, B. R.; Izumi, K.; Jacqmin, T.; Jang, D.H.; Jani, K.; Jaranowski, P.; Jawahar, S.; Jian, L.; Jimenez-Forteza, F.; Johnson, W.; Johnson-McDaniel, N. K.; Jones, I.D.; Jones, R.; Jonker, R. J. G.; Ju, L.; Haris, K.; Kalaghatgi, C. V.; Kalogera, V.; Kandhasamy, S.; Kang, G.H.; Kanner, J. B.; Kapadia, S. J.; Karki, S.; Karvinen, K. S.; Kasprzack, M.; Katsavounidis, E.; Katzman, W.; Kaufer, S.; Kaur, T.; Kawabe, K.; Kefelian, F.; Kehl, M. S.; Keitel, D.; Kelley, D. B.; Kells, W.; Kennedy, R.E.; Key, J. S.; Khalili, F. Y.; Khan, I.; Khan., S.; Khan, Z.; Khazanov, E. A.; Kijbunchoo, N.; Kim, Chi-Woong; Kim, Chunglee; Kim, J.; Kim, K.; Kim, Namjun; Kim, W.; Kim, Y.M.; Kimbrell, S. J.; King, E. J.; King, P. J.; Kissel, J. S.; Klein, B.; Kleybolte, L.; Klimenko, S.; Koehlenbeck, S. M.; Koley, S.; Kondrashov, V.; Kontos, A.; Korobko, M.; Korth, W. Z.; Kowalska, I.; Kozak, D. B.; Kringel, V.; Krolak, A.; Krueger, C.; Kuehn, G.; Kumar, P.; Kumar, R.; Kuo, L.; Kutynia, A.; Lackey, B. D.; Landry, M.; Lange, J.; Lantz, B.; Lasky, P. D.; Laxen, M.; Lazzarini, A.; Lazzaro, C.; Leaci, P.; Leavey, S.; Lebigot, E. O.; Lee, C.H.; Lee, K.H.; Lee, M.H.; Lee, K.; Lenon, A.; Leonardi, M.; Leong, J. R.; Leroy, N.; Letendre, N.; Levin, Y.; Lewis, J. B.; Li, T. G. F.; Libson, A.; Littenberg, T. B.; Lockerbie, N. A.; Lombardi, A. L.; London, L. T.; Lord, J. E.; Lorenzini, M.; Loriette, V.; Lormand, M.; Losurdo, G.; Lough, J. D.; Lousto, C. O.; Lovelace, G.; Lueck, H.; Lundgren, A. P.; Lynch, R.; Ma, Y.; Machenschalk, B.; MacInnis, M.; Macleod, D. M.; Magana-Sandoval, F.; Zertuche, L. Magana; Magee, R. M.; Majorana, E.; Maksimovic, I.; Malvezzi, V.; Man, N.; Mandic, V.; Mangano, V.; Mansell, G. L.; Manske, M.; Mantovani, M.; Marchesoni, F.; Marion, F.; Marka, S.; Markosyan, A. S.; Maros, E.; Martelli, F.; Martellini, L.; Martin, I. W.; Martynov, D. V.; Marx, J. N.; Mason, K.; Masserot, A.; Massinger, T. J.; Masso-Reid, M.; Mastrogiovanni, S.; Matichard, F.; Matone, L.; Mavalvala, N.; Mazumder, N.; McCarthy, R.; McClelland, D. E.; McCormick, S.; McGuire, S. C.; McIntyre, G.; McIver, J.; McManus, D. J.; McRae, T.; McWilliams, S. T.; Meacher, D.; Meadors, G. D.; Meidam, J.; Melatos, A.; Mendell, G.; Mercer, R. A.; Merilh, E. L.; Merzougui, M.; Meshkov, S.; Messenger, C.; Messick, C.; Metzdorff, R.; Meyers, P. M.; Mezzani, F.; Miao, H.; Michel, C.; Middleton, H.; Mikhailov, E. E.; Milano, L.; Miller, A. L.; Miller, A. L.; Miller, B.; Miller, J.; Millhouse, M.; Minenkov, Y.; Ming, J.; Mirshekari, S.; Mishra, C.; Mitra, S.; Mitrofanov, V. P.; Mitselmakher, G.; Mittleman, R.; Moggi, A.; Mohan, M.; Mohapatra, S. R. P.; Montani, M.; Moore, B.C.; Moore, J.C.; Moraru, D.; Gutierrez Moreno, M.; Morriss, S. R.; Mossavi, K.; Mours, B.; Mow-Lowry, C. M.; Mueller, G.; Muir, A. W.; Mukherjee, Arunava; Mukherjee, S.D.; Mukherjee, S.; Mukund, N.; Mullavey, A.; Munch, J.; Murphy, D. J.; Murray, P.G.; Mytidis, A.; Nardecchia, I.; Naticchioni, L.; Nayak, R. K.; Nedkova, K.; Nelemans, G.; Nelson, T. J. N.; Gutierrez-Neri, M.; Neunzert, A.; Newton-Howes, G.; Nguyen, T. T.; Nielsen, A. B.; Nissanke, S.; Nitz, A.; Nocera, F.; Nolting, D.; Normandin, M. E. N.; Nuttall, L. K.; Oberling, J.; Ochsner, E.; O'Dell, J.; Oelker, E.; Ogin, G. H.; Oh, J.; Oh, S. H.; Ohme, F.; Oliver, M. B.; Oppermann, P.; Oram, Richard J.; O'Reilly, B.; O'Shaughnessy, R.; Ottaway, D. J.; Overmier, H.; Owen, B. J.; Pai, A.; Pai, S. A.; Palamos, J. R.; Palashov, O.; Palomba, C.; Pal-Singh, A.; Pan, H.; Pankow, C.; Pannarale, F.; Pant, B. C.; Paoletti, F.; Paoli, A.; Papa, M. A.; Paris, H. R.; Parker, W.S; Pascucci, D.; Pasqualetti, A.; Passaquieti, R.; Passuello, D.; Patricelli, B.; Patrick, Z.; Pearlstone, B. L.; Pedraza, M.; Pedurand, R.; Pekowsky, L.; Pele, A.; Penn, S.; Perreca, A.; Perri, L. M.; Pfeiffer, H. P.; Phelps, M.; Piccinni, O. J.; Pichot, M.; Piergiovanni, F.; Pierro, V.; Pillant, G.; Pinard, L.; Pinto, I. M.; Pitkin, M.; Poe, M.; Poggiani, R.; Popolizio, P.; Post, A.; Powell, J.; Prasad, J.; Predoi, V.; Prestegard, T.; Price, L. R.; Prijatelj, M.; Principe, M.; Privitera, S.; Prix, R.; Prodi, G. A.; Prokhorov, L. G.; Puncken, O.; Punturo, M.; Puppo, P.; Puerrer, M.; Qi, H.; Qin, J.; Qiu, S.; Quetschke, V.; Quintero, E. A.; Quitzow-James, R.; Raab, F. J.; Rabeling, D. S.; Radkins, H.; Raffai, P.; Raja, S.; Rajan, C.; Rakhmanov, M.; Rapagnani, P.; Raymond, V.; Razzano, M.; Re, V.; Read, J.; Reed, C. M.; Regimbau, T.; Rei, L.; Reid, S.; Reitze, D. H.; Rew, H.; Reyes, S. D.; Ricci, F.; Riles, K.; Rizzo, D.M.; Robertson, N. A.; Robie, R.; Robinet, F.; Rocchi, A.; Rolland, L.; Rollins, J. G.; Roma, V. J.; Romano, R.; Romanov, G.; Romie, J. H.; Rosinska, D.; Rowan, S.; Ruediger, A.; Ruggi, P.; Ryan, K.A.; Sachdev, P.S.; Sadecki, T.; Sadeghian, L.; Sakellariadou, M.; Salconi, L.; Saleem, M.; Salemi, F.; Samajdar, A.; Sammut, L.; Sanchez, E. J.; Sandberg, V.; Sandeen, B.; Sanders, J. R.; Sassolas, B.; Sathyaprakash, B. S.; Saulson, P. R.; Sauter, O. E. S.; Savage, R. L.; Sawadsky, A.; Schale, P.; Schilling, R.; Schmidt, J; Schmidt, P.; Schnabel, R.B.; Schofield, R. M. S.; Schoebeck, A.; Schreiber, K.E.C.; Schuette, D.; Schutz, B. F.; Scott, J.; Scott, M.S.; Sellers, D.; Sengupta, A. S.; Sentenac, D.; Sequino, V.; Sergeev, A.; Setyawati, Y.; Shaddock, D. A.; Shaffer, T. J.; Shahriar, M. S.; Shaltev, M.; Shapiro, B.; Shawhan, P.; Sheperd, A.; Shoemaker, D. H.; Shoemaker, D. M.; Siellez, K.; Siemens, X.; Sieniawska, M.; Sigg, D.; Silva, António Dias da; Singer, A; Singer, L. P.; Singh, A.; Singh, R.; Singhal, A.; Sintes, A. M.; Slagmolen, B. J. J.; Smith, R. J. E.; Smith, N.D.; Smith, R. J. E.; Son, E. J.; Sorazu, B.; Sorrentino, F.; Souradeep, T.; Srivastava, A. K.; Staley, A.; Steinke, M.; Steinlechner, J.; Steinlechner, S.; Steinmeyer, D.; Stephens, B. C.; Stevenson-Moore, P.; Stone, J.R.; Strain, K. A.; Straniero, N.; Stratta, G.; Strauss, N. A.; Strigin, S. E.; Sturani, R.; Stuver, A. L.; Summerscales, T. Z.; Sun, L.; Sunil, S.; Sutton, P. J.; Swinkels, B. L.; Szczepanczyk, M. J.; Tacca, M.D.; Talukder, D.; Tanner, D. B.; Tapai, M.; Tarabrin, S. P.; Taracchini, A.; Taylor, W.R.; Theeg, T.; Thirugnanasambandam, M. P.; Thomas, E. G.; Thomas, M.; Thomas, P.; Thorne, K. A.; Thorne, K. S.; Thrane, E.; Tiwari, S.; Tiwari, V.; Tokmakov, K. V.; Toland, K.; Tomlinson, C.; Tonelli, M.; Tornasi, Z.; Torres, C. V.; Torrie, C. I.; Toyra, D.; Travasso, F.; Traylor, G.; Trifiro, D.; Tringali, M. C.; Trozzo, L.; Tse, M.; Turconi, M.; Tuyenbayev, D.; Ugolini, D.; Unnikrishnan, C. S.; Urban, A. L.; Usman, S. A.; Vahlbruch, H.; Vajente, G.; Valdes, G.; Vallisneri, M.; van Bakel, N.; van Beuzekom, M.G.; van den Brand, J. F. J.; Van Den Broeck, C.F.F.; Vander-Hyde, D. C.; van der Schaaf, L.; van der Sluys, M. V.; van Heijningen, J. V.; Vano-Vinuales, A.; van Veggel, A. A.; Vardaro, M.; Vass, S.; Vasuth, M.; Vaulin, R.; Vecchio, A.; Vedovato, G.; Veitch, J.; Veitch, P.J.; Venkateswara, K.; Verkindt, D.; Vetrano, F.; Vicere, A.; Vinciguerra, S.; Vine, D. J.; Vinet, J. -Y.; Vitale, S.; Vo, T.; Vocca, H.; Vorvick, C.; Voss, D. V.; Vousden, W. D.; Vyatchanin, S. P.; Wade, A. R.; Wade, L. E.; Wade, MT; Walker, M.; Wallace, L.; Walsh, S.; Wang, G.; Wang, H.; Wang, M.; Wang, X.; Wang, Y.; Ward, R. L.; Warner, J.; Was, M.; Weaver, B.; Wei, L. -W.; Weinert, M.; Weinstein, A. J.; Weiss, R.; Wen, L.M.; Wessels, P.; Westphal, T.; Wette, K.; Whelan, J. T.; Whiting, B. F.; Williams, D.R.; Williamson, A. R.; Willis, J. L.; Willke, B.; Wimmer, M. H.; Winkler, W.; Wipf, C. C.; Wittel, H.; Woan, G.; Woehler, J.; Worden, J.; Wright, J.L.; Wu, D.S.; Wu, G.; Yablon, J.; Yam, W.; Yamamoto, H.; Yancey, C. C.; Yu, H.; Yvert, M.; Zadrozny, A.; Zangrando, L.; Zanolin, M.; Zendri, J. -P.; Zevin, M.; Zhang, L.; Zhang, M.; Zhang, Y.; Zhao, C.; Zhou, M.; Zhou, Z.; Zhu, X. J.; Zucker, M. E.; Zuraw, S. E.; Zweizig, J.; Boyle, M.; Bruegmann, B.; Campanelli, M.; Chu, I.W.T.; Clark, M.; de Haas, R.; Hemberger, D.; Hinder, I.; Kidder, L. E.; Kinsey, M.; Laguna, P.; Ossokine, S.; Pan, Y.; Roever, C.; Scheel, M.; Szilagyi, B.; Teukolsky, S.; Zlochower, Y.
This paper presents updated estimates of source parameters for GW150914, a binary black-hole coalescence event detected by the Laser Interferometer Gravitational-wave Observatory (LIGO) in 2015 [Abbott et al. Phys. Rev. Lett. 116, 061102 (2016).]. Abbott et al. [Phys. Rev. Lett. 116, 241102 (2016).
Federal Laboratory Consortium — Purpose:The Photometrics Laboratory provides the capability to measure, analyze and characterize radiometric and photometric properties of light sources and filters,...
Federal Laboratory Consortium — FUNCTION: Enables evaluation and characterization of materials ranging from the ultraviolet to the longwave infrared (LWIR).DESCRIPTION: The Blackroom Laboratory is...
Lee, Graham R; Griffin, Alison; Halton, Kieran; Fitzgibbon, Maria C
When laboratory Reference Ranges (RR) do not reflect analytical methodology, result interpretation can cause misclassification of patients and inappropriate management. This can be mitigated by determining and implementing method-specific RRs, which was the main objective of this study. Serum was obtained from healthy volunteers (Male + Female, n > 120) attending hospital health-check sessions during June and July 2011. Pseudo-anonymised aliquots were stored (at - 70 °C) prior t° analysis on Abbott ARCHITECT c16000 chemistry and i 2000SR immunoassay analysers. Data were stratified by gender where appropriate. Outliers were excluded statistically (Tukey method) to generate non-parametric RRs (2.5th + 97.5th percentiles). RRs were compared to those quoted by Abbott and UK Pathology Harmony (PH) where possible. For 7 selected tests, RRs were verified using a data mining approach. For chemistry tests (n = 23), Upper or Lower Reference Limits (LRL or URL) were > 20% different from Abbott ranges in 25% of tests (11% from PH ranges) but in 38% for immunoassay tests (n = 13). RRs (mmol/L) for sodium (138-144), potassium (3.8-4.9) and chloride (102-110) were considerably narrower than PH ranges (133-146, 3.5-5.0 and 95-108, respectively). The gender difference for ferritin (M: 29-441, F: 8-193 ng/mL) was more pronounced than reported by Abbott (M: 22-275, F: 5-204 ng/mL). Verification studies showed good agreement for chemistry tests (mean [SD] difference = 0.4% [1.2%]) but less so for immunoassay tests (27% [29%]), particularly for TSH (LRL). Where resource permits, we advocate using method-specific RRs in preference to other sources, particularly where method bias and lack of standardisation limits RR transferability and harmonisation.
Federal Laboratory Consortium — This laboratory contains a number of commercial off-the-shelf and in-house software packages allowing for both statistical analysis as well as mathematical modeling...
The foundation of a 'National Laboratory' which would support a Research center in synchrotron radiation applications is proposed. The essential features of such a laboratory differing of others centers in Brazil are presented. (L.C.) [pt
Federal Laboratory Consortium — The Geomechanics Laboratory allows its users to measure rock properties under a wide range of simulated service conditions up to very high pressures and complex load...
Herrera, Joshua M. [Sandia National Lab. (SNL-NM), Albuquerque, NM (United States)
This report is an analysis of the means of egress and life safety requirements for the laboratory building. The building is located at Sandia National Laboratories (SNL) in Albuquerque, NM. The report includes a prescriptive-based analysis as well as a performance-based analysis. Following the analysis are appendices which contain maps of the laboratory building used throughout the analysis. The top of all the maps is assumed to be north.
Federal Laboratory Consortium — Purpose: To conduct fundamental studies of highway materials aimed at understanding both failure mechanisms and superior performance. New standard test methods are...
Federal Laboratory Consortium — The NWFSC conducts critical fisheries science research at its headquarters in Seattle, WA and at five research stations throughout Washington and Oregon. The unique...
Federal Laboratory Consortium — The Dynamics Lab replicates vibration environments for every Navy platform. Testing performed includes: Flight Clearance, Component Improvement, Qualification, Life...
Federal Laboratory Consortium — This facility provides testing stations for computer-based assessment of cognitive and behavioral Warfighter performance. This 500 square foot configurable space can...
Federal Laboratory Consortium — FUNCTION: Evaluates and improves the operational effectiveness of existing and emerging electronic warfare systems. By analyzing and visualizing simulation results...
Federal Laboratory Consortium — The Analytical Labspecializes in Oil and Hydraulic Fluid Analysis, Identification of Unknown Materials, Engineering Investigations, Qualification Testing (to support...
Federal Laboratory Consortium — The Propulsion Lab simulates field test conditions in a controlled environment, using standardized or customized test procedures. The Propulsion Lab's 11 cells can...
... Medical Devices Radiation-Emitting Products Vaccines, Blood & Biologics Animal & ... What are lab tests? Laboratory tests are medical devices that are intended for use on samples of blood, urine, or other tissues ...
Federal Laboratory Consortium — FUNCTION: Provides an environment and facilities for auditory display research. A primary focus is the performance use of binaurally rendered 3D sound in conjunction...
Ephraim, D.C.; Pednekar, A.R.
A target laboratory to make stripper foils for the accelerator and various targets for use in the experiments is set up in the pelletron accelerator facility. The facilities available in the laboratory are: (1) D.C. glow discharge setup, (2) carbon arc set up, and (3) vacuum evaporation set up (resistance heating), electron beam source, rolling mill - all for target preparation. They are described. Centrifugal deposition technique is used for target preparation. (author). 3 figs
Federal Laboratory Consortium — The Semiconductor Electrical Measurements Laboratory is a research laboratory which complements the Optical Measurements Laboratory. The laboratory provides for Hall...
This report from the Dutch Ministry of Health is an advisory document concerned with isotope laboratories in hospitals, in connection with the Dutch laws for hospitals. It discusses which hospitals should have isotope laboratories and concludes that as many hospitals as possible should have small laboratories so that emergency cases can be dealt with. It divides the Netherlands into regions and suggests which hospitals should have these facilities. The questions of how big each lab. is to be, what equipment each has, how each lab. is organised, what therapeutic and diagnostic work should be carried out by each, etc. are discussed. The answers are provided by reports from working groups for in vivo diagnostics, in vitro diagnostics, therapy, and safety and their results form the criteria for the licences of isotope labs. The results of a questionnaire for isotope labs. already in the Netherlands are presented, and their activities outlined. (C.F.)
The paper concerns the work of the Kingsbury Laboratories of Fairey Engineering Company, for the nuclear industry. The services provided include: monitoring of nuclear graphite machining, specialist welding, non-destructive testing, and metallurgy testing; and all are briefly described. (U.K.)
Pannus, Pieter; Claus, Maarten; Gonzalez, Maria Mercedes Perez; Ford, Nathan; Fransen, Katrien
The use of dried blood spots (DBS) instead of plasma as a specimen type for HIV-1 viral load (VL) testing facilitates the decentralization of specimen collection and can increase access to VL testing in resource-limited settings. The performance of DBS for VL testing is lower, however, when compared to the gold standard sample type plasma. In this diagnostic accuracy study, we evaluated 3 VL assays with DBS.Participants were recruited between August 2012 and April 2015. Both plasma and DBS specimens were prepared and tested for HIV-1 VL with the Roche CAP/CTM HIV-1 test v2.0, the Abbott RealTime HIV-1, and the bioMérieux NucliSENS EasyQ HIV-1 v2.0. Sensitivity and specificity to detect treatment failure at a threshold of 1000 cps/mL with DBS were determined.A total of 272 HIV-positive patients and 51 HIV-negative people were recruited in the study. The mean difference or bias between plasma and DBS VL was 25% of the specimens differed by >0.5 log cps/mL.All 3 assays had comparable sensitivities around 80% and specificities around 90%. Upward misclassification rates were around 10%, but downward misclassification rates ranged from 20.3% to 23.6%. Differences in between assays were not statistically significant (P > 0.1).The 3 VL assays evaluated had suboptimal performance with DBS but still performed better than immunological or clinical monitoring. Even after the introduction of the much-anticipated point-of-care VL devices, it is expected that DBS will remain important as a complementary option for supporting access to VL monitoring, particularly in rural, resource-limited settings. Manufacturers should accelerate efforts to develop more reliable, sensitive and specific methods to test VL on DBS specimens.
Federal Laboratory Consortium — The Saxton Transportation Operations Laboratory (Saxton Laboratory) is a state-of-the-art facility for conducting transportation operations research. The laboratory...
Our task is to design mined-repository systems that will adequately secure high-level nuclear waste for at least 10,000 yr and that will be mechanically stable for 50 to 100-yr periods of retrievability during which mistakes could be corrected and a valuable source of energy could be reclaimed, should national policy on the reprocessing of spent fuel ever change. The only credible path for the escape of radionuclides from the repository to the biosphere is through ground-water, and in hard rock, bulk permeability is largely governed by natural and artificial fracture systems. Catastrophic failure of an excavation in hard rock is likely to occur at the weakest links - the discontinuities in the rock mass that is perturbed first by mining and then by radiogenic heating. The laboratory can contribute precise measurements of the pertinent thermomechanical, hydrological and chemical properties and improve our understanding of the fundamental processes through careful experiments under well controlled conditions that simulate the prototype environment. Thus laboratory investigations are necessary, but they are not sufficient, for conventional sample sizes are small relative to natural defects like joints - i.e., the rock mass is not a continuum - and test durations are short compared to those that predictive modeling must take into account. Laboratory investigators can contribute substantially more useful data if they are provided facilities for testing large specimens(say one cubic meter) and for creep testing of all candidate host rocks. Even so, extrapolations of laboratory data to the field in neither space nor time are valid without the firm theoretical foundations yet to be built. Meanwhile in-situ measurements of structure-sensitive physical properties and access to direct observations of rock-mass character will be absolutely necessary
The report contains summaries of work carried out under the following headings: fusion research experiments; U.K. contribution to the JET project; supporting studies; theoretical plasma physics, computational physics and computing; fusion reactor studies; engineering and technology; contract research; external relations; staff, finance and services. Appendices cover main characteristics of Culham fusion experiments, staff, extra-mural projects supported by Culham Laboratory, and a list of papers written by Culham staff. (U.K.)
Seamster, A.G.; Weitkamp, W.G.
The lead plating of the prototype resonator has been conducted entirely in the plating laboratory at SUNY Stony Brook. Because of the considerable cost and inconvenience in transporting personnel and materials to and from Stony Brook, it is clearly impractical to plate all the resonators there. Furthermore, the high-beta resonator cannot be accommodated at Stony Brook without modifying the set up there. Consequently the authors are constructing a plating lab in-house
Bettini, A., E-mail: Bettini@pd.infn.i [Padua University and INFN Section, Dipartimento di Fisca G. Galilei, Via Marzolo 8, 35131 Padova (Italy); Laboratorio Subterraneo de Canfranc, Plaza Ayuntamiento n1 2piso, Canfranc (Huesca) (Spain)
Underground laboratories provide the low radioactive background environment necessary to frontier experiments in particle and nuclear astrophysics and other disciplines, geology and biology, that can profit of their unique characteristics. The cosmic silence allows to explore the highest energy scales that cannot be reached with accelerators by searching for extremely rare phenomena. I will briefly review the facilities that are operational or in an advanced status of approval around the world.
Kilgas, Matthew A; Elmer, Steven J
We implemented a team-based activity in our exercise physiology teaching laboratory that was inspired from Abbott et al.'s classic 1952 Journal of Physiology paper titled "The physiological cost of negative work." Abbott et al. connected two bicycles via one chain. One person cycled forward (muscle shortening contractions, positive work) while the other resisted the reverse moving pedals (muscle lengthening contractions, negative work), and the cost of work was compared. This study was the first to link human whole body energetics with isolated muscle force-velocity characteristics. The laboratory activity for our students ( n = 35) was designed to reenact Abbott et al.'s experiment, integrate previously learned techniques, and illustrate differences in physiological responses to muscle shortening and lengthening contractions. Students (11-12 students/laboratory section) were split into two teams (positive work vs. negative work). One student from each team volunteered to cycle against the other for ~10 min. The remaining students in each team were tasked with measuring: 1 ) O 2 consumption, 2 ) heart rate, 3 ) blood lactate, and 4 ) perceived exertion. Students discovered that O 2 consumption during negative work was about one-half that of positive work and all other physiological parameters were also substantially lower. Muscle lengthening contractions were discussed and applied to rehabilitation and sport training. The majority of students (>90%) agreed or strongly agreed that they stayed engaged during the activity and it improved their understanding of exercise physiology. All students recommended the activity be performed again. This activity was engaging, emphasized teamwork, yielded clear results, was well received, and preserved the history of classic physiological experiments. Copyright © 2017 the American Physiological Society.
Donea-Debroise, B; Brocteur, J; Andre, A; Remacle, M B [Liege Univ. (Belgium)
A radioimmunoassay determination of the HBs antigen subtypes is discussed, this simple but effective technique was used in association with the use of the Austria II kit (Abbott Laboratories). This method consists of an inhibition reaction of the Austria II test, by previous incubation of the antigen to be subtyped with a monospecific antibody. With this method we were able to distinguish the y and the d antigens as well as the w1, w3, w4 determinants of hepatitis B surface antigen. We have included a frequency table of the various HBs subtypes found among donor and patient populations in Liege.
Hui, Z.; Coulepis, A.G.; Gust, I.D.
The sensitivity and specificity of two commercially available radioimmunosay tests (Austria II-125, Abbott Laboratories; and International CIS, Commissariat Alenergie Atomique-Oris Laboratoire des Produits Biomedicaux) and a reverse passive haemagglutination test (Hepatest, Wellcome) for detection of hepatitis B surface antigen were evaluated using the Australian hepatitis B reference panel of 25 sera, and a panel of 257 sera collected from patients with acute hepatitis B, chronic carriers of hepatitis B surface antigen and two populations in which hepatitis B virus infection is known to be endemic. The three techniques were found to be generally comparable in sensitivity and specificity. The advantages and disadvantages of each method are discussed
The 32nd National Medicinal Chemistry Symposium, held in Minneapolis, MN, USA, included topics covering new developments in the field of medicinal chemistry. This conference report highlights selected presentations on NR2B subtype-selective NMDA receptor antagonists from Merck; selective neuronal nitric oxide synthase inhibitors from Northwestern University; novel GPR119 agonists, suchas GSK-1292263A (GlaxoSmithKline plc), PSN-821 ((OSI) Prosidion) and MBX-2982 (Metabolex Inc); a small-molecule Bcl inhibitor,navitoclax (Abbott Laboratories); and p53-targeting agents from sanofi-aventis and Ascenta Therapeutics Inc, including AT-219.
Donea-Debroise, B.; Brocteur, J.; Andre, A.; Remacle, M.B.
A radioimmunoassay determination of the HBs antigen subtypes is discussed, this simple but effective technique was used in association with the use of the Austria II kit (Abbott Laboratories). This method consists of an inhibition reaction of the Austria II test, by previous incubation of the antigen to be subtyped with a monospecific antibody. With this method we were able to distinguish the y and the d antigens as well as the w1, w3, w4 determinants of hepatitis B surface antigen. We have included a frequency table of the various HBs subtypes found among donor and patient populations in Liege
This series of papers extends the line-driven wind theory of Castor, Abbott, and Klein (CAK). The present paper develops a physical interpretation of line-driven flows using analytic methods. Numerical results will follow in two subsequent papers
... is necessary to continue protection of the unique natural and ecological research values of the... extension to continue protection of the unique natural and ecological research values at the Abbott Creek...
Objective: This study was to determine eligibility for antiretroviral therapy in resource-limited settings using total lymphocyte .... ART until CD4+ T cell counts fall below 200 cells/mm3 ... (Abbott Cell Dyne Operators manual) were checked for.
Conclusion: Both GC Tooth Mousse (Recaldent and Toothmin Tooth cream (Abbott Healthcare Pvt.Ltd increase the microhardness of bleached enamel. Toothmin tooth cream is a better agent for increasing microhardness, although difference is not significant.
.... Patent No. 5,262,035) from Abbott Diabetes Care Inc., and the Patent and Trademark Office requested FDA's... diabetes mellitus for the purpose of improving diabetes management. Subsequent to this approval, the Patent...
isolated from blood cultures in suspected cases of. lE. Hpwever, his .... dextrin -, maltose +, sucrose -, catalase +,glucose +, mannitol-, trehalose -, nitrate ... by Abbott.9 Whether in acquired or congenital heart disease, the bacteraemia of lE is ...
Tropical Journal of Pharmaceutical Research January 2015; 14 (1): 103-109. ISSN: 1596-5996 (print); .... males were fed with 10 % glucose solution soaked on cotton pad, ... mortality using Abbott's formula . Results from all replicates for ...
Tropical Journal of Pharmaceutical Research January 2015; 14 (1): 111-116 ... Purpose: To determine larvicidal activity of the essential oil derived from .... males were fed with 10 % glucose solution ... mortality using Abbott's formula .
Tropical Journal of Pharmaceutical Research October 2013; 12 (5): 799-804. ISSN: 1596-5996 (print); 1596-9827 .... Japan, Laos, Malaysia, Myanmar, Thailand,. Vietnam . .... mortality using Abbott's formula . Results from all replicates ...
Tropical Journal of Pharmaceutical Research May 2015; 14(5): 831-836. ISSN: 1596-5996 (print); ... Malaysia, Myanmar, Russia, Thailand, Vietnam, and Australia . .... mortality using Abbott's formula . Results from all replicates for the ...
U.S. Environmental Protection Agency — Microarray datasets used in the analysis. This dataset is associated with the following publication: Rosen, M., K. Das, J. Rooney, B. Abbott, C. Lau, and C. Corton....
Distribution. Sincerely, Frank T. Abbott VP of Administration & Finance fta @quantumleap.us cc: Dr. Ganesh Vaidyanathan, Project Manager, Code 34...goals of IWBP. Areas of potential application include health care administration, clinical data analysis and health care research applications
A few French scientists are bringing astronomy to captive audiences, such as the terminally ill and the incarcerated. Alison Abbott joined a group of convicted murderers to learn about gravity (1 page)
Higgins, Victoria; Chan, Man Khun; Nieuwesteeg, Michelle; Hoffman, Barry R; Bromberg, Irvin L; Gornall, Doug; Randell, Edward; Adeli, Khosrow
The Canadian Laboratory Initiative on Pediatric Reference Intervals (CALIPER) has recently established pediatric age- and sex-specific reference intervals for over 85 biochemical markers on the Abbott Architect system. Previously, CALIPER reference intervals for several biochemical markers were successfully transferred from Abbott assays to Roche, Beckman, Ortho, and Siemens assays. This study further broadens the CALIPER database by performing transference and verification for 52 biochemical assays on the Roche cobas 6000 and the Roche Modular P. Using CLSI C28-A3 and EP9-A2 guidelines, transference of the CALIPER reference intervals was attempted for 16 assays on the Roche cobas 6000 and 36 on the Modular P. Calculated reference intervals were further verified using 100 healthy CALIPER samples. Most assays showed strong correlation between assay systems and were transferable from Abbott to the Roche cobas 6000 (81%) and the Modular P (86%). Bicarbonate and magnesium were not transferable on either system and calcium and prealbumin were not transferable to the Modular P. Of the transferable analytes, 62% and 61% were verified on the cobas 6000 and the Modular P, respectively. This study extends the utility of the CALIPER database to two additional analytical systems, which facilitates the broad application of CALIPER reference intervals at pediatric centers utilizing Roche biochemical assays. Transference studies across different analytical platforms can later be collectively analyzed in an attempt to develop common reference intervals across all clinical chemistry instruments to harmonize laboratory test interpretation in diagnosis and monitoring of pediatric disease. Copyright © 2015 The Canadian Society of Clinical Chemists. Published by Elsevier Inc. All rights reserved.
Jain, A.; Bhosle, N
1980; Fletcher and Marshall 1982; Abbott et al. 1983; Frolund et al. 1996; Azeredo and Oliveira 2000; Gubner and Beech 2000). Carbohydrates account for *10 to 50% of dissolved organic carbon in marine waters (Pakulski and Benner 1994; Amon and Benner... in bacterial adhesion to surfaces. Conversely, bacterial adhesion to a protein-rich conditioning film was inhibited (Fletcher 1980; Fletcher and Marshall 1982; Abbott et al. 1983; Husmark and Ronner 1993). It appears that the proteins of the conditioning film...
Kselikova, M; Horejsi, J; Urbankova, J
The presence of the Delta agent within the population was tested by means of the Delta-antibody radioimmunoassay using competitive kits of the firms ABBOTT (ABBOTT-ANTI-DELTA) and SORIN (AB-DELTAK). The Delta-antibody was found in 3.2% HBV patients, 5% HBsAg carriers, and in 20.8% of specific anti-Hbs-immunoglobulin. In hemophiliacs and blood donors no Delta-antibody was seen.
an electrospray ionization time-of-flight (ESI-TOF) mass spectrometer to the lab (PerSeptive Biosystems Mariner), which provides a different...with this Project Faculty: Professor Michael A. Duncan (one month summer salary) Prof. Heather Abbott -Lyons, summer visiting faculty and...Distribution approved for public release. 4. H. L. Abbott , A. D. Brathwaite and M. A. Duncan, "Infrared spectroscopy and structures of mass-selected
James D. Oliver, MC USA∥; COL Kevin C. Abbott , MC USA∥; John A. Jones, BS§; LTC Kevin K. Chung, MC USA§ ABSTRACT Objective: To describe the...Cannon J. W., Zonies D. H., Morrow B. D., Orman J. A., Oliver J. D., Abbott K. C., Jones J. A., Chung K. K., 5d. PROJECT NUMBER 5e. TASK NUMBER...Anuria ( cc/24 hour) Use of vasoactive meds Contrasted radiologic studies Insulin requirement Comorbid diagnoses: diabetes , coronary artery disease
diabetes and hematopoietic stem cell engraftment . Sitagliptin is a DPPIV inhibitor already approved for type 2 diabetes because it has...activation protein (FAP) in hepatitis C virus infection. Adv Exp Med Biol 524:235–243 12. Levy MT, McCaughan GW, Abbott CA, Park JE, Cunningham AM...kb ( Abbott et al., 1994). Three different splice variants of FAP have been observed in mouse embryonic tissues, with all three predicted to encode
Kselikova, M.; Horejsi, J.; Urbankova, J.
The presence of the Delta agent within the population was tested by means of the Delta-antibody radioimmunoassay using competitive kits of the firms ABBOTT (ABBOTT-ANTI-DELTA) and SORIN (AB-DELTAK). The Delta-antibody was found in 3.2% HBV patients, 5% HBsAg carriers, and in 20.8% of specific anti-Hbs-immunoglobulin. In hemophiliacs and blood donors no Delta-antibody was seen. (author)
Five-year clinical and functional multislice computed tomography angiographic results after coronary implantation of the fully resorbable polymeric everolimus-eluting scaffold in patients with de novo coronary artery disease
Onuma, Yoshinobu; Dudek, Dariusz; Thuesen, Leif
This study sought to demonstrate the 5-year clinical and functional multislice computed tomography angiographic results after implantation of the fully resorbable everolimus-eluting scaffold (Absorb BVS, Abbott Vascular, Santa Clara, California).......This study sought to demonstrate the 5-year clinical and functional multislice computed tomography angiographic results after implantation of the fully resorbable everolimus-eluting scaffold (Absorb BVS, Abbott Vascular, Santa Clara, California)....
Abbott, B. P.; Abbott, R.; Adhikari, R. X.; Anderson, S. B.; Arai, K.; Araya, M. C.; Barayoga, J. C.; Barish, B. C.; Berger, B. K.; Billingsley, G.; Blackburn, J. K.; Bork, R.; Brooks, A. F.; Brunett, S.; Cahillane, C.
This paper presents updated estimates of source parameters for GW150914, a binary black-hole coalescence event detected by the Laser Interferometer Gravitational-wave Observatory (LIGO) in 2015 [Abbott et al. Phys. Rev. Lett. 116, 061102 (2016).]. Abbott et al. [Phys. Rev. Lett. 116, 241102 (2016).] presented parameter estimation of the source using a 13-dimensional, phenomenological precessing-spin model (precessing IMRPhenom) and an 11-dimensional nonprecessing effective-one-body (EOB) mode...
Kselikova, M.; Novak, J.; Urbankova, J.
Hepatitis B antigen and antibody were determined simultaneously by radioimmunoassay using the Biotest combRIA-Au kit. The results in determining antigens and antibodies in sera with a known content of these substances by this kit nearly equal those of isolated determinations of antigen by the Abbott AUSRIA II-125 kit, and of antibody by the Abbott AUSAB kit, resp., except for a negligible number of sera with a very low content of either antigen or antibody. (author)
Federal Laboratory Consortium — Description/History: Chemistry and biology laboratoriesThe Bio Engineering Laboratory (BeL) is theonly full spectrum biotechnology capability within the Department...
Federal Laboratory Consortium — This laboratory provides biomechanical and physical analyses for both military and commercial footwear. The laboratory contains equipment that is integral to the us...
Federal Laboratory Consortium — The Nanotechnology Characterization Laboratory (NCL) at the Frederick National Laboratory for Cancer Research performs preclinical characterization of nanomaterials...
Federal Laboratory Consortium — PNNL's Physical Sciences Laboratory (PSL) houses 22 research laboratories for conducting a wide-range of research including catalyst formulation, chemical analysis,...
Federal Laboratory Consortium — The Distributed Energy Technologies Laboratory (DETL) is an extension of the power electronics testing capabilities of the Photovoltaic System Evaluation Laboratory...
Full Text Available Expansion of HIV viral load (VL testing services are required to meet increased targets for monitoring patients on antiretroviral treatment. South Africa currently tests >4million VLs per annum in 16 highly centralised, automated high-throughput laboratories. The Xpert HIV-1 VL assay (Cepheid was evaluated against in-country predicates, the Roche Cobas Taqmanv2 and Abbott HIV-1RT, to investigate options for expanding VL testing using GeneXpert's random access, polyvalent capabilities and already established footprint in South Africa with the Xpert MTB/RIF assay (207 sites. Additionally, the performance of Xpert HIV-1VL on alternative, off-label specimen types, Dried Blood Spots (DBS and whole blood, was investigated.Precision, accuracy (agreement and clinical misclassification (1000cp/ml of Xpert HIV-1VL plasma was compared to Taqmanv2 (n = 155 and Abbott HIV-1 RT (n = 145. Misclassification of Xpert HIV-1VL was further tested on DBS (n = 145 and whole blood (n = 147.Xpert HIV-1VL demonstrated 100% concordance with predicate platforms on a standardised frozen, plasma panel (n = 42 and low overall percentage similarity CV of 1.5% and 0.9% compared to Taqmanv2 and Abbott HIV-1 RT, respectively. On paired plasma clinical specimens, Xpert HIV-1VL had low bias (SD 0.32-0.37logcp/ml and 3% misclassification at the 1000cp/ml threshold compared to Taqmanv2 (fresh and Abbott HIV-1 RT (frozen, respectively. Xpert HIV-1VL on whole blood and DBS increased misclassification (upward by up to 14% with increased invalid rate. All specimen testing was easy to perform and compatible with concurrent Xpert MTB/RIF Tuberculosis testing on the same instrument.The Xpert HIV-1VL on plasma can be used interchangeably with existing predicate platforms in South Africa. Whole blood and DBS testing requires further investigation, but polyvalency of the GeneXpert offers a solution to extending VL testing services.
Federal Laboratory Consortium — Description/History: Chemistry laboratoryThe Advanced Chemistry Laboratory (ACL) is a unique facility designed for working with the most super toxic compounds known...
Federal Laboratory Consortium — The Lincoln Laboratory Grid (LLGrid) is an interactive, on-demand parallel computing system that uses a large computing cluster to enable Laboratory researchers to...
Federal Laboratory Consortium — The Gun Dynamics Laboratory is a research multi-task facility, which includes two firing bays, a high bay area and a second floor laboratory space. The high bay area...
Federal Laboratory Consortium — The NASA Space Radiation Laboratory (NSRL) at Brookhaven National Laboratory is a NASA funded facility, delivering heavy ion beams to a target area where scientists...
Federal Laboratory Consortium — Program CapabilitiesDEN-DO Laboratory is a multi-functional laboratory capable of analyzing most chemical analytes and pathogenic/non-pathogenic microorganisms found...
Fabiansen, C.; Knudsen, J.D.; Lebech, A.M.
Brucellosis is a rare disease in Denmark. We describe one case of laboratory-acquired brucellosis from an index patient to a laboratory technician following exposure to an infected blood culture in a clinical microbiology laboratory Udgivelsesdato: 2008/6/9......Brucellosis is a rare disease in Denmark. We describe one case of laboratory-acquired brucellosis from an index patient to a laboratory technician following exposure to an infected blood culture in a clinical microbiology laboratory Udgivelsesdato: 2008/6/9...
Lenters-Westra, Erna; English, Emma
As a reference laboratory for HbA1c, it is essential to have accurate and precise HbA1c methods covering a range of measurement principles. We report an evaluation of the Abbott Enzymatic (Architect c4000), Roche Gen.3 HbA1c (Cobas c513) and Tosoh G11 using different quality targets. The effect of hemoglobin variants, other potential interferences and the performance in comparison to both the International Federation of Clinical Chemistry and Laboratory Medicine (IFCC) and the National Glycohemoglobin Standardization Program (NGSP) reference systems was assessed using certified evaluation protocols. Each of the evaluated HbA1c methods had CVs HbA1c. Partly based on the result of this study, the Abbott Enzymatic method on the Architect c4000 and the Roche Gen.3 HbA1c on the Cobas c513 are now official, certified IFCC and NGSP SRMPs in the IFCC and NGSP networks. Sigma metrics quality criteria presented in a graph distinguish between good and excellent performance.
Pediatric reference value distributions and covariate-stratified reference intervals for 29 endocrine and special chemistry biomarkers on the Beckman Coulter Immunoassay Systems: a CALIPER study of healthy community children.
Karbasy, Kimiya; Lin, Danny C C; Stoianov, Alexandra; Chan, Man Khun; Bevilacqua, Victoria; Chen, Yunqi; Adeli, Khosrow
The CALIPER program is a national research initiative aimed at closing the gaps in pediatric reference intervals. CALIPER previously reported reference intervals for endocrine and special chemistry markers on Abbott immunoassays. We now report new pediatric reference intervals for immunoassays on the Beckman Coulter Immunoassay Systems and assess platform-specific differences in reference values. A total of 711 healthy children and adolescents from birth to reference intervals calculated in accordance with Clinical and Laboratory Standards Institute (CLSI) EP28-A3c guidelines. Complex profiles were observed for all 29 analytes, necessitating unique age and/or sex-specific partitions. Overall, changes in analyte concentrations observed over the course of development were similar to trends previously reported, and are consistent with biochemical and physiological changes that occur during childhood. Marked differences were observed for some assays including progesterone, luteinizing hormone and follicle-stimulating hormone where reference intervals were higher than those reported on Abbott immunoassays and parathyroid hormone where intervals were lower. This study highlights the importance of determining reference intervals specific for each analytical platform. The CALIPER Pediatric Reference Interval database will enable accurate diagnosis and laboratory assessment of children monitored by Beckman Coulter Immunoassay Systems in health care institutions worldwide. These reference intervals must however be validated by individual labs for the local pediatric population as recommended by CLSI.
Federal Laboratory Consortium — NIST's PV characterization laboratory is used to measure the electrical performance and opto-electronic properties of solar cells and modules. This facility consists...
Federal Laboratory Consortium — The ARDEC Rapid Prototyping (RP) Laboratory was established in December 1992 to provide low cost RP capabilities to the ARDEC engineering community. The Stratasys,...
Federal Laboratory Consortium — The TVA Central Laboratories Services is a comprehensive technical support center, offering you a complete range of scientific, engineering, and technical services....
Federal Laboratory Consortium — For more than 60 years, Sandia has delivered essential science and technology to resolve the nation's most challenging security issues.Sandia National Laboratories...
Federal Laboratory Consortium — The Wireless Emulation Laboratory (WEL) is a researchtest bed used to investigate fundamental issues in networkscience. It is a research infrastructure that emulates...
Federal Laboratory Consortium — This laboratory develops screening assays, tests and modifies biosensor equipment, and optimizes food safety testing protocols for the military and civilian sector...
Federal Laboratory Consortium — The Embedded Processor Laboratory provides the means to design, develop, fabricate, and test embedded computers for missile guidance electronics systems in support...
Federal Laboratory Consortium — FUNCTION: Supports the development of prototype deployment platform vehicles for offboard countermeasure systems.DESCRIPTION: The Vehicle Development Laboratory is...
Federal Laboratory Consortium — This laboratory contains an electro-magnetic worldwide data collection and field measurement capability in the area of acoustic technology. Outfitted by NASA Langley...
Federal Laboratory Consortium — This laboratory conducts basic and applied human research studies to characterize cognitive performance as influenced by militarily-relevant contextual and physical...
Federal Laboratory Consortium — The Space Weather Computational Laboratory is a Unix and PC based modeling and simulation facility devoted to research analysis of naturally occurring electrically...
Federal Laboratory Consortium — The Atmospheric Measurements Laboratory (AML) is one of the nation's leading research facilities for understanding aerosols, clouds, and their interactions. The AML...
Federal Laboratory Consortium — The purpose of the Composites Characterization Laboratory is to investigate new and/or modified matrix materials and fibers for advanced composite applications both...
Federal Laboratory Consortium — The Microgravity Emissions Laboratory (MEL) utilizes a low-frequency acceleration measurement system for the characterization of rigid body inertial forces generated...
Federal Laboratory Consortium — The Semiconductor Laser Measurements Laboratory is equipped to investigate and characterize the lasing properties of semiconductor diode lasers. Lasing features such...
Federal Laboratory Consortium — NETL’s Fuels Processing Laboratory in Morgantown, WV, provides researchers with the equipment they need to thoroughly explore the catalytic issues associated with...
Federal Laboratory Consortium — The Advanced Manufacturing Laboratory at the University of Maryland provides the state of the art facilities for realizing next generation products and educating the...
Federal Laboratory Consortium — The Virtual Training Devices (VTD) Laboratory at the Life Cycle Software Engineering Center, Picatinny Arsenal, provides a software testing and support environment...
Federal Laboratory Consortium — The Intelligent Optics Laboratory supports sophisticated investigations on adaptive and nonlinear optics; advancedimaging and image processing; ground-to-ground and...
Federal Laboratory Consortium — This laboratory contains equipment that performs a broad array of microbiological analyses for pathogenic and spoilage microorganisms. It performs challenge studies...
Pritchard, Jack; Braker, Clifton
Pritchard discusses the opportunities for applied learning afforded by laboratories. Braker describes the evaluation of cognitive, affective, and psychomotor skills in the agricultural mechanics laboratory. (SK)
Federal Laboratory Consortium — This facility provides office space for industry researchers, experimental laboratories, computer facilities for analytical work, and space for assembling components...
Federal Laboratory Consortium — FUNCTION: To process, store, and disseminate geospatial data to the Department of Defense and other Federal agencies.DESCRIPTION: The Geospatial Services Laboratory...
Federal Laboratory Consortium — At NETL’s Thermogravimetric Analysis Laboratory in Morgantown, WV, researchers study how chemical looping combustion (CLC) can be applied to fossil energy systems....
Federal Laboratory Consortium — The Research Combustion Laboratory (RCL) develops aerospace propulsion technology by performing tests on propulsion components and materials. Altitudes up to 137,000...
Federal Laboratory Consortium — The Combustion Research Laboratory facilitates the development of new combustion systems or improves the operation of existing systems to meet the Army's mission for...
Federal Laboratory Consortium — Purpose: The mission of the Coatings and Corrosion Laboratory is to develop and analyze the effectiveness of innovative coatings test procedures while evaluating the...
Federal Laboratory Consortium — Current research in the Laboratory of Chemical Physics is primarily concerned with experimental, theoretical, and computational problems in the structure, dynamics,...
Federal Laboratory Consortium — The Optical Remote Sensing Laboratory deploys rugged, cutting-edge electro-optical instrumentation for the collection of various event signatures, with expertise in...
Federal Laboratory Consortium — The Tactical Systems Integration Laboratory is used to design and integrate computer hardware and software and related electronic subsystems for tactical vehicles....
Federal Laboratory Consortium — As part of the Electrical and Computer Engineering Department and The Institute for System Research, the Neural Systems Laboratory studies the functionality of the...
Federal Laboratory Consortium — The Environmental Microbiology Laboratory, located in Bldg. 644 provides a dual-gas respirometer for measurement of oxygen consumption and carbon dioxide evolution...
The elements (principles) of quality assurance can be applied to the operation of the analytical chemistry laboratory to provide an effective tool for indicating the competence of the laboratory and for helping to upgrade competence if necessary. When used, those elements establish the planned and systematic actions necessary to provide adequate confidence in each analytical result reported by the laboratory (the definition of laboratory quality assurance). The elements, as used at the Hanford Engineering Development Laboratory (HEDL), are discussed and they are qualification of analysts, written methods, sample receiving and storage, quality control, audit, and documentation. To establish a laboratory quality assurance program, a laboratory QA program plan is prepared to specify how the elements are to be implemented into laboratory operation. Benefits that can be obtained from using laboratory quality assurance are given. Experience at HEDL has shown that laboratory quality assurance is not a burden, but it is a useful and valuable tool for the analytical chemistry laboratory
Laboratory diagnosis is auxillary medical discipline studying specific laboratory symptoms of diseases, revealed by investigations of materials taken from patients. The structure of laboratory servie in our country and abroad, items of laboratory investigations, organizational principles are described. Attention is being given to the cost of analyses, the amount of conducted investigations, methods of result presentation, problems of accuracy, quality control and information content
Isajenko, K.A.; Lipinski, P.
The article presents the Mobile Spectrometric Laboratory used by Central Laboratory for Radiological Protection since year 2000. The equipment installed in the Mobile Laboratory and its uses is described. The results of international exercises and intercalibrations, in which the Laboratory participated are presented. (author)
Federal Laboratory Consortium — The Energy Materials Research Laboratory at the Savannah River National Laboratory (SRNL) creates a cross-disciplinary laboratory facility that lends itself to the...
Levidiotou, S; Vrioni, G; Papadogeorgaki, H; Avdeliodi, K; Kada, H; Kaparos, G; Kouskouni, E; Fragouli, E; Legakis, N J
The present retrospective study was initiated to determine the prevalence of Chlamydia trachomatis and to assess the risk factors for infection in adult women and men presenting to general practitioners, gynecologists, dermatologists, and family-planning centers in Greece. The study was carried out in four different Greek hospital centers using highly sensitive nucleic acid amplification techniques. Altogether, 16,834 women and 1,035 men were enrolled from October 1998 to April 2004. Two types of specimens were collected from each patient: cervical swabs from women, urethral swabs from men, and first-catch urine from women and men. All specimens were examined with the Cobas Amplicor C. trachomatis polymerase chain reaction assay (Roche Molecular Systems, Branchburg, NJ, USA) or the LC x C. trachomatis ligase chain reaction assay (Abbott Laboratories, Abbott Park, IL, USA). Demographic and behavioral data were collected by clinicians using a standardized questionnaire. A total of 704 (3.9%) patients were infected with C. trachomatis. The prevalence among female patients was 3.5% and that among male patients 11.2%. Among infected patients, 88% were under 30 years of age, 71% reported more than one sexual partner, and 91% reported a new sexual partner within the last year. In conclusion, the prevalence of C. trachomatis infection in Greece is low. Young age and new and multiple sexual partners within the last year were factors consistently associated with an increased risk of chlamydial infection.
Chavez-Abraham, Victor; Barr, Jason S; Zwiebel, Paul C
Postoperative pain management following aesthetic plastic surgery traditionally has been achieved by systemic administration of several narcotic pain medications. Because this method can lead to undesirable side effects such as sedation, nausea, vomiting, and respiratory depression, a more efficacious method of postoperative analgesia with fewer side effects needs to be implemented in outpatient cosmetic surgery. From March of 2003 until December of 2008, 690 patients underwent augmentation mammaplasty and 215 patients underwent abdominoplasty. All of these patients were equipped with an elastomeric continuous infusion pump postoperatively and were prescribed oral narcotics. Prior to 2003, patients were prescribed only oral narcotics postoperatively. A retrospective chart review of patients before and after implementation of the pain pump was undertaken to review the perceived pain patients experienced postoperatively with and without the pump. The self-administration of oral narcotics was also assessed. Patients equipped with the pain pump experienced a statistically significant decrease in perceived pain compared to those without the pump (augmentation mammaplasty: 2.27 vs. 3.68, p use of the oral narcotic Vicodin™ at 72 h postoperatively (5 mg hydrocodone/500 mg acetaminophen, Abbott Laboratories, Abbott Park, IL) (augmentation mammaplasty: 26.5 mg/2650 mg vs. 49 mg/4900 mg, p reduce both the amount of pain patients experience and the quantity of narcotics used postoperatively.
Angela F Caveney
Full Text Available Angela F Caveney1, Bruno Giordani1, George M Haig21Department of Psychiatry, University of Michigan, Ann Arbor, MI, USA; 2Neurosciences Development, Abbott Laboratories, Abbott Park, IL, USAAbstract: Pagoclone is a novel cyclopyrrolone that acts as a partial GABAA receptor agonist. Preclinical studies suggest that pagoclone may have clinical utility as an anxiolytic agent, as well as a reduced incidence of side-effects. The present study was conducted to determine whether pagoclone would affect healthy individuals’ performances on neuropsychological measures as a function of dose within the projected therapeutic range. Twelve healthy adult subjects were randomly assigned to dosage groups in a 3-way crossover study. Participants were administered neuropsychological measures six hours following dosing on Day 1 and Day 6 of administration of the drug. Dose effects were noted on measures of alertness, learning, and memory and movement time. Significant effects were also noted on measures of alertness, learning and memory, information processing and psychomotor speed. Overall, the results of this small, preliminary study do not support a finding of behavioral toxicity for these doses of pagoclone. Rather, a pattern was found of transient and mild negative effects on learning and memory scores at the highest dose administered, though these changes were small and no longer evident by the sixth day of use.Keywords: pagoclone, cyclopyrrolone, neuropsychological, memory, generalized anxiety disorder
Kari S Anderson
Full Text Available Kari S Anderson, Victoria M Mullally, Linda M Fredrick, Andrew L CampbellAbbott Laboratories, Abbott Park, IL, USAAbstract: Respiratory syncytial virus (RSV is a significant cause of morbidity in high-risk infants. Palivizumab is proven to prevent serious RSV disease, but compliance with prophylaxis (monthly doses during the RSV season is essential to ensure protection. We invited 453 pediatricians to participate in a survey to identify their perspectives of barriers to compliance and interventions to improve compliance with palivizumab prophylaxis schedules. One hundred physicians from five continents completed the survey, identifying caregiver inconvenience, distance to clinic, cost of prophylaxis, and lack of understanding of the severity of RSV as the most common reasons for noncompliance. They recommended provision of educational materials about RSV, reminders from hospital or clinic, and administration of prophylaxis at home to increase compliance. Globally, physicians recognize several obstacles to prophylaxis compliance. This survey suggests that focused proactive interventions such as empowering caregivers with educational materials and reducing caregiver inconvenience may be instrumental to increase compliance.Keywords: medication adherence, respiratory syncytial virus infections, infant, premature, immunization, passive
Raghav, Raj; Middleton, Rachael; BSc, Rinshiya Ahamed; Arjunan, Raji; Caliendo, Valentina
Arterial and venous blood gas analysis is useful in the assessment of tissue oxygenation and ventilation and in diagnosis of metabolic and respiratory derangements. It can be performed with a relatively small volume of blood in avian patients under emergency situations. Arterial and venous blood gas analysis was performed in 30 healthy gyr falcons ( Falco rusticolus ) under anaesthesia to establish temperature-corrected reference intervals for arterial blood gas values and to compare them to temperature-corrected venous blood gas values with a portable point-of-care blood gas analyzer (i-STAT 1, Abbott Laboratories, Abbott Park, IL, USA). Statistically significant differences were observed between the temperature-corrected values of pH, partial pressure of carbon dioxide (Pco2), and partial pressure of oxygen (Po2) and the corresponding nontemperature-corrected values of these parameters in both arterial and venous blood. Values of temperature-corrected pH, temperature-corrected Pco2, bicarbonate concentrations, and base excess of extra cellular fluid did not differ significantly between arterial and venous blood, suggesting that, in anesthetized gyr falcons, venous blood gas analysis can be used in place of arterial blood gas analysis in clinical situations. Values for hematocrit, measured by the point-of-care analyzer, were significantly lower compared with those obtained by the microhematocrit method.
Federal Laboratory Consortium — The Navy Aircraft Protection Laboratory provides complete test support for all Navy air vehicle fire protection systems.The facility allows for the simulation of a...
Federal Laboratory Consortium — The electro-deposition laboratory can electro-deposit various coatings onto small test samples and bench level prototypes. This facility provides the foundation for...
Federal Laboratory Consortium — The Radiochemical Processing Laboratory (RPL)ï¿½is a scientific facility funded by DOE to create and implement innovative processes for environmental clean-up and...
U.S. Department of Health & Human Services — Outpatient clinical laboratory services are paid based on a fee schedule in accordance with Section 1833(h) of the Social Security Act. The clinical laboratory fee...
Skip to main content Argonne National Laboratory Toggle Navigation Toggle Search Energy Environment Laboratory About Safety News Careers Education Community Diversity Directory Energy Environment National Security User Facilities Science Work with Us Environment Atmospheric and Climate Science Ecological
Federal Laboratory Consortium — This laboratory offers the services of highly trained and experienced specialists that have a full complement of measuring equipment. It is equipped with two optical...
Federal Laboratory Consortium — Researchers use computed tomography (CT) scanners at NETL’s Geological Services Laboratory in Morgantown, WV, to peer into geologic core samples to determine how...
Knudsen, Jakob; von Seidlein, Lorenz; Richard, Jean Pierre
An illustrated description of the building of a biomedical research laboratory in Korogwe, Tanzania.......An illustrated description of the building of a biomedical research laboratory in Korogwe, Tanzania....
Federal Laboratory Consortium — The Laboratory of Biological Modeling is defined by both its methodologies and its areas of application. We use mathematical modeling in many forms and apply it to a...
Skip to main content Argonne National Laboratory Toggle Navigation Toggle Search Energy Batteries and Energy Storage Energy Systems Modeling Materials for Energy Nuclear Energy Renewable Energy Smart Laboratory About Safety News Careers Education Community Diversity Directory Energy Environment National
Federal Laboratory Consortium — The Lab has a proud history and heritage of almost 70 years of science and innovation. The people at the Laboratory work on advanced technologies to provide the best...
Federal Laboratory Consortium — This laboratory is a specially constructed facility with elevated (37 feet) ceilings and an overhead catwalk, and which is dedicated to research efforts in reducing...
Federal Laboratory Consortium — Purpose: The mission of the Geometric Design Laboratory (GDL) is to support the Office of Safety Research and Development in research related to the geometric design...
Federal Laboratory Consortium — Program CapabilitiesDET-DO Laboratory is equipped with the usual instrumentation necessary to perform a wide range of analyses of food, drugs and cosmetics. Program...
Federal Laboratory Consortium — This laboratory contains equipment to fabricate and test prototype packages of many types and sizes (e.g., bags, pouches, trays, cartons, etc.). This equipment can...
Federal Laboratory Consortium — Columbia River and groundwater well water sources are delivered to the Aquatic Research Laboratory (ARL), where these resources are used to conduct research on fish...
Federal Laboratory Consortium — Purpose: The purpose of the Human Factors Laboratory is to further the understanding of highway user needs so that those needs can be incorporated in roadway design,...
Federal Laboratory Consortium — Program CapabilitiesPHI-DO Pharmaceutical Laboratory specializes in the analyses of all forms and types of drug products.Its work involves nearly all phases of drug...
Federal Laboratory Consortium — These energetic materials laboratories are equipped with explosion proof hoods with blow out walls for added safety, that are certified for safe handling of primary...
Federal Laboratory Consortium — The Neutral Buoyancy Laboratory (NBL) is an astronaut training facility and neutral buoyancy pool operated by NASA and located at the Sonny Carter Training Facility,...
Federal Laboratory Consortium — This laboratory is a 40 by 28 by 9 foot facility that is equipped with tools for the development of various items of control technology related to the transmission...
U.S. Department of Health & Human Services — This website provides demographic information about laboratories, including CLIA number, facility name and address, where the laboratory testing is performed, the...
Since the keeping of good records is essential in the chemistry laboratory, general guidelines for maintaining a laboratory notebook are provided. Includes rationale for having entries documented or witnessed. (Author/JN)
Defense Systems & Assessments: About Us Sandia National Laboratories Exceptional service in ; Security Weapons Science & Technology Defense Systems & Assessments About Defense Systems & Information Construction & Facilities Contract Audit Sandia's Economic Impact Licensing & Technology
Pazzagli, M.; Malentacchi, F.; Mancini, I.
diagnostic tools and expertise and commands proper state-of-the-art knowledge about Personalized Medicine and Laboratory Medicine in Europe, the joint Working Group "Personalized Laboratory Medicine" of the EFLM and ESPT societies compiled and conducted the Questionnaire "Is Laboratory Medicine ready...... in "omics"; 2. Additional training for the current personnel focused on the new methodologies; 3. Incorporation in the Laboratory of new competencies in data interpretation and counselling; 4. Improving cooperation and collaboration between professionals of different disciplines to integrate information...
The Environmental Protection Agency's (EPA) Chemistry Laboratory (ECL) is a national program laboratory specializing in residue chemistry analysis under the jurisdiction of the EPA's Office of Pesticide Programs in Washington, D.C. At Stennis Space Center, the laboratory's work supports many federal anti-pollution laws. The laboratory analyzes environmental and human samples to determine the presence and amount of agricultural chemicals and related substances. Pictured, ECL chemists analyze environmental and human samples for the presence of pesticides and other pollutants.
Amare, J.; Beltran, B.; Carmona, J.M.; Cebrian, S.; Garcia, E.; Irastorza, I.G.; Gomez, H.; Luzon, G.; Martinez, M.; Morales, J.; Ortiz de Solorzano, A.; Pobes, C.; Puimedon, J.; Rodriguez, A.; Ruz, J.; Sarsa, M.L.; Torres, L.; Villar, J.A.
This paper describes the forthcoming enlargement of the Canfranc Underground Laboratory (LSC) which will allow to host new international Astroparticle Physics experiments and therefore to broaden the European underground research area. The new Canfranc Underground Laboratory will operate in coordination (through the ILIAS Project) with the Gran Sasso (Italy), Modane (France) and Boulby (UK) underground laboratories
Shehabi, Arman [Lawrence Berkeley National Lab. (LBNL), Berkeley, CA (United States); Ganeshalingam, Mohan [Lawrence Berkeley National Lab. (LBNL), Berkeley, CA (United States); DeMates, Lauren [Lawrence Berkeley National Lab. (LBNL), Berkeley, CA (United States); Mathew, Paul [Lawrence Berkeley National Lab. (LBNL), Berkeley, CA (United States); Sartor, Dale [Lawrence Berkeley National Lab. (LBNL), Berkeley, CA (United States)
Laboratories are estimated to be 3-5 times more energy intensive than typical office buildings and offer significant opportunities for energy use reductions. Although energy intensity varies widely, laboratories are generally energy intensive due to ventilation requirements, the research instruments used, and other health and safety concerns. Because the requirements of laboratory facilities differ so dramatically from those of other buildings, a clear need exists for an initiative exclusively targeting these facilities. The building stock of laboratories in the United States span different economic sectors, include governmental and academic institution, and are often defined differently by different groups. Information on laboratory buildings is often limited to a small subsection of the total building stock making aggregate estimates of the total U.S. laboratories and their energy use challenging. Previous estimates of U.S. laboratory space vary widely owing to differences in how laboratories are defined and categorized. A 2006 report on fume hoods provided an estimate of 150,000 laboratories populating the U.S. based in part on interviews of industry experts, however, a 2009 analysis of the 2003 Commercial Buildings Energy Consumption Survey (CBECS) generated an estimate of only 9,000 laboratory buildings. This report draws on multiple data sources that have been evaluated to construct an understanding of U.S. laboratories across different sizes and markets segments. This 2016 analysis is an update to draft reports released in October and December 2016.
Galanti, L M; Cornu, C; Masson, P L; Robert, A R; Becheanu, D; Lamy, M E; Cambiaso, C L
An assay of anti-HBs antibodies based on agglutination of latex particles coated with recombinant HBs-antigen was compared with Abbott radioimmunoassay (Abbott-RIA), which uses a human plasma-derived antigen. The population examined consisted of 76 Abbott-RIA anti-HBs-negative prevaccinated subjects and 1044 serum samples anti-HBs found positive by Abbott-RIA, including 283 samples of subjects vaccinated either with a human plasma-derived vaccine (group A; n = 180) or with a recombinant vaccine (group B; n = 103). Correlation coefficients between the two techniques were respectively r = 0.89 for the whole population (n = 1044), r = 0.98 in group A and r = 0.74 in group B. Anti-HBs titres were higher with latex than with RIA in group B as shown by the regression slopes: latex = 508 + 1.11 RIA in group A and latex = -1138 + 3.97 RIA in group B, suggesting that some vaccinated subjects from group B produced antibodies against epitopes proper to the recombinant antigen. In the prevaccinated population and in group A, the latex results were compared with those of radioimmunoassays (Abbott, Sorin) and enzyme immunoassays (Behring, Roche, Pasteur). Only the Roche-EIA detected anti-HBs in the prevaccinated subjects. The correlation between the various immunoassays was r greater than 0.96 only for values higher than 100 IU/l.
Ham, Melissa R; Okada, Pamela; White, Perrin C
Diabetic ketoacidosis (DKA) is a serious complication of diabetes mellitus marked by characteristic biochemical derangements. Diagnosis and management involve frequent evaluation of these biochemical parameters. Reliable bedside equivalents for these laboratory studies may help reduce the time to treatment and reduce costs. We evaluated the precision and bias of a bedside serum ketone meter in the acute care setting. Serum ketone results using the Precision Xtra glucometer/ketone meter (Abbott Laboratories, MediSense Products Inc., Bedford, MA, USA) correlated strongly with the Children's Medical Center of Dallas' laboratory values within the meter's value range. Meter ketone values steadily decreased during the treatment of DKA as pH and CO(2) levels increased and acidosis resolved. Therefore, the meter may be useful in monitoring therapy for DKA. This meter may also prove useful in identifying patients at risk for DKA in physicians' offices or at home.
Federal Laboratory Consortium — Description: The Heat Flux Instrumentation Laboratory is used to develop advanced, flexible, thin film gauge instrumentation for the Air Force Research Laboratory....
Federal Laboratory Consortium — The Optics/Optical Diagnostics Laboratory supports graduate instruction in optics, optical and laser diagnostics and electro-optics. The optics laboratory provides...
Chapman, W. Max
Out-of-state flat-rate mail order contract laboratories operating from states which have little or no legal control over them can do business in California without obedience to regulations that govern laboratories located within the state. The flat-rate contract principle under which some out-of-state laboratories operate is illegal in California. The use of such laboratories increases physician liability. Legislation for the control of these laboratories is difficult to construct, and laws which might result would be awkward to administer. The best remedy is for California physicians not to use an out-of-state laboratory offering contracts or conditions that it could not legally offer if it were located in California. PMID:14165875
Michigan State Univ., East Lansing. Coll. of Agriculture and Natural Resources Education Inst.
This task-based curriculum guide for medical laboratory assistant is intended to help the teacher develop a classroom management system where students learn by doing. Introductory materials include a Dictionary of Occupational Titles job code and title sheet, a career ladder, a matrix relating duty/task numbers to job titles, and a task list. Each…
Federal Laboratory Consortium — The Fuel Cell Lab (FCL)Established to investigate, integrate, testand verifyperformance and technology readiness offuel cell systems and fuel reformers for use with...
Federal Laboratory Consortium — The HIL uses testing devices to evaluate vehicle interior energy attenuating (EA) technologies for mitigating head injuries resulting from head impacts during mine/...
Federal Laboratory Consortium — The purpose is to increase basic knowledge of metallurgical processing for controlling the microstructure and mechanical properties of metallic aerospace alloys and...
Federal Laboratory Consortium — This biochemistry lab is set up for protein analysis using Western blot, enzyme linked immunosorbent assays, immunohistochemistry, and bead-based immunoassays. The...
Federal Laboratory Consortium — Located at WPAFB, Ohio, the Applied Neuroscience lab researches and develops technologies to optimize Airmen individual and team performance across all AF domains....
Federal Laboratory Consortium — This lab supports cognitive research using rodent models. Capabilities for behavioral assessments include:Morris water maze and Barnes maze (spatial memory)elevate...
Federal Laboratory Consortium — The purpose is to evaluate mechanical properties of materials including metals, intermetallics, metal-matrix composites, and ceramic-matrix composites under typical...
Federal Laboratory Consortium — FUNCTION: Investigates processes and interactions at the air-sea interface, and compares measurements to numerical simulations and field data. Typical phenomena of...
Liu, Xuan; Yang, Yi
Laboratory experiences are essential for optics education. However, college students have limited access to advanced optical equipment that is generally expensive and complicated. Hence there is a need for innovative solutions to expose students to advanced optics laboratories. Here we describe a novel approach, interactive virtual optical laboratory (IVOL) that allows unlimited number of students to participate the lab session remotely through internet, to improve laboratory education in photonics. Although students are not physically conducting the experiment, IVOL is designed to engage students, by actively involving students in the decision making process throughout the experiment.
Federal Laboratory Consortium — FUNCTION: Provides instrumentation and expertise for physical and geoacoustic characterization of marine sediments.DESCRIPTION: The multisensor core logger measures...
Federal Laboratory Consortium — FUNCTION: Performs basic and applied research in interactive 3D computer graphics, including visual analytics, virtual environments, and augmented reality (AR). The...
Federal Laboratory Consortium — FUNCTION: Provides NP-3D aircraft host platforms for Effectiveness of Navy Electronic Warfare Systems (ENEWS) Program antiship missile (ASM) seeker simulators used...
Federal Laboratory Consortium — FUNCTION: Supports experimental research where high-frequency acoustic scattering and surface vibration measurements of fluid-loaded and non-fluid-loaded structures...
Federal Laboratory Consortium — FUNCTION: Supports experimental research where acoustic radiation, scattering, and surface vibration measurements of fluid-loaded and non-fluid-loaded structures are...
Gilliom, Laura R.
Sandia National Laboratories has identified technology transfer to U.S. industry as a laboratory mission which complements our national security mission and as a key component of the Laboratory's future. A number of technology transfer mechanisms - such as CRADA's, licenses, work-for-others, and consortia - are identified and specific examples are given. Sandia's experience with the Specialty Metals Processing Consortium is highlighted with a focus on the elements which have made it successful. A brief discussion of Sandia's potential interactions with NASA under the Space Exploration Initiative was included as an example of laboratory-to-NASA technology transfer. Viewgraphs are provided.
Federal Laboratory Consortium — Structural dynamic testing is performed to verify the survivability of a component or assembly when exposed to vibration stress screening, or a controlled simulation...
The only clear evidence today for physics beyond the standard model comes from underground experiments and the future activity of underground laboratories appears challenging and rich. I review here the existing underground research facilities in Europe. I present briefly the main characteristics, scientific activity and perspectives of these Laboratories and discuss the present coordination actions in the framework of the European Union
This paper presents an overview of the progress up to April 1993 in the development of the Calibration Laboratories Accreditation Program within the framework of the National Voluntary Laboratory Accreditation Program (NVLAP) at the National Institute of Standards and Technology (NIST).
This paper presents an overview of the progress up to April 1993 in the development of the Calibration Laboratories Accreditation Program within the framework of the National Voluntary Laboratory Accreditation Program (NVLAP) at the National Institute of Standards and Technology (NIST)
Kress, R.L.; Love, L.J.
The growth of the Internet has provided a unique opportunity to expand research collaborations between industry, universities, and the national laboratories. The Virtual Robotics Laboratory (VRL) is an innovative program at Oak Ridge National Laboratory (ORNL) that is focusing on the issues related to collaborative research through controlled access of laboratory equipment using the World Wide Web. The VRL will provide different levels of access to selected ORNL laboratory secondary education programs. In the past, the ORNL Robotics and Process Systems Division has developed state-of-the-art robotic systems for the Army, NASA, Department of Energy, Department of Defense, as well as many other clients. After proof of concept, many of these systems sit dormant in the laboratories. This is not out of completion of all possible research topics. but from completion of contracts and generation of new programs. In the past, a number of visiting professors have used this equipment for their own research. However, this requires that the professor, and possibly his/her students, spend extended periods at the laboratory facility. In addition, only a very exclusive group of faculty can gain access to the laboratory and hardware. The VRL is a tool that enables extended collaborative efforts without regard to geographic limitations.
Kress, R.L.; Love, L.J.
The growth of the Internet has provided a unique opportunity to expand research collaborations between industry, universities, and the national laboratories. The Virtual Robotics Laboratory (VRL) is an innovative program at Oak Ridge National Laboratory (ORNL) that is focusing on the issues related to collaborative research through controlled access of laboratory equipment using the World Wide Web. The VRL will provide different levels of access to selected ORNL laboratory equipment to outside universities, industrial researchers, and elementary and secondary education programs. In the past, the ORNL Robotics and Process Systems Division (RPSD) has developed state-of-the-art robotic systems for the Army, NASA, Department of Energy, Department of Defense, as well as many other clients. After proof of concept, many of these systems sit dormant in the laboratories. This is not out of completion of all possible research topics, but from completion of contracts and generation of new programs. In the past, a number of visiting professors have used this equipment for their own research. However, this requires that the professor, and possibly his students, spend extended periods at the laboratory facility. In addition, only a very exclusive group of faculty can gain access to the laboratory and hardware. The VRL is a tool that enables extended collaborative efforts without regard to geographic limitations
Hardt, T.L.; Schutt, S.M.; Doran, K.S.; Dihel, D.L.; Lucas, R.O. II; Eifert, T.K.
A new state of the art radiochemistry laboratory incorporating advanced design and environmental control elements has been constructed in Atlanta, Georgia. The design of the facility is oriented to the efficient production of analytical sample results which meet regulatory requirements while at the same time provides an atmosphere that is pleasurable for analysts and visitors alike. The laboratory building contains two separate and distinct laboratories under one roof. This allows the facility to handle samples with low levels of radioactivity on one side of the lab without fear of contamination of environmental work on the other side. Unlike most laboratories, this facility utilizes a scrubber system and liquid waste holdup system to prevent accidental releases to the environment. The potential spread of radioactive contamination is controlled through the use of negative pressure ventillation zones. Construction techniques, laboratory systems, instrumentation and ergonomic considerations will also be discussed. (author) 1 fig
James R. Wright MD, PhD
Full Text Available Calgary Laboratory Services provides global hospital and community laboratory services for Calgary and surrounding areas (population 1.4 million and global academic support for the University of Calgary Cumming School of Medicine. It developed rapidly after the Alberta Provincial Government implemented an austerity program to address rising health care costs and to address Alberta’s debt and deficit in 1994. Over roughly the next year, all hospital and community laboratory test funding within the province was put into a single budget, fee codes for fee-for-service test billing were closed, roughly 40% of the provincial laboratory budget was cut, and roughly 40% of the pathologists left the province of Alberta. In Calgary, in the face of these abrupt changes in the laboratory environment, private laboratories, publicly funded hospital laboratories and the medical school department precipitously and reluctantly merged in 1996. The origin of Calgary Laboratory Services was likened to an “unhappy shotgun marriage” by all parties. Although such a structure could save money by eliminating duplicated services and excess capacity and could provide excellent city-wide clinical service by increasing standardization, it was less clear whether it could provide strong academic support for a medical school. Over the past decade, iterations of the Calgary Laboratory Services model have been implemented or are being considered in other Canadian jurisdictions. This case study analyzes the evolution of Calgary Laboratory Services, provides a metric-based review of academic performance over time, and demonstrates that this model, essentially arising as an unplanned experiment, has merit within a Canadian health care context.
Horváth, Andrea Rita; Ring, Rózsa; Fehér, Miklós; Mikó, Tivadar
In Hungary, the National Accreditation Body was established by government in 1995 as an independent, non-profit organization, and has exclusive rights to accredit, amongst others, medical laboratories. The National Accreditation Body has two Specialist Advisory Committees in the health care sector. One is the Health Care Specialist Advisory Committee that accredits certifying bodies, which deal with certification of hospitals. The other Specialist Advisory Committee for Medical Laboratories is directly involved in accrediting medical laboratory services of health care institutions. The Specialist Advisory Committee for Medical Laboratories is a multidisciplinary peer review group of experts from all disciplines of in vitro diagnostics, i.e. laboratory medicine, microbiology, histopathology and blood banking. At present, the only published International Standard applicable to laboratories is ISO/IEC 17025:1999. Work has been in progress on the official approval of the new ISO 15189 standard, specific to medical laboratories. Until the official approval of the International Standard ISO 15189, as accreditation standard, the Hungarian National Accreditation Body has decided to progress with accreditation by formulating explanatory notes to the ISO/IEC 17025:1999 document, using ISO/FDIS 15189:2000, the European EC4 criteria and CPA (UK) Ltd accreditation standards as guidelines. This harmonized guideline provides 'explanations' that facilitate the application of ISO/IEC 17025:1999 to medical laboratories, and can be used as a checklist for the verification of compliance during the onsite assessment of the laboratory. The harmonized guideline adapted the process model of ISO 9001:2000 to rearrange the main clauses of ISO/IEC 17025:1999. This rearrangement does not only make the guideline compliant with ISO 9001:2000 but also improves understanding for those working in medical laboratories, and facilitates the training and education of laboratory staff. With the
The practice of surgical pathology is under constant pressure to deliver the highest quality of service, reduce errors, increase throughput, and decrease turnaround time while at the same time dealing with an aging workforce, increasing financial constraints, and economic uncertainty. Although not able to implement total laboratory automation, great progress continues to be made in workstation automation in all areas of the pathology laboratory. This report highlights the benefits and challenges of pathology automation, reviews middleware and its use to facilitate automation, and reviews the progress so far in the anatomic pathology laboratory. Copyright © 2015 Elsevier Inc. All rights reserved.
Klokmose, Clemens Nylandsted; Zander, Pär-Ola
We take digitalization of laboratory work practice as a challenging design domain to explore. There are obvious drawbacks with the use of paper instead of ICT in the collaborative writing that takes place in laboratory notebooks; yet paper persist in being the most common solution. The ultimate aim...... with our study is to produce design relevant knowledge that can envisage an ICT solution that keeps as many advantages of paper as possible, but with the strength of electronic laboratory notebooks as well. Rather than assuming that users are technophobic and unable to appropriate state of the art software...
Grebe, Stefan K G; Kahaly, George J
The clinical diagnosis of hypo- or hyperthyroidism is difficult (full text available online: http://education.amjmed.com/pp1/272). Clinical symptoms and signs are often non-specific, and there is incomplete correlation between structural and functional thyroid gland changes. Laboratory testing is therefore indispensible in establishing the diagnosis of thyrotoxicosis. Similar considerations apply to treatment monitoring. Laboratory testing also plays a crucial role in establishing the most likely cause for a patient's hyperthyroidism. Finally, during pregnancy, when isotopic scanning is relatively contraindicated and ultrasound is more difficult to interpret, laboratory testing becomes even more important. Copyright © 2012. Published by Elsevier Inc.
Smith, Julie; Qvist, Camilla Christine; Jacobsen, Katja Kemp
Previously, biomarker research and development was performed by laboratory technicians working as craftsmen in laboratories under the guidance of medical doctors. This hierarchical structure based on professional boundaries appears to be outdated if we want to keep up with the high performance...... of our healthcare system, and take advantage of the vast potential of future biomarkers and personalized medicine. We ask the question; does our healthcare system benefit from giving the modern medical laboratory scientist (MLS) a stronger academic training in biomarker research, development...
The Simula Research Laboratory, located just outside Oslo in Norway, is rightly famed as a highly successful research facility, despite being, at only eight years old, a very young institution. This fascinating book tells the history of Simula, detailing the culture and values that have been the guiding principles of the laboratory throughout its existence. Dedicated to tackling scientific challenges of genuine social importance, the laboratory undertakes important research with long-term implications in networks, computing and software engineering, including specialist work in biomedical comp
Abbott, Edwin A; Banchoff, Thomas F
Flatland, Edwin Abbott Abbott's story of a two-dimensional universe, as told by one of its inhabitants who is introduced to the mysteries of three-dimensional space, has enjoyed an enduring popularity from the time of its publication in 1884. This fully annotated edition enables the modern-day reader to understand and appreciate the many "dimensions" of this classic satire. Mathematical notes and illustrations enhance the usefulness of Flatland as an elementary introduction to higher-dimensional geometry. Historical notes show connections to late-Victorian England and to classical Greece. Citations from Abbott's other writings as well as the works of Plato and Aristotle serve to interpret the text. Commentary on language and literary style includes numerous definitions of obscure words. An appendix gives a comprehensive account of the life and work of Flatland's remarkable author.
Université de Genève
GENEVA UNIVERSITY École de physique Département de physique nucléaire et corspusculaire 24, quai Ernest-Ansermet 1211 Genève 4 Tél.: (022) 379 62 73 Fax: (022) 379 69 92 Lundi 12 mars 2012 17h00 - Ecole de Physique, Auditoire Stueckelberg « Exploring Flatland with cold atoms » Prof. Jean Dalibard Laboratoire Kastler Brossel, CNRS, Physics Department of Ecole Normale Supérieure, Paris In his world-famous novel "Flatland" published in 1884, the English writer Edwin Abbott imagined a social life in a two-dimensional world. With a very original use of geometrical notions, E. Abbott produced a unique satire of his own society. Long after Abbott's visionary allegory, Microscopic Physics has provided a practical path for the exploration of low-dimensional worlds. With the realization of quantum wells for example, it has been possible to produce two-dimensional gases of electrons. The prope...
Abbott, B P; Abbott, R; Abbott, T D; Abernathy, M R; Acernese, F; Ackley, K; Adams, C; Adams, T; Addesso, P; Adhikari, R X; Adya, V B; Affeldt, C; Agathos, M; Agatsuma, K; Aggarwal, N
PAPER\\ud Effects of waveform model systematics on the interpretation of GW150914\\ud B P Abbott1, R Abbott1, T D Abbott2, M R Abernathy3, F Acernese4,5, K Ackley6, C Adams7, T Adams8, P Addesso9,144, R X Adhikari1, V B Adya10, C Affeldt10, M Agathos11, K Agatsuma11, N Aggarwal12, O D Aguiar13, L Aiello14,15, A Ain16, P Ajith17, B Allen10,18,19, A Allocca20,21, P A Altin22, A Ananyeva1, S B Anderson1, W G Anderson18, S Appert1, K Arai1, M C Araya1, J S Areeda23, N Arnaud24, K G Arun25, S Ascenz...
Wisniewski, Rafal; Bak, Thomas
The Satellite Laboratory at the Department of Control Engineering of Aalborg University (SatLab) is a dynamic motion facility designed for analysis and test of micro spacecraft. A unique feature of the laboratory is that it provides a completely gravity-free environment. A test spacecraft...... of the laboratory is to conduct dynamic tests of the control and attitude determination algorithms during nominal operation and in abnormal conditions. Further it is intended to use SatLab for validation of various algorithms for fault detection, accommodation and supervisory control. Different mission objectives...... can be implemented in the laboratory, e.g. three-axis attitude control, slew manoeuvres, spins stabilization using magnetic actuation and/or reaction wheels. The spacecraft attitude can be determined applying magnetometer measurements...
Federal Laboratory Consortium — FUNCTION: Develops and tests novel user interfaces for 3D virtual simulators and first-person shooter games that make user interaction more like natural interaction...
The laboratory of minerals purification was organized in 1962 where with application of modern physical and chemical methods were investigated the mechanism of flotation reagents interaction with minerals' surface, was elaborated technologies on rising complexity of using of republic's minerals
On 10 May an Agreement was signed at CERN setting up a new European Laboratory. It will be concerned with research in molecularbiology and will be located at Heidelberg in the Federal Republic of Germany.
Laboratory manual 1966 format A3 with the list of equipment cables, electronic tubes, chassis, diodes transistors etc. One of CERN's first material catalogue for construction components for mechanical and electronic chassis.
Shyamprasad, M.; Ramaswamy, V.
The polymetallic nodules occur at an average depth of 4500 m. Adequate equipment and techniques are required for the exploration at such depths. Shipboard and various laboratory equipments for the sampling of polymetallic nodules is described...
... and Drug Administration (FDA) regulates the development and marketing of all laboratory tests that use test kits ... Cancer.gov en español Multimedia Publications Site Map Digital Standards for NCI Websites POLICIES Accessibility Comment Policy ...
Federal Laboratory Consortium — The inorganic Coatings Lab provides expertise to Navy and Joint Service platforms acquisition IPTs to aid in materials and processing choices which balance up-front...
Security Photon Sciences Physical Sciences & Engineering Energy Frontier Research Centers Scientific Publications Researchers Postdocs Exascale Computing Institute for Molecular Engineering at Argonne Work with Us About Safety News Careers Education Community Diversity Directory Argonne National Laboratory
Environmental Management System Pollution Prevention History 60 impacts Diversity Locations Facts & Figures Programs Nuclear Weapons About Nuclear Weapons Safety & Security Weapons Science & Technology Robotics R&D 100 Awards Laboratory Directed Research & Development Technology Deployment Centers
Federal Laboratory Consortium — Features 800,000 lb and 5,000,000 lb universal testing machines, and a dynamic test bed with broad fatigue-testing capabilities, and a wide range of instrumentation....
Federal Laboratory Consortium — Evaluation capabilities for air filtration devicesThe Air Filtration Lab provides testing of air filtration devices to demonstrate and validate new or legacy system...
Federal Laboratory Consortium — The geocentrifuge subjects a sample to a high-gravity field by spinning it rapidly around a central shaft. In this high-gravity field, processes, such as fluid flow,...
Federal Laboratory Consortium — Testing performance claims on heat transfer componentsThe Calorimeter Lab, located in the Ground Systems Power and Energy Lab (GSPEL), is one of the largest in the...
Federal Laboratory Consortium — At NETL’s Alloy Fabrication Facility in Albany, OR, researchers conduct DOE research projects to produce new alloys suited to a variety of applications, from gas...
Federal Laboratory Consortium — FUNCTION: Provides a secure environment to research and develop advanced electronic key management and networked key distribution technologies for the Navy and DoD....
Hellbron, J.L.; Seidel, R.W.
The birthplace of nuclear chemistry and nuclear medicine is the subject of this study of the Radiation Laboratory in Berkeley, California, where Ernest Lawrence used local and national technological, economic, and manpower resources to build the cyclotron
Described is the use of automated control using microcomputers. Covers the development of the microcontroller and describes advantages and characteristics of several brands of chips. Provides several recent applications of microcontrollers in laboratory automation. (MVL)
Boudreau, D A; Scheer, W D; Catrou, P G
The increasing level of technical sophistication and complexity found in clinical laboratory instrumentation today more than ever demands careful attention to maintenance service needs. The time-worn caution for careful definition of requirements for acquisition of a system should also carry over to acquisition of maintenance service. Guidelines are presented for specifications of terms and conditions for maintenance service from the perspective of the laboratorian in the automated clinical laboratory.
This book is in three sections; basic standards of laboratory design and equipment; procedures for safe laboratory practice; and the selection and use of essential biosafety equipment. The intention is that the guidance given in the book should have a broad basis and international application, and that it should be a source from which manuals applicable to local and special conditions can be usefully derived.
Saboe, Thomas J.
This paper discusses the process of managing automated systems through their life cycles within the quality-control (QC) laboratory environment. The focus is on the process of directing and managing the evolving automation of a laboratory; system examples are given. The author shows how both task and data systems have evolved, and how they interrelate. A BIG picture, or continuum view, is presented and some of the reasons for success or failure of the various examples cited are explored. Fina...
Many scientists are using a laboratory notebook when conducting experiments. The scientist documents each step, either taken in the experiment or afterwards when processing data. Due to computerized research systems, acquired data increases in volume and becomes more elaborate. This increases the need to migrate from originally paper-based to electronic notebooks with data storage, computational features and reliable electronic documentation. This talks describes a laboratory notebook bas...
P Junior, Oswaldo A.; Verli, Fernando; Lopes, Humberto E.
Usually, the oily water effluent from petroleum processes needs to be treated prior to its environment discard and/or reuse. The synthesis of such water effluent residues in an Oily Water Laboratory - equipped with Water Treatment Pilot Scale Units - is fundamental to the study and effectiveness comparison among the typical industrial water treatment processes. The Oily Water Laboratory will allow the reproduction - in a small scale - of any oily water effluent produced in the industrial PETROBRAS units - such reproduction can be obtained by using the same fluids, oily concentration, salinity, process temperature, particle size distribution etc. Such Laboratory also allows the performance analysis of typical industrial equipment used throughout the water treatment schemes (e.g., hydro-cyclones), resulting in design and/or operational guidelines for these industrial scale schemes. In the particular niche of very small diameter oil droplet removal, more efficient and non-conventional schemes - such as centrifuges and/or membrane filtration - will be also studied in the Laboratory. In addition, the Laboratory shall be used in the certification of in-line oily water analyzers (e.g., TOC - Total Organic Carbon and OWC - Oil Wax Content). This paper describes the characteristics of such Laboratory and its main operational philosophy. (author)
Almeida, Gevaldo L. de; Esteban, Adolfo; Almeida, Silvio G. de; Araujo, Radier M. de; Rocha, Zildete
A Laboratory Intercomparison Program involving Brazilian and Argentine laboratories, with the special participation of New Brunswick Laboratory - DOE and IAEA Seibersdorf Safeguards Laboratory, was implanted by ABACC having as main purpose to qualify a network to provide analytical services to this Agency on its role as administrator of the Common System of Accountability and Control of Nuclear Materials. For the first round robin of this Program, 15 laboratories were invited to perform elemental analysis on UO 2 samples, by using any desired method. Thirteen confirmed the participation and 10 reported the results. After an evaluation of the results by using a Two-Way Variance Analysis applied to a nested error model, it was found that 5 of them deviate less than 0.1% from the reference value established for the UO 2 uranium contents, being thus situated within the limits adopted for the target values, while the remaining ones reach a maximal deviation of 0.44%. The outcome of this evaluation, was sent to the laboratories, providing them with a feedback to improve their performance by applying corrective actions to the detected sources of errors or bias related to the methods techniques and procedures. (author)
These reports summarize pollution prevention opportunity assessments conducted jointly by EPA and DOE at the Geochemistry Laboratory and the Manufacturing and Fabrication Repair Laboratory at the Department of Energy's Sandia National Laboratories facility in Albuquerque, New Mex...
The grand opening of NASA's new, world-class laboratory for research into future space transportation technologies located at the Marshall Space Flight Center (MSFC) in Huntsville, Alabama, took place in July 2004. The state-of-the-art Propulsion Research Laboratory (PRL) serves as a leading national resource for advanced space propulsion research. Its purpose is to conduct research that will lead to the creation and development of innovative propulsion technologies for space exploration. The facility is the epicenter of the effort to move the U.S. space program beyond the confines of conventional chemical propulsion into an era of greatly improved access to space and rapid transit throughout the solar system. The laboratory is designed to accommodate researchers from across the United States, including scientists and engineers from NASA, the Department of Defense, the Department of Energy, universities, and industry. The facility, with 66,000 square feet of useable laboratory space, features a high degree of experimental capability. Its flexibility allows it to address a broad range of propulsion technologies and concepts, such as plasma, electromagnetic, thermodynamic, and propellant propulsion. An important area of emphasis is the development and utilization of advanced energy sources, including highly energetic chemical reactions, solar energy, and processes based on fission, fusion, and antimatter. The Propulsion Research Laboratory is vital for developing the advanced propulsion technologies needed to open up the space frontier, and sets the stage of research that could revolutionize space transportation for a broad range of applications.
Skinner, E.L.; Watterson, C.A.; Chemerys, J.C.
Safety, defined as 'freedom from danger, risk, or injury,' is difficult to achieve in a laboratory environment. Inherent dangers, associated with water analysis and research laboratories where hazardous samples, materials, and equipment are used, must be minimized to protect workers, buildings, and equipment. Managers, supervisors, analysts, and laboratory support personnel each have specific responsibilities to reduce hazards by maintaining a safe work environment. General rules of conduct and safety practices that involve personal protection, laboratory practices, chemical handling, compressed gases handling, use of equipment, and overall security must be practiced by everyone at all levels. Routine and extensive inspections of all laboratories must be made regularly by qualified people. Personnel should be trained thoroughly and repetitively. Special hazards that may involve exposure to carcinogens, cryogenics, or radiation must be given special attention, and specific rules and operational procedures must be established to deal with them. Safety data, reference materials, and texts must be kept available if prudent safety is to be practiced and accidents prevented or minimized.
The Turkish Government formally requested that the Agency provide for one year the services of an expert in the agricultural applications of radioisotopes. Specifically, they wanted this expert first of all to assist in setting up and equipping a pioneer laboratory for the utilization of radioisotopes in agricultural research. Once the laboratory was in operation, the expert was to initiate various research projects using isotope techniques, and to train personnel to carry on this work. The Agency was also asked to supply various specialized equipment for the laboratory, including some radioisotopes. On 10 December 1960 the first phase was complete - the new laboratory was formally opened. It is foreseen that the research projects which will be initiated at the laboratory will include the following: determination of the effect of fertilizers upon yield and quality of field crops and fruit trees, soil fertility studies, studies of mineral element uptake and localization of nutrients in plant body, studies of the folar application of mineral nutrients, especially in fruit trees, investigation of microelements in field crops and fruit trees, investigation of pollination problems, study of the distribution of mineral elements in different fruit seedlings, study of the uptake of nutrients by fruit trees during the rest period, dispersal studies on insects, insecticide studies
The Turkish Government formally requested that the Agency provide for one year the services of an expert in the agricultural applications of radioisotopes. Specifically, they wanted this expert first of all to assist in setting up and equipping a pioneer laboratory for the utilization of radioisotopes in agricultural research. Once the laboratory was in operation, the expert was to initiate various research projects using isotope techniques, and to train personnel to carry on this work. The Agency was also asked to supply various specialized equipment for the laboratory, including some radioisotopes. On 10 December 1960 the first phase was complete - the new laboratory was formally opened. It is foreseen that the research projects which will be initiated at the laboratory will include the following: determination of the effect of fertilizers upon yield and quality of field crops and fruit trees, soil fertility studies, studies of mineral element uptake and localization of nutrients in plant body, studies of the folar application of mineral nutrients, especially in fruit trees, investigation of microelements in field crops and fruit trees, investigation of pollination problems, study of the distribution of mineral elements in different fruit seedlings, study of the uptake of nutrients by fruit trees during the rest period, dispersal studies on insects, insecticide studies.
The report covers the research activities of the Physics laboratory of H.C. Oersted Institute, University of Copenhagen in the period January 1, 1976 - January 1, 1979. It gives also an idea about the teaching carried out by yhe laboratory. The research - broadly speaking - deals mainly with the interaction of particles (ions, electrons and neutrons) and electromagnetic radiation (X-rays) with matter. Use is made in studies of: atomic physics, radiation effects, surface physics, the electronic and crystallographic structure of matter and some biological problems. The research is carried out partly in the laboratory itself and partly at and in collaboration with other institutes in this country (H.C. Oersted Institute, Chemical Laboratories, Denmark's Technical University, Aarhus University, Institute of Physics and Risoe National Laboratory) and abroad (Federal Republic of Germany, France, India, Sweden, U.K., U.S.A. and U.S.S.R.). All these institutes are listed in the abstract titles. Bibliography comprehends 94 publications. A substantial part of the research is supported by the Danish Natural Sciences Research Council. (author)
Federal Laboratory Consortium — Soil/Rock Properties LaboratoryLocation: Spokane SiteThe Soil/Rock Properties Laboratory is contained in the soils bay, a 4,700 sq. ft. facility that provides space...
Federal Laboratory Consortium — Program CapabilitiesSJN-DO Pharmaceutical Laboratory is an A2LA/ISO/IEC 17025 accredited National Servicing Laboratory specialized in Drug Analysis, is a member of...
Full Text Available Vladimir Tatochenko1, Vasily Uchaikin2, Aleksandr Gorelov3, Konstantin Gudkov4, Andrew Campbell5, Gregory Schulz5, Rebecca Prahl5, Gerard Notario51Scientific Centre of Children’s Health, Russian Academy of Medical Sciences, Lomonosovskiy Prospect, Moscow, Russia; 2Russian State Medical University of Roszdrav, Moscow, Russia; 3Central Scientific Research Institution of Epidemiology, Moscow, Russia; 4Abbott Laboratories LLC, Khimki, Moscow, Russia; 5Abbott Laboratories, Abbott Park, IL, USABackground: Respiratory syncytial virus (RSV is the leading cause of severe lower respiratory tract infections among infants and young children, and is responsible for an estimated four million deaths per year globally. A monthly injection of palivizumab has been used for prophylaxis of serious RSV infections among high-risk children in 71 countries since 1998 and approval for use in the Russian Federation was obtained in February 2010. A recommendation for RSV prophylaxis in the Russian Federation would require knowledge of the prevalence and seasonality of RSV in that country.Methods: In a prospective, multicenter, epidemiological study of the prevalence, seasonality, and peak occurrence of RSV infection, children aged ≤2 years hospitalized for lower respiratory tract infections in three regions of the Russian Federation, from September 2008 through April 2009, were screened and tested for RSV using rapid immunochromatography of nasopharyngeal lavage. For subjects who were tested positive, hospitalization data were collected.Results: Of 519 children aged ≤2 years enrolled from September 11, 2008 through April 26, 2009, 197 tested positive for RSV (38.0%, 95% CI: 33.8, 42.3. The onset of the 2008–2009 RSV season in the Russian Federation occurred in late October 2008, similar to what is observed in other northern temperate zones. Peak activity occurred in early April 2009, when 62% of children enrolled tested positive for RSV.Conclusion: The prevalence
Comparison of semi-automatized assays for anti-T. gondii IgG detection in low-reactivity serum samples: importance of the results in patient counseling Comparação de ensaios semi-automatizados para pesquisa de IgG anti-T. gondii em amostras de soros de baixa reatividade: importância dos resultados no aconselhamento do paciente
Paulo Guilherme Leser
Full Text Available Toxoplasmosis is a disease which can cause severe congenital infection and is normally diagnosed by the detection of T. gondii specific antibodies in the serum of infected patients. Several different tests allow to distinguish recent from past infections and to quantify anti-T. gondii specific IgG, and the results can be used as markers for immunity. In the present study, we compare the performance of two different methodologies, the Elfa (bioMérieux S.A and the Meia (Abbott Laboratories in detecting T. gondii specific IgG in low-reactivity sera. Of 76 analyzed samples, three presented discrepant results, being positive in the Abbott AxSYM Toxo IgG assay, and negative in the bioMérieux Vidas Toxo IgG II assay. By using other tests, the three sera were confirmed to be negative. The results are discussed in the context of their importance for patient management, especially during pregnancy.Toxoplasmose, doença conhecida por sua severidade na infecção congênita é geralmente diagnosticada pela demonstração de anticorpos específicos contra antígenos de T. gondii, presentes no soro de indivíduos infectados. Diferentes testes são disponíveis para diferenciar infecção recente de infecção pregressa, para quantificar anticorpos IgG anti-T. gondii nos soros dos pacientes e utilizar os resultados como marcadores de imunidade. Neste trabalho apresentamos os resultados do estudo comparativo de duas tecnologias, Elfa (bioMérieux S.A. e Meia (Abbott Laboratories, para pesquisa de anticorpos IgG anti-T. gondii em amostras de soros de baixa reatividade. De 76 amostras processadas, três apresentaram resultados discrepantes, reagentes para AxSYM Toxo IgG e não-reagentes para Vidas Toxo IgG II. A confirmação dos resultados, feita por bateria de testes, mostrou que todas as três amostras eram não-reagentes. Os resultados são discutidos em sua importância e orientação clínica, principalmente para a paciente gestante.
to create a new methodology for developing and exploring process models and applications. The paper outlines the process innovation laboratory as a new approach to BPI. The process innovation laboratory is a comprehensive framework and a collaborative workspace for experimenting with process models....... The process innovation laboratory facilitates innovation by using an integrated action learning approach to process modelling in a controlled environment. The study is based on design science and the paper also discusses the implications to EIS research and practice......Most organizations today are required not only to operate effective business processes but also to allow for changing business conditions at an increasing rate. Today nearly every business relies on their enterprise information systems (EIS) for process integration and future generations of EIS...
Wisniewski, Rafal; Bak, Thomas
The Satellite Laboratory at the Department of Control Engineering of Aalborg University (SatLab) is a dynamic motion facility designed for analysis and test of micro spacecraft. A unique feature of the laboratory is that it provides a completely gravity-free environment. A test spacecraft......-axis magnetometer, three piezoelectric gyros, and four reaction wheels in a tetrahedron configuration. The operation of the spacecraft is fully autonomous. The data flow between the transducers and the onboard computer placed physically outside the satellite is provided by a radio link. The purpose...... can be implemented in the laboratory, e.g. three-axis attitude control, slew manoeuvres, spins stabilization using magnetic actuation and/or reaction wheels. The spacecraft attitude can be determined applying magnetometer measurements....
This report covers the activity of the Linear Accelerator Laboratory during the period June 1974-June 1976. The activity of the Laboratory is essentially centered on high energy physics. The main activities were: experiments performed with the colliding rings (ACO), construction of the new colliding rings and beginning of the work at higher energy (DCI), bubble chamber experiments with the CERN PS neutrino beam, counter experiments with CERN's PS and setting-up of equipment for new experiments with CERN's SPS. During this period a project has also been prepared for an experiment with the new PETRA colliding ring at Hamburg. On the other hand, intense collaboration with the LURE Laboratory, using the electron synchrotron radiation emitted by ACO and DCI, has been developed [fr
Soraya E. Morales-López
Full Text Available Tuberculosis (TB is a major cause of worldwide mortality. We report the case of a non-HIV-infected woman with clinical suspicion of pleural tuberculosis and contradictory results between Xpert® MTB/RIF and Abbott RealTime MTB assays from pleural fluid specimen. Liquid and solid cultures for tuberculosis were performed with negative results. The patient received treatment, and clinical improvement was observed. Both techniques detect Mycobacterium tuberculosis complex, but they have different targets and limits of detection. Abbott RealTime MTB results correlated well with the clinical findings of the patient.
Morales-López, Soraya E; Yepes, Jayr A; Anzola, Irina; Aponte, Hernán; Llerena-Polo, Claudia R
Tuberculosis (TB) is a major cause of worldwide mortality. We report the case of a non-HIV-infected woman with clinical suspicion of pleural tuberculosis and contradictory results between Xpert ® MTB/RIF and Abbott RealTime MTB assays from pleural fluid specimen. Liquid and solid cultures for tuberculosis were performed with negative results. The patient received treatment, and clinical improvement was observed. Both techniques detect Mycobacterium tuberculosis complex, but they have different targets and limits of detection. Abbott RealTime MTB results correlated well with the clinical findings of the patient. Copyright © 2016 The Authors. Published by Elsevier Ltd.. All rights reserved.
A false Roche carcinoembryonic antigen (CEA) activity could be detected in all commercial and noncommercial heparin preparations examined. The possibility of 'due to contamination' has been ruled out. Using the Roche procedure, heparin solutions, in the absence of CEA, gave positive CEA activity; on the other hand, no CEA activity was detected in solutions containing only heparin when the Abbott Kit was used. When heparin was present in specimens containing CEA, the Abbott Kit underestimated the CEA activity, whereas the Roche Kit gave false elevated values. However, the negative effect of heparin could be reduced by heat treatment in the presence of plasma proteins. (Auth.)
Naberhuis,Jane K; Hunt,Vivienne; Bell,Jvawnna; Partridge,Jamie; Goates,Scott; Nuijten,Mark
Jane K Naberhuis,1 Vivienne N Hunt,2 Jvawnna D Bell,3 Jamie S Partridge,3 Scott Goates,3 Mark JC Nuijten4 1Division of Nutritional Sciences, University of Illinois at Urbana-Champaign, Urbana, IL, USA; 2Abbott Nutrition, Research and Development, Singapore; 3Abbott Nutrition, Research and Development, Columbus, OH, USA; 4A2M (Ars Accessus Medica), Amsterdam, The Netherlands Background and aims: As policy-makers assess the value of money spent on health care, researc...
Naberhuis JK; Hunt VN; Bell JD; Partridge JS; Goates S; Nuijten MJC
Jane K Naberhuis,1 Vivienne N Hunt,2 Jvawnna D Bell,3 Jamie S Partridge,3 Scott Goates,3 Mark JC Nuijten4 1Division of Nutritional Sciences, University of Illinois at Urbana-Champaign, Urbana, IL, USA; 2Abbott Nutrition, Research and Development, Singapore; 3Abbott Nutrition, Research and Development, Columbus, OH, USA; 4A2M (Ars Accessus Medica), Amsterdam, The Netherlands Background and aims: As policy-makers assess the value of money spent on health care, research in the fie...
Royal, P D
Accreditation or certification is a recognition given to an operation or product that has been evaluated against a standard; be it regulatory or voluntary. The purpose of accreditation is to provide the consumer with a level of confidence in the quality of operation (process) and the product of an organization. Environmental Protection Agency/OCM has proposed the development of an accreditation program under National Environmental Laboratory Accreditation Program for Good Laboratory Practice (GLP) laboratories as a supplement to the current program. This proposal was the result of the Inspector General Office reports that identified weaknesses in the current operation. Several accreditation programs can be evaluated and common components identified when proposing a structure for accrediting a GLP system. An understanding of these components is useful in building that structure. Internationally accepted accreditation programs provide a template for building a U.S. GLP accreditation program. This presentation will discuss the traditional structure of accreditation as presented in the Organization of Economic Cooperative Development/GLP program, ISO-9000 Accreditation and ISO/IEC Guide 25 Standard, and the Canadian Association for Environmental Analytical Laboratories, which has a biological component. Most accreditation programs are managed by a recognized third party, either privately or with government oversight. Common components often include a formal review of required credentials to evaluate organizational structure, a site visit to evaluate the facility, and a performance evaluation to assess technical competence. Laboratory performance is measured against written standards and scored. A formal report is then sent to the laboratory indicating accreditation status. Usually, there is a scheduled reevaluation built into the program. Fee structures vary considerably and will need to be examined closely when building a GLP program.
Sautter, Robert L; Thomson, Richard B
The manner in which medical care is reimbursed in the United States has resulted in significant consolidation in the U.S. health care system. One of the consequences of this has been the development of centralized clinical microbiology laboratories that provide services to patients receiving care in multiple off-site, often remote, locations. Microbiology specimens are unique among clinical specimens in that optimal analysis may require the maintenance of viable organisms. Centralized laboratories may be located hours from patient care settings, and transport conditions need to be such that organism viability can be maintained under a variety of transport conditions. Further, since the provision of rapid results has been shown to enhance patient care, effective and timely means for generating and then reporting the results of clinical microbiology analyses must be in place. In addition, today, increasing numbers of patients are found to have infection caused by pathogens that were either very uncommon in the past or even completely unrecognized. As a result, infectious disease specialists, in particular, are more dependent than ever on access to high-quality diagnostic information from clinical microbiology laboratories. In this point-counterpoint discussion, Robert Sautter, who directs a Charlotte, NC, clinical microbiology laboratory that provides services for a 40-hospital system spread over 3 states in the southeastern United States explains how an integrated clinical microbiology laboratory service has been established in a multihospital system. Richard (Tom) Thomson of the NorthShore University HealthSystem in Evanston, IL, discusses some of the problems and pitfalls associated with large-scale laboratory consolidation. Copyright © 2015, American Society for Microbiology. All Rights Reserved.
Federal Laboratory Consortium — This laboratory evaluates fundamental friction, wear, and lubrication technologies for improved, robust, and power-dense vehicle transmissions. The facility explores...
Federal Laboratory Consortium — These laboratories conduct a wide range of studies to characterize the sensory properties of and consumer responses to foods, beverages, and other consumer products....
Federal Laboratory Consortium — FUNCTION: Provides an environment for developing and evaluating intelligent software for both actual and simulated autonomous vehicles. Laboratory computers provide...
Federal Laboratory Consortium — The Atlantic Oceanographic and Meteorological Laboratory conducts research to understand the physical, chemical, and biological characteristics and processes of the...
Federal Laboratory Consortium — The purpose is to develop advanced, more effective high-resolution micro-NDE materials characterization methods using scanning acoustic microscopy. The laboratory's...
Federal Laboratory Consortium — The Laboratory Animal Sciences Program (LASP) is a comprehensive resource for scientists performing animal-based research to gain a better understanding of cancer,...
Federal Laboratory Consortium — Purpose:The Integrated Support Environment (ISE) Laboratory serves the fleet, in-service engineers, logisticians and program management offices by automatically and...
The various research projects and investigations carried out at the laboratory are briefly described. These include:- hormone investigations (thyroxine and triiodothyronine) by radioimmunology in cattle and swine; the synthesis of fatty acids in sheep digestive juices; vitamin E in pigs; the uptake of phosphorus in cloudberries; the uptake and breaking down of glyphosate in spruce and wild oats; transport and assimilation of MCPA; ground water pollution from sewage; process investigations in fish oil production; cleaning process in dairy piping; soil humidity radiometric gage calibration; mass spectroscopy. The courses held by the laboratory for students and the consumption of radioisotope tracers are summarised. (JIW)
Lin, Shin Ted; Yue, Qian
Deep underground laboratories in Asia have been making huge progress recently because underground sites provide unique opportunities to explore the rare-event phenomena for the study of dark matter searches, neutrino physics and nuclear astrophysics as well as the multi-disciplinary researches based on the low radioactive environments. The status and perspectives of Kamioda underground observatories in Japan, the existing Y2L and the planned CUP in Korea, India-based Neutrino Observatory (INO) in India and China JinPing Underground Laboratory (CJPL) in China will be surveyed
Ramirez J, F.J.
The Radiation detectors laboratory was established with the assistance of the International Atomic Energy Agency which gave this the responsibility to provide its services at National and regional level for Latin America and it is located at the ININ. The more expensive and delicate radiation detectors are those made of semiconductor, so it has been put emphasis in the use and repairing of these detectors type. The supplied services by this laboratory are: selection consultant, detectors installation and handling and associated systems. Installation training, preventive and corrective maintenance of detectors and detection systems calibration. (Author)
Lin, Shin Ted, E-mail: email@example.com [College of Physical Science and Technology, Sichuan University, Chengdu 610064 China (China); Yue, Qian, E-mail: firstname.lastname@example.org [Key Laboratory of Particle and Radiation Imaging (Ministry of Education) and Department of Engineering Physics, Tsinghua University, Beijing 100084 China (China)
Deep underground laboratories in Asia have been making huge progress recently because underground sites provide unique opportunities to explore the rare-event phenomena for the study of dark matter searches, neutrino physics and nuclear astrophysics as well as the multi-disciplinary researches based on the low radioactive environments. The status and perspectives of Kamioda underground observatories in Japan, the existing Y2L and the planned CUP in Korea, India-based Neutrino Observatory (INO) in India and China JinPing Underground Laboratory (CJPL) in China will be surveyed.
Saboe, T J
This paper discusses the process of managing automated systems through their life cycles within the quality-control (QC) laboratory environment. The focus is on the process of directing and managing the evolving automation of a laboratory; system examples are given. The author shows how both task and data systems have evolved, and how they interrelate. A BIG picture, or continuum view, is presented and some of the reasons for success or failure of the various examples cited are explored. Finally, some comments on future automation need are discussed.
Heller, Adam; Cairns, Elton J
Adam Heller, Ernest Cockrell Sr. Chair in Engineering Emeritus of the John J. McKetta Department of Chemical Engineering at The University of Texas at Austin, recalls his childhood in the Holocaust and his contributions to science and technology that earned him the US National Medal of Technology and Innovation in a conversation with Elton J. Cairns, Professor of Chemical and Biomolecular Engineering at the University of California, Berkeley. Dr. Heller, born in 1933, describes the enslavement of his father by Hungarians in 1942; the confiscation of his family's home, business, and all its belongings in 1944; and his incarceration in a brick factory with 18,000 Jews who were shipped by the Hungarians to be gassed by Germans in Auschwitz. Dr. Heller and his immediate family survived the Holocaust and arrived in Israel in 1945. He studied under Ernst David Bergmann at the Hebrew University, and then worked at Bell Laboratories and GTE Laboratories, where he headed Bell Lab's Electronic Materials Research Department. At GTE Laboratories, he built in 1966 the first neodymium liquid lasers and in 1973 with Jim Auborn conceived and engineered the lithium thionyl chloride battery, one of the first to be manufactured lithium batteries, which is still in use. After joining the faculty of engineering of The University of Texas at Austin, he cofounded with his son Ephraim Heller TheraSense, now a major part of Abbott Diabetes Care, which produced a microcoulometer that made the monitoring of glucose painless by accurately measuring the blood glucose concentration in 300 nL of blood. He also describes the electrical wiring of enzymes, the basis for Abbott's state-of-the-art continuous glucose monitoring system. He discusses his perspective of reducing the risk of catastrophic global warming in a wealth-accumulating, more-energy-consuming world and provides advice for students entering careers in science or engineering.
.... FDA-2010-N-0548] Good Laboratory Practice for Nonclinical Laboratory Studies AGENCY: Food and Drug... (FDA) is seeking comment on whether to amend the regulations governing good laboratory practices (GLPs..., 1978 (43 FR 60013). As stated in its scope (Sec. 58.1), this regulation prescribes good laboratory...
Nuclear astrophysics is concerned with nuclear reaction and decay processes from the Big Bang to the present star generation controlling the chemical evolution of our universe. Such nuclear reactions maintain stellar life, determine stellar evolution, and finally drive stellar explosion in the circle of stellar life. Laboratory nuclear astrophysics seeks to simulate and understand the underlying processes using a broad portfolio of nuclear instrumentation, from reactor to accelerator from stable to radioactive beams to map the broad spectrum of nucleosynthesis processes. This talk focuses on only two aspects of the broad field, the need of deep underground accelerator facilities in cosmic ray free environments in order to understand the nucleosynthesis in stars, and the need for high intensity radioactive beam facilities to recreate the conditions found in stellar explosions. Both concepts represent the two main frontiers of the field, which are being pursued in the US with the CASPAR accelerator at the Sanford Underground Research Facility in South Dakota and the FRIB facility at Michigan State University.
Cooper, Robert C.; And Others
This laboratory manual presents information and techniques dealing with aquatic microbiology as it relates to environmental health science, sanitary engineering, and environmental microbiology. The contents are divided into three categories: (1) ecological and physiological considerations; (2) public health aspects; and (3)microbiology of water…
R and D activities on RF Superconductivity have continued at Saclay during the last two years. An important effort has been made to update a picture of the laboratory latest results. A mere 'table of contents' of 19 contributed papers are summarized. (R.P.)
Presents a simple, 10-item quiz designed to make students aware that they must learn laboratory safety. The items include questions on acid/base accidents, several types of fire extinguishers, and safety glassses. Answers and some explanations are included. (DH)
This podcast highlights the importance of integrating laboratory services to maximize service delivery to patients. Created: 4/4/2011 by National Center for HIV/AIDS, Viral Hepatitis, STD, and TB Prevention (NCHHSTP). Date Released: 4/7/2011.
The report summarizes the main activities of the Linear Electron Accelerator Section of the Physics Laboratory of the State University of Ghent. The research fields are relative to: 1. Nuclear fission. 2. Photonuclear reactions. 3. Nuclear spectroscopy and positron annihilation. 4. Dosimetry. 5. Theoretical studies. (MDC)
While nuclear technology continues to expand in all scientific fields, both research and analysis become increasingly important aspects of the work carried out at the IAEA's two principal laboratories at Seibersdorf and Monaco. They also provide training facilities for students and graduates from many Member States. The following outlines give a brief history of their development, and their present work. (author)
Sailor, David Jean [Portland State Univ., Portland, OR (United States)
This project provided support to the Green Building Research Laboratory at Portland State University (PSU) so it could work with researchers and industry to solve technical problems for the benefit of the green building industry. It also helped to facilitate the development of PSU’s undergraduate and graduate-level training in building science across the curriculum.
The report summarizes the main activities of the Linear Electron Accelerator Section of the Physics Laboratory of the State University of Ghent. The research fields are relative to: 1. Nuclear fission. 2. Photonuclear reactions. 3. Nuclear spectroscopy and positron annihilation. 4. Dosimetry. 5. Theoretical studies. (MDC)
Giraud, B. G.
We compare several definitions of the density of a self-bound system, such as a nucleus, in relation with its center-of-mass zero-point motion. A trivial deconvolution relates the internal density to the density defined in the laboratory frame. This result is useful for the practical definition of density functionals.
Ramirez J, F.J.
The National Institute for Nuclear Research has established a Radiation detector laboratory that has the possibility of providing to the consultants on the handling and applications of the nuclear radiation detectors. It has special equipment to repair the radiation detectors used in spectroscopy as the hyper pure Germanium for gamma radiation and the Lithium-silica for X-rays. There are different facilities in the laboratory that can become useful for other institutions that use radiation detectors. This laboratory was created to satisfy consultant services, training and repairing of the radiation detectors both in national and regional levels for Latin America. The laboratory has the following sections: Nuclear Electronic Instrumentation; where there are all kind of instruments for the measurement and characterization of detectors like multichannel analyzers of pulse height, personal computers, amplifiers and nuclear pulse preamplifiers, nuclear pulses generator, aleatories, computer programs for radiation spectra analysis, etc. High vacuum; there is a vacuum escape measurer, two high vacuum pumps to restore the vacuum of detectors, so the corresponding measurers and the necessary tools. Detectors cleaning; there is an anaerobic chamber for the detectors handling at inert atmosphere, a smoke extraction bell for cleaning with the detector solvents. Cryogenic; there are vessels and tools for handling liquid nitrogen which is used for cooling the detectors when they required it. (Author)
While nuclear technology continues to expand in all scientific fields, both research and analysis become increasingly important aspects of the work carried out at the IAEA's two principal laboratories at Seibersdorf and Monaco. They also provide training facilities for students and graduates from many Member States. The following outlines give a brief history of their development, and their present work. (author)
The report summarizes the main activities of the linear Electron Accelerator Section of the Physics Laboratory of the State University of Ghent. The research fields are relative to: 1. Nuclear fission 2. Photonuclear reactions 3. Nuclear spectroscopy and positron annihilation 4. Dosimetry 5. Theoretical studies. (MDC)
Kanare, Howard M.
The purpose of this book is to teach the principles of proper scientific notekeeping. The principles presented in this book are goals for which working scientists must strive. Chapter 1, "The Reasons for Notekeeping," is an overview of the process of keeping a laboratory notebook. Chapter 2, "The Hardware of Notekeeping," is intended especially…
Mustafa, Ajaz; Farooq, A Jan; Qadri, Gj; S A, Tabish
Health and safety in clinical laboratories is becoming an increasingly important subject as a result of emergence of highly infectious diseases such as Hepatitis and HIV. A cross sectional study was carried out to study the safety measures being adopted in clinical laboratories of India. Heads of laboratories of teaching hospitals of India were subjected to a standardized, pretested questionnaire. Response rate was 44.8%. only 60% of laboratories had person in-charge of safety in laboratory. Seventy three percent of laboratories had safety education program regarding hazards. In 91% of laboratories staff is using protective clothing while working in laboratories. Hazardous material regulations are followed in 78% of laboratories. Regular health check ups are carried among laboratory staff in 43.4% of laboratories.Safety manual is available in 56.5% of laboratories. 73.9% of laboratories are equipped with fire extinguishers. Fume cupboards are provided in 34.7% of laboratories and they are regularly checked in 87.5% of these laboratories. In 78.26% of laboratories suitable measures are taken to minimize formation of aerosols.In 95.6% of laboratories waste is disposed off as per bio-medical waste management handling rules. Laboratory of one private medical college was accredited with NABL and safety parameters were better in that laboratory. Installing safety engineered devices apparently contributes to significant decrease in injuries in laboratories; laboratory safety has to be a part of overall quality assurance programme in hospitals. Accreditation has to be made necessary for all laboratories.
Full text: The activities of the Health Physics Laboratory at the Institute of Nuclear Physics in Cracow are principally research in the general area of radiation physics, and radiation protection of the employees of the Institute of Nuclear Physics. Theoretical research concerns modelling of radiation effects in radiation detectors and studies of concepts in radiation protection. Experimental research, in the general area of solid state dosimetry, is primarily concerned with thermoluminescence (TL) dosimetry, and more specifically: development of LiF:Mg, Ti and CVD diamond detectors for medical applications in conventional and hadron radiotherapy and of LiF:Mg, Cu, P for low-level natural external ionising radiation. Environmental radiation measurements (cosmic-rays on aircraft and radon in dwellings and soil) are also performed using track CR-39 and TLD detectors. The Laboratory provides expert advice on radiation protection regulations at national and international levels. Routine work of the Health Physics Laboratory involves design and maintenance of an in-house developed TL-based personnel dosimetry system for over 200 radiation workers at the INP, supervision of radiation safety on INP premises, and advising other INP laboratories on all matters pertaining to radiation safety. We provide personal and environmental TLD dosimetry service for several customers outside the INP, mainly in hospitals and nuclear research institutes in Poland. We also calibrate radiation protection instruments for customers in southern Poland. The year 2000 was another eventful year for the Health Physics Laboratory. We started three new research projects granted by the Polish State Committee of Scientific Research. Mr P. Bilski co-ordinates the project on the measurements of radiation doses on board of commercial aircraft of Polish LOT Airlines. Dr B. Marczewska and I worked on the application of artificial diamonds for dosimetry of ionising radiation. We also participate in a
The activities of the Health Physics Laboratory at the Institute of Nuclear Physics in Cracow are principally research in the general area of radiation physics, and radiation protection of the employees of the Institute of Nuclear Physics. Theoretical research concerns modelling of radiation effects in radiation detectors and studies of concepts in radiation protection. Experimental research, in the general area of solid state dosimetry, is primarily concerned with thermoluminescence (TL) dosimetry, and more specifically: development of LiF:Mg, Ti for medical applications in conventional and hadron radiotherapy, and of LiF:Mg, Cu, P for low-level natural external ionising radiation. Environmental radiation measurements (radon in dwellings and in soil air) are also performed using track detectors. The Laboratory provides expert advice on radiation protection regulations at national and international levels. Routine work of the Health Physics Laboratory involves design and maintenance of an in-house developed TL-based personnel dosimetry system for over 200 radiation workers at the INP, monitoring and supervision of radiation safety on INP premises, and advising other INP laboratories on all matters pertaining to radiation safety. The year 1998 was another eventful year for the Health Physics Laboratory. In retrospective, the main effort in 1998 has been directed towards preparation and participation in the 12th International Conference on Solid State Dosimetry in Burgos, Spain. One of the research projects is aimed at developing novel miniature TLD detectors with improved LET and dose characteristics for precise phantom measurements in eye cancer radiotherapy with proton beams. The second project concerns the application of ultra-sensitive LiF:Mg, Cu, P (MCP-N) TLD detectors in environmental monitoring of gamma ionising radiation. The main objective of this last project is to develop and to test a system for rapid, short-term monitoring of environmental radiation
American Chemical Society, Washington, DC.
A primary goal of the American Chemical Society Task Force on Laboratory Waste Management is to provide laboratories with the information necessary to develop effective strategies and training programs for managing laboratory wastes. This book is intended to present a fresh look at waste management from the laboratory perspective, considering both…
Troxell, Wade [Colorado State Univ., Fort Collins, CO (United States)
The initial federal funding for the Colorado State University Smart Grid Integration Laboratory is through a Congressionally Directed Project (CDP), DE-OE0000070 Smart Grid Integration Laboratory. The original program requested in three one-year increments for staff acquisition, curriculum development, and instrumentation all which will benefit the Laboratory. This report focuses on the initial phase of staff acquisition which was directed and administered by DOE NETL/ West Virginia under Project Officer Tom George. Using this CDP funding, we have developed the leadership and intellectual capacity for the SGIC. This was accomplished by investing (hiring) a core team of Smart Grid Systems engineering faculty focused on education, research, and innovation of a secure and smart grid infrastructure. The Smart Grid Integration Laboratory will be housed with the separately funded Integrid Laboratory as part of CSU's overall Smart Grid Integration Center (SGIC). The period of performance of this grant was 10/1/2009 to 9/30/2011 which included one no cost extension due to time delays in faculty hiring. The Smart Grid Integration Laboratory's focus is to build foundations to help graduate and undergraduates acquire systems engineering knowledge; conduct innovative research; and team externally with grid smart organizations. Using the results of the separately funded Smart Grid Workforce Education Workshop (May 2009) sponsored by the City of Fort Collins, Northern Colorado Clean Energy Cluster, Colorado State University Continuing Education, Spirae, and Siemens has been used to guide the hiring of faculty, program curriculum and education plan. This project develops faculty leaders with the intellectual capacity to inspire its students to become leaders that substantially contribute to the development and maintenance of Smart Grid infrastructure through topics such as: (1) Distributed energy systems modeling and control; (2) Energy and power conversion; (3
Van Mieghem, Nicolas M; Daemen, Joost; den Uil, Corstiaan; Dur, Onur; Joziasse, Linda; Maugenest, Anne-Marie; Fitzgerald, Keif; Parker, Chris; Muller, Paul; van Geuns, Robert-Jan
The HeartMate PHP (percutaneous heart pump) is a second-generation transcatheter axial flow circulatory support system. The collapsible catheter pump is inserted through a 14 Fr sheath, deployed across the aortic valve expanding to 24 Fr and able to deliver up to 5 L/min blood flow at minimum haemolytic risk. As such, this device may be a valuable adjunct to percutaneous coronary intervention (PCI) of challenging lesions in high-risk patients or treatment of cardiogenic shock. This technical report discusses: (i) the HeartMate PHP concept, (ii) the implantation technique, (iii) the haemodynamic performance in an in vitro cardiovascular flow testing set-up, and (iv) preliminary clinical experience. An update on the device, produced by St. Jude Medical/Abbott Laboratories, can be found in the Appendix.
Romero De Pablos, Ana
Patents on antibiotics were introduced in Spain in 1949. Preliminary research reveals diversification in the types of antibiotics: patents relating to penicillin were followed by those relating to streptomycin, erythromycin and tetracycline. There was also diversification in the firms that applied for patents: while Merck & Co. Incorporated and Schenley Industries Inc. were the main partners with Spanish antibiotics manufacturers in the late 1940s, this industrial space also included many others, such as Eli Lilly & Company, Abbott Laboratories, Chas. Pfizer & Co. Incorporated, and American Cyanamid Company in the mid-1970s. The introduction of these drugs in Spain adds new elements to a re-evaluation of the autarkic politics of the early years of the Franco dictatorship.
Fan, Leimin; Lee, Jacob; Hall, Jeffrey; Tolentino, Edward J; Wu, Huaiqin; El-Shourbagy, Tawakol
This article describes validation work for analysis of an Abbott investigational drug (Compound A) in monkey whole blood with dried blood spots (DBS). The impact of DBS spotting volume on analyte concentration was investigated. The quantitation range was between 30.5 and 10,200 ng/ml. Accuracy and precision of quality controls, linearity of calibration curves, matrix effect, selectivity, dilution, recovery and multiple stabilities were evaluated in the validation, and all demonstrated acceptable results. Incurred sample reanalysis was performed with 57 out of 58 samples having a percentage difference (versus the mean value) less than 20%. A linear relationship between the spotting volume and the spot area was drawn. The influence of spotting volume on concentration was discussed. All validation results met good laboratory practice acceptance requirements. Radial spreading of blood on DBS cards can be a factor in DBS concentrations at smaller spotting volumes.
Spitz, J; Braun, J S; Schmidt, M [Krankenhausstiftung Bamberg (Germany, F.R.). Nuklearmedizinische Abt.; Abt fuer Labormedizin, Krankenhausstiftung Bamberg [Germany, F.R.
The results of two radioimmunoassays (RIA, precipitating technique), of a homogenous (EMIT) and a heterogenous (ELISA) enzyme immunoassay (EIA) for ascertaining the amounts of digoxin showed a good correlation in precision and a reasonably AK satisfying correlation in the recovery. However, there was a clear discrepancy in the amounts of digoxin concentrate in the serum of patients. Only the RIA of Abbott and the EIA of Boehringer showed no significant differences. Particularly noticeable was the tendency towards lower values in the EMIT-technique as well as its liability to unspecific serum changes (lipaemia etc.), which often made the detection of digoxin impossible. The routine use of this technique appears problematic. The need for establishing one's own laboratory and test-specific therapeutical range is pointed out.
Ridefelt, Peter; Hilsted, Linda; Juul, Anders; Hellberg, Dan; Rustad, Pål
Reference intervals are crucial tools aiding clinicians when making medical decisions. However, for children such values often are lacking or incomplete. The present study combines data from separate pediatric reference interval studies of Denmark and Sweden in order to increase sample size and to include also pre-school children who were lacking in the Danish study. Results from two separate studies including 1988 healthy children and adolescents aged 6 months to 18 years of age were merged and recalculated. Eighteen general clinical chemistry components were measured on Abbott and Roche platforms. To facilitate commutability, the NFKK Reference Serum X was used. Age- and gender-specific pediatric reference intervals were defined by calculating 2.5 and 97.5 percentiles. The data generated are primarily applicable to a Nordic population, but could be used by any laboratory if validated for the local patient population.